eQuinoxTM Model 4000 Oxygen Concentrator
eQuinoxTM Modèle 4000 Concentrateur d’oxygène
eQuinoxTM Modell 4000 Sauerstoffkonzentrator
eQuinoxTM Modelo 4000 Concentrador de oxígeno
eQuinoxTM Modello 4000 Concentratore di ossigeno
eQuinoxTM Modelo 4000 Concentrador de oxigénio
eQuinoxTM Model 4000 Zuurstofconcentrator

User Manual
Manuel d’utilisation
Bedienungsanleitung
Manual del usuario
Manuale dell’utente
Manual do Utilizador
Gebruikershandleiding

User Controls & System Status Indicators
SymbolA

02

Definition

Definition

Read user manual before operation. See user manual
for instructions.

FAA Approved Symbol: The U.S. Federal Aviation
Administration (FAA) has approved this device foruse on-board commercial aircraft.

No Smoking Icon: Do not smoke near unit.

Flow Setting Indicator

Warnings / ALERT (Yellow) Indicator

Pulse Mode Operation

ALARM (Red) Indicator­

Device operating normally; power button

External Power Indicator

Increase Flow Setting

Battery Pack (battery) Status Gauge

Decrease Flow Setting

Battery Pack Symbol

Use no oil or grease.

Amperes

No open or naked flames.

Alternating Current

No serviceable parts inside. Do not open cover.

IEC Icon for DC Power (Input)

Warm Surface.

Oxygen Output

Do not get wet.

Type BF Applied Part (degree of protection againstelectric shock)

Drip Proof Equipment-IPX1: The eQuinox Battery
Pack (Battery) and AC Power Supply provide
protection against the harmful effects of the ingressof liquids. (IPX1, per IEC 60529)

Recycle Symbol

IP22

SymbolThis symbol is to remind the equipment owners
to return it to a recycling facility at the end of its
life, per Waste Electrical and Electronic Equipment(WEEE) Directive.
Drip and solid ingress proof: The eQuinox provides
protection against the harmful effects of the ingressof liquids and solid foreign objects in accordancewith IEC 60529 at the rating of IP22.
Certified for both the U.S. and Canadian markets, tothe applicable U.S. and Canadian standards.

2 — 4805-C1 C (English)

IPX1

Name and address of manufacturer
Authorized representative in the European
Community

Class II equipment
This device complies with the requirements of
Directive 93/42/EEC concerning medical devices. It
therefore bears the CE marking as shown.

Contents
User Controls & System Status Indicators............................................................................................................................... 2
Quick Start Guide............................................................................................................................................................................ 4
Indications for Use.......................................................................................................................................................................... 7
Contraindications for Use............................................................................................................................................................. 7
Safety Guidelines............................................................................................................................................................................. 7
Recommended Operating Environments.............................................................................................................................. 9Operating Instructions................................................................................................................................................................10
Voice Module Operation.............................................................................................................................................................13First Time eQuinox Operation...................................................................................................................................................14
User Troubleshooting Table ......................................................................................................................................................16
Cleaning, Care & Routine Maintenance.................................................................................................................................19
EMC Tables.......................................................................................................................................................................................21
General Airline Travel Information..........................................................................................................................................25
Battery Pack & Airline Travel......................................................................................................................................................25

eQuinox (Model 4000) User Manual — 3

Quick Start Guide

1

2

Unpack Your eQuinox

Getting to Know Your eQuinox

Review all Warnings, Cautions and additional device
information in the rest of this manual.
Become familiar with the key features of the eQuinox and theUser Control Panel.
OxygenOutlet Port

FRONT
HandleControl
Panel
External Power
Receptacle

AC Power Supply

FAA
Approval
Icon

Battery
Pack Latch

Rating Label &Serial Number
LocationeQuinox Cart

BatteryPack

BACK

12VDC Cable

ServiceCabinetExhaust
Vent

Battery Pack(battery)4 — 4805-C1 C (English)

Air inlet
Filter

The eQuinox User Control Paneldisplays important operating
information.

ON/OFF Button (Green) Indicator:This button powers
the device ON or OFF.The Green Indicator is lit up when
the device is ON and functioning properly.
Increase or Decrease Flow Setting Buttons: Use thesebuttons to set the flow to your prescribed setting.

3

Powering on the eQuinox

To connect to AC power source: Insert the AC Power Supply
cord securely into the External Power Receptacle on the rightside of the device. Plug the AC power cord into the power
supply and into an AC outlet.
Step 1		

Step 2

Step 3

DeliveryModeButtonandIndicator:ThebuttonswitchesbetweenContinuousFlowandPulseDoseMode.The
Pulse Dose Mode activates autoSAT Technology. This
allowsasignificantincreaseintheoperatingtimewhile
powered by the battery pack (battery).When the Pulse
Dose Mode is activated, the green Pulse Dose Mode
Indicator lights up. A pulse of oxygen is delivered andthe green indicator blinks with each inspiratory effort.
ALERT (Yellow) Indicator—When lit up, this indicates alow priority awareness condition or Caution. Continue
to use your system. Refer to theTroubleshootingTable
for the correct action, or contact your Home Care
Provider. A flashing yellow indicates a medium priority
alert. A prompt response is necessary. (Typically low
oxygen or low battery.)
ALARM(Red)Indicator­—Indicatesahighpriorityalarm
condition. An immediate response is necessary. Referto the Troubleshooting Table and contact your homecare provider. (Typically loss of power, low oxygen, or
malfunction.)
Information Display.
• C or P – “C” indicates Continuous Flow mode and“P” indicates Pulse Dose mode.
• Time on battery – The display will cycle every 20
secondswhenrunningonbatterypowertoindicatethe
estimated time left on the battery• Setting (continuous flow or pulse dose) – each time
you power the device ON, the last mode and/or setting
has been saved and will be used at start-up.

When the eQuinox is plugged in properly, a green indicatoron the AC Power Supply will light up. The External Power
Indicator on the user control panel will also light up.
Install the Battery: Slide the battery into the empty compartment
until it locks into position. It will be flush with the front of thedevice.

When the battery is properly installed and the eQuinox is either
ON or plugged into AC power, the Battery Pack Status Gauge
will appear.

Battery Pack (battery) Status Gauge: This indicator
displays the charge remaining in the battery. Each of
the five horizontal gray bars represents approximately
20% of the total battery charge. When the battery is
charging, the charge indicator bars will blink in a waterfall-type fashion. If the battery is not installed, or if it
isincorrectlyinstalled,theBatteryPack(battery)Status
Gauge will not light up or will flash on and off.
ExternalPowerIndicator:WhentheeQuinoxisproperly
plugged in and is using the AC or 12VDC Cable, thisindicator will light up on the User Control Panel.
Buzzer: An audible alarm alerts you to the operating condition of thedevice, either a warning or failure, and confirms a valid key press.
Voice Module: If activated, a voice indicator will alert you of the operating condition of the device, changes in power source, and warning
conditions or failures.

ON/OFF Button (Green) Indicator
Press and hold the “ON/OFF” button for two (2) seconds topower ON your eQuinox. A brief audible and visual power on
self-test will occur. Check that all the indicators are lit up andthe buzzer sounds for three (3) seconds. Press and hold the “ON/OFF” button for two (2) seconds to power OFF your eQuinox.
eQuinox (Model 4000) User Manual — 5

4

Select Flow Delivery Mode

Flow Mode Button and Indicator

Press the Flow Mode button to select your desired flow mode.
Pressing this button repeatedly will switch you back and forth
between continuous flow and pulse dose modes. In Continuous
Flow Mode, the oxygen is provided at a constant flow rate
between 0.5 and 3.0 LPM, indicated by the letter C. In Pulse
Dose Mode, oxygen is supplied in a bolus at the beginning ofeach inhalation. Pulse mode settings 1–9 provide bolus sizes
ranging from 16mL to 192mL and are designated by the letter
P on the display. Pulse dose can be adjusted for sensitivity andbreath bolus delivery. Flow settings in both modes should beprescribed for you by your physician.

5

Adjust Flow Setting to Prescribed Level

6

Begin Using the eQuinox

Air Inlet Filter

Air inlet
FilterYour unit is equipped with an Air Inlet Filter (extra filter
provided). Check daily and clean once a week. Instructions forcleaning are on page 19.
Connect the oxygen supply tubing to the oxygen outlet andconnect the oxygen cannula per the instructions provided withthe cannula, or according to the instructions from your home careprovider.

Breathe normally through the nasal cannula.

Increase or Decrease Flow Setting Button

When operating on battery power, using the Pulse Dose Mode
will result in longer duration-of-use.

Using the + or - buttons, adjust the setting to the prescription
supplied by your physician.

Continuous Flow Mode Display

Pulse Dose Mode Display

WARNING: IT IS VERY IMPORTANT TO SELECT ONLY THE
PRESCRIBED LEVEL OF OXYGEN. DO NOT CHANGE THE
FLOW SELECTION UNLESS YOU HAVE BEEN DIRECTED TO
DO SO BY A LICENSED CLINICIAN. THE PORTABLE OXYGEN CONCENTRATOR MAY BE USED DURING SLEEP UNDER THE RECOMMENDATION OF A QUALIFIED CLINICIAN.
6 — 4805-C1 C (English)

7

eQuinox Cart Operation

Line up the eQuinox with the shaped floor of the cart so that itsits flat and upright. Line up the large threaded screw to the insert
at the back of the eQuinox. Then hand-tighten the knob on the
cart until the cart is tightly secured to the device. Depress thepush-button on the telescoping cart handle and pull upward toadjust the height of the handle.

Important!
Safety Instructions are defined as follows:
WARNING: IMPORTANT SAFETY INFORMATION FOR HAZARDS THAT MIGHT CAUSE SERIOUS INJURY.
CAUTION: Important information for preventing damage
to the eQuinox.
Note: Information needing special attention.

Indications for Use
The eQuinox is intended for the administration of supplemental
oxygen. The device is not intended for life support, nor does it
provide any patient monitoring capabilities.
WARNING: IT IS VERY IMPORTANT TO SELECT ONLY THE
PRESCRIBED LEVEL OF OXYGEN. DO NOT CHANGE THE
FLOW SELECTION UNLESS YOU HAVE BEEN DIRECTED TO
DO SO BY A LICENSED CLINICIAN. THE PORTABLE OXYGEN CONCENTRATOR MAY BE USED DURING SLEEP UNDER THE RECOMMENDATION OF A QUALIFIED CLINICIAN.
WARNING: FEDERAL (USA) LAW RESTRICTS THIS DEVICE TO SALE OR
RENTAL BY ORDER OF A PHYSICIAN OR OTHER LICENSED HEALTH
CARE PROVIDER.
WARNING: THIS UNIT IS NOT TO BE USED FOR LIFE SUPPORT.
GERIATRIC, PEDIATRIC, OR ANY OTHER PATIENT UNABLE TO COMMUNICATE DISCOMFORT WHILE USING THIS DEVICE MAY REQUIREADDITIONAL MONITORING. PATIENTS WITH HEARING AND/OR
SIGHT IMPAIRMENT(S) MAY NEED ASSISTANCE WITH MONITORING
ALARMS.

Contraindications for Use
WARNING: IN CERTAIN CIRCUMSTANCES, THE USE OF
NON-PRESCRIBED OXYGEN CAN BE HAZARDOUS. THIS
DEVICE SHOULD ONLY BE USED WHEN PRESCRIBED BY A
PHYSICIAN.
WARNING: NOT FOR USE IN THE PRESENCE OF FLAMMABLE ANESTHETICS.
WARNING: AS WITH ANY ELECTRICALLY POWERED DEVICE, THE
USER MAY EXPERIENCE PERIODS OF NON-OPERATION AS A RESULTOF ELECTRICAL POWER INTERRUPTION, OR THE NEED TO HAVE THE
EQUINOX SERVICED BY A QUALIFIED TECHNICIAN. THE EQUINOXIS NOT APPROPRIATE FOR ANY PATIENT WHO WOULD EXPERIENCEADVERSE HEALTH CONSEQUENCES AS THE RESULT OF SUCH TEMPORARY INTERRUPTION.

Safety Guidelines
WARNING: NO MODIFICATION OF THIS EQUIPMENT ISPERMITTEDWARNING:THE MANUFACTURER RECOMMENDS AN ALTERNATE SOURCE OF SUPPLEMENTAL OXYGEN IN THE EVENTOF A POWER OUTAGE, ALARM CONDITION, OR MECHANICAL FAILURE. CONSULT YOUR PHYSICIAN OR EQUIPMENTPROVIDER FOR THE TYPE OF RESERVE SYSTEM REQUIRED.
WARNING: THIS DEVICE SUPPLIES HIGH-CONCENTRATION OXYGEN
THAT PROMOTES RAPID BURNING. DO NOT ALLOW SMOKING OR
OPEN FLAMES WITHIN THE SAME ROOM OF (1) THIS DEVICE, OR (2)ANY OXYGEN-CARRYING ACCESSORY. FAILURE TO OBSERVE THISWARNING CAN RESULT IN SEVERE FIRE, PROPERTY DAMAGE, AND/OR CAUSE PHYSICAL INJURY OR DEATH.
WARNING: DO NOT OPERATE UNIT IN A RESTRICTED OR CONFINEDSPACE (I.E., A SMALL CASE OR HANDBAG) WHERE VENTILATIONCAN BE LIMITED. THIS CAN CAUSE THE OXYGEN CONCENTRATOR TO
OVERHEAT AND IMPAIR PERFORMANCE.
WARNING: THE CONCENTRATOR SHOULD BE LOCATED AS TO AVOIDSMOKE, POLLUTANTS OR FUMES.
WARNING: THE USE OF SOME OXYGEN ADMINISTRATING ACCESSORIES NOT SPECIFIED FOR THIS OXYGEN CONCENTRATOR MAY
IMPAIR ITS PERFORMANCE. RECOMMENDED ACCESSORIES ARE
REFERENCED WITHIN THIS MANUAL.
WARNING: IF THE OXYGEN CONCENTRATOR HAS BEEN DROPPED,DAMAGED OR EXPOSED TO WATER, PLEASE CONTACT YOUR HOME
CARE PROVIDER FOR INSPECTION OR POSSIBLE REPAIR OF THE
DEVICE. DO NOT USE THE OXYGEN CONCENTRATOR IF IT HAS ADAMAGED POWER CORD OR PLUG.
WARNING: PROPERLY SECURE, BELT OR OTHERWISE RESTRAIN THE
OXYGEN CONCENTRATOR WHEN IN A VEHICLE DURING TRANSPORTTO PREVENT DAMAGE OR INJURY.
WARNING: DO NOT ALLOW EITHER THE AIR INTAKE OR THE AIR
OUTLET VENTS TO BECOME BLOCKED. DO NOT DROP OR INSERT ANYOBJECTS INTO ANY OPENINGS ON THE DEVICE. THIS CAN CAUSETHE OXYGEN CONCENTRATOR TO OVERHEAT AND IMPAIR PERFORMANCE.
WARNING: DO NOT OVERFILL THE OPTIONAL HUMIDIFIER. FILL THE
OPTIONAL HUMIDIFIER WITH WATER ONLY TO THE LEVEL SHOWN BY
THE MANUFACTURER OF THE HUMIDIFIER.
WARNING: DO NOT OPERATE THE EQUINOX AND AMBULATE WHILEA HUMIDIFIER IS ATTACHED. REMOVE THE HUMIDIFIER BOTTLE BEFORE WALKING. DO NOT LAY THE EQUINOX DOWN WHILE ATTACHEDTO A HUMIDIFIER BOTTLE.
WARNING: THE US DEPARTMENT OF TRANSPORTATION (DOT) AND
UNITED NATIONS (UN) REGULATIONS REQUIRE THE REMOVAL OF
THE BATTERY FROM THE DEVICE FOR ALL INTERNATIONAL AIRLINETRAVEL WHEN THE OXYGEN CONCENTRATOR IS CHECKED AS LUGGAGE. WHEN SHIPPING THE OXYGEN CONCENTRATOR, THE BATTERYMUST ALSO BE REMOVED FROM THE DEVICE AND PACKAGED
PROPERLY.
WARNING: DO NOT OPERATE THIS EQUIPMENT WITHOUT FIRSTREADING AND UNDERSTANDING THIS MANUAL. IF YOU ARE UNABLE
TO UNDERSTAND THE WARNINGS AND INSTRUCTIONS, CONTACTYOUR EQUIPMENT PROVIDER BEFORE ATTEMPTING TO USE THISEQUIPMENT; OTHERWISE, INJURY OR DAMAGE MAY RESULT.
eQuinox (Model 4000) User Manual — 7

WARNING: IF YOU FEEL DISCOMFORT OR ARE EXPERIENCING A MEDICAL EMERGENCY, SEEK MEDICAL ASSISTANCEIMMEDIATELY.

WARNING: SMOKING WHILE USING OXYGEN IS THE
NUMBER ONE CAUSE OF FIRE INJURIES AND RELATED
DEATHS. YOU MUST FOLLOW THESE SAFETY WARNINGS:WARNING: OPERATING THE OXYGEN CONCENTRATOR OUTSIDE OF
THE OPERATIONAL TEMPERATURE SPECIFICATIONS CAN LIMIT THE
CONCENTRATOR’S ABILITY TO MEET OXYGEN CONCENTRATIONSPECIFICATION. REFER TO THE SPECIFICATION SECTION OF THISMANUAL FOR TEMPERATURE LIMITS.

WARNING: DO NOT ALLOW SMOKING, CANDLES, OR OPEN FLAMESIN THE SAME ROOM WITH THE DEVICE OR THE OXYGEN-CARRYINGACCESSORIES.

WARNING: USE NO OIL, GREASE, OR PETROLEUM-BASED OR OTHER
FLAMMABLE PRODUCTS WITH THE OXYGEN-CARRYING ACCESSORIES OR THE OXYGEN CONCENTRATOR. ONLY WATER BASED, OXYGEN COMPATIBLE LOTIONS OR SALVES SHOULD BE USED. OXYGENACCELERATES THE COMBUSTION OF FLAMMABLE SUBSTANCES.

WARNING: IF YOU SMOKE, THESE 3 STEPS MAY SAVE YOUR LIFE:
TURN OFF THE OXYGEN CONCENTRATOR, TAKE OFF THE CANNULA,
AND LEAVE THE ROOM WHERE THIS DEVICE IS LOCATED.

WARNING: THE INCORRECT USE OF THE BATTERY CAN CAUSE THE
BATTERY TO GET HOT, IGNITE, AND MAY CAUSE SERIOUS INJURY. BE
SURE NOT TO PIERCE, STRIKE, STEP ON, DROP THE BATTERY, OR OTHERWISE SUBJECT THE BATTERY TO STRONG IMPACTS OR SHOCKS.
THE USE OF A DAMAGED BATTERY MAY CAUSE PERSONAL INJURY.
WARNING: THE OXYGEN CONCENTRATOR SHOULD NOT BE USEDADJACENT TO OR STACKED WITH OTHER EQUIPMENT. IF ADJACENTOR STACKED USE IS UNAVOIDABLE, THE DEVICE SHOULD BE OBSERVED TO VERIFY NORMAL OPERATION.
CAUTION: In the event of an alarm or you observe the
Oxygen Concentrator is not working properly; consult
the Troubleshooting section in this manual. If you cannot
resolve the problem, consult your Equipment Provider.
CAUTION: To prevent a void warranty, follow all manufacturers’
instructions.
WARNING: DO NOT ATTEMPT ANY MAINTENANCE OTHER
THAN THE POSSIBLE SOLUTIONS LISTED WITHIN THISMANUAL. DO NOT REMOVE COVERS, ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIED SERVICE TECHNICIAN SHOULDREMOVE THE COVERS OR SERVICE THIS DEVICE.
WARNING: ONLY ACCESSORIES RECOMMENDED BY THE MANUFACTURER. USE OF ANY OTHER MAY BE HAZARDOUS, CAUSE SERIOUS
DAMAGE TO YOUR OXYGEN CONCENTRATOR AND WILL VOID THE
WARRANTY .
WARNING:THE MANUFACTURER RECOMMENDS AN ALTERNATE SOURCE OF SUPPLEMENTAL OXYGEN IN THE EVENTOF A POWER OUTAGE, ALARM CONDITION, OR MECHANICAL FAILURE. CONSULT YOUR PHYSICIAN OR EQUIPMENTPROVIDER FOR THE TYPE OF RESERVE SYSTEM REQUIRED.
WARNING: DO NOT ALLOW EITHER THE AIR INTAKE OR THE AIR
OUTLET VENTS TO BECOME BLOCKED. DO NOT DROP OR INSERTANY OBJECTS INTO ANY OPENINGS ON THE DEVICE. THIS CANCAUSE THE OXYGEN CONCENTRATOR TO OVERHEAT AND IMPAIRPERFORMANCE.

8 — 4805-C1 C (English)

WARNING: SMOKING WHILE WEARING AN OXYGEN CANNULA MAY
CAUSE FACIAL BURNS AND POSSIBLY DEATH.

WARNING: “NO SMOKING – OXYGEN IN USE” SIGNS MUST BE
PROMINENTLY DISPLAYED IN THE HOME, OR WHERE OXYGEN IS IN
USE. PATIENTS AND THEIR CAREGIVERS MUST BE INFORMED ABOUT
THE DANGERS OF SMOKING IN THE PRESENCE OF, OR WHILE USING,
MEDICAL OXYGEN.
WARNING: DO NOT USE YOUR OXYGEN CONCENTRATOR IN THE
PRESENCE OF FLAMMABLE GASES. THIS CAN RESULT IN RAPID
BURNING CAUSING PROPERTY DAMAGE, BODILY INJURIES OR
DEATH.
WARNING: REMOVING THE CANNULA AND PUTTING IT ON CLOTHING, BEDDING, SOFAS, OR OTHER CUSHION MATERIAL WILL CAUSEA FLASH FIRE WHEN EXPOSED TO A CIGARETTE, HEAT SOURCE, OR
FLAME.
WARNING: DO NOT LEAVE A NASAL CANNULA ON OR UNDER
CLOTHING, BED COVERINGS OR CHAIR CUSHIONS. IF THE UNIT IS
TURNED ON BUT NOT IN USE, THE OXYGEN WILL MAKE THE MATERIAL FLAMMABLE. SET THE OXYGEN CONCENTRATOR TO THE 0(OFF) POSITION WHEN THE OXYGEN CONCENTRATOR IS NOT IN USE.
CAUTION: Always disconnect AC power supply from thewall before disconnection the AC power supply from theoxygen concentrator.
CAUTION: Always place oxygen supply tubing and power cords in amanner that prevents a trip hazard.
CAUTION: When using the Oxygen Concentrator in an automobile,boat, or on other DC sources with the 12VDC Cable, make sure thatthe vehicle is started and running before connecting the Oxygen
Concentrator. If the external power light on the eQuinox does notilluminate and requires resetting, disconnect the 12VDC Cable fromthe DC outlet, restart your vehicle, and then reconnect your 12VDC
Cable into the DC outlet. Failure to follow these instructions canresult in not supplying power to the Oxygen Concentrator.
CAUTION: When the automobile in which you are using the Oxygen
Concentrator unit is turned off, disconnect and remove the device
from the automobile. Do not store the Oxygen Concentrator in avery hot or cold automobile or in other similar, high-or low-temperature environments.
Note: Portable and mobile RF communications equipment can effectmedical electrical equipment.

Recommended Operating Environments
For correct use of the device refer to the following chart. The
chart provides important information about the recommended
operating environments, or operating conditions.

Specifications
Operating 32º F to 104º F (0º C to 40º C)
Temperature
Operating
Humidity

10% - 95% at an 82.4° F (28° C) dew point

Transport/ -13°F to 158°F (-25°C to 70°C)
Storage
Humidity: Up to 95% Non-Condensing
Temperature-4°F to 122°F (-20°C to 50°C) for battery packs
Electrical

Use no extension cords. Use no electrical outlets controlled by a switch.

Altitude0 – 13,123 feet (0 - 4,000 meters)

Placement

DO NOT block the air inlet (top back cover) or the exhaustvent (back bottom right). Place device a minimum of 3
inches (7.5 cm) away from walls, draperies, furniture, etc.

Environment Must be smoke, pollutant, and fume free.
OperatingTime

24-hours a day when connected to an external AC or DC
power source.

Note:IftheOxygenConcentratorhasbeenstoredforanextendedperiodof time outside its normal operating temperature range, the unit shouldbe allowed to return to normal operating temperature before being
turned on. (Refer to the Specifications section in this manual.)

Proper Placement of the eQuinox
Pick a location for the device that avoids the intake of smoke,
fumes and pollutants. Correct positioning of the device should
allow intake of air through the air inlet filter at the top rear of
the cabinet. It should also allow exhaust air to freely leave theexhaust vent at the bottom right of the device.
Place the device so that the alarms can be heard.
Position the oxygen supply tubing in such a way that it does notkink or occlude.
WARNING: DO NOT USE YOUR OXYGEN CONCENTRATORIN THE PRESENCE OF FLAMMABLE GASES. THIS CANRESULT IN RAPID BURNING CAUSING PROPERTY DAMAGE,
BODILY INJURIES OR DEATH. USE NO OIL, GREASE, OR PETROLEUM-BASED OR OTHER FLAMMABLE PRODUCTS WITH THE OXYGEN-CARRYING ACCESSORIES OR THE OXYGEN CONCENTRATOR.
ONLY WATER BASED, OXYGEN COMPATIBLE LOTIONS OR SALVESSHOULD BE USED. OXYGEN ACCELERATES THE COMBUSTION OF
FLAMMABLE SUBSTANCES.
WARNING: THIS DEVICE SUPPLIES HIGH-CONCENTRATION OXYGEN
THAT PROMOTES RAPID BURNING. DO NOT ALLOW SMOKING OR
OPEN FLAMES WITHIN THE SAME ROOM OF (1) THIS DEVICE, OR (2)ANY OXYGEN-CARRYING ACCESSORY. FAILURE TO OBSERVE THISWARNING CAN RESULT IN SEVERE FIRE, PROPERTY DAMAGE, AND/OR CAUSE PHYSICAL INJURY OR DEATH.
Note: When the automobile in which you are using the Oxygen Concentrator unit is turned off, disconnect and remove the device from theautomobile. Do not store the Oxygen Concentrator in a very hot or cold
automobile or in other similar, high-or low-temperature environments.
DO NOT leave the Oxygen Concentrator or the Power Supply plugged
into the vehicle if the ignition is in the off position. Doing so may drain
the vehicle’s battery.
CAUTION: Do not operate unit in a restricted or confined
space (i.e., a small case or handbag) where ventilation
can be limited. This can cause the Oxygen Concentrator
to overheat and impair performance. Do not allow either the air
intake or the air outlet vents to become blocked. This can cause the
Oxygen Concentrator to overheat and impair performance.

eQuinox (Model 4000) User Manual — 9

Operating Instructions
Before Operating

General Charging Information

This user manual is your reference to help you use and maintainthe device. If you have any questions or concerns please callyour home care provider.

The eQuinox runs on external AC power, DC power, or arechargeable battery. The system includes a “smart battery
charger.” This recharges the battery whenever the eQuinox isconnected to an outside power source. The system monitors andcontrols the power source and the battery charger.

Important! DO NOT try to use the eQuinox without first reading
the Safety Guidelines section of this manual. Please follow allof the operating instructions. Please observe all Warnings on thedevice and in the User Manual. In order to decrease the risk offire, personal injury and serious damage to the eQuinox, please
follow all of the safety precautions.
WARNING: CLEAN THE CABINET, CONTROL PANEL, AND
POWER CORD ONLY WITH A MILD HOUSEHOLD CLEANERAPPLIED WITH A DAMP (NOT WET) CLOTH OR SPONGE,
AND THEN WIPE ALL SURFACES DRY. DO NOT ALLOW ANY LIQUID TOGET INSIDE THE DEVICE. PAY SPECIAL ATTENTION TO THE OXYGENOUTLET FOR THE CANNULA CONNECTION TO MAKE SURE IT REMAINS FREE OF DUST, WATER, AND PARTICLES.
CAUTION: Do not allow either the air intake or the air outlet vents to become blocked. This can cause the Oxygen
Concentrator to overheat and impair performance. Do not
operate the Oxygen Concentrator without the air intake filter inplace. If a second filter is provided, insert the “replacement” filter
before you clean the dirty filter. Clean the dirty filter in a warm
soap and water solution then dry thoroughly prior to use.

The eQuinox charges the battery when using AC or DC power
when the battery temperature is within safe charging temperature.
When external power is removed, the device automaticallyswitches over to the battery, as long as the battery is installed andcharged. When connected to an AC or DC outlet again, the device
runs on the external power source and recharges the battery.
Line up the power plug with the recessed external connector tocorrectly plug in the AC Supply or 12VDC cable. The recessed
power connector is on the right side of the device when facing theControl Panel. Push the power plug into the recessed connector.

AC POWER SUPPLYPowering on your eQuinox
The device can be operated from three (3) different powersources:• AC Power Supply• 12VDC Cable• Rechargeable Battery Pack (Battery)AC Power Supply

The eQuinox includes a universal AC Power Supply for use athome or wherever standard AC power is available. To connect toan AC power source, plug the AC Power Supply cord securely
into the recessed external connector. Also make sure that the cordfrom the AC Power Supply is plugged into an AC outlet. Whenthe eQuinox is plugged in correctly, a green indicator on the AC
Power Supply and the External Power Indicator will light up. The
on the user control panel will also light up. The
indicator
CAIRE AC Power Supply PN 4836 is the main power source forthe eQuinox portable oxygen concentrator. It is a very importantpart of the overall system.
Note: Use only electrical voltage specified on the specification label
affixed to the device.
Step 1		

10 — 4805-C1 C (English)

Step 2

Step 3

WARNING: DO NOT TOUCH THE BATTERY CONTACTS AND
THE PATIENT AT THE SAME TIME WHEN ENERGIZING THE
UNIT WITH THE EXTERNAL POWER SUPPLY (AC POWER
SUPPLY) AND WITH NO BATTERY PACK INSTALLED.
CAUTION: Use only accessories recommended by the
manufacturer. Use of any other may be hazardous, cause
serious damage to your oxygen concentrator and willvoid the warranty.
CAUTION: Do not use extension cords with this unit or connect too
many plugs into the same electrical outlet. The use of extension
cords could adversely affect the performance of the device. Toomany plugs into one outlet can result in an overload to the electrical panel causing the breaker/fuse to activate or fire if the breakeror fuse fails to operate.
CAUTION: Use of cables and adapters other than those specified,with the exception of cables and adapters sold by the manufacturerof the medical electrical equipment as replacement parts for internal components, may result in increased emissions of decreased
immunity of the Oxygen Concentrator.

12VDC Cable
12VDC Cable

A 12VDC Cable allows the system to operate from DC outlets,
such as those found in motor vehicles.
1. Start your vehicle.
2. Insert the 12 Volt DC cord into the recessed power
receptacle on the side of the device.
3. Insert the 12VDC Cable plug into the DC power outlet inyour motor vehicle.
When the device is properly connected and receiving powerfrom the DC power source, the green External Power Indicatoron the control panel will illuminate.

eQuinox DC Supply Connection
When the eQuinox is connected to the DC supply in a vehicle, thefollowing applies:• The eQuinox will run at all continuous flow settings 0.5–
3.0LPM and all pulse settings 1–9 while being operated with
the DC Supply in a car.
• The battery will only charge if there is sufficient power
remaining after the device is running at the prescribed flow
rate. The eQuinox will dedicate the power supplied to run
the unit first. Any power left over will be used as a trickle
charge to re-charge the battery. If the eQuinox battery is being
charged, the battery symbol will waterfall. If the eQuinox
battery is not charging, the battery symbol will be displayedsolid.
WARNING: DO NOT ALLOW EITHER THE AIR INTAKE OR THE
AIR OUTLET VENTS TO BECOME BLOCKED. THIS CAN CAUSETHE OXYGEN CONCENTRATOR TO OVERHEAT AND IMPAIRPERFORMANCE.
WARNING: IF THE VEHICLE’S DC POWER SOURCE DROPS BELOW 11.5
VOLTS, THE EQUINOX WILL REVERT TO THE BATTERY OPERATION (IFTHE BATTERY IS PRESENT AND CHARGED).
CAUTION: When the automobile in which you are using
the Oxygen Concentrator unit is turned off, disconnect
and remove the device from the automobile. Do not store
the Oxygen Concentrator in a very hot or cold automobile or inother similar, high-or low-temperature environments. DO NOT leave
the Oxygen Concentrator or the Power Supply plugged into thevehicle if the ignition is in the OFF position. Doing so may drain thevehicle’s battery.
Note: While operating the device from a 12VDC Cable, the Battery may
not charge.
Note:WhenoperatingtheeQuinoxfroma12VDCCable,itisrecommended that the Pulse Dose Mode be used if prescribed to use a pulse dose
device.
Note: To ensure that the device is utilizing the DC power, check that theExternal Power Indicator is illuminated. If a beep is heard, this could indicate a loss of external power. Check if the green External Power Indicator
LED is on or off. If the battery is removed on DC power, it will ensure noloss of charge while the unit is running. If the indicator is not illuminated,the battery is powering the device.
Note:ActualperformanceoftheeQuinoxmayvarydependinguponusage
conditionsofthemotorvehicle.Somevehiclesmaynotbeabletosupport
full functionality on DC power. In the event the vehicle cannot provide
enough power to operate at the current setting, the eQuinox will firststop charging the battery and continue to run at the prescribed flow rate.
If the vehicle still cannot provide enough power, the eQuinox will thenlower the flow rate if the flow rate is 2.0LPM or greater. If the flow rate isless than 2.0LPM, the eQuinox will shut down.
If low flow rates are experienced, remove all other electronic devices
connected to the power circuit (GPS, cell phone charger, etc.). If sufficient
power is restored, flow rates are not automatically increased back to theoriginal setting; user will need to manually adjust. If the eQuinox is still
not operating at the prescribed flow rate, attempt to remove the battery
pack.
eQuinox (Model 4000) User Manual — 11

Rechargeable Battery Pack (Battery)

Note:OperatingtheOxygenConcentratoroutsideofitsnormaloperating
temperaturerangecanaffectperformanceanddecreasebatteryruntime
and/orincreasebatterychargetime.(RefertotheSpecificationssectionin
this manual.)You will achieve longer operating time on the Battery if you
operate your eQuinox in the Pulse Mode (Physician approved mode).

The eQuinox can be run by the rechargeable
Battery Pack (Battery), which is supplied withthe device.

Installing the Battery: Line up the Battery so it
WARNING: FOR OXYGEN CONCENTRATORS EQUIPPED WITH
will slide into the empty Power Compartment.
BATTERIES: STORE IN A COOL AND DRY LOCATION TO HELP
Push the Battery into the empty PowerENSURE THE LONGEVITY OF YOUR BATTERY. STORING YOUR
Compartment until it locks (clicks) into
OXYGEN CONCENTRATOR FOR EXTENDED PERIODS OF TIME AT HIGH
position. It will be flush with the front of the
TEMPERATURES OR WITH A FULLY CHARGED OR COMPLETELY DISdevice. When the Battery is properly installed
CHARGED BATTERY CAN DEGRADE ITS OVERALL BATTERY LIFE. DO
and the eQuinox is ON, the Battery Status
NOT ATTEMPT TO OPEN THE BATTERY; THERE ARE NO SERVICEABLEPARTS INSIDE THE BATTERY. KEEP BATTERIES AWAY FROM CHILDREN.
Gauge will show on the Control Panel. (If you
DO NOT have a Battery installed, the Battery
Initial Battery Charging
Pack Status Gauge will not light up.) The unit
Rechargeable Battery The new Battery that comes with your eQuinox is not fully
will still run on external power.
Pack (Battery)charged when it is shipped from the factory. Before using yourTo remove the Battery, push the release
eQuinox for the first time, you must first fully charge the Battery.
button on the top of the battery pack. While holding DOWN therelease button, pull the Battery away from the eQuinox.
Plug in the AC Power Supply and correctly install the Battery.
Battery Operating Times: The battery operating time is shownon the Control Panel, in both symbol and alpha-numeric formats.
When running on battery power, the battery status gauge will beshown at all times. The display screen will cycle to display thehours/minutes of battery life left. A number of factors, such as
flow setting, Pulse or Continuous Flow Mode, and breath rate
will affect the operating time. The table Typical New Battery
Pack Operating Times provides operating time estimates forthe eQuinox using a new, fully charged Battery based on flow
settings and indicated operating conditions.
Note:The eQuinox comes with a 12-cell battery when shipped from thefactory. The 24-cell battery is an optional accessory for longer battery
durations.

Battery Type12-cell

24-cell

1 LPM

2.16 hrs (2 hr, 9 min)4.84 hrs (4 hrs, 50 min)

2 LPM

1.2 hours (1 hour, 122.75 hrs (2 hrs, 45 min)

3 LPM

0.8 hrs (48 min)1.83 hrs (1 hr, 49 min)

16 mL (1.0)

2.62 hrs (2 hrs, 37 min)

5.94 hrs (5 hrs, 56 min)

48 mL (3.0)

2.38 hrs (2 hrs, 22 min)5.39 hrs (5 hrs, 23 min)

96 mL (6.0)

1.70 hrs (1 hr, 41 min)3.85 hrs (3 hrs, 51 min)Continuous
Mode

Pulse Mode
(12 BPM)

12 — 4805-C1 C (English)

The eQuinox can be used while the Battery recharges. It may
take up to 4 hours to charge, depending on the flow setting. The
Battery will also charge when the eQuinox is plugged into AC
power and turned off.
CAUTION: Only use the Manufacturer’s provided batteries. For proper battery disposal, contact your Equipment
Provider or your local government agency for disposal
requirements.

* Batteries are recyclable. *Typical New Battery Pack Operating Times
Operating Type

Allow the Battery to completely charge. The Battery is fully
charged when the Battery Pack Status Gauge on the Control
Panel no longer cascades and remains solid.

In the event of an AC or DC power interruption, the eQuinox will
beep and automatically switch to Battery operation, as long as acharged Battery is installed. When external power is returned, theBattery will automatically start charging again. If the Battery is
missing or is fully discharged when the AC power is interrupted,
the eQuinox will shut down and alarm. The Battery is fullydischarged when the Battery Pack Status Gauge on the control
panel appears empty. The device will display a yellow light andsound a low battery alarm. This will continue until the device
completely shuts down with a red light and constant alarm. The
audible alert may continue for up to 5 minutes until the device
shuts down or external power is restored.
Note: Press and hold the eQuinox ON/OFF button for a minimum of 5
seconds, to silence the Loss of Power alarm. Connecting the eQuinox toan external AC power supply for 2 minutes will also eliminate this alarm.

Typical Battery Recharge Time

Enable/Disable the Voice Module

It will normally take between 2 and 5 hours to recharge a fully
drained Battery back to 80% capacity, depending on the flow
setting being used. When the battery is fully charged, the battery
gauge will stop cascading. The battery can be charged in the unit
when the eQuinox is turned off and plugged in.
If the Battery becomes too warm during discharging, recharging
will not begin until the Battery sufficiently cools. Removing theBattery may speed up the cooling process.
The Battery can recharge from both AC and DC power sources.
The Battery will also charge from the optional Desktop Charger.

1. To enable/disable the voice module function, power on theunit by pressing the ON/Standby Key.
2. After the unit has powered on, press the No Smoking Icon
and Hidden Key once.
3. Within 1–2 seconds, the SPKR = ON or SPKR = OFF menu
item will be present.
4. Press the Increase or Decrease Flow Rate Key to toggle theselection to ON (enable voice module) or OFF (disable voice
module).
5. Within 3–5 seconds, the option selected will become active.
Note: This will only disable the voice module function once. When thedevice is turned off and device is reset, the voice module function willautomatically be enabled.

Increase/Decrease the Volume of Vocal Notifications
1. To increase or decrease the volume of vocal notifications,press the No Smoking Icon & Hidden Key twice.
2. The Volume = # - (Volume Setting Menu) will be present.

Voice Module Operation

3. Press the Increase or Decrease Flow Rate Key to increase or
decrease the volume level from 1 to 10.

Voice Module Operational Instructions
The eQuinox can provide verbal notifications or alarms andoperating status. These verbal notifications supplement the
information on the light emitting diode (LED) display. The
following information provides instructions on how to turn thevoice module function on and off, adjust loudness, and change
language.

ON/Standby
Key

ON/Standby
Indicator(Green LED)

Alert Indicator(Yellow LED)

Alarm Indicator(Red LED)
Increase Flow
Rate Key

4. After selecting the desired loudness, the option will become
active/set within 3–5 seconds.
Changing Voice Module Language
1. To change the voice module language, press the No
Smoking Icon & Hidden Key five times to access the language
options.
2. The current activated language will be indicated on thedisplay. For example, if the voice module is speaking English
when accessing the language options menu, “English” will be
the text that appears on the display.
3. Press the Increase or Decrease Flow Rate Key to scroll
through the languages available.
4. Once the desired language is displayed, it will automaticallybe set within 3–5 seconds.

External
Power
Present
Indicator(Green LED)

No
Smoking
Icon &
Hidden Key
Flow Mode
Key

Flow Mode
Indicator(Green LED)

LCD
Display

Decrease Flow
Rate KeyeQuinox (Model 4000) User Manual — 13

First Time eQuinox Operation
Step 1: Positioning Your eQuinox for Use
Place the eQuinox in a well-ventilated, well lit area. Be sure theair inlet and exhaust vents in the back are not blocked.
Position the eQuinox so that all
audible and visual indicators or
alarms can be easily seen andheard.

Air inlet Filter

Be sure the air inlet filter is inplace before operating youreQuinox. If the air inlet filter ismissing, contact your home careprovider. If the air inlet filter isdirty, wash with warm soapy
water. Then rinse it with clear
water and allow it to dry before
placing the air inlet filter back inthe device. If necessary, replace itwith a new, clean air inlet filter.
Exhaust Vent
Plug the device into an AC Power
outlet, a DC Power source, or be sure there is a fully charged
Battery installed.

Step 2: Power ON the Device and Allow it toWarm-Up
Press and hold the “ON/OFF” Button for two (2) seconds topower ON your eQuinox. A brief visual power-on self-test will
occur.
Note: If there are three beeps when the device is first connected to power
orpoweredon,thisindicatesalowbatteryconditionforthe9-voltbattery
insidethecabinetoftheeQuinox.Pleasecontactyourhomecareprovider
and request that the 9-volt battery be replaced. If three beeps continueintermittently after the unit has been power on and running for several
minutes, this is not a low 9V battery condition. Contact your healthcare
provider for service.

The eQuinox has an oxygen concentration status indicator(OCSI) built in to the device. The OCSI continually monitors
the oxygen output of the device. It uses a green light (normal),yellow light (alert), red light (alarm) and an audible alarm.
These indicators will all illuminate during the power-on
self-test at start-up. After first powering ON the device,please allow up to five (5) minutes for the device to reach its
performance specifications. When only the green indicatorlight is illuminated, the device has reached its performance
specifications.

14 — 4805-C1 C (English)

Step 3: Connect the Oxygen Supply Tubing or Nasal
Cannula to the Oxygen Outlet
Replace the cannula and oxygen supply tubing regularly, as
recommended by your home care provider. Ask your home careprovider for cannula and supply tubing cleaning and replacement
information.
Note: Ensure the cannula is fully inserted and secure.This ensures that the
OxygenConcentratorcanproperlydetectinspirationforoxygendelivery.
During inhalation, you should hear or feel oxygen flow to the prongs ofthe nasal cannula.

Step 4: Select the Flow Mode
Press the Flow Mode button to choose the mode. Pressing thisbutton more than once will switch you back and forth between
Pulse Dose Mode and Continuous Flow Mode. The correct modeand setting should be prescribed for you by your physician.
Continuous Flow Mode Operation:
When running in the Continuous Flow Mode, a constant supplyof oxygen will flow through your tubing and nasal cannula. The
oxygen is measured in measured in liters per minute (LPM). You
may use a humidifier when in Continuous Flow mode only. DO
NOT tilt, transport, or lay the eQuinox down while a humidifier
bottle is attached.
Pulse Dose Mode Operation:
When running in Pulse Mode, a bolus of oxygen, measured inmilliliters (mL), is delivered upon inhalation. DO NOT use ahumidifier in the Pulse Dose Mode.
When operating in pulse dose mode, the eQuinox supplies acontinuous flow of oxygen every 15 seconds if a breath is notdetected. The device will stop continuous flow and repeatedly
check for an inspiratory effort every 15 seconds. Once aninspiration is detected, the device will deliver a measured bolus
again. To leave Pulse Dose Mode and start Continuous Flow
Mode, push the flow mode button.
Have your physician prescribe you the correct pulse dose to makesure you receive the proper oxygen saturation. Take into accountyour needs at rest, when you exercise, and when you travel athigh elevation. Contact your care provider for prescription andboarding forms.

Pulse Dose Mode and autoSat Features
Your eQuinox has a feature called autoSAT . It delivers aconsistent-sized bolus of oxygen up to 40 breaths per minute(see Max Breath Rate table below). As your breath rateincreases, such as when you walk, the autoSAT feature servocontrols the device, delivering the set bolus size. autoSAT
Technology is proprietary to CAIRE and is used during Pulse
Mode only.
®

Max Breath Rate

Bolus Volume (+/- 15%)

Pulse Dose Setting
1
2
3
4
5
6
7
8
9

Bolus Size(± 15%) mL

Step 5: Adjust the Flow Setting to the Prescribed
Level
Using the Flow Setting Buttons, adjust the flow setting to theprescribed setting supplied by your physician.
WARNING: IT IS VERY IMPORTANT TO SELECT ONLY THE
PRESCRIBED LEVEL OF OXYGEN. DO NOT CHANGE THE
FLOW SELECTION UNLESS YOU HAVE BEEN DIRECTED TO
DO SO BY A LICENSED CLINICIAN.

Step 6: Begin Using Your eQuinox
Max Breath Rate

16
32
48
64
80
96
128
160
192

40
40
40
40
37
31
23
18
15

Note: Bolus volume decreases as breath rate exceeds published maximum breath rate
Note: The pulse dose setting 1–9 is not equal to LPM.
WARNING: PULSE DOSE MODE SETTINGS MUST BE DETERMINED FOR EACH PATIENT INDIVIDUALLY FOR THEIRNEEDS AT REST, DURING EXERCISE AND WHEN TRAVELING.

Breathe normally through the nasal cannula.
Note: You will achieve longer operating time on the Battery if you operate your eQuinox in the Pulse Mode (Physician approved mode).

Step 7: eQuinox Cart Operation
The eQuinox Cart is designed for use on walking-type surfaces.
Line up the eQuinox with the floor of the cart so it sits uprightand level.
Line up the large threaded screw to the insert at the back of
the eQuinox. Then hand-tighten the knob on the cart until theeQuinox Cart is tightly secured to the device.
Push down the push-button on the eQuinox Cart handle to adjust
the height of the handle.

WARNING: THE EQUINOX MAY NOT BE ABLE TO DETECT ALL INSPIRATORY EFFORTS IN PULSE MODE. IF THE EQUINOX DOES NOTRESPOND TO YOUR INSPIRATION EFFORTS, ASK YOUR HOME CAREPROVIDER TO CHECK THE SENSITIVITY SETTING.
Note:ToEquipmentProvider:Thefollowingoxygenadministrationaccessories are recommended for use with the Oxygen Concentrator:• Nasal Cannula with 7 feet (2.1 m) of tubing: PN 5408-SEQ• For any additional recommended accessories, please see the Accessories Catalog (PN ML-LOX0010) available on www.CAIREmedical.com.
• CAIRE offers the OxySafe™ as an optional accessory. This is intended
tobeusedinconjunctionwiththeeQuinoxconcentrator.Forcustomers
in regions requiring compliance to EN ISO 8359:1996-Ammendment
1:2012, this accessory will meet this need.
The OxySafe™ is a thermal fuse to stop the flow of gas in the eventthat the downstream cannula or oxygen tubing is ignited and burns
to the OxySafe. It is placed in-line with the nasal cannula or oxygen
tubing between the patient and the oxygen outlet of the Equniox.
For proper use of the OxySafe, always refer to the manufacturer’s
instructions (included with each OxySafe kit).
CAIRE offers an OxySafe™ kits that includes a cannula headpiece,
OxySafe™, and 7’ tubing: PN 20628667

Note: DO NOT operate the eQuinox and ambulate while a humidifier is
attached. Remove the humidifier bottle before walking. DO NOT lay theeQuinox down while attached to a humidifier bottle.

Step 8: Power Off the Device
Press and hold the “ON/OFF” Button for two (2) seconds topower OFF your eQuinox.
Store the eQuinox in a cool and dry location for a period nolonger than 1 month. The eQuinox must run for at least 2 hours
per month to ensure proper function of the unit.

eQuinox (Model 4000) User Manual — 15

User Troubleshooting TableDo Not Ignore Alarms
WARNING: THE MANUFACTURER RECOMMENDS AN ALTERNATE SOURCE OF SUPPLEMENTAL OXYGEN IN THE EVENT OF A POWER OUTAGE, ALARM CONDITION, OR MECHANICAL FAILURE. CONSULT YOUR PHYSICIAN OR EQUIPMENT PROVIDER FOR THE TYPE OF RESERVESYSTEM REQUIRED.

Yellow Light
Symptom

Possible Cause

Your Action

The yellow light is on solid.The The eQuinox has detected a 1) Remove battery pack and re-install to ensure that it is secure in the concentrator.
alarm is giving 1 beep every 30 problem with the connec- 2) If possible, install a separate battery pack in the eQuinox.
seconds.
tion to the battery pack.
3) If this problem persists, contact healthcare provider.
The yellow light is flashing.The
battery icon is flashing. The
The eQuinox battery packalarm is giving 1 beeps every 2 is warm.
minutes.

1) Re-attach the eQuinox to external power or install a fully-charged battery pack in theeQuinox. Allow warm battery pack to cool outside the concentrator for 30 minutes.
2) Re-charge the battery pack using AC power
3) Replace the battery pack if another is available.
4) If the condition persists, contact your healthcare provider.

The yellow light is flashing.The
battery icon is flashing. The
The eQuinox battery packalarm is giving 2 beeps every voltage is low.
30 seconds.

1) Plug eQuinox into AC power supply and recharge or replace battery pack with fully
charged battery pack.

1) Clean and replace cabinet inlet filter
2) Ensure that the cannula is not kinked or blocked. If used with a humidifier bottle,
The yellow light is flashing.The The eQuinox has detected a ensure that it is filled properly and not creating a blockage.
alarm is giving 1 beep every 2 flow restriction causing low 3) Ensure that the eQuinox has proper ventilation. It needs to be at least 3 inches fromminutes.
or blocked flow.
any surface to ensure the vents aren’t blocked.
4) If the problem persists, switch to an alternate source of oxygen and contact healthcare provider for assistance.
1) Ensure the air intake filter is not clogged or restricted. Clean and replace the filter ifnecessary.
The yellow light is flashing.
There is no audible alarm.

2) Ensure the eQuinox is in a well ventilated area. Make sure there are at least 3 inches
The eQuinox has detected between the back of the eQuinox and any obstructions (furniture, curtain, etc.)
low oxygen levels.
3) If operating in the car, be sure the back of the eQuinox is facing outward in the seat.
4) If the condition persists, switch to an alternate source of oxygen and contact yourhealthcare provider immediately.

16 — 4805-C1 C (English)

Red Light
Symptom

Possible Cause

Your Action
1) Ensure the air intake filter is not clogged or restricted. Clean and replace the filter ifnecessary.

The red light is flashing.
The eQuinox has detected
The alarm is giving 3 beeps
low oxygen levels
every 2 minutes.

2) Ensure the eQuinox is in a well ventilated area. Make sure there are at least 3 inches
between the back of the eQuinox and any obstructions (furniture, curtain, etc.)
3) If operating in the car, be sure the back of the eQuinox is facing outward in the seat.
4) If the condition persists, switch to an alternate source of oxygen and contact yourhealthcare provider immediately.

1) Inspect the external (AC or DC) power supply and ensure that its connections are
The eQuinox has lost power secure at the eQuinox, the transformer, and the external power outlet.
temporarilywhilerunningon 2) If possible, attempt to use another electrical (AC or DC) outlet to power the eQuinox.
The red light is flashing. The external power.
3) If the condition persists, switch to an alternate source of oxygen and contact youralarm is giving a constanthealthcare provider immediately.
beep. The LCD displays
1) Re-attach the eQuinox to external power or install a fully-charged battery pack in the********. The eQuinox isnotdeliveryoxygenand will The eQuinox has lost power eQuinox. Allow warm battery pack to cool outside the concentrator for 30 minutes.
not power on.
due to a depleted battery
2) Re-charge the battery pack using AC powerpack (battery pack) charge oran overheated battery pack. 3) Replace the battery pack if another is available.
4) If the condition persists, contact your healthcare provider.
1) Remove the battery pack and disconnect the AC or DC power supply from the eQuinox.
Theredlight shoulddisappear andtheFAILmessagewilldisappear fromthescreenwhen
The red light is solid. The
thepowersourcesareremoved.Waitapproximately20seconds,reconnectpowersourcdisplay screen says “FAIL”.
TheeQuinoxhasexperienced es, and attempt to power on the eQuinox again.
The eQuinox is not produca system malfunction.
2) If the eQuinox does power on, monitor it to determine if the FAIL message occurs
ing delivering oxygen andagain. If the FAIL message occurs again, contact your healthcare provider immediately.
will not power on.
3) IftheeQuinoxdoesnotpowerbackon,contactyourhealthcareprovidedimmediately.

eQuinox (Model 4000) User Manual — 17

Other Alarm Conditions
Symptom

While in pulse mode,the pulse mode LED
flashes quickly for 15
seconds.

Possible Cause
The eQuinox isunable to detect
the user’s breathingeffort. See page
15 for pulse mode
information.

Your Action
1) Pulse mode may not work correctly if the user has any condition that causes blocked/restricted nasal
passages. (Illness, deviated septum, etc.)
2) Ensure cannula is no longer than 7 feet in length.
3) Ensure that there are no kinks or blockage in cannula tubing.
4)Ifahumidifierbottleisconnected,removethehumidifierbottleandconnectcannuladirectlytooutlet
port. Pulse mode will not work with a humidifier bottle installed.
5) If the condition persists, contact your healthcare provider.
1) Verify that the outlet is providing power. Do not connect the power supply to a dimmer circuit or a
power strip.
2) Check that cable connections on power supplies are secure to the wall/vehicle and concentrator. If
using the AC power supply, check the cable connection on the power supply transformer as well.

The eQuinox is notBattery pack is drain3) Ensure that the power verification lights are on.
receiving power
ingwhilepluggedinto
from the attached - There will be a green light on the power supply box.
AC/DC power supply
power supply.
- There will be a green light on the eQuinox concentrator control panel in the shape of a power cord.

4) If power verification LEDs are not lit, remove all connections of the power supply for 20 seconds andreconnect.
5) If the condition persists, contact your healthcare provider.
The battery pack
was not fully
charged

1) Connect to AC power to recharge the battery. Verify that the battery charges for 2-5 hours and thebattery icon is full and not flashing before use.
2) Refer to the actions for “Battery Pack is Not Charging” Symptom below.

1) Ensure that you are using the flow rate prescribed by your physician and that pulse or continuous flow
The eQuinox was
The Battery packnot operating at the is being used as required.
(battery) charge is not anticipated flow
lasting as long as it
rate.
should.
1) Attempt to drain the battery pack completely. Do this by running the eQuinox on battery power until
The battery pack is the machine shuts off completely. Then remove the battery and allow it to cool for approximately 30
not preforming to minutes. After this time, re-insert the battery in the eQuinox and recharge it using AC power.
specifications.
2) If the battery still does not last as long as it should after following step 1, contact your healthcare
provider.
1) Verify that the outlet is providing power. Do not connect the power supply to a dimmer circuit or a
power strip.
2) Check that cable connections on power supplies are secure to the wall/vehicle and concentrator. If
using the AC power supply, check the cable connection on the power supply transformer as well.

The eQuinox is notreceiving external 3) Ensure that the power verification lights are on.
AC power to charge - There will be a green light on the power supply box.
the battery.
- There will be a green light on the eQuinox concentrator control panel in the shape of a power cord.
The Battery Pack (battery) is not charging.

4) If power verification LEDs are not lit, remove all connections of the power supply for 20 seconds andreconnect.
5) If the condition persists, contact your healthcare provider.
1) Remove battery pack and re-install to ensure that it is secure in the concentrator.
The eQuinox is notproperly communicating the battery
pack to charge it.

2) Inspect the eQuinox for a solid yellow light when the battery is installed or a flashing battery icon
when attempting to charge. If either of these symptoms is present, continue to step 33) If possible, install a separate battery pack in the eQuinox. If the replacement battery works properly,
then the original battery needs to be replaced.
4) If this problem persists contact healthcare provider.

The eQuinox beeps 3 The internal battery 1) Contact your healthcare provider.
times when it is first that powers thepowered on.
alarms is low
18 — 4805-C1 C (English)

Cleaning, Care & Routine MaintenanceRoutine Maintenance

Air Inlet Filter

Only a service technician trained and qualified by CAIRE should
work on the internal parts of the eQuinox.

Ambient air is drawn into the device through the air inlet locatedon the top rear of the device. Cleaning the air inlet filter is themost important maintenance activity you will perform as a user.
Cleaning the air inlet filter will keep your eQuinox performing
properly. Check this air inlet filter daily and clean it at least oncea week using the following procedure:

Preventative Maintenance (PM) should be done every year.
Talk to your provider to arrange a PM schedule. Your homecare provider or qualified technician will inspect the following
parts only when needed: alarms, internal 9-volt battery, battery,internal filters and other internal parts.
Note: If three beeps sound after the power ON self-test, this indicates the
9v battery is low in power. Contact your home care provider if you hear
three beeps.

User Care and Cleaning of the Device
Cannula Replacement
Replace your supply tubing and cannula as often as yourhome care provider recommends. Your physician or homecare provider will provide you with cleaning, disinfection andreplacement information.
Note: Always follow the cannula manufacturer’s instructions for properuse. Replace the disposable cannula as recommended by the cannula
manufactureroryourEquipmentProvider.Additionalsuppliesareavailable from your Equipment Provider.

1. Remove the air inlet filter
2. Wash the filter in warm tap water using a mild soap
detergent solution
3. Rinse the filter thoroughly with clear tap water and squeeze
out the excess water
4. Allow the filter to air dry
5. Reinsert the filter in the cabinet
Cleaning tip: Keep a clean second filter as a replacement for use
when one filter is drying.

Air inlet
FilterCAUTION: Do not operate the Oxygen Concentrator
without the air intake filter in place. If a second filter isprovided, insert the “replacement” filter before you clean
the dirty filter. Clean the dirty filter in a warm soap and water
solution then dry thoroughly prior to use.
Note: The Manufacturer does not recommend the sterilization of thisequipment.
Note: Do not operate the Oxygen Concentrator without the air intake
filter in place. If a second filter is provided, insert the“replacement”filter
before you clean the dirty filter. Clean the dirty filter in a warm soap andwater solution then dry thoroughly prior to use.

The air inlet filter should be replaced annually. If the eQuinox isused in a dusty environment, the filter may need to be replaced
more frequently. You should not operate the eQuinox without an
air inlet filter installed for more than 5 minutes.

eQuinox (Model 4000) User Manual — 19

AC Power Supply, 12VDC Cable, Power Cord and Outer Case:

Optional Humidifier

Turn OFF the eQuinox and disconnect from AC or DC power
before you clean or disinfect. DO NOT spray right onto the outercase. Use a damp (not wet) cloth or sponge. Spray the cloth orsponge with a mild cleaning solution to clean the cabinet andpower supply. To disinfect the eQuinox, use Lysol® Brand II
disinfectant (or equivalent). Follow the directions of the cleaner
manufacturer.

If used, clean your humidifier every day to decrease thepossibility of contamination. Follow the cleaning directions fromthe home care provider or the manufacturer of the humidifier.
Replace the humidifier every month, or as directed by the
manufacturer or your home care provider. (Do not overfill thehumidifier.)

WARNING: ELECTRICAL SHOCK HAZARD. TURN OFF THE
UNIT AND DISCONNECT THE POWER CORD FROM THE
ELECTRICAL OUTLET BEFORE YOU CLEAN THE UNIT TO PREVENT ACCIDENTAL ELECTRICAL SHOCK AND BURN HAZARD. ONLYYOUR EQUIPMENT PROVIDER OR A QUALIFIED SERVICE TECHNICIANSHOULD REMOVE THE COVERS OR SERVICE THE UNIT. DO NOT USE
LIQUID DIRECTLY ON THE UNIT. A LIST OF UNDESIRABLE CHEMICALAGENTS INCLUDES BUT IS NOT LIMITED TO THE FOLLOWING: ALCOHOL AND ALCOHOL-BASED PRODUCTS, CONCENTRATED CHLORINE-BASED PRODUCTS (ETHYLENE CHLORIDE), AND OIL- BASEDPRODUCTS (PINE-SOL®, LESTOIL®). THESE ARE NOT TO BE USEDTO CLEAN THE PLASTIC HOUSING ON OXYGEN CONCENTRATOR, AS
THEY CAN DAMAGE THE UNIT’S PLASTIC.
WARNING: ELECTRICAL SHOCK HAZARD. DISCONNECT THE POWER
CORD FROM THE ELECTRICAL OUTLET BEFORE YOU CLEAN THE UNIT
TO PREVENT ACCIDENTAL ELECTRICAL SHOCK AND BURN HAZARD.
ONLY YOUR EQUIPMENT PROVIDER OR A QUALIFIED SERVICE TECHNICIAN SHOULD REMOVE THE COVERS OR SERVICE THE UNIT.
WARNING: CARE SHOULD BE TAKEN TO PREVENT THE OXYGEN CONCENTRATOR AND THE BATTERY FROM GETTING WET OR ALLOWINGFLUIDS TO ENTER THE UNIT. THIS CAN CAUSE A MALFUNCTION OR
SHUT DOWN, AND CAUSE AN INCREASED RISK FOR ELECTRICALSHOCK OR BURNS.
WARNING: CLEAN THE CABINET, CONTROL PANEL, AND POWER
CORD ONLY WITH A MILD HOUSEHOLD CLEANER APPLIED WITH A
DAMP (NOT WET) CLOTH OR SPONGE, AND THEN WIPE ALL SURFACES DRY. DO NOT ALLOW ANY LIQUID TO GET INSIDE THE DEVICE. PAY
SPECIAL ATTENTION TO THE OXYGEN OUTLET FOR THE CANNULA
CONNECTION TO MAKE SURE IT REMAINS FREE OF DUST, WATER,
AND PARTICLES.
WARNING: DO NOT USE LIQUID DIRECTLY ON THE UNIT.
A LIST OF UNDESIRABLE CHEMICAL AGENTS INCLUDES
BUT IS NOT LIMITED TO THE FOLLOWING: ALCOHOL AND
ALCOHOL-BASED PRODUCTS, CONCENTRATED CHLORINE-BASEDPRODUCTS (ETHYLENE CHLORIDE), AND OIL- BASED PRODUCTS(PINE-SOL®, LESTOIL®). THESE ARE NOT TO BE USED TO CLEAN
THE PLASTIC HOUSING ON OXYGEN CONCENTRATOR, AS THEY CANDAMAGE THE UNIT’S PLASTIC.

20 — 4805-C1 C (English)

CAUTION: DO NOT operate the eQuinox and ambulate
while a humidifier is attached. Remove the humidifier bottle before walking. DO NOT lay the eQuinox down whileattached to a humidifier bottle.

eQuinox Cart
Before cleaning the eQuinox Cart, remove it from the eQuinox.
Use a damp (not wet) cloth or sponge and mild cleaning solution
to clean the cart.
Battery Pack (Battery)
The Battery in the eQuinox needs special care to make sureit has a longer life and the highest level of performance. The
CAIRE Battery is the only Battery recommended for use withthe eQuinox.
Use a damp (not wet) cloth or sponge to clean the Battery. First
spray the cloth or sponge with a mild detergent, then clean theBattery case and the latch.
CAIRE recommends replacing the battery pack after 2 years.
Battery Disposal
Your Battery is rechargeable and can be recycled. Always returnit to your home care provider for proper disposal. You can also
contact your local city or town offices for directions on proper
disposal of the Battery.
eQuinox Oxygen System Disposal
Local environmental laws may not allow disposal of electricaland/or electronic equipment such as the eQuinox or AC Power
Supply. Contact the local city or town offices for instructions on
the correct way to dispose of electrical or electronic equipment.
You may also contact CAIRE Inc. for disposal information.

EMC Tables
Guidance and Manufacturer’s Declaration­—Electromagnetic Emissions
The eQuinox is intended for use in the electromagnetic environment specified below.The customer or the user of the eQuinox should
assure that it is used in such an environment.
Emissions Test
RF emissions
CISPR 11
RF emissions
CISPR 11

ComplianceElectromagnetic Environment - GuidanceGroup 1The eQuinox uses RF energy only for its internal function. Therefore, its RF
emissions are very low and are not likely to cause any interference in nearby
electronic equipment.

Class B

Harmonic emissions
IEC 61000-3-2Class A

Voltage fluctuations/
flicker emissions
IEC 61000-3-3Complies

The eQuinox is suitable for use in all establishments, including domestic establishmentsandthosedirectlyconnectedtothepubliclow-voltagepowersupply
network that supplies buildings used for domestic purposes.

eQuinox (Model 4000) User Manual — 21

Guidance and Manufacturer’s Declaration—Electromagnetic Immunity
The eQuinox is intended for use in the electromagnetic environment specified below.The customer or the user of the eQuinox should
assure that it is used in such an environment.
Immunity test

IEC 60601 test level

Electromagneticenvi±6 kV contact
ronment – guidance±8 kV air
IEC 61000-4-2Compliance level

Electromagnetic environment – guidance±6 kV contact

Floors should be wood, concrete or ceramic tile.
If floors are covered with synthetic material, therelative humidity should be at least 30 %.

±8 kV airElectrical fast transient/burst±2 kV for power supply lines ±2 kV for power supply lines

IEC 61000-4-4±1 kV for input/output lines N/ASurge
IEC 61000-4-5±1 kV line(s) to line(s)±1 kV line(s) to line(s)±2 kV line(s) to earth<5 % UT<5 % UT(>95 % dip in UT )(>95 % dip in UT )for 0,5 cycle

for 0,5 cycle

40 % UT

40 % UT(60 % dip in UT )Voltage dips, shortfor 5 cycles
interruptions andvoltagevariationson
power supply input 70 % U
Tlines IEC 61000-4-11(30 % dip in UT )(60 % dip in UT )for 5 cycles
70 % UT(30 % dip in UT )for 25 cycles

for 25 cycles<5 % UT<5 % UT(>95 % dip in UT )(>95 % dip in UT )for 5 sec

for 5 sec

Power frequency(50/60 Hz) magnetic 3 A / mfield IEC 61000-4-8

3A/m

NOTE UT is the A.C. mains voltage prior to application of the test level.

22 — 4805-C1 C (English)

Mains power quality should be that of a typicalcommercial or hospital environment.
Mains power quality should be that of a typicalcommercial or hospital environment.

Mains power quality should be that of a typicalcommercialorhospitalenvironment.Iftheuserof
theeQuinoxrequirescontinuedoperationduring
power mains interruptions, it is recommended
thattheeQuinoxbepoweredfromanuninterruptible power supply or a battery.

Powerfrequencymagneticfieldsshouldbeatlevels characteristic of a typical location in a typicalcommercial or hospital environment.

Guidance and Manufacturer’s Declaration–Electromagnetic Immunity
The eQuinox is intended for use in the electromagnetic environment specified below.The customer or the user of the eQuinox should
assure that it is used in such an environment.
Immunity test

IEC 60601 test level

Compliance level

Electromagnetic environment – guidance

Conducted RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz to 80 MHz

Portable and mobile RF communications equipment should be used no closer to any part of theeQuinox,includingcables,thantherecommended
separationdistancecalculatedfromtheequation
applicable to the frequency of the transmitter.

Radiated RF

3 V/m

IEC 61000-4-3

80 MHz to 2,5 GHz

3 V/mRecommended separation distanced  1.2 P
d  1.2 P = 80 MHz to 800 MHz
d  1.2 P = 800 MHz to 2,5 GHz
where P is the maximum output power ratingof the transmitter in watts (W) according to thetransmitter manufacturer and d is the recommended separation distance in metres (m).
Field strengths from fixed RF transmitters, as
determined by an electromagnetic site survey,ashould be less than the compliance level in eachfrequency range. b
Interference may occur in the vicinity of equipment marked with the following symbol:

NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE2Theseguidelinesmaynotapplyinallsituations.Electromagneticpropagationisaffectedbyabsorptionandreflectionfromstructures, objects and people.
Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios,amateur radio, AM and FM radio broadcast andTV broadcast cannot be predicted theoretically with accuracy.To assess the electromagneticenvironmentduetofixedRFtransmitters,anelectromagneticsitesurveyshouldbeconsidered.Ifthemeasuredfieldstrengthinthe
locationinwhichtheeQuinoxisusedexceedstheapplicableRFcompliancelevelabove,theeQuinoxshouldbeobservedtoverifynormal
operation.Ifabnormal performance is observed, additional measures may be necessary, such as re-orienting or relocatingthe eQuinox.
abOver the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

eQuinox (Model 4000) User Manual — 23

Recommended separation distances
between portable and mobile RF communications equipment and the eQuinox
The eQuinox is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled.The customer ortheuseroftheeQuinoxcanhelppreventelectromagneticinterferencebymaintainingaminimumdistancebetweenportableandmobile
RFcommunicationsequipment(transmitters)andtheeQuinoxasrecommendedbelow,accordingtothemaximumoutputpowerofthecommunications equipment.
Rated maximum output
power of transmitter

Separation distance according to frequency of transmitterm
150 kHz to 80 MHz

80 MHz to 800 MHz

800 MHz to 2,5 GHzW

d  1.2 P

d  1.2 P

d  1.2 P

0,01

0.12

0.12

0.23

0,1

0.38

0.38

0.73

1

1.2

1.2

2.3

10

3.8

3.8

7.3

100

12

12

23

For transmitters rated at a maximum output power not listed above, the recommended separation distance d in metres (m) can be estimatedusingtheequationapplicabletothefrequencyofthetransmitter,wherePisthemaximumoutputpowerratingofthetransmitterin watts (W) according to the transmitter manufacturer.
NOTE 1 At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE2Theseguidelinesmaynotapplyinallsituations.Electromagneticpropagationisaffectedbyabsorptionandreflectionfromstructures, objects and people.

24 — 4805-C1 C (English)

General Airline Travel Information
During taxi, take-off, and landing, the eQuinox must be turned
off and stowed under the seat or in another approved stowage
location so as to not block the aisle way or the entry way into therow if the eQuinox will not be used.
During taxi, take-off, and landing, the user must be in a seat
location that does not restrict any other passenger’s access to, or
use of, any required emergency or regular exit, or the aisle(s) inthe passenger compartment of the aircraft if the eQuinox is used.
User’s are not permitted to be seated in an exit row if using theeQuinox.
If the eQuinox is used when decompression of the cabin occurs
and the oxygen cabin oxygen system deploys, then the user is to
discontinue use of the eQuinox and use the aircraft supplemental
oxygen. The eQuinox unit is to be turned off after securing theaircraft supplemental oxygen.
Prior to travel, the user needs to inspect the eQuinox to ensure it
is in good operational condition.

Battery Pack & Airline Travel
Please allow for ground time (preflight checkin, security check in, etc.) and layovers.
Most airlines require that you have an adequate number offully charged batteries to power the device for at least 150% of
the expected maximum trip duration (flight times, all ground
time; before and after flight, and during connections and for
unexpected delays).

It is suggested to always carryan extra battery with you on trips.
Each airline has their own requirements
for traveling with oxygen.
Contact your airline at least 48 hours prior to departure to informthem you will be traveling with the eQuinox.

CAUTION: The US Department of Transportation (DOT) andUnited Nations (UN) Regulations require the removal of
the battery from the device for all international airline
travel when the oxygen concentrator is checked as luggage. Whenshipping the oxygen concentrator, the battery must also be removed
from the device and packaged properly.

eQuinox (Model 4000) User Manual — 25

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107 USA
1-800-482-2473
Worldwide: www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Germany
Worldwide: www.CAIREmedical.com

CAIRE Inc. reserves the right to discontinue its products, or change the prices, materials, equipment, quality, descriptions, specifications and/or processes to its products at any time without prior notice and with no further obligation orconsequence. All rights not expressly stated herein are reserved by us, as applicable.

26 — 4805-C1 C (English)

As of the date of this publication, eQuinox, SeQual and autoSAT and their related
logos (as applicable) are trademarks registered by CAIRE Inc.
Lysol® is a registered trademark of Reckitt Benckiser, UK.
OxySafe® is a registered trademark of Applied Home Healthcare Equipment.
Pine-Sol® and Lestoil® are registered trademarks of The Clorox Company.

Manuel d'utilisation

eQuinoxTM

Modèle 4000
eQuinox (Model 4000) User Manual — 27

Commandes d'utilisation et indicateurs d'état du système
Symbole

A

02

IP22

Définition

SymboleDéfinition

Lire le manuel d'utilisation avant emploi. Se reporterau manuel d'utilisation pour consulter les instructions.

Symbole d'homologation par la FAA : l’ Administration
fédérale de l'aviation (FAA, Federal Aviation
Administration) des États-Unis a homologué l'utilisationde cet appareil à bord des avions commerciaux.

Icone d'interdiction de fumer : ne pas fumer à
proximité de l'unité.

Indicateur de débit

Indicateur d'avertissement/ALERTE (jaune)

Fonctionnement en mode intermittent

Indicateur d'ALARME (rouge)­

Fonctionnement normal de l'appareil ; bouton de misesous tension

Indicateur d'alimentation externe

Augmenter le débit

Indicateur d'état de la batterie

Diminuer le débit

Symbole de batterie

Ne pas utiliser de graisse ou d'huile.

Ampères

Maintenir à l'écart des flammes nues.

Courant alternatif

Ne contient aucune pièce nécessitant un entretien. Ne
pas ouvrir le capot.

Icône CEI pour Alimentation CC (entrée)

Surface chaude.

Sortie d'oxygène

Ne pas exposer à l'eau.

Appareil de Type BF (degré de protection contre
les chocs électriques)

Équipement protégé contre les gouttes d'eau - IPX1 :la batterie eQuinox et l'alimentation CA offrent une
protection contre les effets dangereux de la pénétrationde liquides. (IPX1, conformément à la directive
CEI 60529)

IPX1

Symbole de recyclage

Nom et adresse du fabricant

Ce symbole a pour objet d'inciter le propriétaire del'équipement à le retourner à un centre de recyclage àla fin de son cycle de vie, conformément à la directive
relative aux déchets d'équipements électriques etélectroniques (DEEE).

Représentant agréé au sein de l'Union européenne

Protection contre la pénétration de gouttes d'eau etde solides : le système eQuinox offre une protection
contre les effets dangereux de la pénétration de liquides
et d'objets étrangers solides conformément au code
IP22 défini par la directive CEI 60529.

Équipement de Classe II

Certifié pour les marchés des États-Unis et du Canada,conformément aux normes des États-Unis et du Canada
en vigueur.

Cet appareil est conforme aux exigences de la
Directive 93/42/CEE relative aux dispositifs
médicaux. Il porte par conséquent le marquage
CE illustré ci‑contre.

28 — 4805-C1 C (French)

Contenu
Commandes d'utilisation et indicateurs d'état du système....................................................................................................28
Guide de démarrage rapide...............................................................................................................................................................30
Indications d'utilisation........................................................................................................................................................................33
Contre-indications d'utilisation.........................................................................................................................................................33
Consignes de sécurité...........................................................................................................................................................................33
Conditions de fonctionnement recommandées........................................................................................................................35
Instructions d'utilisation......................................................................................................................................................................36
Utilisation du module vocal...............................................................................................................................................................39
Première utilisation du système eQuinox......................................................................................................................................40
Tableau de dépannage ........................................................................................................................................................................42
Nettoyage, entretien et maintenance régulière..........................................................................................................................45
Tableaux de compatibilité électromagnétique...........................................................................................................................47
Informations générales concernant les voyages en avion......................................................................................................51
Directives sur l'utilisation de batteries durant les voyages en avion...................................................................................51

eQuinox (Model 4000) User Manual — 29

Guide de démarrage rapide

1

Déballage de votre système eQuinox

2

Familiarisation avec votre système eQuinox

Veillez à lire la totalité des avertissements, mises en garde etinformations additionnelles sur l'appareil dans le reste de ce manuel.
Familiarisez-vous avec les principales caractéristiques du systèmeeQuinox et le panneau de commande.
Port de sortied'oxygène

AVANT
Poignée

Panneaude commande

Alimentation CA

Prise d'alimentation externe

Icône d'homologation FAA

Loquet dela batterie

Emplacement
de l'étiquettede spécifications
nominales et dunuméro de série

Chariot eQuinox

BatterieARRIÈRE

Câble 12 V CC

Compartimentd'entretien

Orificed'évacuation

Batterie

30 — 4805-C1 C (French)

Filtre d'admission d'air

Le panneau de commande du systèmeeQuinox affiche des informations
importantes sur le fonctionnement
de l'appareil.
Touche et indicateur MARCHE/ARRÊT (vert) : cette touche permet de mettre l'appareil en marche ou de l'arrêter.
L'indicateur vert s'allume lorsque l'appareil est en marche etfonctionne correctement.
Touches d'augmentation ou de diminution du réglage dedébit : utilisez ces touches pour définir le réglage de débit quivous a été prescrit.

3

Mise en marche du système eQuinox

Pour connecter la source d'alimentation CA : insérez le cordon
d'alimentation CA dans la prise d'alimentation externe encastrée
située sur le côté droit de l'appareil. Connectez le cordon
d'alimentation CA à la source d'alimentation et à une prise secteur
CA.
Étape 1

Étape 2

Étape 3

Touche et indicateur de mode d'administration : cette touche
permetdepasserdumoded'administrationàdébitconstantau
moded'administrationintermittente,etinversement.Lemoded'administrationintermittenteactivelatechnologieautoSAT.
Celle-ci allonge de façon significative la durée de fonctionnementdel'appareillorsqu'ilestalimentéparlabatterie.Lorsque
cemodeestactivé,l'indicateurvertdemoded'administrationintermittentes'allume.Unedosed'oxygèneestdélivréeetl'indicateur vert clignote à chaque effort respiratoire.
Indicateur d'ALERTE (jaune) : cet indicateur s'allume pour
indiquer une condition à signaler ou une mise en garde defaiblepriorité.Vouspouvezcontinueràutiliservotresystème.
Reportez-vous au Tableau de dépannage pour connaître labonnemarcheàsuivre,oucontactervotreorganismedesoinsde santé à domicile. Lorsque l'indicateur jaune clignote, il
indiqueunealertedeprioritémoyenne.Uneréponserapideest
nécessaire(ils'agitgénéralementd'unfaibleniveaud'oxygène
ou de chargement de la batterie).
Indicateurd'ALARME(rouge):indiqueuneconditiond'alarmede haute priorité. Une réponse immédiate est nécessaire.
Reportez-vous au Tableau de dépannage et contactez votreorganismedesoinsdesantéàdomicile(ils'agitgénéralementd'une panne électrique, d'un faible niveau d'oxygène ou d'undysfonctionnement).
Écran d'information• C ou P – La lettre « C » indique le mode d'administration àdébit constant, la lettre « P » indique le mode d'administration intermittente.
• Durée restante sur la batterie – Lorsque le système
fonctionne sur l'alimentation par batterie, l'écran s'actualise
toutesles20secondespourindiquerladuréerestanteestiméede la batterie.
• Réglage (administration à débit constant ou administrationintermittente) - Chaque fois que vous mettez l'appareil soustension, le dernier mode et/ou niveau de réglage utilisé qui aétéenregistréestautomatiquementappliquéaudémarrage.
Indicateur d'état de la batterie : cet indicateur affiche le niveau
d'autonomie restante de la batterie. Chacune des cinq barres
horizontalesgrisesreprésenteenviron20%del'autonomietotale de la batterie. Lorsque la batterie est en cours de charge, lesbarresd'indicationdechargeclignotentencascade.Silabatterien'est pas installée ou est installée incorrectement, l'indicateurd'état de la batterie reste éteint ou clignote.
Indicateurd'alimentationexterne:lorsquelesystèmeeQuinox
estcorrectementconnectéetutilisel'alimentationCAoulecâble
12 V CC, cet indicateur s'affiche sur le panneau de commande.
Alarmesonore:unealarmesonorevousinformedesconditionsdefonctionnement de l'appareil, soit en cas d'avertissement ou de défaillance, soit pourconfirmer que l'utilisateur a appuyé sur une touche valide.
Module vocal : si ce module est activé, un indicateur vocal vous informe desconditionsdefonctionnementdel'appareil,deschangementsdesourced'alimentation et des conditions d'avertissement ou de défaillance.

Lorsque le système eQuinox est correctement connecté, un voyant
vert s'allume sur l'alimentation CA. L'indicateur d'alimentation
externe s'allume également sur le panneau de commande.
Installation de la batterie : faites glisser la batterie dans le compartimentvide jusqu'à ce qu'elle se verrouille en position appropriée. Elle doit
alors être parfaitement alignée sur l'avant de l'appareil.

Lorsque la batterie est correctement installée et que le systèmeeQuinox est en MARCHE ou est connecté à une alimentation CA,
l'indicateur d'état de la batterie s'affiche.

Touche et indicateur MARCHE/ARRÊT (vert)
Appuyez sur la touche « MARCHE/ARRÊT » et maintenez-laenfoncée durant deux (2) secondes pour mettre en MARCHE votre
système eQuinox. Un bref autotest sonore et visuel s'effectue audémarrage. Vérifiez que tous les indicateurs sont allumés et quel'alarme sonore retentit durant trois (3) secondes. Appuyez sur latouche « MARCHE/ARRÊT » et maintenez-la enfoncée durant
deux (2) secondes pour ARRÊTER votre système eQuinox.
eQuinox (Model 4000) User Manual — 31

4

Sélection du mode d'administration

Touche et indicateur de mode d'administration

Appuyez sur la touche de sélection du mode d'administration poursélectionner le mode souhaité. Appuyez sur cette touche de façon
répétée pour sélectionner alternativement le mode d'administrationintermittente et le mode d'administration à débit constant. En moded'administration à débit constant, la lettre « C » s'affiche à l'écran etl'oxygène est délivré à un débit continu compris entre 0,5 et 3 l/min.
En mode d'administration intermittente, la lettre « P » s'affiche à l'écran
et l'oxygène est délivré dans un bolus au début de chaque inspiration.
Les niveaux de réglage 1 à 9 du mode d'administration intermittente
délivrent un volume de bolus de 16 à 192 ml. L'administrationintermittente peut être ajustée en fonction de la sensibilité et de lafréquence respiratoire de délivrance du bolus. Les réglages de débit
appliqués dans ces deux modes doivent vous être prescrits par votre
médecin.

5

Réglage du niveau de débit prescrit

Touches d'augmentation ou de diminution duréglage de débit

À l'aide des touches + ou -, définissez le réglage de débit qui vous aété prescrit par votre médecin.

Affichage en mode d'administration à débit constantAffichage en mode d'administration intermittente

AVERTISSEMENT : IL EST TRÈS IMPORTANT DE SÉLECTIONNER LE DÉBIT D'OXYGÈNE QUI VOUS A ÉTÉ PRESCRIT. NE
MODIFIEZ PAS LE RÉGLAGE DE DÉBIT SANS Y AVOIR ÉTÉINVITÉ PAR UN MÉDECIN. LE CONCENTRATEUR D'OXYGÈNE PEUT
ÊTRE UTILISÉ PENDANT LE SOMMEIL DU PATIENT SOUS LA RECOMMANDATION D'UN CLINICIEN QUALIFIÉ.
32 — 4805-C1 C (French)

6

Utilisation du système eQuinox

Filtre d'admission d'air

Filtre d'admission d'air
Votre unité est équipée d'un filtre d'admission d'air (filtre
supplémentaire fourni). Contrôlez-le quotidiennement et nettoyez-leau moins une fois par semaine. Pour consulter les instructions denettoyage, reportez-vous à la page 19.
Connectez le tube d'alimentation en oxygène à la sortie d'oxygène
et connectez la canule à oxygène conformément aux instructions
fournies avec celle-ci ou selon les instructions de votre organismede soins de santé à domicile.

Respirez normalement au travers de la canule nasale.
Lorsque le système fonctionne sur l'alimentation par batterie, lemode d'administration intermittente offre une durée d'utilisation
supérieure.

7

Utilisation du chariot eQuinox

Alignez le système eQuinox sur le socle du chariot de façon à cequ'il repose à plat sur celui-ci et en position verticale. Alignez lagrande vis filetée sur l'orifice d'insertion situé à l'arrière de l'appareil
eQuinox. Serrez ensuite à la main la molette du chariot jusqu'à
ce que celui-ci soit correctement fixé à l'appareil. Appuyez sur lebouton-poussoir situé sur la poignée télescopique du chariot et tirez
celle-ci vers le haut afin de régler sa hauteur.

Important !
Les instructions de sécurité se définissent comme
indiqué ci-dessous :AVERTISSEMENT : INFORMATIONS IMPORTANTES DESÉCURITÉ RELATIVES AUX DANGERS POUVANT CAUSERDES BLESSURES GRAVES.
MISE EN GARDE : informations importantes pour éviter
tout endommagement du système eQuinox.
Remarque : informations exigeant une attention particulière.

Indications d'utilisation
Le système eQuinox est conçu pour l’administration d’oxygèned’appoint. L’appareil ne doit pas être utilisé à des fins deréanimation, et n’offre aucune capacité de surveillance du patient.
AVERTISSEMENT : IL EST TRÈS IMPORTANT DE SÉLECTIONNER LE DÉBIT D'OXYGÈNE QUI VOUS A ÉTÉ PRESCRIT. NE
MODIFIEZ PAS LE RÉGLAGE DE DÉBIT SANS Y AVOIR ÉTÉINVITÉ PAR UN MÉDECIN. LE CONCENTRATEUR D'OXYGÈNE PEUT
ÊTRE UTILISÉ PENDANT LE SOMMEIL DU PATIENT SOUS LA RECOMMANDATION D'UN CLINICIEN QUALIFIÉ.
AVERTISSEMENT : CONFORMÉMENT À LA LOI FÉDÉRALE DES ÉTATSUNIS, LA VENTE ET LA LOCATION DE CET APPAREIL DOIVENT SE
FAIRE SUR ORDONNANCE D'UN MÉDECIN OU D'UN PRESTATAIRE DE
SOINS MUNI D'UNE LICENCE.
AVERTISSEMENT : CET APPAREIL NE DOIT PAS ÊTRE UTILISÉ À DES
FINS DE RÉANIMATION. IL PEUT ÊTRE NÉCESSAIRE DE SURVEILLER LES PATIENTS DES SERVICES GÉRIATRIQUES, PÉDIATRIQUESOU TOUT AUTRE TYPE DE PATIENT N'ÉTANT PAS EN MESURE DECOMMUNIQUER UN MALAISE OU DE VOIR OU ENTENDRE LESALARMES PENDANT L'UTILISATION DE CET APPAREIL. LES PATIENTSSOUFFRANT DE DÉFICIENCE AUDITIVE OU VISUELLE PEUVENT AVOIRBESOIN D'UNE ASSISTANCE POUR SURVEILLER LES ALARMES.

Contre-indications d'utilisation
AVERTISSEMENT : DANS CERTAINS CAS, L'UTILISATIOND'OXYGÈNE SANS AVIS MÉDICAL PEUT S'AVÉRER DANGEREUSE. CET APPAREIL DOIT ÊTRE UTILISÉ UNIQUEMENTSUR PRESCRIPTION D'UN MÉDECIN.
AVERTISSEMENT : VEILLEZ À NE PAS L'UTILISER EN PRÉSENCE
D'ANESTHÉSIQUES INFLAMMABLES.
AVERTISSEMENT : COMME AVEC TOUT APPAREIL ÉLECTRIQUE,L'UTILISATEUR PEUT ÊTRE CONFRONTÉ À DES PÉRIODES DE
NON-FONCTIONNEMENT DÛ À DES PANNES D'ALIMENTATION
ÉLECTRIQUE OU À LA NÉCESSITÉ DE FAIRE ENTRETENIR LE SYSTÈME
EQUINOX PAR UN TECHNICIEN QUALIFIÉ. LE SYSTÈME EQUINOXN'EST PAS ADAPTÉ AUX PATIENTS POUR LESQUELS UNE TELLE
INTERRUPTIONTEMPORAIRE POURRAIT AVOIR DES CONSÉQUENCESNÉFASTES SUR LA SANTÉ.

Consignes de sécurité
AVERTISSEMENT : AUCUNE MODIFICATION DE CET ÉQUIPEMENT N'EST AUTORISÉE.
AVERTISSEMENT : LE FABRICANT RECOMMANDE DE DISPOSER D'UNE AUTRE SOURCE D'ALIMENTATION EN OXYGÈNE
EN CAS DE COUPURE DE COURANT, DE DÉCLENCHEMENTD'ALARME DE L'APPAREIL OU DE PANNE MÉCANIQUE. CONSULTEZ VOTRE MÉDECIN OU VOTRE FOURNISSEUR AGRÉÉ AU SUJET DU
TYPE DE SYSTÈME DE RÉSERVE À UTILISER.
AVERTISSEMENT : CET APPAREIL DISPENSE UN OXYGÈNE EXTRÊMEMENT PUR, SUSCEPTIBLE DE S'ENFLAMMER TRÈS VITE. NE FUMEZPAS À PROXIMITÉ ET ÉVITEZ TOUTE FLAMME NUE DANS LA MÊMEPIÈCE QUE (1) CET APPAREIL, OU QUE (2) TOUT AUTRE ACCESSOIRE
DE TRANSPORT D'OXYGÈNE. LE NON-RESPECT DE CET AVERTISSEMENT PEUT AVOIR POUR CONSÉQUENCES UN INCENDIE, DES DÉGÂTS
MATÉRIELS ET/OU DES BLESSURES GRAVES VOIRE MORTELLES.
AVERTISSEMENT : N'UTILISEZ PAS CET APPAREIL DANS UN ESPACEFERMÉ OU CONFINÉ (PAR EXEMPLE DANS UNE PETITE VALISE OU UN
SAC) OFFRANT UNE VENTILATION POTENTIELLEMENT INSUFFISANTE.
CECI POURRAIT PROVOQUER UNE SURCHAUFFE DU CONCENTRATEURD'OXYGÈNE ET UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : LE CONCENTRATEUR D'OXYGÈNE DOIT ÊTRE PLACÉÀ L'ÉCART DE TOUTE SOURCE DE FUMÉE, DE POLLUTION OU DEVAPEURS.
AVERTISSEMENT : LE RECOURS À DES ACCESSOIRES D'ADMINISTRATION D'OXYGÈNE NON PRÉVUS POUR UNE UTILISATION AVEC
CE CONCENTRATEUR D'OXYGÈNE EST SUSCEPTIBLE DE NUIRE AUXPERFORMANCES DE L'APPAREIL. LES RÉFÉRENCES DES ACCESSOIRESRECOMMANDÉS SONT INDIQUÉES DANS CE MANUEL.
AVERTISSEMENT : EN CAS DE CHUTE, D'ENDOMMAGEMENT OU
D'EXPOSITION DU CONCENTRATEUR D'OXYGÈNE À DE L'EAU, CONTACTEZ VOTRE ORGANISME DE SOINS DE SANTÉ À DOMICILE AFIN
DE PROCÉDER À L'INSPECTION OU AUX RÉPARATIONS POSSIBLES DE
L'APPAREIL. VEILLEZ À NE PAS UTILISER LE CONCENTRATEUR D'OXYGÈNE SI SA FICHE OU SON CORDON D'ALIMENTATION ÉLECTRIQUE
SONT ENDOMMAGÉS.
AVERTISSEMENT : VEILLEZ À ATTACHER CORRECTEMENT LE CONCENTRATEUR D'OXYGÈNE À L'AIDE D'UNE CEINTURE OU PAR TOUT AUTRE
MOYEN APPROPRIÉ PENDANT LE TRANSPORT DANS UN VÉHICULE
AFIN D'ÉVITER TOUTE BLESSURE OU DÉTÉRIORATION.
AVERTISSEMENT : NE LAISSEZ PAS LES ORIFICES D'ADMISSION
ET D'ÉVACUATION DE L'AIR S'OBSTRUER. VEILLEZ À NE PAS FAIRETOMBER OU INSÉRER DES OBJETS DANS LES OUVERTURES DE L'APPAREIL. CECI POURRAIT PROVOQUER UNE SURCHAUFFE DU CONCENTRATEUR D'OXYGÈNE ET UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : VEILLEZ À NE PAS REMPLIR DE FAÇON EXCESSIVEL'HUMIDIFICATEUR OPTIONNEL. REMPLISSEZ-LE AVEC DE L'EAUUNIQUEMENT, JUSQU'AU NIVEAU INDIQUÉ PAR LE FABRICANT DE
L'HUMIDIFICATEUR.
AVERTISSEMENT : VEILLEZ À NE PAS VOUS DÉPLACER AVEC LESYSTÈME EQUINOX LORSQU'UN HUMIDIFICATEUR EST CONNECTÉ.
RETIREZ LE FLACON DE L'HUMIDIFICATEUR AVANT DE VOUS DÉPLACER. VEILLEZ À NE PAS COUCHER LE SYSTÈME EQUINOX LORSQU'IL ESTCONNECTÉ À UN FLACON D'HUMIDIFICATEUR.
AVERTISSEMENT : LES RÉGLEMENTATIONS DU DÉPARTEMENT DES
TRANSPORTS DES ÉTATS-UNIS ET DES NATIONS UNIES (ONU) EXIGENTQUE LA BATTERIE SOIT RETIRÉE DE L'APPAREIL SUR TOUS LES VOLSINTERNATIONAUX SI CELUI-CI EST ENREGISTRÉ EN TANT QUE BAGAGE.
LORS DE L'EXPÉDITION DU CONCENTRATEUR D'OXYGÈNE, IL ESTÉGALEMENT IMPÉRATIF DE RETIRER LA BATTERIE DE L'APPAREIL ET
DE L'EMBALLER CORRECTEMENT.
AVERTISSEMENT : LISEZ ET ASSUREZ-VOUS D'AVOIR COMPRIS LESINFORMATIONS DE CE MANUEL AVANT D'UTILISER CET APPAREIL. SI
VOUS NE COMPRENEZ PAS LES AVERTISSEMENTS ET LES INSTRUCTIONS, CONTACTEZ VOTRE FOURNISSEUR AGRÉÉ AVANT D'ESSAYER
eQuinox (Model 4000) User Manual — 33

D'UTILISER L'APPAREIL AFIN D'ÉVITER TOUT RISQUE DE
BLESSURE OU DE DÉGÂTS MATÉRIELS.
AVERTISSEMENT : SI VOUS VOUS SENTEZ MAL À L'AISE OU
EN CAS D'URGENCE MÉDICALE, SOLLICITEZ IMMÉDIATEMENT UNEASSISTANCE MÉDICALE.
AVERTISSEMENT : L'UTILISATION DU CONCENTRATEUR D'OXYGÈNE
À UNE TEMPÉRATURE NON CONFORME AUX CARACTÉRISTIQUESTECHNIQUES PEUT LIMITER SA CAPACITÉ À DÉLIVRER LA CONCENTRATION EN OXYGÈNE SPÉCIFIÉE. REPORTEZ-VOUS À LA SECTIONCARACTÉRISTIQUES TECHNIQUES DE CE MANUEL POUR CONSULTERLES LIMITES DE TEMPÉRATURE.
AVERTISSEMENT : VEILLEZ À NE PAS UTILISER D'HUILE, DE GRAISSE,
DE PRODUITS À BASE DE PÉTROLE OU D'AUTRES SUBSTANCESINFLAMMABLES AVEC LES ACCESSOIRES DE TRANSPORT D'OXYGÈNEOU AVEC LE CONCENTRATEUR D'OXYGÈNE. SEULES DES LOTIONS ET
DES CRÈMES À BASE D'EAU ET COMPATIBLES AVEC LA PRÉSENCED'OXYGÈNE DOIVENT ÊTRE UTILISÉES. L'OXYGÈNE ACCÉLÈRE LACOMBUSTION DES SUBSTANCES INFLAMMABLES.
AVERTISSEMENT : EN CAS D'UTILISATION INCORRECTE DE LA BATTERIE, CELLE-CI PEUT SURCHAUFFER, PRENDRE FEU ET PROVOQUERDES BLESSURES GRAVES. VEILLEZ À NE JAMAIS PERCER, COGNEROU FAIRE TOMBER LA BATTERIE, MARCHER DESSUS OU L'EXPOSER
À DES IMPACTS OU DES CHOCS VIOLENTS. L'UTILISATION D'UNE
BATTERIE ENDOMMAGÉE PEUT PROVOQUER DES BLESSURES.
AVERTISSEMENT : LE CONCENTRATEUR D'OXYGÈNE NE DOIT PAS
ÊTRE UTILISÉ LORSQU'IL EST ACCOLÉ À OU EMPILÉ AVEC D'AUTRESÉQUIPEMENTS. SI UN TEL POSITIONNEMENT NE PEUT PAS ÊTREÉVITÉ, IL CONVIENT DE SURVEILLER L'APPAREIL POUR S'ASSURER
QU'IL FONCTIONNE CORRECTEMENT.
MISE EN GARDE : si une alarme se déclenche ou si vousconstatez que le concentrateur d'oxygène ne fonctionne
pas correctement, reportez-vous au Tableau de dépannage figurant dans ce manuel. Si vous ne parvenez pas àrésoudre le problème, consultez votre fournisseur agréé.
MISE EN GARDE : suivez toutes les instructions du fabricant afind'éviter l'annulation de la garantie.
AVERTISSEMENT : NE TENTEZ PAS DE RÉALISER DESOPÉRATIONS DE MAINTENANCE AUTRES QUE LES SOLUTIONS DE DÉPANNAGE DÉTAILLÉES DANS CE MANUEL.
NE RETIREZ PAS LES CAPOTS. SEUL LE FOURNISSEUR DE
L'APPAREIL OU UN TECHNICIEN QUALIFIÉ EST HABILITÉ À RETIRERLES CAPOTS OU RÉPARER L'APPAREIL.
AVERTISSEMENT : UTILISEZ EXCLUSIVEMENT LES ACCESSOIRESRECOMMANDÉS PAR LE FABRICANT. L'UTILISATION D'AUTRESACCESSOIRES POURRAIT S'AVÉRER DANGEREUSE ET ENDOMMAGER GRAVEMENT VOTRE CONCENTRATEUR D'OXYGÈNE, ET PROVOQUERAIT L'ANNULATION DE LA GARANTIE.
AVERTISSEMENT : LE FABRICANT RECOMMANDE DEDISPOSER D'UNE AUTRE SOURCE D'ALIMENTATION ENOXYGÈNE EN CAS DE COUPURE DE COURANT, DE DÉCLENCHEMENT D'ALARME DE L'APPAREIL OU DE PANNEMÉCANIQUE. CONSULTEZVOTRE MÉDECIN OUVOTRE FOURNISSEURAGRÉÉ AU SUJET DU TYPE DE SYSTÈME DE RÉSERVE À UTILISER.
AVERTISSEMENT : NE LAISSEZ PAS LES ORIFICES D'ADMISSION
ET D'ÉVACUATION DE L'AIR S'OBSTRUER. VEILLEZ À NE PAS FAIRETOMBER OU INSÉRER DES OBJETS DANS LES OUVERTURES DE
L'APPAREIL. CECI POURRAIT PROVOQUER UNE SURCHAUFFE DU
CONCENTRATEUR D'OXYGÈNE ET UNE DÉGRADATION DE SES PERFORMANCES.
34 — 4805-C1 C (French)

AVERTISSEMENT : FUMER EN UTILISANT DE L'OXYGÈNE
CONSTITUE LA PREMIÈRE CAUSE DE BLESSURES ET DEDÉCÈS LIÉS AUX INCENDIES. VOUS DEVEZ IMPÉRATIVEMENT RESPECTER LES AVERTISSEMENTS DE SÉCURITÉSUIVANTS :AVERTISSEMENT : NE LAISSEZ PERSONNE FUMER ET N'UTILISEZ PAS
DE BOUGIES OU DE FLAMMES NUES DANS LA MÊME PIÈCE QUE
L'APPAREIL OU QUE LES ACCESSOIRES DE TRANSPORT D'OXYGÈNE.
AVERTISSEMENT : FUMER LORS DU PORT D'UNE CANULE À OXYGÈNE
PEUT PROVOQUER DES BRÛLURES AU VISAGE OU MÊME LA MORT.
AVERTISSEMENT : SI VOUS FUMEZ, CES 3 GESTES PEUVENT VOUS
SAUVER LA VIE : ARRÊTEZ LE CONCENTRATEUR D'OXYGÈNE, RETIREZLA CANULE ET QUITTEZ LA PIÈCE DANS LAQUELLE SE TROUVE L'APPAREIL.
AVERTISSEMENT : DES INDICATIONS « INTERDICTION DE FUMER- UTILISATION D'OXYGÈNE » DOIVENT ÊTRE AFFICHÉES DE FAÇON
BIEN VISIBLE À VOTRE DOMICILE OU DANS TOUT LIEU OU DE L'OXYGÈNE EST UTILISÉ. LES PATIENTS ET LE PERSONNEL SOIGNANT
DOIVENT ÊTRE INFORMÉS DES DANGERS LIÉS AU FAIT DE FUMER ENPRÉSENCE OU PENDANT L'UTILISATION D'OXYGÈNE MÉDICAL.
AVERTISSEMENT : N'UTILISEZ PAS VOTRE CONCENTRATEUR D'OXYGÈNE EN PRÉSENCE DE GAZ INFLAMMABLES, AFIN D'ÉVITER TOUT
RISQUE D'EMBRASEMENT RAPIDE SUSCEPTIBLE D'ENTRAÎNER DESDÉGÂTS MATÉRIELS, DES BLESSURES OU LA MORT.
AVERTISSEMENT : LE FAIT DE RETIRER LA CANULE ET DE LA PLACERSUR UN VÊTEMENT, UN DRAP, UN SOFA OU TOUTE AUTRE MATIÈRETEXTILE PEUT PROVOQUER UN EMBRASEMENT INSTANTANÉ EN CAS
D'EXPOSITION À UNE CIGARETTE, À UNE SOURCE DE CHALEUR OU ÀUNE FLAMME.
AVERTISSEMENT : NE LAISSEZ JAMAIS UNE CANULE NASALE SUR OU
SOUS UN VÊTEMENT, UN DRAP OU UN COUSSIN. SI L'APPAREIL EST
ALLUMÉ SANS ÊTRE UTILISÉ, CES MATIÈRES RISQUENT DE S'ENFLAMMER AU CONTACT DE L'OXYGÈNE. ARRÊTEZ LE CONCENTRATEUR D'OXYGÈNE LORSQU'IL N'EST PAS UTILISÉ.
MISE EN GARDE : déconnectez toujours l'alimentation CA
de la prise secteur murale avant de déconnecter l'alimentation CA du concentrateur d'oxygène.
MISE EN GARDE : disposez toujours le tube d'alimentation en oxygène et les cordons d'alimentation de façon à prévenir tout risquede chute.
MISE EN GARDE : lorsque le concentrateur d'oxygène est utilisé
dans une voiture, sur un bateau ou sur d'autres sources d'alimentation CC avec le câble 12 V CC, assurez-vous que le véhicule est
en marche avant de connecter l'appareil. Si l'indicateur d'alimentation externe du système eQuinox ne s'affiche pas et nécessite uneréinitialisation, déconnectez le câble 12 V CC de la prise d'alimentation CC, redémarrez le véhicule, puis reconnectez votre câble
12 V CC dans la prise CC. Si ces instructions ne sont pas respectées,le concentrateur d'oxygène risque de ne pas être alimenté.
MISE EN GARDE : lorsque le moteur du véhicule dans lequel vous
utilisez le concentrateur d'oxygène est arrêté, déconnectez l'appareil et retirez-le du véhicule. Ne stockez pas le concentrateur d'oxygène dans un véhicule dont l'habitacle peut devenir très chaud ou
très froid, ou dans tout environnement similaire pouvant présenterdes températures particulièrement faibles ou élevées.
Remarque:leséquipementsportablesetmobilesdecommunicationparradiofréquencespeuventaffecterlefonctionnementdeséquipementsmédicauxélectriques.

Conditions de fonctionnement recommandées
Pour une utilisation correcte de l'appareil, reportez-vous au tableausuivant. Ce tableau fournit des informations importantes sur lesconditions de fonctionnement recommandées.

Caractéristiques techniques
Température defonctionnement

0 ºC à 40 ºC (32 ºF à 104 ºF)

Positionnement approprié du système eQuinox
Choisissez un emplacement évitant à l'appareil d'aspirer des fumées,des vapeurs et des polluants. Un positionnement correct de l'appareil
doit permettre l'admission d'air au travers du filtre d'admission d'air
situé en haut du capot arrière du boîtier. Il doit également permettre
à l'air de s'évacuer librement par l'orifice d'évacuation situé en bas àdroite de l'appareil.
Placez l'appareil de façon à pouvoir entendre les alarmes.

Humidité

10 à 95 % à un point de rosée de 28 °C (82,4 °F)

Température detransport/stockage-25 °C à 70 °C (-13 °F à 158 °F)

Caractéristiquesélectriques

N'utilisez pas de rallonges ni de prises secteur contrôlées par un interrupteur.

Altitude

Humidité : Jusqu'à 95 % sans condensation-20 °C à 50 °C (-4 °F à 122 °F) pour les batteries

0 à 4 000 mètres (0 à 13 123 pieds)

Emplacement

VEILLEZ À NE PAS obstruer l'orifice d'admission d'air (enhaut du capot arrière) ou l'orifice d'évacuation (en basà droite du capot arrière). Placez l'appareil à au moins
7,5cm(3pouces)desmurs,desrideaux,desmeubles,etc.

Environnement

Ne doit être exposé à aucun polluant, aucune fumée etaucune vapeur.

Durée defonctionnement

24 heures par jour lorsque l'appareil est connecté à une
source d'alimentation CA ou CC externe.

Remarque : si le concentrateur d'oxygène a été stocké de façon prolongée
à une température non conforme à sa plage de températures normales defonctionnement, il est impératif de laisser l'appareil reprendre une température normale avant de le mettre en marche. (Reportez-vous à la section
Caractéristiques techniques de ce manuel.)

Positionnez le tube d'alimentation en oxygène de façon à ce qu'il neforme pas de coudes ou de plis.
AVERTISSEMENT : N'UTILISEZ PAS VOTRE CONCENTRATEURD'OXYGÈNE EN PRÉSENCE DE GAZ INFLAMMABLES, AFIN
D'ÉVITER TOUT RISQUE D'EMBRASEMENT RAPIDE SUSCEPTIBLE D'ENTRAÎNER DES DÉGÂTS MATÉRIELS, DES BLESSURESOU LA MORT. VEILLEZ À NE PAS UTILISER D'HUILE, DE GRAISSE, DE
PRODUITS À BASE DE PÉTROLE OU D'AUTRES SUBSTANCES INFLAMMABLES AVEC LES ACCESSOIRES DE TRANSPORT D'OXYGÈNE OU
AVEC LE CONCENTRATEUR D'OXYGÈNE. SEULES DES LOTIONS ET
DES CRÈMES À BASE D'EAU ET COMPATIBLES AVEC LA PRÉSENCED'OXYGÈNE DOIVENT ÊTRE UTILISÉES. L'OXYGÈNE ACCÉLÈRE LACOMBUSTION DES SUBSTANCES INFLAMMABLES.
AVERTISSEMENT : CET APPAREIL DISPENSE UN OXYGÈNE EXTRÊMEMENT PUR, SUSCEPTIBLE DE S'ENFLAMMER TRÈS VITE. NE FUMEZPAS À PROXIMITÉ ET ÉVITEZ TOUTE FLAMME NUE DANS LA MÊMEPIÈCE QUE (1) CET APPAREIL, OU QUE (2) TOUT AUTRE ACCESSOIRE
DE TRANSPORT D'OXYGÈNE. LE NON-RESPECT DE CET AVERTISSEMENT PEUT AVOIR POUR CONSÉQUENCES UN INCENDIE, DESDÉGÂTS MATÉRIELS ET/OU DES BLESSURES GRAVES VOIRE MORTELLES.
Remarque : lorsque le moteur du véhicule dans lequel vous utilisez leconcentrateur d'oxygène est arrêté, déconnectez l'appareil et retirez-le duvéhicule. Ne stockez pas le concentrateur d'oxygène dans un véhicule dontl'habitacle peut devenir très chaud ou très froid, ou dans tout environnement
similaire pouvant présenter des températures particulièrement faibles ouélevées.VEILLEZÀNEPASlaisserleconcentrateurd'oxygèneoul'alimentation
connecté dans le véhicule si le moteur est arrêté, afin de ne pas décharger labatterie du véhicule.

MISE EN GARDE : n’utilisez pas cet appareil dans un
espace fermé ou confiné (par exemple dans une petite
valise ou un sac) offrant une ventilation potentiellement
insuffisante. Ceci pourrait provoquer une surchauffe duconcentrateur d'oxygène et une dégradation de ses performances. Ne laissez pas les orifices d'admission et d'évacuation de l'airs'obstruer. Ceci pourrait provoquer une surchauffe du concentrateurd'oxygène et une dégradation de ses performances.

eQuinox (Model 4000) User Manual — 35

Instructions d'utilisation
Avant utilisation

Informations générales sur le chargement

Ce manuel d'utilisation vous sert de référence pour vous aider
à utiliser et à entretenir l'appareil. Pour toute question ou
préoccupation, veuillez contacter votre organisme de soins de santéà domicile.

Le système eQuinox peut fonctionner sur une alimentation CA
externe, une alimentation CC ou une batterie rechargeable. L'appareil
intègre un chargeur de batterie intelligent qui recharge la batteriedès que le système eQuinox est connecté à une source d'alimentation
externe. Le système surveille et contrôle la source d'alimentation etle chargeur de batterie.

Important ! N'utilisez PAS le système eQuinox avant d'avoir lu lasection « Consignes de sécurité » de ce manuel. Veillez à respecter
toutes les instructions d'utilisation ainsi que les avertissements
apposés sur l'appareil ou figurant dans ce manuel. Afin de réduire
les risques d'incendie, de blessures et d'endommagement du systèmeeQuinox, veillez à respecter toutes les précautions de sécurité.
AVERTISSEMENT : NETTOYEZ LE BOÎTIER, LE PANNEAU DE
COMMANDE ET LE CORDON D'ALIMENTATION UNIQUEMENT À L'AIDE D'UN DÉTERGENT DOUX APPLIQUÉ AU
MOYEN D'UN CHIFFON OU D'UNE ÉPONGE LÉGÈREMENT HUMIDE,
PUIS ESSUYEZ TOUTES LES SURFACES POUR LES SÉCHER. ÉVITEZTOUTE PÉNÉTRATION DE LIQUIDE À L'INTÉRIEUR DE L'APPAREIL.
INSPECTEZ TOUT PARTICULIÈREMENT LA SORTIE D'OXYGÈNE AUNIVEAU DU RACCORDEMENT DE LA CANULE AFIN DE VÉRIFIER L'ABSENCE DE POUSSIÈRES, D'EAU OU AUTRES PARTICULES.
MISE EN GARDE : ne laissez pas les orifices d'admission etd'évacuation de l'air s'obstruer. Ceci pourrait provoquer
une surchauffe du concentrateur d'oxygène et une dégradation de ses performances. N'utilisez pas le concentrateur d'oxygène lorsque le filtre d'admission d'air n'est pas en place.
Si un second filtre est fourni, mettez-le en place avant de nettoyer
le filtre sale. Nettoyez le filtre encrassé à l'eau savonneuse tiède,
puis laissez sécher totalement avant de l'utiliser.

Le système eQuinox charge la batterie lorsque l'alimentation CA ou
CC est utilisée et que la température de la batterie est comprise dansla plage de températures correctes de charge. Lorsque l'alimentation
externe est déconnectée, l'appareil bascule automatiquement
sur l'alimentation par batterie, si celle-ci est installée et chargée.
Lorsqu'il est de nouveau connecté à une prise d'alimentation CA ou
CC, l'appareil fonctionne sur cette source d'alimentation externe etrecharge la batterie.
Alignez la fiche du cordon d'alimentation sur le connecteurd'alimentation externe encastré de l'appareil pour connecter
correctement l'alimentation CA ou le câble 12 V CC. Le connecteurd'alimentation externe encastré est situé sur le côté droit de l'appareil
lorsque vous vous trouvez face au panneau de commande. Poussez lafiche du cordon d'alimentation dans le connecteur encastré.

ALIMENTATION CA

Alimentation CA

Mise sous tension du système eQuinox
Il est possible d'utiliser l'appareil depuis trois (3) sourcesd'alimentation différentes :• Alimentation CA• Câble 12 V CC
• Batterie rechargeable

Le système eQuinox est fourni avec une alimentation CA universelle
pouvant être utilisée à votre domicile ou partout où une alimentation
CA standard est disponible. Pour connecter l'appareil à une sourced'alimentation CA, branchez le cordon d'alimentation CA dans leconnecteur externe encastré. Assurez-vous également que le cordon
de l'alimentation CA est branché à une prise secteur CA. Lorsque lesystème eQuinox est correctement connecté, un voyant vert s'allume
sur l'alimentation CA et l'indicateur d'alimentation externe s'affiche.
L'indicateurs'affiche également sur le panneau de commande.
Le dispositif d'alimentation électrique CA CAIRE référence 4836
constitue la source d'alimentation principale du concentrateur d'oxygène
portable eQuinox. Il s'agit d'un élément très important du système.
Remarque:utilisezl'appareiluniquementàlatensionélectriqueindiquéesur
l'étiquette des caractéristiques techniques apposée sur l'appareil.

Étape 1

36 — 4805-C1 C (French)

Étape 2

Étape 3

AVERTISSEMENT : NE TOUCHEZ PAS SIMULTANÉMENT LESCONTACTS DE LA BATTERIE ET LE PATIENT LORSQUE L'APPAREIL EST ALIMENTÉ PAR L'ALIMENTATION EXTERNE (ALIMENTATION CA) ET QUE LA BATTERIE N'EST PAS INSTALLÉE.
MISE EN GARDE : utilisez exclusivement les accessoires
recommandés par le fabricant. L'utilisation d'autres
accessoires pourrait s'avérer dangereuse et endommager
gravement votre concentrateur d'oxygène, et provoquerait l'annulation de la garantie.
MISE EN GARDE : n’utilisez pas de rallonge électrique avec cet
appareil et veillez à ne pas connecter un trop grand nombred'appareils à la même prise secteur. L'utilisation d'une rallongerisque d'affecter les performances de l'appareil. De plus, si vous
effectuez trop de branchements sur une même prise, vous risquezde provoquer une surcharge du tableau électrique et de déclencher
le disjoncteur ou de provoquer un incendie si celui-ci ne fonctionne
pas.
MISE EN GARDE : l’utilisation de câbles et d'adaptateurs autres
que ceux spécifiés, à l'exception des câbles et adaptateurs commercialisés par le fabricant de l'équipement électrique médical
pour le remplacement de composants internes, risque d'entraîner
une augmentation des émissions électromagnétiques ou une
diminution de l'immunité électromagnétique du concentrateurd'oxygène.

Câble 12 V CC

les réglages d'administration intermittente 1 à 9 lorsqu'il est
connecté à l'alimentation CC d'un véhicule.
• la batterie se charge uniquement si l'alimentation restante
est suffisante lorsque l'appareil fonctionne au débit prescrit.
L'alimentation électrique délivrée est consacrée en priorité aufonctionnement du système eQuinox. Si l'alimentation restante
est suffisante, celle-ci est utilisée comme charge de maintien
pour recharger la batterie. Si la batterie du système eQuinox
est en cours de charge, l'indicateur d'état de la batterie clignoteen cascade. Si la batterie du système eQuinox n'est pas en
cours de charge, l'indicateur d'état de la batterie s'affiche encontinu.
AVERTISSEMENT : NE LAISSEZ PAS LES ORIFICES D'ADMISSION ET D'ÉVACUATION DE L'AIR S'OBSTRUER. CECI POURRAIT PROVOQUER UNE SURCHAUFFE DU CONCENTRATEURD'OXYGÈNE ET UNE DÉGRADATION DE SES PERFORMANCES.
AVERTISSEMENT : SI L'ALIMENTATION CC DU VÉHICULE ATTEINT UNE
VALEUR INFÉRIEURE À 11,5 VOLTS, LE SYSTÈME EQUINOX BASCULE
SUR L'ALIMENTATION PAR LA BATTERIE (SI CELLE-CI EST INSTALLÉEET CHARGÉE).
MISE EN GARDE : lorsque le moteur du véhicule danslequel vous utilisez le concentrateur d'oxygène est arrêté,déconnectez l'appareil et retirez-le du véhicule. Ne stockez
pas le concentrateur d'oxygène dans un véhicule dont l'habitacle
peut devenir très chaud ou très froid, ou dans tout environnement
similaire pouvant présenter des températures particulièrement
faibles ou élevées. VEILLEZ À NE PAS laisser le concentrateur d'oxygène ou l'alimentation connecté dans le véhicule si le moteur estarrêté, afin de ne pas décharger la batterie du véhicule.

Câble 12 V CC

Un câble 12 V CC permet au système de fonctionner depuis une

prise d'alimentation CC, telle que celles disponibles dans lesvéhicules motorisés.
1. Démarrez votre véhicule.
2. Insérez le câble 12 V CC dans la prise d'alimentationencastrée située sur le côté de l'appareil.
3. Insérez la fiche du câble 12 V CC dans la prised'alimentation CC de votre véhicule.

Lorsque l'appareil est correctement connecté et est alimenté parune source CC, l'indicateur d'alimentation externes'affiche
sur le panneau de commande.
Connexion du système eQuinox à une alimentation CC
Lorsque le système eQuinox est connecté à l'alimentation CC
d'un véhicule, les remarques suivantes s'appliquent :• le système eQuinox fonctionne à tous les réglagesd'administration à débit constant (de 0,5 à 3 l/min) et à tous

Remarque : lorsque l'appareil est alimenté depuis le câble 12V CC,
il est possible que la batterie ne se charge pas.
Remarque:lorsquelesystèmeeQuinoxestalimentédepuislecâble
12VCC,ilestrecommandéd'utiliserlemoded'administrationintermittente si un tel type d'administration vous a été prescrit.
Remarque : pour vous assurer que l'appareil utilise l'alimentation
CC, vérifiez que l'indicateur d'alimentation externe est affiché.
Lorsquevousentendezunbip,celui-cipeutindiquerunecoupured'alimentation externe. Vérifiez si le voyant vert d'alimentation
externe est allumé ou éteint. Retirer la batterie lorsque l'appareil
fonctionnesurunealimentationCCpermetdes'assurerquecelle-ci
ne subit aucune perte de charge. Si l'indicateur n'est pas allumé,
l'appareil est alimenté par la batterie.
Remarque : les performances réelles du système eQuinox peuvent varier selon les conditions d'utilisation du véhicule motorisé.
Certains véhicules ne permettent pas l'utilisationde latotalité desfonctionnalités lorsque l'appareil fonctionne sur une alimentation CC. Si l'alimentation délivrée par le véhicule n'est pas suffisantepourutiliserl'appareilauréglagedéfini,lesystèmeeQuinox
cesse de charger la batterie et continue de fonctionner au débit
d'administrationprescrit.Sil'alimentationdélivréeparlevéhiculen'est toujours pas suffisante, le système eQuinox réduit le débit
d'administrationsicelui-ciestégalousupérieurà2l/min.Siledébit
d'administrationestinférieurà2l/min.,lesystèmeeQuinoxs'arrête.
Encasdefaibledébitd'administration,débrancheztouslesautres
appareils connectés au circuit d'alimentation (GPS, chargeur detéléphoneportable,etc.).Encasderestaurationd'unealimentation
suffisante, le débit d'administration n'est pas automatiquement
augmentéetrétablisurleréglaged'origine.L'utilisateurdoitalorsle
réglermanuellement.SilesystèmeeQuinoxnefonctionnetoujours
pasaudébitd'administrationprescrit,essayezderetirerlabatterie.
eQuinox (Model 4000) User Manual — 37

Batterie rechargeable

Remarque:l’utilisationduconcentrateurd'oxygèneàunetempératurenon
conforme à sa plage de températures normales de fonctionnement peutnuire aux performances du système et diminuer l'autonomie de la batterie
et/ou augmenter son délai de chargement. (Reportez-vous à la section Caractéristiquestechniquesdecemanuel.)Vousbénéficierezd'unepluslongue
autonomie de la batterie si vous utilisez votre système eQuinox en moded'administration intermittente (mode approuvé par un médecin).

Le système eQuinox peut fonctionner à partir dela batterie rechargeable fournie avec l'appareil.
Installation de la batterie : alignez la batterie
sur le compartiment vide de façon à pouvoir
la glisser à l'intérieur. Poussez la batterie dansle compartiment vide jusqu'à ce qu'elle severrouille en position appropriée. Elle doit alorsêtre parfaitement alignée sur l'avant de l'appareil.
Lorsque la batterie est correctement installéeet que le système eQuinox est en MARCHE,
l'indicateur d'état de la batterie s'affiche sur lepanneau de commande (si AUCUNE batterien'est installée, l'indicateur d'état de la batterie
ne s'affiche pas). L'appareil peut continuer defonctionner depuis une alimentation externe.
Batterie rechargeable

Pour retirer la batterie, appuyez sur le bouton
de déverrouillage situé dans la partie supérieure de celle-ci. Touten maintenant le bouton de déverrouillage ABAISSÉ, tirez sur labatterie pour l'extraire du système eQuinox.

Autonomie de la batterie : l’autonomie de la batterie est indiquée
sur le panneau de commande, sous la forme d'un symbole etde caractères alphanumériques. Lorsque le système fonctionne
sur la batterie, l'indicateur d'état de la batterie s'affiche enpermanence. L'écran d'affichage indique le nombre d'heures/
minutes d'autonomie restante de la batterie. Des facteurs variés,tels que le réglage de débit, l'utilisation du mode d'administrationintermittente ou à débit constant et la fréquence respiratoire
influent sur la durée d'autonomie de la batterie. Le tableau
Durées d'autonomie moyennes d'une batterie neuve indique lesdurées estimées d'autonomie de la batterie du système eQuinox
en fonction des réglages de débit et des conditions d'utilisation
indiquées, avec une batterie neuve et totalement chargée.
Remarque : le système eQuinox est fourni avec une batterie à 12 cellules.
Pouruneautonomiesupérieure,unebatterieà24cellulesestdisponibleen
tant qu'accessoire optionnel.

Durées d'autonomie moyennes d'une batterie neuveType d'utilisation

Type de batterie
12 cellules

24 cellules

1 l/min.

2 h 09 min

4 h 50 min

2 l/min.

1 h 12 min

2 h 45 min

3 l/min.

48 min1 h 49 min

16 ml (1.0)

2 h 37 min

5 h 56 min

48 ml (3.0)

2 h 22 min

5 h 23 min

96 ml (6.0)

1 h 41 min

3 h 51 min

Administration àdébit constant

Administrationintermittente(12 B/min.)

38 — 4805-C1 C (French)

AVERTISSEMENT : POUR LES CONCENTRATEURS D'OXYGÈNE ÉQUIPÉS D'UNE BATTERIE : AFIN DE PRÉSERVERLA LONGÉVITÉ DE VOTRE BATTERIE, STOCKEZ-LA DANSUN LIEU FRAIS ET SEC. LE STOCKAGE DU CONCENTRATEUR D'OXYGÈNE PENDANT DE LONGUES PÉRIODES À DE HAUTESTEMPÉRATURES, AVEC UNE BATTERIE TOTALEMENT CHARGÉE OUENTIÈREMENT ÉPUISÉE, PEUT ÉCOURTER LA DURÉE DE VIE DE
CELLE-CI. NE TENTEZ PAS D'OUVRIR LA BATTERIE. ELLE NE CONTIENT AUCUNE PIÈCE À ENTRETENIR OU À REMPLACER. VEILLEZ À
TENIR LA BATTERIE HORS DE PORTÉE DES ENFANTS.

Chargement initial de la batterie
À la livraison, la batterie neuve fournie avec votre système eQuinox
n'est pas totalement chargée. Avant d'utiliser votre système eQuinox
pour la première fois, vous devez charger totalement la batterie.
Connectez l'alimentation CA et installez correctement la batterie.
Attendez que la batterie se charge totalement. La batterie est
totalement chargée lorsque l'indicateur d'état de la batterie affiché
sur le panneau de commande ne clignote plus en cascade et resteaffiché en continu.
Il est possible d'utiliser le système eQuinox pendant la charge de labatterie. La charge de la batterie peut prendre jusqu'à 4 heures, selon
le réglage de débit défini. Celle-ci se charge également lorsque lesystème eQuinox est connecté à une alimentation CA et est arrêté.
MISE EN GARDE : utilisez uniquement les batteries
fournies par le fabricant. Contactez le fournisseur del'équipement ou votre mairie pour connaître les obligations en matière de recyclage des batteries.

* Les batteries sont recyclables. *En cas de coupure de l'alimentation CA ou CC, le système eQuinoxémet un bip et bascule automatiquement sur l'alimentation parbatterie, à la condition que la batterie soit installée et chargée.
Lorsque l'alimentation externe est rétablie, la batterie recommence
automatiquement à se charger. Si la batterie n'est pas installéeou est totalement déchargée lors d'une coupure de l'alimentation
CA, le système eQuinox s'arrête et émet une alarme. La batterie
est totalement déchargée lorsque l'indicateur d'état de la batterie
affiché sur le panneau de commande apparaît vide. Le voyant jaune
de l'appareil s'allume et une alarme de batterie faible est émise.
Le voyant jaune et l'alarme restent actifs jusqu'à ce que l'appareils'arrête totalement, avec le voyant rouge allumé et une alarmecontinue. L'alerte sonore peut être émise durant 5 minutes, jusqu'à
ce que l'appareil s'arrête ou que l'alimentation externe soit restaurée.
Remarque:pourarrêterl'alarmedeperted'alimentation,appuyezsurlatouche
MARCHE/ARRÊT du système eQuinox et maintenez-la enfoncée durant 5 secondesauminimum.Cettealarmes'arrêteégalementlorsquevousconnectezle
système eQuinox à une source d'alimentation CA externe durant 2 minutes.

Temps moyen de rechargement de la batterie
Selon le réglage de débit, un délai de 2 à 5 heures est normalement
nécessaire pour atteindre une capacité de 80 % à partir d'une batterie
totalement déchargée. Lorsque la batterie est totalement chargée,l'indicateur d'état de la batterie cesse de clignoter en cascade. Il est
possible de charger la batterie dans l'appareil lorsque celui-ci estarrêté et branché.
Si la batterie devient trop chaude durant son déchargement,
le rechargement ne commencera qu'une fois qu'elle se sera
suffisamment refroidie. Il est possible d'accélérer ce processus derefroidissement en retirant la batterie de l'appareil.
Il est possible de recharger la batterie à partir d'une sourced'alimentation CA ou CC. Elle peut aussi être rechargée au moyen
du chargeur de bureau optionnel.

Activer/Désactiver le module vocal
1. Pour activer/désactiver le module vocal, mettez l'appareilen marche en appuyant sur la touche Marche/Veille.
2. Lorsque l'appareil est en marche, appuyez une fois surl'icône Interdiction de fumer et sur la touche masquée.
3. Dans un délai de 1 à 2 secondes, l'élément de menu SPKR
= ON (Haut-parleur activé) ou SPKR = OFF (Haut-parleur
désactivé) s'affiche.
4. Appuyez sur la touche d'augmentation ou de diminutiondu réglage de débit pour passer de l'indication ON (activer lemodule vocal) à l'indication OFF (désactiver le module vocal)
et inversement.
5. L'option sélectionnée est activée dans un délai de 3 à
5 secondes.
Remarque:cetteopérationnedésactivelafonctiondemodulevocalequ'une
seule fois. Lorsque l'appareil est remis en marche et réinitialisé, la fonction demodule vocale est automatiquement réactivée.

Utilisation du module vocal

Augmenter/diminuer le volume des notifications vocales

Instructions d'utilisation du module vocal
Le système eQuinox peut émettre des notifications vocales relatives
aux alarmes et aux conditions de fonctionnement. Ces notifications
vocales complètent les informations affichées par le biais desvoyants DEL. Les sections suivantes fournissent des instructions
pour activer et désactiver le module vocal, régler le volume etchanger la langue des notifications.

1. Pour augmenter ou diminuer le volume des notifications
vocales, appuyez deux fois sur l'icône Interdiction de fumer etla touche masquée.
2. L'indication Volume = # - (menu de réglage du volume)
apparaît.
3. Appuyez sur la touche d'augmentation ou de diminution duréglage de débit pour augmenter ou diminuer le réglage duvolume de 1 à 10.
4. Une fois le volume souhaité sélectionné, celui-ci devient
actif dans un délai de 3 à 5 secondes.

ToucheMarche/VeilleVoyantMarche/Veille(DEL verte)

Voyant d'alarme(DEL rouge)
Voyant d'alerte
Touche d'augmen(DEL jaune)tation du réglage dedébit

Changement de la langue du module vocal
1. Pour changer la langue du module vocal, appuyez cinq fois
sur l'icône Interdiction de fumer et la touche masquée pouraccéder aux options de langue.
2. La langue actuellement activée est indiquée à l'écran.
Si, par exemple, la langue du module vocal est l'anglais,
l'indication « English » (anglais) s'affiche à l'écran.

Voyantd'alimentation
externe (DELverte)

Icône Interdiction de fumer
et touchemasquée
Touche de sélec- Indicateur detion du mode mode d'admind'administration istration (DELverte)Écran
LCD

3. Appuyez sur la touche d'augmentation ou de diminution duréglage de débit pour faire défiler les langues disponibles.
4. Lorsque la langue souhaitée est affichée, elle est
automatiquement activée dans un délai de 3 à 5 secondes.

Touche dediminution duréglage de débit

eQuinox (Model 4000) User Manual — 39

Première utilisation du système eQuinox
Étape 1 : positionnement de votre système eQuinox
Placez le système eQuinox dans un lieu correctement ventilé etéclairé. Vérifiez que les orifices d'admission et d'évacuation de l'air
situés à l'arrière de l'appareil ne sont pas obstrués.
Filtre d'admission d'air
Positionnez le système eQuinox defaçon à ce que les indicateurs ou lesalarmes sonores et visuels soientparfaitement visibles et audibles.
Vérifiez que le filtre d'admission
d'air est en place avant d'utiliser
votre système eQuinox. Si le filtred'admission d'air est manquant,contactez votre organisme de soinsde santé à domicile. Si le filtred'admission d'air est encrassé,lavez-le avec de l'eau savonneusetiède, puis rincez-le à l'eau claire
et laissez-le sécher avant de leremettre en place dans l'appareil.
Remplacez-le par un filtred'admission d'air neuf et propre si
nécessaire.

Remplacez régulièrement la canule et le tube d'alimentation en
oxygène, comme recommandé par votre organisme de soins desanté à domicile. Consultez votre organisme de soins de santéà domicile pour obtenir des informations sur le nettoyage et leremplacement de la canule et du tube d'alimentation en oxygène.
Remarque : vérifiez que la canule est entièrement insérée et convenablementpositionnéeafinqueleconcentrateurd'oxygènepuissecorrectement
détecterchaqueinspirationpourdélivrerl'oxygènerequis.Durantl'inhalation, vous devez entendre ou sentir l'oxygène circuler dans les sondes de lacanule nasale.

Étape 4 : sélection du mode d'administration

Orifice d'évacuation

Connectez l'appareil à une prise d'alimentation secteur CA ou à
une source d'alimentation CC, ou assurez-vous qu'une batterie
totalement chargée est installée.

Étape 2 : mise en MARCHE de l'appareil ettemps de préchauffage
Appuyez sur la touche « MARCHE/ARRÊT » et maintenez-laenfoncée durant deux (2) secondes pour mettre en MARCHE votre
système eQuinox. Un bref autotest visuel s'effectue au démarrage.

Remarque : si l'appareil émet trois bips lorsqu'il est connecté àl'alimentation ou mis en marche pour la première fois, ceux-ci
indiquent un mauvais état de la batterie de 9 volts installée à l'intérieurduboîtierdusystèmeeQuinox.Contactezvotreorganismede soins de santé à domicile pour demander le remplacement dela batterie de 9 volts Si trois bips continuent d'être émis de façon
intermittente une fois que l'appareil a été mis en marche et utilisé
pendant plusieurs minutes, il ne s'agit pas d'un mauvais étatde la batterie de 9 volts. Contactez votre organisme de soins desanté à domicile pour toute réparation.
Le système eQuinox comporte un indicateur de concentrationd'oxygène intégré. Cet indicateur contrôle en permanence la sortied'oxygène de l'appareil. Il intègre des voyants vert (normal), jaune(alerte) et rouge (alarme), ainsi qu'une alarme sonore. Ces voyantss'allument tous durant l'autotest réalisé au démarrage. Après avoir
mis l'appareil en MARCHE pour la première fois, attendez jusqu'à
cinq (5) minutes pour que celui-ci atteigne ses caractéristiques deperformances. Lorsque seul le voyant vert est allumé, l'appareil aatteint ses spécifications de performances.
40 — 4805-C1 C (French)

Étape 3 : connexion du tube d'alimentation en
oxygène, ou canule nasale, au port de sortie d'oxygène

Appuyez sur la touche de sélection du mode d'administration poursélectionner le mode souhaité. Appuyez sur cette touche de façon
répétée pour sélectionner alternativement le mode d'administrationintermittente et le mode d'administration à débit constant. Le
mode et le réglage appropriés doivent vous être prescrits par votre
médecin.

Fonctionnement en mode d'administration à débit constantLorsque vous utilisez l'appareil en mode d'administration à débit
constant, un flux continu d'oxygène circule dans le tube et lacanule nasale. Ce débit d'oxygène est mesuré en litres par minute(l/min.). Il est possible d'utiliser un humidificateur uniquement
en mode d'administration à débit constant. VEILLEZ À NE PAS
incliner, transporter ou coucher le système eQuinox lorsqu'unflacon d'humidificateur est connecté.

Fonctionnement en mode d'administration intermittente
Lorsque vous utilisez l'appareil en mode d'administrationintermittente, un bolus d'oxygène mesuré en millilitres (ml) est
délivré à chaque inspiration. VEILLEZ À NE PAS utiliser unhumidificateur en mode d'administration intermittente.
Lorsqu'il est utilisé en mode d'administration intermittente,le système eQuinox délivre un flux continu d'oxygène toutes
les 15 secondes si aucune inspiration n'est détectée. L'appareilcesse de délivrer un flux continu d'oxygène et vérifie de façon
répétée l'existence d'un effort respiratoire toutes les 15 secondes.
Lorsqu'une inspiration est détectée, l'appareil repasse en mode dedélivrance de bolus mesurés. Pour quitter le mode d'administrationintermittente et activer le mode d'administration à débit constant,appuyez sur la touche de sélection du mode d'administration.
Demandez à votre médecin de vous prescrire la dosed'administration intermittente appropriée pour vous assurer debénéficier du niveau de saturation en oxygène nécessaire. Tenez
compte de vos besoins au repos, durant un effort et lorsque vous
voyagez en altitude. Contactez votre médecin pour obtenir desformulaires de prescription et d'embarquement.

Mode d'administration intermittente et fonctionnalité autoSAT
Votre système eQuinox intègre une fonctionnalité nommée
autoSAT®. Celle-ci délivre un bolus d'oxygène de volume
constant jusqu'à 40 inspirations par minute (voir le tableau defréquence respiratoire max. ci-dessous). Lorsque votre fréquence
respiratoire augmente, par exemple lorsque vous marchez, lafonctionnalité autoSAT contrôle l'appareil par servocommande
afin de délivrer le volume de bolus défini. La technologie autoSAT
est une fonctionnalité exclusive de CAIRE et fonctionne en moded'administration intermittente uniquement.

Volume de bolus (+/- 15 %)Fréquence respiratoire max.
Réglage d'administrationintermittente

Volume de bolus(± 15 %) en ml

Fréquence respiratoire max.

1
2
3
4
5
6
7
8
9

16
32
48
64
80
96
128
160
192

40
40
40
40
37
31
23
18
15

Remarque : le volume de bolus diminue à mesure que la fréquence respiratoire dépasse la plage indiquée.
Remarque : le réglage d'administration intermittente de 1 à 9 n'est pas égalau réglage en l/min.

Étape 5 : réglage du niveau de débit prescrit
À l'aide des touches de réglage du débit, définissez le réglage dedébit qui vous a été prescrit par votre médecin.
AVERTISSEMENT : IL EST TRÈS IMPORTANT DE SÉLECTIONNER LE DÉBIT D'OXYGÈNE QUI VOUS A ÉTÉ PRESCRIT. NE
MODIFIEZ PAS LE RÉGLAGE DE DÉBIT SANS Y AVOIR ÉTÉINVITÉ PAR UN MÉDECIN.

Étape 6 : utilisation de votre système eQuinox
Respirez normalement au travers de la canule nasale.
Remarque : vous bénéficierez d'une plus longue autonomie de la batterie si
vousutilisezvotresystèmeeQuinoxenmoded'administrationintermittente(mode approuvé par un médecin).

Étape 7 : utilisation du chariot eQuinox
Le chariot eQuinox est conçu pour être utilisé sur des surfaces
piétonnes.
Alignez le système eQuinox sur le socle du chariot de façon à ce qu'il
repose sur celui-ci en position verticale et de niveau.
Alignez la grande vis filetée sur l'orifice d'insertion situé à l'arrière
de l'appareil eQuinox. Serrez ensuite à la main la molette du chariot
eQuinox jusqu'à ce que celui-ci soit correctement fixé à l'appareil.
Appuyez sur le bouton-poussoir situé sur la poignée du chariot
eQuinox pour régler la hauteur de celle-ci.

AVERTISSEMENT : LES RÉGLAGES DU MODE D'ADMINISTRATION INTERMITTENT DOIVENT ÊTRE DÉTERMINÉS ENFONCTION DES BESOINS INDIVIDUELS DE CHAQUE PATIENTAU REPOS, DURANT UN EFFORT ET DURANT LES VOYAGES.
AVERTISSEMENT : IL EST POSSIBLE QUE LE SYSTÈME EQUINOXNE DÉTECTE PAS TOUS LES EFFORTS RESPIRATOIRES EN MODE
D'ADMINISTRATION INTERMITTENTE. SI LE SYSTÈME EQUINOX NE
RÉPOND PAS À VOS EFFORTS RESPIRATOIRES, DEMANDEZ À VOTRE
ORGANISME DE SOINS DE SANTÉ À DOMICILE DE VÉRIFIER LERÉGLAGE DE SENSIBILITÉ.
Remarque : à l'attention du fournisseur de l'équipement : les accessoiresd'administration d'oxygène recommandés avec cet appareil sont lessuivants :• Canule nasale et tube de 2,1 mètres de long : Référence 5408-SEQ• Pour consulter des accessoires recommandés supplémentaires, reportez-vousaucatalogued'accessoires(référenceML-LOX0010)disponibleàl'adresse www.CAIREmedical.com.
• CAIREproposel'OxySafe™commeaccessoireoptionnel.Cetaccessoire
estprévupourêtreutiliséenassociationavecleconcentrateureQuinox.Il
permetderépondreauxexigencesdelanormeISO8359:1996/A1:2012.
L'OxySafe™ est un fusible thermique conçu pour stopper l'écoulement
de gaz en cas de combustion remontant depuis la canule ou la tubulured'oxygène. Il se place sur le circuit de la canule nasale/de la tubulured'oxygène, entre le patient et la sortie d'oxygène du système eQuinox.
Pour une utilisation correcte de l'OxySafe, reportez-vous toujours aux
instructions du fabricant (fournies avec chaque kit OxySafe).
CAIRE propose un kit OxySafe™ comprenant un embout de canule,
un OxySafe™ et un tube de 2,1 m (référence 20628667).

Remarque : VEILLEZ À NE PAS vous déplacer avec le système eQuinox lorsqu'unhumidificateurestconnecté.Retirezleflacondel'humidificateuravant
de vous déplacer. VEILLEZ À NE PAS coucher le système eQuinox lorsqu'il est
connecté à un flacon d'humidificateur.

Étape 8 : arrêt de l'appareil
Appuyez sur la touche « MARCHE/ARRÊT » et maintenez-laenfoncée durant deux (2) secondes pour ARRÊTER votre systèmeeQuinox.
Stockez le système eQuinox dans un lieu frais et sec durant une
période n'excédant pas 1 mois. Le système eQuinox doit être mis en
marche au moins 2 heures par mois pour garantir un fonctionnement
approprié de l'appareil.
eQuinox (Model 4000) User Manual — 41

Tableau de dépannage
N'ignorez pas les alarmes.
AVERTISSEMENT : LE FABRICANT RECOMMANDE DE DISPOSER D'UNE AUTRE SOURCE D'ALIMENTATION EN OXYGÈNE EN CAS DE
COUPURE DE COURANT, DE DÉCLENCHEMENT D'ALARME DE L'APPAREIL OU DE PANNE MÉCANIQUE. CONSULTEZ VOTRE MÉDECINOU VOTRE FOURNISSEUR AGRÉÉ AU SUJET DU TYPE DE SYSTÈME DE RÉSERVE À UTILISER.

Voyant jaune
Problème

Cause possible

Action

Le voyant jaune reste allumé de LesystèmeeQuinoxadétecté 1) Retirez la batterie et réinstallez-la de façon appropriée dans le concentrateur.
façon fixe. L'alarme émet 1 bip un problème concernant la 2) Si possible, installez une autre batterie dans le système eQuinox.
toutes les 30 secondes.
connexion de la batterie.
3) Si le problème persiste, contactez votre organisme de soins de santé à domicile.
1) Reconnectez le système eQuinox à une alimentation externe ou installez une batterie
totalement chargée dans l'appareil. Laissez la batterie excessivement chaude se refroidir à
l'extérieur du concentrateur durant 30 minutes.

Le voyant jaune clignote. L'icône
La batterie du système eQuide batterie clignote. L'alarme
2) Rechargez la batterie à l'aide d'une alimentation CA.
nox est trop chaude.
émet 1 bip toutes les 2 minutes.
3) Remplacez la batterie si une autre est disponible.

4) Si le problème persiste, contactez votre organisme de soins de santé à domicile.
Le voyant jaune clignote. L'icônede batterie clignote. L'alarme
La tension de la batterie duémet 2 bips toutes les 30 sec- système eQuinox est faible.
ondes.

1) Branchez le système eQuinox à une alimentation CA et rechargez la batterie ou remplacez-la par une batterie totalement chargée.

1) Nettoyez ou remplacez le filtre d'admission du boîtier.

Le voyant jaune clignote.
L'alarme émet 1 bip toutes les
2 minutes.

LesystèmeeQuinoxadétecté
une restriction du flux d'oxygène provoquant un débit
faible ou nul.

2)Vérifiez que la canule n'est pas coudée ou obstruée. Si vous utilisez un flacon d'humidificateur, vérifiez qu'il est correctement rempli et ne provoque pas d'obstruction.
3)Vérifiez que la ventilation du système eQuinox est appropriée. Le système doit être placé à
aumoins7cm(3pouces)detoutesurfaceafind'évitertouteobstructiondesorificesd'admission et d'évacuation.
4) Si le problème persiste, utilisez une autre source d'oxygène et contactez votre organismede soins de santé à domicile pour obtenir de l'aide.
1) Vérifiez que le filtre d'admission d'air n'est pas colmaté ou obstrué. Nettoyez le filtre ou
remplacez-le si nécessaire.

2)Vérifiez que le système eQuinox se trouve dans un lieu correctement ventilé. Assurez-vousque l'arrière du système eQuinox se trouve à au moins 7 cm (3 pouces) de tout élément
Levoyantjauneclignote.Aucune LesystèmeeQuinoxadétecté pouvant provoquer une obstruction (meuble, rideau, etc.).
alarme n'est émise.
un faible niveau d'oxygène.
3)Sivousutilisezl'appareildansunvéhicule,assurez-vousdepositionnerl'arrièredusystèmeeQuinox vers l'avant sur le siège.
4) Si le problème persiste, utilisez une autre source d'oxygène et contactez immédiatement
votre organisme de soins de santé à domicile.

42 — 4805-C1 C (French)Voyant rouge
Problème

Cause possible

Action1) Vérifiez que le filtre d'admission d'air n'est pas colmaté ou obstrué. Nettoyez le filtre ou
remplacez-le si nécessaire.

Le voyant rouge clignote.
L'alarme émet 3 bips toutes
les 2 minutes.

2)Vérifiez que le système eQuinox se trouve dans un lieu correctement ventilé. Assurez-vousque l'arrière du système eQuinox se trouve à au moins 7 cm (3 pouces) de tout élément pouLe système eQuinox a détecté vant provoquer une obstruction (meuble, rideau, etc.).
un faible niveau d'oxygène.
3) Si vous utilisez l'appareil dans un véhicule, assurez-vous de positionner l'arrière du systèmeeQuinox vers l'avant sur le siège.
4) Si le problème persiste, utilisez une autre source d'oxygène et contactez immédiatement
votre organisme de soins de santé à domicile.
1)Inspectezl'alimentation(CAouCC)externeetvérifiezqu'elleestcorrectementconnectéeau
système eQuinox, au transformateur et à la prise d'alimentation externe.

Une coupure temporaire del'alimentation externe du sys- 2)Sipossible,essayezdeconnecterlesystèmeeQuinoxàuneautreprised'alimentationCAouCC.
Le voyant rouge clignote.
tème eQuinox est survenue. 3) Si le problème persiste, utilisez une autre source d'oxygène et contactez immédiatement
L'alarme émet un bip continu.
votre organisme de soins de santé à domicile.
L'écran LCD affiche les1)ReconnectezlesystèmeeQuinoxàunealimentationexterneouinstallezunebatterietotalecaractères ********. Le
mentchargéedansl'appareil.Laissezlabatterieexcessivementchaudeserefroidiràl'extérieursystème eQuinox ne délivre L'alimentation du systèmepas d'oxygène et ne se met eQuinox a été interrompue du concentrateur durant 30 minutes.
pas en marche.
en raison du déchargement 2) Rechargez la batterie à l'aide d'une alimentation CA.
complet ou d'une surchauffe
3) Remplacez la batterie si une autre est disponible.
de la batterie.
4) Si le problème persiste, contactez votre organisme de soins de santé à domicile.
Le voyant rouge reste alluméde façon fixe. L'écran d'affichage affiche le message «Le système eQuinox a subi un
FAIL » (Échec). Le systèmedysfonctionnement.
eQuinox ne délivre pasd'oxygène et ne se met pas en
marche.

1) Retirez la batterie et déconnectez l'alimentation CA ou CC du système eQuinox. Le voyant
rouge doit s'éteindre et le message « FAIL » (Échec) disparaître de l'écran une fois les sourcesd'alimentationdéconnectées.Patientez20secondesenviron,reconnectezlessourcesd'alimentation et essayez de nouveau de mettre le système eQuinox en marche.
2) Si le système eQuinox se met en marche, surveillez l'écran pour déterminer si le message« FAIL » (Échec) s'affiche de nouveau. Si le message « FAIL » (Échec) s'affiche de nouveau,contactez immédiatement votre organisme de soins de santé à domicile.
3) Si le système eQuinox ne se met pas en marche, contactez immédiatement votre organismede soins de santé à domicile.

eQuinox (Model 4000) User Manual — 43

Autres conditions d'alarme
Problème

Cause possible

Enmoded'administration intermittente,levoyantdu
mode d'administration intermittente clignote
rapidementdurant
15 secondes.

Le système eQuinox
ne parvient pas àdétecter les efforts
respiratoires de l'utilisateur.Reportez-vousà la page 15 pour plus
d'informations sur lemode d'administration intermittente.

Action1)Ilestpossiblequelemoded'administrationintermittentenefonctionnepascorrectementsil'utilisateurprésente
uneconditionprovoquantl'obstruction/l'encombrementdesvoiesnasales(maladie,déviationdelacloisonnasale,etc.).
2) Vérifiez que la longueur de la canule n'excède pas 2,1 m (7 pieds).
3) Vérifiez que le tube de la canule n'est pas coudé ou obstrué.
4) Si un flacon d'humidificateur est connecté, retirez-le et connectez la canule directement au port de sortie. Le
mode d'administration intermittente ne fonctionne pas lorsqu'un flacon d'humidificateur est installé.
5) Si le problème persiste, contactez votre organisme de soins de santé à domicile.
1)Vérifiezquelaprisesecteurfonctionnecorrectement.Veillezànepasconnecterl'alimentationsuruncircuitdotéd'un gradateur ou sur une multiprise.

La batterie se
décharge alors
que le système
est connecté à
une alimentation
CA/CC.

2)Vérifiezquelescâblesdel'alimentationsontcorrectementconnectésàlaprisemurale/duvéhiculeetauconcentrateur. Si vous utilisez l'alimentation CA, vérifiez également la connexion du câble au transformateur.
Le système eQuinox 3) Vérifiez que les voyants de vérification d'alimentation sont allumés.
ne reçoit aucune- Un voyant vert s'allume sur le boîtier de l'alimentation.
alimentation de l'alimentationconnectée. - Un voyant vert représentant un cordon d'alimentation s'allume sur le panneau de commande du concentrateur
eQuinox.
4) Si les voyants de vérification d'alimentation sont éteints, débranchez toutes les connexions de l'alimentation
pendant 20 secondes, puis rebranchez-les.
5) Si le problème persiste, contactez votre organisme de soins de santé à domicile.
1) Connectez l'appareil à une alimentation CA pour recharger la batterie.Vérifiez que la batterie se charge durant
La batterie n'a pas été 2 à 5 heures et que l'icône de la batterie est totalement allumée et ne clignote plus avant d'utiliser l'appareil.
totalement chargée.
2) Reportez-vous aux actions décrites pour le problème « La batterie ne se charge pas » ci-dessous.

Le système eQuinox
L'autonomie denefonctionnaitpasau
la batterie n'est
débit prévu.
pas aussi longue
qu'elle le devrait.
Les performances dela batterie ne sont
pas conformes aux
caractéristiques.

1) Assurez-vous d'utiliser le niveau de débit prescrit par votre médecin et vérifiez que le mode d'administration
approprié (à débit constant ou intermittente) est sélectionné.
1) Essayez de décharger totalement la batterie. Pour ce faire, faites fonctionner le système eQuinox sur l'alimentation par batterie jusqu'à ce que l'appareil s'arrête complètement. Retirez ensuite la batterie et laissez-la refroidir
pendant30minutesenviron.Unefoiscedélaiécoulé,réinstallezlabatteriedanslesystèmeeQuinoxetrechargez-laà l'aide de l'alimentation CA.
2) Si l'autonomie de la batterie est toujours insuffisante après avoir suivi l'étape 1, contactez votre organisme desoins de santé à domicile.
1)Vérifiezquelaprisesecteurfonctionnecorrectement.Veillezànepasconnecterl'alimentationsuruncircuitdotéd'un gradateur ou sur une multiprise.
2)Vérifiezquelescâblesdel'alimentationsontcorrectementconnectésàlaprisemurale/duvéhiculeetauconcentrateur. Si vous utilisez l'alimentation CA, vérifiez également la connexion du câble au transformateur.

Le système eQuinox
3) Vérifiez que les voyants de vérification d'alimentation sont allumés.
ne reçoit aucune- Un voyant vert s'allume sur le boîtier de l'alimentation.
alimentation CA pour
- Un voyant vert représentant un cordon d'alimentation s'allume sur le panneau de commande du concentrateurcharger la batterie.
eQuinox.
4) Si les voyants de vérification d'alimentation sont éteints, débranchez toutes les connexions de l'alimentation
pendant 20 secondes, puis rebranchez-les.

La batterie ne se
charge pas.

5) Si le problème persiste, contactez votre organisme de soins de santé à domicile.
1) Retirez la batterie et réinstallez-la de façon appropriée dans le concentrateur.
Le système eQuinox
ne communique pas
correctement avec
la batterie pour lacharger.

2) Inspectez le système eQuinox pour déterminer si le voyant jaune s'allume de façon continue lorsque la batterie
est installée ou si l'icône de batterie clignote lorsque le système essaie de charger la batterie. Si l'un de ces deux
symptômes est présent, passez à l'étape 3.
3) Si possible, installez une autre batterie dans le système eQuinox. Si la batterie de rechange fonctionne correctement, alors la batterie d'origine doit être remplacée.
4) Si le problème persiste, contactez votre organisme de soins de santé à domicile.

Le système
1) Contactez votre organisme de soins de santé à domicile.
La batterie interne aleQuinox émet3 bips lors de sa imentant les alarmes
première mise en est faible.
marche.
44 — 4805-C1 C (French)

Nettoyage, entretien et maintenance régulière
Maintenance régulière

Filtre d'admission d'air

Seul un technicien qualifié et agréé par CAIRE est habilité àintervenir sur les composants internes du système eQuinox.

L'air ambiant est aspiré à l'intérieur de l'appareil via l'orificed'admission d'air situé dans la partie arrière supérieure de l'appareil.
La tâche de maintenance la plus importante que vous devrez réaliser
en tant qu'utilisateur consiste à nettoyer le filtre d'admission d'air
pour préserver les performances appropriées de votre systèmeeQuinox. Contrôlez le filtre d'admission d'air quotidiennement
et nettoyez-le au moins une fois par semaine en appliquant laprocédure suivante :Une intervention de maintenance préventive doit être réaliséechaque année. Contactez votre organisme de soins de santé àdomicile pour convenir d'un calendrier de maintenance préventive.
Votre organisme de soins de santé à domicile ou un technicien
qualifié inspectera les composants suivants uniquement si
nécessaire : alarmes, batterie interne de 9 volts, batterie, filtresinternes et autres composants internes.
Remarque:sitroisbipssontémisaprèsl'autotestréaliséaudémarrage,ceux-ci
indiquentunfaibleniveaudechargementdelabatterie9V.Contactezvotreorganismedesoinsdesantéàdomicilesivousentendeztroisbipsaudémarrage.

Entretien et nettoyage de l'appareil
Remplacement de la canule
Remplacez régulièrement votre tube d'alimentation et votre canule,comme recommandé par votre organisme de soins de santé àdomicile. Votre médecin ou votre organisme de soins de santé àdomicile vous fournira des informations concernant leur nettoyage,
leur désinfection et leur remplacement.
Remarque:pouruneutilisationcorrectedelacanule,reportez-voustoujours
auxinstructionsdufabricant.Remplacezlacanulejetableconformémentaux
recommandationsdufabricantoudufournisseurdel’équipement.D’autres
fournitures sont disponibles auprès de votre fournisseur.

1. Retirez le filtre d'admission d'air.
2. Nettoyez le filtre à l'eau courante tiède en utilisant une
solution de détergent savonneux doux.
3. Rincez soigneusement le filtre à l'eau claire et comprimezle pour évacuer l'eau en excès.
4. Laissez le filtre sécher à l'air libre.
5. Réinstallez le filtre dans le boîtier.
Conseil de nettoyage : assurez-vous de disposer d'un second filtre
propre utilisable pour remplacer un filtre en cours de séchage.

Filtre d'admission
d'air
MISE EN GARDE : n’utilisez pas le concentrateur d'oxygène lorsque le filtre d'admission d'air n'est pas enplace. Si un second filtre est fourni, mettez-le en place
avant de nettoyer le filtre sale. Nettoyez le filtre encrassé à l'eau
savonneuse tiède, puis laissez sécher totalement avant de l'utiliser.
Remarque : le fabricant ne recommande pas de stériliser cet équipement.
Remarque : n’utilisez pas le concentrateur d'oxygène lorsque le filtred'admission d'air n'est pas en place. Si un second filtre est fourni, mettez-leen place avant de nettoyer le filtre sale. Nettoyez le filtre encrassé à l'eau
savonneuse tiède, puis laissez sécher totalement avant de l'utiliser.

Le filtre d'admission d'air doit être remplacé chaque année. Si lesystème eQuinox est utilisé dans un environnement poussiéreux,
il peut être nécessaire de remplacer le filtre plus fréquemment.
N'utilisez jamais le système eQuinox durant plus de 5 minutes si lefiltre d'admission d'air n'est pas installé.

eQuinox (Model 4000) User Manual — 45

Alimentation CA, câble 12 V CC, cordon d'alimentation etboîtier extérieurArrêtez le système eQuinox et déconnectez-le de l'alimentation
CA ou CC avant toute opération de nettoyage ou de désinfection.
VEILLEZ À NE PAS vaporiser de liquides directement sur leboîtier extérieur. Utilisez une éponge ou un chiffon légèrement
humide. Vaporisez une solution de nettoyage douce sur l'éponge ou
sur le chiffon afin de nettoyer le boîtier et les alimentations. Pourdésinfecter le système eQuinox, utilisez le désinfectant Lysol®
Brand II (ou un produit équivalent). Suivez les instructions dufabricant du produit nettoyant.
AVERTISSEMENT : RISQUE DE CHOC ÉLECTRIQUE ARRÊTEZ
L'APPAREIL ET DÉBRANCHEZ LE CORDON D'ALIMENTATION DE LA PRISE ÉLECTRIQUE AVANT DE NETTOYERL'APPAREIL AFIN D'ÉVITER TOUT CHOC ÉLECTRIQUE ACCIDENTELOU RISQUE DE BRÛLURE. SEUL LE FOURNISSEUR DE L'APPAREILOU UN TECHNICIEN QUALIFIÉ EST HABILITÉ À RETIRER LES CAPOTSOU RÉPARER L'APPAREIL. VEILLEZ À NE PAS UTILISER DE LIQUIDEDIRECTEMENT SUR L'APPAREIL. LES AGENTS CHIMIQUES À ÉVITERINCLUENT DE FAÇON NON EXHAUSTIVE LES PRODUITS ALCOOLISÉSET À BASE D'ALCOOL, LES PRODUITS À BASE DE CHLORE CONCENTRÉ (CHLORURE D'ÉTHYLÈNE) ET LES PRODUITS À BASE D'HUILE(PINE-SOL®, LESTOIL®). CES PRODUITS NE DOIVENT PAS ÊTREUTILISÉS POUR NETTOYER LES PARTIES EN PLASTIQUE DU CONCENTRATEUR D'OXYGÈNE AU RISQUE DE LES ENDOMMAGER.
AVERTISSEMENT : RISQUE DE CHOC ÉLECTRIQUE DÉBRANCHEZLE CORDON D'ALIMENTATION DE LA PRISE ÉLECTRIQUE AVANT
DE NETTOYER L'APPAREIL AFIN D'ÉVITER TOUT CHOC ÉLECTRIQUE
ACCIDENTEL OU RISQUE DE BRÛLURE. SEUL LE FOURNISSEUR DE
L'APPAREIL OU UN TECHNICIEN QUALIFIÉ EST HABILITÉ À RETIRERLES CAPOTS OU RÉPARER L'APPAREIL.
AVERTISSEMENT : PRENEZ TOUTES LES PRÉCAUTIONS NÉCESSAIRES
POUR ÉVITER DE MOUILLER LE CONCENTRATEUR D'OXYGÈNE ET
LA BATTERIE ET ASSUREZ-VOUS QU'AUCUN LIQUIDE NE PUISSEPÉNÉTRER À L'INTÉRIEUR DE L'APPAREIL AFIN D'ÉVITER TOUT
DYSFONCTIONNEMENT OU ARRÊT DE CELUI-CI, AINSI QU'UN RISQUE
ACCRU DE CHOC ÉLECTRIQUE OU DE BRÛLURE.
AVERTISSEMENT : NETTOYEZ LE BOÎTIER, LE PANNEAU DE COMMANDE ET LE CORDON D'ALIMENTATION UNIQUEMENT À L'AIDE
D'UN DÉTERGENT DOUX APPLIQUÉ AU MOYEN D'UN CHIFFON OU
D'UNE ÉPONGE LÉGÈREMENT HUMIDE, PUIS ESSUYEZ TOUTES LESSURFACES POUR LES SÉCHER. ÉVITEZ TOUTE PÉNÉTRATION DE LIQUIDE À L'INTÉRIEUR DE L'APPAREIL. INSPECTEZ TOUT PARTICULIÈREMENT LA SORTIE D'OXYGÈNE AU NIVEAU DU RACCORDEMENT DE LA CANULE AFIN DE VÉRIFIER L'ABSENCE DE
POUSSIÈRES, D'EAU OU AUTRES PARTICULES.
AVERTISSEMENT : VEILLEZ À NE PAS UTILISER DE LIQUIDE DIRECTEMENT SUR L'APPAREIL. LES AGENTS CHIMIQUES À ÉVITER INCLUENT DE FAÇON NON EXHAUSTIVE LES PRODUITS ALCOOLISÉS ET À
BASE D'ALCOOL, LES PRODUITS À BASE DE CHLORE CONCENTRÉ(CHLORURE D'ÉTHYLÈNE) ET LES PRODUITS À BASE D'HUILE (PINESOL®, LESTOIL®). CES PRODUITS NE DOIVENT PAS ÊTRE UTILISÉSPOUR NETTOYER LES PARTIES EN PLASTIQUE DU CONCENTRATEURD'OXYGÈNE AU RISQUE DE LES ENDOMMAGER.

46 — 4805-C1 C (French)

Humidificateur optionnel
Si vous utilisez un humidificateur, nettoyez-le quotidiennement
pour réduire le risque de contamination. Suivez les instructions denettoyage fournies par votre organisme de soins de santé à domicile
ou par le fabricant de l'humidificateur. Remplacez l'humidificateurchaque mois ou selon les recommandations du fabricant ou de votreorganisme de soins de santé à domicile (veillez à ne pas remplir
l'humidificateur de façon excessive).
MISE EN GARDE : VEILLEZ À NE PAS vous déplacer avec
le système eQuinox lorsqu'un humidificateur est connecté. Retirez le flacon de l'humidificateur avant de vousdéplacer. VEILLEZ À NE PAS coucher le système eQuinox lorsqu'il est
connecté à un flacon d'humidificateur.

Chariot eQuinox
Avant de nettoyer le chariot, retirez-le du système eQuinox.
Nettoyez le chariot à l'aide d'une éponge ou d'un chiffon légèrement
humide et d'une solution de détergent doux.

Batterie
La batterie du système eQuinox nécessite un entretien particulier
pour offrir une plus longue durée de vie et des performances
optimales. La batterie CAIRE est la seule batterie approuvée etrecommandée pour une utilisation avec le système eQuinox.
Nettoyez la batterie à l'aide d'une éponge ou d'un chiffon légèrement
humide. Vaporisez un détergent doux sur l'éponge ou sur le chiffon,
puis nettoyez le boîtier de la batterie et le loquet.
CAIRE recommande de remplacer la batterie après 2 ans
d'utilisation.

Recyclage de la batterie
Votre batterie est rechargeable et peut être recyclée. Retournez
toujours la batterie à votre organisme de soins de santé à domicile
pour qu'elle soit correctement recyclée. Vous pouvez également
contacter vos services municipaux locaux pour obtenir desinstructions concernant le recyclage approprié de la batterie.

Mise au rebut du système d'oxygénation eQuinox
Les lois et réglementations environnementales locales peuvent
interdire la mise au rebut des équipements électriques et/ouélectroniques tels que le système eQuinox ou l'alimentation
CA. Contactez vos services municipaux locaux pour obtenir desinstructions concernant le recyclage approprié des équipementsélectriques ou électroniques. Vous pouvez également contacter
CAIRE Inc. pour obtenir des informations sur le recyclage.

Tableaux de compatibilité électromagnétique
Directives et déclaration du fabricant - Émissions électromagnétiques
LesystèmeeQuinoxestprévupouruneutilisationdanslesenvironnementsélectromagnétiquesspécifiésci-dessous.Leclientoul'utilisateurdu
système eQuinox doit s'assurer que celui-ci est utilisé dans un environnement de ce type.
Test d'émissions
Émissions RF
CISPR 11
Émissions RF
CISPR 11

Conformité

Environnement électromagnétique - Directives

Groupe 1

Le système eQuinox utilise de l'énergie RF uniquement pour son fonctionnement
interne. Par conséquent, ses émissions RF sont très faibles et ne sont pas susceptibles de provoquer des interférences avec des équipements électroniques situés àproximité.

Classe B

Émissions de courant harmonique
CEI 61000-3-2Classe A

Fluctuationsdetension/papillotement
CEI 61000-3-3

Conforme

Le système eQuinox convient à une utilisation dans tous les établissements, y comprisleshabitationsetlesétablissementsdirectementconnectésauréseaud'alimentation public basse tension qui alimente les bâtiments à usage d'habitation.

eQuinox (Model 4000) User Manual — 47

Directives et déclaration du fabricant - Immunité électromagnétique
LesystèmeeQuinoxestprévupouruneutilisationdanslesenvironnementsélectromagnétiquesspécifiésci-dessous.Leclientoul'utilisateurdu
système eQuinox doit s'assurer que celui-ci est utilisé dans un environnement de ce type.
Test d'immunité

Niveau de test CEI 60601

Environnementélectromagnétique - Contact ±6 kV
Directives
Air ±8 kV
CEI 61000-4-2

Niveau de conformité

Environnement électromagnétique - Directives

Contact ±6 kV

Les sols doivent être en bois, en ciment ou carrelés.
Si les sols sont revêtus d'un matériau synthétique,le taux d'humidité relative doit être d'au moins 30 %.

Air ±8 kV±2 kV pour les lignes d'alimen±2 kV pour les lignes d'aliTransitoiresélectriques tation
mentation
rapides/Salves
CEI 61000-4-4Surtension
CEI 61000-4-5

Creux de tension, coupures brèves et variations de tension surlescircuitsd'alimentation électrique
CEI 61000-4-11

Champ magnétiqueà la fréquence duréseau (50/60 Hz)
CEI 61000‑4-8±1 kV pour les lignes d'entrée/S/Osortie±1 kV entre fils de ligne±1 kV entre fils de ligne±2 kV entre un fil et la terre<5 % UT<5 % UT(baisse >95 % en UT )(baisse >95 % en UT )pendant 0,5 cycle

pendant 0,5 cycle

40 % UT

40 % UT(baisse de 60 % en UT )(baisse de 60 % en UT )pendant 5 cycles

pendant 5 cycles

70 % UT

70 % UT(baisse de 30 % en UT )(baisse de 30 % en UT )pendant 25 cycles

pendant 25 cycles<5 % UT<5 % UT(baisse >95 % en UT )(baisse >95 % en UT )pendant 5 spendant 5 s

3 A/m

3 A/m

La qualité de l'alimentation secteur doit être celled'un environnement commercial ou médical standard.

La qualité de l'alimentation secteur doit être celled'un environnement commercial ou médical standard.

La qualité de l'alimentation secteur doit être celled'un environnement commercial ou médical standard. Si l'utilisateur du système eQuinox nécessite
un fonctionnement continu de l'appareil en cas decoupure de courant, il est recommandé d'utiliser lesystème eQuinox avec une source d'alimentation
ininterruptible.

Les champs magnétiques à la fréquence du réseau
doivent correspondre à ceux d'un environnement
commercial ou médical standard.

REMARQUE : UT correspond à la tension secteur AC avant l'application du niveau de test.

48 — 4805-C1 C (French)Directives et déclaration du fabricant - Immunité électromagnétique
LesystèmeeQuinoxestprévupouruneutilisationdanslesenvironnementsélectromagnétiquesspécifiésci-dessous.Leclientoul'utilisateurdu
système eQuinox doit s'assurer que celui-ci est utilisé dans un environnement de ce type.
Test d'immunité

Niveau de test CEI 60601

Niveau de conformité

Environnement électromagnétique - Directives

RF conduites

3 Vrms3 Vrms

CEI 61000-4-6

150 kHz à 80 MHz

RF rayonnées

3 V/mLes équipements de communication RF portables
et mobiles ne doivent pas être utilisés à proximitédes composants de l'appareil, y compris les câbles,à une distance inférieure à la distance de séparation
recommandée calculée sur la base de l'équation
applicable à la fréquence de l'émetteur.

CEI 61000-4-3

80 MHz à 2,5 GHz

3 V/mDistance de séparation recommandéed 8 1,2√ P
d 8 1,2√ P = 80 MHz à 800 MHz
d 8 1,2√ P = 800 MHz à 2,5 GHz
où P correspond à la puissance nominale de sortie
maximum de l'émetteur en watts (W), indiquée parle fabricant de l'émetteur, et d correspond à la distance de séparation recommandée en mètres (m).
L'intensité des champs émis par des émetteurs RF
fixes,déterminéeparl'étudeélectromagnétiqued'unsite a, doit être inférieure au niveau de conformité
pour chaque plage de fréquences b.
Des interférences peuvent se produire à proximitédes équipements portant le symbole suivant :REMARQUE 1 : à 80 MHz et 800 MHz, la plage de fréquences supérieure s'applique.
REMARQUE 2 : il est possible que ces directives ne s'appliquent pas à toutes les situations. La propagation électromagnétique dépend desphénomènes d'absorption et de réflexion induits par les structures, les objets et les personnes.
L'intensité des champs produits par des émetteurs fixes, tels que les stations de base pour les radiotéléphones (cellulaires/sans fil) et
les radios mobiles, les radio amateurs, les radios AM et FM et la télévision ne peut pas être prédite théoriquement avec précision. Pour évaluer
l'environnementélectromagnétiquedûauxémetteursRFfixes,uneétudeélectromagnétiquedusitedoitêtreenvisagée.Sil'intensitéduchamp,
mesurée à l'emplacement où le système eQuinox est utilisé, dépasse le niveau de conformité RF applicable indiqué ci-dessus, il conviendra des'assurer du bon fonctionnement de système eQuinox dans un tel environnement. En cas d'anomalie, il pourra être nécessaire de réaliser desmesures supplémentaires, par exemple en modifiant l'orientation ou la position du système eQuinox.

ab

Sur la plage de fréquences de 150 kHz à 80 MHz, l'intensité des champs doit être inférieure à 3 V/m.

eQuinox (Model 4000) User Manual — 49

Distances de séparation recommandée entre des équipements de communication RF portables et mobiles et le système eQuinox
Le système eQuinox est prévu pour une utilisation dans un environnement dans lequel les perturbations dues aux radiofréquences rayonnées
sontcontrôlées.Leclientoul'utilisateurdusystèmeeQuinoxpeutprévenirlesperturbationsélectromagnétiquesenrespectantlesdistancesminimumrecommandéesci-dessouspourlaséparationentreleséquipementsdecommunicationRFportablesetmobiles(émetteurs)etl'appareil,en fonction de la puissance maximum de sortie de l'équipement de communication.
Puissance nominalede sortie maximum del'émetteur

Distance de séparation en fonction de la fréquence de l'émetteurm
150 kHz à 80 MHz

80 MHz à 800 MHz

800 MHz à 2,5 GHz

d 8 1,2√ P

d 8 1,2√ P

d 8 1,2√ P

0,01

0,12

0,12

0,23

0,1

0,38

0,38

0,73

1

1,2

1,2

2,3

10

3,8

3,8

7,3

100

12

12

23

W

Pour lesémetteursavecunepuissancenominaledesortiemaximumnefigurantpasdansletableauci-dessus,ladistancedeséparationrecommandée d en mètres (m) peut être déterminée à l'aide de l'équation applicable à la fréquence de l'émetteur, où P correspond à la puissancenominale de sortie maximum de l'émetteur en watts (W), indiquée par le fabricant de l'émetteur.
REMARQUE 1 : à 80 MHz et 800 MHz, la distance de séparation correspondant à la plage de fréquences supérieure s'applique.
REMARQUE 2 : il est possible que ces directives ne s'appliquent pas à toutes les situations. La propagation électromagnétique dépend desphénomènes d'absorption et de réflexion induits par les structures, les objets et les personnes.

50 — 4805-C1 C (French)

Informations générales concernant les Directives sur l'utilisation de batteries
voyages en avion
durant les voyages en avion
Pendant la circulation au sol, le décollage et l'atterrissage, lesystème eQuinox doit être arrêté et rangé sous le siège ou dans un
autre compartiment de stockage approuvé afin de ne pas bloquerl'allée centrale ou l'entrée s'il n'est pas utilisé.
Pendant la circulation au sol, le décollage et l'atterrissage,
l'utilisateur doit occuper un siège approprié afin de ne pas empêcher
les autres passagers d'accéder à ou d'utiliser toute issue de secours,
toute issue normale ou l'allée du compartiment passagers de l'avion
si le système eQuinox est utilisé.
Les utilisateurs ne sont pas autorisés à s'installer dans une rangée
située près d'une issue de secours s'ils utilisent le système eQuinox.
Si le système eQuinox est utilisé pendant une décompression dela cabine et le déploiement du système d'oxygène de la cabine,
l'utilisateur doit cesser d'utiliser le système eQuinox et utiliser lesystème d'oxygène d'appoint de l'avion. L'unité eQuinox doit êtrearrêtée une fois que l'utilisateur a attaché le système d'oxygèned'appoint de l'avion.
Avant tout déplacement, l'utilisateur doit inspecter le systèmeeQuinox afin de vérifier son bon fonctionnement.

Prévoyez le temps passé au sol (enregistrement, contrôles de sécurité, etc.) et lesescales.
La plupart des compagnies aériennes exigent que l'utilisateur
dispose du nombre adéquat de batteries pleinement chargées pouralimenter l'appareil durant au moins 150 % de la durée maximum
estimée du voyage (temps de vol, temps au sol avant et après le volou durant les correspondances, et retards imprévus).

Il est recommandé de toujours disposer d'une batterie supplémentaire lors de vos déplacements.
Chaque compagnie aérienne applique ses propres exigences
concernant les usagers voyageant avec de l'oxygène.
Contactez votre compagnie aérienne au moins 48 heures avant
le départ afin de l'informer que vous voyagerez avec l'appareil
eQuinox.

MISE EN GARDE : les réglementations du département
des transports des États-Unis et des Nations unies (ONU)
exigent que la batterie soit retirée de l'appareil sur tous
les vols internationaux si celui-ci est enregistré en tant que bagage.
Lors de l'expédition du concentrateur d'oxygène, il est également
impératif de retirer la batterie de l'appareil et de l'emballer correctement.

eQuinox (Model 4000) User Manual — 51

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107, États-Unis
1-800-482-2473
Site Web mondial : www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Allemagne
Site Web mondial : www.CAIREmedical.com

CAIRE Inc. se réserve le droit d'interrompre la commercialisation de ses produits ou de modifier les tarifs, les matériaux,les équipements, les niveaux de qualité, les descriptions, les caractéristiques et/ou les processus de ses produits, et ce à
toutmoment,sansavertissementpréalableetsansaucuneautreobligationniconséquence.Nousnousréservonstous
les droits non expressément stipulés dans le présent document, selon les conditions applicables.

52 — 4805-C1 C (French)À la date de cette publication, eQuinox, SeQual et autoSAT, ainsi que leurs logos associés (le cas échéant), sont des marques déposées de CAIRE Inc.
Lysol® est une marque déposée de Reckitt Benckiser, R-U.
OxySafe® est une marque déposée d'Applied Home Healthcare Equipment.
Pine-Sol® et Lestoil® sont des marques déposées de The Clorox Company.

Bedienungsanleitung

eQuinoxTM

Modell 4000
eQuinox (Model 4000) User Manual — 53

Steuerungselemente und Systemstatusanzeigen
SymbolA

02

Definition

Definition

Lesen Sie vor der Verwendung des Gerätes dieseBedienungsanleitung. Siehe die Anweisungen in derBedienungsanleitung.

Symbol „Zugelassen von FAA“: Die
Bundesluftfahrtbehörde der USA (Federal Aviation
Administration, FAA) hat dieses Gerät für die
Verwendung bei kommerziellen Flügen zugelassen.

Symbol „Rauchen verboten“: Nicht in der Nähe desGeräts rauchen.

Volumenstrom-Einstellungsanzeige

Anzeige für Warnungen/Alarme (gelb)

Pulse-Dose-Betrieb

Anzeige für Alarme (rot)­

Gerät arbeitet normal, Einschalttaster

Anzeige für externe Stromversorgung

Volumenstrom-Einstellung erhöhen

Batteriemodul-Ladeanzeige

Volumenstrom-Einstellung senken

Batteriemodulsymbol

Nicht mit Öl oder Fett schmieren.

AmpereNicht offenem Feuer aussetzen.

Wechselstrom

Keine vom Benutzer wartbaren Teile enthalten.
Abdeckung nicht öffnen.

IEC-Symbol für Gleichstrom (Eingang)

Heiße Oberfläche!

Sauerstoffleistung

Vor Nässe schützen!

Typ BF-geschütztes Gerät (Schutz vor
elektrischem Schlag)

Tropfwassergeschütztes Gerät (Schutzart IPX1): Das
Batteriemodul und das Wechselstromnetzteil deseQuinox sind gegen Schäden durch Eindringen von
Flüssigkeiten geschützt. (IPX1 gemäß IEC 60529)

Recycling-Symbol

IP22

SymbolDas Gerät muss gemäß der Richtlinie 2002/96/EGüber Elektro- und Elektronik-Altgeräte am Ende seiner
Lebensdauer zu einer geeigneten Recyclingstelle
gebracht werden und darf nicht mit dem normalen
Hausmüll entsorgt werden.
Tropfwassergeschützt und sicher vor Eindringen vonfesten Fremdkörpern: Mit der Schutzart IP22 bietet
der eQuinox gemäß IEC 60529 Schutz vor den
schädlichen Einwirkungen von Flüssigkeiten undfesten Fremdkörpern.
Sowohl für den US- als auch für den kanadischen
Markt zugelassen; entspricht den US- und kanadischen
Normen.

54 — 4805-C1 C (German)

IPX1

Name und Adresse des Herstellers
Autorisierter Vertreter in der Europäischen
Gemeinschaft

Gerät der Schutzklasse II

Das Gerät entspricht den Vorgaben der Richtlinie
93/42/EWG für Medizinprodukte und darf daher das
hier dargestellte CE-Zeichen tragen.

Inhalt
Steuerungselemente und Systemstatusanzeigen............................................................................................................54
Kurzanleitung.................................................................................................................................................................................56
Gebrauchshinweise......................................................................................................................................................................59
Gegenanzeigen..............................................................................................................................................................................59
Sicherheitsrichtlinien...................................................................................................................................................................59
Empfohlene Betriebsumgebungen........................................................................................................................................61
Gebrauchsanweisung..................................................................................................................................................................62
Sprachmodulbetrieb....................................................................................................................................................................65
Inbetriebnahme des eQuinox...................................................................................................................................................66
Störungsbehebungstabelle für den Benutzer ...................................................................................................................68
Reinigung, Pflege und regelmäßige Wartung....................................................................................................................71
EMV-Tabellen..................................................................................................................................................................................73
Allgemeine Informationen zu Flugreisen.............................................................................................................................77
Batteriemodul und Flugreisen..................................................................................................................................................77

eQuinox (Model 4000) User Manual — 55

Kurzanleitung

1

Auspacken des eQuinox2

Kennenlernen Ihres eQuinox

Lesen Sie sorgfältig alle Warnhinweise, Vorsichtshinweise undzusätzlichen Informationen zum Gerät im Rest dieses Handbuches.
Machen Sie sich mit den wichtigsten Komponenten des eQuinox
sowie dem Bedienfeld vertraut.

Sauerstoffauslass

VORDERSEITE
Griff

Bedienfeld
Anschluss fürexterne Stromversorgung

Wechselstrom-Netzteil

Symbol„Zugelassen
von FAA“

Batteriemodulriegel

Typenschild &Seriennummer

eQuinox-Trolley

BatteriemodulRÜCKSEITE

12-VDC-Kabel

SchaltschrankAbluftöffnungen

Batteriemodul(Batterie)

56 — 4805-C1 C (German)

Lufteinlassfilter

Das Bedienfeld des eQuinox zeigt
wichtige Betriebsinformationen an.

Ein-/Aus-Taster (grün): Mit diesem Taster wird das Gerätein-bzw.ausgeschaltet.LeuchtetderTastergrün,dannistdas
Gerät eingeschaltet und funktioniert ordnungsgemäß.
TasterzumErhöhenbzw.SenkenderVolumenstrom-Einstellung:ÜberdieseTasterkönnenSiedenVolumenstromaufdenWert
einstellen, der Ihnen von Ihrem Arzt verschrieben wurde.
Betriebsmodus-Wahltaster und -Anzeige: Mit diesem
Taster können Sie zwischen dem Continuous-Flow-Modus(konstanter Volumenstrom) und dem Pulse-Dose-Modus(Sauerstoffbolus)umschalten.DerPulse-Dose-Modusaktiviert
dieautoSAT-Technologie.Dadurcherhöhtsichdiemaximale
BetriebsdauerimBatteriebetrieberheblich.WennderPulseDose-Modusaktiviertist,leuchtetdiegrünePulse-Dose-Modusanzeige.BeijedemAtemvorgangwirdeinSauerstoffbolus
verabreicht und die grüne Anzeige blinkt.
WARNANZEIGE (gelb) – Wenn diese Anzeige leuchtet, sindaufgrund einer Warnung niedriger Priorität erhöhte Wachsamkeit oder Vorsicht erforderlich. Sie können das Systemweiterverwenden.ZiehenSiedieStörungsbehebungstabelle
fürdiegeeigneteMaßnahmeheran,odersetzenSiesichmit
dem Pflegedienst in Verbindung. Blinkt die Anzeige gelb,
zeigtdieseineWarnungmittlererPrioritätan.Essindschnelle
Maßnahmenerforderlich.(TypischeUrsachensindz.B.wenig
Sauerstoff oder eine niedrige Batteriekapazität.)
ALARMANZEIGE(rot)–WeistaufeinenAlarmhoherPriorität
hin. Es sind unverzüglich Maßnahmen zu ergreifen. KonsultierenSiedieStörungsbehebungstabelleundsetzenSiesich
mitIhremPflegedienstinVerbindung.(TypischeUrsachensind
eineUnterbrechungderStromversorgung,wenigSauerstoffoder eine Störung.)
Informationsanzeige.
• C oder P – „C“ bedeutet Continuous-Flow-Modus (konstanter Volumenstrom), „P“ bedeutet Pulse-Dose-Modus(Sauerstoffbolus).
• Batteriebetriebszeit – Im Batteriebetrieb wechselt die Anzeigealle20Sekundenundgibtdieverbleibendegeschätzte
Batteriebetriebszeit an.
• Einstellung (Volumenstrom oder Sauerstoffbolus) – Bei
AbschaltendesGerätswerdenderletzteBetriebsmodussowie
dieletzteEinstellunggespeichert.BeierneutemEinschalten
werden diese wieder aufgerufen.
Batteriemodul-Ladeanzeige: Diese Anzeige zeigt dieverbleibende Batteriekapazität an. Jeder einzelne der fünf
Querbalkenstehtfüretwa20%dergesamtenBatterieladung.
WährenddesLadevorgangsblinkendieLadeanzeigebalken
nacheinanderstufenweiseauf.WennkeinBatteriemoduleingesetztbzw.dasBatteriemodulfalscheingesetztist,leuchtet
die Batteriemodul-Ladeanzeige nicht auf bzw. blinkt.
AnzeigefürexterneStromversorgung:WenndereQuinoxordnungsgemäßüberdasNetzteil(Gleich-oderWechselstrom)
mitdemStromnetz(bzw.einerStromquelle)verbundenist,
erscheint dieses Symbol auf dem Bedienfeld.
AkustischesSignal:EinakustischesSignalweistSieaufdenBetriebszustanddes
Geräteshin,entwedereineWarnungodereineFehlfunktion,undbestätigteinen
gültigen Tastendruck.
Sprachmodul: Falls aktiviert, macht Sie ein gesprochener Hinweis auf den
BetriebszustanddesGeräts,eineÄnderungderStromversorgungundWarnungenoder Fehler aufmerksam.

3

Einschalten des eQuinox

Zum Anschluss des Wechselstrom-Netzteils: Schließen
Sie das Wechselstrom-Netzteilkabel sicher an der externen
Stromanschlussbuchse auf der rechten Seite des Gerätes an.
Verbinden Sie das Wechselstrom-Netzkabel mit dem Netzteil undeiner Wechselstrom-Steckdose.
Schritt 1

Schritt 2

Schritt 3

Wenn das eQuinox ordnungsgemäß angeschlossen ist, leuchtet einegrüne Anzeige auf dem Wechselstrom-Netzteil. Auf dem Bedienfeld
leuchtet zusätzlich die Anzeige für die externe Stromversorgung.
Einsetzen der Batterie: Schieben Sie die Batterie in das leere
Batteriefach, bis sie hörbar einrastet. Sie schließt bündig mit derVorderseite des Geräts ab.

Wenn die Batterie ordnungsgemäß eingesetzt wurde und daseQuinox eingeschaltet oder mit dem Stromnetz verbunden ist, wird
die Batteriemodul-Ladeanzeige

angezeigt.

Ein-/Aus-Taster:Halten Sie den Ein-/Aus-Taster für zwei (2) Sekunden gedrückt, um
das eQuinox einzuschalten. Beim Einschalten wird ein kurzer, hörund sichtbarer Selbsttest durchgeführt. Prüfen Sie, ob alle Lampen
leuchten und das akustische Signal drei (3) Sekunden lang ertönt.
Halten Sie den Ein-/Aus-Taster für zwei (2) Sekunden gedrückt, um
das eQuinox auszuschalten.
eQuinox (Model 4000) User Manual — 57

4

Wählen eines Modus

Moduswahltaster und -Anzeige
Drücken Sie den Moduswahltaster, um die gewünschte Betriebsart
auszuwählen. Durch wiederholtes Drücken kann zwischen dem
Pulse-Dose-Modus (Sauerstoffbolus) und dem Continuous-FlowModus (konstanter Volumenstrom) umgeschaltet werden. Im
Continuous-Flow-Modus strömt der Sauerstoff mit einem konstanten
Volumenstrom von 0,5 und 3,0 Litern pro Minute. Dies wird mit dem
Buchstaben C angezeigt. Im Pulse-Dose-Modus wird der Sauerstoff
zu Beginn jedes Einatmens als Bolus bereitgestellt. Die Pulse-ModusEinstellungen 1–9 entsprechen Bolusgrößen von 16 ml bis 192 ml
und werden auf dem Display durch den Buchstaben P angezeigt.
Der Sauerstoffbolus kann auf Empfindlichkeit und AtembolusVerabreichung eingestellt werden. Die Volumenstrom-Einstellungenin beiden Modi sollten Ihnen von Ihrem Arzt verschrieben werden.

5

E instellen des Volumenstroms auf die verschriebenen Werte

Taster zum Erhöhen bzw. Senken der Volumenstrom-Einstellung:

6

Verwenden des eQuinox beginnen

Lufteinlassfilter

Lufteinlassfilter
Ihr Gerät ist mit einem Lufteinlassfilter ausgestattet. Ein zusätzlicher
Filter wird mitgeliefert. Überprüfen Sie den Filter täglich undreinigen Sie ihn wöchentlich. Anweisungen zum Reinigen finden
Sie auf Seite 19.
Schließen Sie den Sauerstoffversorgungsschlauch an den
Sauerstoffauslass an; schließen Sie die Sauerstoffkanüle gemäß
Herstelleranweisungen oder den Anweisungen Ihres Pflegedienstes
an.

Atmen Sie normal über die Nasenkanüle.
Im Batteriebetrieb ermöglicht der Pulse-Dose-Modus eine längere
Betriebsdauer des Geräts.

Über die Taster + und - können Sie die Einstellung auf die vonIhrem Arzt verschriebene Einstellung ändern.

Anzeige Continuous-Flow-Modus (konstanter Volumenstrom)

Anzeige Pulse-Dose-Modus (Sauerstoffbolus):

WARNUNG: STELLEN SIE UNBEDINGT NUR DIE VERORDNETE
MENGE SAUERSTOFF EIN. ÄNDERN SIE DIE EINSTELLUNG NUR,WENN DIES AUF ANWEISUNG EINES APPROBIERTEN ARZTESERFOLGT.DERGEBRAUCHDESTRAGBARENSAUERSTOFFKONZENTRATORS WÄHREND DES SCHLAFS DARF NUR AUF EMPFEHLUNGDURCH QUALIFIZIERTES KLINISCHES PERSONAL ERFOLGEN.
58 — 4805-C1 C (German)

7

Verwenden des eQuinox-Trolleys

Richten Sie den eQuinox am geformten Boden des Trolleys
aus, sodass er gerade und aufrecht steht. Richten Sie die große
Gewindeschraube auf die Bohrung an der Rückseite des eQuinox
aus. Ziehen Sie nun den Knauf am Trolley handfest an, bis derTrolley fest mit dem Gerät verbunden ist. Drücken Sie den
Druckknopf am Teleskopgriff des Trolleys herunter und ziehen Sie
den Griff nach oben, um die Griffhöhe einzustellen.

Wichtig!
Warn- und Sicherheitshinweise sind folgendermaßen
gekennzeichnet:
WARNUNG: WEIST AUF GEFAHREN HIN, DIE SCHWERE VERLETZUNGEN ZUR FOLGE HABEN KÖNNEN.

VORSICHT: Weist auf Gefahren hin, die Schäden am eQuinox
verursachen können.
Hinweis: Besonders zu beachtende Informationen.

Gebrauchshinweise
Das eQuinox ist für die Verabreichung von Zusatzsauerstoff
bestimmt. Das Gerät ist nicht für lebenserhaltende Maßnahmen
ausgelegt und bietet keine Patientenüberwachung.
WARNUNG: STELLEN SIE UNBEDINGT NUR DIE VERORDNETE
MENGE SAUERSTOFF EIN. ÄNDERN SIE DIE EINSTELLUNG NUR,WENN DIES AUF ANWEISUNG EINES APPROBIERTEN ARZTESERFOLGT.DER GEBRAUCH DESTRAGBAREN SAUERSTOFFKONZENTRATORS WÄHREND DES SCHLAFS DARF NUR AUF EMPFEHLUNGDURCH QUALIFIZIERTES KLINISCHES PERSONAL ERFOLGEN.
WARNUNG: LAUT BUNDESGESETZ (USA) DARF DIESES GERÄT NUR AUF
ANORDNUNG EINES ARZTES ODER DURCH EINEN ARZT ODER EINE LIZENZIERTE PERSON AUS DEM KRANKENPFLEGEBEREICH VERKAUFT ODER
VERMIETET WERDEN.
WARNUNG: DIESES GERÄT IST NICHT FÜR LEBENSERHALTENDEMASSNAHMEN GEEIGNET. BEI GERIATRISCHEN, PÄDIATRISCHEN UND
ANDEREN PATIENTEN, DIE EIN UNWOHLSEIN BEIM GEBRAUCH DESGERÄTS NICHT MITTEILEN KÖNNEN, SIND GGF. ZUSÄTZLICHE ÜBERWACHUNGSMETHODEN NOTWENDIG. PATIENTEN MIT EINGESCHRÄNKTEM
HÖR- UND/ODER SEHVERMÖGEN BENÖTIGEN U. U. HILFE BEI DER
ALARMÜBERWACHUNG.

Gegenanzeigen
WARNUNG: DIE VERWENDUNG VON SAUERSTOFF OHNE
ÄRZTLICHE VERSCHREIBUNG IST UNTER UMSTÄNDEN GEFÄHRLICH. DAS GERÄT DARF NUR AUF VERSCHREIBUNG EINESARZTES VERWENDET WERDEN.
WARNUNG: DAS GERÄT DARF NICHT VERWENDET WERDEN, WENN
BRENNBARE ANÄSTHETIKA VORHANDEN SIND.
WARNUNG: WIE BEI JEDEM ELEKTRISCHEN GERÄT SIND AUCH BEIM
EQUINOX AUSFÄLLE AUFGRUND VON UNTERBRECHUNGEN DER STROMVERSORGUNG MÖGLICH BZW. IST ES EVENTUELL NOTWENDIG, DASGERÄT VON EINEM ENTSPRECHEND GESCHULTEN TECHNIKER WARTENZU LASSEN. DER EQUINOX IST FÜR PATIENTEN, DIE AUFGRUND EINER
SOLCHEN VORÜBERGEHENDEN UNTERBRECHUNG GESUNDHEITLICHESCHÄDEN DAVONTRAGEN WÜRDEN, NICHT GEEIGNET.

Sicherheitsrichtlinien
WARNUNG: AN DIESEM GERÄT DÜRFEN KEINE MODIFIKATIONEN VORGENOMMEN WERDEN.
WARNUNG: FÜR DEN FALL EINES STROMAUSFALLS, ALARMSODER MECHANISCHEN VERSAGENS EMPFIEHLT DER HERSTELLER EINE ALTERNATIVE SAUERSTOFFZUFUHR. KONSULTIEREN SIE IHREN ARZT ODER GERÄTELIEFERANTEN BEZÜGLICHDES TYPS DES BENÖTIGTEN RESERVESYSTEMS.
WARNUNG: DIESES GERÄT LIEFERT HOCHREINEN, HOCHENTZÜNDLICHENSAUERSTOFF. RAUCHEN ODER OFFENE FLAMMEN SIND IM RAUM, IN DEM
SICH (1) DAS GERÄT ODER (2) EIN SAUERSTOFF TRANSPORTIERENDESZUBEHÖRTEIL BEFINDEN, NICHT ERLAUBT. DIE MISSACHTUNG DIESERWARNUNG KANN SCHWERE BRÄNDE, SACH- BZW. PERSONENSCHÄDENODER DEN TOD ZUR FOLGE HABEN.
WARNUNG: BETREIBEN SIE DAS GERÄT NICHT IN EINEM EINGESCHRÄNKTEN ODER ENGEN BEREICH (Z. B. EINEM KLEINEN KOFFER ODER EINERHANDTASCHE), IN DEM DIE BELÜFTUNG UNZUREICHEND SEIN KÖNNTE.
HIERDURCH KANN SICH DER SAUERSTOFFKONZENTRATOR ÜBERHITZENUND DIE LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: DER KONZENTRATOR DARF NICHT MIT RAUCH, SCHADSTOFFEN ODER DÄMPFEN IN KONTAKT KOMMEN.
WARNUNG:DEREINSATZVONSAUERSTOFFBEHANDLUNGSZUBEHÖR,DASFÜR DIE VERWENDUNG MIT DIESEM SAUERSTOFFKONZENTRATOR NICHTVORGESEHEN IST, KANN DIE LEISTUNG BEEINTRÄCHTIGEN. AUF EMPFOHLENES ZUBEHÖR WIRD IN DIESEM HANDBUCH VERWIESEN.
WARNUNG: IST DER SAUERSTOFFKONZENTRATOR HERUNTERGEFALLEN,WURDE ER BESCHÄDIGT ODER WASSER AUSGESETZT, SOLLTEN SIE SICH
MIT IHREM PFLEGEDIENST IN VERBINDUNG SETZEN, UM DAS GERÄT
PRÜFEN ODER EVENTUELL REPARIEREN ZU LASSEN. VERWENDEN SIE DENSAUERSTOFFKONZENTRATOR NICHT,WENN STROMKABEL ODER STECKER
BESCHÄDIGT SIND.
WARNUNG: SICHERN SIE DEN SAUERSTOFFKONZENTRATOR MIT EINEMGURT ODER ÄHNLICHEN VORRICHTUNGEN,WENN ER IN EINEM FAHRZEUG
TRANSPORTIERT WIRD, UM SCHÄDEN BZW. VERLETZUNGEN ZU VERMEIDEN.
WARNUNG: STELLEN SIE SICHER, DASS LUFTEINLASS UND -AUSLASSNICHT BLOCKIERT WERDEN. FÜHREN SIE KEINE GEGENSTÄNDE IN DIEGERÄTEÖFFNUNGEN EIN UND LASSEN SIE NICHTS IN DIESE HINEINFALLEN. HIERDURCH KANN SICH DER SAUERSTOFFKONZENTRATOR ÜBERHITZEN UND DIE LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: DER OPTIONAL ERHÄLTLICHE BEFEUCHTER DARF NICHT ÜBERFÜLLTWERDEN. DER OPTIONAL ERHÄLTLICHE BEFEUCHTER DARF NUR MIT
WASSER BEFÜLLT WERDEN. DAS FÜLLNIVEAU DARF DIE VOM HERSTELLER
DES BEFEUCHTERS ANGEZEIGTE HÖHE NICHT ÜBERSCHREITEN.
WARNUNG: LAUFEN SIE NICHT UMHER, WÄHREND DER EQUINOX INBETRIEB UND EIN BEFEUCHTER ANGESCHLOSSEN IST. NEHMEN SIE DIE
BEFEUCHTERFLASCHE VOR DEM UMHERLAUFEN AB. LEGEN SIE DEN EQUINOX NICHT FLACH HIN, WÄHREND DIESER AN EINE BEFEUCHTERFLASCHEANGESCHLOSSEN IST.
WARNUNG: DIE RICHTLINIEN DES VERKEHRSMINISTERIUMS DER USA(DOT) SOWIE DER VEREINTEN NATIONEN (UN) SEHEN VOR, DASS DIEBATTERIE AUS DEM GERÄT HERAUSGENOMMEN WERDEN MUSS, WENNDIESES ALS GEPÄCK AUF INTERNATIONALEN FLÜGEN AUFGEGEBEN WIRD.
BEI VERSAND DES SAUERSTOFFKONZENTRATORS MÜSSEN DIE BATTERIENEBENFALLS AUS DEM GERÄT ENTNOMMEN UND ORDNUNGSGEMÄSS
VERPACKT WERDEN.
WARNUNG: VERWENDEN SIE DAS GERÄT NUR NACH EINGEHENDEMSTUDIUM UND VERSTÄNDNIS DIESES HANDBUCHS. VERSTEHEN SIE DIE
WARNUNGEN UND ANWEISUNGEN NICHT, WENDEN SIE SICH AN DEN
BEREITSTELLER DIESES GERÄTES, BEVOR SIE VERSUCHEN, ES ZU VERWENDEN. ANDERNFALLS KANN ES ZU PERSONEN- ODER SACHSCHÄDEN
KOMMEN.
eQuinox (Model 4000) User Manual — 59

WARNUNG: ZIEHEN SIE BEI UNWOHLSEIN ODER EINEM MEDIZINISCHEN NOTFALL SOFORT ÄRZTLICHE HILFE ZU RATE.
WARNUNG:WIRD DER SAUERSTOFFKONZENTRATOR AUSSERHALB DES ANGEGEBENEN TEMPERATURBEREICHS VERWENDET, KANN DIESDIE FÄHIGKEIT DES GERÄTS BEEINTRÄCHTIGEN, DIE ANGEGEBENEN SAUERSTOFFKONZENTRATIONEN ZU ERZEUGEN. TEMPERATURGRENZWERTEFINDEN SIE IM ABSCHNITT „TECHNISCHE DATEN" IN DIESEM HANDBUCH.
WARNUNG: VERWENDEN SIE KEINE PRODUKTE AUF ÖL-, FETT- ODERPETROLEUMBASIS ODER ANDERE ENTZÜNDLICHE PRODUKTE MIT DEM
SAUERSTOFFKONZENTRATOR ODER MIT DESSEN SAUERSTOFFTRANSPORTZUBEHÖR. ES SOLLTEN AUSSCHLIESSLICH WASSERBASIERTE, SAUERSTOFFVERTRÄGLICHE LOTIONEN ODER SALBEN VERWENDET WERDEN.
SAUERSTOFF BESCHLEUNIGT DIE VERBRENNUNG ENTZÜNDLICHERSUBSTANZEN.
WARNUNG: EIN NICHT ORDNUNGSGEMÄSSER GEBRAUCH DER BATTERIEKANN ZU EINEM ERHITZEN UND ENTZÜNDEN DER BATTERIE FÜHREN, WAS
SCHWERE VERLETZUNGEN ZUR FOLGE HABEN KANN. DURCHBOHRENSIE DIE BATTERIE NICHT UND SCHLAGEN UND TRETEN SIE NICHT AUF DIE
BATTERIE. LASSEN SIE DIE BATTERIE NICHT FALLEN UND SETZEN SIE DIE
BATTERIE AUCH KEINEN SONSTIGEN HEFTIGEN SCHLÄGEN ODER STÖSSEN
AUS. DIE VERWENDUNG EINER BESCHÄDIGTEN BATTERIE KANN ZU VERLETZUNGEN FÜHREN.
WARNUNG: DER SAUERSTOFFKONZENTRATOR SOLLTE NICHT DIREKTNEBEN ANDEREN GERÄTEN ODER MIT ANDEREN GERÄTEN GESTAPELTVERWENDET WERDEN. IST EINE SOLCHE SITUATION NICHT ZU VERMEIDEN,MUSS DAS GERÄT AUF ORDNUNGSGEMÄSSEN BETRIEB HIN ÜBERWACHTWERDEN.

VORSICHT: Wenn ein Alarm auftritt oder Sie eine Fehlfunktion des Sauerstoffkonzentrators bemerken, schlagen Sie imAbschnitt „Störungsbehebung“ in diesem Handbuch nach.
Wenden Sie sich an den Bereitsteller des Geräts, wenn Sie das Problem
nicht lösen können.
VORSICHT: Befolgen Sie alle Anweisungen des Herstellers, damit die
Garantie ihre Gültigkeit behält.
WARNUNG: VERSUCHEN SIE NICHT, DAS GERÄT – ABGESEHEN VON DEN IN DIESEM HANDBUCH AUFGEFÜHRTENSTÖRUNGSBEHEBUNGEN – ZU WARTEN. ENTFERNEN SIE
KEINE ABDECKUNG. NUR DER BEREITSTELLER DES GERÄTES ODER EINQUALIFIZIERTER WARTUNGSTECHNIKER DÜRFEN DIE ABDECKUNGEN
ENTFERNEN ODER DAS GERÄT WARTEN.
WARNUNG: VERWENDEN SIE NUR VOM HERSTELLER EMPFOHLENESZUBEHÖR. DIE VERWENDUNG ANDEREN ZUBEHÖRS KANN ZU EINERSCHWEREN BESCHÄDIGUNG DES SAUERSTOFFKONZENTRATORS SOWIEZUM ERLÖSCHEN DER GARANTIE FÜHREN.
WARNUNG: FÜR DEN FALL EINES STROMAUSFALLS, ALARMSODER MECHANISCHEN VERSAGENS EMPFIEHLT DER HERSTELLER EINE ALTERNATIVE SAUERSTOFFZUFUHR.KONSULTIEREN SIE IHREN ARZT ODER GERÄTELIEFERANTEN BEZÜGLICH DES TYPSDES BENÖTIGTEN RESERVESYSTEMS.
WARNUNG: STELLEN SIE SICHER, DASS LUFTEINLASS UND -AUSLASSNICHT BLOCKIERT WERDEN. FÜHREN SIE KEINE GEGENSTÄNDE IN DIEGERÄTEÖFFNUNGEN EIN UND LASSEN SIE NICHTS IN DIESE HINEINFALLEN. HIERDURCH KANN SICH DER SAUERSTOFFKONZENTRATORÜBERHITZEN UND DIE LEISTUNG BEEINTRÄCHTIGT WERDEN.

60 — 4805-C1 C (German)

WARNUNG: RAUCHEN WÄHREND DER SAUERSTOFFVERABREICHUNG IST DER HAUPTGRUND FÜR VERLETZUNGEN UNDTODESFÄLLE BEI BRÄNDEN. DIESE SICHERHEITSWARNUNGENMÜSSEN UNBEDINGT BEFOLGT WERDEN:
WARNUNG: RAUCHEN, KERZEN ODER OFFENE FLAMMEN SIND IN
DEM RAUM, IN DEM SICH DAS GERÄT ODER EIN SAUERSTOFF BEFÖRDERNDES ZUBEHÖRTEIL BEFINDEN, VERBOTEN.
WARNUNG: RAUCHEN WÄHREND DES TRAGENS EINER SAUERSTOFFKANÜLE KANN ZU GESICHTSVERBRENNUNGEN UND ZUM TODEFÜHREN.
WARNUNG: WENN SIE RAUCHEN, KÖNNEN DIESE 3 SCHRITTE IHRLEBEN RETTEN: SCHALTEN SIE DEN SAUERSTOFFKONZENTRATOR AUS,
NEHMEN SIE DIE KANÜLE AB UND VERLASSEN SIE DEN RAUM, IN DEM
SICH DAS GERÄT BEFINDET.
WARNUNG: SCHILDER MIT DER AUFSCHRIFT „RAUCHEN VERBOTEN –SAUERSTOFF IN GEBRAUCH“ MÜSSEN IM HAUS ODER AN DEM ORT,
AN DEM DER SAUERSTOFF VERWENDET WIRD, DEUTLICH ERKENNBAR
ANGEBRACHT SEIN. PATIENTEN UND PFLEGEPERSONAL MÜSSEN ÜBERDIE GEFAHREN DES RAUCHENS IN DER NÄHE ODER WÄHREND DESGEBRAUCHSVON MEDIZINISCHEM SAUERSTOFF AUFGEKLÄRTWERDEN.
WARNUNG: VERWENDEN SIE DEN SAUERSTOFFKONZENTRATOR NICHTBEI VORHANDENSEIN ENTZÜNDLICHER GASE. DIES KANN ZU SCHNELLEN BRÄNDEN FÜHREN, DIE SACHSCHÄDEN,VERLETZUNGEN ODER DEN
TOD ZUR FOLGE HABEN KÖNNEN.
WARNUNG: EIN ABNEHMEN DER KANÜLE UND EINE LAGERUNG AUF
KLEIDUNG, BETTWÄSCHE, SOFAS ODER ANDEREM POLSTERMATERIALFÜHRT BEI VORHANDENSEIN EINER ZIGARETTE, WÄRMEQUELLE ODERFLAMME ZU EINER VERPUFFUNG.
WARNUNG: LASSEN SIE EINE KANÜLE NICHT AUF ODER UNTER KLEIDUNG, BETTWÄSCHE ODER STUHLPOLSTERN LIEGEN.WENN DAS GERÄT
EINGESCHALTET IST, ABER NICHT BETRIEBEN WIRD, WIRD DAS MATERIAL DURCH DEN SAUERSTOFF ENTFLAMMBAR. SCHALTEN SIE DENSAUERSTOFFKONZENTRATOR AUS, WENN ER NICHT IN GEBRAUCH IST.
VORSICHT: Trennen Sie den Wechselstrom-Stecker immer zuerst von der Wandsteckdose, bevor Sie das Wechselstrom-Kabel vom Sauerstoffkonzentrator trennen.
VORSICHT: Platzieren Sie Stromkabel und Sauerstoffschläuche so, dasseine Stolpergefahr vermieden wird.
VORSICHT: Wenn Sie den Sauerstoffkonzentrator in einem Auto, Bootoder mit anderen Gleichstromquellen verwenden, stellen Sie sicher,dass der Motor des Fahrzeugs läuft, bevor Sie den Sauerstoffkonzentrator anschließen. Wenn die Anzeige für die externe Stromversorgung
auf dem eQuinox nicht leuchtet und das Gerät neu gestartet werdenmuss, trennen Sie das 12-VDC-Kabel von der Gleichstromquelle, lassen
Sie das Fahrzeug neu an und verbinden Sie das 12-VDC-Kabel wieder
mit der Gleichstromquelle. Die Nichtbeachtung dieser Anweisungenkann dazu führen, dass der Sauerstoffkonzentrator keinen Stromerhält.
VORSICHT: Wenn das Automobil, in dem Sie den Sauerstoffkonzentrator verwenden, nicht läuft, trennen Sie das Gerät vom Automobil undentfernen Sie es. Lagern Sie den Sauerstoffkonzentrator nicht in einem
sehr heißen oder sehr kalten Automobil oder in ähnlichen Umgebungen, die extreme Temperaturen aufweisen.

Hinweis:TragbareundmobileHF-Kommunikationsausrüstungkannmedizinische elektrische Geräte beeinträchtigen.

Empfohlene Betriebsumgebungen
Beachten Sie für die ordnungsgemäße Verwendung des Geräts die
folgende Tabelle. Die Tabelle enthält wichtige Informationen zu den
empfohlenen Betriebsumgebungen oder Betriebsbedingungen.

Technische Daten
Betriebstemperatur
Relative Luftfeuchtigkeit

0 °C bis 40 °C
10 % bis 95 % bei 28 °C Taupunkttemperatur

Transport-/Lagerung- -25 °C bis 70 °C
stemperatur
RelativeLuftfeuchtigkeit:biszu95%,nichtkondensierend

Korrekte Standortwahl für den eQuinox
Wählen Sie den Standort für das Gerät so, dass weder Rauch,
Dämpfe noch Schadstoffe aufgenommen werden können. Bei
korrekter Aufstellung des Geräts muss ausreichend Luft durch den
Lufteinlassfilter oben auf der Rückseite des Geräts einströmen
können. Außerdem muss die Abluft ungehindert aus den
Abluftöffnungen unten rechts am Gerät austreten können.
Platzieren Sie das Gerät so, dass die akustischen Alarme hörbar
sind.
Legen Sie den Sauerstoffversorgungsschlauch so, dass dieser nicht
knicken oder blockiert werden kann.

-20 °C bis 50 °C für Batteriemodule
Stromversorgung

VerwendenSiekeineVerlängerungskabel.Verwenden Sie keine Steckdosen, die über einen Schalterein- und ausgeschaltet werden können.

Höhe über N.N.

0 bis 4.000 Meter

Standort

DieLufteinlassöffnungen(Rückseite,obereAbdeckung) oder die Abluftöffnungen (Rückseite, unten
links) dürfen nicht blockiert werden. Stellen Sie das
Gerät in einer Entfernung von mindestens 7,5 cm zu
Wänden, Vorhängen, Möbeln usw. auf.

Umgebung

Frei von Rauch, Schadstoffen und Dämpfen

Betriebsdauer

Rund um die Uhr bei Anschluss an eine externe
Stromversorgung (Gleich- oder Wechselstrom).

Hinweis: Wurde der Sauerstoffkonzentrator für einen längeren Zeitraum
außerhalbseinerüblichenBetriebstemperaturengelagert,solltedemGerät
vor dem Anschalten Zeit gelassen werden, die übliche Betriebstemperatur
anzunehmen.(SiehedenAbschnitt„TechnischeDaten“indiesemHandbuch).

WARNUNG: VERWENDEN SIE DEN SAUERSTOFFKONZENTRATOR NICHT BEI VORHANDENSEIN ENTZÜNDLICHER GASE. DIES
KANN ZU SCHNELLEN BRÄNDEN FÜHREN, DIE SACHSCHÄDEN,VERLETZUNGEN ODER DEN TOD ZUR FOLGE HABEN KÖNNEN. VERWENDEN SIE KEINE PRODUKTE AUF ÖL-, FETT- ODER PETROLEUMBASISODER ANDERE ENTZÜNDLICHE PRODUKTE MIT DEM SAUERSTOFFKONZENTRATOR ODER MIT DESSEN SAUERSTOFFTRANSPORTZUBEHÖR. ES
SOLLTENAUSSCHLIESSLICHWASSERBASIERTE,SAUERSTOFFVERTRÄGLICHE LOTIONEN ODER SALBEN VERWENDET WERDEN. SAUERSTOFFBESCHLEUNIGT DIE VERBRENNUNG ENTZÜNDLICHER SUBSTANZEN.
WARNUNG: DIESES GERÄT LIEFERT HOCHREINEN, HOCHENTZÜNDLICHEN SAUERSTOFF. RAUCHEN ODER OFFENE FLAMMEN SIND IM RAUM,
IN DEM SICH (1) DAS GERÄT ODER (2) EIN SAUERSTOFF TRANSPORTIERENDES ZUBEHÖRTEIL BEFINDEN, NICHT ERLAUBT. DIE MISSACHTUNGDIESER WARNUNG KANN SCHWERE BRÄNDE, SACH- BZW. PERSONENSCHÄDEN ODER DEN TOD ZUR FOLGE HABEN.

Hinweis: Wenn das Automobil, in dem Sie den Sauerstoffkonzentrator
verwenden,nichtläuft,trennenSiedasGerätvomAutomobilundentfernen
Siees.LagernSiedenSauerstoffkonzentratornichtineinemsehrheißenodersehrkaltenAutomobiloderinähnlichenUmgebungen,dieextremeTemperaturenaufweisen.LassenSiedenSauerstoffkonzentratoroderdasStromkabel
NICHT eingesteckt, wenn die Zündung ausgeschaltet ist. Es kann dabei zu
einer Entladung der Fahrzeugbatterie kommen.
VORSICHT: Betreiben Sie das Gerät nicht in einem eingeschränkten oder engen Bereich (z. B. einem kleinen Koffer odereiner Handtasche), in dem die Belüftung unzureichend sein
könnte. Hierdurch kann sich der Sauerstoffkonzentrator überhitzen und
die Leistung beeinträchtigt werden. Stellen Sie sicher, dass Lufteinlass
und -auslass nicht blockiert werden. Hierdurch kann sich der Sauerstoffkonzentrator überhitzen und die Leistung beeinträchtigt werden.

eQuinox (Model 4000) User Manual — 61

Gebrauchsanweisung
Vor der InbetriebnahmeAllgemeine Ladeinformationen

Diese Bedienungsanleitung soll Sie bei der Verwendung und
Wartung des Geräts unterstützen. Sollten Sie Fragen oder Probleme
haben, setzen Sie sich bitte mit Ihrem Pflegedienst in Verbindung.

Der eQuinox kann mit externer Wechselstrom- oder
Gleichstromversorgung oder einer wiederaufladbaren Batterie
betrieben werden. Zum System gehört ein intelligentes
Batterieladegerät. Damit wird die Batterie aufgeladen, wenn dereQuinox mit einer externen Stromquelle verbunden wird. Das
System überwacht und steuert sowohl die externe Stromversorgung
als auch das Batterieladegerät.

Wichtig! Sie dürfen den eQuinox erst in Betrieb nehmen, wenn Sie
das Kapitel „Sicherheitsrichtlinien“ in dieser Anleitung gelesen
haben. Befolgen Sie alle Gebrauchshinweise. Beachten Sie dabei
insbesondere auch alle Warnhinweise, die auf dem Gerät und in derBedienungsanleitung zu finden sind. Zur Verringerung von Brandund Verletzungsgefahren sowie von schweren Schäden am eQuinox
müssen zudem alle Sicherheitsvorkehrungen beachtet werden.
WARNUNG: REINIGEN SIE GEHÄUSE, BEDIENFELD UND
STROMKABEL NUR MIT EINEM SANFTEN HAUSHALTSREINIGER, DER MIT EINEM FEUCHTEN (NICHT NASSEN) TUCH
ODER SCHWAMM AUFGETRAGEN WIRD. WISCHEN SIE ANSCHLIESSEND
ALLE OBERFLÄCHEN TROCKEN. LASSEN SIE KEINE FLÜSSIGKEITEN IN
DAS INNERE DES GERÄTES GELANGEN. ACHTEN SIE BESONDERS AUF
DEN SAUERSTOFFAUSGANG FÜR DIE VERBINDUNG ZUR KANÜLE UND
STELLEN SIE SICHER, DASS DIESER FREI VON STAUB, WASSER UND
PARTIKELN IST.
VORSICHT: Stellen Sie sicher, dass Lufteinlass und -auslass
nicht blockiert werden. Hierdurch kann sich der Sauerstoffkonzentrator überhitzen und die Leistung beeinträchtigt
werden. Betreiben Sie den Sauerstoffkonzentrator nicht ohne eingesetzten Lufteinlassfilter. Wenn ein zweiter Filter vorhanden ist, setzen
Sie den Ersatzfilter ein, bevor Sie den verschmutzten Filter reinigen.
Reinigen Sie den verschmutzten Filter in warmem Seifenwasser undtrocknen Sie ihn vor der Verwendung gründlich ab.

Die Batterie des eQuinox wird geladen, wenn das Gerät an eine
Wechselstrom- oder Gleichstromquelle angeschlossen ist und die
Temperatur der Batterie ein sicheres Laden der Batterie zulässt.
Wird das Gerät von der externen Stromversorgung getrennt,
schaltet das Gerät selbsttätig auf Batteriebetrieb um, sofern die
Batterie eingesetzt und geladen ist. Wird das Gerät erneut mit derWechselstrom- oder Gleichstromquelle verbunden, wird es mit
der externen Stromversorgung betrieben und die Batterie wird
aufgeladen.
Richten Sie den Stromstecker am versenkten Anschluss aus, um
das Wechselstrom- bzw. das 12-VDC-Kabel richtig anzuschließen.
Der Anschluss für die externe Stromversorgung liegt vertieft auf
der rechten Seite des Geräts, wenn das Bedienfeld zu Ihnen zeigt.
Stecken Sie den Stromstecker in den Anschluss.
WECHSELSTROM-NETZTEIL

Wechselstrom-Netzteil

Einschalten des eQuinox
Das Gerät kann mit drei (3) verschiedenen Stromquellen betrieben
werden:• Wechselstrom-Netzteil• 12-VDC-Kabel• Wiederaufladbares Batteriemodul

Der Lieferumfang des eQuinox umfasst ein UniversalWechselstrom-Netzteil für die Verwendung zuhause bzw. anallen Orten, an denen Wechselstrom zur Verfügung steht. Beim
Anschließen an eine Wechselstromquelle muss das Kabel desWechselstrom-Netzteils sicher mit dem versenkten externen
Anschluss verbunden werden. Außerdem muss das Kabel desWechselstrom-Netzteils mit einer Wechselstrom-Steckdose
verbunden werden. Wenn das eQuinox ordnungsgemäß
angeschlossen ist, leuchten eine grüne Anzeige auf dem
Wechselstrom-Netzteil und die Anzeige für die externe
Stromversorgung. Auf dem Bedienfeld leuchtet zusätzlich die
Anzeige. Das CAIRE-Wechselstrom-Netzteil PN 4836 ist
die Hauptstromquelle für den tragbaren Sauerstoffkonzentrator
eQuinox. Es ist ein sehr wichtiges Element des Gesamtsystems.
Hinweis:VerwendenSienurdieelektrischeSpannung,dieaufdemGeräteetikett mit den technischen Daten angegeben ist.
Schritt 1

62 — 4805-C1 C (German)

Schritt 2

Schritt 3

WARNUNG: BERÜHREN SIE DIE BATTERIEKONTAKTE UND
DEN PATIENTEN NICHT GLEICHZEITIG, WENN SIE DAS GERÄT
MIT DEM EXTERNEN NETZTEIL (WECHSELSTROM-NETZTEIL)BETREIBEN UND KEIN BATTERIEMODUL INSTALLIERT IST.
VORSICHT: Verwenden Sie nur vom Hersteller empfohlenes
Zubehör. Die Verwendung anderen Zubehörs kann zu einer
schweren Beschädigung des Sauerstoffkonzentrators sowie
zum Erlöschen der Garantie führen.
VORSICHT: Verwenden Sie keine Verlängerungskabel für dieses Gerät
und verbinden Sie nicht zu viele Stecker mit derselben Steckdose. Die
Verwendung eines Verlängerungskabels kann die Leistung des Gerätes
beeinträchtigen. Die Verbindung zu vieler Stecker mit einer Steckdose
kann den Schaltschrank überlasten und zum Auslösen von Trennschalter/Sicherung oder bei Fehlfunktion von Trennschalter/Sicherung zum
Ausbruch eines Feuers führen.
VORSICHT: Die Verwendung von anderen Kabeln und Adaptern als den
angegebenen, mit Ausnahme der Kabel und Adapter, die vom Hersteller
des medizinischen elektrischen Geräts als Ersatzteile für interne Komponenten verkauft werden, kann zu erhöhten Emissionen oder einer
reduzierten Störfestigkeit des Sauerstoffkonzentrators führen.

12-VDC-Kabel
12-VDC-Kabel

Mithilfe eines 12-VDC-Kabels kann das System mit
Gleichstromquellen betrieben werden, wie sie z. B. in Fahrzeugen
zu finden sind.
1. Starten Sie das Fahrzeug.
2. Stecken Sie das 12-VDC-Kabel in den versenkten
Stromanschluss an der Seite des Geräts.
3. Stecken Sie den Stecker des 12-VDC-Kabels in die
Gleichstromsteckdose im Fahrzeug.
Wenn das Gerät ordnungsgemäß angeschlossen ist und über die
Gleichstromquelle versorgt wird, leuchtet auf dem Bedienfeld die
grüne Anzeige für die externe Stromversorgung.

eQuinox-Gleichstromanschluss
Ist der eQuinox an den Gleichstromanschluss eines Fahrzeugs
angeschlossen, gilt Folgendes:• Wird der eQuinox mit der Gleichstromversorgung eines Fahrzeugs
betrieben, arbeitet es mit allen Volumenstromeinstellungen von0,5 bis 3,0 Litern pro Minute und allen Impulseinstellungen von
1 bis 9.
• Die Batterie wird nur geladen, wenn bei Betrieb des Geräts
mit dem vorgeschriebenen Volumenstrom genügend Reststrom
verfügbar ist. Der eQuinox nutzt die Stromversorgung zuerst fürden Betrieb des Geräts. Darüber hinaus verfügbare Leistung wird
zum Erhaltungsladen der Batterie genutzt. Wird die eQuinoxBatterie aufgeladen, blinken die Balken des Batteriesymbols
nacheinander stufenweise auf. Wird die eQuinox-Batterie nicht
aufgeladen, bleibt das Batteriesymbol konstant.
WARNUNG: STELLEN SIE SICHER, DASS LUFTEINLASS UND-AUSLASS NICHT BLOCKIERT WERDEN. HIERDURCH KANNSICH DER SAUERSTOFFKONZENTRATOR ÜBERHITZEN UND DIE
LEISTUNG BEEINTRÄCHTIGT WERDEN.
WARNUNG: FÄLLT DIE SPANNUNG DER GLEICHSTROMQUELLE DES
FAHRZEUGS UNTER 11,5 VOLT, SCHALTET DER EQUINOX AUF BATTERIEBETRIEB UM (SOFERN EINE BATTERIE EINGESETZT UND GELADEN IST).

VORSICHT: Wenn das Automobil, in dem Sie den Sauerstoffkonzentrator verwenden, nicht läuft, trennen Sie das
Gerät vom Automobil und entfernen Sie es. Lagern Sie den
Sauerstoffkonzentrator nicht in einem sehr heißen oder sehr kalten
Automobil oder in ähnlichen Umgebungen, die extreme Temperaturen
aufweisen. Lassen Sie den Sauerstoffkonzentrator oder das Stromkabel
NICHT eingesteckt, wenn die Zündung aus ist. Es kann dabei zu einer
Entladung der Fahrzeugbatterie kommen.
Hinweis: Wird das Gerät über ein 12-VDC-Kabel betrieben, wird die Batterie
eventuell nicht geladen.
Hinweis:BeiBetriebdeseQuinoxüberein12-VDC-Kabelwirdempfohlen,den
Pulse-Dose-Moduszuverwenden,soferndieGabevonSauerstoffboliverschrieben wurde.
Hinweis:Umsicherzustellen,dassdasGerättatsächlichüberdieGleichstromquelleläuft,müssenSieprüfen,obdieAnzeigefürdieexterneStromversorgung
leuchtet. Ertönt ein akustisches Signal, kann dies ein Zeichen dafür sein, dassdie externe Stromversorgung unterbrochen wurde. Prüfen Sie, ob die grüne
LED-AnzeigefürdieexterneStromversorgungein-oderausgeschaltetist.Ein
VerlustderBatterieladungwährenddesGerätebetriebsübereineGleichstromquellekannvermiedenwerden,indemdieBatterieentnommenwird.Leuchtet
die Anzeige nicht, läuft das Gerät im Batteriebetrieb.
Hinweis:DietatsächlicheLeistungdeseQuinoxkannjenachdenBetriebsbedingungendesFahrzeugsvariieren.DieGleichstromquellemancherFahrzeuge
unterstützt eventuell nicht den vollen Funktionsumfang des Geräts. Falls das
FahrzeugfürdenBetriebmitderaktuellenEinstellungnichtgenügendStrom
zurVerfügungstellenkann,stopptdereQuinoxzuerstdasAufladenderBatterie
und setzt den Betrieb mit dem vorgeschriebenen Volumenstrom fort. Wenn
das Fahrzeug auch dann nicht genügend Leistung liefert, senkt der eQuinox
denVolumenstrom, wenn derVolumenstrom 2,0 Liter pro Minute oder mehr
beträgt.IstderVolumenstromkleinerals2,0LiterproMinute,wirddereQuinox
ausgeschaltet.
EntfernenSiebeiniedrigemVolumenstromalleanderenelektronischenGeräte,
dieandieStromversorgungdesFahrzeugsangeschlossensind(GPS,Mobiltelefon-Ladegerät usw.). Wird wieder ausreichend Leistung bereitgestellt, wird
derVolumenstromnichtautomatischaufdenursprünglichenWerterhöht.Der
BenutzermussdenVolumenstrommanuelleinstellen.WenndereQuinoxnoch
immernichtmitdemvorgeschriebenenVolumenstromarbeitet,versuchenSie,
das Batteriemodul zu entfernen.
eQuinox (Model 4000) User Manual — 63

Wiederaufladbares Batteriemodul
Der eQuinox kann mit dem mitgelieferten,
wiederaufladbaren Batteriemodul betrieben
werden.
Einsetzen der Batterie: Setzen Sie die Batterieso an das leere Batteriefach an, dass sie
hineingeschoben werden kann. Schieben Sie die
Batterie in das leere Batteriefach, bis sie hörbar
einrastet. Sie schließt bündig mit der Vorderseite
des Geräts ab. Wurde die Batterie ordnungsgemäß
eingesetzt und der eQuinox eingeschaltet,
wird auf dem Bedienfeld die BatteriemodulLadeanzeige angezeigt. (Wenn Sie KEINE
Batterie eingesetzt haben, wird die BatteriemodulLadeanzeige nicht angezeigt.) Das Gerät kann indiesem Fall über eine externe Stromversorgung
betrieben werden.
Wiederaufladbares
Batteriemodul

Um die Batterie zu entfernen, drücken Sie
den Entriegelungsknopf oben auf dem Batteriemodul. Ziehen Sie
die Batterie aus dem eQuinox, während Sie den Entriegelungsknopf
weiterhin nach UNTEN drücken.
Batterielaufzeit: Die Batterielaufzeit wird auf dem Bedienfeld
sowohl als Symbol als auch alphanumerisch angezeigt. Im
Batteriebetrieb wird die Batterieladeanzeige jederzeit angezeigt.
Die Anzeige wechselt, um die Stunden/Minuten der verbleibenden
Batterielaufzeit anzuzeigen. Die Batterielaufzeit wird vonverschiedenen Faktoren beeinflusst, z. B. von der VolumenstromEinstellung, dem gewählten Betriebsmodus (Continuous-Flowoder Pulse-Dose) sowie der Atemfrequenz. Die Tabelle „Typische
Laufzeiten neuer Batteriemodule“ enthält Schätzwerte für den
eQuinox bei Verwendung einer neuen, vollständig aufgeladenen
Batterie unter den angegebenen Volumenstrom-Einstellungen und
Betriebsbedingungen.
Hinweis: Der eQuinox wird werksseitig mit einer Batterie mit 12 Zellenausgeliefert. Eine Batterie mit 24 Zellen ist ein optionales Zubehör, das einen
längeren Batteriebetrieb ermöglicht.

Typische Laufzeiten neuer Batteriemodule
Betriebsart

Batterietyp
12 Zellen

24 Zellen

Konstanter
Volumenstrom
1,0 l/min

2,16 Stunden (2 Std. 9 Min.) 4,84 Stunden (4 Std. 50 Min.)2,0 l/min

1,2 Stunden (1 Std. 12 Min.) 2,75 Stunden (2 Std. 45 Min.)3,0 l/min

0,8 Stunden (48 Min.)

1,83 Stunden (1 Std. 49 Min.)

Pulse-Dose-Modus(bei 12 BPM, d. h.
Boli/min)
16 ml (1,0)

2,62 Stunden (2 Std. 37 Min.) 5,94 Stunden (5 Std. 56 Min.)

48 ml (3,0)

2,38 Stunden (2 Std. 22 Min.) 5,39 Stunden (5 Std. 23 Min.)

96 ml (6,0)

1,70 Stunden (1 Std. 41 Min.) 3,85 Stunden (3 Std. 51 Min.)

64 — 4805-C1 C (German)

Hinweis:WirdderSauerstoffkonzentratoraußerhalbdesüblichenTemperaturbereichs betrieben, kann dies die Leistung beeinträchtigen, die Batterielaufzeit verringern und/oder die Ladezeit der Batterie erhöhen. (Siehe
den Abschnitt „Technische Daten“ in diesem Handbuch). Die maximale
BetriebsdauerimBatteriebetriebkannverlängertwerden,wenndereQuinox
im Pulse-Dose-Modus (ärztlich zugelassener Modus) betrieben wird.
WARNUNG: FÜR MIT BATTERIEN AUSGESTATTETE SAUERSTOFFKONZENTRATOREN: LAGERN SIE DIE BATTERIE KÜHLUND TROCKEN. AUF DIESE WEISE KANN DIE LEBENSDAUER
DER BATTERIE VERLÄNGERT WERDEN. WIRD DER SAUERSTOFFKONZENTRATOR LÄNGERE ZEIT BEI HOHER TEMPERATUR BZW. MIT EINER
VOLL GELADENEN/GANZ LEEREN BATTERIE GELAGERT, KANN DIES DIEBATTERIEHALTBARKEIT BEEINTRÄCHTIGEN. VERSUCHEN SIE NICHT, DIE
BATTERIE ZU ÖFFNEN. SIE ENTHÄLT KEINE WARTBAREN TEILE. HALTENSIE KINDER VON BATTERIEN FERN.

Erstmaliges Laden der Batterie
Die mit dem eQuinox mitgelieferte neue Batterie ist im
Auslieferungszustand nicht voll geladen. Vor der ersten Verwendungdes eQuinox muss die Batterie daher zunächst vollständig geladen
werden.
Schließen Sie das Wechselstrom-Netzteil an und installieren Sie die
Batterie ordnungsgemäß. Warten Sie, bis die Batterie vollständig
geladen ist. Sobald die Batteriemodul-Ladeanzeige auf dem
Bedienfeld nicht mehr in Bewegung ist, ist die Batterie vollständig
geladen.
Der eQuinox kann verwendet werden, während die Batterie geladen
wird. Das Laden kann je nach Volumenstromeinstellung bis zu vierStunden dauern. Die Batterie wird auch geladen, wenn der eQuinox
an eine Wechselstromquelle angeschlossen und ausgeschaltet wird.
VORSICHT: Verwenden Sie nur die vom Hersteller bereitgestellten Batterien. Wenden Sie sich für Anweisungen zur
ordnungsgemäßen Batterieentsorgung an den Bereitsteller
Ihres Gerätes oder eine örtliche zuständige Stelle.

* Batterien und Akkus sind dem Recycling zuzuführen. *Bei Ausfall der Gleich- oder Wechselstromversorgung ertönt
ein akustisches Signal und der eQuinox schaltet automatisch auf
Batteriebetrieb um, sofern eine geladene Batterie installiert ist.
Wenn die externe Stromversorgung wiederhergestellt ist, wird die
Batterie automatisch wieder geladen. Wenn die Batterie bei Ausfall
der Wechselstromversorgung nicht eingesetzt oder vollständig
entladen ist, wird der eQuinox ausgeschaltet und es wird ein
Alarm ausgelöst. Die Batterie ist vollständig entladen, wenn die
Batteriemodul-Ladeanzeige auf dem Bedienfeld leer dargestellt
wird. Auf dem Gerät leuchtet eine gelbe Lampe und es ertönt ein
Alarmsignal für niedrigen Batterieladestand. Dieser Zustand dauertan, bis das Gerät vollständig ausgeschaltet wird. Dann leuchtet einerote Lampe und es ertönt ein dauerndes Alarmsignal. Der akustische
Alarm kann bis zu fünf Minuten dauern, bis das Gerät abschaltet
oder die externe Stromversorgung wiederhergestellt wird.
Hinweis: Halten Sie den Ein-/Aus-Taster des eQuinox für mindestens fünf
Sekunden gedrückt, um den Alarm „kein Strom“ zu quittieren. Der Alarm
wird ebenfalls beendet, wenn der eQuinox für zwei Minuten an eine externe
Wechselstromquelle angeschlossen wird.

Aktivieren/Deaktivieren des Sprachmoduls

Typische Batterieladezeiten
Je nach Volumenstromeinstellung dauert es in der Regel zwei
bis fünf Stunden, um eine vollständig entladene Batterie auf eineKapazität von 80 % aufzuladen. Sobald die Batterie vollständig
geladen ist, hört die Batterieladeanzeige auf sich zu bewegen.
Die Batterie kann im Gerät auch dann geladen werden, wenn dereQuinox ausgeschaltet und an das Stromnetz angeschlossen ist.

1. Um das Sprachmodul zu aktivieren/deaktivieren, schalten Sie
das Gerät mit der AN/Standby-Taste ein.
2. Drücken Sie nach dem Einschalten des Geräts das
Rauchverbotsymbol und die verborgene Taste einmal.
3. Innerhalb von ein bis zwei Sekunden wird die Menüoption„SPKR = ON“ oder „SPKR = OFF“ angezeigt.

Erhitzt sich die Batterie während der Entladung zu stark, kann das
Wiederaufladen erst dann beginnen, wenn die Batterie ausreichendabgekühlt ist. Die Abkühlung kann durch das Entfernen der Batteriebeschleunigt werden.

4. Drücken Sie die Taste zum Erhöhen oder Verringern desVolumenstroms, um die ausgewählte Option einzuschalten(Sprachmodul aktivieren) oder auszuschalten (Sprachmodul
deaktivieren).

Die Batterie kann sowohl mit Wechselstrom- als auch mit
Gleichstromquellen geladen werden. Die Batterie lässt sich ebenfalls
mit dem optionalen Ladegerät aufladen.

5. Die ausgewählte Option wird innerhalb von drei bis fünf
Sekunden aktiviert.
Hinweis:DamitwirddasSprachmodulnureinmaldeaktiviert.WenndasGerät
ausgeschaltet und neu gestartet wird, wird das Sprachmodul automatischwieder aktiviert.

Erhöhen/Verringern der Lautstärke für sprachliche Benachrichtigungen
1. Um die Lautstärke sprachlicher Benachrichtigungen zu
erhöhen bzw. zu verringern, drücken Sie das Rauchverbotsymbol
und die verborgene Taste zweimal.

Sprachmodulbetrieb

2. Das Menü zur Einstellung der Lautstärke „Volume = # -“ wird
angezeigt.

Anweisungen zum Sprachmodulbetrieb
Der eQuinox kann mündliche Benachrichtigungen zu Alarmen undzum Betriebsstatus ausgeben. Diese mündlichen Benachrichtigungenergänzen die Informationen der LED-Anzeigen. Im Folgenden wird
erläutert, wie das Sprachmodul ein- und ausgeschaltet wird, wie die
Lautstärke eingestellt und die Sprache geändert wird.

3. Drücken Sie die Taste zum Erhöhen oder Verringern desVolumenstroms, um die Lautstärke von 1 bis 10 zu erhöhen oderzu verringern.
4. Nach der Auswahl der gewünschten Lautstärke wird die
Option innerhalb von drei bis fünf Sekunden aktiviert.
Ändern der Sprache des Sprachmoduls

AN/Standby-Taste

AN/Standby-Anzeige(grüne LED)

Alarmanzeige(gelbe LED)

Alarmanzeige(rote LED)
Taste zum
Erhöhen desVolumenstroms

1. Wenn Sie die Sprache des Sprachmoduls ändern möchten,
drücken Sie das Rauchverbotsymbol und die verborgene Taste
fünf Mal, um auf die Sprachoptionen zuzugreifen.
2. Auf dem Display wird die aktuell aktivierte Sprache angezeigt.
Wenn das Sprachmodul die englische Sprache verwendet, wird
beim Zugriff auf das Menü mit den Sprachoptionen „English“ aufdem Display angezeigt.
3. Drücken Sie die Taste zum Erhöhen oder Verringern desVolumenstroms, um die verfügbaren Sprachen aufzurufen.
4. Sobald die gewünschte Sprache angezeigt wird, wird sie
innerhalb von drei bis fünf Sekunden automatisch eingestellt.

Anzeige fürexterne Stromversorgung (grüne
LED)

Rauchverbotsymbol
und verborgene
Taste

Flow-Modus-Taste

Modusanzeige(grüne LED)

Taste zum Verringern
des Volumenstroms
LCD-Display

eQuinox (Model 4000) User Manual — 65

Inbetriebnahme des eQuinox
Schritt 3: Schließen Sie den SauerstoffversorgungssStellen Sie den eQuinox in einer gut belüfteten und gut beleuchteten chlauch bzw. die Nasenkanüle an den SauerstofUmgebung auf. Stellen Sie sicher, dass die Lufteinlass- undfauslass an.
Schritt 1: Aufstellen des eQuinox für die Verwendung
Abluftöffnungen auf der Rückseite nicht blockiert sind.
Platzieren Sie den eQuinox so,
dass alle akustischen und visuellen
Signale leicht wahrgenommen
werden können.
Stellen Sie vor der Verwendungdes eQuinox sicher, dass derLufteinlassfilter an seinem Platz
sitzt. Sollte der Lufteinlassfilter
fehlen, setzen Sie sich mit Ihrem
Pflegedienst in Verbindung. Ist
der Lufteinlassfilter verschmutzt,
waschen Sie ihn mit warmem
Seifenwasser. Spülen Sie ihnanschließend mit klarem Wasser
und lassen Sie ihn trocknen, bevor
Sie den Filter wieder im Gerät
montieren. Tauschen Sie den
Lufteinlassfilter ggf. durch einen
neuen, sauberen Filter aus.

Lufteinlassfilter

Hinweis: Stellen Sie sicher, dass die Kanüle vollständig eingeführt ist und fest
sitzt.Damitwirdsichergestellt,dassderSauerstoffkonzentratoreinEinatmen
korrekterkennenundSauerstoffliefernkann.WährenddesEinatmenssollten
Sie den Sauerstofffluss zu den Stutzen der Nasenkanüle hören oder fühlen
können.

Schritt 4: Betriebsart auswählen

Abluftöffnungen

Schließen Sie das Gerät an eine Netzsteckdose oder eine
Gleichstromquelle an bzw. stellen Sie sicher, dass eine vollständig
geladene Batterie eingesetzt ist.

Schritt 2: Schalten Sie das Gerät ein undwarten Sie, bis es sich aufgeheizt hat.
Halten Sie den Ein-/Aus-Taster für zwei (2) Sekunden gedrückt,um den eQuinox einzuschalten. Beim Einschalten wird ein kurzer
sichtbarer Selbsttest durchgeführt.
Hinweis: Wenn beim erstmaligen Anschluss des Geräts an die StromversorgungoderbeimerstmaligenEinschaltendreiakustischeSignaleertönen,ist dies ein Hinweis auf einen niedrigen Batteriestand der 9-Volt-Batterie im
GehäusedeseQuinox.SetzenSiesichbittemitIhremPflegedienstinVerbindung,damitdie9-Volt-Batterievondiesemausgetauschtwerdenkann.Wenn
die drei Signaltöne nach dem Einschalten und dem Betrieb des Geräts fürmehrereMinutenweitermitUnterbrechungenausgegebenwerden,bedeutet
dies nicht, dass die Kapazität der 9-V-Batterie niedrig ist. Setzen Sie sich fürdie Wartung mit dem Pflegedienst in Verbindung.

Der eQuinox verfügt über eine integrierte SauerstoffkonzentrationsStatusanzeige (OCSI). Die OCSI überwacht die Sauerstoffabgabe
des Geräts kontinuierlich. Sie verfügt über eine grüne Lampe(normal), eine gelbe Lampe (Warnung), eine rote Lampe (Alarm)und einen akustischen Alarm. Beim Selbsttest des Geräts während
des Einschaltens leuchten diese Lampen gemeinsam auf. Warten Sie
nach dem ersten Einschalten bis zu fünf (5) Minuten, bis das Gerät
seine Betriebsbereitschaft erreicht hat. Wenn nur noch die grüne
Lampe leuchtet, ist das Gerät betriebsbereit.

66 — 4805-C1 C (German)

Tauschen Sie den Sauerstoffversorgungsschlauch bzw. die
Nasenkanüle regelmäßig entsprechend den Empfehlungen Ihres
Pflegediensts aus. Informationen zur Reinigung und zum Austauschder Kanüle bzw. des Versorgungsschlauchs erhalten Sie von Ihrem
Pflegedienst.

Drücken Sie den Moduswahltaster, um die Betriebsart auszuwählen.
Durch wiederholtes Drücken kann zwischen dem Pulse-DoseModus (Sauerstoffbolus) und dem Continuous-Flow-Modus(konstanter Volumenstrom) hin- und hergeschaltet werden. Welchen
Modus und welche Einstellung Sie wählen müssen, können Sie der
Verschreibung Ihres Arztes entnehmen.
Continuous-Flow-Modus (konstanter Volumenstrom):
Im Continuous-Flow-Modus wird eine konstante Menge Sauerstoff
durch den Schlauch bzw. die Nasenkanüle geleitet. Der Sauerstoff
wird in Litern pro Minute (LPM) gemessen. Einen Befeuchterkönnen Sie nur im Continuous-Flow-Modus verwenden. Sie dürfen
den eQuinox NICHT kippen, transportieren oder flach hinlegen,wenn eine Befeuchterflasche angeschlossen ist.
Pulse-Dose-Modus (Sauerstoffbolus):
Beim Pulse-Dose-Modus wird beim Einatmen ein Sauerstoffbolus
verabreicht. Dieser wird in Millilitern (ml) gemessen. Im PulseDose-Modus darf KEIN Befeuchter verwendet werden!
Wird im Pulse-Dose-Modus kein Atemzyklus erkannt, verabreichtder eQuinox alle 15 Sekunden einen konstanten Sauerstoffstrom.
Alle 15 Sekunden unterbricht das Gerät den Volumenstrom undprüft, ob ein Atemzyklus erkennbar ist. Sobald das Einatmen
wieder registriert wird, schaltet das Gerät zurück auf die Gabe dereingestellten Bolusmenge. Betätigen Sie den Moduswahltaster, um
den Pulse-Dose-Modus zu verlassen und den Continuous-FlowModus zu aktivieren.
Lassen Sie sich von Ihrem Arzt den richtigen Sauerstoffbolus
verschreiben, damit Sie die richtige Sauerstoffsättigung erhalten.
Berücksichtigen Sie Ihren Bedarf bei Ruhe, bei Bewegung und beim
Reisen in größeren Höhen. Wenden Sie sich für eine Verschreibung
und Boarding-Formulare an Ihren Pflegedienst.

Pulse-Dose-Modus und autoSat-Funktionen
Der eQuinox verfügt über eine Funktion mit dem Namen autoSAT®.
Sie liefert einen gleichmäßig großen Sauerstoffbolus für bis
zu 40 Atemzüge pro Minute (siehe unten stehende Tabelle fürmaximale Atemfrequenz). Bei steigender Atemfrequenz, z. B. beim
Gehen, regelt die autoSAT-Funktion automatisch die Bolusmenge,
die vom Gerät verabreicht wird. Die autoSAT-Technologie von
CAIRE ist urheberrechtlich geschützt und kommt nur im PulseDose-Modus zum Einsatz.

Bolusmenge (+/-15 %)Max. Atemfrequenz
Pulse-Dose-Einstellung

Bolusgröße

1
2
3
4
5
6
7
8
9

16
32
48
64
80
96
128
160
192

(± 15 %) ml

Schritt 5: Volumenstrom-Einstellung auf verordnetes Niveau einstellenÜber die Volumenstrom-Einstelltaster können Sie den
Volumenstrom bzw. die Bolusmenge an die von Ihrem Arzt
verordnete Einstellung anpassen.
WARNUNG: STELLEN SIE UNBEDINGT NUR DIE VERORDNETE
MENGE SAUERSTOFF EIN. ÄNDERN SIE DIE EINSTELLUNG NUR,WENN DIES AUF ANWEISUNG EINES APPROBIERTEN ARZTESERFOLGT.

Schritt 6: Verwenden des eQuinox beginnen
Max. Atemfrequenz
40
40
40
40
37
31
23
18
15

Hinweis:ÜberschreitetdieAtemfrequenzdieangegebenenWerte,sinktdie
Bolusmenge.
Hinweis: Die Pulse-Dose-Einstellung 1 bis 9 ist nicht identisch mit LPM.
WARNUNG: DIE EINSTELLUNGEN FÜR DEN PULSE-DOSE-MODUS MÜSSEN FÜR JEDEN PATIENTEN INDIVIDUELL ENTSPRECHEND DEM PERSÖNLICHEN BEDARF IN RUHEPOSITION, BEI
BEWEGUNG ODER AUF REISEN ANGEPASST WERDEN.

Atmen Sie normal über die Nasenkanüle.
Hinweis: Die maximale Betriebsdauer im Batteriebetrieb kann verlängert
werden, wenn der eQuinox im Pulse-Dose-Modus (ärztlich zugelassener
Modus) betrieben wird.

Schritt 7: Verwenden des eQuinox-Trolleys
Der eQuinox-Trolley wurde für die Verwendung auf Oberflächen
konzipiert, die üblicherweise von Fußgängern genutzt werden.
Richten Sie den eQuinox am Boden des Trolleys aus, sodass er
gerade und aufrecht steht.
Richten Sie die große Gewindeschraube auf die Bohrung an derRückseite des eQuinox aus. Ziehen Sie nun den Knauf am Trolleyhandfest an, bis der eQuinox-Trolley fest mit dem Gerät verbundenist.
Drücken Sie den Druckknopf am Griff des eQuinox-Trolleys
herunter und stellen Sie die Griffhöhe ein.

WARNUNG: IM PULSE-MODUS KANN DER EQUINOX EVENTUELL NICHTJEDEN ATEMZYKLUS ERKENNEN. WENN DER EQUINOX AUF IHRENATEMZYKLUS NICHT REAGIERT, BITTEN SIE IHREN PFLEGEDIENST UM
EINE PRÜFUNG DER EMPFINDLICHKEITSEINSTELLUNG.
Hinweis: An den Bereitsteller des Gerätes: Folgendes Sauerstoffanwendungszubehör wird für die Verwendung mit dem Sauerstoffkonzentrator
empfohlen:• Nasenkanüle mit 2,1 m langem Schlauch: PN 5408-SEQ• Weiteres empfohlenes Zubehör finden Sie im Zubehörkatalog(PN ML-LOX0010) auf www.CAIREmedical.com.
• CAIRE bietet die OxySafe™ als optionales Zubehör an. Dieses Zubehör
ist für die Verwendung zusammen mit dem eQuinox-Konzentrator
vorgesehen. Für Kunden in Gebieten, in denen die Einhaltung von EN
ISO 8359:1996-Ergänzung 1:2012 erforderlich ist, erfüllt dieses Zubehör
diese Anforderung.
Die OxySafe™ ist eine thermische Sicherung, die den Gasfluss stoppt,
wenndieflussabwärtsgerichteteKanüleoderderSauerstoffschlauchFeuer
fängt und in Richtung OxySafe abbrennt. Die Sicherung wird linear zur
Nasenkanüle oder dem Sauerstoffschlauch zwischen dem Patienten unddem Sauerstoffauslass des eQuinox eingesetzt.
InformationenzurrichtigenVerwendungderOxySafefindenSieindenAnweisungendesHerstellers(imLieferumfangjedesOxySafe-Kitsenthalten).
CAIRE bietet ein OxySafe™-Kit an, das ein Kanülenkopfstück, OxySafe™
und einen 7’-Schlauch enthält: PN 20628667

Hinweis: Laufen Sie NICHT umher, während der eQuinox in Betrieb und ein
Befeuchter angeschlossen ist. Nehmen Sie die Befeuchterflasche vor dem
Umherlaufen ab. Legen Sie den eQuinox NICHT flach hin, während dieser aneine Befeuchterflasche angeschlossen ist.

Schritt 8: Gerät ausschalten
Halten Sie den Ein-/Aus-Taster für zwei (2) Sekunden gedrückt, um
den eQuinox auszuschalten.
Lagern Sie den eQuinox kühl und trocken. Die Lagerungsdauer darf
einen (1) Monat nicht überschreiten. Der eQuinox muss mindestens
2 Stunden im Monat betrieben werden, um die ordnungsgemäße
Funktion des Geräts sicherzustellen.
eQuinox (Model 4000) User Manual — 67

Störungsbehebungstabelle für den Benutzer
Ignorieren Sie keinen Alarm
WARNUNG: FÜR DEN FALL EINES STROMAUSFALLS, ALARMS ODER MECHANISCHEN VERSAGENS EMPFIEHLT DER HERSTELLER EINEALTERNATIVE SAUERSTOFFZUFUHR. KONSULTIEREN SIE IHREN ARZT ODER GERÄTELIEFERANTEN BEZÜGLICH DES TYPS DES BENÖTIGTEN
RESERVESYSTEMS.

Gelbe Lampe
Symptom

Mögliche Ursache

Maßnahme

1) Entfernen Sie das Batteriemodul und setzen Sie es erneut ein, um sicherzustellen, dass es
Die gelbe Lampe leuchtet
DereQuinoxhateinProblemmit fest im Konzentrator sitzt.
dauerhaft. Alle 30 Sekunden der Verbindung zum Batterie2) Setzen Sie, sofern möglich, ein anderes Batteriemodul in den eQuinox ein.
ertönt ein Alarmsignal.
modul festgestellt.
3)SolltedasProblemfortbestehen,setzenSiesichbittemitIhremPflegedienstinVerbindung.
Die gelbe Lampe blinkt. Das
Batteriesymbol blinkt. Alle
2 Minuten ertönt ein Alarmsignal.

1) Schließen Sie den eQuinox wieder an die externe Stromversorgung an oder setzen Sie ein
vollständig geladenes Batteriemodul in das Gerät ein. Lassen Sie das heiße Batteriemodul
außerhalb des Konzentrators für 30 Minuten abkühlen.

Das eQuinox-Batteriemodul ist
2) Laden Sie das Batteriemodul mittels Wechselstrom auf.
heiß.
3) Ersetzen Sie das Batteriemodul, sofern Ihnen ein anderes zur Verfügung steht.

4) Sollte das Problem fortbestehen, setzen Sie sich mit Ihrem Pflegedienst in Verbindung.
Die gelbe Lampe blinkt. Das
Batteriesymbol blinkt. Alle
30 Sekunden ertönen zwei
Alarmsignale.

1)VerbindenSiedeneQuinoxmitdemWechselstrom-NetzteilundladenSiedasBatteriemodDieSpannungdeseQuinox-Bat- ul auf. Alternativ können Sie das entladene Batteriemodul durch ein vollständig geladenes
teriemoduls ist niedrig.
ersetzen.
1) Reinigen Sie den Lufteinlassfilter bzw. tauschen Sie diesen aus.
2) Stellen Sie sicher, dass die Kanüle weder umgeknickt noch verstopft ist. BeiVerwendung
einer Befeuchterflasche ist sicherzustellen, dass diese ordnungsgemäß befüllt ist und keine
Blockierung verursacht.

Der eQuinox hat eine DurchDie gelbe Lampe blinkt. Alleflussstörungfestgestellt,dieden
2Minutenertönteineinzelnes
Durchfluss beeinträchtigt bzw. 3) Stellen Sie sicher, dass der eQuinox ordnungsgemäß belüftet wird. Das Gerät muss mindAlarmsignal.
estens 7,5 cm von jeder Art von Oberfläche entfernt sein, um ein Blockieren der Lufteinlassverhindert.
und Abluftöffnungen zu verhindern.

4) Sollte das Problem fortbestehen, wechseln Sie zu einer alternativen Sauerstoffquelle undsetzen Sie sich mit Ihrem Pflegedienst in Verbindung.
1) Stellen Sie sicher, dass der Lufteinlassfilter nicht verstopft oder verdeckt ist. Reinigen Sie
den Filter bzw. tauschen Sie diesen ggf. aus.
2)StellenSiesicher,dasssichdereQuinoxineinergutbelüftetenUmgebungbefindet.Stellen
Sie sicher, dass zwischen der Rückseite des eQuinox und anderen Gegenständen (Möbel,Die gelbe Lampe blinkt. Es
Der eQuinox hat ein niedriges Vorhänge usw.) ein Mindestabstand von 7,5 cm eingehalten wird.
ist kein akustischer Alarm zu
Sauerstoffniveau festgestellt.
hören.
3)StellenSiebeiderVerwendungdeseQuinoximAutosicher,dassdieRückseitedeseQuinox
von der Sitzoberfläche weg zeigt.
4) Sollte das Problem fortbestehen, wechseln Sie zu einer alternativen Sauerstoffquelle undsetzen Sie sich umgehend mit Ihrem Pflegedienst in Verbindung.

68 — 4805-C1 C (German)

Rote Lampe
Symptom

Mögliche Ursache

Maßnahme
1) Stellen Sie sicher, dass der Lufteinlassfilter nicht verstopft oder verdeckt ist. Reinigen Sie den
Filter bzw. tauschen Sie diesen ggf. aus.

Die rote Lampe blinkt. AlleDer eQuinox hat ein niedriges
2 Minuten ertönen drei
Sauerstoffniveau festgestellt.
Alarmsignale.

2) Stellen Sie sicher, dass sich der eQuinox in einer gut belüfteten Umgebung befindet. Stellen
Sie sicher, dass zwischen der Rückseite des eQuinox und anderen Gegenständen (Möbel,Vorhänge usw.) ein Mindestabstand von 7,5 cm eingehalten wird.
3) Stellen Sie bei derVerwendung des eQuinox im Auto sicher, dass die Rückseite des eQuinox
von der Sitzoberfläche weg zeigt.
4) Sollte das Problem fortbestehen, wechseln Sie zu einer alternativen Sauerstoffquelle undsetzen Sie sich umgehend mit Ihrem Pflegedienst in Verbindung.
2) Überprüfen Sie die externe Stromversorgung (Gleich- oderWechselstrom) und stellen Sie
sicher,dassalleSteckverbindungenameQuinox,amTransformatorsowieanderSteckdosefest
sitzen.

Der eQuinox hat bei AnschlussaneineexterneStromversorgung3) Wenn Sie die Möglichkeit dazu haben, versuchen Sie, eine andere Steckdose (Gleich- odervorübergehend keinen Strom
Wechselstrom) zu verwenden.
Die rote Lampe blinkt. Eserhalten.
ertönt ein durchgehender3) Sollte das Problem fortbestehen, wechseln Sie zu einer alternativen Sauerstoffquelle und
Alarmton. Auf dem LCD
setzen Sie sich umgehend mit Ihrem Pflegedienst in Verbindung.
wird ******** angezeigt.
Der eQuinox liefert keinen
1) Schließen Sie den eQuinox wieder an die externe Stromversorgung an oder setzen Sie ein
Sauerstoff und lässt sichvollständig geladenes Batteriemodul in das Gerät ein. Lassen Sie das heiße Batteriemodul
nicht einschalten.
Der eQuinox ist aufgrund eines außerhalb des Konzentrators für 30 Minuten abkühlen.
entladenen oder eines überhitz- 2) Laden Sie das Batteriemodul mittels Wechselstrom auf.
ten Batteriemoduls stromlos.
3) Ersetzen Sie das Batteriemodul, sofern Ihnen ein anderes zur Verfügung steht.

4) Sollte das Problem fortbestehen, setzen Sie sich mit Ihrem Pflegedienst in Verbindung.

Die rote Lampe leuchtet
dauerhaft.AufdemDisplay
erscheint die Meldung
Im eQuinox ist eine System„FAIL“. Der eQuinox liefert störung aufgetreten.
keinen Sauerstoff und lässt
sich nicht einschalten.

1) Entnehmen Sie das Batteriemodul und trennen Sie den eQuinox vom Gleich- bzw.Wechselstrom-Netzteil.DieroteLampemusserlöschenunddieFAIL-MeldungvomDisplayverschwinden, wenn die Stromquellen getrennt werden. Warten Sie ca. 20 Sekunden, bevor Sie die
Stromquellen erneut anschließen. Versuchen Sie nun, den eQuinox erneut einzuschalten.
2)Wenn sich der eQuinox wieder einschalten lässt, beobachten Sie das Gerät weiterhin, um zu
prüfen, ob die FAIL-Meldung erneut angezeigt wird. Ist dies der Fall, setzen Sie sich umgehend
mit Ihrem Pflegedienst in Verbindung.
3)Wenn sich der eQuinox nicht wieder einschalten lässt, setzen Sie sich umgehend mit Ihrem
Pflegedienst in Verbindung.

eQuinox (Model 4000) User Manual — 69

Andere AlarmbedingungenSymptom

Mögliche Ursache

Maßnahme

1)MöglicherweisefunktioniertderPulse-Dose-Modusdeshalbnichteinwandfrei,weilsichderBenutzerineinem
Zustandbefindet,beidemderNasenraumverstopftoderverengtist.(DieskannbeiErkrankungen,einerverkrümmten Nasenscheidewand usw. der Fall sein.)
Im Pulse-Modus
2) Stellen Sie sicher, dass die Kanüle eine Länge von 2,1 m nicht überschreitet.
blinkt die Pulse-Mo3) Stellen Sie sicher, dass die Kanüle nicht umgeknickt oder verstopft ist.
dus-LED schnell für4)WenneineBefeuchterflascheangeschlossenist,nehmenSiedieseabundverbindenSiedieKanüledirektmit
15 Sekunden.
demSauerstoffauslassdesGeräts.DerPulse-Dose-ModuskanninVerbindungmiteinerBefeuchterflaschenichtverwendet werden.
5) Sollte das Problem fortbestehen, setzen Sie sich mit Ihrem Pflegedienst in Verbindung.
1) Prüfen Sie, ob auf der Steckdose Strom liegt. Schließen Sie das Netzteil nicht an dimmbare Stromkreise oder
Mehrfachsteckdosen an.
2)PrüfenSie,obdieKabelverbindungenderNetzteilefestinderSteckdosebzw.imFahrzeuganschlusssowieim
Konzentratorsitzen.BeiVerwendungeinesWechselstrom-NetzteilsistzudemdieKabelverbindungamTransformaDas Batteriemodul
tor selbst zu prüfen.
entlädt sich, obwohl
DereQuinoxerhältvon
das Gerät mit einer3) Stellen Sie sicher, dass die Prüflampen für die Stromversorgung leuchten.
der externen StromexternenGleich-oderquelle keinen Strom. - Am Netzteil muss eine grüne Lampe leuchten.
Wechselstromquelle- Am Bedienfeld des eQuinox muss eine grüne Lampe in Form eines Stromkabels leuchten.
verbunden ist.
4)LeuchtendiePrüf-LEDsfürdieStromversorgungnicht,trennenSiealleStromverbindungenfüreineDauervon
20 Sekunden und schließen Sie diese dann erneut an.
5) Sollte das Problem fortbestehen, setzen Sie sich mit Ihrem Pflegedienst in Verbindung.
1) Schließen Sie das Batteriemodul zum Aufladen an eine Wechselstromquelle an. Stellen Sie vor der erneuten
Das Batteriemodul
Verwendung sicher, dass die Batterie zwei bis fünf Stunden aufgeladen wird und das Batteriesymbol voll ist undwurdenichtvollständig nicht mehr blinkt.
aufgeladen.
2)GehenSiezumSymptom„DasBatteriemodullädtnichtauf“undergreifenSiediedortgenanntenMaßnahmen.
Der eQuinox hat nicht 1)StellenSiesicher,dassSiedenvonIhremArztverschriebenenVolumenstrombzw.dieverschriebeneBolusmenge
den gewünschten
verwenden und der richtige Betriebsmodus (Continuous-Flow/Pulse-Dose) eingestellt ist.
Die Ladung desVolumenstrom bzw.
Batteriemoduls hält
diegewünschteBolusnicht so lange an, wie
menge geliefert.
sie sollte.
1)VersuchenSie,dasBatteriemodulvollständigzuentladen.DieskönnenSieerreichen,indemSiedeneQuinoxim
Batteriebetrieblaufenlassen,bissichdasGerätvollständigabschaltet.EntnehmenSiedanndieBatterieundlassen
Das Batteriemodul
Sie diese für etwa 30 Minuten abkühlen. Setzen Sie danach die Batterie wieder in den eQuinox ein und laden Sie
zeigt nicht die spezifi- die Batterie über eine Wechselstromquelle wieder auf.
zierte Leistung.
2)WenndieBatterieauchnachdenMaßnahmenausSchritt1nichtdievorgeseheneLaufzeiterreicht,setzenSie
sich mit Ihrem Pflegedienst in Verbindung.
1) Prüfen Sie, ob auf der Steckdose Strom liegt. Schließen Sie das Netzteil nicht an dimmbare Stromkreise oder
Mehrfachsteckdosen an.
2)PrüfenSie,obdieKabelverbindungenderNetzteilefestinderSteckdosebzw.imFahrzeuganschlusssowieim
Konzentratorsitzen.BeiVerwendungeinesWechselstrom-NetzteilsistzudemdieKabelverbindungamTransformaDer eQuinox erhält
tor selbst zu prüfen.
über die externe3) Stellen Sie sicher, dass die Prüflampen für die Stromversorgung leuchten.
Wechselstromquellekeinen Strom zum- Am Netzteil muss eine grüne Lampe leuchten.
Aufladen der Batterie.
- Am Bedienfeld des eQuinox muss eine grüne Lampe in Form eines Stromkabels leuchten.
Das Batteriemodul
4)LeuchtendiePrüf-LEDsfürdieStromversorgungnicht,trennenSiealleStromverbindungenfüreineDauervon
lädt nicht auf.
20 Sekunden und schließen Sie diese dann erneut an.
5) Sollte das Problem fortbestehen, setzen Sie sich mit Ihrem Pflegedienst in Verbindung.
1)EntfernenSiedasBatteriemodulundsetzenSieeserneutein,umsicherzustellen,dassesfestimKonzentratorsitzt.
DereQuinoxkannden
AtemzyklusdesBenutzers nicht erkennen.
Auf Seite 15 finden Sie
nähereInformationen
zum Pulse-Dose-Modus.

Der eQuinox signalisiertdemBatteriemodul
nichtordnungsgemäß,dass sich dieses nun
aufladen kann.

2) Beobachten Sie, ob am eQuinox bei eingesetzter Batterie eine gelbe Lampe dauerhaft leuchtet oder beim
VersuchzuladeneinBatteriesymbolblinkt.LiegteinesdieserSymptomevor,setzenSiedieStörungsbehebungmit
Schritt 3 fort.
3)SetzenSie,sofernmöglich,einanderesBatteriemodulindeneQuinoxein.WenndieErsatzbatterieordnungsgemäß funktioniert, muss das entnommene Batteriemodul ausgetauscht werden.
4) Sollte das Problem fortbestehen, setzen Sie sich bitte mit Ihrem Pflegedienst in Verbindung.
Beim ersten EinDieintegrierteBatterie, 1) Setzen Sie sich mit Ihrem Pflegedienst in Verbindung.
schaltendeseQuinox die den Alarm mit
ertönen drei Alarm- Strom versorgt, ist
signale.
beinahe leer.
70 — 4805-C1 C (German)

Reinigung, Pflege und regelmäßige Wartung
Regelmäßige Wartung

Lufteinlassfilter

Arbeiten an den internen Teilen des eQuinox dürfen nur von einem
von CAIRE ausgebildeten und qualifizierten Servicetechniker
ausgeführt werden.

Durch die Lufteinlassöffnungen oben an der Rückseite des Geräts
wird Umgebungsluft in das Gerät gesaugt. Die Reinigung desLufteinlassfilters ist die wichtigste Wartungsaufgabe, die Sie als
Benutzer verrichten. Durch die Reinigung des Lufteinlassfilters
bleibt die Funktionsfähigkeit des eQuinox erhalten. Überprüfen Sie
den Lufteinlassfilter täglich und reinigen Sie ihn wenigstens einmal
die Woche. Gehen Sie bei der Reinigung wie folgt vor:

Einmal jährlich sollte eine vorbeugende Wartung durchgeführt
werden. Setzen Sie sich für die Terminabsprache für die
vorbeugende Wartung mit Ihrem Pflegedienst in Verbindung. Ihr
Pflegedienst bzw. der qualifizierte Techniker prüft die folgenden
Teile nur bei Bedarf: Alarme, interne 9-Volt-Batterie, Batterie,interne Filter und andere interne Teile.
Hinweis:WennnachdemEinschaltselbsttestdreiSignaltönezuhörensind,ist
derLadestandder9-V-Batterieniedrig.WendenSiesichanIhrenPflegedienste, wenn Sie drei Signaltöne hören.

Reinigung und Pflege des Geräts durch den Benutzer
KanülenaustauschTauschen Sie den Versorgungsschlauch sowie die Kanüle gemäß
den Empfehlungen Ihres Pflegediensts aus. Informationen zu
Reinigung, Desinfektion und Austausch erhalten Sie von Ihrem Arzt
oder Pflegedienst.
Hinweis:FolgenSiezumordnungsgemäßenGebrauchstetsdenAnleitungendes Kanülenherstellers. Ersetzen Sie die Einweg-Kanüle gemäß den EmpfehlungendesKanülenherstellersoderdesBereitstellersIhresGerätes.Weiteres
Zubehör erhalten Sie vom Bereitsteller Ihres Gerätes.

1. Entfernen Sie den Lufteinlassfilter vom Gerät.
2. Waschen Sie den Filter unter warmem Leitungswasser mit
einem milden Spülmittel ab.
3. Spülen Sie den Filter gründlich mit klarem Leitungswasser aus
und pressen Sie das überschüssige Wasser heraus.
4. Lassen Sie den Filter an der Luft trocknen.
5. Setzen Sie den Filter wieder in das Gehäuse ein.
Tipp für die Reinigung: Halten Sie einen zweiten Filter vorrätig, derverwendet wird, während der erste Filter trocknet.

Lufteinlass-filter

VORSICHT: Betreiben Sie den Sauerstoffkonzentrator nicht
ohne eingesetzten Lufteinlassfilter. Wenn ein zweiter Filter
vorhanden ist, setzen Sie den Ersatzfilter ein, bevor Sie denverschmutzten Filter reinigen. Reinigen Sie den verschmutzten Filter
in warmem Seifenwasser und trocknen Sie ihn vor der Verwendung
gründlich ab.
Hinweis: Der Hersteller empfiehlt, dieses Gerät nicht zu sterilisieren.
Hinweis: Betreiben Sie den Sauerstoffkonzentrator nicht ohne eingesetzten Lufteinlassfilter. Wenn ein zweiter Filter vorhanden ist, setzen Sie den
Ersatzfilterein,bevorSiedenverschmutztenFilterreinigen.ReinigenSieden
verschmutzten Filter in warmem Seifenwasser und trocknen Sie ihn vor der
Verwendung gründlich ab.

Der Lufteinlassfilter sollte jährlich ausgetauscht werden. Wird dereQuinox in staubigen Umgebungen verwendet, muss der Filter
eventuell häufiger ausgetauscht werden. Der eQuinox darf ohne
Lufteinlassfilter nicht länger als 5 Minuten betrieben werden.

eQuinox (Model 4000) User Manual — 71

Wechselstrom-Netzteil, 12-VDC-Kabel, Netzkabel und Gehäuseaußenseite:
Schalten Sie den eQuinox vor dem Reinigen oder Desinfizieren
aus und trennen Sie ihn von der Stromversorgung (Gleich- oder
Wechselstrom). Sprühen Sie das Gehäuse von außen nicht ein.
Verwenden Sie ein feuchtes (nicht nasses) Tuch oder einen
Schwamm. Sprühen Sie das Tuch bzw. den Schwamm zunächst mit
mildem Reinigungsmittel ein und reinigen Sie dann das Gehäusesowie das Netzteil. Verwenden Sie zur Desinfektion des eQuinox
Lysol® Brand II oder ein ähnliches Desinfektionsmittel. Befolgen
Sie die Anweisungen des Reinigungsmittelherstellers.
WARNUNG: STROMSCHLAGGEFAHR. SCHALTEN SIE DASGERÄT AUS UND ZIEHEN SIE DAS NETZKABEL AUS DER STECKDOSE, BEVOR DAS GERÄT GEREINIGT ODER GEWARTET WIRD,
UM DIE GEFAHR EINES STROMSCHLAGS AUSZUSCHLIESSEN. NUR DER
BEREITSTELLER DES GERÄTES ODER EIN QUALIFIZIERTER WARTUNGSTECHNIKER DÜRFEN DIE ABDECKUNGEN ENTFERNEN ODER DAS GERÄTWARTEN. VERWENDEN SIE KEINE FLÜSSIGKEIT DIREKT AUF DEM GERÄT.
ZU DEN NICHT VERTRÄGLICHEN CHEMISCHEN MITTELN ZÄHLEN UNTER
ANDEREM: ALKOHOL UND PRODUKTE AUF ALKOHOLBASIS, KONZENTRIERTE PRODUKTE AUF CHLORBASIS (ETHYLCHLORID) UND PRODUKTE
AUF ÖLBASIS (PINE-SOL®, LESTOIL®). DIESE DÜRFEN NICHT VERWENDET WERDEN, UM DAS KUNSTSTOFFGEHÄUSE DES SAUERSTOFFKONZENTRATORS ZU REINIGEN, DA SIE DEN KUNSTSTOFF BESCHÄDIGENKÖNNEN.
WARNUNG: STROMSCHLAGGEFAHR. ZIEHEN SIE DAS NETZKABEL AUS
DER STECKDOSE, BEVOR DAS GERÄT GEREINIGT ODER GEWARTET WIRD,
UM DIE GEFAHR EINES STROMSCHLAGS AUSZUSCHLIESSEN. NUR DER
BEREITSTELLER DES GERÄTES ODER EIN QUALIFIZIERTER WARTUNGSTECHNIKER DÜRFEN DIE ABDECKUNGEN ENTFERNEN ODER DAS GERÄTWARTEN.
WARNUNG: ES MUSS DARAUF GEACHTET WERDEN, DASS DER SAUERSTOFFKONZENTRATOR UND DIE BATTERIE STETS TROCKEN BLEIBEN UND
KEINERLEI FLÜSSIGKEIT IN DAS GERÄT EINTRITT. ANDERNFALLS KANNES ZU EINER FEHLFUNKTION ODER ZUR ABSCHALTUNG KOMMEN ODER
ES BESTEHT EIN ERHÖHTES RISIKO EINES ELEKTRISCHEN SCHLAGS ODERVON VERBRENNUNGEN.
WARNUNG: REINIGEN SIE GEHÄUSE, BEDIENFELD UND STROMKABELNUR MIT EINEM SANFTEN HAUSHALTSREINIGER, DER MIT EINEM
FEUCHTEN (NICHT NASSEN) TUCH ODER SCHWAMM AUFGETRAGENWIRD. WISCHEN SIE ANSCHLIESSEND ALLE OBERFLÄCHEN TROCKEN.
LASSEN SIE KEINE FLÜSSIGKEITEN IN DAS INNERE DES GERÄTES GELANGEN. ACHTEN SIE BESONDERS AUF DEN SAUERSTOFFAUSGANG FÜR
DIE VERBINDUNG ZUR KANÜLE UND STELLEN SIE SICHER, DASS DIESERFREI VON STAUB, WASSER UND PARTIKELN IST.
WARNUNG: VERWENDEN SIE KEINE FLÜSSIGKEIT DIREKT AUF
DEM GERÄT. ZU DEN NICHT VERTRÄGLICHEN CHEMISCHENMITTELN ZÄHLEN UNTER ANDEREM: ALKOHOL UND PRODUKTE AUF ALKOHOLBASIS, KONZENTRIERTE PRODUKTE AUF CHLORBASIS(ETHYLCHLORID) UND PRODUKTE AUF ÖLBASIS (PINE-SOL®, LESTOIL®).
DIESE DÜRFEN NICHT VERWENDET WERDEN, UM DAS KUNSTSTOFFGEHÄUSE DES SAUERSTOFFKONZENTRATORS ZU REINIGEN, DA SIE DEN
KUNSTSTOFF BESCHÄDIGEN KÖNNEN.

72 — 4805-C1 C (German)

Optional erhältlicher Befeuchter
Wenn Sie einen Befeuchter verwenden, muss dieser täglich gereinigt
werden, um eine Kontaminationsgefahr zu vermeiden. Befolgen Sie
die Reinigungsempfehlungen des Pflegediensts bzw. des Herstellers
des Befeuchters. Tauschen Sie den Befeuchter jeden Monat bzw.
in den vom Hersteller bzw. Ihrem Pflegedienst empfohlenen
Intervallen aus. (Der Befeuchter darf nicht überfüllt werden.) ORSICHT: Laufen Sie NICHT umher, während der eQuinox
Vin Betrieb und ein Befeuchter angeschlossen ist. Nehmen
Sie die Befeuchterflasche vor dem Umherlaufen ab. Legen
Sie den eQuinox NICHT flach hin, während dieser an eine
Befeuchterflasche angeschlossen ist.

eQuinox-Trolley
Entfernen Sie den eQuinox-Trolley vor der Reinigung vom
eQuinox. Verwenden Sie zum Reinigen ein feuchtes (nicht nasses)
Tuch oder einen Schwamm sowie ein mildes Reinigungsmittel.
Batteriemodul (Batterie)
Die in den eQuinox eingesetzte Batterie benötigt für eine lange
Lebensdauer und bestmögliche Leistung eine besondere Pflege.
Es wird empfohlen, den eQuinox nur mit CAIRE-Batterien zu
betreiben.
Verwenden Sie zum Reinigen der Batterie ein feuchtes (nicht
nasses) Tuch oder einen Schwamm. Sprühen Sie das Tuch bzw. den
Schwamm zunächst mit einem milden Reinigungsmittel ein undreinigen Sie dann das Batteriegehäuse sowie den Riegel.
CAIRE empfiehlt, das Batteriemodul nach 2 Jahren auszutauschen.
Entsorgung der Batterie
Die Batterie kann aufgeladen und wiederverwertet werden. Geben
Sie die Batterie zur ordnungsgemäßen Entsorgung an Ihren
Pflegedienst zurück. Für Anweisungen zur ordnungsgemäßen
Entsorgung der Batterie können Sie sich zudem an die zuständigen
Stellen Ihres Wohnorts wenden.
Entsorgung des eQuinox-Sauerstoffsystems
Lokale Umweltvorschriften verbieten möglicherweise die Entsorgung
elektrischer und/oder elektronischer Geräte wie des eQuinox oder desWechselstrom-Netzteils. Wenden Sie sich an Ihre Stadtverwaltung,um weitere Informationen zur ordnungsgemäßen Entsorgung
elektrischer oder elektronischer Geräte zu erhalten. Informationen zur
Entsorgung können Sie auch von CAIRE Inc. anfordern.

EMV-Tabellen
Leitlinien und Herstellererklärung – elektromagnetische Aussendungen
DereQuinoxistfürdenGebrauchineinerwieimFolgendenbeschriebenenelektromagnetischenUmgebungbestimmt.DerKundeoderBenutzer
des eQuinox muss sicherstellen, dass das Gerät in einer entsprechenden Umgebung verwendet wird.
Störaussendungsmessungen
HF-Aussendung
CISPR 11
HF-Aussendung
CISPR 11Übereinstimmung

Elektromagnetische Umgebung – Leitlinien

Gruppe 1

Der eQuinox verwendet HF-Energie ausschließlich für den internen Betrieb. Daher
ist die HF-Aussendung sehr gering, und es ist unwahrscheinlich, dass benachbarte
elektronische Geräte gestört werden.

Klasse B

Oberwellenaussendungen
IEC 61000-3-2

Klasse A

Spannungsschwankungen/
Flicker-Aussendung
IEC 61000-3-3

stimmt überein

Der eQuinox ist für den Gebrauch in allen Einrichtungen einschließlich denen im
Wohnbereichundsolchengeeignet,dieunmittelbaraneinöffentlichesVersorgungsnetzangeschlossensind,dasauchGebäudeversorgt,diezuWohnzweckenbenutzt
werden.

eQuinox (Model 4000) User Manual — 73

Leitlinien und Herstellererklärung – elektromagnetische Störfestigkeit
DereQuinoxistfürdenGebrauchineinerwieimFolgendenbeschriebenenelektromagnetischenUmgebungbestimmt.DerKundeoderBenutzer
des eQuinox muss sicherstellen, dass das Gerät in einer entsprechenden Umgebung verwendet wird.
Störfestigkeitsprüfung

IEC 60601 Prüfpegel

Elektromagnetische±6 kV Kontakt
Umgebung – Leitlinien±8 kV LuftIEC 61000-4-2±2 kV für Netzleitungenschnelle transiente
elektrische Störgrößen/Bursts
±1 kV für Eingangs- und
IEC 61000-4-4
Ausgangsleitungen
Stoßspannungen
IEC 61000-4-5±6 kV Kontakt±8 kV Luft
±2 kV für Netzleitungenentfällt±1 kV Gegentaktspannung ±1 kV Gegentaktspannung±2 kV Gleichtaktspannung<5 % UT<5 % UT(>95 % Einbruch der UT )(>95 % Einbruch der UT )für 0,5 Perioden

für 0,5 Perioden

40 % UT

40 % UT(60 % Einbruch der UT )Spannungseinbrüche,Kurzzeitunterbrechungen für 5 Perioden
und Schwankungen der
Versorgungsspannung 70 % U
Tgemäß IEC 61000-4-11(30 % Einbruch der UT )Magnetfeld bei der
Versorgungsfrequenz(50/60 Hz) gemäß IEC
61000-4-8

Übereinstimmungspegel(60 % Einbruch der UT )für 5 Perioden
70 % UT(30 % Einbruch der UT )für 25 Perioden

für 25 Perioden<5 % UT<5 % UT(>95 % Einbruch der UT )(>95 % Einbruch der UT )für 5 Sekundenfür 5 Sekunden

3 A/m

3 A/m

HINWEIS UT ist die Netzwechselspannung vor der Anwendung des Prüfpegels.

74 — 4805-C1 C (German)

Elektromagnetische Umgebung – Leitlinien
Fußböden sollten aus Holz oder Beton bestehenoder mit Keramikfliesen versehen sein. Wenn derFußboden mit synthetischem Material versehenist, muss die relative Luftfeuchte mindestens 30 %betragen.
Die Qualität der Versorgungsspannung sollte dereinertypischenGeschäfts-oderKrankenhausumgebung entsprechen.
Die Qualität der Versorgungsspannung sollte dereinertypischenGeschäfts-oderKrankenhausumgebung entsprechen.

Die Qualität der Versorgungsspannung sollte dereinertypischenGeschäfts-oderKrankenhausumgebungentsprechen.WennderAnwenderdeseQuinox
auchbeiUnterbrechungenderNetzstromversorgung
aufeinenfortgesetztenBetriebangewiesenist,wird
empfohlen,deneQuinoxübereineunterbrechungsfreieStromversorgungodereineBatteriezuspeisen.

Magnetfelder bei der Netzfrequenz sollten den typischen Werten, wie sie in der Geschäfts- und Krankenhausumgebungvorzufindensind,entsprechen.

Leitlinien und Herstellererklärung – elektromagnetische Störfestigkeit
DereQuinoxistfürdenGebrauchineinerwieimFolgendenbeschriebenenelektromagnetischenUmgebungbestimmt.DerKundeoderBenutzer
des eQuinox muss sicherstellen, dass das Gerät in einer entsprechenden Umgebung verwendet wird.
Störfestigkeitsprüfung

IEC 60601 PrüfpegelÜbereinstimmungspegel

Elektromagnetische Umgebung – Leitlinien

Geleitete HF

3 Veff

3 Veff

IEC 61000-4-6

150 kHz bis 80 MHz

Tragbare und mobile Funkgeräte sollten in keinem
geringerenAbstandzumeQuinoxeinschließlichder
Leitungenverwendetwerdenalsdemempfohlenen
Schutzabstand, der nach der für die Sendefrequenz
zutreffenden Gleichung berechnet wird.

Gestrahlte HF

3 V/m

IEC 61000-4-3

80 MHz bis 2,5 GHz

3 V/m
Empfohlener Schutzabstandd =1,2
1.2 P
d =1,2
1.2 P = 80 MHz bis 800 MHz
d =1,2
1.2 P = 800 MHz bis 2,5 GHz
Mit P als der Nennleistung des Senders in Watt (W)gemäß Angaben des Senderherstellers und d als
empfohlenem Schutzabstand in Metern (m).
DieFeldstärkestationärerFunksendersolltebeiallen
Frequenzen gemäß einer Untersuchung vor Ort ageringer als der Übereinstimmungspegel sein. b
In der Umgebung von Geräten, die das folgende
Bildzeichen tragen, sind Störungen möglich:

ANMERKUNG 1 Bei 80 MHz und 800 MHz gilt der höhere Frequenzbereich.
ANMERKUNG 2 Diese Leitlinien sind möglicherweise nicht in allen Fällen anwendbar. Die Ausbreitung elektromagnetischerWellen wird durch
Absorptionen und Reflexionen von Gebäuden, Gegenständen und Menschen beeinflusst.
Die Feldstärke stationärer Sender, z. B. Basisstationen von Funktelefonen und mobilen Landfunkgeräten, Amateurfunkstationen,
AM-undFM-Rundfunk-undFernsehsendern,könnentheoretischnichtgenauvorherbestimmtwerden.UmdieelektromagnetischeUmgebung
hinsichtlichderstationärenSenderzuermitteln,sollteeineStandortanalyseerwogenwerden.WenndiegemesseneFeldstärkeandemStandort,
an dem der eQuinox benutzt wird, die obigen Übereinstimmungspegel überschreitet, sollte der eQuinox beobachtet werden, um die bestimmungsgemäßeFunktionnachzuweisen.WenneineanormaleLeistungfestgestelltwird,könnenzusätzlicheMaßnahmenerforderlichsein,z.B.
eine veränderte Ausrichtung oder ein anderer Standort des eQuinox.
abÜber den Frequenzbereich von 150 kHz bis 80 MHz muss die Feldstärke geringer als 3 V/m sein.

eQuinox (Model 4000) User Manual — 75

Empfohlene Schutzabstände
zwischen tragbaren und mobilen HF-Kommunikationsgeräten und dem eQuinox
DereQuinoxistfürdenBetriebineinerelektromagnetischenUmgebungbestimmt,inderdieHF-Störgrößenkontrolliertsind.DerKundeoderder
BenutzerdeseQuinoxkanndazubeitragen,elektromagnetischeStörungenzuvermeiden,indemerdenMindestabstandzwischentragbarenund
mobilenHF-Telekommunikationsgeräten(Sendern)unddemeQuinox–abhängigvonderAusgangsleistungdesKommunikationsgeräts–wie
unten angegeben einhält.
Nennleistung des Senders

Schutzabstand abhängig von der SendefrequenzmW

150 kHz bis 80 MHz

80 MHz bis 800 MHz

800 MHz bis 2,5 GHz

0,01

0,12

0,12

0,23

0,1

0,38

0,38

0,73

1

1,2

1,2

2,3

10

3,8

3,8

7,3

100

12

12

23

d = 11,2.2 P

d  =1.1,2
2 P

d  1=.21,2P

FürSender,derenmaximaleNennleistunginobigerTabellenichtangegebenist,kannderempfohleneSchutzabstanddinMetern(m)unterVerwendungderGleichungermitteltwerden,diefürdiejeweiligenSendefrequenzgilt.DabeiistPdiemaximaleNennleistungdesSendersinWatt(W) gemäß Angabe des Senderherstellers.
ANMERKUNG 1 Bei 80 MHz und 800 MHz gilt der Schutzabstand für den höheren Frequenzbereich.
ANMERKUNG2DieseLeitliniensindmöglicherweisenichtinallenFällenanwendbar.DieAusbreitungelektromagnetischerWellenwirddurch
Absorptionen und Reflexionen von Gebäuden, Gegenständen und Menschen beeinflusst.

76 — 4805-C1 C (German)

Allgemeine Informationen zu Flugreisen Batteriemodul und Flugreisen
Wenn der eQuinox bei Rangiervorgängen sowie während desStarts und der Landung nicht verwendet wird, muss der eQuinox
ausgeschaltet bleiben und unter dem Sitz bzw. in einem anderenzugelassenen Stauraum so verstaut werden, dass weder derMittelgang noch der Zugang in die Sitzreihe blockiert werden.
Wenn der eQuinox bei Rangiervorgängen sowie während des Starts
und der Landung verwendet wird, muss sich der Benutzer in einer
Sitzposition befinden, die andere Passagiere nicht am Zugang bzw.
an der Nutzung von Notausgängen, normalen Ausgängen sowie
Mittelgängen in der Passagierkabine des Flugzeugs hindert.
Eine Person, die einen eQuinox verwendet, darf nicht in einer
Sitzreihe platziert werden, in der sich ein Ausgang befindet.
Kommt es während der Verwendung des eQuinox in der Kabine zu
einem Druckabfall und fallen die Sauerstoffmasken heraus, mussdie Nutzung des eQuinox beendet und das Zusatzsauerstoffsystem
des Flugzeugs genutzt werden. Der eQuinox muss nach dem
Anlegen der Sauerstoffmasken abgeschaltet werden.
Vor dem Antritt der Reise muss der eQuinox vom Benutzer auf
einwandfreien Betrieb geprüft werden.

Kalkulieren Sie die Zeit am Boden (Check-in vor
dem Flug, Sicherheitskontrollen usw.) sowie
eventuelle Wartezeiten mit ein.
Die meisten Fluggesellschaften fordern, dass eine ausreichende
Zahl an vollständig geladenen Batterien zur Stromversorgung
des Geräts mitgeführt werden muss, die für 150 % der erwarteten
maximalen Reisezeit (Flugdauer, gesamte Zeit am Boden, d. h.
Zeiten vor und nach dem Flug sowie während des Umsteigens undunerwartete Verspätungen) ausreicht.

Es wird empfohlen, stets eine zusätzliche Batterie auf
die Reise mitzunehmen.
Jede Fluggesellschaft hat ihre eigenen Bestimmungen bezüglichder Mitnahme von Sauerstoff.
Setzen Sie sich mindestens 48 Stunden vor dem Abflug mit der
Fluggesellschaft in Verbindung und teilen Sie dieser mit, dass Sie
den eQuinox mit auf die Reise nehmen.

VORSICHT: Die Richtlinien des Verkehrsministeriums der USA(DOT) sowie der Vereinten Nationen (UN) sehen vor, dassdie Batterie aus dem Gerät herausgenommen werden muss,
wenn dieses als Gepäck auf internationalen Flügen aufgegeben wird.
Bei Versand des Sauerstoffkonzentrators müssen die Batterien ebenfalls aus dem Gerät entnommen und ordnungsgemäß verpackt werden.

eQuinox (Model 4000) User Manual — 77

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107 USA
1-800-482-2473
Weltweit: www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Deutschland
Weltweit: www.CAIREmedical.com

Chart Industries behält sich das Recht vor, dieVermarktung seiner Produkte einzustellen bzw. Preise,Werkstoffe, Ausrüstungsteile,
Qualität,Beschreibung,Spezifikationenbzw.ProzesseohneVorankündigungzueinembeliebigenZeitpunktzuändern,ohnedasshierausirgendwelcheVerpflichtungenoderRechtsfolgenentstehen.AllehiernichtausdrücklichgenanntenRechtebleibenimgesetzlich
zulässigen Rahmen Chart Industries vorbehalten.

78 — 4805-C1 C (German)

Zum Zeitpunkt der Veröffentlichung dieses Handbuchs sind eQuinox, SeQual und autoSAT
sowie die zugehörigen Logos (falls zutreffend) von CAIRE Inc. eingetragene Marken.
Lysol® ist eine eingetragene Marke von Reckitt Benckiser (Großbritannien).
OxySafe® ist eine eingetragene Marke von Applied Home Healthcare Equipment.
Pine-Sol® und Lestoil® sind eingetragene Marken der The Clorox Company.

Manual del usuario

eQuinoxTM

Modelo 4000
eQuinox (Model 4000) User Manual — 79

Controles del usuario e indicadores de estado del sistema
SímboloA

02

Definición

Definición

Lea el manual del usuario antes de utilizar el equipo.
Lea el manual del usuario para obtener instrucciones.

Símbolo de Aprobado por la FAA: La Administración
Federal de Aviación (FAA) de los Estados Unidos haaprobado el uso de este dispositivo a bordo de aviones
comerciales.

Icono de Prohibido fumar: No fumar en las
proximidades de la unidad.

Indicador del ajuste de flujo

Indicador de Advertencias/ALERTA (amarillo)

Funcionamiento en modo de dosis pulsada

Indicador de ALARMA (rojo)

El dispositivo opera normalmente; botón de encendido

Indicador de alimentación externa

Aumento del ajuste de flujo

Indicador de estado del paquete de baterías (batería)

Reducción del ajuste de flujo

Símbolo del paquete de baterías

No utilizar sustancias oleosas ni grasas.

Amperios

No utilizar con llamas vivas.

Corriente alterna

No contiene piezas reparables en el interior. No abrir lacubierta.

Icono de CEI para la alimentación de CC (entrada)

Superficie caliente.

Salida de oxígeno

No mojar.

Pieza aplicada de tipo BF (grado de protección contra
descargas eléctricas)

Equipo IPX1 a prueba de goteo: El paquete de baterías(batería) de eQuinox y la fuente de alimentación de CA
proporcionan protección contra los efectos dañinos por
ingreso de líquidos. (IPX1, según CEI 60529)

Símbolo de reciclaje

IP22

Símbolo

Este símbolo recuerda a los propietarios de los equipos
que deben devolverlos a una instalación de reciclado al
final de su vida útil, según la Directiva de Residuos deAparatos Eléctricos y Electrónicos (RAEE).
Prueba de ingreso de goteo y elementos sólidos:
El sistema eQuinox proporciona protección contra losefectos dañinos por ingreso de objetos extraños sólidos
y líquidos, de conformidad con la normativa CEI
60529 con calificación de IP22.
Certificado para los mercados de los EE. UU. y el
Canadá, según las normas aplicables de ambos países.

80 — 4805-C1 C (Spanish)

IPX1

Nombre y dirección del fabricante
Representante autorizado en la Comunidad Europea

Equipo de clase II

Este dispositivo cumple con los requisitos de la
Directiva 93/42/CEE sobre dispositivos médicos.
Por consiguiente, lleva la marca CE como se muestra.

Índice
Controles del usuario e indicadores de estado del sistema...........................................................................................80
Guía de inicio rápido....................................................................................................................................................................82
Indicaciones de uso......................................................................................................................................................................85
Contraindicaciones de uso........................................................................................................................................................85
Directrices de seguridad.............................................................................................................................................................85
Entornos de operación recomendados.................................................................................................................................87
Instrucciones de funcionamiento...........................................................................................................................................88
Funcionamiento del módulo de voz......................................................................................................................................91
Operación del sistema eQuinox por primera vez..............................................................................................................92
Tabla de solución de problemas del usuario ......................................................................................................................94
Limpieza, cuidados y mantenimiento periódico...............................................................................................................97
Tablas de CEM.................................................................................................................................................................................99
Información general para viajes aéreos............................................................................................................................. 103
Paquete de baterías y viajes aéreos..................................................................................................................................... 103

eQuinox (Model 4000) User Manual — 81

Guía de inicio rápido

1

Desembalaje de su sistema eQuinox

2

Conocimiento de su sistema eQuinox

Revise todas las advertencias, las precauciones y la información
adicional del dispositivo en el resto de este manual.
Familiarícese con las funciones clave del sistema eQuinox y conel panel de control del usuario.

Puerto desalida deoxígenoPARTE DELANTERA
Asa

Panel decontrol
Toma de alimentación externa

Fuente de alimentación de CA

Icono deAprobado
por la FAA

Cierre delpaquete debaterías

Ubicación dela etiqueta declasificación
y del número
de serie

Carrito para el eQuinox

Paquete debaterías

PARTE TRASERA

Cable de 12 VCC

Gabinete deservicio

Aberturas
de escapePaquete debaterías (batería)

82 — 4805-C1 C (Spanish)

Filtro de entrada de aireEl panel de control del usuario delsistema eQuinox muestra informaciónde funcionamiento importante.

Indicador(verde)ybotóndeENCENDIDO/APAGADO:Este
botónENCIENDEyAPAGAeldispositivo.Elindicadorverde
seenciendecuandoeldispositivoestáencendidoyfunciona de forma correcta.
Botonesdeaumentoodisminucióndelajustedeflujo:Utilice
estos botones para ajustar el flujo al parámetro prescrito.

3

Encender el sistema eQuinox

Para conectar a una fuente de alimentación de CA: Inserte el cablede la fuente de alimentación de CA firmemente, en el receptáculo
de alimentación externa, en el lado derecho del dispositivo. Conecte
el cable de alimentación de CA en la fuente de alimentación en lasalida de CA.
Paso 1

Paso 2

Paso 3

Indicadorybotóndelmododesuministro:Elbotóncambia
entre flujo continuo y modo de dosis pulsada. El modo dedosispulsadaactivalatecnologíaautoSAT.Estopermiteun
aumento significativo en el tiempo operativo mientras es
accionado por el paquete de baterías (batería). Cuando se
activa el modo de dosis pulsada, se enciende la luz verdedel indicador del modo de dosis pulsada. Se suministra un
pulso de oxígeno y el indicador verde parpadea con cadaesfuerzo inspiratorio.
Indicador de ALERTA (amarillo): cuando está encendido,
indica una condición de atención de baja prioridad oprecaución. Siga utilizando su sistema. Consulte la tabla
de solución de problemas para conocer la acción correcta
opóngaseencontactoconelproveedordeatencióndomiciliaria. Una luz parpadeante amarilla indica una alerta deprioridad media. Se necesita una respuesta rápida. (Normalmente, indica un nivel bajo de oxígeno o de la batería).
Indicador de ALARMA (rojo): indica una condición de alarma de alta prioridad. Se necesita una respuesta inmediata.
Consulte la tabla de solución de problemas y póngase
en contacto con su proveedor de atención domiciliaria.
(Normalmente, indica una pérdida de energía, un nivel deoxígeno bajo o un funcionamiento defectuoso).
Pantalla de información:• C o P: "C" indica modo de flujo continuo y "P" indica
modo de dosis pulsada.
• Tiempo en la batería: la pantalla se apagará y se encenderá cada 20 segundos cuando funcione con alimentaciónde la batería para indicar el tiempo restante estimado enella.
• Ajuste (flujo continuo o dosis pulsada): cada vez que
enciendaeldispositivo,sehaguardadoelúltimomodoy/oajuste y se utilizará en el inicio.

Cuando el sistema eQuinox se conecta debidamente, se enciende
un indicador verde en la fuente de alimentación de CA. También
se encenderá el indicador de alimentación externa en el panel decontrol del usuario.
Para instalar la batería: Deslice la batería hacia el interior delcompartimiento vacío hasta que se trabe en su posición. Quedará
nivelada con la parte delantera del dispositivo.

Cuando la batería esté instalada correctamente y el sistema eQuinox
esté encendido o conectado a una alimentación de CA, aparecerá elindicador de estado del paquetede baterías.

Indicador de estado del paquete de baterías (batería): Este
indicador muestra la carga restante de la batería. Cada
una de las cinco barras grises horizontales representa
aproximadamente el 20 % de la carga total de la batería.
Cuando se carga la batería, las barras del indicador decarga parpadearán en forma continua. Si la batería noestá instalada, o si está instalada de forma incorrecta, el
indicadordeestadodelpaquetedebaterías(batería)nose
iluminará ni parpadeará.
Indicador de alimentación externa: Cuando el sistema
eQuinox esté correctamente enchufado y utilice la fuentede alimentación de CA o de 12 VCC, este indicador se
iluminará en el panel de control del usuario.
Timbre:Unaalarmasonoraleadvierteacercadelacondicióndefuncionamientodel dispositivo, ya sea una advertencia o falla, y confirma la pulsación de una
tecla válida.
Módulo de voz: Si está activado, un indicador de voz le advertirá acerca
de la condición de funcionamiento del dispositivo, cambios en la fuente dealimentación y condiciones de advertencias o fallas.

Indicador (verde) y botón de ENCENDIDO/APAGADO
Mantenga pulsado el botón de ENCENDIDO/APAGADO durante
dos (2) segundos para encender su sistema eQuinox. Se iniciará una
breve prueba automática visual y sonora de encendido. Verifiqueque todos los indicadores estén iluminados y que el timbre suene
por tres (3) segundos. Mantenga pulsado el botón de ENCENDIDO/APAGADO durante dos (2) segundos para apagar el sistema
eQuinox.
eQuinox (Model 4000) User Manual — 83

4

Seleccionar el modo de entrega de flujo

Indicador y botón del modo de flujo

Pulse el botón de modo de flujo para seleccionar el modo de flujo
deseado. Al pulsar este botón repetidas veces, alternará entre elmodo de flujo continuo y el modo de dosis pulsada. En el modo deflujo continuo, el oxígeno es proporcionado en una frecuencia deflujo constante entre 0,5 y 3,0 LPM, indicada por la letra C. En elmodo de dosis pulsada, el oxígeno es suministrado en un bolo al
comienzo da cada inhalación. Los ajustes del modo de dosis pulsada
1–9 proporcionan tamaños de bolos que van de 16 ml a 192 ml
y están designados con la letra P en la pantalla. La dosis pulsada
puede ajustarse en relación a la sensibilidad y al suministro del bolo
para la respiración. Los ajustes de flujo en ambos modos deben serprescritos por su médico.

5Adecuar el ajuste de flujo al nivel indicado

6

Comenzar a usar el sistema eQuinox

Filtro de entrada de aire

Filtro deentradade aireSu unidad está equipada con un filtro de entrada de aire (filtro
adicional provisto). Compruébelo diariamente y límpielo una vez ala semana. Las instrucciones para la limpieza están en la página 19.
Conecte el tubo de suministro de oxígeno a la salida de oxígeno yconecte la cánula de oxígeno según las instrucciones provistas conla cánula, o según las instrucciones de su proveedor de atención
domiciliaria.

Respire con normalidad a través de la cánula nasal.
Al operar con la energía de la batería, con la utilización del modo dedosis pulsada se obtendrá una mayor duración de uso.

Botones de aumento o disminución del ajuste deUsando los botones + o -, adapte el ajuste según la indicación
provista por su médico.

Pantalla de modo de flujo continuo

Pantalla de modo de dosis pulsada

ADVERTENCIA: ES MUY IMPORTANTE SELECCIONAR SOLO ELNIVEL DE OXÍGENO PRESCRITO. NO CAMBIE LA SELECCIÓN
DE FLUJO A MENOS QUE UN PROFESIONAL MÉDICO ACREDITADO ASÍ SE LO HAYA INDICADO. EL CONCENTRADOR DEOXÍGENO PORTÁTIL PUEDE USARSE AL DORMIR SI SE LO HA RECOMENDADO UN PROFESIONAL CUALIFICADO.
84 — 4805-C1 C (Spanish)

7

Funcionamiento del carrito para el eQuinox

Alinee el sistema eQuinox con la plataforma moldeada del carrito,
de modo que quede apoyado de forma plana y vertical. Alinee eltornillo roscado grande con la inserción de la parte posterior delsistema eQuinox. A continuación, ajuste manualmente la manijadel carrito hasta que el carrito esté firmemente sujeto al dispositivo.
Apriete el botón del asa telescópica del carrito y tire hacia arriba
para ajustar la altura del asa.

¡Importante!
Las instrucciones de seguridad se definen de la siguiente manera:
ADVERTENCIA: INFORMACIÓN DE SEGURIDAD IMPORTANTESOBRE PELIGROS QUE PUEDEN CAUSAR LESIONES GRAVES.
PRECAUCIÓN: Información importante sobre la prevención
de daños en el sistema eQuinox.

Nota: Información que requiere atención especial.

Indicaciones de uso
El sistema eQuinox está destinado para la administración deoxígeno complementario. El dispositivo no está destinado parasoporte vital ni tiene capacidad de monitoreo del paciente.
ADVERTENCIA: ES MUY IMPORTANTE SELECCIONAR SOLO ELNIVEL DE OXÍGENO PRESCRITO. NO CAMBIE LA SELECCIÓN
DE FLUJO A MENOS QUE UN PROFESIONAL MÉDICO ACREDITADO ASÍ SE LO HAYA INDICADO. EL CONCENTRADOR DEOXÍGENO PORTÁTIL PUEDE USARSE AL DORMIR SI SE LO HA RECOMENDADO UN PROFESIONAL CUALIFICADO.
ADVERTENCIA: LAS LEYES FEDERALES (DE LOS ESTADOS UNIDOS)
RESTRINGEN ESTE DISPOSITIVO A SU VENTA O ALQUILER POR PARTE DE
UN MÉDICO U OTRO PROVEEDOR DE ATENCIÓN MÉDICA ACREDITADO.
ADVERTENCIA: ESTA UNIDAD NO DEBE UTILIZARSE PARA SOPORTEVITAL. EN LOS PACIENTES GERIÁTRICOS, PEDIÁTRICOS O CUALQUIEROTRO PACIENTE INCAPAZ DE COMUNICAR SU INCOMODIDAD MIENTRASUSA ESTE DISPOSITIVO, PODRÍA SER NECESARIO UTILIZAR UN SISTEMA
DE VIGILANCIA ADICIONAL. LOS PACIENTES CON DISCAPACIDADES AUDITIVAS Y/O VISUALES PUEDEN NECESITAR AYUDA CON LAS ALARMAS
DE VIGILANCIA.

Directrices de seguridad
ADVERTENCIA: NO SE PERMITEN MODIFICACIONES A ESTEEQUIPO.
ADVERTENCIA: EL FABRICANTE RECOMIENDA DISPONER DE
UNA FUENTE ALTERNATIVA DE OXÍGENO COMPLEMENTARIOEN CASO DE CORTE ELÉCTRICO, CONDICIÓN DE ALARMA O
FALLA MECÁNICA. CONSULTE A SU MÉDICO O PROVEEDOR
DEL EQUIPO PARA DETERMINAR EL TIPO DE SISTEMA DE RESERVA QUE
NECESITA.
ADVERTENCIA: ESTE DISPOSITIVO SUMINISTRA OXÍGENO DE ALTA CONCENTRACIÓN, LO QUE LO HACE MUY INFLAMABLE. NO PERMITA QUE
SE FUME NI LA PRESENCIA DE LLAMAS VIVAS DENTRO DE LA MISMAHABITACIÓN EN LA QUE SE ENCUENTRA (1) ESTE DISPOSITIVO NI (2)NINGÚN ACCESORIO QUE CONTENGA OXÍGENO. SI NO SE TIENE EN
CUENTA ESTA ADVERTENCIA, PUEDE PRODUCIRSE UN INCENDIO GRAVE,
DAÑOS MATERIALES Y/O TENER COMO RESULTADO LESIONES FÍSICASO LA MUERTE.
ADVERTENCIA: NO OPERE LA UNIDAD EN UN ESPACIO RESTRINGIDO O CONFINADO (P. EJ., UNA CAJA O BOLSO PEQUEÑO) DONDE LA
VENTILACIÓN PUEDE SER LIMITADA. ESTO PUEDE OCASIONAR QUE EL
CONCENTRADOR DE OXÍGENO SE SOBRECALIENTE Y PERJUDIQUE EL
RENDIMIENTO.
ADVERTENCIA: EL CONCENTRADOR DEBE UBICARSE PARA EVITAR EL
HUMO, LOS CONTAMINANTES O LOS GASES TÓXICOS.
ADVERTENCIA: EL USO DE ALGUNOS ACCESORIOS PARA LA ADMINISTRACIÓN DE OXÍGENO NO ESPECIFICADOS PARA ESTE CONCENTRADORDE OXÍGENO PUEDE AFECTAR SU FUNCIONAMIENTO. LOS ACCESORIOSRECOMENDADOS SE MENCIONAN DENTRO DE ESTE MANUAL.
ADVERTENCIA: SI EL CONCENTRADOR DE OXÍGENO SE HA CAÍDO, HA
SUFRIDO DAÑOS O SE HA EXPUESTO AL AGUA, PÓNGASE EN CONTACTOCON EL PROVEEDOR DE ATENCIÓN DOMICILIARIA PARA SU INSPECCIÓN
O POSIBLE REPARACIÓN. NO USE EL CONCENTRADOR DE OXÍGENO SI
TIENE EL CABLE DE ALIMENTACIÓN O EL ENCHUFE DAÑADOS.
ADVERTENCIA: ASEGURE FIRMEMENTE, COLÓQUELE EL CINTURÓN
DE SEGURIDAD O AFIRME DE OTRA FORMA EL CONCENTRADOR DEOXÍGENO EN UN VEHÍCULO DURANTE SU TRANSPORTE PARA PREVENIRDAÑOS O LESIONES.
ADVERTENCIA: NO PERMITA QUE SE BLOQUEE LA ENTRADA DE AIRE
NI LAS ABERTURAS DE SALIDA DE AIRE. NO DEJE CAER NI INSERTEOBJETOS EN NINGUNA ABERTURA DEL DISPOSITIVO. ESTO PUEDE
OCASIONAR QUE EL CONCENTRADOR DE OXÍGENO SE SOBRECALIENTEY PERJUDIQUE EL RENDIMIENTO.
ADVERTENCIA: NO LLENE DEMASIADO EL HUMIDIFICADOR OPCIONAL.
LLENE EL HUMIDIFICADOR OPCIONAL CON AGUA SOLO HASTA EL NIVEL
INDICADO POR EL FABRICANTE DEL HUMIDIFICADOR.

Contraindicaciones de uso
ADVERTENCIA: EN ALGUNAS CIRCUNSTANCIAS, EL USO DE
OXÍGENO NO PRESCRITO PUEDE SER PELIGROSO. ESTEDISPOSITIVO SOLO DEBERÁ UTILIZARSE CUANDO LO
PRESCRIBA UN MÉDICO.
ADVERTENCIA: NO UTILIZAR EN PRESENCIA DE ANESTÉSICOS INFLAMABLES.
ADVERTENCIA: AL IGUAL QUE CON CUALQUIER OTRO DISPOSITIVOELÉCTRICO, EL USUARIO PODRÍA EXPERIMENTAR PERÍODOS DE NO
FUNCIONAMIENTO, DEBIDO A INTERRUPCIONES ELÉCTRICAS O A LA
NECESIDAD DE QUE EL TÉCNICO CUALIFICADO REVISE EL SISTEMAEQUINOX. EL SISTEMA EQUINOX NO ES ADECUADO PARA PACIENTESQUE PUEDAN SUFRIR CONSECUENCIAS DAÑINAS PARA LA SALUD A
CAUSA DE DICHAS INTERRUPCIONES TEMPORALES.

ADVERTENCIA: NO CAMINE Y UTILICE EL SISTEMA EQUINOX MIENTRASEL HUMIDIFICADOR ESTÉ ACOPLADO. QUITE LA BOTELLA HUMIDIFICADORA ANTES DE ANDAR. NO RECUESTE EL SISTEMA EQUINOX MIENTRAS LA BOTELLA HUMIDIFICADORA ESTÉ ACOPLADA.
ADVERTENCIA: LOS REGLAMENTOS DEL DEPARTAMENTO DE TRANSPORTE (DOT, POR SUS SIGLAS EN INGLÉS) ESTADOUNIDENSE Y DE LASNACIONES UNIDAS (ONU) REQUIEREN LA EXTRACCIÓN DE LA BATERÍA
DEL DISPOSITIVO PARA TODOS LOS VIAJES EN AEROLÍNEAS INTERNACIONALES CUANDO SE FACTURE EL CONCENTRADOR DE OXÍGENO
COMO EQUIPAJE. CUANDO SE ENVÍA EL CONCENTRADOR DE OXÍGENO,
LA BATERÍA TAMBIÉN DEBE EXTRAERSE DEL DISPOSITIVO Y EMBALARSEADECUADAMENTE.
ADVERTENCIA: NO UTILICE ESTE EQUIPO SIN LEER PRIMERO Y COMPRENDER ESTE MANUAL. SI NO COMPRENDE LAS ADVERTENCIAS E
INSTRUCCIONES, CONTACTE A SU PROVEEDOR DEL EQUIPO ANTES DE
INTENTAR USAR EL EQUIPO; DE LO CONTRARIO, PODRÍAN PRODUCIRSELESIONES O DAÑOS.

eQuinox (Model 4000) User Manual — 85

ADVERTENCIA: SI SIENTE MOLESTIAS O SUFRE UNA
EMERGENCIA MÉDICA, BUSQUE ASISTENCIA MÉDICA DE
INMEDIATO.
ADVERTENCIA: LA OPERACIÓN DEL CONCENTRADOR DEOXÍGENO FUERA DE LAS ESPECIFICACIONES DE LA TEMPERATURA DEFUNCIONAMIENTO PUEDE LIMITAR LA CAPACIDAD DEL CONCENTRADOR DE CUMPLIR CON LAS ESPECIFICACIONES DE CONCENTRACIÓNDE OXÍGENO. CONSULTE LA SECCIÓN DE ESPECIFICACIONES DE ESTEMANUAL PARA CONOCER LOS LÍMITES DE TEMPERATURA.
ADVERTENCIA: NO USE ACEITE, GRASA, PRODUCTOS CON BASE DEPETRÓLEO NI OTROS PRODUCTOS INFLAMABLES CON LOS ACCESORIOSDE TRANSPORTE DE OXÍGENO O CON EL CONCENTRADOR DE OXÍGENO.
SOLO DEBEN USARSE LOCIONES O BÁLSAMOS COMPATIBLES CON ELOXÍGENO, CON BASE DE AGUA. EL OXÍGENO ACELERA LA COMBUSTIÓN
DE LAS SUSTANCIAS INFLAMABLES.
ADVERTENCIA: EL USO INCORRECTO DE LA BATERÍA PUEDE OCASIONARQUE ESTA SE CALIENTE O SE ENCIENDA, Y PUEDE CAUSAR LESIONESSERIAS. ASEGÚRESE DE NO PERFORAR, GOLPEAR, PISAR, DEJAR CAER
O SOMETER DE OTRA FORMA LA BATERÍA A IMPACTOS O GOLPESFUERTES. EL USO DE BATERÍAS QUE ESTÉN DAÑADAS PODRÍA CAUSARLESIONES FÍSICAS.
ADVERTENCIA: EL CONCENTRADOR DE OXÍGENO NO DEBE USARSEADYACENTE A O APILADO CON OTROS EQUIPOS. SI EL USO ADYACENTEO APILADO ES INEVITABLE, EL DISPOSITIVO DEBE OBSERVARSE PARAVERIFICAR SU FUNCIONAMIENTO NORMAL.
PRECAUCIÓN: En el caso de que se produzca una alarma o si
observa que el concentrador de oxígeno no funciona
debidamente, consulte la sección de solución de problemasde este manual. Si no puede resolver el problema, consulte
con su proveedor del equipo.

ADVERTENCIA: FUMAR MIENTRAS SE USA OXÍGENO ES LA
CAUSA PRINCIPAL DE LESIONES POR INCENDIOS Y MUERTESRELACIONADAS. DEBE SEGUIR LAS ADVERTENCIAS DE
SEGURIDAD:
ADVERTENCIA: NO DEJE CIGARRILLOS ENCENDIDOS, VELAS O LLAMASVIVAS EN LA MISMA HABITACIÓN EN QUE SE ENCUENTRA EL DISPOSITIVO O LOS ACCESORIOS QUE TRANSPORTAN OXÍGENO.
ADVERTENCIA: FUMAR MIENTRAS SE USA UNA CÁNULA DE OXÍGENO
PUEDE CAUSAR QUEMADURAS FACIALES Y POSIBLEMENTE LA MUERTE.
ADVERTENCIA: SI FUMA, ESTOS 3 PASOS PUEDEN SALVAR SU VIDA:
APAGUE EL CONCENTRADOR DE OXÍGENO, RETIRE LA CÁNULA Y SALGADE LA HABITACIÓN DONDE SE UBICA EL DISPOSITIVO.
ADVERTENCIA: DEBEN EXHIBIRSE LETREROS DE "NO FUMAR: OXÍGENO
EN USO" EN EL HOGAR, O EN EL SITIO DONDE SE USE EL OXÍGENO. LOSPACIENTES Y LOS CUIDADORES SANITARIOS DEBEN RECIBIR INFORMACIÓN ACERCA DE LOS PELIGROS DE FUMAR AL USAR OXÍGENO
MEDICINAL O EN PRESENCIA DE ESTE.
ADVERTENCIA: NO USE SU CONCENTRADOR DE OXÍGENO EN PRESENCIA DE GASES INFLAMABLES. ESTO PUEDE RESULTAR EN UN FUEGORÁPIDO QUE PUEDE CAUSAR DAÑOS MATERIALES, LESIONES CORPORALES O LA MUERTE.
ADVERTENCIA: RETIRAR LA CÁNULA Y COLOCARLA SOBRE LA ROPA,
CAMA, SOFÁ U OTRO MATERIAL ACOLCHADO CAUSARÁ UNA LLAMARADA SI SE LA EXPONE A UN CIGARRILLO, FUENTE DE CALOR O LLAMA.
ADVERTENCIA: NO DEJE UNA CÁNULA NASAL SOBRE O DEBAJO DE LA
ROPA, DE LOS COBERTORES DE LA CAMA O ALMOHADONES DE UNASILLA. SI LA UNIDAD ESTÁ ENCENDIDA PERO NO EN USO, EL OXÍGENO PODRÍA VOLVER INFLAMABLES ESOS MATERIALES. COLOQUE EL
CONCENTRADOR DE OXÍGENO EN LA POSICIÓN 0 (APAGADO) CUANDONO ESTÉ EN USO.

PRECAUCIÓN: Para evitar la anulación de la garantía, siga todas las
instrucciones del fabricante.
ADVERTENCIA: NO INTENTE REALIZAR TAREAS DE MANTENIMIENTO DIFERENTES DE LAS POSIBLES SOLUCIONESENUMERADAS EN ESTE MANUAL. NO RETIRE LAS CUBIERTAS, SOLO SU PROVEEDOR DEL EQUIPO O UN TÉCNICO DESERVICIO CUALIFICADO DEBEN REVISAR O RETIRAR LAS CUBIERTAS DE
ESTE DISPOSITIVO.
ADVERTENCIA: USE SOLO ACCESORIOS RECOMENDADOS POR EL FABRICANTE. EL USO DE OTRO ACCESORIO PUEDE SER PELIGROSO, CAUSARÁ
DAÑOS GRAVES A SU CONCENTRADOR DE OXÍGENO Y ANULARÁ LA
GARANTÍA.
ADVERTENCIA: EL FABRICANTE RECOMIENDA DISPONER DE
UNA FUENTE ALTERNATIVA DE OXÍGENO COMPLEMENTARIOEN CASO DE CORTE ELÉCTRICO, CONDICIÓN DE ALARMAO FALLA MECÁNICA. CONSULTE A SU MÉDICO O PROVEEDOR
DEL EQUIPO PARA DETERMINAR EL TIPO DE SISTEMA DE RESERVA QUE
NECESITA.
ADVERTENCIA: NO PERMITA QUE SE BLOQUEE LA ENTRADA DE AIRE
NI LAS ABERTURAS DE SALIDA DE AIRE. NO DEJE CAER NI INSERTEOBJETOS EN NINGUNA ABERTURA DEL DISPOSITIVO. ESTO PUEDE
OCASIONAR QUE EL CONCENTRADOR DE OXÍGENO SE SOBRECALIENTEY PERJUDIQUE EL RENDIMIENTO.

86 — 4805-C1 C (Spanish)

PRECAUCIÓN: Siempre desconecte la fuente de alimentaciónde CA de la pared antes de desconectar la fuente dealimentación de CA del concentrador de oxígeno.
PRECAUCIÓN: Siempre coloque el tubo de suministro de oxígeno y los
cables de corriente de forma tal que evite el riesgo de tropiezos.
PRECAUCIÓN: Al usar el concentrador de oxígeno en un automóvil, unbote o en otras fuentes de CC con el cable de 12 VCC, asegúrese deque el vehículo esté funcionando antes de conectar el concentradorde oxígeno. Si la luz de alimentación externa del sistema eQuinox nose ilumina y requiere un reajuste, desconecte el cable de 12 VCC de latoma de CC, vuelva a iniciar su vehículo y luego reconecte su cable de
12 VCC a la toma de salida de CC. Si no se siguen estas instrucciones, lafuente de alimentación podría no suministrar energía al concentradorde oxígeno.
PRECAUCIÓN: Cuando el automóvil en el que se usa la unidad del concentrador de oxígeno se apague, desconecte y retire el dispositivo delautomóvil. No almacene el concentrador de oxígeno en un automóvil
muy caliente o muy frío o en otro ambiente similar de alta o baja
temperatura.

Nota: Los equipos de comunicaciones portátiles y móviles de radiofrecuencia pueden afectar los equipos eléctricos médicos.

Entornos de operación recomendadosPara el uso correcto del dispositivo, consulte el siguiente cuadro.
El cuadro presenta información importante sobre los entornos
operativos recomendados o las condiciones operativas.

Especificaciones
Temperatura defuncionamiento

De 0 °C a 40 °C (de 32 °F a 104 °F)

Humedad defuncionamiento

Del 10 % al 95 % en un punto de condensación de 28 °C(82,4 °F)

Temperatura de -25 °C a 70 °C (-13 °F a 158 °F)
transporte/almaHumedad: Hasta el 95 % sin condensación
cenamiento-20 °C a 50 °C (-4 °F a 122 °F) para paquetes de batería
Especificaciones
eléctricas
Altitud

Noutilicecablesdeextensión.Noutilicetomaseléctricas
controladas por un interruptor.
De 0 a 4000 metros (de 0 a 13 123 pies)

Colocación

NO bloquee la entrada de aire (cubierta de la parte superior trasera) ni las aberturas de escape (parte inferior
izquierda de la parte trasera). Coloque el dispositivo a
una distancia mínima de 7,5 cm (3 pulg.) de paredes,
cortinas, muebles, etc.

Ambiente

No debe haber humo, contaminantes ni gases.

Tiempo de funcionamiento

24 horas al día cuando está conectado a una fuente dealimentación de CA o CC.

Nota: Si el concentrador de oxígeno se ha almacenado por un período
prolongadofueradesuintervalodetemperaturadefuncionamientonormal,
debe permitirse que la unidad vuelva a la temperatura de funcionamiento
normalantesdeencenderla.(Consultelaseccióndeespecificacioneseneste
manual).

Colocación adecuada del sistema eQuinox
Seleccione una ubicación para el dispositivo en la que no hayaentrada de humo, gases ni contaminantes. La colocación correctadel dispositivo debería permitir la entrada de aire a través del filtro
de entrada de aire en la parte superior trasera del gabinete. También
debería permitir que el aire de escape deje libremente las aberturasde escape en la parte inferior derecha del dispositivo.
Coloque el dispositivo de forma que puedan oírse las alarmas.
Coloque el tubo de suministro de oxígeno de tal forma que no securve ni ocluya.
ADVERTENCIA: NO USE SU CONCENTRADOR DE OXÍGENO EN
PRESENCIA DE GASES INFLAMABLES. ESTO PUEDE RESULTAREN UN FUEGO RÁPIDO QUE PUEDE CAUSAR DAÑOS
MATERIALES, LESIONES CORPORALES O LA MUERTE. NO USE
ACEITE, GRASA, PRODUCTOS CON BASE DE PETRÓLEO NI OTROSPRODUCTOS INFLAMABLES CON LOS ACCESORIOS DE TRANSPORTE DEOXÍGENO O CON EL CONCENTRADOR DE OXÍGENO. SOLO DEBENUSARSE LOCIONES O BÁLSAMOS COMPATIBLES CON EL OXÍGENO, CON
BASE DE AGUA. EL OXÍGENO ACELERA LA COMBUSTIÓN DE LASSUSTANCIAS INFLAMABLES.
ADVERTENCIA: ESTE DISPOSITIVO SUMINISTRA OXÍGENO DE ALTA CONCENTRACIÓN, LO QUE LO HACE MUY INFLAMABLE. NO PERMITA QUE
SE FUME NI LA PRESENCIA DE LLAMAS VIVAS DENTRO DE LA MISMAHABITACIÓN EN LA QUE SE ENCUENTRA (1) ESTE DISPOSITIVO NI (2)NINGÚN ACCESORIO QUE CONTENGA OXÍGENO. SI NO SE TIENE EN
CUENTA ESTA ADVERTENCIA, PUEDE PRODUCIRSE UN INCENDIO GRAVE,
DAÑOS MATERIALES Y/O TENER COMO RESULTADO LESIONES FÍSICASO LA MUERTE.
Nota: Cuando el automóvil en el que se usa la unidad del concentrador deoxígenoseapague,desconecteyretireeldispositivodelautomóvil.Noalmacene el concentrador de oxígeno en un automóvil muy caliente o muy frío oen otro ambiente similar de alta o baja temperatura. NO deje el concentradorde oxígeno ni la fuente de alimentación enchufados en el vehículo si seencuentra apagado. De lo contrario, podría agotar la batería del vehículo.

PRECAUCIÓN: No opere la unidad en un espacio restringido
o confinado (p. ej., una caja o bolso pequeño) donde laventilación puede ser limitada. Esto puede ocasionar que elconcentrador de oxígeno se sobrecaliente y perjudique el rendimiento.
No permita que se bloquee la entrada de aire ni las aberturas de salidade aire. Esto puede ocasionar que el concentrador de oxígeno sesobrecaliente y perjudique el rendimiento.

eQuinox (Model 4000) User Manual — 87

Instrucciones de funcionamiento
Antes de utilizar

Información general de carga

Este manual del usuario sirve como referencia para ayudarle a usar
el dispositivo y llevar a cabo su mantenimiento. Si tiene alguna
pregunta o duda, llame al proveedor de atención domiciliaria.

El sistema eQuinox funciona con alimentación de CA, alimentaciónde CC o una batería recargable. El sistema incluye un "cargador debatería inteligente". Este recarga la batería cada vez que el sistema
eQuinox se conecta a una fuente de alimentación externa. El sistema
supervisa y controla la fuente de alimentación y el cargador de labatería.

¡Importante! NO intente poner en funcionamiento el sistema
eQuinox sin leer primero la sección Directrices de seguridad de este
manual. Siga todas las instrucciones de funcionamiento. Observe
todas las Advertencias del dispositivo y del Manual del usuario.
Para reducir el riesgo de incendios, lesiones físicas y daños graves
en el sistema eQuinox, preste atención a todas las precauciones deseguridad.
ADVERTENCIA: LIMPIE EL GABINETE, EL PANEL DE CONTROLY EL CABLE DE ALIMENTACIÓN SOLO CON UN LIMPIADORDOMÉSTICO APLICADO CON UN PAÑO HÚMEDO (NOMOJADO) O ESPONJA, Y LUEGO SEQUE TODAS LAS SUPERFICIES. NO PERMITA QUE NINGÚN LÍQUIDO ENTRE EN EL DISPOSITIVO.
PRESTE ESPECIAL ATENCIÓN A LA SALIDA DE OXÍGENO PARA LACONEXIÓN DE LA CÁNULA PARA ASEGURARSE DE QUE PERMANEZCA
LIBRE DE POLVO, AGUA Y PARTÍCULAS.
PRECAUCIÓN: No permita que se bloquee la entrada de aireni las aberturas de salida de aire. Esto puede ocasionar que
el concentrador de oxígeno se sobrecaliente y perjudiqueel rendimiento. No opere el concentrador de oxígeno sin el
filtro de entrada de aire colocado. Si dispone de un segundo filtro, coloque este filtro de "reemplazo" antes de limpiar el filtro sucio. Limpie
el filtro sucio con una solución tibia de agua y jabón, y luego seque porcompleto antes de usar.

El sistema eQuinox carga la batería mientras se utiliza la fuente dealimentación de CA o CC y la temperatura de la batería se encuentra
en un intervalo seguro. Cuando la fuente de alimentación externa
se desconecta, el dispositivo pasa automáticamente al uso conbatería, si esta está instalada y cargada. Cuando vuelve a conectarse
a una salida de CA o CC, el dispositivo funciona con la fuente dealimentación externa y recarga la batería.
Alinee el enchufe de alimentación con el toma de corriente externapara conectarlo correctamente a la fuente de CA o al cable de
12 VCC. El toma de corriente empotrado se encuentra en el lado
derecho del dispositivo al mirar de frente el panel de control.
Introduzca el enchufe en el toma de corriente empotrado.

FUENTE DE ALIMENTACIÓN DE CA

Fuente de alimentación de CA

Encendido del sistema eQuinox
El dispositivo puede operarse con tres (3) fuentes de alimentación
diferentes:• Fuente de alimentación de CA• Cable de 12 VCC• Paquete de baterías recargable (batería)

El sistema eQuinox incluye una fuente de alimentación deCA universal para su uso en el hogar o donde esté disponible
la alimentación de CA estándar. Para conectar a una fuentede alimentación de alimentación de CA, enchufe el cable dealimentación de CA de forma segura en el toma de corriente
externa. También asegúrese de que el cable de la fuente dealimentación de CA esté enchufado a una salida de CA. Cuando
el sistema eQuinox sea enchufado correctamente, se iluminarán elindicador verde en la fuente de alimentación de CA y el indicador
de alimentación externa. El indicador
en el panel de control
del usuario también se iluminará. La fuente de alimentación de
CA CAIRE PN 4836 es la fuente de alimentación principal parael concentrador de oxígeno portátil eQuinox. Es una parte muy
importante del sistema en general.
Nota:Useúnicamenteelvoltajeeléctricoindicadoenlaetiqueta deespecificaciones adherida al dispositivo.
Paso 1

88 — 4805-C1 C (Spanish)

Paso 2

Paso 3

ADVERTENCIA: NO TOQUE LOS CONTACTOS DE LA BATERÍA
Y AL PACIENTE AL MISMO TIEMPO CUANDO ENERGICE LA
UNIDAD CON LA FUENTE DE ALIMENTACIÓN EXTERNA(FUENTE DE ALIMENTACIÓN DE CA) Y SIN NINGÚN PAQUETE
DE BATERÍAS INSTALADO.
PRECAUCIÓN: Use solo accesorios recomendados por el
fabricante. El uso de cualquier otra fuente de alimentación
puede ser peligroso, provocar daños graves en el concentrador de oxígeno y anulará la garantía.

Conexión de la fuente de CC de eQuinox
Cuando el sistema eQuinox se conecta a una fuente de CC en un
vehículo, se cumplen las siguientes condiciones:• El sistema eQuinox funcionará con todos los ajustes de flujo
continuos de 0,5–3,0 LPM y todos los ajustes de pulso de
1–9 mientras funciona con la fuente de CC en un automóvil.

PRECAUCIÓN: No utilice cables de extensión en esta unidad ni conecte
demasiados enchufes a la misma toma eléctrica. El uso de cables de extensión podría afectar de forma adversa el rendimiento del dispositivo.
Demasiados enchufes en una toma pueden ocasionar una sobrecarga al
panel eléctrico y hacer que el disyuntor/fusible se active o dispare si eldisyuntor o el fusible no funciona.

• La batería solo se cargará si queda suficiente alimentación
después de que el dispositivo funcione con la frecuencia deflujo indicada. El sistema eQuinox aplicará la alimentación
suministrada para hacer funcionar la unidad primero. La
alimentación que quede se utilizará como una carga decompensación para recargar la batería. Si se carga la batería delsistema eQuinox, el símbolo de la batería parpadeará. Si la bateríadel sistema eQuinox no carga, el símbolo de la batería se mostrarápermanentemente encendido.

PRECAUCIÓN: El uso de cables y adaptadores diferentes a los especificados, con excepción de los cables y de los adaptadores vendidos por
el fabricante del equipo médico eléctrico como piezas de repuesto para
componentes internos, puede resultar en una mayor emisión o una
menor inmunidad del concentrador de oxígeno.

ADVERTENCIA: NO PERMITA QUE SE BLOQUEE LA ENTRADA
DE AIRE NI LAS ABERTURAS DE SALIDA DE AIRE. ESTO PUEDE
OCASIONAR QUE EL CONCENTRADOR DE OXÍGENO SE
SOBRECALIENTE Y PERJUDIQUE EL RENDIMIENTO.

Cable de 12 VCC
Cable de 12 VCC

ADVERTENCIA: SI LA FUENTE DE ALIMENTACIÓN DE CC DEL VEHÍCULO
CAE POR DEBAJO DE LOS 11,5 VOLTIOS, EL SISTEMA EQUINOX PASARÁAL FUNCIONAMIENTO CON BATERÍA (SI ESTA ESTÁ INSTALADA Y CARGADA).
PRECAUCIÓN: Cuando el automóvil en el que se usa la
unidad del concentrador de oxígeno se apague, desconecte
y retire el dispositivo del automóvil. No almacene elconcentrador de oxígeno en un automóvil muy caliente o
muy frío o en otro ambiente similar de alta o baja temperatura. NO
deje el concentrador de oxígeno o la fuente de alimentación enchufados en el vehículo si se encuentra apagado. De lo contrario, podría
agotar la batería del vehículo.

Una fuente de alimentación de 12 VCC permite el funcionamientodel sistema desde tomas de CC, como las de los vehículos
motorizados.
1. Arranque el vehículo.
2. Introduzca el cable de 12 VCC en la toma de corriente dellateral del dispositivo.
3. Introduzca el enchufe del cable de 12 VCC en la toma dealimentación de CC del vehículo motorizado.
Cuando el dispositivo esté correctamente conectado y reciba energía
de la fuente de alimentación de CC, se iluminará un indicador verdede energía externa
en el panel de control.

Nota: Mientras utilice el dispositivo con un cable de 12VCC, la batería puedeno cargarse.
Nota: Cuando utilice el sistema eQuinox con un cable de 12 VCC, serecomiendaelusodelmododedosispulsada,siempreque sehaya prescrito
el uso de un dispositivo de dosis pulsada.
Nota: Para asegurarse de que el dispositivo está utilizando la alimentación
deCC,compruebequeelindicadordealimentaciónexternaestéiluminado.
Si oye un pitido, puede ser que exista una pérdida de alimentación externa.
Compruebe si la LED verde del indicador de alimentación externa está encendidaoapagada.Simientraslaunidadestáfuncionandoconalimentaciónde CC se quita la batería, esto asegurará que no se produzca una pérdidade carga. Si el indicador no está iluminado, entonces el dispositivo está
funcionando con la batería.
Nota: El rendimiento real del sistema eQuinox puede variar en función de lascondicionesdeusodelvehículomotorizado.EsposiblequealgunosvehículosnopuedanadmitirlafuncionalidadcompletaconalimentacióndeCC.En
elcasodequeunvehículonopuedasuministrarsuficientealimentaciónpara
funcionar con el ajuste actual, el sistema eQuinox primero dejará de cargar
la batería y seguirá funcionando con la frecuencia de flujo indicada. Si elvehículoaúnnopuedesuministrarsuficienteenergía,elsistemaeQuinoxreducirá la frecuencia de flujo si esta es de 2,0 LPM o superior. Si la frecuencia
de flujo es inferior a 2,0 LPM, el sistema eQuinox se apagará.
Si se perciben frecuencias de flujo bajas, quite todos los demás dispositivos electrónicos conectados al circuito de alimentación (GPS, cargador deteléfono celular, etc.). Si se recupera alimentación suficiente, las frecuencias
de flujo no aumentarán automáticamente al ajuste original; el usuario debe
hacer el ajuste de forma manual. Si el sistema eQuinox aún no funciona conla frecuencia de flujo indicada, intente quitar el paquete de baterías.
eQuinox (Model 4000) User Manual — 89

Paquete de baterías recargable (batería)
El sistema eQuinox puede funcionar con elpaquete de baterías (batería) recargable que sesuministra con el dispositivo.
Instalación de la batería: Alinee la batería paraque se deslice hacia el interior del compartimentode la alimentación que está vacío. Empuje labatería hacia el interior del compartimentode alimentación hasta que se trabe (se ajuste)
en su posición. Quedará nivelada con la partedelantera del dispositivo. Cuando la batería estécorrectamente instalada y el sistema eQuinox esté
encendido, el indicador de estado de la batería
aparecerá en el panel de control. (Si la batería NO
está instalada, el indicador de estado del paquetede baterías se encenderá). La unidad seguirá
funcionando con una fuente de alimentación
Paquete de baterías
externa.
recargable (batería)
Para quitar la batería, presione el botón de desbloqueo en la partesuperior del paquete de baterías. Mantenga la palanca de desbloqueo
hacia ABAJO y tire de la batería para extraerla del sistema eQuinox.
Tiempo de funcionamiento de la batería: El tiempo defuncionamiento de la batería se muestra en el panel de control,
en los formatos simbólico y alfanumérico. Cuando el dispositivo
funcione con alimentación de la batería, el indicador de estado dela batería se mostrará todo el tiempo. La pantalla programará las
horas/minutos del tiempo restante de la batería. Distintos factores,
como el ajuste de flujo, el modo de flujo continuo o de dosis pulsada
y la frecuencia respiratoria, afectarán el tiempo de funcionamientode la batería. La tabla Tiempo de funcionamiento típico del nuevo
paquete de baterías ofrece cálculos del tiempo de funcionamiento
estimativo del sistema eQuinox, para cuando utilice una batería
nueva con carga completa de acuerdo con el ajuste de flujo y lascondiciones de funcionamiento indicadas.
Nota: El sistema eQuinox viene con una batería de 12 celdas cuando se envía
desde fábrica. La batería de 24 celdas es un accesorio opcional para una
duración más prolongada de la batería.

Nota: Si el concentrador de oxígeno se opera fuera de su intervalo de temperatura de funcionamiento normal, se puede ver afectado el rendimiento y
reducirseeltiempodefuncionamientodelabateríay/oaumentareltiempodecargadelabatería.(ConsultelasecciónEspecificacionesenestemanual).Obtendráunmayortiempodefuncionamientoconlabatería,siutilizaelsistema
eQuinox con el modo de dosis pulsada (modo aprobado por el médico).
ADVERTENCIA: PARA CONCENTRADORES DE OXÍGENO
EQUIPADOS CON BATERÍAS: PARA GARANTIZAR LA DURACIÓN DE LA VIDA DE LA BATERÍA, MANTÉNGALA ALMACENADA EN UN LUGAR FRESCO Y SECO. GUARDAR EL CONCENTRADOR DE OXÍGENO DURANTE UN LAPSO PROLONGADO A ALTASTEMPERATURAS O CON UNA BATERÍA TOTALMENTE CARGADA/DESCARGADA PUEDE REDUCIR LA VIDA ÚTIL GENERAL DE LA BATERÍA.
NO INTENTE ABRIR LA BATERÍA, YA QUE NO DISPONE DE PIEZAS QUEPUEDAN REPARARSE EN SU INTERIOR. MANTENGA LAS BATERÍAS LEJOS
DE LOS NIÑOS.

Carga inicial de la batería
La batería nueva suministrada con el sistema eQuinox no estácargada completamente cuando se envía de fábrica. Antes de utilizar
el sistema eQuinox la primera vez, primero debe cargar la bateríacompletamente.
Conecte la fuente de alimentación de CA e instale la batería
correctamente. Permita que la batería se cargue completamente.
La batería está cargada completamente cuando el indicador deestado del paquete de baterías en el panel de control deja deparpadear en forma continua y queda permanentemente encendido.
El sistema eQuinox puede usarse mientras la batería se recarga.
Puede tardar hasta 4 horas para cargarse, dependiendo del ajuste delflujo. La batería también se cargará al enchufar el sistema eQuinox
a una toma de CA y al apagarlo.
PRECAUCIÓN: Solo use las baterías provistas por el fabricante. Para la eliminación correcta de la batería, contacte a su
proveedor del equipo o a su organismo gubernamental local
a fin de conocer los requisitos de eliminación.

* Las baterías son reciclables. *Tiempo de funcionamiento típico del nuevo paquete de baterías
Tipo de funcionamiento

Tipo de bateríade 12 celdas

de 24 celdas

1 LMP

2,16 h (2 h, 9 min)

4,84 h (4 h, 50 min)

2 LPM

1,2 horas (1 hora, 12 min)

2,75 h (2 h, 45 min)

3 LPM

0,8 h (48 min)

1,83 h (1 h, 49 min)

16 ml (1,0)

2,62 h (2 h, 37 min)

5,94 h (5 h, 56 min)

48 ml (3,0)

2,38 h (2 h, 22 min)

5,39 h (5 h, 23 min)

96 ml (6,0)

1,70 h (1 h, 41 min)3,85 h (3 h, 51 min)Modo
continuo

Modo de dosis pulsada (a 12 bpm)

90 — 4805-C1 C (Spanish)

En caso de que se produzca una interrupción de la alimentaciónde CA o de CC, el sistema eQuinox emitirá un pitido y pasará
automáticamente al funcionamiento con batería, siempre que esta
esté instalada y cargada. Cuando se devuelve la alimentación
externa, la batería comenzará a cargarse nuevamente de forma
automática. Si falta la batería o si esta está completamente
descargada cuando se interrumpe la alimentación de CA, el sistema
eQuinox se apagará y emitirá una alarma. La batería está descargada
completamente cuando el indicador de estado del paquete debaterías en el panel de control aparece vacío. El dispositivo
mostrará una luz amarilla y emitirá una alarma de batería baja. Esto
continuará hasta que el dispositivo se apague por completo con una
luz roja y una alarma constante. La alerta sonora puede continuar
por un máximo de 5 minutos hasta que el dispositivo se apague o serestituya la alimentación externa.
Nota: Mantenga pulsado el botón de ENCENDIDO/APAGADO del sistema
eQuinox durante un mínimo de 5 segundos para silenciar la alarma de pérdidadeenergía.LaconexióndelsistemaeQuinoxaunafuentedealimentaciónde CA externa durante 2 minutos también eliminará esta alarma.

Tiempo típico de recarga de la batería

Activación/desactivación del módulo de voz

Por lo general, se tarda de 2 a 5 horas para recargar una bateríacompletamente agotada a un 80 % de su capacidad, según el ajustede flujo que se utilice. Cuando la batería esté completamente
cargada, el indicador de la batería dejará de parpadear en forma
continua. La batería puede cargarse en la unidad cuando el sistema
eQuinox esté apagado y enchufado.
Si la batería se calienta demasiado durante la descarga, la recarga no
se iniciará hasta que la batería se haya enfriado lo suficiente. Quitar
la batería puede acelerar el proceso de enfriamiento.
La batería puede recargarse a partir de fuentes de alimentaciónde CA o CC. La batería también se cargará desde el cargador de
escritorio opcional.

1. Para activar/desactivar la función del módulo de voz, encienda
la unidad presionando la tecla de ENCENDIDO/Espera.
2. Después de que se encienda la unidad, presione el ícono deProhibido fumar y la tecla Ocultar una vez.
3. En 1 o 2 segundos, aparecerá el elemento del menú SPKR =ON (ALTAVOZ = ENCENDIDO) o SPKR = OFF (ALTAVOZ=APAGADO).
4. Presione la tecla de aumento o reducción de la frecuencia deflujo para cambiar la selección a ENCENDIDO [ON] (activar
el módulo de voz) o APAGADO [OFF] (desactivar el módulo
de voz).
5. En un tiempo de 3 a 5 segundos, se activará la opción
seleccionada.
Nota: Esto solo desactivará la función del módulo de voz una vez. Cuando eldispositivo se apague y se reinicie, la función del módulo de voz se activará
automáticamente.

Aumento/disminución del volumen de las notificaciones vocales
1. Para subir o bajar el volumen de las notificaciones vocales,presione el ícono de Prohibido fumar y la tecla Ocultar dos veces.

Funcionamiento del módulo de voz
Instrucciones operativas del módulo de voz
El sistema eQuinox puede suministrar notificaciones verbaleso alarmas y el estado operativo. Estas notificaciones verbales
complementan la información en la pantalla de diodos emisores
de luz (LED). La siguiente información proporciona instrucciones
sobre cómo encender y apagar la función del módulo de voz, ajustar
el volumen y cambiar el idioma.
Tecla deENCENDIDO/
EsperaIndicador de ENIndicador de alarCENDIDO/Espera Indicador dema (LED rojo)
alerta (LED am(LED verde)Tecla de frecuenarillo)
cia de aumentode flujo

2. Aparecerá Volume = # - (Menú de ajuste del volumen).
3. Presione la tecla de aumento o reducción de la tasa de flujo
para subir o bajar el nivel del volumen de 1 a 10.
4. Después de seleccionar el nivel de volumen deseado, la opción
se activará/ajustará después de 3 a 5 segundos.
Cambiar el idioma del módulo de voz
1. Para cambiar el idioma del módulo de voz, presione el ícono
de Prohibido fumar y la tecla Ocultar cinco veces para acceder alas opciones de idioma.
2. El idioma activado actualmente estará indicado en la pantalla.
Por ejemplo, si el módulo de voz está en idioma inglés cuando se
accede al menú de opciones de idioma, en la pantalla aparece eltexto "English" (inglés).
3. Presione la tecla de aumento o reducción de la frecuencia deflujo para desplazarse por los idiomas disponibles.
4. Cuado aparezca el idioma deseado, este se ajustaráautomáticamente en un tiempo de 3 a 5 segundos.

Indicador depresencia dealimentación
externa(LED verde)Ícono de Prohibido fumar ytecla Ocultar

Tecla de modode flujo

Indicador demodo de flujo(LED verde)

Pantalla
LCD

Tecla de frecuencia de reducciónde flujo

eQuinox (Model 4000) User Manual — 91

Operación del sistema eQuinox por primera vez
Paso 1: Colocación del sistema eQuinox para su uso
Coloque el sistema eQuinox en una zona iluminada y con buena
ventilación. Asegúrese de que la entrada de aire y las aberturas deventilación en la parte trasera no estén bloqueadas.
Coloque el sistema eQuinox demodo que los indicadores visuales
y sonoros puedan verse y oírse
fácilmente.

Filtro de entrada de aire

Asegúrese de que el filtro de entrada
de aire esté bien colocado antes deutilizar el sistema eQuinox. Si no
se incluye el filtro de entrada deaire, póngase en contacto con su
proveedor de atención domiciliaria.
Si el filtro de entrada de aireestá sucio, lávelo con agua tibia
jabonosa. Luego, enjuáguelo con
agua limpia y déjelo secar antes decolocar el filtro de entrada de aireen el dispositivo. Si fuera necesario,
sustitúyalo por uno nuevo y limpio.
Enchufe el dispositivo en una
Aberturas de escapetoma de CA o en una fuente dealimentación de CC, o asegúrese de que haya instalada una bateríacompletamente cargada.

Paso 2: Encendido del dispositivo y su preparación en reposo
Mantenga pulsado el botón de "ENCENDIDO/APAGADO" durante
dos (2) segundos para encender el sistema eQuinox. Se producirá
una breve prueba automática visual.
Nota: Si se producen tres pitidos cuando el dispositivo se conecta primeroa la alimentación o es activado, esto indica una condición de nivel bajo de labatería para la batería de 9 voltios que está dentro del gabinete del sistema
eQuinox. Póngase en contacto con su proveedor de atención domiciliaria ypídale que sustituya la batería de 9 voltios. Si los tres pitidos continúan intermitentementedespuésdequelaunidadhasidoactivadayhafuncionadoporvarios minutos, no se trata de una condición de nivel bajo de la batería de 9 V.
Comuníquese con su proveedor de atención médica para obtener el servicio.

El sistema eQuinox posee un indicador de estado de laconcentración de oxígeno (OCSI, del inglés oxygen concentration
status indicator) integrado en el dispositivo. El OCSI continuamente
supervisa la salida de oxígeno del dispositivo. Utiliza una luz verde(normal), una luz amarilla (alerta), una luz roja (alarma) y una
alarma sonora. Estos indicadores se iluminarán durante la prueba
automática de encendido al iniciarse el sistema. Tras el encendido
inicial del dispositivo, deje un período de cinco (5) minutos paraque el dispositivo alcance sus especificaciones de rendimiento. Solocuando se encienda la luz verde del indicador, el dispositivo habrá
alcanzado las especificaciones de rendimiento adecuadas.

92 — 4805-C1 C (Spanish)

Paso 3: Conexión del tubo de suministro de oxígeno
o la cánula nasal a la salida de oxígeno
Reemplace la cánula y el tubo de suministro de oxígeno con
regularidad, como recomiende su proveedor de atención domiciliaria.
Solicite al proveedor de atención domiciliaria información sobre lalimpieza y la sustitución de la cánula y del tubo de suministro.
Nota: Asegúrese de que la cánula esté bien insertada y esté firme. Esto
aseguraqueelconcentradordeoxígenopuedadetectarapropiadamentela
inspiraciónparalaentregadeoxígeno.Durantelainhalación,debeescucharo sentir el flujo de oxígeno hacia el extremo de la cánula nasal.

Paso 4: Selección del modo de flujo
Presione el botón del modo de flujo para seleccionar
el modo. Al pulsar este botón más de una vez, alternará entre elmodo de dosis pulsada y el modo de flujo continuo. El médico leprescribirá el modo y el ajuste adecuados.
Funcionamiento del modo de flujo continuo:
Cuado el dispositivo funcione en el modo de flujo continuo, unsuministro constante de oxígeno fluirá a través de la cánula nasal
y el tubo. El oxígeno se mide en litros por minuto (LPM). Puede
utilizar un humidificador cuando se use el modo de flujo continuoúnicamente. NO incline, transporte ni recueste el sistema eQuinox
mientras la botella del humidificador esté acoplada.
Funcionamiento del modo de dosis pulsada:
Cuando utilice el modo de dosis pulsada, se suministrará con cadainhalación un bolo de oxígeno medido en mililitros (ml). NO utilice
un humidificador con el modo de dosis pulsada.
Cuando utilice el modo de dosis pulsada, al no detectar respiración,
el sistema eQuinox suministra un flujo continuo de oxígeno cada
15 segundos. El dispositivo detendrá el flujo continuo y comprobará
continuamente cada 15 segundos el esfuerzo inspiratorio. Una vez
detectada la inspiración, el dispositivo volverá a suministrar un bolo
medido. Para salir del modo de dosis pulsada e iniciar el modo deflujo continuo, pulse el botón de modo de flujo.
Solicite a su médico que le prescriba la dosis pulsada correcta para
asegurarse de recibir la saturación de oxígeno adecuada. Tenga encuenta sus necesidades al descansar, al ejercitarse y cuando viaja en
zonas elevadas. Contacte a su proveedor de servicios sanitarios paraobtener una receta y los formularios de embarque.

Paso 5: Regulación del ajuste de flujo al nivel pre-Funciones del modo de dosis pulsada y de autoSat

Su sistema eQuinox tiene una función denominada autoSAT®.
scritoEsta suministra un bolo de tamaño consistente de oxígeno de hasta
Con los botones de ajuste de flujo, regule el ajuste de flujo
40 inhalaciones por minuto (consulte la tabla de frecuencia respiratoria
y defina los parámetros prescritos por el médico.
máxima). A medida que su frecuencia respiratoria aumenta, como
cuando camina, la función autoSAT servo-controla el dispositivo,
ADVERTENCIA: ES MUY IMPORTANTE SELECCIONAR SOLO EL
suministrando el tamaño de bolo definido. La tecnología autoSAT es
NIVEL DE OXÍGENO PRESCRITO. NO CAMBIE LA SELECCIÓN
propiedad de CAIRE y se utiliza durante el modo de dosis pulsada.

DE FLUJO A MENOS QUE UN PROFESIONAL MÉDICO ACREDITADO ASÍ SE LO HAYA INDICADO.

Volumen del bolo (+/- 15 %)Frecuencia respiratoria máxima
Ajuste de dosis
pulsadaTamaño del bolo(± 15 %) ml

Frecuenciarespiratoria máxima

1
2
3
4
5
6
7
8
9

16
32
48
64
80
96
128
160
192

40
40
40
40
37
31
23
18
15

Nota: El volumen del bolo disminuye a medida que la frecuencia respiratoria
supera la frecuencia respiratoria máxima publicada.
Nota: El ajuste de dosis pulsada (bolo) 1-9 no es equivalente a LPM.
ADVERTENCIA: LOS AJUSTES DEL MODO DE DOSIS PULSADADEBEN DETERMINARSE PARA CADA PACIENTE INDIVIDUALMENTE, SEGÚN SUS NECESIDADES EN ESTADO DE REPOSO, AL
REALIZAR EJERCICIO Y AL VIAJAR.

Paso 6: Comienzo de la utilización del sistema eQuinox
Respire con normalidad a través de la cánula nasal.
Nota:Obtendráunmayortiempodefuncionamientoconlabatería,siutiliza
el sistema eQuinox con el modo de dosis pulsada (modo aprobado por elmédico).

Paso 7: Funcionamiento del carrito eQuinox
El carrito eQuinox está diseñado para el uso en superficies para
caminar.
Alinee el sistema eQuinox con la plataforma del carrito de modoque quede vertical y nivelado.
Alinee el tornillo roscado grande con la inserción de la parteposterior del sistema eQuinox. A continuación, apriete a mano la
manija del carrito hasta que el carrito eQuinox esté firmemente
sujeto al dispositivo.
Apriete el botón del asa del carrito eQuinox para ajustar la altura delasa.

ADVERTENCIA: ES POSIBLE QUE EL SISTEMA EQUINOX NO PUEDA DETECTAR TODOS LOS ESFUERZOS RESPIRATORIOS EN EL MODO DE DOSISPULSADA. SI EL SISTEMA EQUINOX NO RESPONDE A SUS ESFUERZOS DERESPIRACIÓN, SOLICITE A SU PROVEEDOR DE ATENCIÓN DOMICILIARIAQUE VERIFIQUE EL AJUSTE DE LA SENSIBILIDAD.
Nota:Alproveedordelequipo:Serecomiendanlossiguientesaccesoriosparala administración de oxígeno para usar con el concentrador de oxígeno:• Cánula nasal con un tubo de 2,1 m (7 pies): NP 5408-SEQ• Para ver accesorios recomendados adicionales, consulte el catálogo deaccesorios (NP ML-LOX0010), disponible en www.CAIREmedical.com.
• CAIRE ofrece el OxySafe™ como accesorio adicional. Está destinado para
serutilizadoconjuntamente con el concentradoreQuinox.Paraclientesen
regiones que requieren el cumplimiento con EN ISO 8359:1996-Enmienda
1:2012, este accesorio cumplirá esta necesidad.
El OxySafe™ es un fusible térmico que detiene el flujo de gas en caso deque la cánula descendente o el tubo de oxígeno se enciendan y ardan hacia
el OxySafe. Se coloca en línea con la cánula nasal o el tubo de oxígeno
entre el paciente y la salida de oxígeno del Equinox.
Para el uso correcto del OxySafe, consulte siempre las instrucciones delfabricante (incluidas con cada conjunto de OxySafe).
	CAIRE ofrece un conjunto de OxySafe™ que incluye un accesorio de lacánula para la cabeza, OxySafe ™ y un tubo de 7’: NP 20628667

Nota: NOcamineyutiliceelsistemaeQuinoxmientraselhumidificadoresté
acoplado. Quite la botella humidificadora antes de andar. NO recueste elsistema eQuinox mientras la botella humidificadora esté acoplada.

Paso 8: Apagado del dispositivo
Mantenga pulsado el botón de ENCENDIDO/APAGADO durante
dos (2) segundos para apagar el sistema eQuinox.
Almacene el sistema eQuinox en un lugar fresco y seco durante un
período no superior a 1 mes. El sistema eQuinox debe funcionar
un mínimo de 2 horas al mes para garantizar un funcionamiento
adecuado de la unidad.
eQuinox (Model 4000) User Manual — 93

Tabla de solución de problemas del usuario
No ignore las alarmas
ADVERTENCIA: EL FABRICANTE RECOMIENDA DISPONER DE UNA FUENTE ALTERNATIVA DE OXÍGENO COMPLEMENTARIO EN CASO DE
CORTE ELÉCTRICO, CONDICIÓN DE ALARMA O FALLA MECÁNICA. CONSULTE A SU MÉDICO O PROVEEDOR DEL EQUIPO PARA DETERMINAREL TIPO DE SISTEMA DE RESERVA QUE NECESITA.

Luz amarilla
Síntoma

Causa posible

Acción1) Extraiga el paquete de baterías y vuelva a instalarlo para asegurarse de que esté bien
La luz amarilla está fija.
ElsistemaeQuinoxhadetectado sujeto en el concentrador.
La alarma emite 1 pitido cada un problema de conexión con el2) De ser posible, instale un paquete de baterías separado en el sistema eQuinox.
30 segundos.
paquete de baterías.
3) Si el problema persiste, póngase en contacto con el proveedor de atención sanitaria.
1) Vuelva a enchufar el sistema eQuinox a una fuente de alimentación externa o instale unpaquetedebateríascompletamentecargadoeneleQuinox.Esperedurante30minutoshasta
que el paquete de baterías tibio se enfríe fuera del concentrador.

La luz amarilla está parpadeando. El ícono de la batería El paquete de baterías del eQui2) Vuelva a cargar el paquete de baterías usando alimentación de CA.
está parpadeando. La alarma nox está tibio.
emite 1 pitido cada 2 minutos.
3) Sustituya el paquete de baterías si hay otro disponible.

4) Si la condición persiste, póngase en contacto con el proveedor de atención sanitaria.
La luz amarilla está parpadeando. El ícono de la batería El voltaje del paquete deestá parpadeando. La alarma baterías del sistema eQuinox
emite 2 pitidos cada 30
es bajo.
segundos.

1) Enchufe el sistema eQuinox a una fuente de alimentación de CA y vuelva a cargar o sustituya el paquete de baterías por un paquete de baterías completamente cargado.

1) Limpie y sustituya el filtro de entrada de aire del gabinete.
2) Asegúrese de que la cánula no esté retorcida ni bloqueada. Si está utilizando el sistema
con una botella humidificadora, asegúrese de que la haya llenado correctamente sin crear
ElsistemaeQuinoxhadetectado una obstrucción.
La luz amarilla está parpadeuna restricción del flujo que
ando. La alarma emite 1 pitido
provoca un bloqueo o un nivel 3)AsegúresedequeelsistemaeQuinoxtengaunaventilaciónadecuada.Debeestarsituado,cada 2 minutos.
al menos, a 7,5 cm (3 pulg.) de cualquier superficie para asegurarse de que las aberturas debajo de flujo.
escape no estén bloqueadas.
4) Si el problema persiste, cambie a otra fuente de oxígeno alternativa y póngase en contacto
con el proveedor de atención sanitaria para obtener ayuda.
1) Asegúrese de que el filtro de entrada de aire no esté obstruido ni comprimido. Limpie
y sustituya el filtro si fuera necesario.
2) Asegúrese de que el sistema eQuinox se encuentre en un lugar con buena ventilación.
Asegúrese de que exista una distancia de 7,5 cm (3 pulg.) entre la parte posterior del sistema
La luz amarilla está parpade- ElsistemaeQuinoxhadetectado eQuinox y cualquier obstáculo posible (muebles, cortinas, etc.).
ando. No hay alarma sonora. niveles bajos de oxígeno.
3) Si utiliza el sistema eQuinox en el coche, asegúrese de que la parte posterior de este mirehacia afuera en el asiento.
4) Si la condición persiste, cambie a otra fuente de oxígeno alternativa y póngase en contacto
con el proveedor de atención sanitaria inmediatamente.

94 — 4805-C1 C (Spanish)

Luz roja
Síntoma

Causa posible

Acción1) Asegúrese de que el filtro de entrada de aire no esté obstruido ni comprimido. Limpie y sustituya el filtro si fuera necesario.

La luz roja está parpadeando. La alarma emite 3
pitidos cada 2 minutos.

2) Asegúrese de que el sistema eQuinox se encuentre en un lugar con buena ventilación.
Asegúrese de que exista una distancia de 7,5 cm (3 pulg.) entre la parte posterior del sistema
El sistema eQuinox ha detectado eQuinox y cualquier obstáculo posible (muebles, cortinas, etc.).
niveles bajos de oxígeno.
3) Si utiliza el sistema eQuinox en el coche, asegúrese de que la parte posterior de este mirehacia afuera en el asiento.
4) Si la condición persiste, cambie a otra fuente de oxígeno alternativa y póngase en contacto
con el proveedor de atención sanitaria inmediatamente.

El sistema eQuinox ha perdido
alimentación temporalmentemientras funcionaba con aliLa luz roja está parpadeanmentación externa.
do. La alarma emite un pitido constante. La pantalla
de LCD muestra ********.
El sistema eQuinox no está El eQuinox ha perdido alisuministrandooxígenoyno mentacióndebidoalagotamiense enciende.
to de la carga del paquete debaterías (paquete de baterías)o el sobrecalentamiento delpaquete de baterías.

2) Compruebe la fuente de alimentación externa (CA o CC) y asegúrese de que las conexiones están bien sujetas al sistema eQuinox. Examine también el transformador y la toma dealimentación externa.
3) Si fuera posible, utilice otra toma eléctrica (CA o CC) para encender el sistema eQuinox.
3) Si la condición persiste, cambie a otra fuente de oxígeno alternativa y póngase en contacto
con el proveedor de atención sanitaria inmediatamente.
1) Vuelva a enchufar el sistema eQuinox a una fuente de alimentación externa o instale unpaquete de baterías completamente cargado en el eQuinox. Espere durante 30 minutos hasta
que el paquete de baterías tibio se enfríe fuera del concentrador.
2) Vuelva a cargar el paquete de baterías usando alimentación de CA.
3) Sustituya el paquete de baterías si hay otro disponible.
4) Si la condición persiste, póngase en contacto con el proveedor de atención sanitaria.
1) Quite el paquete de baterías y desconecte la fuente de alimentación de CA o de CC del sistema eQuinox. La luz roja debería apagarse y el mensaje "FAIL" (ERROR) tendría que desaparecer de la pantalla al quitar la fuente de alimentación. Espere 20 segundos aproximadamente,
vuelva a conectar la fuente de alimentación e intente encender de nuevo el sistema eQuinox.

La luz roja está fija.
La pantalla muestra el
El sistema eQuinox ha experimensaje "FAIL" (ERROR).
mentadounmalfuncionamiento 2) Si el sistema eQuinox se enciende, obsérvelo para determinar si el mensaje FAIL (ERROR)
El sistema eQuinox no estádel sistema.
vuelve a producirse. Si el mensaje FAIL (ERROR) se produce de nuevo, póngase en contacto con
suministrandooxígenoyno
el proveedor de atención sanitaria inmediatamente.
se enciende.
3) Si el sistema eQuinox no se vuelve a encender, póngase en contacto con el proveedor deatención sanitaria inmediatamente.

eQuinox (Model 4000) User Manual — 95

Otras condiciones de alarma
Síntoma

Causa posible

AcciónMientras utilice elmodo de dosis pulsada, el LED del modode dosis pulsada parpadea rápidamente
por 15 segundos.

El sistema eQuinox
no puede detectar elesfuerzo respiratoriodel usuario. Consultela página 15 para obtenermásinformación
sobre el modo de dosis
pulsada.

1)Esposiblequeelmododedosispulsadanofuncionecorrectamentesielusuariopadecealgunaafecciónque
puedaprovocarelbloqueo/obstaculizacióndelasfosasnasales.(Enfermedad,desviacióndemucosidad,etc.).
2) Asegúrese de que la cánula no tenga más de 2 m (7 pies) de largo.
3) Asegúrese de que los tubos de la cánula no estén retorcidos ni obstruidos.
4) Si hay una botella humidificadora conectada, quítela y conecte la cánula directamente al puerto de salida.
El modo de dosis pulsada no funcionará con una botella humidificadora instalada.
5) Si la condición persiste, póngase en contacto con el proveedor de atención sanitaria.
1) Compruebe que el toma de corriente esté suministrando energía. No conecte la fuente de alimentación a un
circuito de atenuación ni a una regleta.

El paquete debaterías se está
descargandomientras
está enchufado a
una fuente de alimentacióndeCA/CC.

2) Compruebe que las conexiones de los cables de las fuentes de alimentación estén fijas a la pared/vehículo
y al concentrador. Si está utilizando una fuente de alimentación de CA, compruebe también la conexión de loscables del transformador de la fuente de alimentación.

El sistema eQuinox
no está recibiendo
3) Asegúrese de que las luces de verificación de energía estén encendidas.
energía de la fuente de- Habrá una luz verde en la caja de la fuente de alimentación.
alimentaciónacoplada.
- Habrá una luz verde en el panel de control del concentrador eQuinox con forma de cable de alimentación.

4) Si los LED de verificación de energía no están encendidos, quite todas las conexiones de la fuente de alimentación durante 20 segundos y, a continuación, vuelva a conectarlas.
5) Si la condición persiste, póngase en contacto con el proveedor de atención sanitaria.
El paquete de baterías 1) Conéctese a una fuente de alimentación de CA para recargar la batería. Compruebe que la batería cargue
durante 2-5 horas y, antes de utilizarla, compruebe que el icono de batería esté lleno y que no parpadee.
no estaba cargado
completamente.
2) Consulte las acciones para el síntoma "El paquete de baterías no carga" que se muestra a continuación.
El sistema eQuinox noestaba funcionando
La carga del paquetea la velocidad de flujo
de baterías (batería)
prevista.
no dura todo lo que
debería.
El paquete de baterías
no está funcionando
según las especificaciones.

1) Asegúrese de que está utilizando la velocidad de flujo prescrita por el médico, y de que ha seleccionado elmodo de flujo continuo o de dosis pulsada, según sea necesario.

1) Intente descargar el paquete de baterías por completo. Para ello, encienda el sistema eQuinox con la alimentación de la batería hasta que la máquina se apague por completo. A continuación, quite la batería y deje
queseenfríedurante30minutosaproximadamente.Transcurridoestetiempo,vuelvaaintroducirlabateríaen
el sistema eQuinox y recárguela utilizando alimentación de CA.
2) Si la batería sigue sin durar el tiempo estimado después de llevar a cabo el paso 1, póngase en contacto conel proveedor de atención sanitaria.
1) Compruebe que el toma de corriente esté suministrando energía. No conecte la fuente de alimentación a un
circuito de atenuación ni a una regleta.

El paquete debaterías (batería) no
carga.

2) Compruebe que las conexiones de los cables de las fuentes de alimentación estén fijas a la pared/vehículo
y al concentrador. Si está utilizando una fuente de alimentación de CA, compruebe también la conexión de los
El sistema eQuinox no cables del transformador de la fuente de alimentación.
recibealimentaciónde 3) Asegúrese de que las luces de verificación de energía estén encendidas.
CA externa para cargar- Habrá una luz verde en la caja de la fuente de alimentación.
la batería.
- Habrá una luz verde en el panel de control del concentrador eQuinox con forma de cable de alimentación.
4) Si los LED de verificación de energía no están encendidos, quite todas las conexiones de la fuente de alimentación durante 20 segundos y, a continuación, vuelva a conectarlas.
5) Si la condición persiste, póngase en contacto con el proveedor de atención sanitaria.
El sistema eQuinox nose está comunicando
adecuadamente con
el paquete de baterías
para cargarlo.

1) Extraiga el paquete de baterías y vuelva a instalarlo para asegurarse de que esté bien sujeto en el concentrador.
2) Compruebe el sistema eQuinox para ver, cuando se instale la batería, si se enciende una luz amarilla fija o si
el icono de batería parpadea al intentar cargarla. Si se presenta alguno de estos síntomas, siga con el paso 3.
3) De ser posible, instale un paquete de baterías separado en el sistema eQuinox. Si la batería de recambio
funciona correctamente, entonces debe reemplazar la batería original.
4) Si el problema persiste, póngase en contacto con el proveedor de atención sanitaria.

El sistema eQuinox
1) Póngase en contacto con el proveedor de atención sanitaria.
emite un pitido 3
La batería interna queveces cuando seenciende la alarma
enciendeporprimera tiene una carga baja.
vez.
96 — 4805-C1 C (Spanish)Limpieza, cuidados y mantenimiento periódico
Mantenimiento periódico

Filtro de entrada de aireÚnicamente un técnico de servicio capacitado y cualificado por
CAIRE debe trabajar con las piezas internas del sistema eQuinox.

El aire ambiente entra en el dispositivo a través de la entrada de aire
situada en la parte superior trasera del dispositivo. Limpiar el filtro
de entrada de aire es la actividad de mantenimiento más importanteque debe realizar como usuario. Limpiar el filtro de entrada deaire mantendrá su sistema eQuinox funcionando correctamente.
Compruebe el filtro de entrada de aire a diario y límpielo, al menos,
una vez a la semana de la siguiente manera:

El mantenimiento preventivo (MP) debe realizarse anualmente.
Consulte al proveedor para organizar el programa de MP. Su
proveedor de atención domiciliaria o técnico cualificado revisaránlas siguientes piezas cuando sea necesario: alarmas, batería internade 9 V, batería, filtros internos y otras piezas internas.
Nota: Si se perciben tres pitidos después de la prueba automática de encendido, esto indica que la batería de 9 V tiene poca energía. Si escucha tres
pitidos, comuníquese con su proveedor de atención domiciliaria.

Cuidados y limpieza del dispositivo por parte delusuario
Reemplazo de la cánula
Sustituya el tubo de suministro y la cánula con la frecuencia queel proveedor de atención domiciliaria recomiende. El médico oproveedor de atención domiciliaria le facilitarán información sobrela limpieza, desinfección y sustitución de piezas.
Nota: Siempre siga las instrucciones del fabricante de la cánula para un uso
correcto. Reemplace la cánula desechable de acuerdo con las recomendacionesdelfabricantedelacánulaodesuproveedordelequipo.Suproveedordel equipo dispone de accesorios de repuesto.

1. Extraiga el filtro de entrada de aire.
2. Lave el filtro con agua del grifo tibia y una solución jabonosa
suave.
3. Aclare el filtro en profundidad con agua del grifo limpia,
y apriete hasta que salga toda el agua.
4. Deje que el filtro se seque al aire.
5. Vuelva a introducir el filtro en la caja.
Sugerencia de limpieza: Conserve un segundo filtro limpio como
reemplazo para su uso cuando un filtro se está secando.

Filtro deentradade aire

PRECAUCIÓN: No opere el concentrador de oxígeno sin el
filtro de entrada de aire colocado. Si dispone de un segundo
filtro, coloque este filtro de "reemplazo" antes de limpiar elfiltro sucio. Limpie el filtro sucio con una solución tibia deagua y jabón, y luego seque por completo antes de usar.
Nota: El fabricante no recomienda la esterilización de este equipo.
Nota: No opere el concentrador de oxígeno sin el filtro de entrada de aire
colocado. Si dispone de un segundo filtro, coloque este filtro de "reemplazo"antes de limpiar el filtro sucio. Limpie el filtro sucio con una solución tibia deagua y jabón, y luego seque por completo antes de usar.

El filtro de entrada de aire debe reemplazarse anualmente. Si se
utiliza el sistema eQuinox en un entorno polvoriento, es posible que
haya que reemplazar el filtro con mayor frecuencia. No debe utilizar
el sistema eQuinox durante más de 5 minutos sin que haya un filtro
de entrada de aire instalado.

eQuinox (Model 4000) User Manual — 97

Fuente de alimentación de CA, cable de 12 VCC, cable de alimentación y carcasa exterior:
APAGUE el sistema eQuinox y desconéctelo de la alimentaciónde CA o de CC antes de limpiarlo o desinfectarlo. NO pulverice
directamente sobre la carcasa externa. Utilice un paño húmedo
(no mojado) o una esponja. Pulverice el paño o la esponja con una
solución de limpieza suave para limpiar el gabinete y la fuentede alimentación. Para desinfectar el sistema eQuinox, utilice
el desinfectante Lysol® Brand II (o uno equivalente). Siga las
indicaciones del fabricante del producto de limpieza.
ADVERTENCIA: RIESGO DE DESCARGA ELÉCTRICA. APAGUELA UNIDAD Y DESCONECTE EL CABLE DE ALIMENTACIÓN DE
LA TOMA ELÉCTRICA ANTES DE LIMPIAR LA UNIDAD PARAPREVENIR UNA DESCARGA ELÉCTRICA ACCIDENTAL ORIESGO DE QUEMADURAS. ÚNICAMENTE SU PROVEEDOR DE EQUIPOS O
UN TÉCNICO DE SERVICIO CUALIFICADO DEBEN QUITAR LAS TAPAS ODAR SERVICIO A LA UNIDAD. NO USE LÍQUIDO DIRECTAMENTE EN LAUNIDAD. A CONTINUACIÓN, SE INCLUYE UNA LISTA DE ALGUNOSAGENTES QUÍMICOS DESACONSEJABLES, ENTRE OTROS: ALCOHOL Y
PRODUCTOS A BASE DE ALCOHOL, PRODUCTOS A BASE DE CLORO
CONCENTRADO (CLORURO DE ETILENO) Y PRODUCTOS A BASE DE
ACEITE (PINE-SOL®, LESTOIL®). ESTOS NO DEBEN USARSE PARA
LIMPIAR LA CARCASA DE PLÁSTICO DEL CONCENTRADOR DE OXÍGENO,YA QUE PUEDEN DAÑAR EL PLÁSTICO DE LA UNIDAD.
ADVERTENCIA: RIESGO DE DESCARGA ELÉCTRICA. DESCONECTE EL CABLE DE ALIMENTACIÓN DE LA TOMA ELÉCTRICA ANTES DE LIMPIAR LA
UNIDAD PARA EVITAR UNA DESCARGA ELÉCTRICA ACCIDENTAL O RIESGOS DE QUEMADURAS. ÚNICAMENTE SU PROVEEDOR DE EQUIPOS O
UN TÉCNICO DE SERVICIO CUALIFICADO DEBEN QUITAR LAS TAPAS ODAR SERVICIO A LA UNIDAD.
ADVERTENCIA: DEBE TENERSE CUIDADO DE EVITAR QUE EL CONCENTRADOR DE OXÍGENO Y LA BATERÍA SE MOJEN Y NO DEBE PERMITIRSEQUE NINGÚN LÍQUIDO ENTRE EN LA UNIDAD. ESTO PUEDE CAUSAR UNA
FALLA O DESACTIVACIÓN Y PRODUCIR UN MAYOR RIESGO DE DESCARGA ELÉCTRICA O QUEMADURAS.
ADVERTENCIA: LIMPIE EL GABINETE, EL PANEL DE CONTROL Y EL CABLE
DE ALIMENTACIÓN SOLO CON UN LIMPIADOR DOMÉSTICO APLICADOCON UN PAÑO HÚMEDO (NO MOJADO) O ESPONJA, Y LUEGO SEQUETODAS LAS SUPERFICIES. NO PERMITA QUE NINGÚN LÍQUIDO ENTRE EN
EL DISPOSITIVO. PRESTE ESPECIAL ATENCIÓN A LA SALIDA DE OXÍGENO
PARA LA CONEXIÓN DE LA CÁNULA PARA ASEGURARSE DE QUE PERMANEZCA LIBRE DE POLVO, AGUA Y PARTÍCULAS.
ADVERTENCIA: NO USE LÍQUIDO DIRECTAMENTE EN LA UNIDAD.
A CONTINUACIÓN, SE INCLUYE UNA LISTA DE ALGUNOS AGENTESQUÍMICOS DESACONSEJABLES, ENTRE OTROS: ALCOHOL Y
PRODUCTOS A BASE DE ALCOHOL, PRODUCTOS A BASE DE
CLORO CONCENTRADO (CLORURO DE ETILENO) Y PRODUCTOS A BASE DE ACEITE (PINE-SOL®, LESTOIL®). ESTOS NO
DEBEN USARSE PARA LIMPIAR LA CARCASA DE PLÁSTICO DELCONCENTRADOR DE OXÍGENO, YA QUE PUEDEN DAÑAR EL PLÁSTICO DE
LA UNIDAD.

98 — 4805-C1 C (Spanish)

Humidificador opcional
Si utiliza un humidificador, límpielo a diario para reducir laposibilidad de contaminación. Siga las indicaciones de limpieza
indicadas por el proveedor de atención domiciliaria o por
el fabricante del humidificador. Sustituya el humidificador
mensualmente, o según lo indique el fabricante o el proveedor
de atención domiciliaria. (No llene demasiado el humidificador).
PRECAUCIÓN: NO camine y utilice el sistema eQuinox
mientras el humidificador esté acoplado. Quite la botella
humidificadora antes de andar. NO recueste el sistema
eQuinox mientras la botella humidificadora esté acoplada.

Carrito para el eQuinox
Antes de limpiar el carrito eQuinox, quítelo del sistema eQuinox.
Utilice un paño húmedo (no mojado) o una esponja con una
solución de limpieza suave para limpiar el carrito.
Paquete de baterías (Batería)
La batería del sistema eQuinox requiere un cuidado especial para
prolongar su vida útil y garantizar el mayor nivel de rendimiento.
La batería CAIRE es la única batería recomendada para el uso con
el sistema eQuinox.
Utilice un paño húmedo (no mojado) o una esponja para limpiar labatería. Primero pulverice el paño o la esponja con un detergente
suave y, a continuación, limpie la carcasa de la batería y el cierre.
CAIRE recomienda la sustitución del paquete de baterías cada
2 años.
Eliminación de la batería
La batería es recargable y puede reciclarse. Devuelva siempre
la batería al proveedor de atención domiciliaria para su correcta
eliminación. También puede ponerse en contacto con las oficinas desu ciudad para obtener instrucciones sobre la correcta eliminaciónde la batería.
Eliminación del sistema de oxígeno eQuinox
Es posible que por las leyes ambientales locales no se permita laeliminación de equipos electrónicos y/o eléctricos como el sistema
eQuinox o la fuente de alimentación de CA. Póngase en contacto
con las oficinas de su ciudad para obtener instrucciones sobre lamanera correcta de eliminar equipos eléctricos o electrónicos.
También puede ponerse en contacto con CAIRE Inc. para obtener
información sobre la eliminación.

Tablas de CEM
Orientación y declaración del fabricante: emisiones electromagnéticas
El sistema eQuinox está destinado para su uso en el ambiente electromagnético que se especifica a continuación. El cliente o el usuario delsistema eQuinox deben verificar que se use en tal ambiente.
Prueba de Emisiones
Emisiones de radiofrecuencia
CISPR 11
Emisiones de radiofrecuencia
CISPR 11

Cumplimiento

Ambiente electromagnético: orientación

Grupo 1El sistema eQuinox usa energía de radiofrecuencia solo para su función interna. En
consecuencia,susemisionesderadiofrecuenciasonmuybajasynoesprobablequecausen interferencias en equipos electrónicos cercanos.

Clase B

Emisiones armónicas
IEC 61000-3-2

Clase A

Fluctuacionesdevoltaje/emisiones de flicker
IEC 61000-3-3

CumpleEl sistema eQuinox es adecuado para usar en todos los establecimientos, incluidos
losestablecimientosdomésticosyaquellosconectadosdirectamentealaredpública
deenergíaeléctricadebajovoltajequealimentaedificiosusadosparafinesdomésticos.

eQuinox (Model 4000) User Manual — 99

Orientación y declaración del fabricante: inmunidad electromagnética
El sistema eQuinox está destinado para su uso en el ambiente electromagnético que se especifica a continuación. El cliente o el usuario delsistema eQuinox deben verificar que se use en tal ambiente.
Prueba de inmunidad

Nivel de prueba IEC 60601

AmbienteelectromagContacto de ± 6 kV
nético: orientación
Aire de ± 8 kV
IEC 61000-4-2
Ráfagas eléctricas
de transitorios muy
rápidos
IEC 61000-4-4
Onda errante
IEC 61000-4-5

Caídas de voltaje,interrupciones por
cortocircuitos y variaciones de voltaje enlas líneas de entrada
de las fuentes de alimentación IEC 610004-11

Campo magnético
a frecuencia industrial (50/60 Hz)
IEC 61000‑4-8± 2 kV para las líneas dealimentación

Nivel de cumplimiento

Ambiente electromagnético: orientación

Contacto de ± 6 kV

Lospisosdebenserdemadera,concretoocerámica.
Si los pisos están cubiertos con material sintético, la
humedad relativa debe ser de al menos un 30 %.

Aire de ± 8 kV
± 2 kV para las líneas dealimentación± 1 kV para líneas de entrada/
N/Dsalida± 1 kV línea(s) a línea(s)± 1 kV línea(s) a línea(s)± 2 kV línea(s) a tierra< 5 % UT

< 5 % UT

(Caída > 95 % en UT )(Caída > 95 % en UT )para 0,5 ciclos

para 0,5 ciclos

40 % UT

40 % UT(Caída del 60 % en UT )(Caída del 60 % en UT )para 5 ciclos

para 5 ciclos

70 % UT

70 % UT(Caída del 30 % en UT )(Caída del 30 % en UT )para 25 ciclos

para 25 ciclos< 5 % UT

< 5 % UT

(Caída > 95 % en UT )(Caída > 95 % en UT )para 5 segundospara 5 segundos

3 A/m

3 A/m

La calidad de la red eléctrica debe ser la de un ambiente comercial u hospitalario típico.

La calidad de la red eléctrica debe ser la de un ambiente comercial u hospitalario típico.

La calidad de la red eléctrica debe ser la de un ambiente comercial u hospitalario típico. Si el usuario delsistemaeQuinoxrequiereunfuncionamientocontinuo durante cortes de la red eléctrica, se recomienda
que el eQuinox se alimente desde una fuente dealimentaciónsininterrupcionesodesdeunabatería.

Los campos magnéticos a frecuencia industrial
debentenerunnivelcaracterísticodeunaubicación
típicaenunambientecomercialuhospitalariotípico.

NOTA: UT es el voltaje de CA de la red eléctrica antes de la aplicación del nivel de prueba.

100 — 4805-C1 C (Spanish)Orientación y declaración del fabricante: inmunidad electromagnética
El sistema eQuinox está destinado para su uso en el ambiente electromagnético que se especifica a continuación. El cliente o el usuario delsistema eQuinox deben verificar que se use en tal ambiente.
Prueba de inmunidad

Nivel de prueba IEC 60601

Nivel de cumplimiento

Ambiente electromagnético: orientación

Radiofrecuencia
conducida

3 Vrms3 VrmsLosequiposportátilesymóvilesdecomunicaciones
por radiofrecuencia no deben usarse más cerca delas piezas del sistema eQuinox, incluidos los cables,
que la distancia de separación recomendada calculada a partir de la ecuación aplicable a la frecuenciadel transmisor.

IEC 61000-4-6

150 kHz a 80 MHz

RF irradiada

3 V/m

IEC 61000-4-3

80 MHz a 2,5 GHz

3 V/mDistancia recomendada de separaciónd = 1.2 P
d = 1.2 P = 80 MHz a 800 MHz
d = 1.2 P = 800 MHz a 2,5 GHz
donde P es la clasificación máxima de potencia desalida del transmisor en vatios (W) de acuerdo conel fabricante del transmisor y d es la distancia deseparación recomendada en metros (m).
Lasconcentracionesdecampodetransmisoresfijosde RF, determinadas por un relevamiento electromagnético del sitio, a deberían ser inferiores al nivel
de cumplimiento en cada rango de frecuencia. b
Puede producirse interferencia en la cercanía de losequipos marcados con el símbolo siguiente:

NOTA 1: A 80 MHz y 800 MHz, se aplica el rango de mayor frecuencia.
NOTA 2: Estos lineamientos pueden no ser de aplicación en todas las situaciones. La propagación electromagnética se ve afectada por la adsorción y la reflexión de estructuras, objetos y personas.
Las concentraciones de campo de los transmisores fijos, tales como las estaciones de base para teléfonos de radio (móviles/inalámbricos) y radios móviles terrestres, radios de aficionados y las transmisiones de radio de AM y FM y la transmisión deTV, no pueden predecirse
teóricamente con precisión. Para evaluar el ambiente electromagnético debido a transmisores fijos de RF, debe considerarse un relevamiento
electromagnético del sitio. Si la concentración de campo medida en la ubicación donde se usa el sistema eQuinox supera el nivel correspondientedecumplimientodeRFanterior,elsistemaeQuinoxdebeobservarseparaverificarsufuncionamientonormal.Siseobservaunrendimiento
anormal, pueden necesitarse medidas adicionales, tales como la reorientación o la reubicación del sistema eQuinox.
ab

En el rango de frecuencia de 150 kHz a 80 MHz, las concentraciones de campo deben ser inferiores a 3 V/m.

eQuinox (Model 4000) User Manual — 101

Distancias recomendadas de separación
entre equipos portátiles y móviles de comunicaciones por radiofrecuencia y el sistema eQuinox.
ElsistemaeQuinoxestádiseñadoparaserutilizadoenunambienteelectromagnéticoenelquesecontrolanlasinterferenciasderadiofrecuencia
irradiada.ElclienteoelusuariodelsistemaeQuinoxpuedenayudaraprevenirlainterferenciaelectromagnéticaalmantenerunadistanciamínimaentrelosequiposportátiles ymóviles decomunicaciónpor radiofrecuencia(transmisores) y eleQuinox,talcomoserecomienda a continuación, de acuerdo con la potencia máxima de salida de los equipos de comunicaciones.
Potencia de salida nominal máxima del transmisor

Distancia de separación de acuerdo con la frecuencia del transmisorm
150 kHz a 80 MHz

80 MHz a 800 MHz

800 MHz a 2,5 GHz

d = 1,2
1.2 P

d =11,2.2 P

d  1=.21,2P

0,01

0,12

0,12

0,23

0,1

0,38

0,38

0,73

1

1,2

1,2

2,3

10

3,8

3,8

7,3

100

12

12

23

W

Para transmisores con una potencia nominal máxima de salida no mencionada anteriormente, la distancia de separación recomendada d en
metros (m) puede calcularse usando la ecuación aplicable a la frecuencia del transmisor, donde P es la potencia nominal máxima de salida deltransmisor en vatios (W) de acuerdo con el fabricante del transmisor.
NOTA 1: A 80 MHz y 800 MHz, se aplica la distancia de separación para el rango de mayor frecuencia.
NOTA 2: Estos lineamientos pueden no ser de aplicación en todas las situaciones. La propagación electromagnética se ve afectada por la adsorción y la reflexión de estructuras, objetos y personas.

102 — 4805-C1 C (Spanish)

Información general para viajes aéreos
Si no va a utilizar el sistema eQuinox, durante el carreteo, el
despegue y el aterrizaje del avión, debe apagarlo y guardarlo debajodel asiento o en otra ubicación de almacenamiento aprobada para no
bloquear el pasillo ni la entrada a la fila.
Si va a utilizar el sistema eQuinox, durante el carreteo, el despegue
o el aterrizaje del avión, el usuario debe situarse en un asiento
que no impida el acceso o el uso por parte de otros pasajeros delas puertas de salida o de emergencia ni del (de los) pasillo(s) del
habitáculo de pasajeros del avión.
Si el usuario utiliza el sistema eQuinox, no podrá estar sentado en
una fila de salida.
Si el usuario está utilizando el sistema eQuinox cuando se produce
la descompresión de la cabina y salta el sistema de suministro deoxígeno de la cabina, entonces deberá dejar de utilizar el sistema
eQuinox para hacer uso del oxígeno suplementario del avión. La
unidad eQuinox deberá apagarse una vez que se haya asegurado eloxígeno suplementario del avión.
Antes de viajar, el usuario debe inspeccionar el sistema
eQuinox para asegurarse de que esté en buenas condiciones defuncionamiento.

Paquete de baterías y viajes aéreos
Tenga en cuenta el tiempo en tierra (facturación
antes del vuelo, control de seguridad, etc.) y las
escalas.
La mayoría de las aerolíneas requieren que cuente con un número
adecuado de baterías cargadas completamente para alimentar eldispositivo por al menos 150 % del tiempo máximo de duración
prevista del viaje (tiempos de vuelo, todo el tiempo en tierra;antes y después del vuelo y durante las conexiones y para demoras
imprevistas).

Se recomienda viajar siempre con una batería
adicional.
Cada aerolínea tiene sus propios requisitos para poder viajar
con oxígeno.
Póngase en contacto con su aerolínea al menos 48 horas antes dela salida del vuelo para informarles de que viajará con un sistema
eQuinox.

PRECAUCIÓN: Los reglamentos del Departamento deTransporte (DOT, por sus siglas en inglés) estadounidensey de las Naciones Unidas (ONU) requieren la extracción de labatería del dispositivo para todos los viajes en aerolíneas
internacionales cuando se facture el concentrador de oxígeno comoequipaje. Cuando se envía el concentrador de oxígeno, la batería
también debe extraerse del dispositivo y embalarse adecuadamente.

eQuinox (Model 4000) User Manual — 103

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107 EE. UU.
1-800-482-2473
En todo el mundo: www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Alemania
En todo el mundo: www.CAIREmedical.com

CAIRE Inc. se reserva el derecho a suspender sus productos o cambiar sus precios, materiales, equipo, calidad, descripciones,
especificacionesy/oprocesosdesusproductosencualquiermomento,sinprevioaviso,ysinningúncompromisooconsecuencia.
Queda reservado cualquier otro derecho que no se encuentre indicado explícitamente en este documento, según proceda.

104 — 4805-C1 C (Spanish)

A partir de la fecha de publicación, eQuinox, SeQual y autoSAT y sus logotipos relacionados(según corresponda) son marcas registradas de CAIRE Inc.
Lysol® es una marca registrada de Reckitt Benckiser, del Reino Unido.
OxySafe® es una marca registrada de Applied Home Healthcare Equipment.
Pine-Sol® y Lestoil® son marcas registradas de The Clorox Company.

Manuale dell'utente

eQuinoxTM

Modello 4000
eQuinox (Model 4000) User Manual — 105

Controlli utente e indicatori di stato del sistema
SimboloA

02

Definizione

Definizione

Leggere il manuale dell'utente prima dell'uso.
Consultare il manuale dell'utente per le istruzioni.

Simbolo di approvazione FAA: la Federal Aviation
Administration (FAA) ha approvato l'uso di questo
dispositivo a bordo di aerei commerciali.

Icona Vietato fumare: non fumare vicino all'unità.

Indicatore dell'impostazione di flusso

Avvertenze / Indicatore di AVVISO (giallo)

Funzionamento in modalità Impulso

Indicatore di ALLARME (rosso)

Funzionamento normale del dispositivo; pulsante diaccensione

Indicatore di alimentazione esterna

Aumentare l'impostazione di flusso

Indicatore di stato della batteria

Ridurre l'impostazione di flusso

Simbolo della batteria

Non usare oli o grassi.

AmpereNon usare fiamme libere.

Corrente alternata

Non contiene componenti riparabili. Non aprire lacopertura.

Icona IEC per alimentazione CC (ingresso)

Superficie calda.

Uscita ossigeno

Non bagnare.

Parte applicata di tipo BF (grado di protezione dascossa elettrica)

Apparecchiatura IPX1 a tenuta stagna: la batteria
eQuinox e l'alimentatore CA offrono la protezione
contro gli effetti dannosi causati dall'ingresso diliquidi (IPX1, in conformità allo standard IEC
60529).

Riciclabile

IP22

Simbolo

Questo simbolo ha lo scopo di ricordare ai
proprietari dell'apparecchiatura di consegnarlapresso un centro per il riciclaggio al termine dellasua vita utile, in conformità alla Direttiva sui
rifiuti di apparecchiature elettriche ed elettroniche(RAEE).
Protezione contro l'ingresso di solidi e liquidi: ildispositivo eQuinox offre protezione contro gli
effetti dannosi causati dall'ingresso di liquidi e
corpi estranei solidi in conformità allo standard IEC
60529 con classificazione IP22.
Certificata per i mercati statunitense e canadese inconformità ai relativi standard applicabili.

106 — 4805-C1 C (Italian)

IPX1

Nome e indirizzo del produttore

Rappresentante autorizzato per la Comunità Europea

Apparecchiatura di classe II
Il presente dispositivo è conforme ai requisiti della
Direttiva 93/42/CEE relativa ai dispositivi medici.
Pertanto, riporta il marchio CE come illustrato.

Sommario
Controlli utente e indicatori di stato del sistema............................................................................................................ 106
Guida introduttiva...................................................................................................................................................................... 108
Indicazioni per l'uso................................................................................................................................................................... 111
Controindicazioni all'uso......................................................................................................................................................... 111
Linee guida per la sicurezza................................................................................................................................................... 111
Ambienti operativi consigliati............................................................................................................................................... 113
Istruzioni per l'uso...................................................................................................................................................................... 114
Funzionamento del modulo vocale..................................................................................................................................... 117
Primo utilizzo del dispositivo eQuinox............................................................................................................................... 118
Tabella di risoluzione dei problemi dell'utilizzatore ..................................................................................................... 120
Pulizia, cura e manutenzione ordinaria.............................................................................................................................. 123
Tabelle CEM.................................................................................................................................................................................. 125
Informazioni generali sui viaggi in aereo.......................................................................................................................... 129
Batteria e per viaggi aerei........................................................................................................................................................ 129

eQuinox (Model 4000) User Manual — 107

Guida introduttiva

1Disimballaggio del dispositivo eQuinox

2

Descrizione del dispositivo eQuinox

Leggere con attenzione tutti i messaggi di avvertenza e diattenzione e le informazioni aggiuntive sul dispositivo riportati
nel resto del manuale.
Acquisire dimestichezza con le funzionalità principali deldispositivo eQuinox e del pannello di controllo utente.
Uscita dell'ossigeno

PARTE ANTERIORE
Maniglia

Pannello dicontrollo
Presa peralimentazione
esterna

Alimentatore CA

Icona di approvazione FAA

Fermo della batteria

Posizionedell'etichetta diclassificazione e delnumero di serie

Carrello eQuinox

Batteria

PARTE POSTERIORE

Cavo 12 VCC

Alloggiamento
di servizio

Sfiato discaricoBatteria

108 — 4805-C1 C (Italian)

Filtro diingressodell'aria

Sul pannello di controllo del dispositivo
eQuinox sono visualizzate importanti
informazioni sul funzionamento.

Indicatore del pulsante ON/OFF (verde): questo pulsanteconsentediaccendereospegnereildispositivo.L'indicatoreverde si accende quando il dispositivo è acceso e funziona
correttamente.
Pulsantiperl'aumentooladiminuzionedell'impostazionedi
flusso:consentonodiimpostareilflussosulvaloreprescritto.
Indicatore e pulsante della modalità di erogazione: ilpulsanteconsentedipassaredallamodalitàFlussocontinuo
alla modalità Dosaggio impulsi e viceversa. La modalità
DosaggioimpulsiattivalatecnologiaautoSAT,chepermetteunnotevoleaumentodeltempodifunzionamentoquando
il dispositivo è alimentato a batteria. Quando è attivata lamodalitàDosaggioimpulsi,l'indicatoreverdecorrispondenteèacceso;duranteciascunosforzoinspiratoriovieneerogato
un impulso di ossigeno e l'indicatore verde lampeggia.
IndicatorediAVVISO(giallo):seacceso,indicaunmessaggio
diattenzioneounacondizionediprioritàbassa.Continuare
a utilizzare il sistema. Fare riferimento alla Tabella per larisoluzionedeiproblemiperconoscerel'interventoadeguatoo rivolgersi al fornitore di assistenza domiciliare. Il giallo
lampeggianteindicaunavvisodiprioritàmedia.Ènecessaria
una risposta rapida (in genere questi avvisi indicano livelli
bassi di ossigeno o di carica della batteria).
Indicatore di ALLARME (rosso): indica una condizione diallarmediprioritàalta.Ènecessariaunarispostaimmediata.
Fare riferimento alla Tabella per la risoluzione dei problemi
e rivolgersi al fornitore di assistenza domiciliare (in genere
questi allarmi indicano perdita di potenza, bassi livelli diossigeno o malfunzionamenti).
Display informativo.
• C o P: "C" indica la modalità Flusso continuo, "P" lamodalità Dosaggio impulsi.
• Tempo di funzionamento a batteria: durante il funzionamento a batteria, il display segnala ogni 20 secondi circa ladurata rimanente stimata della batteria.
• Impostazione (flusso continuo o dosaggio impulsi): ogni
voltachesiaccendeildispositivo,all'avviovengonoutilizzate
l'ultima modalità e/o impostazione salvate in precedenza.
Indicatoredistatodellabatteria:questoindicatorevisualizzala carica residua della batteria. Ciascuna delle cinque barre
grigie orizzontali rappresenta circa il 20% della carica totaledellabatteria.Quandolabatteriaèincarica,lebarredell'indicatore di carica lampeggiano a cascata. Se la batteria non è
presenteononèinseritacorrettamente,l'indicatoredistatodella batteria rimane spento o lampeggia a intermittenza.
Indicatore di alimentazione esterna: quando il dispositivo
eQuinox è collegato correttamente all'alimentazione CA oal cavo 12 V CC, questo indicatore sul pannello di controllo
utente si accende.
Cicalino:unallarmeacusticoavvisadellecondizionidifunzionamentodeldispositivo(adesempioavvertenzeoguasti)epuòessereconfermatopremendo
un tasto valido.
Modulo vocale: se attivato, un indicatore vocale segnala all'utente le condizionidifunzionamentodeldispositivo,levariazionialivellodialimentazione, le condizioni di avvertenza o i guasti.

3

Accensione del dispositivo eQuinox

Per collegare una sorgente di alimentazione CA: inseriresaldamente il cavo dell'alimentatore CA nella presa peralimentazione esterna sul lato destro del dispositivo. Inserire ilcavo di alimentazione CA nell'alimentatore e in una presa CA.
Fase 1

Fase 2

Fase 3

Quando il dispositivo eQuinox è collegato correttamente,l'indicatore verde sull'alimentatore CA si accende. Anchel'indicatore di alimentazione esterna
sul pannello dicontrollo utente si accende.
Inserire la batteria: far scorrere la batteria nel vano vuoto finoa bloccarla in posizione, allineandola alla parte anteriore deldispositivo.

Quando la batteria è inserita correttamente e il dispositivo
eQuinox è acceso e collegato all'alimentazione CA, viene
visualizzato l'indicatore di stato della batteria.

Indicatore del pulsante ON/OFF (verde)
Tenere premuto il pulsante "ON/OFF" per due (2) secondi peraccendere il dispositivo eQuinox. Viene eseguito un brevetest POST acustico e visivo. Verificare che tutti gli indicatori
siano accesi e che il cicalino emetta un segnale acustico pertre (3) secondi. Tenere premuto il pulsante "ON/OFF" per due(2) secondi per spegnere il dispositivo eQuinox.
eQuinox (Model 4000) User Manual — 109

4S elezione della modalità di erogazionedel flusso

Indicatore e pulsante della modalità di flusso

Premere il pulsante della modalità di flusso per selezionare lamodalità desiderata. Premendo ripetutamente questo pulsanteè possibile passare tra le modalità Dosaggio impulsi e Flusso
continuo. Nella modalità Flusso continuo, indicata dalla lettera C,
l'ossigeno viene erogato a un flusso costante compreso tra 0,5 e
3,0 lpm. Nella modalità Dosaggio impulsi, l'ossigeno viene
fornito in un bolo all'inizio di ogni inalazione. Le impostazioni
1-9 della modalità Impulso, indicata dalla lettera P sul display,
permettono di scegliere la dimensione del bolo tra 16 ml e
192 ml. Il dosaggio impulsi può essere regolato a livello disensibilità ed erogazione del bolo d'aria. Le impostazioni di flusso
in entrambe le modalità devono essere prescritte dal medico per5 egolazione dell'impostazione di flusso al
Rlivello prescritto

Pulsante per l'aumento o la diminuzione dell'impostazione di flusso

6

Uso iniziale del dispositivo eQuinox

Filtro di ingresso dell'aria

Filtro diingressodell'aria

L'unità è dotata di un filtro di ingresso dell'aria (filtro aggiuntivo
in dotazione). Controllarlo quotidianamente e pulirlo una volta asettimana. Le istruzioni per la pulizia sono a pagina 19.
Collegare il tubo di erogazione dell'ossigeno all'uscita
dell'ossigeno e collegare la cannula dell'ossigeno in conformità
alle istruzioni fornite con la cannula stessa o dal fornitore di
assistenza domiciliare.

Respirare normalmente attraverso la cannula nasale.
Durante il funzionamento a batteria, l'utilizzo della modalità
Dosaggio impulsi aumenta la durata d'uso.

È possibile regolare l'impostazione alla prescrizione del medico
tramite i pulsanti + e –.

Display della modalità Flusso continuo

Display della modalità Dosaggio impulsi

AVVERTENZA: È MOLTO IMPORTANTE SELEZIONARE SOLOIL LIVELLO DI OSSIGENO PRESCRITTO. NON MODIFICARELA SELEZIONE DEL FLUSSO SE NON IN CASO DI PRECISEISTRUZIONI DA PARTE DI UN MEDICO AUTORIZZATO. È POSSIBILEUTILIZZARE IL CONCENTRATORE PORTATILE DI OSSIGENO DURANTEIL SONNO SU RACCOMANDAZIONE DI UN MEDICO QUALIFICATO.
110 — 4805-C1 C (Italian)7

Uso del carrello eQuinox

Allineare il dispositivo eQuinox al pavimento sagomato delcarrello in modo che risulti in posizione piana e verticale.
Allineare la vite filettata grande alla guarnizione sul retrodel dispositivo eQuinox. Infine, serrare a mano la manopola
sul carrello finché il carrello non risulta saldamente fissato al
dispositivo. Premere il pulsante sulla maniglia telescopica delcarrello e tirarla verso l'alto per regolarne l'altezza.

Importante!
Le istruzioni per la sicurezza sono definite comeindicato di seguito:
AVVERTENZA: IMPORTANTI INFORMAZIONI PER LA SICUREZZA RELATIVE AI PERICOLI CHE POSSONO CAUSAREGRAVI LESIONI.
ATTENZIONE: informazioni importanti per evitare danni al
dispositivo eQuinox.
Nota: informazioni che richiedono particolare attenzione.

Indicazioni per l'uso
eQuinox è previsto per la somministrazione di ossigeno supplementare. Il dispositivo non è previsto per il supporto alla vita, né
offre alcuna capacità di monitoraggio del paziente.
AVVERTENZA: È MOLTO IMPORTANTE SELEZIONARE SOLOIL LIVELLO DI OSSIGENO PRESCRITTO. NON MODIFICARELA SELEZIONE DEL FLUSSO SE NON IN CASO DI PRECISEISTRUZIONI DA PARTE DI UN MEDICO AUTORIZZATO. È POSSIBILEUTILIZZARE IL CONCENTRATORE PORTATILE DI OSSIGENO DURANTEIL SONNO SU RACCOMANDAZIONE DI UN MEDICO QUALIFICATO.
AVVERTENZA: LA LEGGE FEDERALE (USA) LIMITA LA VENDITA O IL
NOLEGGIO DI QUESTO DISPOSITIVO DIETRO PRESCRIZIONE DI UN
MEDICO O DI UN ALTRO PROFESSIONISTA SANITARIO QUALIFICATO.
AVVERTENZA: QUESTA UNITÀ NON DEVE ESSERE UTILIZZATACOME SUPPORTO SALVAVITA. I PAZIENTI GERIATRICI, PEDIATRICI OINCAPACI DI COMUNICARE I PROPRI DISTURBI DURANTE L'USO DI
QUESTO DISPOSITIVO POTREBBERO NECESSITARE DI UN MONITORAGGIO AGGIUNTIVO. I PAZIENTI CON DANNI ALL'UDITO E/O ALLAVISTA POTREBBERO NECESSITARE DI ASSISTENZA PER MONITORAREGLI ALLARMI.

Controindicazioni all'uso

MEDICA.

AVVERTENZA: IN ALCUNE CIRCOSTANZE, L'USO DI OSSIGENO NON PRESCRITTO PUÒ ESSERE PERICOLOSO. QUESTODISPOSITIVO VA UTILIZZATO SOLO DIETRO PRESCRIZIONEAVVERTENZA: NON UTILIZZARE IN PRESENZA DI ANESTETICI INFIAMMABILI.
AVVERTENZA: ANALOGAMENTE A QUALSIASI ALTRO DISPOSITIVO ELETTRICO, POTREBBERO VERIFICARSI PERIODI DI INATTIVITÀ
DOVUTI ALL'INTERRUZIONE DELL'ALIMENTAZIONE ELETTRICA OALLA NECESSITÀ DI FAR RIPARARE IL DISPOSITIVO EQUINOX DA UNTECNICO QUALIFICATO. IL DISPOSITIVO EQUINOX NON È INDICATOPER PAZIENTI CHE POTREBBERO SUBIRE CONSEGUENZE NEGATIVESULLA PROPRIA SALUTE A CAUSA DI TALE INTERRUZIONE TEMPORANEA.

Linee guida per la sicurezza
AVVERTENZA: NON È CONSENTITO APPORTARE ALCUNAMODIFICA ALL'APPARECCHIATURA.
AVVERTENZA: IL PRODUTTORE CONSIGLIA DI TENEREA DISPOSIZIONE UN'ALTRA SORGENTE DI OSSIGENO
SUPPLEMENTARE IN CASO DI INTERRUZIONI DI CORRENTE,
CONDIZIONI DI ALLARME O GUASTI MECCANICI. PER CONOSCERE IL TIPO DI SISTEMA DI RISERVA NECESSARIO, RIVOLGERSIAL PROPRIO MEDICO O AL FORNITORE DELL'APPARECCHIATURA.
AVVERTENZA: QUESTO DISPOSITIVO EROGA OSSIGENO A ELEVATA
CONCENTRAZIONE, CHE FAVORISCE IL RAPIDO SVILUPPO DI INCENDI.
NON CONSENTIRE AL PAZIENTE DI FUMARE O DI USARE FIAMME
LIBERE ALL'INTERNO DELLA STANZA IN CUI SI TROVA (1) QUESTO DISPOSITIVO OPPURE (2) QUALSIASI ALTRO ACCESSORIO PER L'EROGAZIONE DELL'OSSIGENO. IL MANCATO RISPETTO DI QUESTA AVVERTENZAPUÒ COMPORTARE GRAVI INCENDI, DANNI MATERIALI E/O LESIONIFISICHE O IL DECESSO.
AVVERTENZA: NON UTILIZZARE L'UNITÀ IN SPAZI CONFINATI O
ANGUSTI (AD ESEMPIO UNA BORSA O UNA CUSTODIA DI PICCOLEDIMENSIONI) IN CUI LA VENTILAZIONE POSSA ESSERE LIMITATA. IL
CONCENTRATORE DI OSSIGENO POTREBBE SURRISCALDARSI PREGIUDICANDO LE PRESTAZIONI.
AVVERTENZA: IL CONCENTRATORE VA POSIZIONATO IN MODO DA
EVITARE FUMO, AGENTI INQUINANTI O VAPORI.
AVVERTENZA: L'USO DI ALCUNI ACCESSORI PER LA SOMMINISTRAZIONE DI OSSIGENO NON ESPRESSAMENTE INDICATI PER L'USO CONQUESTO CONCENTRATORE DI OSSIGENO PUÒ PREGIUDICARNE LE
PRESTAZIONI. GLI ACCESSORI RACCOMANDATI SONO SEGNALATI
ALL'INTERNO DEL PRESENTE MANUALE.
AVVERTENZA: SE IL CONCENTRATORE DI OSSIGENO VIENE FATTOCADERE, DANNEGGIATO O ESPOSTO ALL'ACQUA, RIVOLGERSI AL
FORNITORE DI ASSISTENZA DOMICILIARE PER UN'ISPEZIONE O
UN'EVENTUALE RIPARAZIONE DEL DISPOSITIVO. NON UTILIZZARE IL
CONCENTRATORE DI OSSIGENO SE IL CAVO O LA SPINA DI ALIMENTAZIONE SONO DANNEGGIATI.
AVVERTENZA: DURANTE IL TRASPORTO ALL'INTERNO DI UN VEICOLO,FISSARE IL CONCENTRATORE DI OSSIGENO IN MANIERA ADEGUATA,
MEDIANTE UNA CINTURA O UN FERMO DI ALTRO TIPO, PER EVITAREDANNI O LESIONI.
AVVERTENZA: NON OSTRUIRE L'INGRESSO O LE USCITE DELL'ARIA.
NON FAR CADERE O INSERIRE ALCUN OGGETTO NELLE APERTURE
PRESENTI SUL DISPOSITIVO. IL CONCENTRATORE DI OSSIGENO POTREBBE SURRISCALDARSI PREGIUDICANDO LE PRESTAZIONI.
AVVERTENZA: NON RIEMPIRE ECCESSIVAMENTE L'UMIDIFICATOREOPZIONALE. RIEMPIRE L'UMIDIFICATORE OPZIONALE CON ACQUAESCLUSIVAMENTE FINO AL LIVELLO INDICATO DAL PRODUTTORE.
AVVERTENZA: NON UTILIZZARE IL DISPOSITIVO EQUINOX E NON DEAMBULARE MENTRE È COLLEGATO UN UMIDIFICATORE. RIMUOVEREIL FLACONE DELL'UMIDIFICATORE PRIMA DI DEAMBULARE. NON
DISTENDERE IL DISPOSITIVO EQUINOX QUANDO È COLLEGATO A UN
FLACONE UMIDIFICATORE.
AVVERTENZA: IL DEPARTMENT OF TRANSPORTATION (DOT) DEGLISTATI UNITI E I REGOLAMENTI DELLE NAZIONI UNITE RICHIEDONOLA RIMOZIONE DELLA BATTERIA DAL DISPOSITIVO PER L'INTERO
VIAGGIO AEREO INTERNAZIONALE QUANDO IL CONCENTRATOREDI OSSIGENO VIENE REGISTRATO COME BAGAGLIO. DURANTE LA
SPEDIZIONE DEL CONCENTRATORE DI OSSIGENO, È NECESSARIO RIMUOVERE LA BATTERIA DAL DISPOSITIVO E IMBALLARLO IN MANIERAADEGUATA.

eQuinox (Model 4000) User Manual — 111

AVVERTENZA: NON UTILIZZARE QUESTA APPARECCHIATURA SENZA PRIMA AVER LETTO E COMPRESO IL PRESENTEMANUALE. IN CASO DI DIFFICOLTÀ A COMPRENDERE LE AVVERTENZE E LE ISTRUZIONI, RIVOLGERSI AL FORNITORE DELL'APPARECCHIATURA PRIMA DI TENTARE DI UTILIZZARLA; IN CASO CONTRARIO,POSSONO VERIFICARSI DANNI O LESIONI.
AVVERTENZA: SE SI MANIFESTANO DISTURBI O IN CASO DI EMERGENZA MEDICA, RIVOLGERSI IMMEDIATAMENTE A UN MEDICO.
AVVERTENZA: L'UTILIZZO DEL CONCENTRATORE DI OSSIGENO AL DI
FUORI DELLE SPECIFICHE DELLA TEMPERATURA DI ESERCIZIO PUÒ
LIMITARE LA CAPACITÀ DEL CONCENTRATORE DI SODDISFARE I VALORI
SPECIFICATI PER LA CONCENTRAZIONE DELL'OSSIGENO. PER I LIMITI DI
TEMPERATURA, FARE RIFERIMENTO ALLA SEZIONE DELLE SPECIFICHE
NEL PRESENTE MANUALE.
AVVERTENZA: NON UTILIZZARE OLIO, GRASSO, PRODOTTI A BASE DIPETROLIO O INFIAMMABILI DI ALTRO TIPO CON ACCESSORI PER IL
TRASPORTO DELL'OSSIGENO O CON IL CONCENTRATORE DI OSSIGENO.
UTILIZZARE ESCLUSIVAMENTE LOZIONI O UNGUENTI A BASE D'ACQUACOMPATIBILI CON L'OSSIGENO. L'OSSIGENO ACCELERA LA COMBUSTIONE DELLE SOSTANZE INFIAMMABILI.
AVVERTENZA: L'USO NON CORRETTO DELLA BATTERIA PUÒ CAUSARNEIL SURRISCALDAMENTO E L'ACCENSIONE, PROVOCANDO GRAVI LESIONI. PRESTARE ATTENZIONE A NON BUCARE, COLPIRE, SALIRE SOPRAO FAR CADERE LA BATTERIA, E NON ESPORLA A IMPATTI O SOLLECITAZIONI FORTI. L'USO DI UNA BATTERIA DANNEGGIATA PUÒ CAUSARELESIONI PERSONALI.
AVVERTENZA: NON UTILIZZARE IL CONCENTRATORE DI OSSIGENO ACCANTO O IMPILATO SU ALTRE APPARECCHIATURE. SE NON È POSSIBILEEVITARE TALE MODALITÀ DI UTILIZZO, OSSERVARE IL DISPOSITIVO PERVERIFICANE IL NORMALE FUNZIONAMENTO.
ATTENZIONE: in caso di allarme o qualora si osservi
un funzionamento non corretto del concentratore diossigeno, consultare la sezione Risoluzione dei problemi
nel presente manuale. Se non si riesce a risolvere il problema,rivolgersi al fornitore dell'apparecchiatura.
ATTENZIONE: per evitare l'annullamento della garanzia, attenersi
alle istruzioni del produttore.
AVVERTENZA: NON TENTARE DI ESEGUIRE INTERVENTI DI
MANUTENZIONE, A ECCEZIONE DELLE SOLUZIONI POSSIBILI ELENCATE ALL'INTERNO DEL PRESENTE MANUALE.
NON RIMUOVERE LE COPERTURE; TALE OPERAZIONE O GLI INTERVENTI DI ASSISTENZA DEVONO ESSERE ESEGUITI ESCLUSIVAMENTEDAL FORNITORE DELL'APPARECCHIATURA O DA UN TECNICO DI
ASSISTENZA QUALIFICATO.
AVVERTENZA: UTILIZZARE ESCLUSIVAMENTE GLI ACCESSORI RACCOMANDATI DAL PRODUTTORE. L'USO DI ALTRI ACCESSORI PUÒ
ESSERE PERICOLOSO, CAUSARE GRAVI DANNI AL CONCENTRATOREDI OSSIGENO E RENDERE NULLA LA GARANZIA.
AVVERTENZA: IL PRODUTTORE CONSIGLIA DI TENERE ADISPOSIZIONE UN'ALTRA SORGENTE DI OSSIGENO SUPPLEMENTARE IN CASO DI INTERRUZIONI DI CORRENTE,
CONDIZIONI DI ALLARME O GUASTI MECCANICI. PER CONOSCERE IL
TIPO DI SISTEMA DI RISERVA NECESSARIO, RIVOLGERSI AL PROPRIOMEDICO O AL FORNITORE DELL'APPARECCHIATURA.
AVVERTENZA: NON OSTRUIRE L'INGRESSO O LE USCITE DELL'ARIA.
NON FAR CADERE O INSERIRE ALCUN OGGETTO NELLE APERTURE
PRESENTI SUL DISPOSITIVO. IL CONCENTRATORE DI OSSIGENO
POTREBBE SURRISCALDARSI PREGIUDICANDO LE PRESTAZIONI.
112 — 4805-C1 C (Italian)

AVVERTENZA: FUMARE DURANTE L'USO DELL'OSSIGENO
È LA PRIMA CAUSA DI USTIONI E MORTI CORRELATE AGLIINCENDI. È NECESSARIO ATTENERSI ALLE SEGUENTI AVVERTENZE DI SICUREZZA:
AVVERTENZA: NON CONSENTIRE AL PAZIENTE DI FUMARE O DI USARECANDELE O FIAMME LIBERE ALL'INTERNO DELLA STANZA IN CUI SI
TROVA IL DISPOSITIVO OPPURE QUALSIASI ALTRO ACCESSORIO PERL'EROGAZIONE DELL'OSSIGENO.
AVVERTENZA: FUMARE MENTRE SI INDOSSA UNA CANNULA PERL'OSSIGENO PUÒ PROVOCARE USTIONI AL VISO E PROBABILMENTE LAMORTE.
AVVERTENZA: SE SI FUMA, QUESTI 3 PASSAGGI POSSONO SALVARE
LA VITA: SPEGNERE IL CONCENTRATORE DI OSSIGENO, TOGLIERE LACANNULA E LASCIARE LA STANZA IN CUI SI TROVA IL DISPOSITIVO.
AVVERTENZA: IN CASA O NEL LUOGO IN CUI È UTILIZZATO L'OSSIGENO OCCORRE AFFIGGERE CARTELLI "NON FUMARE - OSSIGENO IN
USO" BEN VISIBILI. I PAZIENTI E COLORO CHE PRESTANO ASSISTENZA
DEVONO ESSERE INFORMATI DEI PERICOLI DEL FUMO IN PRESENZA O
DURANTE L'UTILIZZO DI OSSIGENO.
AVVERTENZA: NON UTILIZZARE IL CONCENTRATORE DI OSSIGENOIN PRESENZA DI GAS INFIAMMABILI, POICHÉ POSSONO VERIFICARSIFENOMENI DI COMBUSTIONE RAPIDA TALI DA COMPORTARE DANNIMATERIALI, LESIONI CORPOREE O DECESSO.
AVVERTENZA: RIMUOVENDO LA CANNULA E POSIZIONANDOLA SUINDUMENTI, BIANCHERIA, DIVANI O ALTRI MATERIALI IMBOTTITI SI
POSSONO VERIFICARE LAMPI DI FUOCO IN CASO DI ESPOSIZIONE ASIGARETTE, FONTI DI CALORE O FIAMME.
AVVERTENZA: NON LASCIARE LA CANNULA NASALE SOPRA O SOTTO
INDUMENTI, COPERTE O CUSCINI DI SEDIE. SE L'UNITÀ È ACCESA MA
NON IN USO, L'OSSIGENO PUÒ RENDERE INFIAMMABILE IL MATERIALE. QUANDO IL CONCENTRATORE DI OSSIGENO NON È IN USO, IMPOSTARE IL CONCENTRATORE DI OSSIGENO NELLA POSIZIONE 0 (OFF).
ATTENZIONE: scollegare sempre l'alimentatore CA dallapresa di rete prima di scollegarlo dal concentratore diossigeno.
ATTENZIONE: posizionare sempre il tubo di erogazione dell'ossigeno e i cavi di alimentazione in modo che non vi sia pericolo diinciampare.
ATTENZIONE: quando si utilizza il concentratore di ossigeno in
auto, in barca o con altre sorgenti CC tramite il cavo 12 VCC, accertarsi di avviare e mettere in funzione il veicolo prima di collegare
il concentratore di ossigeno. Se la spia di alimentazione esterna
sul dispositivo eQuinox non si accende ed è necessario ripristinare
il dispositivo, scollegare il cavo 12 VCC dalla presa CC, riavviare
il veicolo e ricollegare il cavo 12 VCC alla presa CC. La mancata
osservanza di tali istruzioni può comportare la mancata erogazionedell'alimentazione al concentratore di ossigeno.
ATTENZIONE: scollegare e rimuovere il concentratore di ossigeno
quando si spegne l'automobile in cui si utilizza l'unità. Non conservare il concentratore di ossigeno in un'auto molto calda o molto
fredda o in altri ambienti simili a temperatura molto alta o bassa.
Nota: le apparecchiature mobili di comunicazione a RF possono avere
effetti sui dispositivi elettromedicali.

Ambienti operativi consigliati
Fare riferimento alla tabella di seguito per un uso corretto deldispositivo. Nella tabella sono riportate informazioni importanti
sulle condizioni d'uso o sugli ambienti operativi consigliati.

SpecificheTemperatura diesercizio
Umidità diesercizio
Temperatura ditrasporto/stoccaggio

Da 0 °C a 40 °C (da 32 °F a 104 °F)

Posizionamento corretto del dispositivo eQuinox
Scegliere una posizione per il dispositivo in modo da evitarel'ingresso di fumo, vapori e agenti inquinanti. Il corretto
posizionamento del dispositivo deve consentire l'ingresso di aria
attraverso il filtro di ingresso dell'aria nella parte superiore sul
retro dell'alloggiamento, nonché l'uscita libera dell'aria di scarico
dallo sfiato di scarico nella parte inferiore destra del dispositivo.

Dal 10% al 95% a un punto di rugiada di 28 °C(82,4 °F)

Posizionare il dispositivo in modo da poter udire gli allarmi.

Da -25 °C a 70 °C (da -13 °F a 158 °F)

Posizionare il tubo di erogazione dell'ossigeno in modo che nonsi annodi e non sia ostruito.

Umidità: fino al 95% senza condensa
Da -20 °C a 50 °C (da -4 °F a 122 °F) per le batterie

Specifiche
elettriche

Non utilizzare prolunghe. Non utilizzare prese
elettriche controllate da un interruttore.

Altitudine

Da 0 a 4.000 metri (da 0 a 13.123 piedi)

Posizionamento NON bloccare l'ingresso dell'aria (copertura superiore
posteriore) o lo sfiato di scarico (parte inferiore destra
posteriore). Posizionare il dispositivo a una distanza dialmeno7,5cm(3pollici)dapareti,tendaggi,mobili,ecc.
Ambiente

Deve essere privo di fumo, agenti inquinanti e vapori.

Durata di esercizio

24 ore al giorno se collegato a una sorgente di alimentazione CA o CC esterna.

Nota: se il concentratore di ossigeno è rimasto per periodi prolungati aldi fuori dell'intervallo normale della temperatura di esercizio, è necessario lasciare che raggiunga la normale temperatura operativa prima diaccenderlo (consultare la sezione Specifiche in questo manuale).

AVVERTENZA: NON UTILIZZARE IL CONCENTRATORE DIOSSIGENO IN PRESENZA DI GAS INFIAMMABILI, POICHÉPOSSONO VERIFICARSI FENOMENI DI COMBUSTIONE RAPIDA TALI DA COMPORTARE DANNI MATERIALI, LESIONI CORPOREE
O DECESSO. NON UTILIZZARE OLIO, GRASSO, PRODOTTI A BASE DIPETROLIO O INFIAMMABILI DI ALTRO TIPO CON ACCESSORI PER IL
TRASPORTO DELL'OSSIGENO O CON IL CONCENTRATORE DI OSSIGENO. UTILIZZARE ESCLUSIVAMENTE LOZIONI O UNGUENTI A BASE
D'ACQUA COMPATIBILI CON L'OSSIGENO. L'OSSIGENO ACCELERA LACOMBUSTIONE DELLE SOSTANZE INFIAMMABILI.
AVVERTENZA: QUESTO DISPOSITIVO EROGA OSSIGENO A ELEVATA CONCENTRAZIONE, CHE FAVORISCE IL RAPIDO SVILUPPO DI
INCENDI. NON CONSENTIRE AL PAZIENTE DI FUMARE O DI USARE
FIAMME LIBERE ALL'INTERNO DELLA STANZA IN CUI SI TROVA (1)QUESTO DISPOSITIVO OPPURE (2) QUALSIASI ALTRO ACCESSORIO
PER L'EROGAZIONE DELL'OSSIGENO. IL MANCATO RISPETTO DI
QUESTA AVVERTENZA PUÒ COMPORTARE GRAVI INCENDI, DANNIMATERIALI E/O LESIONI FISICHE O IL DECESSO.
Nota: scollegare e rimuovere il concentratore di ossigeno quando si
spegnel'automobileincuisiutilizzal'unità.Nonconservareilconcentratore di ossigeno in un'auto molto calda o molto fredda o in altri ambienti
simili a temperatura molto alta o bassa. NON lasciare il concentratore diossigeno e l'alimentatore collegati al veicolo se il sistema di accensione è
spento. Si potrebbe scaricare la batteria del veicolo.
ATTENZIONE: non utilizzare l'unità in spazi confinati oangusti (ad esempio una borsa o una custodia di piccole
dimensioni) in cui la ventilazione possa essere limitata.
Il concentratore di ossigeno potrebbe surriscaldarsi pregiudicando le prestazioni. Non ostruire l'ingresso o le uscite dell'aria. Il
concentratore di ossigeno potrebbe surriscaldarsi pregiudicando leprestazioni.

eQuinox (Model 4000) User Manual — 113

Istruzioni per l'uso
Prima dell'uso

Informazioni generali per il caricamento

Il presente manuale dell'utente è una guida di riferimento perl'uso e la manutenzione del dispositivo. In caso di dubbi odomande, rivolgersi al fornitore di assistenza domiciliare.

Il dispositivo eQuinox può essere alimentato mediante
alimentazione CA esterna, alimentazione CC o batteria
ricaricabile. Il sistema include un "caricabatterie intelligente",
che ricarica la batteria ogni volta che il dispositivo eQuinox
è collegato a un'alimentazione esterna. Il sistema monitora econtrolla la sorgente di alimentazione e il caricabatterie.

Importante! NON tentare di utilizzare eQuinox senza prima
aver letto la sezione Linee guida per la sicurezza del presentemanuale. Seguire tutte le istruzioni per l'uso. Osservare tutte leavvertenze indicate sul dispositivo e nel manuale dell'utente.
Per ridurre il rischio di incendio, lesioni personali e gravi danni
al dispositivo eQuinox, osservare tutte le precauzioni per lasicurezza.
AVVERTENZA: PULIRE L'ALLOGGIAMENTO, IL PANNELLO DI
CONTROLLO E IL CAVO DI ALIMENTAZIONE SOLO CON UN
DETERGENTE DELICATO PER USO DOMESTICO, APPLICANDOLO CON UNA SPUGNA O UN PANNO UMIDO (NON BAGNATO),POI ASCIUGARE TUTTE LE SUPERFICI. NON FAR ENTRARE LIQUIDIALL'INTERNO DEL DISPOSITIVO. PRESTARE PARTICOLARE ATTENZIONE ALL'USCITA DELL'OSSIGENO PER IL COLLEGAMENTO DELLACANNULA E ACCERTARSI CHE RESTI PRIVA DI POLVERE, ACQUA EPARTICELLE.
ATTENZIONE: non ostruire l'ingresso o le uscite dell'aria.
Il concentratore di ossigeno potrebbe surriscaldarsi pregiudicando le prestazioni. Non utilizzare il concentratore
di ossigeno senza il filtro di ingresso dell'aria in posizione. Se è
disponibile un secondo filtro, inserire il filtro "di ricambio" prima dipulire quello sporco. Pulire il filtro sporco con una soluzione tiepida
di acqua e sapone e asciugare a fondo prima dell'uso.

Accensione del dispositivo eQuinox
Il dispositivo può essere alimentato da tre (3) diverse sorgenti dialimentazione:• Alimentatore CA• Cavo 12 VCC• Batteria ricaricabile

Il dispositivo eQuinox carica la batteria durante l'utilizzo
dell'alimentazione CA o CC e quando la temperatura dellabatteria rientra nell'intervallo di temperatura di sicurezza.
Quando l'alimentazione esterna è scollegata, il dispositivo
passa automaticamente alla batteria, se quest'ultima è inserita e
carica. Dopo il collegamento a una presa CA o CC, il dispositivo
viene alimentato dall'alimentazione esterna e la batteria viene
ricaricata.
Allineare la spina di alimentazione al connettore esterno
incassato in modo da collegare adeguatamente l'alimentazione
CA o il cavo 12 VCC. Il connettore di alimentazione incassato
si trova sul lato destro del dispositivo guardando il pannello
di controllo. Inserire la spina di alimentazione nel connettore
ALIMENTATORE CA
incassato.

Alimentatore CA

Il dispositivo eQuinox è dotato di un alimentatore CA universale
per uso domestico o dovunque sia disponibile un'alimentazione
CA standard. Per il collegamento a un'alimentazione CA, inseriresaldamente il cavo dell'alimentatore CA nel connettore esterno
incassato. Assicurarsi inoltre che il cavo dell'alimentatore CA
sia inserito in una presa CA. Quando il dispositivo eQuinox
è collegato correttamente, si accendono un indicatore verde
sull'alimentatore CA e l'indicatore di alimentazione esterna. Anche
sul pannello di controllo utente si accende.
l'indicatore
L'alimentatore CA CAIRE, cod. art. 4836, è la sorgente dialimentazione principale per il concentratore di ossigeno portatile
eQuinox. È una parte molto importante dell'intero sistema.
Nota: utilizzare solo la tensione elettrica specificata sull'etichetta delle
specifiche applicata sul dispositivo.
Fase 1

114 — 4805-C1 C (Italian)

Fase 2

Fase 3

AVVERTENZA: NON TOCCARE CONTEMPORANEAMENTE I
CONTATTI DELLA BATTERIA E IL PAZIENTE DURANTE L'ALIMENTAZIONE DELL'UNITÀ CON L'ALIMENTATORE ESTERNO(ALIMENTATORE CA) E SENZA UNA BATTERIA INSERITA.
ATTENZIONE: utilizzare esclusivamente gli accessori
raccomandati dal produttore. L'uso di altri accessori può
essere pericoloso, causare gravi danni al concentratore diossigeno e rendere nulla la garanzia.
ATTENZIONE: non utilizzare prolunghe con l'unità né collegare
troppe spine alla stessa presa elettrica. L'uso di prolunghe può
avere effetti avversi sulle prestazioni del dispositivo. Il collegamento di troppe spine a un'unica presa può provocare un sovraccaricodel quadro elettrico, causando l'attivazione dell'interruttore/fusibile, o un incendio in caso di mancato funzionamento dell'interruttore/fusibile.
ATTENZIONE: l'uso di cavi e adattatori diversi da quelli specificati, ad eccezione di quelli venduti dal produttore del dispositivo
elettromedicale come parti di ricambio per i componenti interni,può provocare un aumento delle emissioni e una riduzione dell'immunità del concentratore di ossigeno.

Cavo 12 VCC
Cavo 12 VCC

Un cavo 12 VCC consente al sistema di funzionare mediante
prese CC, ad esempio quelle che si trovano nei veicoli a motore.
1. Avviare il veicolo.
2. Inserire il cavo 12 VCC nella presa di alimentazione
incassata sul lato del dispositivo.
3. Inserire la spina del cavo 12 VCC nella presa elettrica CCdel veicolo a motore.
Se il dispositivo è collegato correttamente ed è alimentato da una
sorgente di alimentazione CC, l'indicatore verde di alimentazione
esterna
sul pannello di controllo si accende.

Collegamento del dispositivo eQuinox all'alimentazione CC
Quando il dispositivo eQuinox è collegato all'alimentatore CC in
un veicolo, sono applicabili le seguenti caratteristiche:• Il dispositivo eQuinox utilizzerà tutte le impostazioni di flusso
continuo da 0,5 a 3,0 lpm e tutte le impostazioni di impulso da
1 a 9 mentre è alimentato dall'alimentazione CC di un veicolo.
• La batteria sarà ricaricata solo se rimane energia sufficiente
dopo il funzionamento del dispositivo alla portata prescritta.
Il dispositivo eQuinox riserverà l'alimentazione fornita
prima al funzionamento dell'unità. L'alimentazione residua
sarà utilizzata per ricaricare la batteria. Se la batteria deldispositivo eQuinox è in carica, il simbolo della batteria
lampeggia a cascata. Se la batteria del dispositivo eQuinox
non è in carica, il simbolo della batteria è fisso.
AVVERTENZA: NON OSTRUIRE L'INGRESSO O LE USCITE
DELL'ARIA. IL CONCENTRATORE DI OSSIGENO POTREBBESURRISCALDARSI PREGIUDICANDO LE PRESTAZIONI.
AVVERTENZA: SE LA SORGENTE DI ALIMENTAZIONE CC DEL VEICOLOSCENDE AL DI SOTTO DI 11,5 V, EQUINOX RITORNA AL FUNZIONAMENTO A BATTERIA (SE LA BATTERIA È PRESENTE E CARICA).
ATTENZIONE: scollegare e rimuovere il concentratore diossigeno quando si spegne l'automobile in cui si utilizza
l'unità. Non conservare il concentratore di ossigeno in
un'auto molto calda o molto fredda o in altri ambienti simili atemperatura molto alta o bassa. NON lasciare il concentratore diossigeno e l'alimentatore collegati al veicolo se il sistema di accensione è spento. Si potrebbe scaricare la batteria del veicolo.
Nota: durante l'alimentazione del dispositivo mediante un cavo 12 VCC, labatteria potrebbe non essere caricata.
Nota: durante l'alimentazione del dispositivo eQuinox mediante un cavo
12 VCC, si consiglia di utilizzare la modalità Dosaggio impulsi se è statoprescritto di utilizzare un dispositivo di dosaggio impulsi.
Nota:perassicurarsicheildispositivostiautilizzandol'alimentazioneCC,verificarechel'indicatoredialimentazioneesternasiaacceso.Unsegnaleacustico
potrebbeindicareunaperditadell'alimentazioneesterna.ControllareseilLED
verdedell'indicatoredialimentazioneesternaèaccesoospento.Larimozionedella batteria durante l'alimentazione CC assicura che non vi sia alcuna
perdita di carica mentre l'unità è in funzione. Se l'indicatore non è acceso, labatteria sta alimentando il dispositivo.
Nota:leprestazionieffettivedeldispositivoeQuinoxpossonovariareinbase
allecondizionidiutilizzodelveicoloamotore.Alcuniveicolipotrebberonon
essereingradodigarantirelefunzionalitàcompletedurantel'alimentazione
CC. Qualora il veicolo non possa fornire alimentazione sufficiente per il funzionamentoconl'impostazionecorrente,ildispositivoeQuinoxinterromperà
perprimacosalaricaricadellabatteriaecontinueràafunzionareallaportata
prescritta. Se l'alimentazione fornita dal veicolo è ancora insufficiente, il
dispositivo eQuinox ridurrà la portata se è pari o superiore a 2,0 lpm. Se laportata è inferiore a 2,0 lpm, il dispositivo eQuinox verrà spento.
Sesiriscontranoportatelimitate,rimuoveretuttiglialtridispositivielettronici
collegatialcircuitodialimentazione(GPS,caricabatteriepercellulare,ecc.).
Sel'alimentazionerisultaorasufficiente,leportatenonvengonoautomaticamenteaumentatefinoall'impostazioneoriginale;saràl'utilizzatoreadovereseguirelaregolazionemanuale.SeildispositivoeQuinoxancoranonfunziona
alla portata prescritta, provare a rimuovere la batteria.
eQuinox (Model 4000) User Manual — 115

Batteria ricaricabile

Nota:ilfunzionamentodelconcentratorediossigenoaldifuoridell'intervallo normale di temperatura di esercizio è in grado di incidere sulle prestazioniediridurreiltempodifunzionamentodellabatteriae/odiaumentarne
il tempo di ricarica (consultare la sezione Specifiche in questo manuale).
È possibile ottenere un tempo di funzionamento a batteria superiore se si
utilizzaeQuinoxnellamodalitàImpulso(modalitàapprovatadalmedico).

Il dispositivo eQuinox può essere alimentato dalla
batteria ricaricabile fornita con il dispositivo.
Inserimento della batteria: allineare labatteria in modo da farla scorrere nel vanodi alimentazione vuoto. Spingere la batteria
nel vano vuoto fino a bloccarla in posizione,
allineandola alla parte anteriore del dispositivo.
Quando la batteria è correttamente inserita e il
dispositivo eQuinox è acceso, viene visualizzato
l'indicatore di stato della batteria sul pannello
di controllo. Se NON è inserita una batteria,
l'indicatore di stato della batteria non si accende
e l'unità continua a funzionare mediante
alimentazione esterna.
Per rimuovere la batteria, premere il pulsantedi sblocco nella parte superiore della batteria.
Tenendo premuto il pulsante di sblocco,
estrarre la batteria dal dispositivo eQuinox.

AVVERTENZA: PER I CONCENTRATORI DI OSSIGENO DOTATIDI BATTERIE: CONSERVARE IN UN LUOGO FRESCO E ASCIUTTO PER CONTRIBUIRE A GARANTIRE LA LONGEVITÀ DELLABATTERIA. LA CONSERVAZIONE DEL CONCENTRATORE DI OSSIGENOPER PERIODI PROLUNGATI A TEMPERATURE ELEVATE O CON LA BATTERIA COMPLETAMENTE CARICA O COMPLETAMENTE SCARICA PUÒ
RIDURRE LA DURATA COMPLESSIVA DELLA BATTERIA. NON TENTAREDI APRIRE LA BATTERIA, CHE NON CONTIENE COMPONENTI RIPARABILI. TENERE LE BATTERIE LONTANO DAI BAMBINI.

Batteria ricaricabile Carica iniziale della batteria

La nuova batteria fornita con il dispositivo eQuinox non ècompletamente carica quando viene spedita dalla fabbrica. Primadi utilizzare il dispositivo eQuinox per la prima volta, è necessario
caricare completamente la batteria.

Tempi di funzionamento a batteria: il tempo di funzionamento abatteria è indicato sul pannello di controllo sia con un simbolo
sia in formato alfanumerico. Durante l'alimentazione a batteria,
l'indicatore di stato della batteria è sempre visibile. Lo schermodel display viene aggiornato per mostrare la durata rimanentedella batteria in ore e minuti. Diversi fattori influiscono sui tempi
di funzionamento, tra cui l'impostazione di flusso, la modalità
Impulso o Flusso continuo e la frequenza respiratoria. Nella
tabella Tempi di funzionamento tipici di una batteria nuova sono
riportate le stime dei tempi di funzionamento del dispositivo
eQuinox con una batteria nuova, completamente carica, in basealle impostazioni di flusso e alle condizioni operative indicate.
Nota: il dispositivo eQuinox è fornito dalla fabbrica con una batteria a
12 celle. La batteria a 24 celle è un accessorio opzionale che consente
una durata della batteria superiore.

Collegare l'alimentatore CA e inserire correttamente la batteria.
Attendere la completa ricarica della batteria. La batteria ècompletamente carica quando l'indicatore di stato della batteria sul
pannello di controllo non lampeggia più a cascata e rimane fisso.
Il dispositivo eQuinox può essere utilizzato durante la ricaricadella batteria. La ricarica può richiedere fino a 4 ore, in baseall'impostazione di flusso. La batteria viene caricata anche quando
il dispositivo eQuinox è collegato all'alimentazione CA e spento.
ATTENZIONE: utilizzare solo batterie fornite dal produttore. Per il corretto smaltimento della batteria, rivolgersi
al fornitore dell'apparecchiatura o all'ente governativo
preposto per i requisiti di smaltimento.

* Le batterie sono riciclabili. *Tempi di funzionamento tipici di una batteria nuova
Tipo di funzionamento

Tipo di batteria
12 celle

24 celle

Modalità Flusso
continuo
1 lpm

2,16 ore (2 ore, 9 minuti) 4,84 ore (4 ore, 50 minuti)

2 lpm

1,2 ore (1 ora, 12 minuti) 2,75 ore (2 ore, 45 minuti)

3 lpm

0,8 ore (48 minuti)

1,83 ore (1 ora, 49 minuti)

Modalità Impulso(12 bpm)
16 ml (1,0)

2,62 ore (2 ore, 37 minuti) 5,94 ore (5 ore, 56 minuti)

48 ml (3,0)

2,38 ore (2 ore, 22 minuti) 5,39 ore (5 ore, 23 minuti)

96 ml (6,0)

1,70 ore (1 ora, 41 minuti) 3,85 ore (3 ore, 51 minuti)

116 — 4805-C1 C (Italian)

In caso di interruzione dell'alimentazione CA o CC, il dispositivo
eQuinox emette un segnale acustico e passa automaticamente
al funzionamento a batteria, se quest'ultima è inserita e carica.
Quando si ritorna all'alimentazione esterna, la ricarica dellabatteria viene ripresa automaticamente. Durante l'interruzionedell'alimentazione CA, se la batteria non è presente o ècompletamente scarica, il dispositivo eQuinox si spegne
emettendo un allarme. La batteria è completamente scarica
quando l'indicatore di stato della batteria sul pannello di controllo
appare vuoto. Il dispositivo mostra una spia gialla ed emette
un allarme di batteria scarica. Questa condizione prosegue finoallo spegnimento del dispositivo con una spia rossa e un allarmecontinuo. L'avviso acustico può durare fino a 5 minuti finché ildispositivo non si spegne o non viene ripristinata l'alimentazione
esterna.
Nota:persilenziarel'allarmediperditadialimentazione,tenerepremuto
il pulsante ON/OFF del dispositivo eQuinox per almeno 5 secondi. È
possibile eliminare l'allarme anche collegando il dispositivo eQuinox a
un alimentatore CA esterno per 2 minuti.

Tempo di ricarica tipico della batteria

Abilitazione/disabilitazione del modulo vocale

Di solito sono necessarie da 2 a 5 ore per ricaricare una
batteria completamente scarica a una capacità dell'80%, inbase all'impostazione di flusso utilizzata. Quando la batteria
è completamente carica, il relativo indicatore smette dilampeggiare a cascata. È possibile caricare la batteria all'internodell'unità quando il dispositivo eQuinox è spento e collegato.

1. Per abilitare/disabilitare la funzione del modulo vocale,accendere l'unità premendo il tasto ON/Standby.
2. Dopo aver acceso l'unità, premere una volta il tasto
nascosto con l'icona Vietato fumare.
3. Entro 1 o 2 secondi, viene visualizzata la voce di menu
SPKR = ON o SPKR = OFF.

Se la batteria diventa troppo calda durante la fase discaricamento, la ricarica non viene avviata finché la batteria
non è sufficientemente fredda. Per accelerare il processo diraffreddamento è possibile rimuovere la batteria.

4. Premere il tasto per aumentare o ridurre l'impostazione
di flusso per abilitare (ON) o disabilitare (OFF) il modulo
vocale.

La batteria può essere ricaricata da sorgenti di alimentazione CAe CC. La batteria può inoltre essere caricata con il caricatore da
scrivania facoltativo.

Nota: questa operazione consente di disabilitare una sola volta la funzione del modulo vocale. Dopo aver spento e reimpostato il dispositivo, lafunzione del modulo vocale viene abilitata automaticamente.

5. Entro 3-5 secondi, l'opzione selezionata diventa attiva.

Aumento/riduzione del volume delle notifiche locali
1. Per aumentare o ridurre il volume delle notifiche locali,premere due volte il tasto nascosto con l'icona Vietato fumare.

Funzionamento del modulo vocale

3. Premere il tasto per aumentare o ridurre l'impostazione diflusso per impostare il volume su un livello da 1 a 10.

Istruzioni per l'uso del modulo vocale
Il dispositivo eQuinox può fornire verbalmente notifiche oallarmi e informazioni sullo stato operativo. Queste notificheverbali integrano le informazioni trasmesse dal display a LED.
Di seguito sono disponibili le istruzioni su come accendere
e spegnere il modulo vocale, regolare il volume e cambiare
la lingua.

Tasto ON/
Standby

Indicatore
ON/Standby(LED verde)

2. Viene visualizzato Volume = # - (menu di impostazione delvolume).

Indicatore di allarme(LED rosso)
Indicatore di avviso
Tasto per(LED giallo)
aumentarel'impostazione
di flusso

4. Dopo aver selezionato il volume desiderato, l'opzione viene
attivata e impostata entro 3-5 secondi.
Modifica della lingua del modulo vocale
1. Per cambiare la lingua del modulo vocale, premere cinquevolte il tasto nascosto con l'icona Vietato fumare per accedere
alle opzioni della lingua.
2. La lingua attiva viene indicata sul display. Ad esempio, se
il modulo vocale è impostato sull'inglese all'accesso al menudelle opzioni della lingua, sul display sarà visualizzato il testo"English".
3. Premere il tasto per aumentare o ridurre l'impostazione diflusso per scorrere le lingue disponibili.
4. Dopo aver visualizzato la lingua desiderata, l'opzione viene
impostata automaticamente entro 3-5 secondi.

Indicatore dialimentazione esternapresente (LEDverde)

Tasto nascosto con
icona Vietato
fumareTasto dellamodalità diflussoIndicatore dellamodalità di flusso(LED verde)

Display
LCD

Tasto per ridurre
l'impostazione diflussoeQuinox (Model 4000) User Manual — 117

Primo utilizzo del dispositivo eQuinox
Fase 1: posizionamento del dispositivo eQuinox
per l'uso
Posizionare il dispositivo eQuinox in un'area ben ventilata e
ben illuminata. Assicurarsi che
Filtro di ingresso dell'aria
l'ingresso dell'aria e gli sfiati discarico sul retro non siano ostruiti.
Posizionare il dispositivo eQuinox
in modo che tutti gli indicatori o
gli allarmi acustici e visivi possano
essere visti e uditi facilmente.
Prima di utilizzare il dispositivo
eQuinox, assicurarsi che il filtro diingresso dell'aria sia in posizione.
Se il filtro di ingresso dell'aria nonè presente, rivolgersi al fornitore
di assistenza domiciliare. Se il
filtro di ingresso dell'aria è sporco,
lavarlo con acqua tiepida e sapone.
Risciacquarlo con acqua pulita e
Sfiato di scarico
aspettare che si asciughi prima diriposizionarlo nel dispositivo. Se necessario, sostituirlo con un
filtro di ingresso dell'aria nuovo e pulito.
Collegare il dispositivo a una presa di rete CA, a una sorgente dialimentazione CC oppure assicurarsi che sia inserita una batteria
completamente carica.

Fase 2: accensione e riscaldamento del dispositivo
Tenere premuto il pulsante "ON/OFF" per due (2) secondi peraccendere il dispositivo eQuinox. Viene eseguito un breve test
POST visivo.
Nota: l'emissione di tre segnali acustici quando il dispositivo viene perla prima volta collegato all'alimentazione o acceso indica che la batteria
da 9 V presente all'interno dell'alloggiamento del dispositivo eQuinox
è scarica. Rivolgersi al fornitore di assistenza domiciliare e richiedere lasostituzione della batteria da 9 V. Se i tre segnali acustici proseguono
in maniera intermittente dopo che l'unità è stata accesa ed è rimasta infunzione per qualche minuto, il problema non riguarda la batteria da 9V.
Rivolgersi al fornitore di assistenza sanitaria per la riparazione.

Il dispositivo eQuinox dispone di un indicatore di stato dellaconcentrazione di ossigeno (OCSI, Oxygen Concentration Status
Indicator) integrato nel dispositivo. L'indicatore OCSI monitora
continuamente l'erogazione di ossigeno del dispositivo e utilizza
una spia verde (normale), una spia gialla (avviso), una spia
rossa (allarme) e un allarme acustico. Questi indicatori sono tutti
accesi durante il test POST all'avvio del dispositivo. Una volta
acceso il dispositivo, attendere fino a cinque (5) minuti affinché
il dispositivo possa raggiungere le specifiche prestazionali.
Quando è accesa solo la spia dell'indicatore verde, il dispositivo haraggiunto le specifiche prestazionali.
118 — 4805-C1 C (Italian)

Fase 3: collegamento del tubo di erogazionedell'ossigeno o della cannula nasale all'uscita
dell'ossigeno
Sostituire regolarmente la cannula e il tubo di erogazionedell'ossigeno, come raccomandato dal fornitore di assistenza
domiciliare. Consultare il fornitore di assistenza domiciliare perinformazioni sulla pulizia e sulla sostituzione della cannula e deltubo di erogazione.
Nota: accertarsi che la cannula sia completamente inserita e salda. In
questo modo il concentratore di ossigeno è in grado di rilevare correttamentel'inspirazioneperl'erogazionedell'ossigeno.Durantel'inalazione,
si deve udire o avvertire il flusso di ossigeno attraverso la biforcazionedella cannula nasale.

Fase 4: selezione della modalità di flusso
Premere il pulsante della modalità di flusso per scegliere lamodalità. Premendo più volte questo pulsante è possibile passare
dalla modalità Dosaggio impulsi alla modalità Flusso continuo eviceversa. Le modalità e le impostazioni corrette devono essere
prescritte dal medico per ciascun paziente.
Funzionamento in modalità Flusso continuo:
Durante il funzionamento in modalità Flusso continuo,
un'erogazione costante di ossigeno passa attraverso il tubo e lacannula nasale. L'ossigeno è misurato in litri al minuto (lpm).
È possibile utilizzare un'umidificatore solo nella modalità Flusso
continuo. NON inclinare, trasportare o distendere il dispositivo
eQuinox quando è collegato a un flacone umidificatore.
Funzionamento in modalità Dosaggio impulsi:
Durante il funzionamento in modalità Impulso, un bolo diossigeno, misurato in millilitri (ml), viene erogato al momentodell'inspirazione. NON utilizzare un umidificatore nella modalità
Dosaggio impulsi.
Durante il funzionamento nella modalità Dosaggio impulsi, il
dispositivo eQuinox eroga un flusso continuo di ossigeno ogni
15 secondi quando non viene rilevato un respiro. Il dispositivo
interrompe il flusso continuo e verifica costantemente lapresenza di sforzi inspiratori ogni 15 secondi. Una volta rilevata
un'inspirazione, il dispositivo riprende a erogare il bolo misurato.
Per uscire dalla modalità Dosaggio impulsi e attivare la modalità
Flusso continuo, premere il pulsante della modalità di flusso.
Occorre chiedere al medico di prescrivere il dosaggio impulsi
corretto per ricevere la saturazione di ossigeno adeguata, tenendo
conto delle proprie esigenze a riposo, durante l'esercizio fisico e
per i viaggi ad alta quota. Rivolgersi al fornitore di assistenza perla prescrizione e i moduli di imbarco.

Modalità Dosaggio impulsi e funzionalità autoSAT
Il dispositivo eQuinox è dotato di una funzionalità chiamata
autoSAT®, che consente di erogare un bolo di ossigeno didimensioni costanti per un massimo di 40 respiri al minuto (vederela tabella di seguito per la frequenza respiratoria massima).
All'aumentare della frequenza respiratoria, ad esempio mentre
si deambula, la funzionalità autoSAT controlla il dispositivo
erogando il bolo di dimensione impostata. La tecnologia autoSAT è
di proprietà di CAIRE ed è utilizzata solo nella modalità Impulso.

Volume bolo (+/- 15%)

Impostazione Dosaggio impulsi
1
2
3
4
5
6
7
8
9

(± 15%) ml
16
32
48
64
80
96
128
160
192

Utilizzando i pulsanti di impostazione di flusso, regolare
l'impostazione di flusso al livello prescritto dal medico.
AVVERTENZA: È MOLTO IMPORTANTE SELEZIONARE SOLOIL LIVELLO DI OSSIGENO PRESCRITTO. NON MODIFICARELA SELEZIONE DEL FLUSSO SE NON IN CASO DI PRECISEISTRUZIONI DA PARTE DI UN MEDICO AUTORIZZATO.

Fase 6: uso iniziale del dispositivo eQuinox

Frequenza respiratoria massima
Dimensionebolo

Fase 5: Regolazione dell'impostazione di flusso
al livello prescritto

Frequenzarespiratoria
massima
40
40
40
40
37
31
23
18
15

Nota: il volume del bolo diminuisce quando la frequenza respiratoria
supera il valore massimo indicato.
Nota:l'impostazionedidosaggioimpulsi1-9noncorrispondealvalorelpm.
AVVERTENZA: LE IMPOSTAZIONI DELLA MODALITÀ DOSAGGIO IMPULSI DEVONO ESSERE DEFINITE PER CIASCUN PAZIENTE INDIVIDUALMENTE, IN FUNZIONE DELLE RELATIVEESIGENZE A RIPOSO, DURANTE L'ESERCIZIO FISICO E DURANTE I
VIAGGI.

Respirare normalmente attraverso la cannula nasale.
Nota: è possibile ottenere un tempo di funzionamento a batteria superiore se si utilizza eQuinox nella modalità Impulso (modalità approvata dal
medico).

Fase 7: uso del carrello eQuinox
Il carrello eQuinox è progettato per l'uso sulle superfici dideambulazione.
Allineare il dispositivo eQuinox al pavimento del carrello inmodo che risulti in posizione piana e verticale.
Allineare la vite filettata grande alla guarnizione sul retro deldispositivo eQuinox. Infine, serrare a mano la manopola sul
carrello finché il carrello eQuinox non risulta saldamente fissato
al dispositivo.
Premere il pulsante sulla maniglia del carrello eQuinox per
regolarne l'altezza.

AVVERTENZA: IL DISPOSITIVO EQUINOX POTREBBE NON ESSERE IN
GRADO DI RILEVARE TUTTI GLI SFORZI INSPIRATORI NELLA MODALITÀIMPULSO. SE IL DISPOSITIVO EQUINOX NON RISPONDE AGLI SFORZI DIINSPIRAZIONE, RICHIEDERE AL FORNITORE DI ASSISTENZA DOMICILIARE DI CONTROLLARE L'IMPOSTAZIONE DELLA SENSIBILITÀ.
Nota: per il fornitore dell'apparecchiatura: per l'uso con il concentratore
di ossigeno sono raccomandati i seguenti accessori di somministrazionedell'ossigeno:• Cannula nasale con tubicino da 2,1 m (7 piedi): cod. art. 5408-SEQ.
• Per eventuali altri accessori consigliati, consultare il catalogo
degli accessori (cod. art. ML-LOX0010) disponibile all'indirizzo
www.CAIREmedical.com.
• CAIRE offre OxySafe™ come accessorio facoltativo. Tale accessorio
deveessereutilizzatounitamentealconcentratoreeQuinox.Periclienti
ubicati in zone ove sia richiesta la conformità alla normativa EN ISO
8359:1996-Rettifica1:2012,ilpresenteaccessoriosoddisfataleesigenza.
	OxySafe™ è un fusibile termico che serve a interrompere il flusso di gas
nel caso in cui il tubo della cannula a valle o quello dell'ossigeno prenda
fuoco e bruci in direzione di OxySafe. È posizionato in linea con la cannula nasale o il tubo dell'ossigeno, tra il paziente e l'uscita dell'ossigenodel dispositivo eQuinox.
Per un uso corretto di OxySafe, attenersi sempre alle istruzioni delproduttore (incluse con ogni kit OxySafe).
CAIRE propone kit OxySafe™ comprendenti cuffia della cannula, OxySafe™ e tubi da 2,1 m (7 piedi): cod. art. 20628667.

Nota: NON utilizzare il dispositivo eQuinox e non deambulare mentre ècollegatounumidificatore.Rimuovereilflaconedell'umidificatoreprima
dideambulare.NONdistendereildispositivoeQuinoxquandoècollegato
a un flacone umidificatore.

Fase 8: spegnimento del dispositivo
Tenere premuto il pulsante "ON/OFF" per due (2) secondi per
spegnere il dispositivo eQuinox.
Conservare il dispositivo eQuinox in un luogo fresco e asciutto
per un periodo non superiore a 1 mese. Per garantire il corretto
funzionamento del dispositivo eQuinox, è necessario utilizzarlo
per almeno 2 ore al mese.
eQuinox (Model 4000) User Manual — 119

Tabella di risoluzione dei problemi dell'utilizzatore
Non ignorare gli allarmi
AVVERTENZA: IL PRODUTTORE CONSIGLIA DI TENERE A DISPOSIZIONE UN'ALTRA SORGENTE DI OSSIGENO SUPPLEMENTARE IN CASODI INTERRUZIONI DI CORRENTE, CONDIZIONI DI ALLARME O GUASTI MECCANICI. PER CONOSCERE IL TIPO DI SISTEMA DI RISERVA
NECESSARIO, RIVOLGERSI AL PROPRIO MEDICO O AL FORNITORE DELL'APPARECCHIATURA.

Spia gialla
SintomoPossibile causa

La spia gialla è accesa fissa. Il dispositivo eQuinox ha
L'allarme emette 1 segnale rilevato un problema con ilacustico ogni 30 secondi. collegamento alla batteria.

La spia gialla lampeggia.
L'icona della batteria
La batteria del dispositivo
lampeggia.L'allarmeemetteeQuinox è calda.
1 segnale acustico ogni
2 minuti.

Azione1) Rimuovere la batteria e reinserirla per assicurarsi che sia posizionata saldamente nel
concentratore.
2) Se possibile, inserire una batteria separata nel dispositivo eQuinox.
3) Se il problema persiste, rivolgersi al fornitore di assistenza sanitaria.
1)RicollegareildispositivoeQuinoxall'alimentazioneesternaoppureinserireunabatteria completamente carica. Lasciare raffreddare la batteria calda fuori dal concentratore
per 30 minuti.
2) Ricaricare la batteria mediante alimentazione CA.
3) Se è disponibile un'altra batteria, sostituirla.
4) Se la condizione persiste, rivolgersi al fornitore di assistenza sanitaria.

1) Collegare il dispositivo eQuinox all'alimentatore CA e ricaricare o sostituire la batteria
La spia gialla lampeggia.
con una completamente carica.
L'icona della batteria
La tensione della batteria dellampeggia.L'allarmeemettedispositivo eQuinox è bassa.
2 segnali acustici ogni
30 secondi.
1) Pulire e sostituire il filtro di ingresso dell'alloggiamento.
Il dispositivo eQuinox ha rileLa spia gialla lampeggia.
vato una restrizione di flusso
L'allarme emette 1 segnale
che ha causato un flusso basso
acustico ogni 2 minuti.
o bloccato.

2) Assicurarsi che la cannula non sia piegata o bloccata. Se si utilizza un flacone umidificatore, assicurarsi che sia riempito correttamente e non crei un blocco.
3)AssicurarsicheildispositivoeQuinoxdispongadiunaventilazioneadeguata.Ènecessario che sia almeno a 8 cm (3 pollici) da qualsiasi superficie per garantire che gli sfiati
non siano bloccati.
4) Se il problema persiste, passare a una sorgente di ossigeno alternativa e rivolgersi al
fornitore di assistenza sanitaria.
1) Assicurarsi che il filtro di ingresso dell'aria non sia ostruito o ristretto. Pulire e sostituire il filtro, se necessario.

2) Assicurarsi che il dispositivo eQuinox sia in un'area ben ventilata. Assicurarsi chevi siano almeno 8 cm (3 pollici) tra il retro del dispositivo eQuinox e qualsiasi tipo di
La spia gialla lampeggia.
Il dispositivo eQuinox ha rile- ostacolo (accessori, tende, ecc.).
Non vi sono allarmi acustici. vato bassi livelli di ossigeno.
3) Se utilizzato in auto, assicurarsi che il retro del dispositivo eQuinox sia rivolto verso
l'esterno nel sedile.
4) Se la condizione persiste, passare a una sorgente di ossigeno alternativa e rivolgersi
immediatamente al fornitore di assistenza sanitaria.

120 — 4805-C1 C (Italian)Spia rossa
SintomoPossibile causa

Azione1) Assicurarsi che il filtro di ingresso dell'aria non sia ostruito o ristretto. Pulire e sostituire
il filtro, se necessario.

2) Assicurarsi che il dispositivo eQuinox sia in un'area ben ventilata. Assicurarsi che vi siano almeno 8 cm (3 pollici) tra il retro del dispositivo eQuinox e qualsiasi tipo di ostacolo
La spia rossa lampeggia.
IldispositivoeQuinoxharileva- (accessori, tende, ecc.).
L'allarmeemette3segnato bassi livelli di ossigeno.
li acustici ogni 2 minuti.
3) Se utilizzato in auto, assicurarsi che il retro del dispositivo eQuinox sia rivolto verso
l'esterno nel sedile.
4) Se la condizione persiste, passare a una sorgente di ossigeno alternativa e rivolgersi
immediatamente al fornitore di assistenza sanitaria.
1) Controllare l'alimentatore esterno (CA o CC) e verificare che i collegamenti al dispositivo eQuinox, al trasformatore e alla presa elettrica esterna siano ben saldi.

Durante il funzionamento
con alimentazione esterna, il 2) Se possibile, provare a utilizzare un'altra presa elettrica (CA o CC) per alimentare il
La spia rossa lampegdispositivo eQuinox ha perso dispositivo eQuinox.
gia. L'allarme emette
temporaneamente potenza. 3) Se la condizione persiste, passare a una sorgente di ossigeno alternativa e rivolgersi
un segnale acustico
immediatamente al fornitore di assistenza sanitaria.
costante. Sul display LCD
1)RicollegareildispositivoeQuinoxall'alimentazioneesternaoppureinserireunabatteria
è visualizzato ********.
completamente carica. Lasciare raffreddare la batteria calda fuori dal concentratore per
IldispositivoeQuinoxnon
eroga ossigeno e non si Il dispositivo eQuinox ha perso 30 minuti.
accende.
potenza a causa di una batteria 2) Ricaricare la batteria mediante alimentazione CA.
scarica o surriscaldata.
3) Se è disponibile un'altra batteria, sostituirla.
4) Se la condizione persiste, rivolgersi al fornitore di assistenza sanitaria.
1) Rimuovere la batteria e scollegare l'alimentatore CA o CC dal dispositivo eQuinox. La
spia rossa e il messaggio di errore non dovrebbero più essere visualizzati quando le sorgentidialimentazionevengonorimosse.Attenderecirca20secondi,ricollegarelesorgenti
di alimentazione e provare a riaccendere il dispositivo eQuinox.

La spia rossa è accesa fissa. Sullo schermo è visuIl dispositivo eQuinox haalizzato "FAIL" (ERRORE).
rilevatounmalfunzionamento 2) Se il dispositivo eQuinox si accende, verificare che il messaggio di errore non venga viIldispositivoeQuinoxnon
del sistema.
sualizzatonuovamente.Seilmessaggiodierrorevienevisualizzatonuovamente,rivolgersi
eroga ossigeno e non si
immediatamente al fornitore di assistenza sanitaria.
accende.
3) Se il dispositivo eQuinox non si riaccende, rivolgersi immediatamente al fornitore di
assistenza sanitaria.

eQuinox (Model 4000) User Manual — 121

Altre condizioni di allarme
SintomoPossibile causa

Nella modalità
Impulso, il LED
della modalità
Impulso lampeggia
rapidamente per
15 secondi.

Il dispositivo eQuinox
non è in grado di rilevare lo sforzo di respirazione dell'utente.
Per informazioni sulla
modalità Impulso, vedere a pagina 15.

La batteria si scarica
quandoildispositivo
è collegato all'alimentatore CA/CC.

Lacaricadellabatteria non dura quanto
dovrebbe.

La batteria non si
carica.

Azione1)ÈpossibilechelamodalitàImpulsononfunzionicorrettamentesesussisteunaqualsiasicondizioneche
causi il blocco o la restrizione dei passaggi nasali (malattia, setto deviato, ecc.).
2) Assicurarsi che la cannula non superi 2 metri circa (7 piedi) di lunghezza.
3) Assicurarsi che il tubo della cannula non sia piegato o bloccato.
4) Se è collegato un flacone umidificatore, rimuoverlo e collegare la cannula direttamente all'uscita. La
modalità Impulso non funziona se è installato un flacone umidificatore.
5) Se la condizione persiste, rivolgersi al fornitore di assistenza sanitaria.
1)Verificarechelapresaeroghil'alimentazione.Noncollegarel'alimentatoreauncircuitodimmeroauna
ciabatta.
2)Verificarecheicavideglialimentatorisianosaldamentecollegatiallaparete/veicoloealconcentratore.
Sesiutilizzal'alimentatoreCA,verificareancheilcollegamentodeicavisultrasformatoredialimentazione.
Il dispositivo eQuinox 3) Assicurarsi che le spie di verifica dell'alimentazione siano accese.
non riceve alimentazi- È presente una spia verde sull'alimentatore.
one dall'alimentatore- È presente una spia verde a forma di cavo di alimentazione sul pannello di controllo del concentratore
collegato.
eQuinox.
4)SeiLEDdiverificadell'alimentazionenonsonoaccesi,rimuoveretuttiicollegamentidell'alimentatore
per 20 secondi, quindi ricollegarli.
5) Se la condizione persiste, rivolgersi al fornitore di assistenza sanitaria.
La batteria non è stata 1) Collegare il dispositivo all'alimentazione CA per ricaricare la batteria.Verificare che la batteria si carichi
per 2-5 ore e che l'icona corrispondente sia piena e non lampeggi prima dell'uso.
caricata completamente.
2) Fare riferimento alle azioni per il sintomo "La batteria non si carica" riportate di seguito.
Il dispositivo eQuinox 1)Assicurarsidiutilizzarelaportataprescrittadalmedicoecheilflussocontinuooaimpulsivengautilizzanon funzionava alla
to come richiesto.
portata prevista.
1) Provare a far scaricare completamente la batteria. Effettuare questa operazione alimentando il dispositivo eQuinox a batteria finché non si spegne completamente. Rimuovere la batteria e lasciare che siLabatterianonfunzio- raffreddipercirca30minuti.Dopoquestoperiodo,reinserirelabatterianeldispositivoeQuinoxericaricarlanaconformementealle mediante alimentazione CA.
specifiche.
2) Se dopo aver seguito le istruzioni della fase 1 la batteria continua a non durare quanto dovrebbe, rivolgersi al fornitore di assistenza sanitaria.
1)Verificarechelapresaeroghil'alimentazione.Noncollegarel'alimentatoreauncircuitodimmeroauna
ciabatta.
2)Verificarecheicavideglialimentatorisianosaldamentecollegatiallaparete/veicoloealconcentratore.
Sesiutilizzal'alimentatoreCA,verificareancheilcollegamentodeicavisultrasformatoredialimentazione.
Il dispositivo eQuinox 3) Assicurarsi che le spie di verifica dell'alimentazione siano accese.
non riceve alimentazi- È presente una spia verde sull'alimentatore.
one CA esterna per- È presente una spia verde a forma di cavo di alimentazione sul pannello di controllo del concentratore
caricare la batteria.
eQuinox.
4)SeiLEDdiverificadell'alimentazionenonsonoaccesi,rimuoveretuttiicollegamentidell'alimentatore
per 20 secondi, quindi ricollegarli.
5) Se la condizione persiste, rivolgersi al fornitore di assistenza sanitaria.
1) Rimuovere la batteria e reinserirla per assicurarsi che sia posizionata saldamente nel concentratore.

2)ControllarechenonsiapresenteunaspiagiallafissasuldispositivoeQuinoxquandolabatteriaèinseriIl dispositivo eQuinox
ta oppure che non sia presente un'icona della batteria lampeggiante quando si tenta di caricarla.
nonècollegatocorretSe è presente uno di questi sintomi, procedere alla fase 3.
tamente alla batteria
3) Se possibile, inserire una batteria separata nel dispositivo eQuinox. Se la batteria di ricambio funziona
per la ricarica.
correttamente, è necessario sostituire quella originale.
4) Se il problema persiste, rivolgersi al fornitore di assistenza sanitaria.
1) Contattare il fornitore di assistenza sanitaria.
Il dispositivo
eQuinox emette un Il livello di carica dellasegnale acustico
batteria interna che
3voltequandoviene alimenta gli allarmi èacceso per la prima basso.
volta.
122 — 4805-C1 C (Italian)Pulizia, cura e manutenzione ordinaria
Manutenzione ordinaria

Filtro di ingresso dell'aria

Gli interventi sui componenti interni del dispositivo eQuinox
devono essere eseguiti solo da un tecnico dell'assistenzaqualificato e addestrato da CAIRE.

L'aria ambientale viene aspirata nel dispositivo attraverso il
condotto di ingresso dell'aria situato nella parte posteriore
superiore del dispositivo. La pulizia del filtro di ingresso dell'aria
è l'attività di manutenzione più importante che l'utente deve
effettuare. La pulizia del filtro di ingresso dell'aria consente dipreservare il corretto funzionamento del dispositivo eQuinox.
Controllare il filtro di ingresso dell'aria quotidianamente epulirlo almeno una volta a settimana utilizzando la seguente
procedura:

La manutenzione preventiva deve essere eseguita ogni
anno. Consultare il fornitore per definire un programma dimanutenzione preventiva. Il fornitore di assistenza domiciliare oil tecnico qualificato controlleranno i seguenti componenti solo
quando necessario: allarmi, batteria interna da 9 V, batteria, filtri
interni e altri componenti interni.
Nota: se vengono emessi tre segnali acustici dopo il test POST, significa
che la batteria da 9 V è quasi scarica. Rivolgersi al fornitore di assistenza
domiciliare se si sentono tre segnali acustici.

Cura e pulizia del dispositivo da parte dell'utilizzatore
Sostituzione della cannula
Sostituire la cannula e il tubo di erogazione con la frequenza
raccomandata dal fornitore di assistenza domiciliare. Il medicoo il fornitore di assistenza domiciliare fornirà le informazioni
riguardanti la pulizia, la disinfezione e la sostituzione.
Nota: per l'utilizzo adeguato, attenersi sempre alle istruzioni del produttore della cannula. Sostituire la cannula monouso come consigliato dal
produttoredellacannulaodalfornitoredell'apparecchiatura.Ilfornitore
dell'apparecchiatura è in grado di fornire ulteriori pezzi di ricambio.

1. Rimuovere il filtro di ingresso dell'aria
2. Lavare il filtro in acqua di rubinetto tiepida utilizzando una
soluzione detergente con sapone delicato
3. Sciacquare accuratamente il filtro con acqua di rubinetto
pulita ed eliminare l'acqua in eccesso
4. Far asciugare il filtro all'aria
5. Reinserire il filtro nell'alloggiamento
Suggerimento di pulizia: tenere a portata di mano un secondo
filtro pulito da utilizzare in sostituzione a quello in fase diasciugatura.

Filtro diingressodell'aria

ATTENZIONE: non utilizzare il concentratore di ossigeno
senza il filtro di ingresso dell'aria in posizione. Se è disponibile un secondo filtro, inserire il filtro "di ricambio"
prima di pulire quello sporco. Pulire il filtro sporco con una soluzione tiepida di acqua e sapone e asciugare a fondo prima dell'uso.
Nota: il produttore non raccomanda la sterilizzazione di questa apparecchiatura.
Nota: non utilizzare il concentratore di ossigeno senza il filtro di ingressodell'aria in posizione. Se è disponibile un secondo filtro, inserire il filtro"di ricambio" prima di pulire quello sporco. Pulire il filtro sporco con una
soluzione tiepida di acqua e sapone e asciugare a fondo prima dell'uso.

Il filtro di ingresso dell'aria deve essere sostituito con
cadenza annuale. Se il dispositivo eQuinox viene utilizzato
in un ambiente polveroso, è necessario sostituire il filtro più
frequentemente. Non utilizzare il dispositivo eQuinox per più di
5 minuti senza un filtro di ingresso dell'aria installato.

eQuinox (Model 4000) User Manual — 123

Alimentatore CA, cavo 12 VCC, cavo di alimentazione e custodia esterna:
Spegnere il dispositivo eQuinox e scollegarlo dall'alimentazione
CA o CC prima di qualsiasi attività di pulizia o disinfezione. NON
spruzzare direttamente sul telaio esterno. Utilizzare una spugna
o un panno umido (non bagnato). Per pulire l'alloggiamento el'alimentatore, spruzzare una soluzione detergente delicata sul
panno o sulla spugna. Per disinfettare il dispositivo eQuinox,
utilizzare il disinfettante Lysol® Brand II (o equivalente).
Attenersi alle indicazioni del produttore del detergente.
AVVERTENZA: PERICOLO DI SCOSSA ELETTRICA. SPEGNERE L'UNITÀ E SCOLLEGARE IL CAVO DI ALIMENTAZIONEDALLA PRESA ELETTRICA PRIMA DI PULIRE L'UNITÀ PER
PREVENIRE IL PERICOLO DI SCOSSE ELETTRICHE E USTIONI ACCIDENTALI. GLI INTERVENTI DI RIPARAZIONE E LA RIMOZIONE DEI
COPERCHI DEVONO ESSERE ESEGUITI ESCLUSIVAMENTE DAL FORNITORE DELL'APPARECCHIATURA O DA UN TECNICO DI ASSISTENZAQUALIFICATO.NON UTILIZZARE LIQUIDI DIRETTAMENTE SULL'UNITÀ.
NELL'ELENCO DELLE SOSTANZE CHIMICHE NON DESIDERATE SONOCOMPRESI, IN VIA NON LIMITATIVA: ALCOL E PRODOTTI A BASE
ALCOLICA, PRODOTTI CONCENTRATI A BASE DI CLORO (CLORURODI ETILENE) E PRODOTTI A BASE OLEOSA (PINE-SOL®, LESTOIL®).
QUESTE SOSTANZE NON VANNO UTILIZZATE PER PULIRE L'ALLOGGIAMENTO DI PLASTICA SUL CONCENTRATORE DI OSSIGENO, POICHÉ POSSONO DANNEGGIARE LE PARTI IN PLASTICA DELL'UNITÀ.
AVVERTENZA: PERICOLO DI SCOSSA ELETTRICA. SCOLLEGARE IL
CAVO DI ALIMENTAZIONE DALLA PRESA ELETTRICA PRIMA DI PULIRE L'UNITÀ PER PREVENIRE IL PERICOLO DI SCOSSE ELETTRICHEE USTIONI ACCIDENTALI. GLI INTERVENTI DI RIPARAZIONE E LA
RIMOZIONE DEI COPERCHI DEVONO ESSERE ESEGUITI ESCLUSIVAMENTE DAL FORNITORE DELL'APPARECCHIATURA O DA UN TECNICODI ASSISTENZA QUALIFICATO.
AVVERTENZA: EVITARE CON ATTENZIONE CHE IL CONCENTRATOREDI OSSIGENO E LA BATTERIA SI BAGNINO E CHE I FLUIDI ENTRINO
NELL'UNITÀ. POTREBBERO VERIFICARSI PROBLEMI DI FUNZIONAMENTO O ARRESTI, NONCHÉ UN AUMENTO DEL RISCHIO DI USTIONIO SCOSSE ELETTRICHE.
AVVERTENZA: PULIRE L'ALLOGGIAMENTO, IL PANNELLO DI CONTROLLO E IL CAVO DI ALIMENTAZIONE SOLO CON UN DETERGENTE
DELICATO PER USO DOMESTICO, APPLICANDOLO CON UNA SPUGNAO UN PANNO UMIDO (NON BAGNATO), POI ASCIUGARE TUTTE LE
SUPERFICI. NON FAR ENTRARE LIQUIDI ALL'INTERNO DEL DISPOSITIVO. PRESTARE PARTICOLARE ATTENZIONE ALL'USCITA DELL'OSSIGENO PER IL COLLEGAMENTO DELLA CANNULA E ACCERTARSI CHERESTI PRIVO DI POLVERE, ACQUA E PARTICELLE.
AVVERTENZA:NON UTILIZZARE LIQUIDI DIRETTAMENTE SULL'UNITÀ.
NELL'ELENCO DELLE SOSTANZE CHIMICHE NON DESIDERATE SONO COMPRESI, IN VIA NON LIMITATIVA: ALCOL E
PRODOTTI A BASE ALCOLICA, PRODOTTI CONCENTRATI A
BASE DI CLORO (CLORURO DI ETILENE) E PRODOTTI A BASE OLEOSA(PINE-SOL®, LESTOIL®). QUESTE SOSTANZE NON VANNO UTILIZZATE PER PULIRE L'ALLOGGIAMENTO DI PLASTICA SUL CONCENTRATORE DI OSSIGENO, POICHÉ POSSONO DANNEGGIARE LE PARTI IN
PLASTICA DELL'UNITÀ.
124 — 4805-C1 C (Italian)

Umidificatore opzionale
Se utilizzato, pulirlo quotidianamente per ridurre la possibilità
di contaminazione. Attenersi alle raccomandazioni di pulizia
del fornitore di assistenza domiciliare o del produttoredell'umidificatore. Sostituire l'umidificatore mensilmente ocome consigliato dal produttore o dal fornitore di assistenza
domiciliare (non riempire eccessivamente l'umidificatore).
ATTENZIONE: NON utilizzare il dispositivo eQuinox e nondeambulare mentre è collegato un umidificatore. Rimuovere il flacone dell'umidificatore prima di deambulare.
NON distendere il dispositivo eQuinox quando è collegato a un
flacone umidificatore.

Carrello eQuinox
Prima di pulire il carrello eQuinox, rimuoverlo dal dispositivo
eQuinox. Utilizzare un panno o una spugna umidi (non bagnati)
e una soluzione detergente delicata per pulire il carrello.
Batteria
La batteria del dispositivo eQuinox richiede una cura particolare
per garantire una maggiore durata e il livello di prestazioni
massimo. La batteria CAIRE è l'unica batteria raccomandata perl'uso con il dispositivo eQuinox.
Utilizzare una spugna o un panno umido (non bagnato) per
pulire la batteria. Spruzzare prima un detergente delicato sul
panno o sulla spugna, quindi pulire l'alloggiamento della batteria
e il fermo.
CAIRE consiglia di sostituire la batteria dopo 2 anni.
Smaltimento della batteria
La batteria è ricaricabile e può essere riciclata. Restituire sempre
la batteria al fornitore di assistenza domiciliare per il corretto
smaltimento. È inoltre possibile rivolgersi agli uffici comunali
preposti per ottenere informazioni sul corretto smaltimento dellabatteria.
Smaltimento del sistema per ossigeno eQuinox
Le leggi locali a tutela dell'ambiente possono vietare losmaltimento delle apparecchiature elettriche e/o elettroniche, ad
esempio il dispositivo eQuinox o l'alimentatore CA. Rivolgersi
agli uffici comunali preposti per ottenere informazioni sul
corretto smaltimento delle apparecchiature elettriche oelettroniche. In alternativa è possibile rivolgersi a CAIRE, Inc.
per ottenere informazioni sullo smaltimento.

Tabelle CEM
Linee guida e dichiarazione del produttore - Emissioni elettromagnetiche
Il dispositivo eQuinox è destinato all'uso nell'ambiente elettromagnetico descritto di seguito. Il cliente o l'utilizzatore del dispositivo
eQuinox deve garantirne l'uso all'interno di un ambiente di questo tipo.
Test delle emissioni
Emissioni RF
CISPR 11
Emissioni RF
CISPR 11

Conformità

Ambiente elettromagnetico - Linee guida

Gruppo 1

IldispositivoeQuinoxusaenergiainradiofrequenzasoloperlefunzioniinterne.
Pertanto, le sue emissioni RF sono molto basse ed è improbabile che possanocausare interferenze nelle apparecchiature elettroniche nelle vicinanze.

Classe B

Emissioni armoniche
IEC 61000-3-2Classe A

Fluttuazioniditensione/emissioni di sfarfallii IEC 610003-3

Conforme

Il dispositivo eQuinox è idoneo all'uso in tutti gli edifici, compresi quelli ad uso
abitativo e quelli collegati direttamente alla rete pubblica di alimentazione abassa tensione utilizzata per rifornire gli edifici ad uso abitativo.

eQuinox (Model 4000) User Manual — 125

Linee guida e dichiarazione del produttore - Immunità elettromagnetica
Il dispositivo eQuinox è destinato all'uso nell'ambiente elettromagnetico descritto di seguito. Il cliente o l'utilizzatore del dispositivo
eQuinox deve garantirne l'uso all'interno di un ambiente di questo tipo.
Test dell'immunità

Livello di test IEC 60601

Livello di conformità

Ambiente elettromagnetico - Lineeguida±6 kV a contatto±6 kV a contatto±8 kV nell'aria±8 kV nell'aria

Transitori elettrici
veloci/burst±2 kV per linee di alimentazione±2 kV per linee di alimentazione

IEC 61000-4-4±1 kV per linee di ingresso/uscita

IEC 61000-4-2

Sovratensione tran- ±1 kV da linea a linea
sitoria
IEC 61000-4-5

Calidi tensione,brevi
interruzioni e variazioni della tensione
di alimentazione
sullelineediingressodell'alimentazione
IEC 61000-4-11Campo magneticoalla frequenza direte (50/60 Hz)
IEC 61000-4-8

N/D±1 kV da linea a linea±2 kV da linea a terra<5% UT<5% UT(>95% di calo in UT )(>95% di calo in UT )per 0,5 cicli

per 0,5 cicli

40% UT

40% UT(60% di calo in UT )(60% di calo in UT )per 5 cicli

per 5 cicli

70% UT

70% UT(30% di calo in UT )(30% di calo in UT )per 25 cicli

per 25 cicli<5% UT<5% UT(>95% di calo in UT )(>95% di calo in UT )per 5 secondi

per 5 secondi

3 A/m

3 A/m

NOTA: UT indica la tensione di rete CA prima dell'applicazione del livello di test.

126 — 4805-C1 C (Italian)Ambiente elettromagnetico - Lineeguida
Il pavimento deve essere di legno, cementoo piastrelle in ceramica. Se i pavimenti sono
ricoperti di materiale sintetico, l'umidità
relativa deve essere almeno del 30%.
La qualità dell'alimentazione di rete deveesserequellatipicadegliambienticommerciali e/o ospedalieri.
La qualità dell'alimentazione di rete deveesserequellatipicadegliambienticommerciali e/o ospedalieri.

La qualità dell'alimentazione di rete deveessere quella tipica degli ambienti commerciali e/o ospedalieri. Se l'utilizzatoredel dispositivo eQuinox richiede continuità
di funzionamento durante le interruzioni
dell'alimentazione elettrica, è consigliabile
alimentare il dispositivo eQuinox con un
gruppo di continuità o una batteria.

I campi magnetici alla frequenza di retedevono essere ai livelli tipici degli ambienti
commerciali e/o ospedalieri.

Linee guida e dichiarazione del produttore - Immunità elettromagnetica
IldispositivoeQuinoxèdestinatoall'usonell'ambienteelettromagneticodescrittodiseguito.Ilclienteol'utilizzatoredeldispositivoeQuinox deve garantirne l'uso all'interno di un ambiente di questo tipo.
Test dell'immunità

Livello di test IEC 60601

Livello di conformità

Ambiente elettromagnetico - Linee guida

RF condotta

3 Vrms3 VrmsIEC 61000-4-6

Da 150 kHz a 80 MHz

RF irradiata

3 V/m

Le apparecchiature di comunicazione a RF mobili
e portatili devono essere utilizzate a una distanzadaqualsiasicomponentedeldispositivoeQuinox,
compresi i cavi, superiore a quella calcolata inbase all'equazione applicabile alla frequenza deltrasmettitore.

IEC 61000-4-3

Da 80 MHz a 2,5 GHz

3 V/mDistanza di separazione consigliata

=1,2
=1,2
=1,2

= Da 80 MHz a 800 MHz= Da 800 MHz a 2,5 GHz

dove P è la potenza di uscita nominale massima
del trasmettitore in watt (W) secondo le indicazioni del produttore e d è la distanza di separazione
consigliata in metri (m).
Le intensità di campo dai trasmettitori RF fissi,
determinatemedianteun'indagineelettromagnetica nella sedea, devono essere inferiori al livello diconformità in ogni intervallo di frequenzab.
Le interferenze possono verificarsi nei pressidelle apparecchiature contrassegnate da questo
simbolo:

NOTA 1: a 80 MHz e 800 MHz, si applica l'intervallo di frequenza superiore.
NOTA 2: queste linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica è influenzata
dall'assorbimento e dalla riflessione di strutture, oggetti e persone.
Le intensità di campo da trasmettitori fissi, quali stazioni base per telefoni via radio (cellulari e cordless) e radio mobili terrestri,radioamatoriali,trasmissioniradiofonicheAM/FMetrasmissionitelevisive,nonpossonoessereprevisteconaccuratezzaalivelloteorico.Per
valutarel'ambienteelettromagneticoprodottodatrasmettitoriRFfissi,èopportunoeseguireun'indagineelettromagneticadellasede.Se
l'intensitàdicampomisuratanelpuntoincuièutilizzatoildispositivoeQuinoxsuperaillivelloapplicabilediconformitàRFsopraindicato,è
opportunoappurarecheildispositivoeQuinoxfunzionicorrettamente.Qualorafosseriscontratounfunzionamentofuoridallanorma,potrebbe essere necessario adottare ulteriori provvedimenti, ad esempio cambiando l'orientamento o la posizione del dispositivo eQuinox.

ab

Oltre l'intervallo di frequenza da 150 kHz a 80 MHz, le intensità di campo devono essere inferiori a 3 V/m.

eQuinox (Model 4000) User Manual — 127

Distanze di separazione consigliate
tra le apparecchiature di comunicazione RF mobili e portatili e il dispositivo eQuinox
Il dispositivo eQuinox è destinato all'uso in un ambiente elettromagnetico in cui i disturbi da RF irradiata sono tenuti sotto controllo.
Ilclienteol'utilizzatoredeldispositivoeQuinoxpuòprevenireleinterferenzeelettromagnetichemantenendounadistanzaminimatrale
apparecchiature di comunicazione RF portatili e mobili (trasmettitori) e il dispositivo eQuinox come indicato di seguito, calcolando tale
distanza in base alla potenza massima di uscita dell'apparecchiatura di comunicazione.
Potenza di uscita
nominale massima deltrasmettitore
W
0,01
0,1

Distanza di separazione in base alla frequenza del trasmettitorem
Da 150 kHz a 80 MHz

Da 80 MHz a 800 MHz

Da 800 MHz a 2,5 GHz

=1,2

=1,2

=1,2

0,12
0,38

0,12

0,23

0,38

0,73

1

1,2

1,2

2,3

10

3,8

3,8

7,3

100

12

12

23

Per i trasmettitori con una potenza di uscita massima non elencata sopra, la distanza di separazione consigliata d in metri (m) può essere
stimata utilizzando l'equazione applicabile alla frequenza del trasmettitore, dove P è la potenza di uscita massima del trasmettitore in
watt (W) ottenuta in base alle informazioni fornite dal produttore.
NOTA 1: a 80 MHz e 800 MHz, si applica la distanza di separazione per l'intervallo di frequenza superiore.
NOTA 2: queste linee guida potrebbero non essere applicabili in tutte le situazioni. La propagazione elettromagnetica è influenzata
dall'assorbimento e dalla riflessione di strutture, oggetti e persone.

128 — 4805-C1 C (Italian)

Informazioni generali sui viaggi in aereo Batteria e per viaggi aerei
Se non utilizzato durante il rullaggio, il decollo e l'atterraggio, il
dispositivo eQuinox deve essere spento e riposto sotto il sedile oin un altro luogo di stivaggio approvato, in modo da non bloccare
il passaggio o l'ingresso nelle file.
Se il dispositivo eQuinox viene utilizzato durante il rullaggio, il
decollo e l'atterraggio, l'utente deve trovarsi in un posto a sedere
che non limiti l'accesso a qualsiasi altro passeggero, l'uso diqualsiasi uscita normale o di emergenza richiesta o il passaggio
nei corridoi dell'abitacolo dell'aereo.
Agli utilizzatori del dispositivo eQuinox non è consentito sedersi
nelle file corrispondenti alle uscite di emergenza.
Se si utilizza il dispositivo eQuinox quando si verifica ladecompressione della cabina e viene distribuito ossigeno
mediante il sistema della cabina di ossigeno, è necessariointerrompere l'uso di eQuinox e utilizzare l'ossigeno
supplementare dell'aereo. Dopo essersi assicurati della presenzadell'ossigeno supplementare dell'aereo, spegnere l'unità eQuinox.
Prima del viaggio, è necessario che l'utente ispezioni ildispositivo eQuinox per assicurarsi che sia in buono stato difunzionamento.

Tenere conto del tempo a terra (check-in
prima del volo, check-in di sicurezza, ecc.)
e degli scali.
La maggior parte delle compagnie aeree richiede di disporre
di un numero adeguato di batterie completamente cariche
per alimentare il dispositivo per almeno il 150% della durata
massima del viaggio prevista (ore di volo, tempo complessivo aterra, prima e dopo il volo e durante i collegamenti, nonché perritardi imprevisti).

Si consiglia di portare sempre con sé una batteria
aggiuntiva durante i viaggi.
Ciascuna compagnia aerea dispone di propri requisiti riguardo alla possibilità di viaggiare con ossigeno.
Contattare la compagnia aerea almeno 48 ore prima dellapartenza per informarla che si intende viaggiare con il
dispositivo eQuinox.

ATTENZIONE: il Department of Transportation (DOT) degli
Stati Uniti e i regolamenti delle Nazioni Unite richiedonola rimozione della batteria dal dispositivo per l'intero
viaggio aereo internazionale quando il concentratore di ossigeno
viene registrato come bagaglio. Durante la spedizione del concentratore di ossigeno, è necessario rimuovere la batteria dal dispositivo e imballarlo in maniera adeguata.

eQuinox (Model 4000) User Manual — 129

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107 USA
1-800-482-2473
Nel mondo: www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Germania
Nel mondo: www.CAIREmedical.com

CAIRE Inc. si riserva il diritto di interrompere la produzione o di modificare prezzi, materiali, apparecchiature, qualità,
descrizioni,specifichee/oprocessideipropriprodottiinqualsiasimomento,senzapreavvisoesenzaulterioriobblighio conseguenze.Tutti i diritti non espressamente indicati nel presente documento sono riservati, laddove applicabili.

130 — 4805-C1 C (Italian)Alla data della presente pubblicazione, eQuinox, SeQual, autoSAT e i relativi loghi(come applicabile) sono marchi registrati di CAIRE Inc.
Lysol® è un marchio registrato di Reckitt Benckiser, Regno Unito.
OxySafe® è un marchio registrato di Applied Home Healthcare Equipment.
Pine-Sol® e Lestoil® sono marchi registrati di The Clorox Company.

Manual de Instruções

eQuinoxTM

Modelo 4000
eQuinox (Model 4000) User Manual — 131

Controlos do Utilizador e Indicadores de Estado do Sistema
Símbolo

A

02

IP22

Definição

Símbolo

Definição

Ler o manual do utilizador antes de utilizar. Consultar as
instruções no manual do utilizador.

Símbolo de Aprovação da FAA: *** A Administração
Federal de Aviação Norte-Americana (FAA) aprovou este
aparelho para uso a bordo de aviões comerciais.

Ícone de Proibido Fumar: Não fumar perto da unidade.

Indicador da Configuração de Fluxo

Avisos / Indicador de ALERTA (Amarelo)

Operação no Modo Pulsátil

Indicador de ALARME (Vermelho)­

Dispositivo em funcionamento normal; botão ligar/desligar

Indicador de Alimentação Externa

Aumentar Configuração de Fluxo

Indicador de Estado da Unidade de alimentação (bateria)

Diminuir Configuração de Fluxo

Símbolo da Unidade da Bateria

Não utilizar lubrificante ou gordura.

Amperes

Chamas proibidas.

Corrente alternada

Não existem peças reparáveis no interior. Não abrira cobertura.

Ícone IEC para Alimentação CC (entrada)

Superfície Quente.

Saída de Oxigénio

Não molhar.

Peça aplicada tipo BF (nível de proteção contra
choque elétrico)Equipamento IPX1 à prova de gotejamento: A Unidadede Alimentação do eQuinox (Bateria) e a Fonte deAlimentação CA proporcionam proteção contra os efeitos
prejudiciais da penetração de líquidos. (IPX1, conforme
IEC 60529)

IPX1

Símbolo de Reciclagem

Nome e endereço do fabricante

Este símbolo destina-se a lembrar os proprietários do
equipamento que devem devolvê-lo a uma instalação
de reciclagem no final da sua vida útil, de acordo com a
Diretiva relativa aos Resíduos de Equipamentos Elétricose Eletrónicos (WEEE).

Representante autorizado na Comunidade Europeia

À prova de gotejamento e de penetração de sólidos: O
eQuinox fornece proteção contra os efeitos nocivos dapenetração de objetos estranhos, líquidos e sólidos, em
conformidade com a IEC 60529 na classificação de IP22.

Equipamento de Classe II

Certificado para os mercados norte-americano e
canadiano, de acordo com os padrões norte-americanos e
canadianos aplicáveis.

Este dispositivo está em conformidade com osrequisitos da Diretiva 93/42/CEE relativamente
aos dispositivos médicos. Possui a Marcação CE,
conforme indicado. Dit apparaat voldoet aan de eisen vanrichtlijn 93/42/EEG betreffende medische hulpmiddelen.
Daarom draagt het de CE-markering zoals afgebeeld.

132 — 4805-C1 C (Portuguese)Índice
Controlos do Utilizador e Indicadores de Estado do Sistema.................................................................................... 132
Guia de iniciação rápida........................................................................................................................................................... 134
Indicações de Utilização.......................................................................................................................................................... 137
Contraindicações da utilização............................................................................................................................................. 137
Diretrizes de Segurança........................................................................................................................................................... 137
Ambientes de Operação Recomendados.......................................................................................................................... 139
Instruções de utilização........................................................................................................................................................... 140
Utilização do módulo de voz................................................................................................................................................. 143
Utilização do eQuinox pela primeira vez........................................................................................................................... 144
Quadro de resolução de problemas do utilizador ........................................................................................................ 146
Limpeza, Conservação e Manutenção de Rotina........................................................................................................... 149
Quadros EMC............................................................................................................................................................................... 151
Informações Gerais das Companhias Aéreas................................................................................................................... 155
Unidade da Bateria e a Viagens em Companhias Aéreas............................................................................................. 155

eQuinox (Model 4000) User Manual — 133

Guia de iniciação rápida

1

Desembalar o eQuinox

2Conhecer o seu eQuinox

Rever todos os Avisos, Cuidados e informações do dispositivo
adicionais no resto deste manual.
Familiarize-se com as principais características do eQuinox
e com o Painel de Controlo do Utilizador.
Porta de saídade oxigénio

FRENTE
Pega

Painel decontrolo
Recetáculo da
alimentação
externaTransformador de CA
Carrinho de transporte
do eQuinoxÍcone deAprovação
da FAA

Trinco da
unidadeda bateria

Local da Etiquetade Classificação
e N.º de Série

Unidade dabateria

TRASEIRA

Cabo 12VDC

Caixa demanutenção

Saída deExaustão

Unidade dealimentação
(bateria)

134 — 4805-C1 C (Portuguese)

Filtro deadmissãode ar

O Painel de Controlo do Utilizador
eQuinox apresenta importantes
informações de operação.

Indicador do Botão LIGAR/DESLIGAR (verde): Este botão
LIGA ou DESLIGA o aparelho. O Indicador Verde acende
quandooaparelhoestáLIGADOefuncionacorretamente.
Botões para Aumentar ou Diminuir a Configuração deFluxo: Utilize estes botões para ajustar o fluxo para a
configuração prescrita.
Botão e Indicador do Modo de Administração: O botão
alterna entre o Modo de Fluxo Contínuo e o Modo deDosagem Pulsátil. O Modo de Dosagem Pulsátil ativa a
TecnologiaautoSAT.Istopermiteumaumentosignificativodoperíododefuncionamentoquandoestáaserusadaa Unidade de alimentação (bateria). Quando o Modo deDosagemPulsátiléativado,oIndicadorverdedeModode
DosagemPulsátilacende-se.Éadministradoumpulsode
oxigénio e o indicador verde pisca com cada esforço deinspiração.
IndicadordeALERTA(Amarelo)—Quandoestáligado,indica
umacondiçãodebaixaprioridadeouCuidado.Continuea
usar o sistema. Consulte o Quadro de Resolução de problemas para se informar sobre as medidas corretas a tomar,
oucontacteoseuPrestadordecuidadosdomiciliários.Um
amarelointermitenteindicaumalertadeprioridademédia.
Énecessáriaumarespostarápida.(Geralmente,nívelbaixo
de oxigénio ou bateria fraca.)

3

Ligar o eQuinox

Para ligar a uma fonte de alimentação CA: Introduza firmementeo cabo de alimentação CA no recetáculo da alimentação externa,no lado direito do dispositivo. Ligue o cabo de alimentação CA
ao transformador e a uma tomada de CA.
Passo 1

Passo 2

Passo 3

Quando o eQuinox está corretamente ligado, acende-se um
indicador verde no transformador de CA. O Indicador dealimentação externa no painel de controlo do utilizador também
se acende.
Instalar a Bateria: Insira a bateria no compartimento vazio até
encaixar no lugar. Ficará nivelada com a parte frontal do aparelho.

Indicador de ALARME (Vermelho)­—Indica uma situação
de alarme de alta prioridade. É necessária uma resposta imediata. Consulte a o Quadro de Resolução deproblemas e entre em contacto com o seu prestador decuidados domiciliários. (Geralmente perda de potência,
baixo nível de oxigénio ou anomalia.)
Visor de informação.
• C ou P – "C" indica modo de Fluxo Contínuo e "P" indica
modo de Dosagem Pulsátil.
• Tempo em bateria – Quando a alimentação da bateria
está a ser usada, o visor efetua um ciclo a cada 20 segundospara indicar a duração prevista da carga da bateria•Configuração(fluxocontínuooudosagempulsátil)–cadavezqueLIGAodispositivo,oúltimomodoe/oudefiniçãofoi
guardado e é usado no arranque.

Quando a bateria está corretamente instalada e o eQuinox está
LIGADO ou conectado a uma fonte de alimentação CA, surgeo Indicador de estado da unidade da bateria.

Indicadordeestadodaunidadedealimentação(bateria):
Este indicador mostra a carga restante da bateria. Cada
uma das cinco barras horizontais cinzentas representa
aproximadamente20%dacargatotaldabateria.Quando
a bateria está a carregar, as barras indicadoras de carga
ficamintermitentesnumaespéciedecascata.Seabateria
não estiver instalada ou se tiver sido incorretamente
instalada, o Indicador de Estado da Unidade de Alimentação (bateria) não acende ou fica intermitente.
Indicador de Alimentação Externa: Se o eQuinox estiver
corretamente ligado e utilizar a Fonte de alimentação CA
ou o Cabo 12VDC, este indicador acende-se no Painel deControlo do Utilizador.
Sinalsonoro:Umalarmesonoroalerta-oacercadacondiçãodefuncionamento do dispositivo, um aviso ou falha, e confirma que foi premida uma
tecla válida.
Módulo de voz: Se for ativado, um indicador de voz alerta-o acerca dacondiçãodefuncionamentododispositivo,alteraçõesnafontedealimentação, advertências ou falhas.

Indicador de Botão LIGAR/DESLIGAR (verde):
Prima e mantenha premido o botão "LIGAR/DESLIGAR" durante
dois (2) segundos para LIGAR o eQuinox. Ocorre um breve sinal
sonoro e visual no autoteste. Verifique se todos os indicadoresestão acesos e se o sinal sonoro soa durante três (3) segundos.
Prima e mantenha premido o botão "LIGAR/DESLIGAR" durante
dois (2) segundos para DESLIGAR o eQuinox.
eQuinox (Model 4000) User Manual — 135

4

Selecionar o modo de administração dofluxoBotão e Indicador do Modo de Fluxo

Prima o botão de Modo de Fluxo para selecionar o modo
desejado. Ao premir este botão repetidamente, alterna entre os
modos de fluxo contínuo e de dosagem pulsátil. Em Modo defluxo contínuo, o oxigénio é fornecido num caudal constanteentre 0,5 e 3,0 LPM, indicado pela letra C. Em Modo dedosagem pulsátil, o oxigénio é fornecido num bolus no iníciode cada inalação. As configurações do modo pulsátil 1–9
fornecem bolos com dimensões entre os 16mL e os 192mL esão designadas pela letra P no visor. A dosagem pulsátil podeser ajustada para sensibilidade e administração de bolo por
respiração. As configurações de fluxo de ambos os modos devem
ser prescritas pelo seu médico.

5

Ajuste a Configuração de Fluxo Para oNível PrescritoBotão para aumentar ou diminuir a configuração
de fluxo:
Utilizando os botões + ou -, ajuste a configuração paraa prescrição fornecida pelo seu médico.

Visualização do modo de fluxo contínuo

Visualização do modo de dosagem pulsátil

AVISO: É MUITO IMPORTANTE SELECIONAR APENAS ONÍVEL DE OXIGÉNIO PRESCRITO. NÃO ALTERE A SELEÇÃO
DO FLUXO, A MENOS QUE TAL LHE TENHA SIDO INDICADOPOR UM MÉDICO AUTORIZADO. O CONCENTRADOR DE OXIGÉNIOPORTÁTIL PODE SER USADO DURANTE O SONO QUANDO TAL FOR
RECOMENDADO POR UM MÉDICO QUALIFICADO.
136 — 4805-C1 C (Portuguese)

6

Começar a usar o eQuinox

Filtro de Admissão de Ar

Filtro deadmissão
de ar

A sua unidade está equipada com Filtro de admissão de ar (filtro
extra fornecido). Verificar diariamente e limpar uma vez por
semana. As instruções de limpeza encontram-se na página 19.
Ligue o tubo de fornecimento de oxigénio à saída de oxigénio
e ligue a cânula de oxigénio de acordo com as instruções
fornecidas com a cânula, ou com as instruções do seu prestador
de cuidados domiciliários.

Respire normalmente através da cânula nasal.
Quando é usada a alimentação da bateria, a utilização do
Modo de dosagem pulsátil resulta numa duração de utilização
mais longa.

7

Utilização do carrinho de transporte
do eQuinox

Alinhe o eQuinox com a forma da base do carrinho de modo aficar nivelado e na vertical. Alinhe o parafuso roscado grande
com a entrada na parte traseira do eQuinox. Em seguida, aperte
manualmente o manípulo do carrinho até este ficar bem fixo ao
aparelho. Prima o botão na pega do carrinho telescópico e puxe
para cima para ajustar a altura da pega.

Importante!
As instruções de segurança estão definidas da
seguinte forma:
AVISO: INFORMAÇÕES IMPORTANTES DE SEGURANÇA
PARA PERIGOS QUE PODEM CAUSAR FERIMENTOSGRAVES.
CUIDADO: Informações importantes para evitar danos noeQuinox.
Nota: Informações que necessitam de atenção especial.

Indicações de Utilização
O eQuinox destina-se à administração de exigénio suplementar.
O dispositivo não se destina a suporte de vida, nem fornece capacidades de monitorização de pacientes.
AVISO: É MUITO IMPORTANTE SELECIONAR APENAS ONÍVEL DE OXIGÉNIO PRESCRITO. NÃO ALTERE A SELEÇÃO
DO FLUXO, A MENOS QUE TAL LHE TENHA SIDO INDICADOPOR UM MÉDICO AUTORIZADO. O CONCENTRADOR DE OXIGÉNIOPORTÁTIL PODE SER USADO DURANTE O SONO QUANDO TAL FOR
RECOMENDADO POR UM MÉDICO QUALIFICADO.
AVISO: A LEI FEDERAL DOS E.U.A. LIMITA A VENDA OU ALUGUER
DESTE DISPOSITIVO A PEDIDO DE UM MÉDICO OU DE OUTROPRESTADOR DE CUIDADOS DE SAÚDE AUTORIZADO.
AVISO: ESTE UNIDADE NÃO SE DESTINA A SER USADA COMOEQUIPAMENTO DE SUPORTE À VIDA. OS DOENTES GERIÁTRICOS,PEDIÁTRICOS OU OUTROS DOENTES INCAPAZES DE COMUNICARDESCONFORTO QUANDO USAM ESTA UNIDADE PODEM REQUERER
MONITORIZAÇÃO ADICIONAL. OS DOENTES COM DEFICIÊNCIA(S)
AUDITIVAS E/OU VISUAIS PODEM NECESSITAR DE AUXÍLIO PARAMONITORIZAR OS ALARMES.

Contraindicações da utilizaçãoAVISO: EM DETERMINADAS CIRCUNSTÂNCIAS, A UTILIZAÇÃO DE OXIGÉNIO SEM PRESCRIÇÃO MÉDICA PODESER PERIGOSA. ESTE APARELHO SÓ DEVE SER UTILIZADOQUANDO PRESCRITO POR UM MÉDICO.
AVISO: NÃO DEVE SER UTILIZADO NA PRESENÇA DE ANESTÉSICOSINFLAMÁVEIS.
AVISO: TAL COMO ACONTECE COM QUALQUER OUTRO APARELHOALIMENTADO A ENERGIA ELÉTRICA, O UTILIZADOR PODE EXPERIENCIAR PERÍODOS DE NÃO FUNCIONAMENTO COMO RESULTADODE INTERRUPÇÕES DA ENERGIA ELÉTRICA OU NECESSITAR QUE OEQUINOX SEJA REPARADO POR UM TÉCNICO QUALIFICADO. O EQUINOX NÃO É ADEQUADO PARA PACIENTES QUE POSSAM SOFRERCONSEQUÊNCIAS ADVERSAS PARA A SAÚDE COMO RESULTADO DE
TAIS INTERRUPÇÕES TEMPORÁRIAS.

Diretrizes de Segurança
AVISO: NÃO É PERMITIDA QUALQUER MODIFICAÇÃO
DESTE EQUIPAMENTOAVISO: O FABRICANTE RECOMENDA UMA FONTE ALTERNATIVA DE OXIGÉNIO SUPLEMENTAR PARA O CASO DE
FALHA DE ENERGIA, SITUAÇÃO DE ALARME OU FALHA
MECÂNICA. CONSULTE O SEU MÉDICO OU O FORNECEDORDO EQUIPAMENTO PARA SE INFORMAR SOBRE O TIPO DE SISTEMA
DE RESERVA REQUERIDO.
AVISO: ESTE DISPOSITIVO FORNECE UMA ALTA CONCENTRAÇÃO DEOXIGÉNIO QUE PROMOVE A COMBUSTÃO RÁPIDA. NÃO PERMITAQUE SE FUME OU A PRESENÇA DE CHAMAS NA MESMA DIVISÃO
ONDE SE ENCONTRA ESTE (1) DISPOSITIVO, OU (2) QUALQUERACESSÓRIO DE TRANSPORTE DE OXIGÉNIO. A NÃO OBSERVAÇÃODESTA ADVERTÊNCIA PODE RESULTAR EM INCÊNDIOS GRAVES,DANOS DE PROPRIEDADE E/OU PROVOCAR LESÕES FÍSICASOU MORTE.
AVISO: NÃO UTILIZE A UNIDADE NUM ESPAÇO RESTRITO OU CONFINADO (OU SEJA, NUMA CAIXA PEQUENA OU NUMA BOLSA) ONDEA VENTILAÇÃO POSSA FICAR LIMITADA. FAZÊ-LO PODE PROVOCARO SOBREAQUECIMENTO DO CONCENTRADOR DE OXIGÉNIO E PREJUDICAR O DESEMPENHO.
AVISO: O CONCENTRADOR DEVE FICAR LOCALIZADO DE MODO AEVITAR FUMO, AGENTES POLUENTES OU VAPORES.
AVISO: A UTILIZAÇÃO DE ALGUNS ACESSÓRIOS DE ADMINISTRAÇÃODE OXIGÉNIO NÃO ESPECIFICADOS PARA ESTE CONCENTRADOR DEOXIGÉNIO PODE PREJUDICAR O SEU DESEMPENHO. OS ACESSÓRIOS
RECOMENDADOS SÃO INDICADOS NESTE MANUAL.
AVISO: SE O CONCENTRADOR DE OXIGÉNIO TIVER SOFRIDO UMAQUEDA, ESTIVER DANIFICADO OU TIVER SIDO EXPOSTO À ÁGUA,CONTACTE O SEU PRESTADOR DE CUIDADOS DOMICILIÁRIOS PARAEFETUAR UMA INSPEÇÃO OU EVENTUAL REPARAÇÃO DO APARELHO.
NÃO USE O CONCENTRADOR DE OXIGÉNIO SE O RESPETIVO CABOOU FICHA DE ALIMENTAÇÃO ESTIVEREM DANIFICADOS.
AVISO: PRENDA DEVIDAMENTE O CONCENTRADOR DE OXIGÉNIOQUANDO O TRANSPORTAR NUM VEÍCULO PARA PREVENIR DANOSOU FERIMENTOS.
AVISO: NÃO PERMITA QUE AS ENTRADAS DE AR OU AS SAÍDAS DE
VENTILAÇÃO FIQUEM BLOQUEADAS. NÃO DEIXE CAIR NEM INSIRAOBJETOS NAS ABERTURAS DO APARELHO. FAZÊ-LO PODE PROVOCAR O SOBREAQUECIMENTO DO CONCENTRADOR DE OXIGÉNIO E
PREJUDICAR O DESEMPENHO.
AVISO: NÃO ENCHA EXCESSIVAMENTE O HUMIDIFICADOR OPCIONAL. ENCHA O HUMIDIFICADOR OPCIONAL COM ÁGUA APENAS ATÉAO NÍVEL INDICADO PELO FABRICANTE DO HUMIDIFICADOR.
AVISO: NÃO UTILIZE O EQUINOX ENQUANTO SE DESLOCA SE
ESTIVER LIGADO UM HUMIDIFICADOR. REMOVA A GARRAFAHUMIDIFICADORA ANTES DE SE DESLOCAR. NÃO POUSE O EQUINOX ENQUANTO ESTE SE ENCONTRAR LIGADO A UMA GARRAFAHUMIDIFICADORA.
AVISO: OS REGULAMENTOS DO DEPARTAMENTO NORTE-AMERICANO DOS TRANSPORTES (DOT) E DAS NAÇÕES UNIDAS (NU)
EXIGEM A REMOÇÃO DA BATERIA DO APARELHO EM TODAS AS
VIAGENS AÉREAS INTERNACIONAIS QUANDO O CONCENTRADORDE OXIGÉNIO É REGISTADO COMO BAGAGEM. PARA TRANSPORTARO CONCENTRADOR DE OXIGÉNIO, A BATERIA TAMBÉM TEM DE SERREMOVIDA DO DISPOSITIVO E EMBALADA ADEQUADAMENTE.

eQuinox (Model 4000) User Manual — 137

AVISO: NÃO UTILIZE ESTE EQUIPAMENTO SEM LER E COMPREENDER ESTE MANUAL PRIMEIRO. SE NÃO CONSEGUIRENTENDER AS ADVERTÊNCIAS E INSTRUÇÕES, CONTACTEO FORNECEDOR DO EQUIPAMENTO ANTES DE TENTAR USÁ-LO; CASOCONTRÁRIO, PODEM OCORRER LESÕES OU DANOS.
AVISO: SE SENTIR DESCONFORTO OU SE ESTIVER A PASSAR POR
UMA EMERGÊNCIA MÉDICA, PROCURE ASSISTÊNCIA MÉDICA IMEDIATAMENTE.
AVISO: A UTILIZAÇÃO DO CONCENTRADOR DE OXIGÉNIO FORA DASESPECIFICAÇÕES DE TEMPERATURA OPERACIONAIS PODE LIMITARA CAPACIDADE DO CONCENTRADOR PARA CORRESPONDER ÀSESPECIFICAÇÕES DE CONCENTRAÇÃO DE OXIGÉNIO. CONSULTE ASECÇÃO DE ESPECIFICAÇÕES DESTE MANUAL PARA SE INFORMARSOBRE OS LIMITES DE TEMPERATURA.
AVISO: NÃO USE ÓLEO, GORDURA, OU OUTROS PRODUTOS
INFLAMÁVEIS À BASE DE PETRÓLEO COM OS ACESSÓRIOS DE
TRANSPORTE DE OXIGÉNIO OU COM O CONCENTRADOR DE OXIGÉNIO. DEVEM USAR-SE APENAS LOÇÕES OU UNGUENTOS À BASE DEÁGUA E COMPATÍVEIS COM O OXIGÉNIO. O OXIGÉNIO ACELERA ACOMBUSTÃO DE SUBSTÂNCIAS INFLAMÁVEIS.
AVISO: A UTILIZAÇÃO INCORRETA DA BATERIA PODE PROVOCARO AQUECIMENTO E INFLAMAÇÃO DA MESMA E CAUSAR LESÕESGRAVES. CERTIFIQUE-SE DE NÃO PERFURAR, ATINGIR, PISAR OU
DEIXAR CAIR A BATERIA, OU DE A SUBMETER A IMPACTOS OU CHOQUES FORTES. A UTILIZAÇÃO DE UMA BATERIA DANIFICADA PODEPROVOCAR FERIMENTOS.
AVISO: O CONCENTRADOR DE OXIGÉNIO NÃO DEVE SER USADO
ADJACENTE A, OU EMPILHADO COM OUTRO EQUIPAMENTO.. SE AUTILIZAÇÃO DE FORMA ADJACENTE OU EMPILHADA FOR INEVITÁVEL, O CONCENTRADOR DE OXIGÉNIO DEVE SER OBSERVADOPARA VERIFICAR O SEU NORMAL FUNCIONAMENTO.
CUIDADO: Em caso de alarme, ou se notar que o Concentrador de oxigénio não está a funcionar corretamente;
consulte a secção de Resolução de problemas deste
manual. Se não conseguir resolver o problema, consulte oFornecedor do equipamento.
CUIDADO: Para evitar a anulação da garantia, siga todas as instruções dos fabricantes.
AVISO: NÃO TENTE EXECUTAR QUAISQUER MEDIDAS DE
MANUTENÇÃO PARA ALÉM DAS SOLUÇÕES POSSÍVEISLISTADAS NESTE MANUAL. NÃO REMOVA AS TAMPAS.
APENAS O SEU FORNECEDOR DO EQUIPAMENTO OU UMTÉCNICO DE MANUTENÇÃO QUALIFICADO DEVERÃO REMOVER ASTAMPAS OU EFETUAR A MANUTENÇÃO DESTE APARELHO.
AVISO: APENAS ACESSÓRIOS RECOMENDADOS PELO FABRICANTE.
A UTILIZAÇÃO DE OUTROS ACESSÓRIOS PODE SER PERIGOSA,CAUSAR GRAVES DANOS NO SEU CONCENTRADOR DE OXIGÉNIOE INVALIDAR A GARANTIA.
AVISO: O FABRICANTE RECOMENDA UMA FONTE ALTERNATIVA DE OXIGÉNIO SUPLEMENTAR PARA O CASO DE
FALHA DE ENERGIA, SITUAÇÃO DE ALARME OU FALHA
MECÂNICA. CONSULTE O SEU MÉDICO OU O FORNECEDORDO EQUIPAMENTO PARA SE INFORMAR SOBRE O TIPO DE SISTEMA
DE RESERVA REQUERIDO.
AVISO: NÃO PERMITA QUE AS ENTRADAS DE AR OU AS SAÍDAS DE
VENTILAÇÃO FIQUEM BLOQUEADAS. NÃO DEIXE CAIR NEM INSIRAOBJETOS NAS ABERTURAS DO APARELHO. FAZÊ-LO PODE PROVOCAR O SOBREAQUECIMENTO DO CONCENTRADOR DE OXIGÉNIO E
PREJUDICAR O DESEMPENHO.
138 — 4805-C1 C (Portuguese)

AVISO: FUMAR ENQUANTO SE USA O OXIGÉNIO É A PRIMEIRA CAUSA DE INCÊNDIOS E DE MORTES ASSOCIADAS.
É NECESSÁRIO QUE SIGA ESTES AVISOS DE SEGURANÇA:AVISO: NÃO PERMITA QUE SE FUME NEM QUE SE UTILIZEM VELAS OU CHAMAS ABERTAS NA MESMA DIVISÃO ONDE SE
ENCONTRA O DISPOSITIVO OU OS ACESSÓRIOS DE TRANSPORTE DEOXIGÉNIO.
AVISO: FUMAR ENQUANTO SE USA UMA CÂNULA DE OXIGÉNIOPODE CAUSAR QUEIMADURAS FACIAIS E POSSIVELMENTE A
MORTE.
AVISO: SE É FUMADOR, ESTES 3 PASSOS PODEM SALVAR A SUAVIDA: DESLIGUE O CONCENTRADOR DE OXIGÉNIO , RETIRE A CÂNULA E ABANDONE A DIVISÃO ONDE O APARELHO ESTÁ SITUADO.
AVISO: OS SINAIS "PROIBIDO FUMAR – PRESENÇA DE OXIGÉNIO"
DEVEM SER CLARAMENTE EXIBIDOS NO LAR ONDE O OXIGÉNIOESTÁ A SER USADO. OS DOENTES E OS PRESTADORES DE CUIDADOSTÊM DE SER INFORMADOS ACERCA DOS PERIGOS DE FUMAR NA
PRESENÇA DE, OU ENQUANTO SE USA OXIGÉNIO MÉDICO.
AVISO: NÃO USE O CONCENTRADOR DE OXIGÉNIO NA PRESENÇA
DE GASES INFLAMÁVEIS. FAZÊ-LO PODE RESULTAR EM COMBUSTÕES RÁPIDAS E CAUSAR DANOS DE PROPRIEDADE, LESÕESCORPORAIS OU MORTE.
AVISO: REMOVER A CÂNULA E COLOCÁ-LA EM CIMA DE VESTUÁRIO,ROUPA DE CAMA, SOFÁS OU OUTROS MATERIAIS ACOLCHOADOS
PODE PROVOCAR UM INCÊNDIO SÚBITO SE EXISTIR EXPOSIÇÃO A
CIGARROS, FONTES DE CALOR OU CHAMAS.
AVISO: NÃO DEIXE CÂNULAS NASAIS SOBRE OU SOB VESTUÁRIO,ROUPA DE CAMA OU ALMOFADAS DE CADEIRAS. SE A UNIDADE
ESTIVER LIGADA MAS NÃO ESTIVER A SER UTILIZADA, O OXIGÉNIOTORNARÁ O MATERIAL INFLAMÁVEL. COLOQUE O CONCENTRADORDE OXIGÉNIO NA POSIÇÃO 0 (DESLIGADO) QUANDO NÃO ESTIVERA SER UTILIZADO.
CUIDADO: Desligue sempre a fonte de alimentação CA daparede antes de desligar a a fonte de alimentação CA doconcentrador de oxigénio.
CUIDADO: Posicione sempre o tubo de fornecimento de oxigénio eos cabos de alimentação de modo a evitar tropeções.
CUIDADO: Quando usar o Concentrador de oxigénio num automóvel, barco, ou noutras fontes de CC com o cabo 12VDC,
certifique-se de que o veículo é posto em funcionamento antes deligar o Concentrador de oxigénio. Se a luz de alimentação externa do eQuinox não se acender e necessitar de uma redefinição,desligue o Cabo 12VDC da tomada de CC, volte a ligar o veículo e,em seguida, ligue de novo o Cabo 12VDC à tomada de CC. O não
seguimento destas instruções pode resultar na falta de fornecimento de energia ao Concentrador de oxigénio.
CUIDADO: Quando o automóvel onde está a usar o Concentradorde oxigénio for desligado, desligue e remova o aparelho do automóvel. Não guarde o Concentrador de oxigénio num automóvel
muito quente ou frio ou noutros ambientes similares de altas ou
baixas temperaturas.
Nota:OsequipamentosdecomunicaçãodeRFportáteisemóveispodem
afetar os equipamentos médicos elétricos.

Ambientes de Operação RecomendadosPara uma utilização correta do dispositivo, consulte a tabela
seguinte. A tabela fornece informações importantes acerca dos
ambientes e condições de funcionamento recomendados.

Especificações
Temperatura de De 0°C a 40°C (32°F a 104°F)
funcionamento
Humidade deOperação

10% - 95% a um ponto de condensação de
28°C (82,4° F)

Temperatura de De -25°C a 70°C (-13°F a 158°F)
Transporte/Ar- Humidade: Até 95% sem condensação
mazenamento
De -20°C a 50°C (-4°F a 122°F) para as unidades
de bateria
ElétricaNão utilize extensões. Não utilize tomadas eléctricas
controladas por um interruptor.

Altitude0 – 13.123 pés (0 - 4.000 metros)

Posicionamento NÃObloqueieaentradadear(coberturasuperiortraseira) nem a saída de exaustão (parte inferior direita
traseira). Posicione o aparelho a, no mínimo, 7,5 cm(3 polegadas) de distância de paredes, cortinados,móveis, etc.
Ambiente

Deve ser livre de fumo, poluentes e vapores.

Tempo de Funcionamento

24 horas por dia quando ligado a uma fonte dealimentação externa CA ou CC.

Nota: Se o Concentrador de oxigénio tiver sido armazenado por um
períododetempoalargadoforadosseuslimitesnormaisdetemperatura
defuncionamento,deverápermitirqueaunidadevolteàtemperaturade
funcionamento normal antes de a ligar. (Consulte a secção de Especificações deste manual.)

Posicionamento Correto do eQuinox
Selecione uma localização para o aparelho onde seja evitada aentrada de fumo, vapores e poluentes. O posicionamento correto
do aparelho deve permitir a entrada de ar através do filtro deadmissão de ar situado na parte superior traseira da caixa. Devepermitir também que o ar de exaustão saia livremente pela saídade exaustão na parte inferior direita do aparelho.
Posicione o equipamento de forma a que os alarmes sejam
audíveis.
Posicione o tubo de fornecimento de oxigénio de modo a não
ficar torcido ou obstruído.
AVISO: NÃO USE O CONCENTRADOR DE OXIGÉNIO NA
PRESENÇA DE GASES INFLAMÁVEIS. FAZÊ-LO PODERESULTAR EM COMBUSTÕES RÁPIDAS E CAUSAR DANOSDE PROPRIEDADE, LESÕES CORPORAIS OU MORTE. NÃO USE ÓLEO,GORDURA, OU OUTROS PRODUTOS INFLAMÁVEIS À BASE DEPETRÓLEO COM OS ACESSÓRIOS DE TRANSPORTE DE OXIGÉNIO OUCOM O CONCENTRADOR DE OXIGÉNIO. DEVEM USAR-SE APENASLOÇÕES OU UNGUENTOS À BASE DE ÁGUA E COMPATÍVEIS COM OOXIGÉNIO. O OXIGÉNIO ACELERA A COMBUSTÃO DE SUBSTÂNCIASINFLAMÁVEIS.
AVISO: ESTE DISPOSITIVO FORNECE UMA ALTA CONCENTRAÇÃO DEOXIGÉNIO QUE PROMOVE A COMBUSTÃO RÁPIDA. NÃO PERMITAQUE SE FUME OU A PRESENÇA DE CHAMAS NA MESMA DIVISÃO
ONDE SE ENCONTRA ESTE (1) DISPOSITIVO, OU (2) QUALQUERACESSÓRIO DE TRANSPORTE DE OXIGÉNIO. A NÃO OBSERVAÇÃODESTA ADVERTÊNCIA PODE RESULTAR EM INCÊNDIOS GRAVES,DANOS DE PROPRIEDADE E/OU PROVOCAR LESÕES FÍSICASOU MORTE.
Nota:Quando o automóvelondeestáausar o Concentradordeoxigénio
for desligado, desligue e remova o aparelho do automóvel. Não guardeo Concentrador de oxigénio num automóvel muito quente ou frio ou
noutrosambientessimilaresdealtasoubaixastemperaturas.NÃOdeixeo Concentrador de oxigénio ou o transformador ligados ao veículo sea ignição estiver na posição de desligado. Fazê-lo pode descarregar
a bateria do veículo.
CUIDADO: Não utilize a unidade num espaço restrito ou
confinado (ou seja, numa caixa pequena ou numa bolsa)
onde a ventilação possa ficar limitada. Fazê-lo podeprovocar o sobreaquecimento do Concentrador de oxigénio e
prejudicar o desempenho. Não permita que as entradas de ar ou
as saídas de ventilação fiquem bloqueadas. Fazê-lo pode provocar
o sobreaquecimento do Concentrador de oxigénio e prejudicar odesempenho.

eQuinox (Model 4000) User Manual — 139

Instruções de utilização
Antes de Utilizar

Informações Gerais de Carregamento

Este manual de instruções serve como referência para o auxiliar
na utilização e manutenção do aparelho. Se tiver dúvidas
ou preocupações, entre em contacto com o seu prestador decuidados domiciliários.

O eQuinox funciona com alimentação externa CA, alimentação
CC ou com uma bateria recarregável. O sistema inclui um"carregador de bateria inteligente." Este sistema recarrega
a bateria sempre que o eQuinox está ligado a uma fonte dealimentação externa. O sistema monitoriza e controla a fontede energia e o carregador da bateria.

Importante! NÃO tente operar o eQuinox sem primeiro ler asecção Diretrizes de Segurança deste manual. Siga todas as
instruções de operação. Tenha em consideração todos os Avisos
no aparelho e no Manual de Instruções. De forma a reduziro risco de incêndio, ferimentos ou graves danos materiais
no eQuinox, tenha em consideração todas as precauções desegurança.
AVISO: LIMPE A CAIXA, O PAINEL DE CONTROLO E OCABO DE ALIMENTAÇÃO APENAS COM UM AGENTE DE
LIMPEZA DOMÉSTICO LIGEIRO APLICADO COM UM PANOOU ESPONJA HÚMIDOS (E NÃO MOLHADOS) E, EM SEGUIDA, SEQUETODAS AS SUPERFÍCIES. NÃO PERMITA QUE ENTREM LÍQUIDOS NODISPOSITIVO. PRESTE ESPECIAL ATENÇÃO À SAÍDA DE OXIGÉNIOPARA A LIGAÇÃO DA CÂNULA, PARA SE CERTIFICAR DE QUE SE
MANTÉM LIVRE DE PÓ, ÁGUA E PARTÍCULAS.
CUIDADO: Não permita que as entradas de ar ou as saídasde ventilação fiquem bloqueadas. Fazê-lo pode provocar
o sobreaquecimento do Concentrador de oxigénio e prejudicar o desempenho. Não utilize o Concentrador de oxigénio sem
que o filtro de admissão de ar esteja no devido lugar. Se lhe forfornecido um segundo filtro, insira o filtro de "substituição" antes
de limpar o filtro sujo. Limpe o o filtro sujo numa solução com águae sabão e, em seguida, seque-o completamente antes de utilizar.

O eQuinox carrega a bateria durante a utilização de energia
CA ou CC quando a temperatura da bateria se situa dentro deuma amplitude segura para o carregamento. Se a alimentação
externa for removida, o aparelho muda automaticamente paraa alimentação da bateria, caso esta se encontre instalada ecarregada. Quando é novamente ligado a uma tomada de AC ou
CC, o aparelho funciona através da fonte de alimentação externa
e recarrega a bateria.
Alinhe a ficha com o conector externo embutido para ligar
corretamente o transformador CA ou o Cabo 12VDC. O
conector de alimentação embutido encontra-se no lado direito
do aparelho, olhando para o Painel de controlo. Insira a ficha noconector embutido.
FONTE DE ALIMENTAÇÃO CA

Transformador de CA

Ligar o eQuinox
O dispositivo pode ser utilizado com recurso a três (3) fontes dealimentação diferentes:• Transformador de CA• Cabo 12VDC• Unidade de alimentação recarregável (bateria)

O eQuinox inclui um transformador CA universal para utilização
em casa ou onde se encontre uma fonte alimentação de CApadrão disponível. Para efetuar a ligação a uma fonte dealimentação de CA, ligue o cabo de alimentação CA firmemente
ao conector externo embutido. Certifique-se também de queo cabo do transformador de CA está ligado a uma tomada de
CA. Quando o eQuinox está corretamente ligado, acende-se oindicador verde no transformador e o indicador de alimentação
externa. O indicador
no painel de controlo do utilizador
também se acende. O transformador de CA CAIRE, referência
4836 é a principal fonte de alimentação do concentradorde oxigénio portátil eQuinox. Trata-se de uma parte muito
importante do sistema geral.
Nota: Utilize apenas a tensão elétrica especificada na etiqueta de especificações afixada no dispositivo.
Passo 1

140 — 4805-C1 C (Portuguese)

Passo 2

Passo 3

AVISO: NÃO TOQUE NOS CONTACTOS DA BATERIA E NO
DOENTE AO MESMO TEMPO ENQUANTO ESTIVER A FORNECER ENERGIA À UNIDADE COM A FONTE DE ALIMENTAÇÃO EXTERNA (TRANSFORMADOR DE CA) E SE NÃO ESTIVER
INSTALADA NENHUMA UNIDADE DE BATERIA.
CUIDADO: Use apenas acessórios recomendados pelo fabricante. A utilização de outros pode ser perigosa, causar
graves danos no seu concentrador de oxigénio e invalidar
a garantia.
CUIDADO: Não utilize extensões elétricas com esta unidade nem
ligue demasiadas fichas à mesma tomada eléctrica. A utilização deextensões pode afetar negativamente o desempenho do dispositivo. A existência de demasiadas fichas numa tomada pode resultar
numa sobrecarga do quadro elétrico levando à ativação do disjuntor/fusível ou a um incêndio caso o disjuntor ou o fusível deixem defuncionar.
CUIDADO: A utilização de outros cabos e adaptadores além dos
especificados, à exceção dos cabos e adaptadores vendidos pelo
fabricante do equipamento médico elétrico como peças de substituição para componentes internos, pode resultar em emissões
acrescidas ou imunidade reduzida do Concentrador de oxigénio.

Cabo 12VDC
Cabo 12VDC

O cabo 12VDC permite ao sistema funcionar com baseem tomadas CC, como aquelas encontradas em veículos
motorizados.
1. Ligue o seu veículo.
2. Insira o cabo CC de 12 Volts na tomada embutida, na partelateral do aparelho.
3. Encaixe a ficha do cabo 12VDC na tomada CC do seu
veículo motorizado.
Quando o dispositivo está corretamente ligado e a receber
energia da fonte de alimentação de CC, o Indicador dealimentação externa verde
no painel de controlo acende-se.

Ligação do fornecimento de CC do eQuinox
Quando o eQuinox for ligado a uma fonte de CC de um veículo,
aplica-se o seguinte:• O eQuinox funcionará em todas as configurações de fluxo
contínuo 0,5–3,0LPM e configurações pulsáteis 1–9 enquanto
estiver a ser utilizado com o fornecimento de CC de um
automóvel.
• A bateria só será carregada se restar energia suficiente apóso dispositivo estar a funcionar à taxa de fluxo prescrita. O
eQuinox dará prioridade ao funcionamento da unidade noque respeita à energia fornecida. A energia restante é usadacomo carga lenta para recarregar a bateria. Se a bateria do
eQuinox estiver a ser carregada, o símbolo da bateria moverse-á em forma de cascata. Se a bateria do eQuinox não estivera carregar, o símbolo da bateria é exibido imóvel.
AVISO: NÃO PERMITA QUE AS ENTRADAS DE AR OU AS
SAÍDAS DE VENTILAÇÃO FIQUEM BLOQUEADAS. FAZÊ-LO
PODE PROVOCAR O SOBREAQUECIMENTO DO CONCENTRADOR DE OXIGÉNIO E PREJUDICAR O DESEMPENHO.
AVISO: SE A FONTE DE ALIMENTAÇÃO CC DO VEÍCULO DESCERABAIXO DOS 11,5 VOLTS, O EQUINOX MUDA PARA O FUNCIONAMENTO A BATERIA (SE A BATERIA ESTIVER INSTALADA E CARREGADA).
CUIDADO: Quando o automóvel onde está a usar o Concentrador de oxigénio for desligado, desligue e remova
o aparelho do automóvel. Não guarde o Concentradorde oxigénio num automóvel muito quente ou frio ou noutros
ambientes similares de altas ou baixas temperaturas. NÃO deixe oConcentrador de oxigénio ou o transformador ligados ao veículo sea ignição estiver na posição de DESLIGADO. Fazê-lo pode descarregar a bateria do veículo.
Nota: Ao operar o aparelho a partir de um cabo 12VDC, a Bateria não irá
carregar.
Nota: Ao operar o eQuinox a partir de um cabo 12VDC, recomendamos
que utilize o Modo de Dosagem Pulsátil se lhe tiver sido prescrita autilização de um equipamento de dosagem pulsátil.
Nota: Para garantir que o aparelho está a utilizar a alimentação CC, verifique se o Indicador de Alimentação Externa está aceso. Se for audível
um sinal sonoro, tal pode indicar uma perda de alimentação externa.
Verifiqueseo LEDverdeIndicador dealimentação externaestáacesoou
apagado. Se a bateria for removida durante a utilização de alimentação
CC, não haverá perda de carga enquanto a unidade se encontrar em funcionamento. Se o indicador não estiver aceso, a bateria está a alimentar
o aparelho.
Nota: O desempenho real do eQuinox pode variar dependendo das
condiçõesdeutilizaçãodoveículomotorizado.Algunsveículospoderão
não suportar uma funcionalidade total utilizando a alimentação CC.
Se o veículo for incapaz de fornecer energia suficiente para funcionar na
atual configuração, o eQuinox interromperá o carregamento da bateria
e continuará a funcionar à taxa de fluxo prescrita. Se, ainda assim, o veículo não fornecer energia suficiente, o eQuinox reduzirá a taxa de fluxo
se esta for igual ou superior a 2,0LPM. Se a taxa de fluxo for inferior a2,0LPM, o eQuinox desliga-se.
Se se verificarem taxas de fluxo baixas, remova todos os outros dispositivos eletrónicos ligados ao circuito de energia (GPS, carregador detelemóvel, etc.). Se o fornecimento de energia suficiente for reposto,
as taxas de fluxo não aumentarão automaticamente para voltar à
configuraçãooriginal;outilizadorterádeefetuaroajustemanualmente.
Se, ainda assim, o eQuinox não funcionar à taxa de fluxo prescrita, tenteremover a unidade da bateria.
eQuinox (Model 4000) User Manual — 141

Unidade de alimentação recarregável (bateria)
O eQuinox pode ser alimentado pela Unidade deAlimentação (Bateria) recarregável, fornecida
com o aparelho.

Nota: A utilização do Concentrador de oxigénio fora dos limites detemperatura de funcionamento normais pode afetar o desempenho ereduzir a duração da bateria e/ou aumentar o período de carregamento
da mesma. (Consulte a secção de Especificações neste manual.) ObteráumperíododefuncionamentomaisprolongadocomaBateriaseoperaro eQuinox no Modo Pulsátil (modo aprovado por médicos).

Instalação da Bateria: Alinhe a Bateria, demodo a fazê-la deslizar para o interior do
Compartimento de Alimentação vazio. Introduza
AVISO: PARA OS CONCENTRADORES DE OXIGÉNIO EQUIPADOS COM BATERIAS: ARMAZENE EM LOCAL FRESCO E
a bateria no Compartimento de Alimentação
SECO PARA AJUDAR A GARANTIR UMA MAIOR VIDA ÚTILvazio até encaixar no devido lugar (ouvindo-se
DA SUA BATERIA. O ARMAZENAMENTO DO CONCENum clique). Ficará nivelada com a parte frontal
TRADOR DE OXIGÉNIO POR PERÍODOS DE TEMPO PROLONGADOS
do aparelho. Se a Bateria estiver corretamente
A ALTAS TEMPERATURAS OU COM UMA BATERIA TOTALMENTE
instalada e o eQuinox estiver LIGADO, o
CARREGADA OU DESCARREGADA PODE DEGRADAR A VIDA ÚTILIndicador de Estado da Bateria aparece no
DA BATERIA. NÃO TENTE ABRIR A BATERIA: NÃO EXISTEM PEÇASREPARÁVEIS NO SEU INTERIOR. MANTENHA AS BATERIAS AFASTAPainel de Controlo. (Se NÃO estiver instaladaDAS DO ALCANCE DAS CRIANÇAS.
uma bateria, o Indicador de Estado da Unidadeda bateria não acende). A unidade continuará a
Unidade de alimen- Carregamento Inicial da Bateria
funcionar com alimentação externa.
tação recarregável A Bateria nova fornecida com o seu eQuinox não se encontra
Para retirar a Bateria, pressione o botão detotalmente carregada quando é entregue de fábrica. Antes de(bateria)
desbloqueio na parte superior da unidade dautilizar o eQuinox pela primeira vez, deve carregar a Bateria nabateria. Mantendo o botão de desbloqueio em BAIXO, afaste
totalidade.
a Bateria do eQuinox.
Ligue a fonte de alimentação de CA e instale corretamente a
Tempos de funcionamento da Bateria: O período de
Bateria. Deixe a Bateria carregar completamente. A Bateria está
totalmente carregada quando o Indicador de estado da unidade
funcionamento da bateria é apresentado no Painel de controlo,da bateria no Painel de controlo deixar de se mover em cascata
em forma de símbolo e em formato alfanumérico. Quando ae ficar imóvel.
alimentação da bateria está a ser usada, o indicador de estado
da bateria está sempre presente. O ecrã efetuará um ciclo para
O eQuinox pode ser usado enquanto a Bateria carrega.
apresentar as horas/minutos de bateria restantes. Uma variedade O carregamento pode demorar até 4 horas, dependendo daconfiguração de fluxo. A Bateria também carrega quando
de fatores, como a configuração do fluxo, o Modo de Fluxo
o eQuinox está desligado mas conectado a uma fonte de
Pulsátil ou Contínuo e a frequência respiratória, influenciam oalimentação CA.
tempo de funcionamento. O quadro Tempos de funcionamento
típicos de uma unidade de bateria nova, fornece estimativas dos
CUIDADO: Use apenas as baterias fornecidas pelo Fabritempos de funcionamento do eQuinox usando uma Bateria nova
cante. Para a eliminação adequada da bateria, entre em
e totalmente carregada, com base nas configurações de fluxo econtacto com o seu Fornecedor do Equipamento ou a sua
nas condições de utilização indicadas.
Nota: O eQuinox está equipado com uma bateria de 12 células quando é
enviado da fábrica. A bateria de 24 células é um acessório opcional paradurações de bateria mais longas.

Tempos de funcionamento típicos de uma unidade de bateria nova
Tipo de Funcionamento

Tipo de bateria
12-células

24-células

Modo contínuo
1 LPM

2,16 horas (2 h, 9 min)

4,84 horas (4 h, 50 min)

2 LPM

1,2 hours (1 hora, 12

2,75 hrs (2 horas, 45 min)

3 LPM

0,8 h (48 min)

1,83 h (1 hr, 49 min)Modo pulsátil(12 BPM)
16 mL (1,0)

2,62 horas (2 h, 37 min) 5,94 horas (5 h, 56 min)

48 mL (3,0)

2,38 horas (2 h, 22 min) 5,39 horas (5 h, 23 min)

96 mL (6,0)

1,70 horas (1 h, 41 min) 3,85 horas (3 h, 51 min)

142 — 4805-C1 C (Portuguese)agência governamental local para se informar sobre os requisitos
de eliminação.

* As baterias são recicláveis. *No caso de uma interrupção da alimentação CA ou CC, oeQuinox emite um sinal sonoro e muda automaticamente para ofuncionamento a Bateria, desde que a bateria se encontre instaladae carregada. Quando a alimentação externa é retomada, a Bateria
começa a carregar novamente de modo automático. Se a Bateria
não estiver presente ou se estiver totalmente descarregada quando
a alimentação CA for interrompida, o eQuinox encerra e faz
soar o alarme. A Bateria está totalmente descarregada quando
o Indicador de estado da unidade de alimentação no painel decontrolo aparecer vazio. O dispositivo exibe uma luz amarela efaz soar um alarme de bateria fraca. Esta condição mantém-se até
o aparelho encerrar completamente com uma luz vermelha e um
alarme constante. O alerta sonoro pode continuar durante cerca de
5 minutos até o dispositivo encerrar ou até a alimentação externa
ser reposta.
Nota: Prima e mantenha premido o botão LIGAR/DESLIGAR do eQuinox
durante um mínimo de 5 segundos para silenciar o alarme de Perda deEnergia. Ligar o eQuinox a uma fonte de alimentação CA externa duran-Período Típico de Recarga da Bateria

Ativar/Desativar o módulo de voz

Normalmente, são necessárias entre 2 e 5 horas para recarregar
uma Bateria totalmente esgotada até 80% da sua capacidade,dependendo da configuração de fluxo usada. Quando a bateria
estiver totalmente carregada, o indicador da bateria pára demover-se em cascata. A bateria pode ser carregada na unidade
quando o eQuinox se encontra desligado mas conectado à fontede alimentação.
Se a Bateria aquecer demasiado durante a descarga, o processo
de carregamento não começa até que a Bateria tenha arrefecidoo suficiente. Remover a Bateria pode acelerar o processo dearrefecimento.
A Bateria pode recarregar a partir de fontes de alimentação CA e
CC. A Bateria também carrega utilizando o Carregador de mesa
opcional.

1. Para ativar/desativar a funcionalidade do módulo de voz,
ligue a unidade premindo o botão LIGAR/Modo de espera.
2. Depois de ligar a unidade, prima o Ícone de Proibido fumar
e botão oculto uma vez.
3. Após 1–2 segundos, surge o item do menu SPKR = ON ou
SPKR = OFF.
4. Prima o botão Aumentar ou Reduzir a taxa de fluxo paraalternar entre as opções LIGAR (ativar módulo de voz) ou
DESLIGAR (desativar módulo de voz).
5. Após 3–5 segundos, a opção selecionada fica ativa.
Nota: Isto só desativará a função do módulo de voz uma vez. Quando odispositivo for desligado e depois reiniciado, a função do módulo de voz
será automaticamente ativada.

Aumentar/Reduzir o Volume das Notificações Vocais
1. Para aumentar ou diminuir o volume das notificações
vocais, prima o Ícone de Proibido fumar e Botão Ocultoduas vezes.

Utilização do módulo de voz

2. Surge o Volume = # - (Menu de configuração do volume).
3. Prima o botão Aumentar ou Reduzir a taxa de fluxo paraaumentar ou diminuir o nível de volume de 1 a 10.

Instruções operacionais do módulo de voz
O eQuinox tem a capacidade de fornecer notificações, alarmes
e estados de funcionamento verbais. Estas notificações verbais
complementam as informações do visor LED (díodo emissor deluz). As informações seguintes fornecem instruções sobre comoligar e desligar a funcionalidade do módulo de voz, regular ovolume e alterar a língua.
Indicador de
IndicadorBotão LIGADO/ LIGADO/Modo Indicador de alarme (Led verModo dealerta (LED ammelho)de esperaBotão para
esperaarelo)(LED verde)aumentar a
taxa de fluxo

4. Depois de selecionar o volume sonoro pretendido, a opção
ficará ativa/definida no espaço de 3–5 segundos.
Alterar o idioma do módulo de voz
1. Para alterar o idioma do módulo de voz, prima o Ícone deProibido fumar e Botão Oculto cinco vezes para aceder àsopções de idioma.
2. A língua atualmente ativada é indicada no visor. Por
exemplo, se o módulo de voz estiver a falar em Inglês, o texto
que aparece no visor ao aceder ao menu de opções de idioma
é, "English".
3. Prima o botão para Aumentar ou Reduzir a taxa de fluxo
para navegar pelas línguas disponíveis.
4. A língua pretendida é definida automaticamente
3–5 segundos depois de surgir no visor.

Indicador dealimentação
externa presente (LEDverde)

Ícone deProibido fumar e Botão
ocultoBotão demodo defluxo

Indicador domodo de fluxo(LED verde)

Visor LCD

Botão para
reduzir a taxa
de fluxo

eQuinox (Model 4000) User Manual — 143

Utilização do eQuinox pela primeira vez
Passo 1: Posicionar o Seu eQuinox para Utilização
Coloque o eQuinox numa área bem ventilada e iluminada.
Certifique-se de que a entrada de ar e as saídas de exaustão na
parte traseira não estão bloqueadas.
Filtro de admissão de ar
Posicione o eQuinox de forma
a que todos os indicadores ou
alarmes visuais e sonoros possam
ser facilmente vistos e ouvidos.
Certifique-se de que o filtro deadmissão de ar está instalado antes
de utilizar o seu eQuinox. Se ofiltro de admissão de ar estiverem falta, contacte o seu prestador
de cuidados domiciliários. Se ofiltro de admissão de ar estiversujo, lave-o com água quente com
sabão. Em seguida, enxaguar com
água limpa e permitir que sequeantes de voltar a colocar o filtro
Saída de Exaustão
de admissão de ar no dispositivo.
Se necessário, substitua-o por um filtro de admissão de ar novo
e limpo.

Passo 3: Ligue o Tubo de Fornecimento de Oxigénio
ou a Cânula Nasal à Saída de Oxigénio
Substitua a cânula e o tubo de fornecimento de oxigénio
regularmente, conforme recomendado pelo seu prestador decuidados domiciliários. Consulte o seu prestador de cuidados
domiciliários para se informar acerca da limpeza e substituição
da cânula e do tubo de alimentação.
Nota: Certifique-se de que a cânula está total e firmemente inserida. Issogarante que o Concentrador de oxigénio deteta corretamente a inspiração para administração do oxigénio. Durante a inalação, deverá ouvir
ou sentir o fluxo de oxigénio pelos tubos da cânula nasal.

Passo 4: Seleccione o Modo de Fluxo
Prima o botão de Modo de fluxo para selecionar o modo.
Ao premir este botão mais de uma vez alternará entre o Modode Dosagem Pulsátil e o Modo de Fluxo Contínuo. O modo e
configuração corretos devem-lhe ser prescritos pelo seu médico.
Operação do Modo de Fluxo Contínuo:Em Modo de fluxo contínuo, o oxigénio é administrado de forma
constante através do tubo e cânula nasal. O oxigénio é medido
em litros por minuto (LPM). Pode usar um humidificador apenas
quando em Modo de fluxo contínuo. NÃO incline, transporte,
Ligue o aparelho a uma tomada CA, a uma Fonte de alimentação ou deite o eQuinox enquanto houver uma garrafa humidificadoraCC, ou certifique-se de que está instalada uma Bateria totalmente ligada.
carregada.

Passo 2: LIGUE o Aparelho e Deixe-o Aquecer
Prima e mantenha premido o botão "LIGAR/DESLIGAR" durante dois (2) segundos para LIGAR oeQuinox. É executado um breve autoteste de ligação visual.
Nota: Se ouvir três sinais sonoros (bips) quando o dispositivo for ligado
ou conectado à fonte de alimentação pela primeira vez, isso indica um
estado de bateria fraca referente à bateria de 9-volts no interior da
caixa do eQuinox. Contacte o seu prestador de cuidados domiciliários esolicite a substituição da bateria de 9 volts. Se os três "bips" continuarem de forma intermitente depois de a unidade ter sido ligada e estara funcionar à vários minutos, isso significa que não se trata de um alerta
de bateria fraca referente à bateria de 9V. Contacte o seu prestador decuidados de saúde para providenciar a reparação.

O eQuinox possui um indicador do estado de concentração deoxigénio (OCSI) integrado no aparelho. O OCSI monitoriza
de forma contínua a saída de oxigénio do dispositivo. Utiliza
uma luz verde (normal), luz amarela (alerta), luz vermelha(alarme) e um alarme audível. Estes indicadores acendem todos
durante o autoteste de ligação no arranque. Após LIGAR oaparelho, aguarde cinco (5) minutos para que este alcance assuas especificações a nível de desempenho. O aparelho alcança
as suas especificações a nível de desempenho quando se acende
apenas o indicador verde.
144 — 4805-C1 C (Portuguese)

Operação no Modo de Dosagem Pulsátil:Ao funcionar em Modo Pulsátil, um bolo de oxigénio, medido
em mililitros (ml), é fornecido ao inspirar. NÃO use um
humidificador no Modo de Dosagem Pulsátil.
Ao operar no modo de dosagem pulsátil, o eQuinox fornece um
fluxo contínuo de oxigénio a cada 15 segundos, quando não é
detetada uma inalação. O aparelho interrompe o fluxo contínuo
e verifica repetidamente a presença de um esforço inspiratório
a cada 15 segundos. Se for detetada uma inalação, o aparelho
fornece um bolo medido novamente. Para sair do Modo deDosagem Pulsátil e ativar o Modo de Fluxo Contínuo, primao botão de modo de fluxo.
Solicite ao seu médico que lhe prescreva a dose pulsátil correta
para se certificar de que recebe a saturação de oxigénio adequada.
Tenha em consideração as suas necessidades durante o repouso,
quando faz exercício e quando viaja a altitudes elevadas.
Contacte o seu prestador de cuidados para obter a prescrição
e os formulários de embarque.

Modo de Dosagem Pulsátil e Funções autoSat
O seu eQuinox possui uma funcionalidade denominada autoSAT®.
Esta funcionalidade administra um bolo de oxigénio de dimensões
consistentes de até 40 respirações por minuto (consultar o quadro
Frequência respiratória máx. abaixo). À medida que a frequência
da sua respiração aumenta, como acontece ao caminhar, a função
autoSAT servo-controla o aparelho, administrando o tamanho debolo definido. A tecnologia autoSAT é exclusiva da CAIRE e éusada apenas durante o Modo pulsátil.

Volume do Bolo (+/- 15%)Frequência respiratória máx.
Configuração da
Dosagem Pulsátil

Tamanho do Bolo(± 15%) mL

Frequência respiratória máx.

1
2
3
4
5
6
7
8
9

16
32
48
64
80
96
128
160
192

40
40
40
40
37
31
23
18
15

Nota: O volume do bolo diminui à medida que a frequência respiratória
excede o valor máximo definido.
Nota: A configuração da dosagem pulsátil 1–9 não é igual a LPM.
AVISO: AS CONFIGURAÇÕES DO MODO DE DOSAGEM
PULSÁTIL DEVEM SER DETERMINADAS INDIVIDUALMENTE PARA CADA PACIENTE, DE ACORDO COM AS SUAS
NECESSIDADES EM REPOUSO, DURANTE O EXERCÍCIO E
DURANTE VIAGENS.
AVISO: O EQUINOX PODE NÃO CONSEGUIR DETETAR TODOS OS
ESFORÇOS INSPIRATÓRIOS EM MODO PULSÁTIL. SE O EQUINOX
NÃO RESPONDER AOS SEUS ESFORÇOS INSPIRATÓRIOS, SOLICITEAO SEU PRESTADOR DE CUIDADOS DOMICILIÁRIOS QUE VERIFIQUE
A CONFIGURAÇÃO DE SENSIBILIDADE.

Passo 5: Ajuste a Configuração de Fluxo Parao Nível Prescrito
Utilizando os Botões de Configuração do Fluxo, ajuste o fluxo
para o valor prescrito fornecido pelo seu médico.
AVISO: É MUITO IMPORTANTE SELECIONAR APENAS ONÍVEL DE OXIGÉNIO PRESCRITO. NÃO ALTERE A SELEÇÃO
DO FLUXO, A MENOS QUE TAL LHE TENHA SIDO INDICADOPOR UM MÉDICO AUTORIZADO.

Passo 6: Começar a utilizar o seu eQuinox
Respire normalmente através da cânula nasal.
Nota:ObteráumperíododefuncionamentomaisprolongadocomaBateria
se operar o eQuinox no Modo Pulsátil (modo aprovado por médicos).

Passo 7: Utilização do carrinho de transporte
do eQuinox
O carrinho de transporte do eQuinox foi concebido para ser
usado em superfícies próprias para caminhar.
Alinhe o eQuinox com a base do carrinho de modo a ficar na
vertical e nivelado.
Alinhe o parafuso roscado grande com a entrada na parte traseirado eQuinox. Em seguida, aperte manualmente o manípulo do
carrinho, até que o Carrinho do eQuinox fique bem fixo ao
aparelho.
Prima o botão na pega do Carrinho eQuinox para ajustar a alturada pega.

Nota:Paraofornecedordoequipamento:Recomenda-seautilizaçãodos
seguintesacessóriosdeadministraçãodeoxigéniocomoConcentradorde oxigénio:• Cânula nasal com 2,1 m (7 pés) de tubo: REF 5408-SEQ• Para se informar sobre os acessórios adicionais recomendados, consulte o Catálogo de acessórios (REF ML-LOX0010) disponível em www.
CAIREmedical.com.
• A CAIRE faculta o OxySafe™ como acessório opcional. Destina-se a serutilizadoconjuntamentecomoconcentradoreQuinox.Paraosclientes
emregiõesqueexijamaconformidadecomaENISO8359:1996-Emenda 1:2012, este acessório colmata esta necessidade.
O OxySafe™ consiste num fusível térmico para interromper o fluxo
gasoso na eventualidade de uma cânula ou tubo de oxigénio a jusanteincendiar e queimar o OxySafe. É colocado em linha com a cânula nasal
ou tubo de oxigénio entre o doente e a saída de oxigénio do eQuinox.
Para uma utilização apropriada do OxySafe, consulte sempre as instruções do fabricante (incluídas com cada kit OxySafe).
A CAIRE oferece um kit OxySafe™ que inclui uma máscara com cânula,o OxySafe ™ e tubos de 2,1 m (7’): N.º de ref.ª 20628667

Nota: NÃO utilize o eQuinox enquanto se desloca se estiver ligado um
humidificador. Remova a garrafa humidificadora antes de se deslocar.
NÃO pouse o eQuinox enquanto este se encontrar ligado a uma garrafa
humidificadora.

Passo 8: Desligar o Aparelho
Prima e mantenha premido o botão "LIGAR/DESLIGAR"
durante dois (2) segundos para DESLIGAR o eQuinox.
Armazene o eQuinox. num local fresco e seco durante um
período de tempo não superior a 1 mês. O eQuinox. deve
funcionar um mínimo de 2 horas por mês para garantir o correto
funcionamento da unidade.
eQuinox (Model 4000) User Manual — 145

Quadro de resolução de problemas do utilizador
Não ignore os alarmes
AVISO: O FABRICANTE RECOMENDA UMA FONTE ALTERNATIVA DE OXIGÉNIO SUPLEMENTAR PARA O CASO DE FALHA DE ENERGIA,SITUAÇÃO DE ALARME OU FALHA MECÂNICA. CONSULTE O SEU MÉDICO OU O FORNECEDOR DO EQUIPAMENTO PARA SE INFORMARSOBRE O TIPO DE SISTEMA DE RESERVA REQUERIDO.

Luz Amarela
Sintoma

Causa Possível

O eQuinox detetou um
A luz amarela está acesa.
problema com a ligação
O alarme emite 1 sinal sonoro
à unidade de alimentação(bip) a cada 30 segundos.
bateria.
A luz amarela está intermitente. O ícone da bateria está
A unidade de alimentação
intermitente. O alarme emite
do eQuinox está quente.
1 sinal sonoro (bip) a cada 2
minutos.

A sua ação1) Remova a unidade da bateria e volte a instalá-la para garantir que se encontra bem
encaixada no concentrador.
2) Se possível, instale outra unidade de alimentação no eQuinox.
3) Se este problema persistir, contacte o seu prestador de cuidados de saúde.
1)Volte a ligar o eQuinox à alimentação externa ou instale uma unidade de alimentação
totalmentecarregadanoeQuinox.Deixeaunidadedealimentaçãoquentearrefecerfora
do concentrador durante 30 minutos.
2) Recarregue a unidade da bateria utilizando a fonte de alimentação CA
3) Substitua a unidade de alimentação se possuir outra disponível.
4) Se a condição persistir, contacte o seu prestador de cuidados de saúde.

A luz amarela está intermi1) Ligue o eQuinox à fonte de alimentação CA e recarregue ou substitua a unidade datente. O ícone da bateria está A voltagem da unidade de bateria por uma totalmente carregada.
intermitente. O alarme emite 2 alimentação do eQuinox
sinais sonoros (bips) a cada 30 está baixa.
segundos.
1) Limpe e substitua o filtro de admissão da caixa
2) Certifique-se de que a cânula não está dobrada nem obstruída. Se usado com uma
garrafahumidificadora,certifique-sedequeestaestácorretamentecheiaenãoprovoca
um bloqueio.

O eQuinox detetou uma
A luz amarela está intermirestrição no fluxo, resultantente. O alarme emite 1 sinal
do num fluxo reduzido ou 3) Assegure uma ventilação adequada do eQuinox. Este deve estar, no mínimo, a 7,5 cm
sonoro a cada 2 minutos.
(3 polegadas) de distância de qualquer superfície, de forma a garantir que as saídas debloqueado.
ventilação não ficam obstruídas.

4) Seo problemapersistir, mudeparaumafontealternativadeoxigénio econtacteoseu
prestador de cuidados de saúde para obter assistência.
1) Certifique-se de que o filtro de admissão de ar não está obstruído ou limitado. Limpee substitua o filtro se necessário.
2) Certifique-se de que o eQuinox se encontra numa área bem ventilada. Certifique-sede que existe uma distância mínima de 7,5 cm (3 polegadas) entre a parte traseira do
A luz amarela está intermiO eQuinox detectou níveis eQuinox e quaisquer potenciais obstruções (móveis, cortinas, etc.)
tente.Nenhumalarmesonoro. baixos de oxigénio.
3) Se o utilizar no carro, certifique-se de que a parte traseira do eQuinox fica virada paraa parte exterior no banco.
4) Se a situação persistir, mude para uma fonte alternativa de oxigénio e contacte o seu
prestador de cuidados de saúde de imediato.

146 — 4805-C1 C (Portuguese)Luz Vermelha
Sintoma

Causa Possível

A sua ação1) Certifique-se de que o filtro de admissão de ar não está obstruído ou limitado. Limpee substitua o filtro se necessário.

A luz vermelha está intermitente.Oalarmeemite3 O eQuinox detetou níveis
sinais sonoros a cada 2 baixos de oxigénio
minutos.

2) Certifique-se de que o eQuinox se encontra numa área bem ventilada. Certifique-sede que existe uma distância mínima de 7,5 cm (3 polegadas) entre a parte traseira doeQuinox e quaisquer potenciais obstruções (móveis, cortinas, etc.)
3) Se o utilizar no carro, certifique-se de que a parte traseira do eQuinox fica virada paraa parte exterior no banco.
4) Se a situação persistir, mude para uma fonte alternativa de oxigénio e contacte o seu
prestador de cuidados de saúde de imediato.

1) Inspecione a fonte de alimentação externa (CA ou CC) e certifique-se de que as suas
OeQuinoxperdeutemporaria- ligações estão bem fixas ao eQuinox, ao transformador e à tomada externa.
menteaenergiaenquantofun2) Se possível, tente utilizar outra tomada elétrica (CA ou CC) para alimentar o eQuinox.
cionavaatravésdealimentação
3) Se a situação persistir, mude para uma fonte alternativa de oxigénio e contacte o seu
externa.
prestador de cuidados de saúde de imediato.

A luz vermelha estáintermitente. O alarmeemite um sinal sonoro
constante. O visor LCD
OeQuinoxperdeuenergiadevindica ********. O eQuiido ao esgotamento da carganoxnãoforneceoxigénio
da unidade de alimentação
e não liga.
(unidade da bateria) ou ao
sobreaquecimentodaunidadede alimentação.

A luz vermelha está acesa. O ecrã do visor indica
O eQuinox apresenta uma"FAIL" (Falha). O eQuinox
anomalia no sistema.
não está a administrar
oxigénio e não liga.

1)Volte a ligar o eQuinox à alimentação externa ou instale uma unidade de alimentação
totalmentecarregadanoeQuinox.Deixeaunidadedealimentaçãoquentearrefecerfora
do concentrador durante 30 minutos.
2) Recarregue a unidade da bateria utilizando a fonte de alimentação CA.
3) Substitua a unidade de alimentação se possuir outra disponível.
4) Se a condição persistir, contacte o seu prestador de cuidados de saúde.
1) Remova a unidade de alimentação e desligue a fonte de alimentação CA ou CC doeQuinox. A luz vermelha deverá desaparecer e a mensagem FAIL desaparecerá do ecrãquandoasfontesdealimentaçãoforemremovidas.Aguardeaproximadamente20segundos, volte a ligar as fontes de alimentação e tente ligar novamente o eQuinox.
2) Se o eQuinox ligar, monitorize-o para determinar se a mensagem FAIL volta a aparecer. Se a mensagem FAIL voltar a aparecer, contacte imediatamente o seu prestador decuidados de saúde.
3) Se o eQuinox não voltar a ligar, contacte imediatamente o seu prestador de cuidadosde saúde.

eQuinox (Model 4000) User Manual — 147

Outras Condições de Alarme
Sintoma

Causa Possível

O eQuinox não é capaz
Quando se encontra em
de detetar o esforço demodopulsátil,oLEDdomodo inspiração do utilizador.
pulsátil pisca rapidamente Consulte a página 15 para
durante 15 segundos.
mais informações sobre omodo pulsátil.

A sua ação1)Omodopulsátilpodenãofuncionarcorretamenteseoutilizadorapresentarumacondição
que provoque fossas nasais obstruídas/com restrições. (Doença, desvio no septo nasal, etc.)
2) Certifique-se de que a cânula não possui mais de 2 metros (7 pés) de comprimento.
3) Certifique-se da ausência de dobras ou bloqueios no tubo ou cânula.
4)Seestiverligadaumagarrafahumidificadora,remova-aeligueacânuladiretamenteàporta
de saída. O modo pulsátil não funciona com uma garrafa humidificadora instalada.
5) Se a condição persistir, contacte o seu prestador de cuidados de saúde.
1)Verifiqueseatomadaestáafornecerenergia.Nãoligueafontedealimentaçãoaumcircuito
atenuador ou a uma ficha múltipla.
2)Verifiqueseasligaçõesdoscabosnasfontesdealimentaçãoestãofixasàparede/aoveículo
eaoconcentrador.SeutilizarafontedealimentaçãoCA,verifiquetambémaligaçãodocaboaotransformador de energia.
3) Certifique-se de que as luzes de verificação de alimentação estão acesas.

A Unidade da bateria desO eQuinox não recebe
carrega quando está ligada à energiadafontedealimen- Será visível uma luz verde na caixa da fonte de alimentação.
fonte de alimentação CA/CC tação ligada.
- Será visível uma luz verde no painel de controlo do concentrador eQuinox na forma de um
cabo de alimentação.
4)SeosLEDsdeverificaçãodaalimentaçãonãoestiveremacesos,removatodasasligaçõesda
fonte de alimentação durante 20 segundos e volte a ligar.
5) Se a condição persistir, contacte o seu prestador de cuidados de saúde.
A unidade de alimentação 1) Ligue à alimentação CA para recarregar a bateria. Assegure-se de que a bateria carrega
durante2a5horaseverifiqueseoíconedabateriaestácheioenãointermitenteantesdeusar.
não estava totalmentecarregada.
2) Consulte as ações para o sintoma "Unidade de Alimentação Não Carrega" mais abaixo.
O eQuinox não estava
1) Certifique-se de que está a usar a taxa de fluxo prescrita pelo seu médico e que o modo dea funcionar com o fluxo
fluxo pulsátil ou contínuo está a ser usado conforme requerido.
A carga da Unidade de aliprevisto.
mentação (bateria) não dura
1)Tente esvaziar a unidade da bateria por completo. Faça-o utilizando o eQuinox alimentadoo tempo que deveria.
A unidade da bateria não pela bateria até a máquina se desligar por completo. De seguida, remova a bateria e deixe-a
apresentaumdesempenho arrefecerdurantecercade30minutos.Apósesteperíododetempo,volteacolocarabaterianode acordo com as especifi- eQuinox e recarregue-a, utilizando a alimentação CA.
cações.
2) Se a bateria continuar a não durar o tempo previsto após seguir o passo 1, contacte o seu
prestador de cuidados de saúde.
1)Verifiqueseatomadaestáafornecerenergia.Nãoligueafontedealimentaçãoaumcircuito
atenuador ou a uma ficha múltipla.
2)Verifiqueseasligaçõesdoscabosnasfontesdealimentaçãoestãofixasàparede/aoveículo
eaoconcentrador.SeutilizarafontedealimentaçãoCA,verifiquetambémaligaçãodocaboaotransformador de energia.
O eQuinox não está areceber alimentação CA 3) Certifique-se de que as luzes de verificação de alimentação estão acesas.
externa para carregar abateria.

- Será visível uma luz verde na caixa da fonte de alimentação.

- Será visível uma luz verde no painel de controlo do concentrador eQuinox na forma de um
cabo de alimentação.
A Unidade de alimentação
4)SeosLEDsdeverificaçãodaalimentaçãonãoestiveremacesos,removatodasasligaçõesda
(bateria) não está a carregar.
fonte de alimentação durante 20 segundos e volte a ligar.
5) Se a condição persistir, contacte o seu prestador de cuidados de saúde.
1)Removaaunidadedabateriaevolteainstalá-laparagarantirqueseencontrabemencaixada no concentrador.
2) Inspecione o eQuinox quanto a uma luz amarela contínua quando a bateria se encontrar
O eQuinox não está acomunicar corretamente instaladaouumíconeintermitentedabateriaaotentarcarregar.Seseverificaraexistênciade
com a unidade da bateria um destes sintomas, avance para o passo 3.
de modo a carregá-la.
3)Sepossível,instaleoutraunidadedebaterianoeQuinox.Seabateriadesubstituiçãofuncionar corretamente, a bateria original deve ser substituída.
4) Se este problema persistir contacte o seu prestador de cuidados de saúde.
1) Contacte o seu prestador de cuidados de saúde.
O eQuinox emite um sinal
A bateria interna que alisonoro(bip)3vezesquandoé
menta o alarme está fraca
ligado pela primeira vez.
148 — 4805-C1 C (Portuguese)Limpeza, Conservação e Manutenção de Rotina
Manutenção de Rotina

Filtro de Admissão de Ar

As peças internas do eQuinox só devem ser sujeitas amanutenção por parte de um técnico qualificado e formado
pela CAIRE.

O ar ambiente é atraído para o equipamento através da entrada
de ar localizada na parte superior traseira do aparelho. A
limpeza do filtro de admissão de ar é a ação de manutenção
mais importante que irá efetuar enquanto utilizador. A limpezado filtro de admissão de ar fará com que o eQuinox mantenhaum desempenho adequado. Verifique diariamente o filtro deadmissão de ar e limpe-o uma vez por semana, pelo menos,
utilizando o seguinte procedimento:

A manutenção preventiva (MP) deve ser efetuada anualmente.
Contacte o seu prestador para obter um planeamento da MP.
O seu prestador de cuidados domiciliários ou um técnico
qualificado inspecionarão as peças seguintes, apenas quando
necessário: alarmes, bateria interna de 9-volts, bateria, filtrosinternos e outras peças internas.
Nota: Se ouvir três bips após o autoteste de ligação, isso indica que abateria de 9v está fraca. Se ouvir três sinais sonoros (bips), contacte o seu
prestador de cuidados domiciliários.

Conservação e Limpeza do Aparelho pelo Utilizador
Substituição da Cânula

1. Remova o filtro de admissão de ar
2. Lave o filtro em água corrente morna, utilizando uma
solução de detergente suave
3. Enxague bem o filtro com água corrente limpa e removaa água em excesso
4. Deixe o filtro secar ao ar
5. Volte a colocar o filtro na caixa

Substitua o tubo e a cânula de fornecimento com a frequência
Conselho de limpeza: Mantenha um segundo filtro limpo
recomendada pelo seu prestador de cuidados domiciliários. O seu disponível como substituição, para utilizar enquanto o outro filtro
médico ou prestador de cuidados domiciliários irá fornecer-lhe as se encontra a secar.
informações de limpeza, desinfeção e substituição.
Nota: Siga sempre as instruções do fabricante da cânula para uma
utilizaçãoapropriada.Substituaacânuladescartávelconformerecomendado pelo fabricante da cânula ou pelo fornecedor do equipamento.
Estão disponíveis consumíveis adicionais junto do seu Fornecedor do
equipamento.

Entrada de
Ar Filtro

CUIDADO: Não utilize o Concentrador de oxigénio sem queo filtro de admissão de ar esteja no devido lugar. Se lhe
for fornecido um segundo filtro, insira o filtro de "substituição" antes de limpar o filtro sujo. Limpe o o filtro sujo numa
solução com água e sabão e, em seguida, seque-o completamente
antes de utilizar.
Nota: O Fabricante não recomenda a esterilização deste equipamento.
Nota: Não utilize o Concentrador de oxigénio sem que o filtro deadmissão de ar esteja no devido lugar. Se lhe for fornecido um segundo
filtro, insira o filtro de "substituição" antes de limpar o filtro sujo. Limpeo o filtro sujo numa solução com água e sabão e, em seguida, seque-ocompletamente antes de utilizar.

O filtro de admissão de ar deve ser substituído anualmente.
Se o eQuinox for utilizado num ambiente poeirento, pode sernecessário substituir o filtro com maior frequência. Não deve
operar o eQuinox durante mais de 5 minutos sem um filtro deadmissão de ar instalado.

eQuinox (Model 4000) User Manual — 149

Fonte de alimentação CA, Cabo 12VDC, Cabo de alimentação
e Caixa exterior:
DESLIGUE o eQuinox e desconecte-o da alimentação CA ou
CC antes de limpar ou desinfetar. NÃO pulverize soluções
diretamente sobre a caixa exterior. Utilize um pano ou esponja
húmidos (não molhados). Pulverize o pano ou a esponja com
uma solução de limpeza suave para limpar a caixa e as fontes
de alimentação. Para desinfetar o eQuinox,, utilize desinfetante
Lysol® Brand II (ou equivalente). Siga as instruções dofabricante do agente de limpeza.

AVISO: PERIGO DE CHOQUE ELÉTRICO. DESLIGUE (OFF)A UNIDADE E RETIRE O CABO DE ALIMENTAÇÃO DA
TOMADA ELÉTRICA ANTES DE LIMPAR A UNIDADE, DE
MODO A EVITAR CHOQUES ELÉTRICOS ACIDENTAIS E O RISCO DE
QUEIMADURAS. APENAS O SEU FORNECEDOR DO EQUIPAMENTO OU
UM TÉCNICO DE MANUTENÇÃO QUALIFICADO DEVERÃO REMOVER AS TAMPAS OU EFETUAR A MANUTENÇÃO DA UNIDADE. NÃO
UTILIZE LÍQUIDOS DIRETAMENTE SOBRE A UNIDADE. A LISTA DE
AGENTES QUÍMICOS INDESEJÁVEIS INCLUI MAS NÃO SE LIMITA AOSSEGUINTES: ÁLCOOL E PRODUTOS À BASE DE ÁLCOOL, PRODUTOS
CONCENTRADOS À BASE DE CLORO (CLORETO DE ETILENO ), E
PRODUTOS À BASE DE ÓLEO (PINE-SOL®, LESTOIL®). ESTAS SUBSTÂNCIAS NÃO DEVEM SER USADAS PARA LIMPAR O INVÓLUCRODE PLÁSTICO DO CONCENTRADOR DE OXIGÉNIO, JÁ QUE PODEMDANIFICAR O PLÁSTICO DA UNIDADE.
AVISO: PERIGO DE CHOQUE ELÉTRICO. DESLIGUE O CABO DE ALIMENTAÇÃO DA TOMADA ELÉTRICA ANTES DE LIMPAR A UNIDADE
DE MODO A EVITAR CHOQUES ELÉTRICOS ACIDENTAIS E O RISCO DE
QUEIMADURAS. APENAS O SEU FORNECEDOR DO EQUIPAMENTO OU
UM TÉCNICO DE MANUTENÇÃO QUALIFICADO DEVERÃO REMOVER
AS TAMPAS OU EFETUAR A MANUTENÇÃO DA UNIDADE.
AVISO: DEVE TER-SE CUIDADO PARA EVITAR QUE O CONCENTRADORDE OXIGÉNIO E A BATERIA SE MOLHEM OU QUE ENTREM FLUIDOSNA UNIDADE. ISSO PODE CAUSAR UMA AVARIA OU PARALISAÇÃO
E RESULTAR NUM MAIOR RISCO DE CHOQUES ELÉTRICOS OU QUEIMADURAS.
AVISO: LIMPE A CAIXA, O PAINEL DE CONTROLO E O CABO DE
ALIMENTAÇÃO APENAS COM UM AGENTE DE LIMPEZA DOMÉSTICO LIGEIRO APLICADO COM UM PANO OU ESPONJA HÚMIDOS(E NÃO MOLHADOS) E, EM SEGUIDA, SEQUE TODAS AS SUPERFÍCIES.
NÃO PERMITA QUE ENTREM LÍQUIDOS NO DISPOSITIVO.
PRESTE ESPECIAL ATENÇÃO À SAÍDA DE OXIGÉNIO PARAA LIGAÇÃO DA CÂNULA, PARA SE CERTIFICAR DE QUE SE
MANTÉM LIVRE DE PÓ, ÁGUA E PARTÍCULAS.
AVISO: NÃO UTILIZE LÍQUIDOS DIRETAMENTE SOBRE A UNIDADE.
A LISTA DE AGENTES QUÍMICOS INDESEJÁVEIS INCLUI MAS NÃO SE
LIMITA AOS SEGUINTES: ÁLCOOL E PRODUTOS À BASE DE ÁLCOOL,
PRODUTOS CONCENTRADOS À BASE DE CLORO (CLORETO DE
ETILENO ), E PRODUTOS À BASE DE ÓLEO (PINE-SOL®, LESTOIL®).
ESTAS SUBSTÂNCIAS NÃO DEVEM SER USADAS PARA LIMPAR O
INVÓLUCRO DE PLÁSTICO DO CONCENTRADOR DE OXIGÉNIO, JÁQUE PODEM DANIFICAR O PLÁSTICO DA UNIDADE.

150 — 4805-C1 C (Portuguese)

Humidificador opcional:
Se o utilizar, limpe o humidificador diariamente para reduzir
a possibilidade de contaminação. Siga as recomendações delimpeza fornecidas pelo seu prestador de cuidados domiciliários
ou pelo fabricante do humidificador. Substitua mensalmente o
seu humidificador ou conforme recomendado pelo fabricanteou pelo seu prestador de cuidados domiciliários. (Não enchao humidificador em excesso.)
CUIDADO: NÃO utilize o eQuinox enquanto se desloca seestiver ligado um humidificador. Remova a garrafa humidificadora antes de se deslocar. NÃO pouse o eQuinox
enquanto este se encontrar ligado a uma garrafa humidificadora.

Carrinho de transporte do eQuinox
Antes de limpar o Carrinho de transporte do eQuinox remova-odo eQuinox. Use um pano ou esponja húmidos (não molhados)e uma solução de limpeza ligeira para limpar o carrinho.
Unidade de alimentação (Bateria)
A bateria no eQuinox necessita de especial cuidado para garantir
uma vida útil mais longa e o melhor nível de desempenho. A
Bateria CAIRE é a única bateria recomendada para utilização
com o eQuinox.
Use um pano ou esponja húmidos (não molhados) para limpar a
Bateria. Em primeiro lugar, pulverize o pano ou a esponja com
um detergente suave e, em seguida, limpe a caixa da Bateria eo fecho.
A CAIRE recomenda que a unidade da bateria seja substituída
após 2 anos.
Eliminação da bateria
A sua Bateria é recarregável e pode ser reciclada. Devolva-a
sempre ao seu prestador de cuidados domiciliários para garantir
uma eliminação adequada. Pode também entrar em contracto
com as entidades locais para obter instruções acerca da correta
eliminação da Bateria.
Eliminação do sistema de oxigénio eQuinox
A legislação ambiental local pode não permitir a eliminação deequipamentos elétricos e/ou eletrónicos, tal como o eQuinox ou
a fonte de alimentação CA. Entre em contacto com as entidades
responsáveis locais para obter instruções sobre a eliminação deequipamentos elétricos ou eletrónicos. Pode também contactar
a CAIRE Inc. para obter informações sobre a eliminação.

Quadros EMC
Orientações e Declaração do Fabricante­—Emissões eletromagnéticas
OeQuinoxdestina-seaserusadonoambienteeletromagnéticoespecificadoabaixo.OclienteouutilizadordoeQuinoxdevecertificar-sede que este é usado no ambiente descrito.
Teste de Emissões
Emissões RF
CISPR 11
Emissões RF
CISPR 11

Conformidade

Ambiente eletromagnético - Orientações

Grupo 1O eQuinox utiliza energia de RF apenas para o seu funcionamento interno.
Por conseguinte, as suas emissões de RF são muito baixas e não é provável
que causem interferências nos equipamentos eletrónicos próximos.

Classe B

Emissões harmónicas
IEC 61000-3-2Classe A

Oscilações da voltagem/
emissões oscilantes
IEC 61000-3-3

Em conformidade

A utilização do eQuinox é adequada em todos os estabelecimentos, incluindo
os estabelecimentos domésticos e os que se encontram ligados à rede pública
defornecimentodeenergiadebaixavoltagemqueabasteceosedifíciosusados
para fins domésticos.

eQuinox (Model 4000) User Manual — 151

Orientações e Declaração do Fabricante—Imunidade eletromagnética
OeQuinoxdestina-seaserusadonoambienteeletromagnéticoespecificadoabaixo.OclienteouutilizadordoeQuinoxdevecertificar-sede que este é usado no ambiente descrito.
Teste de imunidade Nível de teste IEC 60601
AmbienteeletromagContacto ±6 kV
nético – orientações
Ar ±8 kV
IEC 61000-4-2
Transiente/disparo
rápido elétrico
IEC 61000-4-4

Sobretensão
IEC 61000-4-5

Quedas de tensão,interrupções curtas
e variações detensão nas linhas deentrada de fornecimento de energia
IEC 61000-4-11Campo magnético
de frequência deenergia (50/60 Hz)
IEC 61000-4-8

Nível de conformidade

Ambiente eletromagnético – orientações

Contacto ±6 kV

O piso deve ser de madeira, cimento ou azulejo.
Se o piso estiver coberto de material sintético, ahumidade relativa deve ser de 30%, no mínimo.

Ar ±8 kV

±2 kV para linhas de forneci±2 kV para linhas de fornecmento energético
imento energético
A qualidade da rede elétrica deve ser idêntica à
de um ambiente comercial ou hospitalar típico.
±1 kV para linhas de entrada/
N/A
saída
±1 kV linha(s) para linha(s)±1 kV linha(s) para linha(s)±2 kV linha(s) para terra<5% UT<5% UT(>95% de queda na UT )(>95% de queda na UT )para 0,5 ciclos

para 0,5 ciclos

40% UT

40% UT(60% de queda na UT )(60% de queda na UT )para 5 ciclos

para 5 ciclos

70% UT

70% UT(30% de queda na UT )(30% de queda na UT )para 25 ciclos

para 25 ciclos<5% UT<5% UT(>95% de queda na UT )(>95% de queda na UT )durante 5 seg.

durante 5 seg.

3A/m

3A/m

NOTA UT é a tensão de CA antes da aplicação do nível de teste.

152 — 4805-C1 C (Portuguese)

A qualidade da rede elétrica deve ser idêntica à
de um ambiente comercial ou hospitalar típico.

A qualidade da rede elétrica deve ser idêntica à
de um ambiente comercial ou hospitalar típico.
Se o utilizador do eQuinox necessitar de um
funcionamentocontinuadoduranteinterrupções
de energia, recomendamos que o eQuinox sejaalimentadoporumafontedealimentaçãoininterrupta ou bateria.

Os campos magnéticos de frequência de energia
devem estar a um nível característico de um local
típico, num ambiente comercial ou hospitalar
típico.

Orientações e Declaração do Fabricante–Imunidade eletromagnética
OeQuinoxdestina-seaserusadonoambienteeletromagnéticoespecificadoabaixo.OclienteouutilizadordoeQuinoxdevecertificar-sede que este é usado no ambiente descrito.
Teste de imunidade Nível de teste IEC 60601

Nível de conformidade

Ambiente eletromagnético – orientações

RF Conduzida

3 Vrms3 VrmsIEC 61000-4-6

150 kHz a 80 MHz

RF irradiada

3 V/mOequipamentoportátilemóveldecomunicações
RF não deve ser utilizado mais perto de qualquerpeça do eQuinox, incluindo cabos, do que adistância de separação recomendada calculadaa partir da equação aplicável à frequência dotransmissor.

IEC 61000-4-3

80 MHz a 2,5 GHz

3 V/mDistância de separação recomendadad = 1,.2 P
d = 1,.2 P = 80 MHz a 800 MHz
d = 1,.2 P = 800 MHz a 2,5 GHz
onde P é a potência nominal máxima de saída
do transmissor em Watts (W), de acordo com ofabricante do transmissor, e d é a distância deseparação recomendada em metros (m).
As forças de campo provenientes de transmissores de RF fixos, conforme determinado por um
levantamento eletromagnético do local a devem
ser inferiores ao nível de conformidade em cadaintervalo de frequência. b
Poderão ocorrer interferências nas imediações
de equipamentos assinalados com o seguinte
símbolo:

NOTA 1 A 80 MHz e 800 MHz, aplica-se o intervalo de frequência superior.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela absorção e reflexo de
estruturas, objetos e pessoas.
As forças de campo de transmissores fixos, tais como estações de base para telefones de rádio (telemóveis/sem fios) e rádios
móveis terrestres, rádio amador, transmissão de rádio AM e FM e transmissão de televisão não se podem prever teoricamente com
precisão. Para avaliar o ambiente eletromagnético para transmissores de RF fixos, deverá ser considerado um estudo eletromagnético
do local. Se a força do campo medido no local onde o eQuinox é utilizado exceder o nível de conformidade RF aplicável indicado acima,o eQuinox deve ser observado para determinar o seu funcionamento normal. Se for observado um desempenho anormal, poderão sernecessárias medidas adicionais, tais como reorientar ou relocalizar o eQuinox.
ab

No intervalo de frequência de 150 kHz a 80 MHz, as forças de campo devem ser inferiores a 3 V/m.

eQuinox (Model 4000) User Manual — 153

Distâncias de separação recomendadas
entre os equipamentos de comunicações RF portáteis e móveis e o eQuinox
O eQuinox destina-se a ser utilizado num ambiente eletromagnético no qual as perturbações de RF irradiadas sejam controladas. O
cliente ou o utilizador do eQuinox pode ajudar a evitar interferências eletromagnéticas mantendo uma distância mínima entre os equipamentos de comunicações por RF portáteis e móveis (transmissores) e o eQuinox, tal como se recomenda em baixo, de acordo com apotência máxima de saída do equipamento de comunicações.
Potência nominal máxima de saída do transmissor

Distância de separação de acordo com a frequência do transmissorm
150 kHz a 80 MHz

80 MHz a 800 MHz

800 MHz a 2,5 GHzW

d = 1., 2 P

d =1.2, P

d  1=.2 , P

0,01

0,12

0,12

0,23

0,1

0,38

0,38

0,73

1

1,2

1,2

2.3

10

3,8

3,8

7,3

100

12

12

23

Paratransmissorescomumapotêncianominaldesaídamáximanãoindicadaacima,adistânciadeseparaçãorecomendadademmetros(m) pode ser estimada utilizando a equação aplicável à frequência do transmissor, em que P é a potência nominal máxima de saída dotransmissor em Watts (W) segundo o fabricante do transmissor.
NOTA 1 A 80 MHz e 800 MHz, aplica-se a distância de separação para o intervalo de frequência superior.
NOTA 2 Estas diretrizes podem não se aplicar em todas as situações. A propagação eletromagnética é afetada pela absorção e reflexo de
estruturas, objetos e pessoas.

154 — 4805-C1 C (Portuguese)

Informações Gerais das Companhias Aéreas
Durante os movimentos de táxi, a descolagem e a aterragem, seo eQuinox não for usado, deve ser desligado e acondicionado
sob o assento ou em outro local de acondicionamento aprovado,de forma a não obstruir o corredor ou os pontos de acesso à fila.
Durante os movimentos de táxi, a descolagem e aterragem, seo eQuinox for usado, o utilizador deve situar-se num assento
que não restrinja o acesso dos outros passageiros às saídas deemergência e saídas normais necessárias, ou ao(s) corredor(es)na área de passageiros da aeronave.
Durante a utilização do eQuinox, os utilizadores não podem
sentar-se numa fila junto à saída.
Se o eQuinox estiver em utilização durante uma situação dedescompressão da cabina e acionamento do sistema de oxigénioda cabina, o utilizador deve interromper a utilização do eQuinox
e utilizar o oxigénio complementar da aeronave. A unidade doeQuinox deve ser desligada após estar assegurado o oxigénio
complementar da aeronave.
Antes da viagem, o utilizador deve inspecionar o eQuinox paragarantir o seu bom estado de funcionamento.

Unidade da Bateria e a Viagens em
Companhias Aéreas
Tenha em consideração o tempo em terra(check-in antes do voo, check in de segurança, etc.) e transbordos.
A maioria das companhias aéreas exigem que se disponha deum número adequado de baterias totalmente carregadas paraalimentar o dispositivo durante, pelo menos, 150% da duração
máxima esperada para a viagem (tempos de voo, todo o tempo
em terra; período anterior e posterior ao voo e durante asligações e atrasos inesperados).

É recomendado transportar sempre uma bateria
extra nas suas viagens.
Cada companhia aérea possui os seus próprios requisitos
para efetuar viagens com oxigénio.
Contacte a sua companhia aérea pelo menos 48 horas antes
da partida para informar que irá viajar com o eQuinox.

CUIDADO: Os regulamentos do Departamento
Norte-americano dos Transportes (DOT) e das Nações
Unidas (NU) exigem a remoção da bateria do aparelho em
todas as viagens aéreas internacionais quando o concentrador deoxigénio é registado como bagagem. Para transportar o concentrador de oxigénio, a bateria também tem de ser removida dodispositivo e embalada adequadamente.

eQuinox (Model 4000) User Manual — 155

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107 USA
1-800-482-2473
A nível mundial: www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Alemanha
A nível mundial: www.CAIREmedical.com

A CAIRE Inc. reserva-se o direito de suspender os seus produtos, ou alterar os preços, os materiais, o equipamento,
a qualidade, as descrições, as especificações e/ou os processos relativos aos seus produtos a qualquer momento sem
aviso prévio e sem qualquer outra obrigação ou consequência. Todos os direitos não indicados expressamente nopresente documento são reservados por nós, conforme aplicável.

156 — 4805-C1 C (Portuguese)Até à data desta publicação, eQuinox, SeQual e autoSAT e os seus respetivos logótipos (conforme aplicável) são marcas comerciais registadas pela CAIRE Inc.
Lysol® é uma marca registada da Reckitt Benckiser, UK.
OxySafe® é uma marca registada da Applied Home Healthcare Equipment.
Pine-Sol® e Lestoil® são marcas registadas da The Clorox Company.

Gebruikershandleiding

eQuinoxTM

Model 4000
eQuinox (Model 4000) User Manual — 157

Gebruikersbediening en indicatoren van de systeemstatus
SymboolA

02

Betekenis

Betekenis

Lees de gebruikershandleiding voordat u
het apparaat in gebruik neemt. Raadpleeg
de gebruikershandleiding voor instructies.

Door de FAA goedgekeurd symbool: De
Amerikaanse Federal Aviation Administration (FAA)
heeft dit apparaat goedgekeurd voor gebruik aan
boord van een commercieel vliegtuig.

Het pictogram "verboden te roken": rook niet inde nabijheid van het apparaat.

Indicator voor debietinstelling

Waarschuwingen/WAARSCHUWINGS-indicator(geel)

Bediening in pulsmodus

ALARM-indicator (rood) ­Apparaat werkt normaal; aan-uitknop

Externe aan-uitindicator

Instelling om de dosering te verhogen

Accustatusmeter (batterij)Instelling om de dosering te verlagen

Accusymbool

Gebruik geen olie of smeer.

AmpèreGeen open of naakte vlammen.

Wisselstroom

Dit apparaat bevat geen onderdelen die door degebruiker kunnen worden onderhouden. Open het
deksel niet.

IEC-pictogram voor gelijkstroom (invoer)

Warm oppervlak.

Zuurstofuitvoer

Niet nat laten worden.

Type BF-apparatuur (mate van bescherming tegenelektrische schokken)

Druipwaterdichte apparatuur (IPX1): de eQuinoxaccu (batterij) en de wisselstroomvoeding biedenbescherming tegen de schadelijke effecten van
indringende vloeistoffen. (IPX1, overeenkomstig
IEC 60529)

Recyclingsymbool

IP22

Symbool

Dit symbool herinnert gebruikers eraan om
de apparatuur aan het einde van de levensduur
naar een recyclingbedrijf te brengen, conformde AEEA‑richtlijn (afgedankte elektrische enelektronische apparatuur).
Bestand tegen binnendringen van druipwater envoorwerpen: de eQuinox biedt bescherming tegende schadelijke effecten van indringende vloeistoffen
en vreemde voorwerpen overeenkomstig IEC 60529
volgens klasse IP22.
Gecertificeerd voor zowel de Canadese alsde Amerikaanse markt, overeenkomstig detoepasselijke Amerikaanse en Canadese normen.

158 — 4805-C1 C (Dutch)

IPX1

Naam en adres van de fabrikant
Gevolmachtigde vertegenwoordiger in de Europese
Gemeenschap

Klasse II-apparatuur
Dit apparaat voldoet aan de eisen van richtlijn
93/42/EEG betreffende medische hulpmiddelen.
Daarom draagt het de CE-markering zoals
afgebeeld.

Inhoud
Gebruikersbediening en indicatoren van de systeemstatus...................................................................................... 158
Snelstartgids................................................................................................................................................................................ 160
Indicaties voor gebruik............................................................................................................................................................. 163
Contra-indicaties voor gebruik.............................................................................................................................................. 163
Veiligheidsrichtlijnen................................................................................................................................................................ 163
Aanbevolen bedrijfsomgevingen........................................................................................................................................ 165
Gebruiksaanwijzing................................................................................................................................................................... 166
Bediening spraakmodule........................................................................................................................................................ 169
Het eerste gebruik van de eQuinox..................................................................................................................................... 170
Probleemoplossingstabel voor de gebruiker ................................................................................................................. 172
Reiniging, zorg en routineonderhoud................................................................................................................................ 175
EMC-tabel...................................................................................................................................................................................... 177
Algemene informatie voor vliegreizen............................................................................................................................... 181
Accu & vliegreizen...................................................................................................................................................................... 181

eQuinox (Model 4000) User Manual — 159

Snelstartgids

1

Uw eQuinox uitpakken

2

Kennismaking met uw eQuinox

Lees alle waarschuwingen, voorzorgsmaatregelen en
aanvullende informatie in verband met het apparaat in deze
handleiding.
Leer de voornaamste functies van de eQuinox en het
gebruikersbedieningspaneel kennen.
Zuurstofuitgang

VOORZIJDE
HendelBedieningspaneel
Ingang voor externe voeding

Wisselstroomvoeding

Pictogram
FAA‑goedkeuring

Veerslotaccu

Specificatielabel &serienummer

eQuinox-trolley

Accu

ACHTERZIJDE

12V-gelijkstroomkabel

Onderhoudspoort

Afvoerrooster

Accu (batterij)

160 — 4805-C1 C (Dutch)

Luchtinlaatfilter

Het gebruikersbedieningspaneel van deeQuinox geeft belangrijke informatie
weer in verband met de werking.

IndicatorAAN/UIT-knop(groen):metdezeknopwordthet
apparaat in- of uitgeschakeld. De groene indicator brandt
wanneer het apparaat AAN staat en correct werkt.
Regelknoppenomdedoseringteverhogenofteverlagen:gebruik deze knoppen om het debiet te regelen op devoorgeschreven instelling.
Knop toedieningsmodus en indicator: De knop schakelt
tussen "continue dosering" en "pulsdosering". De
pulsdoseringsmodusactiveertdeautoSAT-technologie.Dit
maakt een aanzienlijke toename in bedrijfstijd mogelijk
wanneer aangedreven door de accu (batterij). Wanneer
de pulsdoseringsmodus is geactiveerd, brandt de groene
indicatorvoordepulsdoseringsmodus.Erwordteenzuurstofpulstoegediendendegroeneindicatorknippertbijelke
ademhalingsinspanning.
WAARSCHUWINGS-indicator(geel):wanneerdezebrandt,
wijstditopeensituatieofwaarschuwingmetlageprioriteit.
Ukuntuwsysteemblijvengebruiken.Raadpleegdeprobleemoplossingstabelvoordecorrectehandelingofneem
contact op met uw thuiszorgverlener. Een geel knipperlampjeduidtopeenwaarschuwingmetmiddelhogeprioriteit.
U dient zo snel mogelijk te reageren. (Doorgaans voor een
laag zuurstofpeil of een laag batterijniveau.)
ALARM-indicator (rood): wijst op een alarmsituatie met
hogeprioriteit.Udientonmiddellijktereageren.Raadpleeg
deprobleemoplossingstabelenneemcontactopmetuw
thuiszorgverlener. (Doorgaans doorvoedingsverlies,eenlaag zuurstofpeil of een defect.)
Informatieweergave.
• C of P: "C" staat voor continue doseringsmodus en "P"voor pulsdoseringsmodus.
•Resterendebatterijtijd:wanneerhetapparaatopbatterijvoedingwerkt,vernieuwthetschermelke20secondenmeteen
schatting van de resterende batterijtijd.
•Instelling(continuedoseringofpulsdosering):telkensuhet
apparaatinschakelt,wordtdelaatstemodusen/ofinstellingbewaard en gebruikt bij het opstarten.

3

De eQuinox inschakelen

Aansluiting op de wisselstroomvoeding: steek het
wisselstroomsnoer in de ingang voor externe voeding aan derechterkant van het apparaat. Sluit het wisselstroomsnoer aan op
het stroomnet en steek hem in een wisselstroomstopcontact.
Stap 1		

Stap 2

Stap 3

Wanneer de eQuinox naar behoren is aangesloten
op het stroomnet brandt een groene indicator op dewisselstroomvoeding. De indicator voor externe voeding op
het gebruikersbedieningspaneel brandt eveneens.
Installatie van de batterij: schuif de batterij in het lege vak totdat
deze vastklikt. De batterij ligt dan vlak met de voorkant van het
apparaat.

Wanneer de batterij naar behoren is aangesloten en deeQuinox AAN staat of is aangesloten op het stroomnet,
verschijnt de accustatusmeter.

Accustatusmeter (batterij): deze indicator geeft het resterende batterijvermogen weer. Elk van de vijf horizontale
grijzestavenvertegenwoordigtongeveer20%vandetotale
laadcapaciteit van de batterij. Wanneer de batterij wordt
opgeladen,knipperendelaadindicatorstaventrapsgewijs.
Alsdebatterijnietofnietcorrectisgeïnstalleerd,brandtde
accustatusmeter (batterij) niet of knippert hij aan en uit.
Externe aan-/uitindicator: wanneer de eQuinox naarbehorenisaangeslotenophetstroomnetengebruikmaakt
van wisselstroomvoeding of een 12V-gelijkstroomkabel,brandtdezeindicatorophetgebruikersbedieningspaneel.
Zoemer:eengeluidsalarmwijstuopdebedrijfstoestandvanhetapparaat,bij een waarschuwing of defect, of ter bevestiging van een geldige drukop een knop.
Spraakmodule:indiengeactiveerd,brengteenstemuopdehoogtevande
bedrijfstoestandvanhetapparaat,veranderingeninstroomtoevoerenvan
waarschuwingen of defecten.

Indicator AAN/UIT-knop (groen)
Houd de "AAN/UIT"-knop twee (2) seconden lang ingedrukt
om uw eQuinox AAN te zetten. Er wordt een korte hoor- en
zichtbare zelftest uitgevoerd. Controleer of alle indicatoren
branden en of de zoemer drie (3) seconden lang luidt. Houdde "AAN/UIT"-knop twee (2) seconden lang ingedrukt om uw
eQuinox UIT te schakelen.
eQuinox (Model 4000) User Manual — 161

4

De toedieningsmodus selecteren

Knop doseringsmodus en indicator

Druk op de doseringsmodusknop om de gewenste
doseringsmodus te selecteren. Door herhaaldelijk op deze knop
te drukken, schakelt u tussen continue dosering en pulsdosering.
In de continue doseringsmodus, aangegeven door de letter C,
wordt er zuurstof toegediend met een constant debiet tussen
0,5 en 3,0 l/‌min. In de pulsdoseringsmodus wordt zuurstof
toegediend ineen bolus bij aanvang van elke inademing. Instellingen 1-9 vande pulsdoseringsmodus, aangegeven door de letter P op het
scherm, dienen bolussen toe gaande van 16 ml tot 192 ml. De
gevoeligheid en de toediening van de ademhalingsbolus van depulsdosering kunnen worden aangepast. De doseringsinstellingvoor beide modi moet door uw arts worden voorgeschreven.

5

De toedieningsinstellingen volgens
het voorgeschreven niveau regelen

6

Ingebruikneming van de eQuinox

Luchtinlaatfilter

Luchtinlaatfilter

Uw apparaat is uitgerust met een luchtinlaatfilter (extra
filter meegeleverd). Controleer het filter dagelijks en maak
het wekelijks schoon. Reinigingsinstructies zijn terug tevinden op pagina 19.
Sluit de slang voor zuurstoftoevoer aan op de zuurstofuitlaat
en sluit de zuurstofcanule aan volgens de instructies die bij
de canule werden geleverd, of volgens de instructies van
uw thuiszorgverlener.

Adem normaal door de neuscanule.

Regelknoppen om de dosering te verhogen of te verlagen
Regel de instelling met de knoppen + of - overeenkomstig het
voorschrift van uw arts.

Weergave continue doseringsmodus

Weergave pulsdoseringsmodus

WAARSCHUWING: HET IS ERG BELANGRIJK OM UITSLUITEND HET VOORGESCHREVEN ZUURSTOFNIVEAU TESELECTEREN.WIJZIG DETOEDIENINGSSELECTIE NIETTENZIJDIT WERD AANGEGEVEN DOOR EEN BEVOEGDE CLINICUS. DE DRAAGBARE ZUURSTOFCONCENTRATOR MAG GEBRUIKTWORDEN TIJDENS HET SLAPEN ALS DIT WERD AANBEVOLEN DOOREEN BEVOEGDE CLINICUS.
162 — 4805-C1 C (Dutch)

Gebruik van de pulsdoseringsmodus in combinatie met debatterij leidt tot een langere gebruiksduur.

7

Bediening eQuinox-trolley

Breng de eQuinox op één lijn met de voorgevormde bodem van detrolley zodat deze plat en rechtop staat. Plaats de grote schroef met
schroefdraad in de daartoe voorziene opening aan de achterzijde
van de eQuinox. Draai vervolgens de knop op de trolleyhandmatig vast tot de trolley stevig vastzit aan het apparaat. Drukde drukknop op de telescopische trolleyhendel in en trek hem naar
boven om de hoogte van de hendel aan te passen.

Belangrijk!
Veiligheidsinstructies zijn als volgt vermeld:
WAARSCHUWING: BELANGRIJKE VEILIGHEIDSINFORMATIE
VOOR GEVAREN DIE ERNSTIG LETSEL KUNNEN VEROORZAKEN.
OPGELET: belangrijke informatie om beschadiging van
de eQuinox te voorkomen.
Opmerking: deze informatie vereist extra aandacht.

Indicaties voor gebruik
De eQuinox is bestemd voor de toediening van aanvullende
zuurstof. De apparatuur is niet bedoeld ter ondersteuningof instandhouding van de levensfuncties en biedt ook geenvoorzieningen voor het bewaken van patiënten.
WAARSCHUWING: HET IS ERG BELANGRIJK OM UITSLUITEND HET VOORGESCHREVEN ZUURSTOFNIVEAU TESELECTEREN.WIJZIG DETOEDIENINGSSELECTIE NIETTENZIJDIT WERD AANGEGEVEN DOOR EEN BEVOEGDE CLINICUS. DE DRAAGBARE ZUURSTOFCONCENTRATOR MAG GEBRUIKTWORDEN TIJDENS HET SLAPEN ALS DIT WERD AANBEVOLEN DOOREEN BEVOEGDE CLINICUS.
WAARSCHUWING: VOLGENS DE AMERIKAANSE FEDERALE WETGEVING (VSA) MAG DIT APPARAAT ALLEEN VERKOCHT OF VERHUURDWORDEN OP VOORSCHRIFT VAN EEN ARTS OF EEN ANDERE ERKENDE
PROFESSIONELE ZORGVERLENER.
WAARSCHUWING: DIT APPARAAT MAG NIET ALS LEVENSINSTANDHOUDINGSSYSTEEM WORDEN GEBRUIKT. BIJ GERIATRISCHE, PEDIATRISCHE OF ENIGE ANDERE PATIËNTEN DIE NIET IN STAAT ZIJN EEN
ONGEMAK KENBAAR TE MAKEN, KAN BIJKOMEND TOEZICHT VEREISTZIJN. PATIËNTEN MET AUDITIEVE EN/OF VISUELE BEPERKING(EN)VEREISEN MOGELIJK HULP OP HET VLAK VAN ALARMBEWAKING.

Contra-indicaties voor gebruik
WAARSCHUWING: IN BEPAALDE OMSTANDIGHEDEN
KAN GEBRUIK VAN NIET-VOORGESCHREVEN ZUURSTOFGEVAARLIJK ZIJN. DIT APPARAAT IS BESTEMD VOOR GEBRUIK OP VOORSCHRIFT VAN EEN ARTS.
WAARSCHUWING: NIET BESTEMD VOOR GEBRUIK IN DE NABIJHEIDVAN ONTVLAMBARE ANESTHETICA.
WAARSCHUWING: NET ZOALS BIJ ELK ELEKTRISCH AANGEDREVENAPPARAAT, KAN HET APPARAAT MOGELIJK TIJDELIJK NIET WERKENTEN GEVOLGE VAN EEN STROOMONDERBREKING OF DOOR DE
BEHOEFTE AAN ONDERHOUD DOOR EEN BEVOEGDE MONTEURVAN EQUINOX. DE EQUINOX IS NIET GESCHIKT VOOR PATIËNTEN
DIE SCHADELIJKE GEVOLGEN KUNNEN ONDERVINDEN DOOR EEN
DERGELIJKE ONDERBREKING.

Veiligheidsrichtlijnen
WAARSCHUWING: ER MOGEN GEEN WIJZIGINGEN
WORDEN AANGEBRACHT AAN DEZE APPARATUUR.
WAARSCHUWING: DE FABRIKANT RAADT AAN OM EEN
ALTERNATIEVE BRON VOOR EXTRA ZUURSTOF TE GEBRUIKEN IN GEVAL VAN EEN STROOMUITVAL, ALARMSITUATIESOF MECHANISCHE STORINGEN. RAADPLEEG UW ARTS OF
DE LEVERANCIER VAN UW APPARATUUR VOOR HET VEREISTE SOORTRESERVESYSTEEM.
WAARSCHUWING: DIT APPARAAT LEVERT HOOGZUIVERE ZUURSTOF
DIE EEN SNELLE VERBRANDING BEVORDERT. ZORG ERVOOR DATMEN NIET ROOKT EN DAT ER ZICH GEEN OPEN VLAMMEN BEVINDEN
IN DEZELFDE RUIMTE VAN (1) DIT APPARAAT OF (2) ALLE ZUURSTOFBEVATTENDE ACCESSOIRES. NIET-NALEVING VAN DEZE WAARSCHUWING KAN LEIDEN TOT ERNSTIGE BRAND, MATERIËLE SCHADE
EN/OF FYSIEK LETSEL OF DE DOOD.
WAARSCHUWING: GEBRUIK HET APPARAAT NIET IN EEN BEPERKTEOF BESLOTEN RUIMTE (BIJV. EEN KLEINE KOFFER OF HANDTAS) MET
ONVOLDOENDEVENTILATIE.DIT KAN DE ZUURSTOFCONCENTRATOR
DOEN OVERVERHITTEN EN DE GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: DE CONCENTRATOR MOET GEPLAATST WORDEN
OP EEN LOCATIE ZONDER ROOK, VERVUILENDE STOFFEN OF DAMPEN.
WAARSCHUWING:BEPAALDEZUURSTOFTOEDIENINGSACCESSOIRESDIE NIET ZIJN GENOEMD VOOR GEBRUIK IN COMBINATIE MET
DEZE ZUURSTOFCONCENTRATOR, KUNNEN DE WERKING VAN HETAPPARAAT NADELIG BEÏNVLOEDEN. DE AANBEVOLEN ACCESSOIRESSTAAN VERMELD IN DEZE HANDLEIDING.
WAARSCHUWING: ALS DE ZUURSTOFCONCENTRATOR GEVALLEN, BESCHADIGD OF BLOOTGESTELD IS AAN WATER, NEEM DANCONTACT OP MET UW THUISZORGVERLENER VOOR CONTROLE EN
EVENTUEEL HERSTEL VAN HET APPARAAT. GEBRUIK DE ZUURSTOFCONCENTRATOR NIET ALS HET SNOER OF DE STEKKER BESCHADIGDIS.
WAARSCHUWING: VOORKOM SCHADE OF LETSEL TIJDENS VERVOER
IN EEN VOERTUIG DOOR DE ZUURSTOFCONCENTRATOR STEVIG TE
BEVESTIGEN MET EEN GORDEL OF ANDERSZINS.
WAARSCHUWING: BLOKKEER NOOIT DE OPENINGEN VOOR LUCHTTOEVOER/-AFVOER. STEEK GEEN VOORWERPEN IN DE OPENINGEN
VAN HET APPARAAT EN VOORKOM DAT ER VOORWERPEN IN VALLEN.
DIT KAN DE ZUURSTOFCONCENTRATOR DOEN OVERVERHITTEN EN
DE GOEDE WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: VUL DE OPTIONELE LUCHTBEVOCHTIGER NIETOVERMATIG. VUL DE OPTIONELE LUCHTBEVOCHTIGER MET WATERTOT AAN HET PEIL DAT WORDT OPGEGEVEN DOOR DE FABRIKANTVAN DE LUCHTBEVOCHTIGER.
WAARSCHUWING: U DIENT DE EQUINOX NIET TE GEBRUIKEN EN
ERMEE ROND TE WANDELEN ALS ER EEN LUCHTBEVOCHTIGER AAN
IS BEVESTIGD. VERWIJDER DE FLES VAN DE LUCHTBEVOCHTIGER
VOOR U ZICH ERMEE VERPLAATST. LEG DE EQUINOX NIET NEER MET
DE FLES VAN EEN LUCHTBEVOCHTIGER ERAAN BEVESTIGD.
WAARSCHUWING: DE REGELGEVING VAN HET AMERIKAANSE DEPARTMENT OF TRANSPORTATION (DOT - MINISTERIE VAN VERVOER)
EN DE VERENIGDE NATIES (VN) EISEN DAT DE BATTERIJ UIT HETAPPARAATWORDTVERWIJDERDTIJDENS INTERNATIONALEVLUCHTEN ALS DE ZUURSTOFCONCENTRATOR WORDT INGECHECKT ALSBAGAGE. WANNEER DE ZUURSTOFCONCENTRATOR WORDT VERVOERD, MOET DE BATTERIJ OOK VERWIJDERD EN CORRECT VERPAKTWORDEN.

eQuinox (Model 4000) User Manual — 163

WAARSCHUWING: LEES DEZE HANDLEIDING EN GA NA OF U
DE INHOUD BEGRIJPT VOOR U DE APPARATUUR IN GEBRUIKNEEMT. ALS U DE WAARSCHUWINGEN EN INSTRUCTIES NIETBEGRIJPT, DIENT U CONTACT OP TE NEMEN MET DE LEVERANCIER
VAN UW APPARATUUR ALVORENS U DEZE IN GEBRUIK NEEMT; DITOM LETSEL OF SCHADE TE VERMIJDEN.
WAARSCHUWING: VRAAG ONMIDDELLIJK MEDISCHE HULP ALS U
ONGEMAK OF EEN MEDISCHE NOODTOESTAND ERVAART.
WAARSCHUWING: GEBRUIK DE ZUURSTOFCONCENTRATOR NIET BIJ
TEMPERATUREN DIE BUITEN DE BEDRIJFSSPECIFICATIES LIGGEN, DIT
KAN HET VERMOGEN VAN HET APPARAAT OM DE GESPECIFICEERDEZUURSTOFCONCENTRATIETE BEREIKEN, BEÏNVLOEDEN. RAADPLEEGHET DEEL "SPECIFICATIES" IN DEZE HANDLEIDING VOOR MEER
INFORMATIE OVER HET TEMPERATUURBEREIK.
WAARSCHUWING: GEBRUIK GEEN OLIE, SMEER, PRODUCTEN OP
BASIS VAN PETROLEUM OF ANDERE ONTVLAMBARE PRODUCTENIN COMBINATIE MET ZUURSTOFHOUDENDE ACCESSOIRES OF DEZUURSTOFCONCENTRATOR. GEBRUIK ALLEEN LOTIONS OF ZALVEN
OP WATERBASIS DIE COMPATIBEL ZIJN MET ZUURSTOF. ZUURSTOF
VERSNELT DE VERBRANDING VAN ONTVLAMBARE STOFFEN.
WAARSCHUWING: DOOR INCORRECT GEBRUIK VAN DE BATTERIJKAN DEZE VERHITTEN, VLAM VATTEN EN ERNSTIG LETSEL VEROORZAKEN. DOORBOOR DE BATTERIJ NIET, KLOP ER NIET OP, LAAT HEM
NIET VALLEN, STAP ER NIET OP EN STEL DE BATTERIJ NIET BLOOTAAN STERKE STOTEN OF SCHOKKEN. HET GEBRUIK VAN EEN BESCHADIGDE BATTERIJ KAN LEIDEN TOT PERSOONLIJK LETSEL.
WAARSCHUWING: DE ZUURSTOFCONCENTRATOR MAG NIETWORDEN GEBRUIKT NAAST OF GESTAPELD OP ANDERE APPARATUUR. ALS AANPALEND OF GESTAPELD GEBRUIK ONVERMIJDELIJK
IS, MOET GECONTROLEERD WORDEN OF HET APPARAAT NORMAALWERKT.
OPGELET: raadpleeg de probleemoplossingstabel in deze
handleiding in het geval van een alarm of als u vaststelt
dat de zuurstofconcentrator niet naar behoren werkt.
Neem contact op met de leverancier van uw apparatuur als u er
niet in slaagt om het probleem op te lossen.
OPGELET: om te voorkomen dat de garantie zijn geldigheid verliest, dient u alle aanwijzingen van de fabrikant in acht te nemen.
WAARSCHUWING: VOER GEEN ONDERHOUD UIT DATGEEN DEEL UITMAAKT VAN DE MOGELIJKE OPLOSSINGEN IN DEZE HANDLEIDING. VERWIJDER GEEN DEKSELS.
ALLEEN DE LEVERANCIER VAN UW APPARATUUR OF EEN
BEVOEGDE ONDERHOUDSTECHNICUS MAG DEKSELSVERWIJDEREN
OF ONDERHOUD UITVOEREN.
WAARSCHUWING: GEBRUIK ALLEEN ACCESSOIRES DIE DOOR DE
FABRIKANT ZIJN AANBEVOLEN. GEBRUIK VAN ANDERE ACCESSOIRES KAN GEVAARLIJK ZIJN, ERNSTIGE SCHADE TOEBRENGEN
AAN UW ZUURSTOFCONCENTRATOR EN DE GARANTIE ANNULEREN.
WAARSCHUWING: DE FABRIKANT RAADT AAN OM EEN
ALTERNATIEVE BRON VOOR EXTRA ZUURSTOF TE GEBRUIKEN IN GEVAL VAN EEN STROOMUITVAL, ALARMSITUATIES OF MECHANISCHE STORINGEN. RAADPLEEG UW ARTS OF DE
LEVERANCIER VAN UW APPARATUUR VOOR HET VEREISTE SOORTRESERVESYSTEEM.
WAARSCHUWING: BLOKKEER NOOIT DE OPENINGEN VOOR LUCHTTOEVOER/-AFVOER. STEEK GEEN VOORWERPEN IN DE OPENINGEN
VAN HET APPARAAT EN VOORKOM DAT ER VOORWERPEN IN VALLEN. DIT KAN DE ZUURSTOFCONCENTRATOR DOEN OVERVERHITTENEN DE GOEDE WERKING ERVAN BEÏNVLOEDEN.
164 — 4805-C1 C (Dutch)

WAARSCHUWING: ROKEN TIJDENS GEBRUIK VAN ZUURSTOF IS DE VOORNAAMSTE OORZAAK VAN BRANDWONDEN EN DAARMEE GEPAARD GAAND OVERLIJDEN. VOLG
DEZE VEILIGHEIDSWAARSCHUWINGEN:WAARSCHUWING: STA NIET TOE DAT ER GEROOKT WORDT, DAT ERKAARSEN BRANDEN OF OPEN VUUR IS IN DE KAMER WAAR HET APPARAATOFDEZUURSTOFHOUDENDEACCESSOIRESZICHBEVINDEN.
WAARSCHUWING: ROKEN TERWIJL U EEN ZUURSTOFCANULEDRAAGT, KAN LEIDEN TOT BRANDWONDEN IN HET GELAAT OFZELFS DE DOOD.
WAARSCHUWING: ALS U ROOKT, KUNNEN DE VOLGENDE 3 STAPPENUW LEVEN REDDEN: SCHAKEL DE ZUURSTOFCONCENTRATOR UIT,VERWIJDER DE CANULE EN VERLAAT DE KAMER WAAR HET APPARAAT ZICH BEVINDT.
WAARSCHUWING: PLAATS DUIDELIJKE BORDEN MET "VERBODEN
TE ROKEN - GEBRUIK VAN ZUURSTOF" IN DE WONING OF ELDERS
WAAR ZUURSTOF WORDT GEBRUIKT. PATIËNTEN EN HUN ZORGVERLENERS MOETEN INGELICHT WORDEN OVER DE GEVAREN DIE
ROKEN IN DE AANWEZIGHEID VAN OF BIJ GEBRUIK VAN ZUURSTOFINHOUDT.
WAARSCHUWING: GEBRUIK UW ZUURSTOFCONCENTRATOR NIET
IN DE AANWEZIGHEID VAN ONTVLAMBARE GASSEN. DIT KAN SNEL
BRAND VEROORZAKEN EN LEIDEN TOT MATERIËLE SCHADE, LICHAMELIJK LETSEL OF DE DOOD.
WAARSCHUWING: ALS U DE CANULE AFNEEMT EN PLAATST OP
KLEDING, BEDLINNEN, EEN SOFA OF KUSSENS ONTSTAAT ER EENSTEEKVLAM BIJ BLOOTSTELLING AAN EEN SIGARET, WARMTEBRONOF VLAM.
WAARSCHUWING: PLAATS GEEN NEUSCANULE OP OF ONDER KLEDING, BEDLINNEN OF KUSSENS. ALS HET APPARAAT INGESCHAKELDIS MAAR NIET GEBRUIKT WORDT, ZAL DE ZUURSTOF HET MATERIAALONTVLAMBAAR MAKEN. ZET DE ZUURSTOFCONCENTRATOR IN DE
0-STAND (UIT) WANNEER HET APPARAAT NIET WORDT GEBRUIKT.

OPGELET: koppel altijd de wisselstroomvoeding los vanhet contact voor u de wisselstroomvoeding loskoppelt van
de zuurstofconcentrator.
OPGELET: plaats de zuurstoftoevoerslang en voedingssnoeren altijd
zo dat ze geen gevaar voor struikelen vormen.
OPGELET: bij gebruik van de zuurstofconcentrator in een auto, bootof met andere gelijkstroombronnen met behulp van de 12V-gelijkstroomkabel, moet het voertuig zijn opgestart en draaien alvorens
u de zuurstofconcentrator aansluit. Als het externe voedingslampje
op de eQuinox niet brandt en moet worden teruggesteld, dient u
de 12V-gelijkstroomkabel los te koppelen van het gelijkstroomstopcontact, uw voertuig opnieuw op te starten en vervolgens uw
12V-gelijkstroomkabel opnieuw aan te sluiten op het gelijkstroomstopcontact. Als deze instructies niet worden nageleefd, is het
mogelijk dat de voeding geen stroom levert aan de zuurstofconcentrator.
OPGELET: als de auto waarin u de zuurstofconcentrator gebruikt
uitgeschakeld is, moet het apparaat worden losgekoppeld en uit deauto worden verwijderd. Berg de zuurstofconcentrator niet op in
een heel warme of koude auto of andere, gelijkaardige omgevingen
met een hoge of lage temperatuur.
Opmerking: draagbare en mobiele RF-communicatie-apparatuur
kan elektrische medische apparatuur beïnvloeden.

Aanbevolen bedrijfsomgevingen
Raadpleeg de onderstaande tabel voor het correct gebruik van
het apparaat. De tabel bevat belangrijke informatie over deaanbevolen bedrijfsomgevingen of bedrijfsomstandigheden.

Correcte plaatsing van de eQuinox

Bedrijfstemperatuur 0 ºC tot 40 ºC (32 ºF tot 104 ºF)

Kies een locatie waar geen rook, dampen of luchtverontreiniging
is. Bij een correcte plaatsing van het apparaat kan er voldoende
lucht door het luchtinlaatfilter bovenaan aan de achterkant
van de behuizing stromen. De plaatsing dient ook een vrije
luchtuitstroom mogelijk te maken rond het afvoerrooster rechts
onderaan het apparaat.

Bedrijfsvochtigheid

10% - 95% bij een dauwpunt van 28 °C (82,4 °F)

Plaats het apparaat dusdanig dat de alarmen hoorbaar zijn.

Temperatuur voor
vervoer/opslag-25 °C tot 70 °C (-13 °F tot 158 °F)

Plaats de zuurstoftoevoerslang dusdanig dat er geen plooienof blokkeringen zijn.

Specificaties

Vochtigheid: tot 95% niet condenserend-20 °C tot 50 °C (-4 °F tot 122 °F) voor accu's

Elektriciteit

Gebruik geen verlengsnoer. Gebruik geen elektrische stopcontacten met een schakelknop.

Hoogte0 – 4.000 meter (0 - 13.123 voet)

PlaatsingBlokkeer de luchtinlaat (bovenzijde achteraan) of
deluchtafvoer(achterzijderechtsonderaan)NIET.
Plaats het apparaat op minimaal 7,5 cm (3 inch)
van muren, gordijnen, meubels enz.

Omgeving

Vrij van rook, verontreiniging en dampen.

Bedrijfsduur

24uurperdagindienaangeslotenopeenexterne
wissel- of gelijkstroomvoeding.

Opmerking: als de zuurstofconcentrator voor lange tijd buiten denormalebedrijfstemperatuurwerdopgeborgen,moethetapparaateerst
terug de normale bedrijfstemperatuur bereiken alvorens opnieuw in teschakelen. (Raadpleeg de Specificaties in deze handleiding.)

WAARSCHUWING: GEBRUIK UW ZUURSTOFCONCENTRATOR NIET IN DE AANWEZIGHEID VAN ONTVLAMBAREGASSEN. DIT KAN SNEL BRAND VEROORZAKEN EN LEIDEN
TOT MATERIËLE SCHADE, LICHAMELIJK LETSEL OF DE DOOD. GEBRUIK GEEN OLIE, SMEER, PRODUCTEN OP BASIS VAN PETROLEUM OF
ANDERE ONTVLAMBARE PRODUCTEN IN COMBINATIE MET ZUURSTOFHOUDENDE ACCESSOIRES OF DE ZUURSTOFCONCENTRATOR.
GEBRUIK ALLEEN LOTIONS OF ZALVEN OP WATERBASIS DIE COMPATIBEL ZIJN MET ZUURSTOF. ZUURSTOF VERSNELT DE VERBRANDING
VAN ONTVLAMBARE STOFFEN.
WAARSCHUWING: DIT APPARAAT LEVERT HOOGZUIVERE ZUURSTOF
DIE EEN SNELLE VERBRANDING BEVORDERT. ZORG ERVOOR DATMEN NIET ROOKT EN DAT ER ZICH GEEN OPEN VLAMMEN BEVINDEN
IN DEZELFDE RUIMTE VAN (1) DIT APPARAAT OF (2) ALLE ZUURSTOFBEVATTENDE ACCESSOIRES. NIET-NALEVING VAN DEZE WAARSCHUWING KAN LEIDEN TOT ERNSTIGE BRAND, MATERIËLE SCHADE
EN/OF FYSIEK LETSEL OF DE DOOD.
Opmerking: als de auto waarin u de zuurstofconcentrator gebruikt
uitgeschakeldis,moethetapparaatwordenlosgekoppeldenuitdeauto
worden verwijderd. Berg de zuurstofconcentrator niet op in een heel
warmeofkoudeautoofandere,gelijkaardigeomgevingenmeteenhoge
of lage temperatuur. Laat de zuurstofconcentrator of de voedingsbron
NIET aangesloten op het voertuig als de ontsteking in de uit-stand staat.
Dit kan de batterij van het voertuig ontladen.
OPGELET: gebruik het apparaat niet in een beperkte of
besloten ruimte (bijv. een kleine koffer of handtas) met
onvoldoende ventilatie. Dit kan de zuurstofconcentrator doen oververhitten en de goede werking ervan beïnvloeden.
blokkeer nooit de openingen voor luchttoevoer/-afvoer. Dit kan
de zuurstofconcentrator doen oververhitten en de goede werking
ervan beïnvloeden.

eQuinox (Model 4000) User Manual — 165

Gebruiksaanwijzing
Voor gebruik

Algemene oplaadinformatie

Deze gebruikershandleiding dient als referentie om u het
apparaat te helpen gebruiken en onderhouden. Als u vragen of
twijfels hebt, kunt u contact opnemen met uw thuiszorgverlener.

De eQuinox werkt op externe wisselstroomvoeding,gelijkstroomvoeding of een herlaadbare batterij. Het systeem
bevat een "slimme batterijlader". Deze herlaadt de batterij
wanneer de eQuinox is aangesloten op een externe voedingsbron.
Het systeem bewaakt en controleert de voedingsbron en debatterijlader.

Belangrijk! Probeer NIET om de eQuinox te gebruiken zonder
eerst de Veiligheidsinstructies van deze handleiding te hebben
gelezen. Volg alle gebruiksaanwijzingen. Houd rekening met deWaarschuwingen op het apparaat en in de gebruikershandleiding.
Volg alle veiligheidsmaatregelen om gevaar voor brand,
persoonlijk letsel of ernstige beschadiging van de eQuinox
te voorkomen.
WAARSCHUWING: MAAK DE BEHUIZING, HET BEDIENINGSPANEEL EN HET VOEDINGSSNOER ALLEEN SCHOONMET EEN MILD SCHOONMAAKMIDDEL EN EEN VOCHTIGE(GEEN NATTE) DOEK OF SPONS EN VEEG VERVOLGENS
ALLE OPPERVLAKKEN DROOG. LAAT GEEN VLOEISTOF DOORDRINGEN IN HET APPARAAT. BESTEED BIJZONDERE AANDACHT AAN DEZUURSTOFUITLAAT VOOR DE CANULEAANSLUITING OM TE VERZEKEREN DAT DEZE VRIJ IS VAN STOF-, WATER- EN ANDERE DEELTJES.

OPGELET: blokkeer nooit de openingen voor luchttoevoer/-afvoer. Dit kan de zuurstofconcentrator doen oververhitten en de goede werking ervan beïnvloeden. gebruikde zuurstofconcentrator niet als het luchtinlaatfilter niet
is geïnstalleerd. Als een tweede filter is meegeleverd, plaatst u het"vervangingsfilter" voordat u het vervuilde filter reinigt. Maakhet filter schoon in een warm sopje en laat het volledig drogenvóór gebruik.

Uw eQuinox inschakelen
Het apparaat kan werken met drie (3) verschillendestroombronnen:• Wisselstroomvoeding• 12V-gelijkstroomkabel• Herlaadbare accu (batterij)

De eQuinox laadt de batterij op wanneer wisselstroom- ofgelijkstroomvoeding wordt gebruikt en de batterijtemperatuur
binnen een veilig laadbereik valt. Wanneer de externe
voedingsbron wordt losgekoppeld, schakelt het apparaat
automatisch over op de batterij, zolang de batterij geïnstalleerd en
opgeladen is. Wanneer het apparaat opnieuw wordt aangesloten op
een wisselstroom- of gelijkstroomstopcontact, maakt het apparaat
gebruik van de externe voedingsbron en herlaadt hij de batterij.
Verbind de stroomstekker met de verzonken connector
voor externe stroom om een correcte aansluiting met dewisselstroomvoeding of de 12V-gelijkstroomkabel te bekomen.
De verzonken connector voor externe stroom bevindt zich aan derechterzijde van het apparaat met het bedieningspaneel naar voren.
Duw de stekker in de verzonken connector.
WISSELSTROOMVOEDING

Wisselstroomvoeding

De eQuinox beschikt over een universele wisselstroomvoedingvoor thuisgebruik of elders waar wisselstroom beschikbaar is.
Om een aansluiting met een wisselstroombron te bekomen,sluit u het wisselstroomsnoer stevig aan op de verzonken
connector voor externe stroom. Zorg er eveneens voor dat
het snoer van de wisselstroomvoeding is aangesloten op
een wisselstroomstopcontact. Wanneer de eQuinox naarbehoren is aangesloten, brandt een groene indicator op dewisselstroomvoeding, alsook de indicator voor externe voeding.
De indicator
op het gebruikersbedieningspaneel zal
eveneens branden. De CAIRE wisselstroomvoeding onderdeelnr.
4836 is de voornaamste stroombron voor de eQuinox draagbare
zuurstofconcentrator. Het is een zeer belangrijk onderdeel van het
algehele systeem.
Opmerking:gebruikalleendespanningdiewordtvermeldophettypeplaatje van het apparaat.
Stap 1

166 — 4805-C1 C (Dutch)

Stap 2

Stap 3

WAARSCHUWING: RAAK DE BATTERIJCONTACTEN EN
DE PATIËNT NIET GELIJKTIJDIG AAN WANNEER U HETAPPARAAT AANSLUIT OP DE EXTERNE STROOMVOEDING(WISSELSTROOMVOEDING) EN ER GEEN ACCU IS GEÏNSTALLEERD.
OPGELET: gebruik alleen accessoires die door de fabrikantzijn aanbevolen. Gebruik van andere voorzieningen kan
gevaarlijk zijn, ernstige schade toebrengen aan uw zuurstofconcentrator en de garantie annuleren.
OPGELET: gebruik geen verlengsnoeren in combinatie met dit apparaat en beperk het aantal apparaten dat is aangesloten op hetzelfde
stopcontact. Het gebruik van verlengsnoeren kan de prestaties van
het apparaat negatief beïnvloeden. Indien er toch te veel apparaten
op een stopcontact worden aangesloten, bestaat het risico op overbelasting waardoor de zekeringen kunnen springen. Als de zekeringen niet goed werken, kan dit zelfs brand veroorzaken.
OPGELET: gebruik van niet-gespecificeerde kabels en adapters, metuitzondering van kabels en adapters die worden verkocht door de
fabrikant van de medische elektrische apparatuur als vervangende
onderdelen voor interne onderdelen, kan resulteren in een verhoogde uitstoot van verminderde immuniteit van de zuurstofconcentrator.

12V-gelijkstroomkabel
12V-gelijkstroomkabel

Met een 12V-gelijkstroomkabel kan het systeem werken via
gelijkstroomstopcontacten, zoals die van motorvoertuigen.
1. Start uw voertuig.
2. Steek de 12V-gelijkstroomkabel in de verzonken
voedingsingang aan de zijkant van het apparaat.
3. Sluit de 12V-gelijkstroomkabel aan op het
gelijkstroomstopcontact van uw voertuig.
Wanneer het apparaat naar behoren is aangesloten en stroom
ontvangt van de gelijkstroombron, brandt de indicator voor
externe voeding
op het bedieningspaneel.

eQuinox-gelijkstroomvoedingsaansluiting
Als de eQuinox aangesloten is op gelijkstroomvoeding in een
voertuig, is het volgende van toepassing:• De eQuinox werkt op alle instellingen voor continue dosering
0,5-3,0 l/min en op alle instellingen voor pulsdoseringen 1-9
wanneer het apparaat is aangesloten op de gelijkstroomvoedingin een auto.
• De batterij laadt alleen op wanneer er voldoende stroom overblijft
nadat het apparaat werkt met het voorgeschreven debiet. De
eQuinox gebruikt de stroom in de eerste plaats om het apparaat
te laten werken. Mocht er stroom overblijven, dan zal dit worden
aangewend om de batterij op te laden. Tijdens het opladen van deeQuinox-batterij, is het batterijsymbool trapsgewijs in beweging.
Als de eQuinox-batterij niet oplaadt, blijft het batterijsymbool
vast weergegeven.
WAARSCHUWING: BLOKKEER NOOIT DE OPENINGEN
VOOR LUCHTTOEVOER/-AFVOER. DIT KAN DE ZUURSTOFCONCENTRATOR DOEN OVERVERHITTEN EN DE GOEDE
WERKING ERVAN BEÏNVLOEDEN.
WAARSCHUWING: ALS DE GELIJKSTROOMVOEDINGSBRONVAN HET
VOERTUIG ONDER 11,5 VOLT DAALT, SCHAKELT DE EQUINOX OVER
OP DE BATTERIJ (ALS DE BATTERIJ AANWEZIG EN OPGELADEN IS).
OPGELET: als de auto waarin u de zuurstofconcentrator
gebruikt uitgeschakeld is, moet het apparaat worden
losgekoppeld en uit de auto worden verwijderd. Bergde zuurstofconcentrator niet op in een heel warme of
koude auto of andere, gelijkaardige omgevingen met een hoge oflage temperatuur. Laat de zuurstofconcentrator of de voedingsbron
NIET aangesloten op het voertuig als de ontsteking in de uit-standstaat. Dit kan de batterij van het voertuig ontladen.
Opmerking:alshetapparaatwerktopeen12V-gelijkstroomkabel,kanhetzijn dat de batterij niet wordt opgeladen.
Opmerking: als de eQuinox werkt op een 12V-gelijkstroomkabel, is het
aanbevolen om de pulsdoseringsmodus te gebruiken indien voorgeschreven om een pulsdoseringsapparaat te gebruiken.
Opmerking: om na te gaan of het apparaat gelijkstroom gebruikt, controleertuofdeindicatorvoorexternevoedingbrandt.Alsueenpieptoon
hoort, kan dit duiden op verlies van externe voeding. Controleer of degroeneledlampindicatorvoorexternevoedingaanofuitis.Alsdebatterij
van de gelijkstroom wordt losgekoppeld, is er geen ladingverlies tijdens
gebruikvanhetapparaat.Alsdeindicatornietbrandt,wordthetapparaat
gevoed door de batterij.
Opmerking: de effectieve prestaties van de eQuinox kunnen variëren
naargelangdegebruiksvoorwaardenvanhetmotorvoertuig.Sommigevoertuigenzijnnietinstaatomdevolledigefunctionaliteitteondersteunen
op gelijkstroomvoeding. In het geval dat het voertuig niet in staat is om
voldoende stroom te voorzien om te werken met de huidige instelling,
stopt de eQuinox eerst en vooral met het opladen van de batterij en blijft
hijwerkenaanhetvoorgeschrevendebiet.Alshetvoertuigdannogonvoldoende stroom biedt, verlaagt de eQuinox het debiet als die 2,0 l/min of
hoger is. Als het debiet lager is dan 2,0 l/min, schakelt de eQuinox uit.
Indien u een laag debiet vaststelt, dient u alle andere elektronische
apparatuur van de stroomkring los te koppelen (gps, oplader mobiele
telefoon,enz.).Indienditvoorvoldoendestroomzorgt,wordthetdebiet
nietautomatischverhoogdnaardeoorspronkelijkeinstelling,degebruiker
moetdithandmatigaanpassen.AlsdeeQuinoxnogsteedsnietwerktaan
het voorgeschreven debiet, probeer dan om de accu te verwijderen.
eQuinox (Model 4000) User Manual — 167

Herlaadbare accu (batterij)
De eQuinox kan worden gevoed met demeegeleverde herlaadbare accu (batterij).
Installatie van de batterij: Plaats de batterij
dusdanig dat hij in het lege voedingsvak glijdt.
Duw de batterij in het lege voedingsvak totdat
hij op zijn plaats vastklikt. De batterij ligt dan
vlak met de voorkant van het apparaat. Wanneer
de batterij correct is geïnstalleerd en de eQuinox
AAN staat, wordt de batterijstatusmeter
weergegeven op het bedieningspaneel. (Alser GEEN batterij is geïnstalleerd, brandt deaccustatusmeter niet.) Het apparaat werkt dan
nog steeds op de externe voeding.
Om de batterij te verwijderen, duwt u op
de ontgrendelingsknop bovenaan de accu.
Trek de batterij uit de eQuinox terwijl u deontgrendelingsknop INGEDRUKT houdt.

Herlaadbare accu
(batterij)

Gebruiksduur van de batterij: De gebruiksduur van de batterij
wordt weergegeven op het bedieningspaneel, zowel met een
symbool als in een alfanumeriek formaat. Zolang het apparaat
werkt op batterijvoeding, wordt de batterijstatusmeter te allen
tijde weergegeven. Op het scherm worden de uren/minuten
van de resterende batterijduur weergegeven. De gebruiksduur
wordt beïnvloed door verschillende factoren, zoals dedoseringsregeling, pulsdosering of continue dosering en het
ademhalingsritme. De tabel "Standaard bedrijfsduur voor nieuweaccu" geeft ramingen weer voor de gebruiksduur van de eQuinox
met een nieuwe, volledig opgeladen batterij, op basis van dedebietinstellingen en aangegeven bedrijfsvoorwaarden.
Opmerking:deeQuinoxwordtgeleverdmeteen12-celbatterijbijvertrek
uit de fabriek. De 24-celbatterij is een optioneel accessoire voor een
langere batterijduur.

Batterijtype12-celbatterij

24-celbatterij

1 l/min

2,16 uur (2u, 9 min.)

4,84 uur (4u, 50 min.)2 l/min

1,2 uur (1u, 12 min.)

2,75 uur (2u, 45 min.)3 l/min

0,8 uur (48 min.)

1,83 uur (1u, 49 min.)

16 ml (1,0)

2,62 uur (2u, 37 min.)

5,94 uur (5u, 56 min.)

48 ml (3,0)

2,38 uur (2u, 22 min.)

5,39 uur (5u, 23 min.)

96 ml (6,0)

1,70 uur (1u, 41 min.)3,85 uur (3u, 51 min.)Continue dosering:

Pulsdoseringsmodus (12 BPM)

168 — 4805-C1 C (Dutch)

WAARSCHUWING: VOOR ZUURSTOFCONCENTRATORSMET BATTERIJEN: BEWAAR DE BATTERIJ OP EEN KOELE EN
DROGE PLAATS OM DE LEVENSDUUR ERVAN TE VERZEKEREN. LANGDURIGE OPBERGING VAN DE ZUURSTOFCONCENTRATOR BIJ HOGE TEMPERATUREN OF MET EEN VOLLEDIG GELADEN
OF ONTLADEN BATTERIJ, KAN DE LEVENSDUUR VAN DE BATTERIJBEPERKEN. PROBEER NIET OM DE BATTERIJ TE OPENEN. DE BATTERIJ BEVAT GEEN ONDERDELEN DIE DOOR DE GEBRUIKER KUNNEN
WORDEN ONDERHOUDEN. HOUD BATTERIJEN BUITEN HET BEREIKVAN KINDEREN.

Eerste oplaadbeurt van de batterij
De nieuwe batterij die bij uw eQuinox wordt geleverd, is niet
volledig opgeladen bij vertrek uit de fabriek. Voor u de eQuinox
voor de eerste keer in gebruik neemt, moet de batterij volledig
worden opgeladen.
Sluit aan op de wisselstroomvoeding en installeer de batterij naarbehoren. Laat de batterij volledig opladen. De batterij is volledig
opgeladen wanneer de accustatusmeter op het bedieningspaneel
niet langer trapsgewijs knippert, maar blijft branden.
De eQuinox kan worden gebruikt terwijl de batterij wordt
opgeladen. Het opladen kan tot 4 uur duren, afhankelijk van dedebietinstelling. De batterij laadt eveneens op als de eQuinox isaangesloten op wisselstroomvoeding en is uitgeschakeld.
OPGELET: gebruik alleen batterijen die afkomstig zijn vande fabrikant. Neem contact op met de leverancier van
uw apparatuur of met de bevoegde lokale autoriteiten
voor de afdankingsrichtlijnen voor een correcte afdanking
van batterijen.

* De batterijen zijn recyclebaar. *Standaard bedrijfsduur voor nieuwe accu
Bedrijfsduur

Opmerking: gebruik van de zuurstofconcentrator buiten het normale
bedrijfstemperatuurbereikkandeprestatiesbeïnvloedenendezelfstandige
gebruiksduur van de batterij verminderen en/of de oplaadduur van debatterij verlengen. (Raadpleeg de Specificaties in deze handleiding.) uw
batterijzaleenlangerezelfstandigegebruiksduurgenietenalsudeeQuinox
gebruikt in pulsdoseringsmodus (goedgekeurd door de arts).

Als de gelijk- of wisselstroomvoeding wordt onderbroken,
laat de eQuinox een pieptoon horen en wordt er automatischovergeschakeld op batterijvoeding, zolang de batterij aanwezig
en opgeladen is. Wanneer er opnieuw externe voeding beschikbaar
is, begint de batterij automatisch te herladen. Als de batterij
ontbreekt of volledig ontladen is wanneer de wisselstroom wordt
onderbroken, luidt de eQuinox een alarm en schakelt het apparaat
uit. De batterij is volledig ontladen wanneer de accustatusmeter
op het bedieningspaneel leeg is. Het apparaat geeft een geel
lampje weer en laat een alarm voor een laag batterijniveau luiden.
Dit gaat door totdat het apparaat volledig is uitgeschakeld met een
rood lampje en een constant alarm. Het hoorbare alarm kan tot
5 minuten blijven luiden totdat het apparaat is uitgeschakeld oftotdat externe voeding wordt aangesloten.
Opmerking:houddeAAN/UIT-knopminstens5secondenlangingedrukt
om het hoorbare alarm voor stroomverlies uit te schakelen. Het alarm
wordtookuitgeschakeldwanneerudeeQuinox2minutenaansluitopeen
externe wisselstroomvoedingsbron.

Standaard oplaadduur van de batterij

Spraakmodule in- en uitschakelen

Het duurt gewoonlijk tussen 2 en 5 uur om een volledig ontladenbatterij op te laden tot een capaciteit van 80%, afhankelijk van
de gebruikte debietinstelling. Wanneer de batterij volledig is
opgeladen, branden de staafjes van de batterijmeter niet langer
trapsgewijs. De batterij kan in het apparaat worden opgeladen
wanneer de eQuinox is uitgeschakeld en aangesloten op het
stroomnet.
Als de batterij te warm wordt tijdens het ontladen, start het
opladen pas wanneer de batterij voldoende is afgekoeld. Het
verwijderen van de batterij kan het afkoelen versnellen.
Voor het opladen van de batterij kan men zowel gelijk- alswisselstroombronnen gebruiken. De batterij kan ook worden
opgeladen via de optionele desktoplader.

1. Om de spraakmodule in of uit te schakelen, schakelt u het
apparaat in door op de AAN/Stand-by-knop te drukken.
2. Zodra het apparaat is opgestart, drukt u eenmaal op
het Verboden te roken-pictogram en verborgen toets.
3. Binnen 1-2 seconden verschijnt de menukeuze
SPKR = ON (d.w.z. Luidspreker = aan) of SPKR = OFF(d.w.z. Luidspreker = uit).
4. Druk op de toets voor doseringsverhoging of -verlaging
om te kiezen voor ON (spraakmodule inschakelen) of OFF(spraakmodule uitschakelen).
5. Binnen 3-5 seconden wordt de geselecteerde optie
geactiveerd.
Opmerking: het uitschakelen van de spraakmodule is slechts eenmalig.
Als het apparaat wordt uitgeschakeld en het apparaat wordt teruggesteld, wordt de spraakmodule automatisch geactiveerd.

Volume van mondelinge kennisgevingen verhogen/verlagen
1. Om het volume van de mondelinge kennisgevingen teverhogen of te verlagen, drukt u tweemaal op het Verboden
te roken-pictogram en verborgen toets.

Bediening spraakmodule

2. "Volume = #" (menu voor volume-instelling) verschijnt.

Gebruiksaanwijzing spraakmodule
De eQuinox kan mondelinge kennisgevingen of alarmen en
gebruiksstatussen verschaffen. Deze mondelinge kennisgevingenzijn een aanvulling op de informatie op het ledscherm. De
volgende informatie biedt aanwijzingen om de spraakmodule inen uit te schakelen, het volume te regelen en de taal te wijzigen.

AAN/Stand-by-inAAN/Stand- dicator (groene
ledlamp)by-knop

Alarmindicator(gele ledlamp)

Alarmindicator(rode ledlamp)
Toets voor doseringsverhoging

3. Druk op de toets voor doseringsverhoging of -verlaging
om het volume te verhogen of te verlagen tussen 1 en 10.
4. Zodra de gewenste geluidssterkte werd geselecteerd,
wordt de optie geactiveerd/ingesteld binnen 3-5 seconden.
Taal van de spraakmodule wijzigen
1. Om de taal van de spraakmodule te wijzigen, drukt u vijf
keer op het Verboden te roken-pictogram en verborgen toets
om de taalkeuzes te bekijken.
2. De huidige geactiveerde taal wordt weergegeven op het
scherm. Als de spraakmodule bijvoorbeeld op Engels is
ingesteld wanneer u de taalkeuzemenu opent, verschijnt"English" op het scherm.
3. Druk op de toets voor doseringsverhoging of -verlaging
om te scrollen door de beschikbare talen.

Verboden teroken-pictogram
en verborgen
toets
Doseringsmodustoets
Doseringsmodusindicator(groene ledlamp)

Lcdscherm

Indicator
van externe
voeding
aanwezig(groeneledlamp)Toets voor doseringsverlaging

4. Zodra de gewenste taal wordt weergegeven, wordt deinstelling automatisch ingesteld binnen 3-5 seconden.

eQuinox (Model 4000) User Manual — 169

Het eerste gebruik van de eQuinox
Stap 1: Uw eQuinox plaatsen voor gebruik
Plaats de eQuinox in een goed geventileerde, goed verlichte
ruimte. Zorg ervoor dat de luchttoevoer- en afvoerroosters niet
worden geblokkeerd.
Luchtinlaatfilter
Plaats de eQuinox dusdanig
dat alle hoorbare en visueleindicatoren of alarmen zichtbaar
en hoorbaar zijn.
Controleer of het luchtinlaatfilter
aanwezig is voordat u de eQuinox
inschakelt. Als het luchtinlaatfilter
ontbreekt, neem dan contact op
met uw thuiszorgverlener. Als het
luchtinlaatfilter vuil is, dient u die
te wassen met een warm sopje.
Spoel het filter vervolgens met
schoon water en laat die drogen
alvorens het luchtinlaatfilter
terug in het apparaat te plaatsen.
Vervang het filter indien
nodig door een nieuw, schoon
luchtinlaatfilter.

Vervang de canule en zuurstoftoevoerslang regelmatig,
overeenkomstig de aanbevelingen van uw thuiszorgverlener.
Raadpleeg uw thuiszorgverlener voor informatie over dereiniging en vervanging van slangen en canules.
Opmerking: controleerofdecanulecorrectisgeplaatstengoedvastzit.
Zokandezuurstofconcentratorcorrectdeademhalingdetecterenvoor
detoedieningvanzuurstof.Tijdenshetinademen,moetudezuurstoftoediening via de takken van de neuscanule horen of voelen.

Stap 4: De doseringsmodus selecteren
Druk op de doseringsmodusknop om de modus te selecteren.
Door herhaaldelijk op deze knop te drukken, schakelt u tussen
Pulsdosering en Continue dosering. De correcte modus en
instelling worden voor u voorgeschreven door uw arts.
Werking continue doseringsmodus:
Afvoerrooster

Sluit het apparaat aan op een wisselstroomcontact, eengelijkstroomvoedingsbron of controleer of er een volledig
opgeladen batterij is geïnstalleerd.

Stap 2: Schakel het apparaat IN en laat het
opwarmen.
Houd de "AAN/UIT"-knop twee (2) seconden lang ingedrukt
om uw eQuinox IN te schakelen. Er wordt een korte zichtbare
zelftest uitgevoerd.
Opmerking: als u drie pieptonen hoort bij het aansluiten van het apparaat op de voeding of bij het inschakelen, wijst dit op een lage batterijstand van de 9V-batterij in de behuizing van de eQuinox. Neem contact
opmetuwthuiszorgverlenerenvraagomde9volt-batterijtevervangen.
Alsdedriepieptonenafwisselendblijvenluidennadathetapparaatwerd
ingeschakeld en een aantal minuten werkt, gaat het niet om een lage
batterijstand van de 9V-batterij. Neem contact op met uw zorgverlenervoor een onderhoud.

De eQuinox heeft een ingebouwde indicator voor
zuurstofconcentratie (IZC). De IZC houdt voortdurend
het zuurstofdebiet van het apparaat in de gaten. Er wordt
gebruikgemaakt van een groen lampje (normaal), geel lampje(waarschuwing), rood lampje (alarm) en een hoorbaar alarm.
Deze indicatoren branden allemaal tijdens de zelftest bij het
opstarten. Wacht vijf (5) minuten na de eerste inschakeling van
het apparaat tot het de gespecificeerde prestaties bereikt. Als
alleen het groene indicatorlampje brandt, heeft het apparaat
de gespecificeerde prestaties bereikt.

170 — 4805-C1 C (Dutch)

Stap 3: Sluit de zuurstoftoevoerslang of de neuscanule aan op de zuurstofuitgang.

In de continue doseringsmodus wordt een constantezuurstoftoevoer toegediend door uw slang en neuscanule. De
zuurstof wordt gemeten in liter per minuut (l/min). U mag alleen
een luchtbevochtiger gebruiken in de continue doseringsmodus.
U dient de eQuinox NIET te kantelen, vervoeren of neer teleggen wanneer een fles voor luchtbevochtiging is bevestigd.
Werking pulsdoseringsmodus:
In de pulsdoseringsmodus wordt een bolus van zuurstof(gemeten in milliliter - ml) toegediend bij inademing. Gebruik
GEEN luchtbevochtiger in de Pulsdoseringsmodus.
In de pulsdoseringsmodus dient de eQuinox elke 15 seconden
een continu zuurstofdebiet toe wanneer er geen ademhaling
wordt gedetecteerd. Het apparaat onderbreekt elke 15 seconden
automatisch de continue zuurstoftoevoer om te controlerenof er wordt ingeademd. Wanneer er een inademing wordt
gedetecteerd, dient het apparaat opnieuw een gemeten bolus toe.
Duw op de doseringsmodusknop om de pulsdosering te verlatenen over te gaan naar continue dosering.
Laat uw arts de juiste pulsdosering voorschrijven om teverzekeren dat u de juiste zuurstofverzadiging krijgt. Houdrekening met uw behoeften in rust, bij lichaamsbeweging en
wanneer u reist op grote hoogte. Neem contact op met uw
zorgverlener voor een voorschrift en reisformulieren.

Pulsdoseringsmodus en autoSat-functies
Uw eQuinox is uitgerust met een functie die autoSAT® heet.
De functie dient evenredige zuurstofbolussen toe tot en met
40 ademhalingen per minuut (zie Tabel max. ademhalingsritme
hieronder). Als uw ademhaling versnelt, bijvoorbeeld
bij het wandelen, dient de autoSAT aan de hand van een
servomechanisme de aangewezen bolushoeveelheid toe. autoSATtechnologie is eigendom van CAIRE en wordt uitsluitend gebruikt
in de pulsdoseringsmodus.
Max. ademhalingsritme

Bolusvolume (+/- 15%)

Pulsdoseringsinstelling
1
2
3
4
5
6
7
8
9

Bolushoeveelheid(± 15%) ml

Max. ademhalingsritme

16
32
48
64
80
96
128
160
192

40
40
40
40
37
31
23
18
15

Opmerking:hetbolusvolumevermindertwanneerhetademhalingsritme
het opgegeven bereik overschrijdt.
Opmerking: de instelling van de pulsdosering (bolus) 1-9 is niet gelijk aan l/min.
WAARSCHUWING: VOOR ELKE PATIËNT DIENEN INDIVIDUELE INSTELLINGEN VOOR PULSDOSERING TE WORDENBEPAALD OVEREENKOMSTIG DE BEHOEFTE IN RUST, IN
BEWEGING EN TIJDENS REIZEN.

Stap 5: Regel de debietinstelling op het voorgeschreven niveau
Regel de debietinstelling aan de hand van deDoseerregelingsknoppen overeenkomstig het voorschrift van uw
arts.
WAARSCHUWING: HET IS ERG BELANGRIJK OM UITSLUITEND HET VOORGESCHREVEN ZUURSTOFNIVEAU TE SELECTEREN. WIJZIG DE TOEDIENINGSSELECTIE NIET TENZIJDIT WERD AANGEGEVEN DOOR EEN BEVOEGDE CLINICUS.

Stap 6: Ingebruikneming van uw eQuinox
Adem normaal door de neuscanule.
Opmerking: uwbatterijzaleenlangerezelfstandigegebruiksduurgenieten
als u de eQuinox gebruikt in pulsdoseringsmodus (goedgekeurd door dearts).

Stap 7: Bediening eQuinox-trolley
De eQuinox-trolley is ontworpen voor gebruik op bewandelbare
oppervlakken.
Plaats de eQuinox gelijk met de bodem van de trolley zodat dezevlak en rechtop staat.
Plaats de grote schroef met schroefdraad in de daartoe voorziene
opening aan de achterzijde van de eQuinox. Draai de knop op detrolley vervolgens handmatig vast tot de eQuinox-trolley stevig
vastzit aan het apparaat.
Duw op de drukknop van de hendel van de eQuinox-trolley om
de hoogte ervan te regelen.

WAARSCHUWING: DE EQUINOX IS MOGELIJK NIET IN STAAT ALLE
ADEMHALINGEN WAAR TE NEMEN IN DE PULSDOSERINGSMODUS.
ALS DE EQUINOX NIET REAGEERT OP UW ADEMHALINGEN, VRAAGDAN AAN UW THUISZORGVERLENER OM DE GEVOELIGHEIDSINSTELLING TE CONTROLEREN.
Opmerking:bestemdvoordeleveranciervandeapparatuur:Devolgende
accessoiresvoortoedieningvanzuurstofwordenaanbevolenvoorgebruik
met de zuurstofconcentrator:• Neuscanule met 2,1 m (7 ft.) lange slang: Onderdeelnr. 5408-SEQ•ZiedeAccessoirecatalogus(Onderdeelnr.ML-LOX0010)opwww.CAIREmedical.com voor extra aanbevolen accessoires.
• CAIRE biedt de OxySafe™ als optioneel accessoire. Dit accessoire isbestemdvoorgebruikincombinatiemetdeeQuinox-concentrator.Dit
accessoireisinovereenstemmingmetENISO8359:1996-Amendement
1:2012 en is daarom geschikt voor klanten in regio's waar naleving vandeze norm verplicht is.
DeOxySafe™iseenthermischezekeringomdegasstroomtestoppenals
destroomafwaartsecanuleofzuurstofslanginbrandraaktenverbrandt
tot aan de OxySafe. De OxySafe™ wordt in lijn met de neusadembril ofzuurstofslang geplaatst, tussen de patiënt en de zuurstofuitlaat van deeQuinox.
VooreencorrectgebruikvandeOxySafe,dientusteedsdeaanwijzingenvan de fabrikant te raadplegen (inbegrepen in elke OxySafe-kit).
CAIRE biedt een OxySafe™-kit bestaande uit een hoofdband voor decanule, OxySafe™ en een slang van 2,1 m (7'): Onderdeelnr. 20628667

Opmerking: udientdeeQuinoxNIETtegebruikenenermeerondtewandelenalsereenluchtbevochtigeraanisbevestigd.Verwijderdeflesvande
luchtbevochtiger voor u zich ermee verplaatst. Leg de eQuinox NIET neermet de fles van een luchtbevochtiger eraan bevestigd.

Stap 8: Schakel het apparaat uit
Houd de "AAN/UIT"-knop twee (2) seconden lang ingedrukt om
de eQuinox UIT te schakelen.
Berg de eQuinox maximaal 1 maand lang op een koele en droge
plaats op. De eQuinox moet minstens 2 uur per maand werken
om een goede werking van het apparaat te garanderen.
eQuinox (Model 4000) User Manual — 171

Probleemoplossingstabel voor de gebruiker
Negeer geen alarmen
WAARSCHUWING: DE FABRIKANT RAADT AAN OM EEN ALTERNATIEVE BRON VOOR EXTRA ZUURSTOF TE GEBRUIKEN IN GEVAL VAN
EEN STROOMUITVAL, ALARMSITUATIES OF MECHANISCHE STORINGEN. RAADPLEEG UW ARTS OF DE LEVERANCIER VAN UW APPARATUUR VOOR HET VEREISTE SOORT RESERVESYSTEEM.

Geel lampje
Symptoom

Mogelijke oorzaak

Vast geel lampje. Het alarm
laat elke 30 seconden
1 pieptoon horen.

1) Verwijder de accu en installeer deze opnieuw om zeker te zijn dat hij stevig vastzit
De eQuinox heeft een
in de concentrator.
probleemgedetecteerdmet
2) Installeer indien mogelijk een andere accu in de eQuinox.
de aansluiting op de accu.
3) Neem contact op met uw zorgverlener als dit probleem blijft aanhouden.

Knipperend geel lampje. Het
batterijpictogram knippert. De eQuinox-accu heeft
Het alarm laat elke 2 minuten warm.
1 pieptoon horen.

Wat kunt u doen

1) Sluit de eQuinox opnieuw aan op de externe voedingsbron of plaats een volledig
geladen accu in de eQuinox. Laat de accu 30 minuten afkoelen buiten de concentrator.
2) Herlaad de accu met behulp van wisselstroomvoeding.
3) Vervang de accu als er een ander exemplaar beschikbaar is.
4) Neem contact op met uw zorgverlener als het probleem blijft aanhouden.

Knipperend geel lampje. Het
1) Sluit de eQuinox aan op wisselstroomvoeding en herlaad de accu of vervang dezebatterijpictogramknippert.Het De spanning van de eQui- met en volledig opgeladen accu.
alarm laat elke 30 seconden nox‑accu is laag.
2 pieptonen horen.
1) Maak het inlaatfilter van de behuizing schoon en bevestig die opnieuw.
De eQuinox heeft een
Knipperend geel lampje. Het
belemmeringgedetecteerd
alarm laat elke 2 minuten
die het debiet vermindert
1 pieptoon horen.
of blokkeert.

2) Controleer of de canule niet geplooid of geblokkeerd is. Controleer bij gebruik van
een luchtbevochtiger of de fles correct gevuld is en geen blokkering veroorzaakt.
3) Controleer of de eQuinox naar behoren wordt geventileerd. Het apparaat moet minstens 7,5 cm (3 inch) van andere oppervlakken verwijderd zijn om de openingen niet tebelemmeren.
4) Als het probleem blijft aanhouden, schakel dan over naar een andere zuurstofbronen neem contact op met uw zorgverlener voor ondersteuning.
1) Controleer of het luchtinlaatfilter niet verstopt zit of belemmerd is. Maak het filter
schoon en vervang die indien nodig.

Knipperend geel lampje.
Er is geen hoorbaar alarm.

2) Controleer of de eQuinox in een voldoende geventileerde ruimte geplaatst is.
Controleer of er minstens 7,5 cm (3 inch) vrij is tussen de achterzijde van de eQuinox
De eQuinox heeft een laag en andere voorwerpen (meubels, gordijnen, enz.).
zuurstofpeil gedetecteerd.
3) Controleer bij gebruik in een auto of de achterzijde van de eQuinox niet naar de autostoel gericht is.
4) Als het probleem blijft aanhouden, schakel dan over naar een andere zuurstofbronen neem onmiddellijk contact op met uw zorgverlener.

172 — 4805-C1 C (Dutch)

Rood lampje
Symptoom

Mogelijke oorzaak

Wat kunt u doen
1) Controleer of het luchtinlaatfilter niet verstopt zit of belemmerd is. Maak het filter
schoon en vervang die indien nodig.

2) Controleer of de eQuinox in een voldoende geventileerde ruimte geplaatst is. Controleer of er minstens 7,5 cm (3 inch) vrij is tussen de achterzijde van de eQuinox en andeKnipperendroodlampje.Het
De eQuinox heeft een laag re voorwerpen (meubels, gordijnen, enz.).
alarm laat elke 2 minuten
zuurstofpeil gedetecteerd.
3 pieptonen horen.
3) Controleer bij gebruik in een auto of de achterzijde van de eQuinox niet naar de autostoel gericht is.
4) Als het probleem blijft aanhouden, schakel dan over naar een andere zuurstofbronen neem onmiddellijk contact op met uw zorgverlener.
De eQuinox zat tijdelijk
zonder stroom terwijl het
apparaat werkte op een
externe voedingsbron.

Knipperend rood lampje.
Het alarm laat een constante pieptoon horen. De
lcd-schermen ********. De
eQuinoxdientgeenzuurstof
toe en kan niet worden
De eQuinox verloor stroom
ingeschakeld.
ten gevolge van een ontladen accu (batterij) of een
oververhitte accu.

1) Controleer de externe voedingsbron (wissel- of gelijkstroom) en of de aansluitingenstevig vastzitten op de eQuinox, de transformator en het extern stopcontact.
2) Probeer indien mogelijk of u de eQuinox kunt voeden via een ander stopcontact (wissel- of gelijkstroom).
3) Als het probleem blijft aanhouden, schakel dan over op een andere zuurstofbronen neem onmiddellijk contact op met uw zorgverlener.
1)SluitdeeQuinoxopnieuwaanopdeexternevoedingsbronofplaatseenvollediggeladen accu in de eQuinox. Laat de accu 30 minuten afkoelen buiten de concentrator.
2) Herlaad de accu met behulp van wisselstroomvoeding.
3) Vervang de accu als er een ander exemplaar beschikbaar is.
4) Neem contact op met uw zorgverlener als het probleem blijft aanhouden.

Vast rood lampje. Op het
scherm wordt "FAIL"
De eQuinox ondervindt(mislukt) weergegeven.
DeeQuinoxdientgeenzuur- een systeemfout.
stof toe en kan niet worden
ingeschakeld.

1)Verwijder de accu en koppel de eQuinox los van de wissel- of gelijkstroomvoeding. Het
rodelampjeenhetFAIL-berichtverdwijnenvanhetschermnaloskoppelingvandevoedingsbronnen.Wacht ongeveer 20 seconden, sluit de voedingsbronnen opnieuw aan enprobeer de eQuinox in te schakelen.
2) Als de eQuinox kan worden ingeschakeld, controleer dan of het FAIL-bericht opnieuw
wordt weergegeven. Als dit het geval is, neem dan onmiddellijk contact op met uw
zorgverlener.
3)AlsdeeQuinoxnietopnieuwkanwordeningeschakeld,neemdanonmiddellijkcontact
op met uw zorgverlener.

eQuinox (Model 4000) User Manual — 173

Andere alarmomstandigheden
Symptoom
In de pulsdoseringsmodus knippert het
ledlampje voor pulsdoseringgedurende15
seconden op een snel
tempo.

Mogelijke oorzaak

Wat kunt u doen1) De pulsdoseringsmodus werkt mogelijk niet correct als de gebruiker lijdt aan een conditie die deneusdoorgangen belemmert/blokkeert. (ziekte, afwijkingen van het septum, enz.)

DeeQuinoxkangeenina- 2) Controleer of de canule niet langer is dan 2,1 meter (7 ft.).
deming van de gebruiker
detecteren.Ziepagina15 3) Controleer of de canule geen plooien of blokkering vertoont.
voormeerinformatieover4)Alsereenluchtbevochtigerflesisaangesloten,verwijderdandeflesensluitdecanulerechtstreeks
pulsdosering.
opdeuitgangaan.Depulsdoseringsmoduswerktnietalsereenluchtbevochtigerflesisgeïnstalleerd.
5) Neem contact op met uw zorgverlener als het probleem blijft aanhouden.
1)Controleerofhetstopcontactstroomlevert.Sluitdevoedingsbronnietaanopeendimmercircuitof
meervoudige contactdoos.
2)Controleerofdekabelaansluitingenopdevoedingsbronstevigvastzitteninhetwandcontact/het
voertuigendeconcentrator.Alsuwisselstroomvoedinggebruikt,controleerdanookdeaansluiting
op de transformator van de voedingsbron.

De accu ontlaadt
terwijl het apparaat
De eQuinox ontvangt3) Controleer of de indicatorlampjes van de voeding branden.
is aangesloten op een geenstroomvandeaang- Er brandt een groen lampje op de behuizing van de voeding.
wissel-ofgelijkstroom- esloten voedingsbron.
-ErbrandteengroenlampjeophetbedieningspaneelvandeeQuinox-concentratorindevormvaneen
voedingsbron.
stroomsnoer.

4)Alsdeledlampjesnietbranden,koppeldanalleaansluitingenvandevoeding20secondenlanglos
en sluit het geheel opnieuw aan.
5) Neem contact op met uw zorgverlener als het probleem blijft aanhouden.
1) Sluit de accu aan op de wisselstroomvoeding om de batterij op te laden. Laat de batterij 2 tot 5 uur
De accu was niet volledig opladen en controleer of het batterijpictogram een vol niveau weergeeft en niet knippert.
opgeladen
2) Zie de handelingen voor het symptoom "Accu laadt niet op" hierna.
De zelfstandige geb- De eQuinox werkt niet
ruiksduur van de accu methetverwachtedebiet.
(batterij) is niet zo langals zou moeten.
De accu levert niet degespecificeerdeprestaties.

1) Controleer of u het debiet gebruikt dat werd voorgeschreven door uw arts en dat puls- of continue
dosering wordt gebruikt zoals vereist.
1) Probeer de accu volledig te ontladen. Doe dit door de eQuinox op batterijvoeding te laten werken
tothetapparaatvollediguitschakelt.Verwijdervervolgensdebatterijenlaathemongeveer30minuten
afkoelen. Plaats de batterij terug in de eQuinox en laat hem opladen op wisselstroomvoeding.
2) Als de batterij na het uitvoeren van stap 1 nog steeds niet de verwachte zelfstandige gebruiksduur
biedt, neem dan contact op met uw zorgverlener.
1)Controleerofhetstopcontactstroomlevert.Sluitdevoedingsbronnietaanopeendimmercircuitof
meervoudige contactdoos.
2)Controleerofdekabelaansluitingenopdevoedingsbronstevigvastzitteninhetwandcontact/het
voertuigendeconcentrator.Alsuwisselstroomvoedinggebruikt,controleerdanookdeaansluiting
opxde transformator van de voedingsbron.

DeeQuinoxontvangtgeen3) Controleer of de indicatorlampjes van de voeding branden.
externe wisselstroomvoedingomdebatterijop - Er brandt een groen lampje op de behuizing van de voeding.
te laden.
-ErbrandteengroenlampjeophetbedieningspaneelvandeeQuinox-concentratorindevormvaneen
stroomsnoer.
De accu (batterij) laadt
niet op.

4) Als de ledlampjes niet branden, koppel dan alle aansluitingen van de voeding 20 seconden langlos en sluit het geheel opnieuw aan.
5) Neem contact op met uw zorgverlener als het probleem blijft aanhouden.
1)Verwijderdeaccueninstalleerdezeopnieuwomzekertezijndathijstevigvastzitindeconcentrator.
2) Controleer of u een vast geel lampje ziet op de eQuinox wanneer de batterij is geïnstalleerd of een
DeeQuinoxmaaktonvol- knipperendbatterijpictogramwanneerudebatterijprobeertopteladen.Alséénvanbeidesymptomen
doendeverbindingmetde aanwezig is, ga dan over naar stap 3.
accu om deze op te laden. 3) Installeer indien mogelijk een andere accu in de eQuinox. Als de reservebatterij correct werkt,
dient de oorspronkelijke batterij te worden vervangen.
4) Neem contact op met uw zorgverlener als het probleem blijft aanhouden.

De interne batterij die de 1) Neem contact op met uw zorgverlener.
De eQuinox laat 3
pieptonenhorenbijde alarmen voedt, is bijna
leeg.
eerste inschakeling.
174 — 4805-C1 C (Dutch)

Reiniging, zorg en routineonderhoud
Routineonderhoud

Luchtinlaatfilter

Alleen een door CAIRE opgeleide en bevoegde
onderhoudstechnicus mag werkzaamheden uitvoeren aan deinterne onderdelen van de eQuinox.

Via de luchtinlaatopening bovenaan de achterzijde van
het apparaat wordt omgevingslucht opgenomen. Het filter
schoonmaken is het belangrijkste onderhoud dat u als gebruiker
zult uitvoeren. Door het luchtinlaatfilter schoon te houden,
zal uw eQuinox correct blijven werken. Controleer het
luchtinlaatfilter dagelijks en maak het filter minstens eenmaal
per week schoon aan de hand van de volgende procedure:

Elk jaar dient een preventief onderhoud (PO) te worden
uitgevoerd. Raadpleeg uw zorgverlener om een PO-schema
op te stellen. Uw thuiszorgverlener of een bevoegde technicus
zal de volgende onderdelen alleen inspecteren indien vereist:
alarmen, interne 9V-batterij, batterij, interne filters en andere
interne onderdelen.
Opmerking: als het apparaat 3 pieptonen laat horen na de zelftest bij
inschakeling, wijst dit op een bijna lege 9V-accu. Neem contact op met
uw zorgverlener als u drie pieptonen hoort.

Zorg en reiniging van het apparaat door de gebruiker
Vervanging van de canule

1. Verwijder het luchtinlaatfilter;
2. Was het filter in een sopje van warm kraantjeswater en een
milde zeep;
3. Spoel het filter grondig met helder kraantjeswater en knijp
het overtollige water uit;
4. Laat het filter drogen aan de lucht;
5. Plaats het filter terug in de behuizing.
Reinigingstip: voorzie een tweede schone filter ter vervanging
voor het eerste terwijl die droogt.

Vervang uw toevoerslang en canule net zo vaak als aangeraden
door uw thuiszorgverlener. Uw arts of thuiszorgverlener zal u
informatie geven over reiniging, ontsmetting en vervanging.
Opmerking: volg altijd de instructies voor correct gebruik van defabrikant van de canule. Plaats de wegwerpcanule terug volgens de
instructies van de fabrikant ervan of van de leverancier van uw apparatuur.Bijkomendebenodigdhedenzijnbeschikbaarbijdeleveranciervan
uw apparatuur.

Luchtinlaat
Filter

OPGELET: gebruik de zuurstofconcentrator niet als het
luchtinlaatfilter niet is geïnstalleerd. Als een tweede filter
is meegeleverd, plaatst u het "vervangingsfilter" voordat
u het vervuilde filter reinigt. Maak het filter schoon in een warm
sopje en laat het volledig drogen vóór gebruik.
Opmerking: de fabrikant raadt niet aan om deze apparatuur te steriliseren.
Opmerking:gebruikdezuurstofconcentratornietalshetluchtinlaatfilter
niet is geïnstalleerd. Als een tweede filter is meegeleverd, plaatst u het"vervangingsfilter" voordat u het vervuilde filter reinigt. Maak het filter
schoon in een warm sopje en laat het volledig drogen vóór gebruik.

Het luchtinlaatfilter moet jaarlijks worden vervangen. Als deeQuinox in een stoffige omgeving wordt gebruikt, kan het nodig
zijn om het filter vaker te vervangen. Laat de eQuinox niet
langer dan 5 minuten werken zonder luchtinlaatfilter.

eQuinox (Model 4000) User Manual — 175

Wisselstroomvoedingsbron, 12V-gelijkstroomkabel, stroomsnoer en behuizing:
Schakel de eQuinox UIT en koppel los van de gelijk- ofwisselstroom alvorens het apparaat schoon te maken of tedesinfecteren. Spuit NIET rechtstreeks op de behuizing.
Gebruik een vochtige (geen natte) doek of spons. Bespuit dedoek of spons met een milde zeepoplossing om de behuizing
en voedingsbron schoon te maken. Gebruik Lysol® Brand IIontsmettingsmiddel (of een gelijkaardig product) om de eQuinox
te ontsmetten. Volg de aanwijzingen van de fabrikant van het
schoonmaakproduct.
WAARSCHUWING: RISICO OP ELEKTRISCHE SCHOKKEN.
SCHAKEL HET APPARAAT UIT EN HAAL DE STEKKERUIT HET STOPCONTACT VOORDAT U HET APPARAATSCHOONMAAKT. ZO VOORKOMT U MOGELIJKE ELEKTRISCHE SCHOKKEN OF RISICO OP BRANDWONDEN. ALLEEN UWLEVERANCIER VAN DE APPARATUUR OF EEN BEVOEGD ONDERHOUDSTECHNICUS MAG DE BEHUIZING VERWIJDEREN OF HET APPARAAT ONDERHOUDEN. GEBRUIK GEENVLOEISTOF RECHTSTREEKSOP HET APPARAAT. DE LIJST MET AFGERADEN CHEMISCHE STOFFENOMVAT, MAAR IS NIET BEPERKT TOT: ALCOHOL EN PRODUCTEN OP
ALCOHOLBASIS,GECONCENTREERDEPRODUCTENOPCHLOORBASIS(ETHYLEENCHLORIDE) EN PRODUCTEN OP OLIEBASIS (PINE-SOL®,LESTOIL®). DEZE STOFFEN MOGEN NIET GEBRUIKT WORDEN OM DE
PLASTIC BEHUIZINGVAN DE ZUURSTOFCONCENTRATOR SCHOONTE
MAKEN AANGEZIEN ZE HET PLASTIC KUNNEN BESCHADIGEN.
WAARSCHUWING: RISICO OP ELEKTRISCHE SCHOKKEN. TREK HETVOEDINGSSNOER UIT HET WANDSTOPCONTACT VOORDAT U HETAPPARAAT SCHOONMAAKT OM ELEKTRISCHE SCHOKKEN EN
BRANDGEVAAR TE VOORKOMEN. ALLEEN UW LEVERANCIER VAN DE
APPARATUUR OF EEN BEVOEGD ONDERHOUDSTECHNICUS MAG DE
BEHUIZING VERWIJDEREN OF HET APPARAAT ONDERHOUDEN.
WAARSCHUWING:VOORKOM DAT DE ZUURSTOFCONCENTRATOR EN
DE BATTERIJ NAT WORDEN OF DAT ER VLOEISTOF BINNENDRINGTIN HET APPARAAT. DIT KAN LEIDEN TOT EEN STORING OF KAN HETAPPARAAT UITSCHAKELEN EN EEN VERHOOGD RISICO OP ELEKTRISCHE SCHOKKEN OF BRANDWONDEN MET ZICH MEEBRENGEN.
WAARSCHUWING: MAAK DE BEHUIZING, HET BEDIENINGSPANEELEN HET VOEDINGSSNOER ALLEEN SCHOON MET EEN MILDSCHOONMAAKMIDDEL EN EEN VOCHTIGE (GEEN NATTE) DOEKOF SPONS EN VEEG VERVOLGENS ALLE OPPERVLAKKEN DROOG.
LAAT GEEN VLOEISTOF DOORDRINGEN IN HET APPARAAT. BESTEEDBIJZONDERE AANDACHT AAN DE ZUURSTOFUITLAAT VOOR DE CANULEAANSLUITING OM TE VERZEKEREN DAT DEZE VRIJ IS VAN STOF-,WATER- EN ANDERE DEELTJES.

WAARSCHUWING: GEBRUIK GEEN VLOEISTOF RECHTSTREEKS OP HET APPARAAT. DE LIJST MET AFGERADEN
CHEMISCHE STOFFEN OMVAT, MAAR IS NIET BEPERKT TOT:ALCOHOL EN PRODUCTEN OP ALCOHOLBASIS,GECONCENTREERDE PRODUCTEN OP CHLOORBASIS (ETHYLEENCHLORIDE) EN
PRODUCTEN OP OLIEBASIS (PINE-SOL®, LESTOIL®). DEZE STOFFEN
MOGEN NIET GEBRUIKT WORDEN OM DE PLASTIC BEHUIZING VAN
DE ZUURSTOFCONCENTRATOR SCHOON TE MAKEN AANGEZIEN ZE
HET PLASTIC KUNNEN BESCHADIGEN.

176 — 4805-C1 C (Dutch)

Optionele luchtbevochtiger
indien in gebruik maakt u de luchtbevochtiger dagelijks
schoon om mogelijke besmetting te voorkomen. Volg dereinigingsaanbevelingen van de thuiszorgverlener of defabrikant van de luchtbevochtiger. Vervang de luchtbevochtiger
maandelijks of overeenkomstig de aanbevelingen van defabrikant of uw thuiszorgverlener. (Vul de luchtbevochtiger
niet overmatig.)
OPGELET: u dient de eQuinox NIET te gebruiken en ermeerond te wandelen als er een luchtbevochtiger aan isbevestigd. Verwijder de fles van de luchtbevochtiger voor
u zich ermee verplaatst. Leg de eQuinox NIET neer met defles van een luchtbevochtiger eraan bevestigd.

eQuinox-trolley
Haal de trolley van de eQuinox voordat u de trolleyschoonmaakt. Gebruik een vochtige (geen natte) doek of sponsen een milde zeepoplossing om de trolley schoon te maken.
Accu (batterij)
De batterij in de eQuinox dient met speciale zorg te worden
behandeld om een langere levensduur en optimale prestaties
te garanderen. De CAIRE-batterij is de enige batterij die wordt
aanbevolen voor gebruik met de eQuinox.
Maak de batterij schoon met een vochtige (geen natte)
doek of spons. Bespuit de doek of spons eerst met een mild
schoonmaakmiddel en maak vervolgens de behuizing en devergrendeling van de accu schoon.
CAIRE raadt aan de accu om de 2 jaar te vervangen.
Afdanking van de batterij
Uw batterij is herlaadbaar en recyclebaar. Bezorg afgedankte
batterijen aan uw thuiszorgverlener voor een correcte afdanking.
U kunt ook contact opnemen met de lokale autoriteiten voor
inlichtingen over een correcte afdanking van de batterij.
Afdanking van eQuinox-zuurstofsysteem
Het kan zijn dat plaatselijke milieuvoorschriften u niet toestaanelektrische en/of elektronische apparatuur zoals de eQuinox
of wisselstroomvoeding af te danken. Neem contact op met delokale autoriteiten voor inlichtingen over het correcte afdanken
van elektrische en elektronische apparatuur. U kunt ook contact
opnemen met CAIRE Inc. voor informatie over afdanking.

EMC-tabel
Leidraad en Verklaring van de fabrikant — Elektromagnetische emissies
DeeQuinoxisbestemdvoorgebruikindehierondergespecificeerdeelektromagnetischeomgeving.DeklantofdegebruikervandeeQuinox
dient te verzekeren dat het in een dergelijke omgeving wordt gebruikt.
Emissietest
RF-emissies
CISPR 11
RF-emissies
CISPR 11

Naleving

Elektromagnetische omgeving - Leidraad

Groep 1

DeeQuinoxgebruiktalleenRF-energievoorzijninternewerking.DeRF-emissies
zijn daarom erg laag en het is onwaarschijnlijk dat deze enige storing zouden
veroorzaken in elektronische apparatuur die zich dichtbij bevindt.

Klasse B

Harmonische emissies
IEC 61000-3-2

Klasse A

Spanningsfluctuaties/flikkeremissies
IEC 61000-3-3

Conform

De eQuinox is geschikt voor gebruik in alle instellingen, waaronder instellingen
met een woonfunctie en instellingen die rechtstreeks zijn aangesloten op een
laagspanningsnetwerkdatgebouwenmeteenwoonfunctievanstroomvoorziet.

eQuinox (Model 4000) User Manual — 177

Leidraad en Verklaring van de fabrikant — Elektromagnetische immuniteit
DeeQuinoxisbestemdvoorgebruikindehierondergespecificeerdeelektromagnetischeomgeving.DeklantofdegebruikervandeeQuinox dient te verzekeren dat het in een dergelijke omgeving wordt gebruikt.
Immuniteitstest

IEC 60601-testniveau

Nalevingsniveau

Elektromagnetischeomgeving - Leidraad±6 kV contact±6 kV contact±8 kV lucht±8 kV lucht

IEC 61000-4-2
Snelle elektrische transiënten/burst
IEC 61000-4-4
Stroomstoten
IEC 61000-4-5

Spanningsdalingen,korte onderbrekingenenspanningsfluctuaties
bij ingangslijnen van
de stroomvoorziening
IEC 61000-4-11

Netfrequentie (50/60
Hz) magnetisch veld
IEC 61000-4-8±2 kV voor voedingskabels±1 kV voor ingangs-/uitgangsleidingen±2 kV voor voedingskabels
N.v.t.

±1 kV lijn(en) naar lijn(en) ±1 kV lijn(en) naar lijn(en)±2 kV lijn(en) naar aarde<5% UT<5% UT(>95% daling in UT )(>95% daling in UT )voor 0,5 cyclus

voor 0,5 cyclus

40% UT

40% UT(60% daling in UT )(60% daling in UT )voor 5 cycli

voor 5 cycli

70% UT

70% UT(30% daling in UT )(30% daling in UT )voor 25 cycli

voor 25 cycli<5% UT<5% UT(>95% daling in UT )(>95% daling in UT )voor 5 secvoor 5 sec

3 A/m

3 A/m

Elektromagnetische omgeving - Leidraad
Houten, betonnen of keramische tegelvloer
vereist. Indien de vloerbedekking bestaat uit
synthetisch materiaal, dient de relatieve luchtvochtigheid minimaal 30% te zijn.
De netvoedingskwaliteit dient die van een typischecommerciëleofziekenhuisomgevingtezijn.

De netvoedingskwaliteit dient die van een typischecommerciëleofziekenhuisomgevingtezijn.

De netvoedingskwaliteit dient die van een typischecommerciëleofziekenhuisomgevingtezijn.
Indiendegebruikereenconstantewerkingvereist
van de eQuinox gedurende stroomonderbrekingen, is het aanbevolen de eQuinox aan te sluiten
opeenniet-onderbreekbarestroomtoevoerofeen
batterij.

De magnetische velden van de netfrequentie
dienen die van een typische commerciële of ziekenhuisomgeving te zijn.

OPMERKING UT is de spanning van de wisselstroomnetvoeding voorafgaand aan de toepassing van het testniveau.

178 — 4805-C1 C (Dutch)

Leidraad en verklaring fabrikant – Elektromagnetische immuniteit
DeeQuinoxisbestemdvoorgebruikindehierondergespecificeerdeelektromagnetischeomgeving.DeklantofdegebruikervandeeQuinox dient te verzekeren dat het in een dergelijke omgeving wordt gebruikt.
Immuniteitstest

IEC 60601-testniveau

Nalevingsniveau

Elektromagnetische omgeving - Leidraad

Geleide RF

3 Vrms3 VrmsIEC 61000-4-6

150 kHz tot 80 MHz

Uitgestraalde RF

3 V/mDraagbareenmobieleRF-communicatieapparatuur mag niet dichter bij onderdelen van de eQuinox, inclusief de kabels, worden gebruikt dan deaanbevolenscheidingsafstanddiewerdberekend
a.d.h.v. de vergelijking die toepasselijk is op defrequentie van de zender.

IEC 61000-4-3

80 MHz tot 2,5 GHz

3 V/mAanbevolen scheidingsafstandd 8 1,2 √ P= 80 MHz tot 800 MHz

d 8 1,2 √ P= 800 MHz tot 2,5 GHz

d 8 1,2 √ P

waarbijPstaatvoorhetnominalemaximumvermogenvandezenderinwatt(W)volgensdefabrikant
van de zender en d voor de aanbevolen afstand inmeter (m).
Veldsterktes van stationaire RF-zenders, zoals
bepaald door een
overzichtvanhetelektromagnetischveld,adienen
lagertezijndanhetnalevingsniveauinelkfrequentiebereik. b
Interferentie kan voorkomen in de nabijheid van
apparatuur waarop het volgende symbool staat:

OPMERKING 1 Bij 80 MHz en 800 MHz geldt het hogere frequentiebereik.
OPMERKING 2 Het kan zijn dat deze leidraad niet op alle situaties toepasselijk is. Elektromagnetische propagatie wordt beïnvloed door
absorptie en reflectie van gebouwen, voorwerpen en mensen.
Veldsterktesvanstationairezenderszoalsbasisstationsvoor(mobiele/draadloze)radiotelefoonsenlandmobieleradio's,amateurradio-,AM-enFM-radio-uitzendingenentelevisie-uitzendingen,kunnennietnauwkeurigtheoretischwordenvoorspeld.OmdeelektromagnetischeomgevingvanstationaireRF-zendersteevalueren,kanmeneenelektromagnetischterreinonderzoekuitvoeren.Wanneer
degemetenveldsterkteopdeplekwaardeeQuinoxgebruiktwordtbovenhethierbovenweergegevenRF-nalevingsniveauuitkomt,moetde eQuinox in de gaten worden gehouden om te controleren of hij normaal werkt. Als u constateert dat het apparaat niet normaal werkt,
kan het nodig zijn om extra maatregelen te treffen, zoals het anders richten of het verplaatsen van de eQuinox.

abBinnen het frequentiebereik van 150 kHz tot 80 MHz dienen de veldsterktes minder dan 3 V/m te zijn.

eQuinox (Model 4000) User Manual — 179

Aanbevolen scheidingsafstandtussen draagbare en mobiele RF-communicatieapparatuur en de eQuinox.
DeeQuinoxisbestemdvoorgebruikineenelektromagnetischeomgevingwaarinuitgestraaldeRF-storingenwordenbeheerd.Deklantof
degebruikervandeeQuinoxkanelektromagnetischestoringhelpenvoorkomendooreenminimalescheidingsafstandtussendedraagbareenmobieleRF-communicatieapparatuur(zenders)endeeQuinoxinachttenemenzoalshieronderaanbevolen,overeenkomstighet
maximumvermogen van de communicatieapparatuur.
Nominaal maximumvermogen van zender

Scheidingsafstand volgens frequentie van de zenderm
150 kHz tot 80 MHz

80 MHz tot 800 MHz

W

800 MHz tot 2,5 GHz

d 8 1,2 √ P

d 8 1,2 √ P

0,01

0,12

0,12

0,23

0,1

0,38

0,38

0,73

1

1,2

1,2

2,3

10

3,8

3,8

7,3

100

12

12

23

d 8 1,2 √ P

Voorzendersmeteenandernominaalmaximumvermogendanhierbovenvermeld,kandeaanbevolenscheidingsafstandinmeter(m)worden geschat met behulp van de vergelijking die van toepassing is op de frequentie van de zender, waarbij P staat voor het nominale
maximumvermogen van de zender in watt (W) volgens de fabrikant van de zender.
OPMERKING 1 Bij 80 MHz en 800 MHz geldt het hogere frequentiebereik.
OPMERKING 2 Het kan zijn dat deze leidraad niet op alle situaties toepasselijk is. Elektromagnetische propagatie wordt beïnvloed door
absorptie en reflectie van gebouwen, voorwerpen en mensen.

180 — 4805-C1 C (Dutch)

Algemene informatie voor vliegreizen
Tijdens het taxiën, opstijgen en landen, moet de eQuinox
uitgeschakeld en opgeborgen worden onder de stoel of op een
andere geschikte plaats, zodat de gang of de doorgang voor derij niet geblokkeerd is wanneer de eQuinox niet wordt gebruikt.
Tijdens het taxiën, opstijgen en landen, moet de gebruiker zich
op een plaats bevinden die de toegang tot of het gebruik van
een gewone of nooduitgang, evenals de rij met zitplaatsen voor
andere passagiers niet belemmert wanneer de eQuinox in gebruikis.
De gebruiker mag zich niet in een uitgangrij bevinden bij gebruik
van de eQuinox.
Als de eQuinox in gebruik is bij decompressie van de cabine
en het zuurstofsysteem van de cabine wordt ingeschakeld, dientde gebruiker het gebruik van de eQuinox te stoppen en overte gaan op het zuurstofsysteem van het vliegtuig. De eQuinox
moet worden uitgeschakeld wanneer het zuurstofsysteem van
het vliegtuig in gebruik is.
Voor de reis moet de gebruiker controleren of de eQuinox zichin goede staat van werking bevindt.

Accu & vliegreizen
Houd ook rekening met de tijd buiten devlucht zelf (check-in, veiligheidscontrole enz.)
en reisonderbrekingen.
De meeste luchtvaartmaatschappijen vereisen dat u over een
geschikt aantal volledig opgeladen batterijen beschikt om het
toestel van stroom te voorzien gedurende minstens 150% vande verwachte maximale duur van de reis (vluchttijden, tijd op
de grond, voor en na de vlucht, tijdens verbindingen en voor
onverwachte vertragingen).

Het is aanbevolen om steeds een extra batterij
te voorzien wanneer u reist.
Elke luchtvaartmaatschappij heeft specifieke vereisten voor
reizen met zuurstof.
Neem minstens 48 uur voor vertrek contact op met deluchtvaartmaatschappij om te laten weten dat u met eQuinox
zult reizen.

OPGELET: de regelgeving van het Amerikaanse Department of Transportation (DOT - Ministerie van Vervoer)
en de Verenigde Naties (VN) eisen dat de batterij uit
het apparaat wordt verwijderd tijdens internationale
vluchten als de zuurstofconcentrator wordt ingecheckt als bagage.
Wanneer de zuurstofconcentrator wordt vervoerd, moet de batterij
ook verwijderd en correct verpakt worden.

eQuinox (Model 4000) User Manual — 181

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107 VSA
1-800-482-2473
Wereldwijd: www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Duitsland
Wereldwijd: www.CAIREmedical.com

CAIREInc.behoudtzichhetrechtvooromzijnproductietebeëindigenendeprijzen,materialen,apparatuur,kwaliteit,
beschrijvingen, specificaties en/of processen van zijn producten op elk moment te wijzigen, zonder voorafgaande
kennisgeving en zonder verdere verplichtingen of gevolgen. Alle hierin niet uitdrukkelijk genoemde rechten zijn
voorbehouden, waar van toepassing.

182 — 4805-C1 C (Dutch)

Met ingang van de datum van deze publicatie zijn eQuinox, SeQual en autoSAT en hun
bijhorende logo's (indien van toepassing) gedeponeerde handelsmerken van CAIRE Inc.
Lysol® is een gedeponeerd handelsmerk van Reckitt Benckiser, VK.
OxySafe® is een gedeponeerd handelsmerk van Applied Home Healthcare Equipment.
Pine-Sol® en Lestoil® zijn gedeponeerde handelsmerken van The Clorox Company.

eQuinox (Model 4000) User Manual — 183

CAIRE Inc.®
2200 Airport Industrial Drive, Ste. 500
Ball Ground, GA 30107 USA
1-800-482-2473
Worldwide: www.CAIREmedical.com

Medical Product Services GmbH
Borngasse 20
35619 Braunfels, Germany
Worldwide: www.CAIREmedical.com

CAIRE Inc. reserves the right to discontinue its products, or change the prices, materials, equipment, quality, descriptions, specifications and/or processes to its products at any time without prior notice and with no further obligation orconsequence. All rights not expressly stated herein are reserved by us, as applicable.

184 — 4805-C1 C (English)

As of the date of this publication, eQuinox, SeQual and autoSAT and their related
logos (as applicable) are trademarks registered by CAIRE Inc.
Lysol® is a registered trademark of Reckitt Benckiser, UK.
®
OxySafe is a registered trademark of Applied Home Healthcare Equipment.
Pine-Sol® and Lestoil® are registered trademarks of The Clorox Company.
PN 4805-C1 C