for ElectrotherapyCompany Profile
02 Message from the President
03 Corporate Philosophy
04 There is a reason why people choose ITO
06 Research & Development
08 Superior Quality & Production
10 ITO & Sports

Products
12 Overview of our lineup
16 EU-941 / EU-921
20 ES-5400 / ES-5200
24 TENS 120ZCONTENTS

26 D function
28 US-751
30 US-101L / US-103S
32 OSTEOTRON Ⅳ
34 RSK-600
36 LAZR-207 / LAZR-215 / LAZR-115
38 DIA-TKR 800
40 SW-1000
42

PM-810+44 TM-400
46 ES-160
48 ES-130
50 EasyAngle
52 Best View Stand

Message from the President

Drawing on an outstanding
history and track record to movetoward a new future in physiotherapy
Since our founding in 1916, based on the clear vision of founder Kenji Ito,
ITO has made steady progress in physiotherapy over a long history of
more than 100 years. Our electric physiotherapy devices have earned astrong reputation for technology, results, and usability, a reputation webelieve is rooted in our focus on contributing to society rather than mere
sales results—another legacy dating from the time of our founder.
In its essence, physiotherapy involves not just technical capabilities, butestablishing an approach founded on evidence-based medicine (EBM). For
this reason, through ongoing joint research with universities and research
institutions, we’ve moved quickly to demonstrate the evidence and backing
for the efficacy of our products. This represents the foundations of ourvision for mastering physiotherapy and a key element in the support
we provide not just to medical institutions, but to the health of general
consumers and conditioning efforts among athletes. With societies growing
increasingly older around the world, we expect to see the sphere of ouractivities continue to expand, based on our grounding in fields in which wehave already achieved significant expertise.
We also plan to continue pursuing research
efforts to ensure comprehensive reliability andquality controls for our products and to develop
new domains for physiotherapy.

Tsukasa Kurahashi, President
2Company Profile

C orporate Philosophy

Spirit of Altruism
Since the foundation in 1916, we havealways kept one corporate philosophy. It
is the“Spirit of Altruism”practiced by
Kenji Ito, the founder of ITO. Altruism isthinking and acting for the sake of others
more than yourself. It is also known as thepractice of Bodhisattva.
“Contribution to society by providing
our devises to more people”. This is our
corporate mission and firm management

Kenji Ito, Founderphilosophy. We dedicate ourselves inpursuit of this our mission forever.

3

There is a reason
why people choose ITO
An international reputation and track record
accumulated over 40 years in 100 countries
Since our founding in 1916, alongside its own research and development, ITO has
sought out and adopted technologies from around the world to design numerous
physiotherapy devices. Drawing on this legacy, we launched full-fledged international
expansion efforts in the 1970s and opened offices in China and Vietnam during the
1990s. Since then, we’
ve rapidly grown our businesses in Asian markets.
In 1996, we began exhibiting in MEDICA, in Dusseldorf, Germany, the world’s largest
medical device trade show. We’ve also exhibited in other leading trade shows aroundthe world, including Arab Health in the UAE and CMEF in China.
Our more than 40 years of activities worldwide
have helped build deep-rooted global trustin ITO. Today, we export our high-quality
products to more than 100 countries.

4Company Profile

A spirit that inspires us tocontinue pursuing challenges
posed by new domains
ITO’s product development history is characterized by an eagerness

Head office relocatedKawaguchi, Saitama,
2018 toJapanto tackle the challenges of miniaturization, sophisticated features, andITO Manufacturing Vietnam

Ltd. established in Hanoi,
2017 Co.,
Vietnamhigh performance. This spirit emerges from our founder’s vision—to
help people live healthier lives by broadening the range of therapy

2016 100th anniversary

devices available for use at home.

US-101L / US-103S, a new styleportable ultrasound therapy
2010 insystems,
launched

Since successfully miniaturizing shortwave therapy devices
about 80 years ago, ITO has continually striven to make ourdevices as compact as possible while retaining the features andperformance of larger therapy devices. Products born fromwins 2005 iF Design
2005 EU-940
Award

our unique expertise, including portable ultrasound therapy
devices, portable interferential electrotherapy devices, andControl Center2004 Reliability
establishedultracompact hand-carried electrostimulator equipment,
have been welcomed with wonder in markets aroundES-420 and ES-160 win 2003 iF

Award
2003 Design
ISO 14001 certification acquired

the world.

2001 ISO 13485 certification acquired

Grounded in this unique approach and offering
advanced features, ITO’s compact high-1999 ISO 9001 certification acquired

performance products are used today
across a wide range of domains, fromrehabilitation to athletic conditioning
and therapy to in-home care.

90th anniversary of founding

US-750 wins 2006 iF
2006 celebrated
Design Award

1998

Hanoi Office opened in Vietnam Trio 300, world’s firstminiature multi-mode stimulator, launched
PM-800 Series, world’s first inverter-controlled
microwave therapy unit, launched

1994 Shanghai Office opened in P.R.China
Japan’s first bone-growth stimulator, launched
1989 Osteotron,
Tsukuba Factory opened

1983 R&D Center opened
name changed to ITO Co., Ltd.
1957 Company
Wide range of electrotherapy equipment developed

1934 Japan’s first shortwave therapy unit, developedand Experiments” monthly magazine, Japan’s first specialist radio technology magazine,
1924 “Radio
first published (Now published by Seibundo Shinkosha publishing company)
Medical Electric Co., Ltd. founded by Kenji Ito
1916 TokyoJapan’s first AC X-ray machine and other devices developed5Research & Development
Developing highly reliable and effective products based on the latest
information and comprehensive EBM

6Company Profile

Advancing joint research withnumerous research institutions
The concept of evidence-based medicine (EBM) has drawn
significant attention in the field of physiotherapy.
Focusing on the significance of EBM, at ITO, we, too, consider
it essential to demonstrate the effectiveness of our products
through scientific and clinical approaches. We constantly
and proactively collect the latest information from academic
associations around the world, including the World Confederation
for Physical Therapy (WCPT). We pursue advanced joint
research with research institutions and physiotherapists in Japanand around the world, incorporating the results into our products.
Our products have also begun to draw attention in dental field,
and research seeking to demonstrate the efficacy of ultrasoundtechnology in dental therapy is advancing rapidly.

Pursuing research and development
through multiple development sections
We established a general R&D section in 1983. We strive to develop
devices that will contribute to society, reflecting societal and corporateneeds identified by sales personnel. In recent years, for example, demandhas grown for compact devices in the field of in-home care. We offer
extensive and proprietary expertise in compact, sophisticated, and highperformance devices with advanced features, a natural result of our
focus on device miniaturization over the years. Building on this trackrecord, we’re striving to develop even more advanced products.

Quality certified underinternational standards
We obtained ISO certification for the first time in 1998.
Since then, we’ve gained certification under numerous
rigorous international standards, including ISO 13485(Medical devices – Quality management systems) and ISO
14001 (Environmental management systems). We remain
dedicated to pursuing new possibilities in physiotherapy
by strengthening not just our technical capabilities, but thereliability of our products. In this way, we seek to continue
delivering products that contribute to society.

7Superior Quality & Production
Producing trusted devices backed by advanced testing andinspection functions

8

Company Profile

One-person cellular production system
improves quality and efficiency
In 2010, we adopted a one-person cellular production system under which
each product is assembled, inspected, and packed by the same individual.
Under this system, which is especially well-suited to producing a large
variety of products in small lots, each worker handles all aspects ofthe production line, strengthening his or her sense of responsibility
and improving skills dramatically. The system has also been highly
successful in terms of work efficiency.

Rigorous quality controls withadvanced testing equipment
To implement even more thorough quality controls, theReliability Control Center established at the Tsukuba Factoryin 2004 is fully equipped with some of the industry’s mostadvanced testing equipment, including intelligent load testing
robots, thermal shock chambers, low temperature thermohygrostats, hydrophones, vibration testers, drop impact testers,
bending testers, micro focus X-ray inspection systems, andphotomicroscopy systems. Before, we had outsourced quality
control; now, with these in-house facilities, we are capable ofapplying maximum loads to each part and implementing evenmore thorough testing. We carry out repeated and rigorous
testing to maintain reliable quality for the physical therapy
devices used not just in medical care, but in homes.

Establishing traceability for thoroughproduct control
Stressing thorough traceability for each and every product, we maintain astructure that makes it possible to precisely ascertain information on every
aspect of production, from the parts used to make a product through theassembly process and the shipping route. In the rare event of a productdefect, this makes it possible to verify the causative factors from multiple
angles. Since this approach can also be used to clearly ascertain information
ranging from production records to the number of years for which eachproduct has been in use, it allows swift response to any problems withproducts used for extended periods.
By delivering high-quality products at all times, we strive to earn the trustof our customers and to reward the high regard in which they hold us.

9ITO & Sports

The ITO brand: A leader inathletic conditioning in JapanFor some 20 years, ITO has pursued efforts to support
treatment and conditioning for athletes throughphysiotherapy. Today, thanks to the conditioning support
we provide in numerous international competitions
and our track record with many professional sports
franchises, including soccer and baseball teams, the ITO
name is known throughout the Japanese sports world.
We’ve been designated an official sponsor or supplier
for more than 30 athletic associations and teams,
including the All Japan Judo Federation, the Japan
Cycling Federation, the Japan Association of Athletics
Federations, and the Japan Wheelchair Tennis
Association.

10

Company Profile

Quality and technologies recognized byathletes worldwide
Today, ITO’s initiatives in support of athletic endeavor extend around the world.
Brazil’s Sao Paulo FC, Chonburi FC in the Thai Premier League, Balmazújvárosi FC
in Hungary professional football clubs have formally adopted ITO’s physiotherapy
devices, which they use to optimize the health and conditioning of their athletes. Our
palm-sized ultrasound devices in particular have earned a sterling reputation forenabling conditioning away from team training centers.
Our treatment devices are widely used not just by sports teams, but hospitals,
rehabilitation centers, and other facilities around the world, supporting athletesseeking to recover quickly from injuries. As requested by the Ministry of Culture,
Sports and Tourism of Vietnam, we donated our devices to support the athletes onthat nation’s national teams. We have also hosted seminars on treatment methods
at the Vietnam Sports Hospital. In these ways, our treatment devices today playmajor roles on the sports scene worldwide.

11O ver view o f o u r lineup

Combination Therapy

EU-941

Multi-channel Electrotherapy /Ultrasound Combo
P.16

EU-921

Multi-channel Electrotherapy /Ultrasound Combo
P.16

Electrotherapy

TENS 120Z
Dual Channel TENS
P.24

ES-5400

Multi-channel Electrotherapy Unit
P.20

ES-5200

Multi-channel Electrotherapy Unit
P.20

12

Electrotherapy for dental use

D function

Multi-current Stim for TMJ
P.26

Ultrasound Therapy

US-751

Multi-frequency Ultrasound
P.28

US-101L / US-103S
Palm-sized Ultrasound
P.30

LIPUS Therapy (for Bone Growth)

OSTEOTRON Ⅳ

Shockwave Therapy

RSK-600

Radial Shockwave
P.34

Ultrasound Bone Growth
Stimulator
P.32

13

O ver view o f o u r lineup

High Power Laser Therapy

LAZR-207 / LAZR-215 / LAZR-115
High Power Laser
P.36

DIA-TKR 800
Radio Frequency
P.38

Shortwave Diathermy

SW-1000

Pulsed & Continuous Shortwave
P.40

14

Radio Frequency Therapy

Microwave Diathermy

PM-810+Microwave Therapy
P.42

Traction Therapy

Electroacupuncture

ES-1606-channel Programmable
Electroacupuncture
P.46

TM-400Powered Traction Unit
P.44

ES-1303-channel Palm-sizedElectroacupuncture
P.48

Digital Goniometer

OthersEasyAngle

Accurate and Single Hand
Operation ROM Measuring Device

Best View Stand
Triangular Tilt Stand
P.52

P.50

15

Combination Therapy

EU-941 / EU-921

Multi-channel Electrotherapy / Ultrasound Combo

EU-941

EU-921

Combination therapy can be applied by providing electrotherapy and ultrasoundtherapy simultaneously, providing ver y efficient therapy.
47 effective clinical programs are installed to our combination unit. You can simply
choose the area to be treated from Human Body Diagram.
The BNR of EU-941 and EU-921 is 2.4 to 4.6 (IEC standards), a significantly low
ratio. Low beam non-uniformity ratio prevents hot spots and tissue damages.
Features4 or 2 independent electrotherapy channels+ 1 ultrasound channel
Low BNR (Beam Non-uniformity Ratio)
47 effective clinical programs with Human Body
Diagram for over 30 types of pathologies
13 types of frequently used current modes installed
Visual probe contact status
210 memorable program numbers
Sequential mode to apply two designatedtherapeutic courses without having to modify unit
parameters

16

Constant current / constant voltage modes
Exclusive vacuum unit available (Optional)
Multi-frequency treatment probes (1 MHz and
3 MHz)
Full-color, 7˝ LCD touch screen
9 languages available (English, German, French,
Italian, Spanish, Portuguese, Turkish, Vietnamese& Chinese)
2 ultrasound probes connectable
Selectable coupling sensitivity for ultrasound gel ormedicated gel

Current Mode

Output mode

Maximum
amplitude (peak)Constant, Sweep

100voltage
mA

Constant, Sweep

100 mA

Waveform image

Frequency

Output voltage

CH1

Outputvoltage

IF-4

Output1–250 HzCH2

Output voltage

Output voltage

CH1

CH1

Outputvoltage
Outputvoltage

Outputvoltage

IF-2

Output voltage

CH2
CH1

CH2
Output voltage

Outputvoltage

Output voltage

CH1

CH1
OutputOutputvoltage

CH2

OutputvoltageCH2
voltage

OutputCH2

voltage

OutputOutput voltage
voltage

OutputOutputvoltagevoltage

EMS

H1
CH1

ut

voltage
Constant,
Sweep, Burst,
Surge Independent, Surge
Co-Cont, Surge Alternate

Continuous

OutputOutputvoltagevoltage

Output voltage

CH1
Output(Alternate(Alternate polarity)
polarity)
Continuous
voltage

Outputvoltage

MF
Outputvoltage

Alternate polarity)

GalvanicDF

Triangularwave
wave
Triangular

Outputermittent
OutputAlternatepolarity)
Alternatepolarity)
voltagevoltage
utput
oltageOutputOutputvoltagevoltage

gular
wave
riangular
wave

CP(Alternate polarity)

CP-ISO

Outputvoltage

Outputvoltage
Rectangular wave
Surge wave

CP-ISO

Outputvoltage

LP
Outputvoltage

Triangular wave
Outputvoltageectangularectangular wave
wave

Rectangular wave

tangular wave

put
tage

Outputvoltage

Outputvoltage

CP

OutputTriangular wave
voltage

Outputvoltage

OutputOutputvoltagevoltageMF

voltage

Triangular wave

Traebert
Rectangular wave

Outputvoltage

Outputvoltage

Surge
Surge wave
wave
Rectangularwave
Triangular wave
Outputvoltage
OutputOutputOutputvoltagevoltagevoltage
Surge wave
OutputOutputvoltage

Outputvoltage
Rectangular wave
Outputvoltage

Outputvoltage

Outputvoltage

Outputvoltage

OutputvoltageRS

DF, voltage
MF, CP, CP-ISO, LP, RSMF

OutputOutputvoltagevoltage

RS

Outputvoltage142 Hz

70 mA

I/T Curve: 0.33–0.5 Hz
AQ: 0.33 Hz

CP
Outputvoltage

RS
RS

Outputvoltage

OutputOutputvoltagevoltage
Surge wave

Constant

CP-ISO
Outputvoltage

LP
Outputvoltage

Outputvoltage

Rectangular wave

70 mA

voltage
OutputOutputvoltagevoltage

Triangular wave

I/T Curve・AQ

50 Hz or 100 Hz

LP
LP
OutputOutputvoltage

Outputvoltage

Outputvoltage

Outputvoltage70mA

Outputvoltage

Triangular wave

Rectangular
Rectangular wave
wave

Outputvoltage

RS

Outputvoltage Output

RSLP

20–250 Hz

LP

Outputvoltage

OutputOutputRectangular wave
voltagevoltageIntermittent

Outputvoltage

70 mA

DF

voltage
CP-ISO
CP-ISO
voltage

CP-ISO

voltage(Alternate
polarity)
Rectangular
wave

RS

OutputOutputCP

Outputvoltage

Surge Independent

Continuous

LP

Outputvoltage

RS

Outputvoltage

LP

RS

Outputvoltage

Outputvoltage

Outputvoltage

RS

Intermittent

DF

MF

Outputvoltage

OutputDiadynamicOutput
voltage

Outputvoltage

Outputvoltagevoltagevoltagevoltage

DF

Outputvoltage

OutputOutput Rectangular wave
voltagevoltage

Intermittent

OutputOutputvoltagevoltage

Outputvoltage

Outputvoltage

OutputOutputOutputvoltagevoltage
OutputOutputOutputCP-ISO

CP-ISO

LP

voltage

OutputOutput CP
CP
voltage voltage

0.95–15.8 Hz

Outputvoltage

Rectangularwave
OutputRS

CP

Output20
mA
voltage

LP

Surge wave

OutputLP

Continuous(Alternate polarity)e wavevoltagewave
ectangularput
Output wave
riangular
riangular
wave
age
voltage
OutputOutputvoltagevoltage

OutputMF
Output MF
voltage voltage

voltage

Rectangularwave
OutputOutputvoltage

OutputOutputvoltage

OutputOutput
voltagevoltage

CP

OutputCP-ISO
voltage

OutputOutputvoltagevoltage

Rectangularwave
voltage
Rectangularwave
Outputvoltage
Outputvoltagewave
Surge
Outputvoltage

OutputOutputvoltagevoltage

Triangularwave
Triangularwave
Triangularwave
Continuous
voltage

Outputvoltage

Outputvoltage(Alternate polarity)(Alternate polarity)

Rectangular wave

Continuous, Interrupted

voltage
Triangular wave

LP

Outputvoltage

Output voltage
voltage

RS

Surge wave

Output wave
Triangular

Faradicput
Outputage
voltagevoltage

0.2–400 Hz

MF

Outputvoltage

Outputvoltage
CP-ISO

voltagevoltage voltage(Alternate
Outputpolarity)

Output

MFCP-ISOLP

Output voltage Output
Output Rectangularvoltagewave
voltage Output
OutputOutputvoltagevoltagevoltage

750 μA

Outputvoltage

Outputvoltage

Outputvoltage

Outputvoltage

OutputCPvoltage

CP-ISO DF
Output Surge wave
DF

Triangularwave
voltage

Outputvoltage

Constant

Triangular voltage
wave

MF Output
Triangular wave

Triangularwave
OutputOutputvoltage

OutputvoltageMF

CPIntermittent
Intermittent

CP
OutputOutputvoltage

OutputDF

CP-ISO

Outputvoltage Intermittent

Outputvoltage

OutputContinuous
Continuous
voltage

Outputvoltage

Outputvoltage

OutputOutputvoltage

Surge wave
OutputOutputRectangularwave
voltagevoltage
OutputOutputOutputvoltagevoltagevoltage

Output voltage

MF
OutputvoltageContinuous

Outputvoltage

CP

Intermittent(Alternate
polarity)
OutputOutputOutputvoltage
Outputvoltagevoltagevoltage

Output voltage

CH2

DF

voltage

Triangularwave
wave
Triangular

0.5–250 Hz

Outputvoltage

Intermittent

DF

100 mACP

Surge
wavevoltage
Output(Alternate polarity)

CH1

Outputvoltagevoltage

Outputvoltage

Outputvoltage

DF

OutputContinuous
voltage
OutputOutputvoltage

MF

OutputvoltageContinuous

MicrocurrentDF
Outputvoltage(MCR)

Outputvoltagevoltage

Outputvoltage

Outputvoltage
Output(Alternate
polarity)
voltage

Intermittent

Surge wave
Rectangularwave
Output0.5–200 Hz

Outputvoltage

Constant, Sweep, Burst,
Outputvoltage
Surge
Independent, Surge
Intermittent
Co-Cont, Surge Alternate
Triangular wave

Intermittent(Alternate polarity)

CH2

600 mAMF

OutputContinuous
voltage

DF

OutputOutputvoltagevoltage
CH2

TENS

ernate polarity)

utput
oltageOutputvoltage

Outputvoltage(Alternate polarity)

CH1

Outputvoltage

Intermittent

utge

OutputOutputoltagevoltageontinuous

DF

Outputvoltage

OutputOutputvoltagevoltage

utput
OutputoltageoltageOutputOutputvoltage

Output(Alternate
polarity)polarity)(Alternate

Outputvoltage
Output(Alternate polarity)
voltage(Alternate polarity)

Output voltage
oltage2.5 kHz

OutputOutputvoltagevoltage

OutputvoltageContinuous

100 mA(Alternate polarity)

OutputOutputvoltagevoltage

Outputvoltage

20–250 Hz

Outputvoltage

Surge Independent, Surge
Co-Cont, Surge Alternate(Alternate polarity)

High Voltage(HV)

Output100voltage
mA

Outputvoltage
Outputvoltage

RussianOutputvoltage

Outputvoltage

OutputOutputvoltagevoltage

OutputOutputvoltagevoltage

utput
oltageOutputvoltage(Alternate polarity)

Outputvoltage

Outputvoltage

1–250 Hz
Outputvoltage

Surge Independent, Surge
Co-Cont, Surge Alternate

OutputOutputvoltagevoltage

H2
CH2
ge

Outputvoltage

Combination Therapy

Current Details

Constant
Rectangular wave
Outputvoltage

RS
Outputvoltage17

Combination Therapy

Specifications
EU-941

EU-921

Power supply

AC 100–240 V, 50/60 Hz

Power consumption

190 VA

140 VA

Number of channels

5 independent(4 for electrotherapy, 1 for ultrasound)

3 independent(2 for electrotherapy, 1 for ultrasound)

Display

Full-color, 7  LCD touch screen

Safety class according to IEC 60601-1

Class I, Type BF

Dimensions

350 (W) × 270 (D) × 145 (H) mm

Weight4 kg

3.5 kgElectrotherapy
IF carrier frequency

2 kHz, 4 kHz, 5 kHz, 8 kHz, 10 kHz

Vector sweep (for IF-4)

0°, 15°, 30°, 45°

User-defined protocol memories

120

Pre-programmed clinical protocols

28

Timer

Max. 60 min.
UltrasoundUltrasound mode

Continuous, Pulsed (5%, 10%, 20%, 30%, 40%, 50%)
Continuous: 2 W/cm2, Pulsed: 3 W/cm2

Ultrasound intensity (max.)
Ultrasound frequency

1 MHz, 3 MHz

Pulse frequency

16 Hz, 48 Hz, 100 Hz

User-defined protocol memories

10 (for US) / 80 (for Combo)

Pre-programmed clinical protocols

18 (for US) / 1 (for Combo)

TimerMax. 30 min.

Number of US Probe connections

2

Probe head diameter[Large] 37.5 mm / [Small] 16 mm

BNR US Probe head (IEC Standards)[Large] 1 MHz: 4.6 / 3 MHz: 2.4[Small] 1 MHz: 2.9 / 3 MHz: 2.4

ERA US Probe head (IEC Standards)[Large] 1 MHz: 5.0 cm2 / 3 MHz: 5.0 cm2[Small] 1 MHz: 0.7 cm2 / 3 MHz: 0.5 cm2

Probe cable length2m

Degree of protection against
harmful ingress of water

IPX7 (US Probe)

Ordering Data
Standard Kit *Standard kit includes main unit.

EU-941

EU-921012418Ultrasound Probe (L)1×1×B180534

Electrode Cable (Brown)

1×

1×

B180535

Electrode Cable (Red)

1×—

B180536

Electrode Cable (Orange)

1×

—
1×

B180537

Electrode Cable (Yellow)

1×011151

Rubber Electrode (M), 60 × 50 mm, 2pcs/pack

4×

2×

011148

Electrode Sponge A (M), 80 × 65 mm, 2pcs/pack

4×

2×011655

Strap (L), 80 × 1200 mm

4×2×

011654

Strap (S), 80 × 600 mm

4×2×

012298

Probe Holder

1×

1×

120612Ultrasound Gel (250 ml)

1×

1×

180562

Power Supply Cord (220–240 V, Type F) or

180566

Power Supply Cord (110–120 V, Type B)

1×

1×

012418011654
B180537
011148B180536120612

Optional Accessories *Available to both EU-941 and EU-921

18012416Ultrasound Probe (S)011152

Rubber Electrode (L), 100 × 60 mm, 2pcs/pack

011150

Rubber Electrode (S), 50 × 30 mm, 2pcs/pack

011149

Electrode Sponge A (L), 120 × 80 mm, 2pcs/pack

011147

Electrode Sponge A (S), 70 × 45 mm, 2pcs/pack010306

Self-adhesive Electrode, 49 × 49 mm, 4pcs/pack

010747

Self-adhesive Electrode, 89 × 51 mm, 4pcs/pack

010889

Self-adhesive Electrode, ø32, 4pcs/pack

011356

HV/DC Probe

011172

MCR (Microcurrent) Probe, 2pcs/pack—Vacuum Unit (Please see next page.)012298011655

B180534

B180535

011151011356011172010306012416011147011149011152 011150 010889

010747

Combination Therapy

SU-540 / SU-520<BK>
Blow-out System Vacuum Unit

SU-540 with EU-941

SU-520<BK> with EU-921

SU-540 for EU-941 and SU-520<BK> for EU-921, both models adopt blow-out
vacuum system, and therefore no water reservoir is needed. It is virtually
maintenance-free, since dust or moisture does not adhere inside the tube or on theconnections, resulting in preventing blockages and oxidation.
Specifications
Power supply

SU-540SU-520 <BK>AC 100–240 V, 50/60 Hz

AC 110, 120 or 220–240 V, 50/60 Hz

150 VA110 and 120 V: 70 VA
220–240 V: 90 VA

Power consumption
Number of channels
Dimensions

4 independent

2 independent

345 (W) × 315 (D) × 115 (H) mm

345 (W) × 267 (D) × 116 (H) mm

Weight8.5 kg

8 kg

Suction pressure

0 to -100 mmHg

Ordering Data
Standard Kit *Standard kit includes main unit.

SU-540SU-520 <BK>

011771

Electrode Hose (Blue)

2×

1×011772

Electrode Hose (Gray)

2×

1×012399Vacuum Electrode B (S), ø80 (Gray)

4×

2×012400Vacuum Electrode B (S), ø80 (Blue)

4×

2×011276

Electrode Sponge B (S), ø70, 4pcs/pack

4×2×180562

Power Supply Cord (220–240 V, Type F) or

180566

Power Supply Cord (110–120 V, Type B)

1×

1×

011276012400012399011771011772

Optional Accessories *Available to both SU-540 and SU-520 <BK>
012401

Vacuum Electrode B (L), ø100 (Gray)

012402

Vacuum Electrode B (L), ø100 (Blue)

120974

Suction Cup (L), ø100 (Gray) (cup only)

120975

Suction Cup (L), ø100 (Blue) (cup only)

120976

Suction Cup (S), ø80 (Gray) (cup only)

120977

Suction Cup (S), ø80 (Blue) (cup only)

151157

Suction Steel Plate (L), ø55

151083

Suction Steel Plate (S), ø45121359Suction Head

121360

Suction Head Cap

011277

Electrode Sponge B (L), ø90, 4pcs/pack220278Paper Disk (L), ø90, 100pcs/pack220279Paper Disk (S), ø70, 100pcs/pack

012401012402120974120975011277220279151157 151083

12135912097612136022027812097719

Electrotherapy

ES-5400 / ES-5200
Multi-channel Electrotherapy Unit

ES-5400

ES-5200

Electrotherapy is a treatment that promotes or represses vital reactions through theapplication of electric stimulation to the neuromuscular system or sensor y ner ves.
Intended use is to reduce pain, improve muscle strength, increase joint mobility,
suppress spasms, promote blood circulation and reduce the severity of edema.
28 pre-programmed therapy parameters are installed in ES-5400 and ES-5200 for
over 20 types of typical pathologies. You can simply choose the area to be treatedfrom Human Body Diagram.
Features4 or 2 independent electrotherapy channels
28 effective clinical programs with Human Body
Diagram
13 types of frequently used current modes installed
Sequential mode to apply two designatedtherapeutic courses without having to modify unit
parameters

20

Exclusive vacuum unit available (optional)
Constant current / constant voltage modes
Full-color, 7˝ LCD touch screen
120 memorable program numbers
9 languages available (English, German, French,
Italian, Spanish, Portuguese, Turkish, Vietnamese& Chinese)Current Mode

Output mode

Maximum
amplitude (peak)Constant, Sweep

100voltage
mA

Constant, Sweep

100 mA

Waveform image

Frequency

Output voltage

CH1

Outputvoltage

IF-4

Output1–250 HzCH2

Output voltage

Output voltage

CH1

CH1

Outputvoltage
Outputvoltage

Outputvoltage

IF-2

Output voltage

CH2
CH1

CH2
Output voltage

Outputvoltage

Output voltage

CH1

CH1
OutputOutputvoltage

CH2

OutputvoltageCH2
voltage

OutputCH2

voltage

OutputOutput voltage
voltage

OutputOutputvoltagevoltage

EMS

H1
CH1

ut

voltage
Constant,
Sweep, Burst,
Surge Independent, Surge
Co-Cont, Surge Alternate

Continuous

OutputOutputvoltagevoltage

Output voltage

CH1
Output(Alternate(Alternate polarity)
polarity)
Continuous
voltage

Outputvoltage

MF
Outputvoltage

Alternate polarity)

GalvanicDF

Triangularwave
wave
Triangular

Outputermittent
OutputAlternatepolarity)
Alternatepolarity)
voltagevoltage
utput
oltageOutputOutputvoltagevoltage

gular
wave
riangular
wave

CP(Alternate polarity)

CP-ISO

Outputvoltage

Outputvoltage
Rectangular wave
Surge wave

CP-ISO

Outputvoltage

LP
Outputvoltage

Triangular wave
Outputvoltageectangularectangular wave
wave

Rectangular wave

tangular wave

put
tage

Outputvoltage

Outputvoltage

CP

OutputTriangular wave
voltage

Outputvoltage

OutputOutputvoltagevoltageMF

voltage

Triangular wave

Traebert
Rectangular wave

Outputvoltage

Outputvoltage

Surge
Surge wave
wave
Rectangularwave
Triangular wave
Outputvoltage
OutputOutputOutputvoltagevoltagevoltage
Surge wave
OutputOutputvoltage

Outputvoltage
Rectangular wave
Outputvoltage

Outputvoltage

Outputvoltage

Outputvoltage

OutputvoltageRS

DF, voltage
MF, CP, CP-ISO, LP, RSMF

OutputOutputvoltagevoltage

RS

Outputvoltage142 Hz

70 mA

I/T Curve: 0.33–0.5 Hz
AQ: 0.33 Hz

CP
Outputvoltage

RS
RS

Outputvoltage

OutputOutputvoltagevoltage
Surge wave

Constant

CP-ISO
Outputvoltage

LP
Outputvoltage

Outputvoltage

Rectangular wave

70 mA

voltage
OutputOutputvoltagevoltage

Triangular wave

I/T Curve・AQ

50 Hz or 100 Hz

LP
LP
OutputOutputvoltage

Outputvoltage

Outputvoltage

Outputvoltage70mA

Outputvoltage

Triangular wave

Rectangular
Rectangular wave
wave

Outputvoltage

RS

Outputvoltage Output

RSLP

20–250 Hz

LP

Outputvoltage

OutputOutputRectangular wave
voltagevoltageIntermittent

Outputvoltage

70 mA

DF

voltage
CP-ISO
CP-ISO
voltage

CP-ISO

voltage(Alternate
polarity)
Rectangular
wave

RS

OutputOutputCP

Outputvoltage

Surge Independent

Continuous

LP

Outputvoltage

RS

Outputvoltage

LP

RS

Outputvoltage

Outputvoltage

Outputvoltage

RS

Intermittent

DF

MF

Outputvoltage

OutputDiadynamicOutput
voltage

Outputvoltage

Outputvoltagevoltagevoltagevoltage

DF

Outputvoltage

OutputOutput Rectangular wave
voltagevoltage

Intermittent

OutputOutputvoltagevoltage

Outputvoltage

Outputvoltage

OutputOutputOutputvoltagevoltage
OutputOutputOutputCP-ISO

CP-ISO

LP

voltage

OutputOutput CP
CP
voltage voltage

0.95–15.8 Hz

Outputvoltage

Rectangularwave
OutputRS

CP

Output20
mA
voltage

LP

Surge wave

OutputLP

Continuous(Alternate polarity)e wavevoltagewave
ectangularput
Output wave
riangular
riangular
wave
age
voltage
OutputOutputvoltagevoltage

OutputMF
Output MF
voltage voltage

voltage

Rectangularwave
OutputOutputvoltage

OutputOutputvoltage

OutputOutput
voltagevoltage

CP

OutputCP-ISO
voltage

OutputOutputvoltagevoltage

Rectangularwave
voltage
Rectangularwave
Outputvoltage
Outputvoltagewave
Surge
Outputvoltage

OutputOutputvoltagevoltage

Triangularwave
Triangularwave
Triangularwave
Continuous
voltage

Outputvoltage

Outputvoltage(Alternate polarity)(Alternate polarity)

Rectangular wave

Continuous, Interrupted

voltage
Triangular wave

LP

Outputvoltage

Output voltage
voltage

RS

Surge wave

Output wave
Triangular

Faradicput
Outputage
voltagevoltage

0.2–400 Hz

MF

Outputvoltage

Outputvoltage
CP-ISO

voltagevoltage voltage(Alternate
Outputpolarity)

Output

MFCP-ISOLP

Output voltage Output
Output Rectangularvoltagewave
voltage Output
OutputOutputvoltagevoltagevoltage

750 μA

Outputvoltage

Outputvoltage

Outputvoltage

Outputvoltage

OutputCPvoltage

CP-ISO DF
Output Surge wave
DF

Triangularwave
voltage

Outputvoltage

Constant

Triangular voltage
wave

MF Output
Triangular wave

Triangularwave
OutputOutputvoltage

OutputvoltageMF

CPIntermittent
Intermittent

CP
OutputOutputvoltage

OutputDF

CP-ISO

Outputvoltage Intermittent

Outputvoltage

OutputContinuous
Continuous
voltage

Outputvoltage

Outputvoltage

OutputOutputvoltage

Surge wave
OutputOutputRectangularwave
voltagevoltage
OutputOutputOutputvoltagevoltagevoltage

Output voltage

MF
OutputvoltageContinuous

Outputvoltage

CP

Intermittent(Alternate
polarity)
OutputOutputOutputvoltage
Outputvoltagevoltagevoltage

Output voltage

CH2

DF

voltage

Triangularwave
wave
Triangular

0.5–250 Hz

Outputvoltage

Intermittent

DF

100 mACP

Surge
wavevoltage
Output(Alternate polarity)

CH1

Outputvoltagevoltage

Outputvoltage

Outputvoltage

DF

OutputContinuous
voltage
OutputOutputvoltage

MF

OutputvoltageContinuous

MicrocurrentDF
Outputvoltage(MCR)

Outputvoltagevoltage

Outputvoltage

Outputvoltage
Output(Alternate
polarity)
voltage

Intermittent

Surge wave
Rectangularwave
Output0.5–200 Hz

Outputvoltage

Constant, Sweep, Burst,
Outputvoltage
Surge
Independent, Surge
Intermittent
Co-Cont, Surge Alternate
Triangular wave

Intermittent(Alternate polarity)

CH2

600 mAMF

OutputContinuous
voltage

DF

OutputOutputvoltagevoltage
CH2

TENS

ernate polarity)

utput
oltageOutputvoltage

Outputvoltage(Alternate polarity)

CH1

Outputvoltage

Intermittent

utge

OutputOutputoltagevoltageontinuous

DF

Outputvoltage

OutputOutputvoltagevoltage

utput
OutputoltageoltageOutputOutputvoltage

Output(Alternate
polarity)polarity)(Alternate

Outputvoltage
Output(Alternate polarity)
voltage(Alternate polarity)

Output voltage
oltage2.5 kHz

OutputOutputvoltagevoltage

OutputvoltageContinuous

100 mA(Alternate polarity)

OutputOutputvoltagevoltage

Outputvoltage

20–250 Hz

Outputvoltage

Surge Independent, Surge
Co-Cont, Surge Alternate(Alternate polarity)

High Voltage(HV)

Output100voltage
mA

Outputvoltage
Outputvoltage

RussianOutputvoltage

Outputvoltage

OutputOutputvoltagevoltage

OutputOutputvoltagevoltage

utput
oltageOutputvoltage(Alternate polarity)

Outputvoltage

Outputvoltage

1–250 Hz
Outputvoltage

Surge Independent, Surge
Co-Cont, Surge Alternate

OutputOutputvoltagevoltage

H2
CH2
ge

Outputvoltage

Electrotherapy

Current Details

Constant
Rectangular wave
Outputvoltage

RS
Outputvoltage21

Electrotherapy

Specifications
ES-5400

EU-5200

Power supply

AC 100–240 V, 50/60 Hz

Power consumption

150 VA

Number of channels

4

100 VA
2

Display

Full-color, 7  LCD touch screen

Safety class according to IEC 60601-1

Class I, Type BF

Dimensions

345 (W) × 270 (D) × 145 (H) mm

Weight2.5 kg

2.3 kgIF carrier frequency

2 kHz, 4 kHz, 5 kHz, 8 kHz, 10 kHz

Vector sweep (for IF-4)

0°, 15°, 30°, 45°

User-defined protocol memories

120

Pre-programmed clinical protocols

28

Timer

Max. 60 min.

Ordering Data
Standard Kit *Standard kit includes main unit.

ES-5400

ES-5200

B180534

Electrode Cable (Brown)

1×

1×

B180535

Electrode Cable (Red)

1×

1×
—

B180536

Electrode Cable (Orange)

1×

B180537

Electrode Cable (Yellow)

1×—011151

Rubber Electrode (M), 60 × 50 mm, 2pcs/pack

4×

2×

011148

Electrode Sponge A (M), 80 × 65 mm, 2pcs/pack

4×

2×011655

Strap (L), 80 × 1200 mm

4×2×

011654

Strap (S), 80 × 600 mm

4×

2×

1×

1×

180562

Power Supply Cord (220–240 V, Type F) or

180566

Power Supply Cord (110–120 V, Type B)

011655

B180536

22

Rubber Electrode (L), 100 × 60 mm, 2pcs/pack

011150

Rubber Electrode (S), 50 × 30 mm, 2pcs/pack

011149

Electrode Sponge A (L), 120 × 80 mm, 2pcs/pack

011147

Electrode Sponge A (S), 70 × 45 mm, 2pcs/pack010306

Self-adhesive Electrode, 49 × 49 mm, 4pcs/pack

010747

Self-adhesive Electrode, 89 × 51 mm, 4pcs/pack

010889

Self-adhesive Electrode, ø32, 4pcs/pack

011356

HV/DC Probe

011172

MCR (Microcurrent) Probe, 2pcs/pack—Vacuum Unit (Please see next page.)

011148B180537

011151
B180534
B180535

Optional Accessories *Available to both ES-5400 and ES-5200
011152011654011356011172011149010306011147011152011150 010889

010747Electrotherapy

SU-540 / SU-520<BK>
Blow-out System Vacuum Unit

SU-540 with ES-5400

SU-520<BK> with ES-5200

SU-540 for ES-5400 and SU-520<BK> for ES-5200, both models adopt blow-out
vacuum system, and therefore no water reservoir is needed. It is virtually
maintenance-free, since dust or moisture does not adhere inside the tube or on theconnections, resulting in preventing blockages and oxidation.
Specifications
Power supply

SU-540SU-520 <BK>AC 100–240 V, 50/60 Hz

AC 110, 120 or 220–240 V, 50/60 Hz

150 VA110 and 120 V: 70 VA
220–240 V: 90 VA

Power consumption
Number of channels
Dimensions

4 independent

2 independent

345 (W) × 315 (D) × 115 (H) mm

345 (W) × 267 (D) × 116 (H) mm

Weight8.5 kg

8 kg

Suction pressure

0 to -100 mmHg

Ordering Data
Standard Kit *Standard kit includes main unit.

SU-540SU-520 <BK>

011771

Electrode Hose (Blue)

2×

1×011772

Electrode Hose (Gray)

2×

1×012399Vacuum Electrode B (S), ø80 (Gray)

4×

2×012400Vacuum Electrode B (S), ø80 (Blue)

4×

2×011276

Electrode Sponge B (S), ø70, 4pcs/pack

4×2×180562

Power Supply Cord (220–240 V, Type F) or

180566

Power Supply Cord (110–120 V, Type B)

1×

1×

011276012400012399011771011772

Optional Accessories *Available to both SU-540 and SU-520 <BK>
012401

Vacuum Electrode B (L), ø100 (Gray)

012402

Vacuum Electrode B (L), ø100 (Blue)

120974

Suction Cup (L), ø100 (Gray) (cup only)

120975

Suction Cup (L), ø100 (Blue) (cup only)

120976

Suction Cup (S), ø80 (Gray) (cup only)

120977

Suction Cup (S), ø80 (Blue) (cup only)

151157

Suction Steel Plate (L), ø55

151083

Suction Steel Plate (S), ø45121359Suction Head

121360

Suction Head Cap

011277

Electrode Sponge B (L), ø90, 4pcs/pack220278Paper Disk (L), ø90, 100pcs/pack220279Paper Disk (S), ø70, 100pcs/pack

012401012402120974120975011277220279151157 151083

12135912097612136022027812097723

Electrotherapy

TENS 120Z
Dual Channel TENSTENS 120Z has been sold for over 30 years. As the weight of the unit is less than
200 g and the unit size is ver y compact, you can easily bring it anywhere you like.
You can fix parameters by using the safe-lock lever for safe treatment.
Features
Portable size and weight
Safe-lock lever for control knobs for safe treatment
Longtime seller

24Power supply

Electrotherapy

Specifications
DC 9 V

Number of channels

2

Safety class according to IEC 60601-1

Internally powered equipment, Type BF

Dimensions

61 (W) × 27 (D) × 96 (H) mm

Weight

120 g (without a battery)

Current mode

TENS

Output mode

Constant, Burst, Modulation

Maximum amplitude

20 mArms

Frequency

2–200 Hz

Ordering Data
Standard Kit *Standard kit includes main unit.

Optional Accessories

B120671

Rubber Electrode, 40 × 41 mm, 2pcs/pack, 2×

011148

Electrode Sponge A (M), 80 × 65 mm, 2pcs/pack8003220

Gel Pad, 40 × 41 mm, 4pcs/pack010306

Self-adhesive Electrode, 49 × 49 mm, 4pcs/pack

B180376

Electrode Cable, 2×010747

Self-adhesive Electrode, 89 × 51 mm, 4pcs/pack260182Carrying Bag

010889

Self-adhesive Electrode, ø32, 4pcs/packB010807

Strap ( L), 45 × 1200 mm

010820Strap (S), 45 × 600 mm

NOTE: A battery is not included.

010306
B180376

2601820108898003220010747010820011148B120671

B01080725

Electrotherapy for dental use

D function
Multi-current Stim for TMJ

Electrical stimulation is helpful to maintain and control the balance of the tension ofner ves and muscles around the head and neck. It allows to effectively relieve pain
associated with TMJ (temporomandibular joint).
D function is dual channel electrical muscle stimulator which is equipped withfunctional electric waveforms and programs to match the multiple needs for TMJ
treatment.
Features
Safe default output intensity for applications to thesensitive regions, such as head and neckSpecifically developed single-use electrode forhygienic concern
2 channels with flexible control settings to treat
lateral and bilateral TMJ

26

The handy size ensures no restrictions in usage
environment
Also effective when combined with other clinicalapplications, such as splint therapy
Wide range of treatment options to match themultiple needs for TMJ treatment

Power supply

DC 7.4 V (battery) / DC 12 V (AC Adaptor)

Power consumption

30 VA

Number of channels

2

Display

Full-color, 3.5  LCD touch screen

Safety class according to IEC 60601-1

Class II / Internally powered equipment, Type BF

Dimensions

84 (W) × 23.5 (D) × 151 (H) mm

Weight

230 g (including a battery)

Treatment mode

CARE, PAIN, MCR

Output mode

Surge, Sweep, Burst, Constant

Maximum amplitude (peak)

80 mA

Frequency

CARE: 0.67 Hz (fixed), PAIN: 200 Hz (max.), MCR: 400 Hz (max.)

Timer

Max. 60 min. (Max. 480 min. in MCR mode)

Ordering Data
Standard Kit *Standard kit includes main unit.
012954

D function pad, 8 pcs/pack, 5×180531

Electrode cable (1.15 m, Type A, black)

180530Electrode cable (1.15 m, Type A, gray)080767

AC adaptor

180885

Power cable012953

Lithium ion battery *Installed in main unit before shipping

260724

Soft case

180531012954080767180885180530Electrotherapy for dental use

Specifications

012953260724Treatment Images

TMJ and Masseter muscles

TMJ and Cervical spine

27

Ultrasound Therapy

US-751

Multi-frequency UltrasoundUltrasound therapy is based on sonic energy inaudible to the human ear which
generates micro vibration in body tissues. This vibration contributes to pain relief,
localized increase in blood flow and stimulation of tissue repair. Furthermore,
ultrasound generates thermal changes in the tissues which help to reduce muscle
spasms and decrease joint contractures.
The BNR of US-751 is 2.4 to 3.0 (IEC standards), a significantly low ratio. Low beam
non-uniformity ratio prevents hot spots and tissue damages. You can simply choose
the area to be treated from Human Body Diagram. 42 parameters are pre-set. All the
pre-set parameters can be modified to suit your particular needs.
FeaturesFull-color, 7˝ LCD touch screen
Low BNR (Beam Non-uniformity Ratio)
42 pre-set parameters with human body diagram
Visual probe contact status

28

Selectable coupling sensitivity
8 languages available (English, German, French,
Spanish, Portuguese, Turkish, Swedish & Chinese)
2 ultrasound probes connectable

Power supply

AC 100–240 V, 50/60 Hz

Power consumption

85 VA

Number of channels

1

Display

Full-color, 7  LCD touch screen

Safety class according to IEC 60601-1

Class I, Type BF

Dimensions

290 (W) × 233 (D) × 96 (H) mm

Weight3 kg

Ultrasound mode

Continuous, Pulsed (5%, 10%, 20%, 30%, 40%, 50%)
Continuous: 2 W/cm2, Pulsed: 3 W/cm2

Ultrasound intensity (max.)
Ultrasound frequency

1 MHz, 3 MHz

Pulse frequency

16 Hz, 48 Hz, 100 Hz

Pre-set parameters

42

Timer

Max. 30 min.

Number of US Probe connections

2

Probe head diameter[Large] 37.5 mm / [Small] 16 mm

BNR US Probe head (IEC Standards)[Large] 1 MHz: 3.0 / 3 MHz: 2.4[Small] 1 MHz: 2.9 / 3 MHz: 2.4

ERA US Probe head (IEC Standards)[Large] 1 MHz: 5.0 cm2 / 3 MHz: 5.0 cm2[Small] 1 MHz: 0.7 cm2 / 3 MHz: 0.5 cm2

Probe cable length

Ultrasound Therapy

Specifications

2m

Degree of protection against
harmful ingress of water

IPX7 (US Probe)

Ordering Data
Standard Kit *Standard kit includes main unit.

Optional Accessories

012297Ultrasound Probe (L) or

012296Ultrasound Probe (S)012330Ultrasound Probe (L) (Canada version)

012329Ultrasound Probe (S) (Canada version)

120612Ultrasound Gel (250 ml)

012298

Probe Holder

080611

Core Filter

180672

Power Supply Cord (220–240 V, Type F) or

180676

Power Supply Cord (110–120 V, Type A) or

180673

Power Supply Cord (110–120 V, only North America, Type B)

01229601229801229712061208061129

Ultrasound Therapy

US-101L / US-103S
Palm-sized UltrasoundUltrasound treatment is available in both acute and chronic phases for deep andsuperficial tissues. Difficult body parts to be treated such as toes or fingers can be treatedby immersing a probe and the body part in water.
The BNR of US-101L / US-103S is 2.4 to 3.5 (IEC standards), a significantly low ratio.
Low beam non-uniformity ratio prevents hot spots and tissue damages. As the weightof these units is less than 200 g and the unit size is very compact, you can easily bringthem anywhere you like. Furthermore, it is possible to use these models in the areas withunstable supply of electricity if you charge battery in advance.
Features
Easily portable
One-hand operation
Waterproof design
Low BNR (Beam Non-uniformity Ratio)
Rechargeable battery available (optional)
5 pre-set parameters installed (modifiable)

US-101L30

US-103S

US-101L
Power supply

US-103S

AC 100–240 V, 50/60 Hz or rechargeable battery (optional)

Power consumption

50 VA

17 VA

Display

1.44  color LCD

Safety class according to IEC 60601-1
Dimensions

Class I / Internally powered equipment, Type BF
134 (H) × 59 (W) × 55 (D) mm

134 (H) × 59 (W) × 60.5 (D) mm

Weight

200 g

190 g

Ultrasound mode

Continuous, Pulsed (5%, 10%, 20%, 30%, 40%, 50%)
Continuous: 2 W/cm2, Pulsed: 3 W/cm2

Ultrasound intensity (max.)
Ultrasound frequency

1 MHz

3 MHz

Pulse frequency

100 HzPre-set parameters

5

Timer

Max. 30 min.

Probe head diameter

37.5 mm

16 mm

BNR US Probe head (IEC Standards)

3.5

2.4

ERA US Probe head (IEC Standards)

5.0 cm2

0.5 cm2

Probe cable length

Ultrasound Therapy

Specifications

1.5 m

Degree of protection against
harmful ingress of water

IPX7 (except for Intermediate box)

Ordering Data
Standard Kit *Standard kit includes main unit.

Optional Accessories

US-101L

012608Battery Charger

012055

Protective Cap for US-101L Probe

012061

Rechargeable Battery (except for North America) or

120612Ultrasound Gel (250 ml)

012173

Rechargeable Battery (only North America)

012606AC Adaptor (100−240 V)

012062

Battery Case

012057

Power Supply Cord (220−240 V, Type F) or

012063

Unit Stand

180676

Power Supply Cord(110−120 V, except for North America, Type B) or

012064Pouch

180673

Power Supply Cord (110−120 V, only North America, Type B)

012608*This cord uses along with an AC Adaptor.
US-103S
012058

Protective Cap for US-103S Probe

120612Ultrasound Gel (250 ml)

012606AC Adaptor (100−240 V)

012057

Power Supply Cord (220−240 V, Type F) or

180676

Power Supply Cord(110−120 V, except for North America, Type B) or

180673

Power Supply Cord (110−120 V, only North America, Type B)

012061012062012063012064*This cord uses along with an AC Adaptor.

01260601205501205812061231

LIPUS Therapy (for Bone Growth)

OSTEOTRON Ⅳ
Ultrasound Bone Growth Stimulator

The sound pressure stimulation provided by the LIPUS (Low Intensity Pulsed Ultrasound)system accelerates the formation and re-union of fractured bone, and contributes to rapid
recovery. LIPUS therapy is also effective to fresh fracture, delayed unions and nonunion.
As OSTEOTRON IV has 2 output channels, simultaneous therapy for multiple areas isavailable. Only three-step output control to start treatment. You can easily bring the unit
anywhere you like as the weight of main unit is approx. 300 g and the unit size is very
compact.
Features
2 output channels
Low BNR (Beam Non-uniformity Ratio)
Dual frequency (1.5 MHz and 750 kHz)Easy probe attachment with exclusive probe retainers
Easily portable
AC or battery operation is available
Simple operation (Only three-step output control to start treatment)
Easy to carry with exclusive carrying case32

Power supply

LIPUS Therapy (for Bone Growth)Specifications
DC 4.8 V (battery) / DC 5 V (AC Adaptor)

Power consumption

8 VA

Number of channels

2

Display

3  LCD

Safety class according to IEC 60601-1

Class I / Internally powered equipment, Type BF

Dimensions

98 (W) × 40 (D) × 145 (H) mm

Weight

240 g (without batteries)

Ultrasound mode

Pulsed (20% fixed)
30 mW/cm2, 45 mW/cm2, 60 mW/cm2 (SATA)

Ultrasound intensity (max.)
Ultrasound frequency

1.5 MHz, 750 kHz (optional)

Pulse frequency

100 Hz, 1000 Hz

Timer

20 min., 30 min.

Probe head diameter

32 mm

BNR US Probe head (IEC Standards)[1.5 MHz] 3.5[750 kHz] 3.0

ERA US Probe head (IEC Standards)[1.5 MHz] 3.9 cm2[750 kHz] 3.5 cm2

Probe cable length

0.5 m

Degree of protection against
harmful ingress of water

IPX7 (US Probe)

Ordering Data
Standard Kit *Standard kit includes main unit.

Optional Accessories

012096Ultrasound Probe 1.5 MHz, 2×

012101

Ultrasound Probe 750 kHz

012097

Probe Retainer with 700 mm Belt, for use with Cast

012102

Probe Retainer with 250 mm Belt, for use with Cast

012098

Probe Retainer with 450 mm Belt, for use with Cast

012103Extension Belt, 600 mm

012099Probe Retainer with 470 mm Belt, for use without Cast

012104

Solid Gel Pad , 8pcs/sheet

012609AC Adaptor (100−240 V)

012057

Power Cable for AC Adaptor (220−240 V, Type F)

120612Ultrasound Gel (250 ml)

260700Carrying Bag

012101012102012104

NOTE: Nickel metal hydride rechargeable batteries and a charger are notincluded. Purchase them locally.
01209701210301209801209912061201209601260926070033

Shockwave Therapy

RSK-600
Radial Shockwave

Shockwave therapy is mainly used for the treatment of muscles, tendons, bones and soft
tissues. Radial shockwaves are high energy acoustic waves. The kinetic energy of the
projectile, created by the electromagnetic generator, is transferred to skin through theapplicator of the handpiece. The collision generates shockwaves which diffuse, expanding
radially, through skin and the first layer of tissue below.
The innovative electromagnetic generator of RSK-600 guarantees to generate up to
5,000,000 shots. The model contains a large interactive database containing over 60
pathologies which guide the operator through the treatment. The model does not need an
air compressor as it applies an electromagnetic system. Because of its compact size, thedevice does not occupy much space.
Features5 million shots guaranteed
Clinical protocols for over 60 pathologies
3D protocols by phases
Innovative swing mode
Unique pulsed “ultra soft” shockwave

34

6 languages available (English, German, French,
Spanish, Italian & Russian)Compact body for space saving
Easy to carry with exclusive carrying casePower supply

AC 100–240 V, 50/60 Hz

Power consumption

250 VA

Display

Full-color, 5.7  LCD touch screen

Safety class according to IEC 60601-1

Class I, Type B

Dimensions

320 (W) × 245 (D) × 130 (H) mm
2 kg (main unit)
0.95 kg (handpiece)

WeightOutput mode

Continuous, Burst, Swing

Output power

50–200 mJ (almost equivalent to 1–5 bar)
*Adjustable in step of 10 mJ

Frequency

1–25 HzSpecific protocols for

Bio-stimulant, Analgesic, Anti-inflammatory, Anti-edema, Tension Relief

Pre-programmed clinical protocols

For over 60 pathologies

Shockwave technology

Compressor–free ballistic radial shockwave with electromagnetic generator

Shockwave Therapy

Specifications

Ordering Data
Standard Kit *Standard kit includes main unit.
8004461

Handpiece

8004462

Applicator ø68004463

Applicator ø158004464

Applicator ø248004467

Foot switch

8004468

Silicone cap

8004472

Caliber

8004473

Holder8004424

Bag

80044618004462800446880044638004464800447280044678004473800442435

High Power Laser Therapy

LAZR-207 / LAZR-215 / LAZR-115
High Power Laser

An advantage of laser therapy is its ability to work at deep area, and to resolve the cause of
the pathology at its point of origin. This characteristic can be attributed to two propertiesof the laser: wavelength and power. The emission power also increases the effectivenessof the laser, transmitting the beneficial effects to great depth. Different wavelengths havedifferent properties in terms of diffusion and absorption by human tissues.
As LAZR series has two wavelength modes, you can select suitable one for eachtherapeutic objective. Pathology Library includes over 60 pathologies with relativeinteractive protocols, sub-divided by phase.
Features
The 2 wavelength modes, COMBINATION(810 + 980 nm) and SINGLE (1064 nm)
Effective 7 emission modes (Pulsed, Single Pulse,
Continuous Wave, Custom, AntInf, Burst and E2C)
Innovative patented “E2C” emission mode installed

36

Clinical protocols for over 60 pathologies
Specific protocols for 5 main effects
6 languages available (English, German, French,
Spanish, Italian & Russian)Easy to carry with exclusive carrying case

LAZR-207
Power supply

LAZR-215

LAZR-115

AC 100–240 V, 50/60 Hz

Power consumption

160 VA

Display

Full-color, 5.7  LCD touch screen

Safety class according to IEC 60601-1

Class I, Type B

Dimensions

320 (W) × 245 (D) × 130 (H) mm

Weight3 kg

Wavelength

810 nm + 980 nm

Guide light1064 nm

650 nm (wavelength) / 3 mW (power)

Laser power

up to 7 W

up to 15 W

Continuous (CW), E2C, AntInf, Pulsed, Single Pulse, Burst, Custom

Emission mode
Operation mode

Joule, Timer, Trigger Point

Specific protocols for

Bio-stimulant, Analgesic, Anti-inflammatory, Anti-edema, Tension Relief

Pre-programmed clinical protocols

For over 60 pathologies

Laser class

IV

High Power Laser Therapy

Specifications

Ordering Data
Standard Kit *Standard kit includes main unit.
8004442

Applicator ø40

8004441

Applicator ø30

8004447

Foot switch

8004448

Foot switch with Cover

8004444

Laser Warning Sign

8004450

Meter

8004446

Safety Glasses, 2×8004443

3 Pole Interlock

8004424

Bag

80044428004448800444180044448004447

8004450

800444680044438004424

37

Radio Frequency Therapy

DIA-TKR 800
Radio Frequency

We use innovative Switching Real Time technology for DIA-TKR 800: thanks to thisimportant step forward in technology, it is possible to reach high power and increased
therapeutic performance. Switching Real Time technology has also allowed the dimensionsof the device to be reduced.
DIA-TKR 800 is compact, high power, and easily transportable. This important innovation
has been made possible by the know-how which we have gained over the years, designingand building the components of its machines according to therapeutic results. The modelfeatures a single handpiece for both capacitive and resistive modes. With one simple
command, it is possible to change from one mode to the other: simplicity and functionality
in cutting-edge therapeutic technology.
Features
Single handpiece for both Capacitive and Resistive modes
AV mode (Automatic Mode and Automatic Pulsed Mode)
Clinical protocols for over 60 pathologies
Bio-transfer system
Adjustable pulsed mode for effective thermal control
6 languages available (English, German, French, Spanish, Italian & Russian)Easy to carry with exclusive carrying case

38

Single Application HandpiecePower supply

AC 100–240 V, 50/60 Hz

Power consumption

160 VA

Display

Full-color, 5.7  LCD touch screen

Safety class according to IEC 60601-1

Class I, Type BF

Dimensions

320 (W) × 245 (D) × 130 (H) mm

Weight

1.8 kgOutput power

120 W effective (350 Wpp)

Operation mode

Joule, Timer, Adjustable Pulsed, AV (Automatic, Automatic Pulsed, Manual)Frequency

470–560 kHz

Emission mode

Capacitive and Resistive

Pre-programmed clinical protocols

For over 60 pathologies

Ordering Data
Standard Kit *Standard kit includes main unit.

Optional Accessories

8004411

Handpiece

8004431

Round Plate ø358004412

Common Cable

8004432

Round Plate ø60

8004413

Triax Cable

8004433

Round Plate ø808004419

Patient Button

8004414

Conductive Creme

8004415Creme Diffusor

8004416

Flat Plate ø358004417

Flat Plate ø60

8004418

Flat Plate ø808004421

Common Plate 150 × 200 mm

8004420

Common Plate 180 × 230 mm

8004422

Adhesive Plate 25pcs/box

8004423

Adhesive Electrode 120pcs/box

8004424

Bag

800441180044318004412800441680044138004419800441780044188004432

Radio Frequency Therapy

Specifications

80044338004414800442180044158004420800442380044228004424

39

Shortwave Diathermy

SW-1000

Pulsed & Continuous Shortwave

Shortwave therapy uses a frequency of 27.12 MHz. The electric-fields methods usedin shortwave therapy comprise Capacitive (Condenser-field) and Inductive (Coil-field)
methods. Shortwave in the form of electromagnetic radiation is absorbed by the body,
generating molecular rotational movement and displacement current within the body,in accordance with the Maxwell equation. The unique wave motion of electromagnetic
radiation propagates deep inside the body allowing tissue to be heated.
SW-1000 is pre-programmed with 10 therapy parameters for typical pathologies. You can
use both continuous and pulsed output modes in one unit. 1000 W in pulsed mode and
400 W in continuous mode can be performed as maximum output power.
Features
Continuous and pulsed modes selectable
10 effective clinical programs
16 memorable program numbers
3 types of pulsed output modes selectable

40

Full color large touch screen interface
High power output (1000 W in pulsed mode and
400 W in continuous mode)
Auto-tuning function installedPower supply

AC 100–240 V, 50/60 Hz

Power consumption

1 kVA

Display

Full-color, 7  LCD touch screen

Safety class according to IEC 60601-1

Class I, Type BF

Dimensions

470 (W) × 470 (D) × 940 (H) mm (main unit)

Weight38 kg (without electrodes)

Output mode

Continuous, Pulsed (3 in 3, 2 in 3 or 1 in 3)
1000 W (peak) in pulsed mode
400 W (mean) in continuous mode

Maximum output power
Frequency

27.12 MHz

Pulse frequency

5 Hz, 10 Hz, 20 Hz, 30 Hz, 50 Hz, 80 Hz, 100 Hz, 200 Hz, 400 Hz, 600 Hz, 800 Hz

Pulse width

20 μs, 40 μs, 65 μs, 100 μs, 200 μs, 400 μs

TimerMax. 30 min.

Shortwave DiathermySpecifications

Ordering Data
Standard Kit *Standard kit includes main unit.
8004809

100 mm Capacitive Electrodes and Cables, 2pcs/pack8004808

Output Tester

80048098004808

Screens41

Microwave Diathermy

PM-810+Microwave Therapy

Microwave diathermy is classified as high frequency therapy which utilizes electromagnetic
waves. As the energy from these waves is absorbed in high water content such as muscle
and will make them warm, microwave diathermy will be practical and suitable for
reduction of pain, relaxation of muscle and improvement of range of motion and treatment
for pathologies in the muscles.
The smaller dimensions of PM-810+ do not occupy much space of treatment room. You
can easily switch between Pulsed and Continuous modes by pressing button on theoperating panel.
Features
Effortless transition between pulsed andcontinuous mode
Lockable caster wheels help to safely install the unit
Space-saving design

42

Just right size of rectangular applicator covers both
small and medium treatment areas
High quality well-shielded HF cable provides
trouble-free operation

Power supply

AC 110, 220 or 230 V, 50/60 Hz
110 V: 850 VA
220 V: 760 VA
230 V: 770 VA

Power consumption
Safety class according to IEC 60601-1

Class I, Type B

Dimensions

330 (W) × 395 (D) × 790 (H) mm (main unit)

Weight28 kg (without applicator)

Output mode

Continuous, Pulsed
Continuous: 100 W
Pulsed: 100 W mean power, a fixed peak power of 1500 W

Maximum output power
Frequency

2450 MHz

Timer

Max. 30 min.

Applicator size

293 × 150 mm

Microwave DiathermySpecifications

Ordering Data
Standard Kit *Standard kit includes main unit.
012040

Rectangular Applicator with HF Cable, 290 mm in width

151196Tool for Tightening Arm Joint

180350Power Supply Cord (220−240 V, Type F) or

180349Power Supply Cord (110−120 V, Type A)

151196012040Treatment Images

43

Traction Therapy

TM-400Powered Traction Unit

Traction therapy is one of the easiest and most effective methods for spinal decompression.
Traction therapy influences the pressure between the vertebrae. Because of the pressure
decrease during traction, a mechanical effect on the nerves is generated and a biophysicaleffect on the circulation and oxygen supply to the local structures occurs. This can resultin immediate pain relief, but also in a structural improvement in the long run.
TM-400 is installed with most common effective 8 traction modes. Parameter of eachmode can be modified to suit your particular needs. Innovative automatic calibration
function for traction force is installed. It contributes to maintaining accurate and suitabletraction force.
Features8 traction modes
30 memorable program numbers
Automatic calibration for traction force
Large and high-visibility LCD screen
Selectable pulling speed
Exclusive traction table available (optional)
Various types of traction harnesses can be suppied

44

10 languages available (English, German, French,
Italian, Spanish, Portuguese, Danish, Dutch,
Swedish & Chinese)
Set and actual values of traction force are displayedon the same screen
Patient switch for safe treatment

Power supply

AC 100–240 V, 50/60 Hz

Power consumption

75 VA

Display116 (W) × 87 (H) mm

Safety class according to IEC 60601-1

Class I, Type B

Dimensions

260 (W) × 350 (D) × 295 (H) mm

Weight14 kg

Force range

1–90 kg (1–198 lbs)

Traction mode

Intermittent, Static, Progressive-intermittent, Progressive-static,
Progressive-regressive, Cyclic-intermittent, Cyclic-static, Cyclic

User-defined protocol memories

30

Timer

Max. 99 min.

Traction Therapy

Specifications

Ordering Data
Standard Kit *Standard kit includes main unit.

Traction Packages

010850Patient Switch

Fixed Traction Table Package, including TM-400 traction unit

180350Power Supply Cord (220−240 V, Type F) or

180290Power Supply Cord (110−120 V, Type A)

0108508005704

Fixed Height Traction Table including 2 Armpit Holders,
600 × 2500 mm

011977

Cervical Harness

011978Pelvic Harness

012003

Spreader Bar

Optional Accessories
011843

Traction Rope Kit

011975

Triangular Flexion Stool

011976

Pillow

Electric Traction Table Package, including TM-400 traction unit

011977

Cervical Harness

8005701

Electric Variable Height Traction Table, 640 × 2540 mm

011978Pelvic Harness

011977

Cervical Harness

011990Thoracic Harness

011978Pelvic Harness

012003

Spreader Bar

011990Thoracic Harness

8005760

Armpit Holder, 2pcs/set

012003

Spreader Bar

8005704

Fixed Height Traction Table including 2 Armpit Holders,
600 × 2500 mm

8005701

Electric Variable Height Traction Table, 640 × 2540 mm

011977

TM-400 + 8005704

012003

TM-400 + 8005701
011975
Cervical Traction

8005760

011990011978Lumbar Traction

45

Electroacupuncture

ES-1606-channel Programmable Electroacupuncture

Electroacupuncture is to insert needles and apply electrical stimulation at ver yprecise acupuncture points. Needling the acupuncture points stimulates the ner vous
system to release chemicals in the muscles, spinal cord and brain. These chemicals
either change the experience of pain, or they trigger the release of other chemicals
and hormones that influence the body’s own internal regulating system.
ES-160 has various safety functions for safe treatment. Semi-independent 6 electro
stimulation channels allow to treat various parts of body simultaneously. You can save
16 treatment parameters and quickly load & modify them as you like. The previous
treatment saving function is also installed.
Features
Semi-independent 6 electro stimulation channels
Various stimulation modes installed (Constant,
Burst, Surge, Fast+Slow, Sweep, Random-1,
Random-2, Randam-3)
Various safety functions for safe treatment

46

Previous treatment parameter saving function & 16
memorable program numbers
Acupuncture point search function with exclusiveprobePower supply

DC 6 V

Number of channels

6 semi-independent

Safety class according to IEC 60601-1

Internally powered equipment, Type BF

Dimensions

239 (W) × 174 (D) × 41 (H) mm

Weight

600 g (without batteries)

Output mode

Constant, Burst, Surge, Fast+Slow, Sweep, Random prog. 1, Random prog. 2, Random prog. 3

Maximum amplitude

8 mArms

Frequency

0.5 Hz, 0.7 Hz, 1–500 Hz

Pulse shape

Symmetric biphasic rectangular pulse

Phase duration

50–400 μs

TimerMax. 60 min.

Electroacupuncture

Specifications

Ordering Data
Standard Kit *Standard kit includes main unit.
011380

Search/Stimulation Probe

011324

Electrode Cable (Black)011325

Electrode Cable (Brown)

011326

Electrode Cable (Red)011327

Electrode Cable (Green)

011328

Electrode Cable (Blue)

011329

Electrode Cable (Gray)

B060999

Exclusive Clips (Black), 6×

B061000

Exclusive Clips (Red), 6×

NOTE: Batteries are not included.

Main unit

011380B061000

Electrode
Cable
B060999

47

Electroacupuncture

ES-1303-channel Palm-sized Electroacupuncture

Electroacupuncture is a method of encouraging the body to promote natural healing
and to improve functioning.
As the weight of ES-130 is less than 150 g (without batter y) and the unit size isver y compact, you can easily bring it anywhere you like. Semi-independent 3 electro
stimulation channels allow to treat for various parts of body simultaneously. “Semiindependent channels” means that you can apply different current intensity for eachchannel. Low/High intensity can be easily set by Selector Switch located at the backside of main unit.
Features
Semi-independent 3 electro stimulation channels
Easily portable
Dual intensity setting (Low and High)48

Frequency range selector switch
Frequency table & fine frequency adjustment dialfor particular settingPower supply

DC 9 V

Number of channels

3 semi-independent

Safety class according to IEC 60601-1

Internally powered equipment, Type BF

Dimensions

63 (W) × 27 (D) × 96 (H) mm

Weight

120 g (without battery)

Maximum amplitude

14 mArms

Frequency

1–500 Hz

Pulse shape

Biphasic rectangular waveform

Phase duration

100 μsOrdering Data

Electroacupuncture

Specifications

Standard Kit *Standard kit includes main unit.
260182Carrying Bag

B180432

Electrode Cable with Clips (White)

B180433

Electrode Cable with Clips (Yellow)

B180434

Electrode Cable with Clips (Green)

NOTE: A battery is not included.

B180433

260182

B180432

B180434

49

Digital Goniometer

EasyAngle

Accurate and Single Hand Operation ROM Measuring Device

Measuring ROM is one of the most common measurements in physical therapy. It needs tobe done on daily and treatment basis as the measuring progress is an important motivator
for the patients. EasyAngle is a very innovative device which can help physicians fortheir daily work. The device is not just a device to measure ROM, it is a device that helps
professionals become more professional.
Features
Measurement can be quickly done with one hand,making it possible to support the patient with theother hand
Can be used for measurement of all joints, replacing
other devices such as the goniometer, inclinometerand CROM device

50

High precision sensor and many studies show highreliability and validity
Measurement is done with three quick clicks andonly one alignment needs to be done at a time
Ergonomic design and a clear display where thelast five measurements are saved

Power supply

DC 3.7 V (battery) / DC 5 V (USB cable)

Safety class according to IEC 60601-1

Internally powered equipment, Type B

Dimensions

86 (W) × 38 (D) × 37 (H) mm (main unit only)

Weight69 g (including alignment guide)

Degree of protection against
harmful ingress of particulate matter

IP4X

Operative duration time

2 weeks (12 min. use / day)Standard Kit *Standard kit includes main unit.
——

Alignment Guide——

USB cable

Alignment Guide

Digital Goniometer

Specifications

USB cable

Measurement Images

51

Others

Best View Stand
Triangular Tilt Stand

Best View Stand dramatically improves visibility and operability for therapists.

without standwith standSpecifications
Available with52

EU-941, EU-921, ES-5400, ES-5200

3-1-8 Sakae-cho, Kawaguchi-shi, Saitama 332-0017, Japan
TEL: 81-48-254-1031 FAX: 81-48-254-1033
URL: https://www.itocoltd.com/ E-Mail: itocoltd@itolator.co.jp＊Designs and specifications are subject to change without notice.
No.217 C0c000107-2306 T2307DATA