MET ONE 3400+
Automate environmental monitoring for GMP
cleanroom compliance with ONE easy decision

Our new MET ONE 3400+ Series Portable
Airborne Particle Counter features automatedintegrated electronic SOPs and sampling
maps that reduce training requirements—andsampling errors.

MET ONE 3400+

It’s all in the box

PORTABLE AIRBORNE
PARTICLE COUNTER

Customized Electronic SOP Maps

Secure Electronic Records

Using a web-browser interface, load your SOP
map into the counter, mark sample locations, anddefine SOP requirements for each.

An electronic signature is attached to all approved
final reports, which can then be exported insecure electronic format.

Interactive Tracking
Onscreen instructions help operators navigate
your map and perform sampling according to theSOP. Location names turn green after sampling,making it easy to track progress at a glance.
Managers can review daily monitoring progressincluding all sample results and alarms, via a webbrowser.

Barcode Function (optional)
Connect a barcode reader to automatically
capture sample location, production batch IDand more.

Automated SOP Version Control
To simplify SOP changes/updates, administrators
use a web-based tool to approve each updated
SOP via electronic signature. Changes areautomatically replicated across all instruments.

Plug-n-Play Networking

Review and Approve

Sampling Options

All instruments automatically create a network,
which minimizes involvement of your IT
department.

Supervisors can log-on to the instrument via a
web-browser to review sampling results, alarms
and comments and approve the final electronic
record, attaching their electronic signature.

1. Configure network
2 |Available in 1 ft3/min, 50 L/min and 100 L/minversions. The MET ONE 3400+ can sample 1m3 inas little as 10 minutes

2. Choose LDAP settings including Active Directory

For Research Use Only. Not for use in diagnostic procedures

3. SOP version control inside the counter4. Confirm all changes using built-in audit trail

5. Review and approve monitoring results via web-browser

Characterized by Ingenuity | 3

Simplified, secure data management eases audits

Designed for life in your cleanroom

The MET ONE 3400+ helps you manage accurate cleanroom monitoring data via safe and secure internal
software and web access—there’s no external software needed. All data is encrypted, and users are never permitted
to delete records.
And because it uses Microsoft Active Directory for Username and Password Control for log-on and electronic
signatures, it supports 21 CFR Part 11 compliance and ALCOA guidance.

Lighter to carry
At 12.4 lbs (5.6 kg)*, it’s up to 26% lighter than earlier models.

Attributable

Legible records

Contemporaneously
createdMulti-level, Microsoft Active Directory Username and Password control creates unique electronic
signatures for users

Legible, secure PDF electronic record export for measurement results

Secure PDF electronic records created in the counter - no manual data transcription or scanning of paper
reportOriginal

Original electronic record created directly inside the counter

Accurate

Your SOPs and sampling maps uploaded into the counter with version control. Completed records
reviewed and approved for accuracy in the counter remotely via web-browser. No manual counter
configuration or sample location entry

Easier to use
Sensitive, 10-inch touchscreen enables reliable data entry even whenusers are double-gloved.
Optional barcode reader expedites data entry and minimizes potential
for operator errors.

Quieter
New blower technology helps make the MET ONE 3400+whisper quiet.
To help you quickly locate data you need during
an audit, you can use filters to review multiple
parameters, including:• DateSecure electronic records (Excel or PDF) can beexported directly to your network or to a USB
device.
Each record contains:• User• SOP version number• Location• User name• Failed logins• Location name• Action type• Time• Alert level exceeded• DateEasier to clean
Because it has a sealed touchscreen and stainless steel case,
you can spray cleaners (VHP, bleach, alcohol) directly onto
the instrument.

• Alarms• Counter configuration• Sampling results• Unlimited user defined fields, e.g. production
batch ID, or room occupancy state, etc.
*Weight is without batteries
4 |For Research Use Only. Not for use in diagnostic procedures

Characterized by Ingenuity | 5

Networks so well so you don’t have to

Product specifications

We live in an interconnected world, but for many GMP regulated devices to share data requires huge software
integrations and validation. The MET ONE 3400+ has built-in, secure networking capabilities that are validated out ofthe box using our IQ / OQ protocols. A lightweight networking protocol securely protected via transport layer security(TLS) in the MET ONE 3400+ interconnects multiple devices and shares configuration and sample data in a secure andcompliant manner. Ease the burden on your IT department with the new MET ONE 3400+ air particle counter.

MET ONE 3400+ Specifications

Identical web browser
experience to instrumentfor SOP configuration, user
management, or training
through a projector.
An instrument can beinstalled into a remotelocation and fully controlled
for sampling via the remoteweb interface, allowing
for central control orremote operation in areas
where technicians are notpermitted.

Particle Counting Performance

MET ONE 3413+

MET ONE 3415+

MET ONE 3423+

MET ONE 3425+

MET ONE 3445+

Standard Sizes

0.3, 0.5, 1.0, 3.0, 5.0,
10.0 µm

0.5, 1.0, 2.0, 3.0, 5.0,
10.0 µm

0.3, 0.5, 1.0, 3.0, 5.0,10.0 µm

0.5, 1.0, 2.0, 3.0, 5.0,
10.0 µm

0.5, 1.0, 2.0, 3.0, 5.0,10.0 µm

Flow Rate

28.3 LPM (1.0 CFM)

28.3 LPM (1.0 CFM)

50 LPM (1.77 CFM)

50 LPM (1.77 CFM)

100 LPM (3.53 CFM)

20 minutes

10 minutes

Flow Rate Tolerance
Time to sample 1 m³

±5%
35.3 minutes

Machine Groups and Data
Replication

Each MET ONE 3400+ assigned to a machine group will share user configuration, SOPconfiguration, and sample data to enable any machine in the group to have a complete view of theenvironmental monitoring status. A light weight protocol connects the instruments to exchange
true uniquely identified copies of data and to insure all instruments are in sync.

Coincidence Loss

10% at 36,000,000particles per m³

10% at 36,000,000
particles per m³

10% at 20,000,000
particles per m³

10% at 20,000,000
particles per m³

10% at 10,000,000
particles per m³

Counting Efficiency - 50%

50% ±20% for 0.3 µm

50% ±20% for 0.5 µm

50% ±20% for 0.3 µm

50% ±20% for 0.5 µm

50% ±20% for 0.5µm

Counting Efficiency - 100%

100% ±10% for0.45 µm

100% ±10% for
0.75 µm

100% ±10% for0.45 µm

100% ±10% for
0.75 µm

100% ±10% for0.75 µm

Light SourceLong Life Laser™ diode with 10-year Mean Time To FailurePump Type

Filtered to ISO 3 specifications, 3/8” NPT pipe fitting

Exhaust3/8" NPT pipe fitting available

Physical and Electrical Specifications

Wireless Connectivity

Network Shares

Audit Trail

Within the MET ONE 3400+, Windows Network Shares can be configured to allow a technician
to generate a report and upload it to that directory. Windows authentication controls rights andaccess to the share using tools already commonly deployed in most enterprises.

A GMP grade audit trail is included in the instrument, including information on logins, logouts,power events, sampling, electronic signatures, SOP updates, and other critical actions taken on theinstrument. Built in filtering and reporting simplifies review and approval of audit events.

3 hours

Battery Life (Continuous Sampling) w/ 2 Battery

6.25 hours

6.25 hours

6.5 hours

6.5 hours

4 hours

Battery Life (Continuous Sampling) w/ 3 Battery

9.25 hours

9.25 hours

9.75 hours

9.75 hours

5.5 hours

Maximum Battery Recharge Time

Weight (without battery)
Weight (with included batteries)
Enclosure Material

3 hours

3 hours

Pump will notoperate

3 hours

6.75 hours(with 3 batteries)

3.5 hours (with 2 batteries)
26.7 cm W x 26 cm H x 21 cm D (10.5" W x 10.25" H x 8.25" D)
5.6 kg (12.4 lbs)
6.5 kg (14.3 lbs) with 2 batteries

6.9 kg (15.3 lbs)

316 Stainless SteelOperating Temperature and Humidity

0 to 40°C (32 to 104°F), 10 to 90% relative humidity, non-condensing

Storage Temperature and Humidity-20 to 50 degree C (-4 to 122 F), 0 to 98% relative humidity, non-condensing

User Interface
Screen10" High Resolution LCDInterface

High sensitivity capacitive touch screen - works with dual gloves

AlarmsUnlimited user defined alarms per channel or external environmental sensor

Sampling Configuration Options

Sample by time or volume, up to 100 cycles, with user defined time intervals

Electrical Interfaces
USB HostEthernet
WiFi
Environmental Probe

For Research Use Only. Not for use in diagnostic procedures

3 Li Ion Rechargeable Smart Battery
Packs

1, 2 or 3 Li Ion Rechargeable Smart Battery Packs

Battery Life (Continuous Sampling) w/ 1 Battery

USB Device

6 |Output: 24 VDC, 5.0 A Maximum; Input: 80-264 VAC, 47-63 Hz, 1.2 A Maximum

Battery Type and Quantity

Dimensions

The MET ONE 3400+ includes an 802.11b/g wireless interface with full security capability, including
enterprise grade 802.1x certificate-based authentication using WPA EAP (PEAP, TLS, and TTLSvariants) and WPA2/PSK.

20 minutes

1 count or less in 5 minutes, 95% confidence level

AC Power Supply

Better networking features

35.3 minutes

Zero Count3 Type A USB 2.0 Host Ports
1 Type B USB Device Port
1 802.3 Compliant 10/100 BASE-TX Auto negotiating with Auto MDI-X
802.11 b/g 2.4 GHz WiFi Radio
Proprietary DIN port for Relative Humidity and Temperature sensor probe PN 2088373-01

Characterized by Ingenuity | 7Ordering information
Sample FlowRateMinimum
Particle Size

APC, Port, MET ONE 3413+

28.3 LPM

0.3 μm

C38678

APC, Port, MET ONE 3415+

28.3 LPM

0.5 μm

C38681

APC, Port, MET ONE 3423+

50 LPM

0.3 μm

C38675

C38684APC, Port, MET ONE 3425+

50 LPM

0.5 μm

C38687

APC, Port, MET ONE 3445+

100 LPM0.5 μmEach part number includes the instrument, 2 batteries, power supply, ISO probe and stand, 10 feet of tubing, anda cleaning brush. C38687 includes 3 batteries.
Part Number

Description

C64707

Probe, Isokinetic, SS, for 100 LPM

C64708

Probe, Isokinetic, SS, for 28.3 LPM

C64709Probe, Isokinetic, SS, for 50 LPM

C64713Tubing, Hytrel®, 3/8”ID, 1/2”OD (10’ Long)

2088000-02

Filter Scan Probe, No Electronics. 28.3 LPM*

C64705Paper, Thermal 58mm Wide Roll (Pack of 10)

2088373-01

Probe, Relative Humidity & Temperature

C64706

Battery. Power Supply, External 24VDC 3400+

C50400Battery, 3400+

C64693External Dual Battery Charger

C64408

Cordset, IEC Type I, Australia

C64410Cordset, IEC Type N, Brazil

C64412

Cordset, IEC Type E, Europe

C64413

Cordset, IEC Type D, India

C64414

Cordset, IEC Type F, Korea

C64415

Cordset, IEC Type G, UK

2080732-13

High Pressure Diffuser, 28.3 LPM

2080732-12

High Pressure Diffuser, 50 or 100 LPM

C50352

Cap, I/O Connectors 3400+

C56591

Wheeled, Hard Sided Transit Case 3400+*not available in EU

Interested in talking to someone? Visit www.beckman.com/contact-us
Learn more about MET ONE solutions at beckman.com/metone© 2021 Beckman Coulter, Inc. All rights reserved. Beckman Coulter, the stylized logo, and the Beckman Coulter product and service
marks mentioned herein are trademarks or registered trademarks of Beckman Coulter, Inc. in the United States and other countries.
For Beckman Coulter’s worldwide office locations and phone numbers, please visit Contact Us at beckman.com
PART-8058SB10.20