Communication analyzer with touch panel and color LCD

Supports RS-232C, RS-422/485, TTL (UART, I2C, SPI)

Multi-protocol analyzerLE-2500XR

NEW

Arbitrary speed up to 2 Mbps
100MB buffer memory
Up to 32GB auto save function
Wi-Fi remote control
The screen can be scrolled byswiping. The key operation is alsoavailable

HDLC/SYNC/ASYNC standard compatible, CAN/CAN FD/CXPI/LIN optional.
● Multi interface / Multi-Protocol supported
Equipped with RS-232C, RS-422/485, TTL (1.8/2.5/3.3/5V)
measurement ports.

Standard support:
ASYNC (start-stop synchronization), SYNC/BSC,HDLC/SDLC, PPP(ASYNC), I2C, SPI, BURST
Optional support:
CAN FD/CAN, CXPI,
LIN, X.20/21,
RS-449, V.35,
RS-530,Current loop

● Safe and reliable Auto-Save function
The measurement log file can be continuously recorded for along time into a SDHC card or a USB flash drive. Even if thebattery runs out suddenly, the log file being written is safely
closed and measurement is completed, so valuable records
can be reliably saved.
DUT
speed(bps)9600
115.2K
1M

Estimated continuous recording time *100MB
Main memory only

32GB SDHC card
Using SD-32GX

Approx. 6hrs.
Approx. 28min.
Approx. 200sec.

Approx. 80days
Approx. 6.5days
Approx. 20hrs.

*: Calculated when the 1 Kbyte data transmission
period and 1 msec idle time are repeated.

● Convenient signal amplitude measurement
function and logic analyzer function
Expandable by replacing the board
or adding a dedicated cable● Monitor, simulation, BERT available
Many test functions and modes are available that are suitable
for analysis purposes and test scenes.

● Utilize measurement data on a PC
Includes PC link software (light version). Wi-Fi connection
between analyzer and PC is also available.

LE-2500XR

specifications

Standardmeasurement port
Expansion
measurement port
Standard protocol
Expansion protocol
Capture memory
Measurable speed
Line signaling mode
Data display code
Measurement
condition setting
Online monitor function

Error check function
Idle time measurement

Time stamp record
Line status record
Interval timer
General counter
Data counter
Trigger function
Trigger condition

Trigger action

Monitor conditions auto
setting

Auto run/stop function
Auto save function

Delay time function

RS-232C, RS-422/485,
TTL (1.8V/2.5V/3.3V/5V level)
CAN, CAN FD, CXPI, LIN, X.20/21,
RS-449, V.35, RS-530, current loop
ASYNC, SYNC/BSC, HDLC/SDLC,
PPP(ASYNC), I2C, SPI, MODBUS, BURST
CAN, DeviceNet *1, CAN FD, CXPI, LIN
100 MB *2
Can be split into 250bps ~ 1.000Mbps(Available with 4 significant figures)
NRZ, NRZI, FM0, FM1
ASCII, EBCDIC, JIS, Baudot, Transcode,
IPARS, EBCD, EBCDIK, HEX
Protocol, data/parity/stop bit, sync clock, frameend, address filter, bit transmission order, bitpolarity, etc. can be specified according to theDUT
Raw data display, protocol translation display,error display, bit shift display possible
Parity, Fleming, Break, BCC, Abort, Shortframe, Non-ACK (I2C)
Resolution 1 msec., 10 msec., 100 msec., andOFF can be specified up to 999.9 sec
Minimum 10 msec. (day:minute:second.10msec.), 6 time units and OFF can be specified
Record RTS, CTS, DTR, DSR, DCD, RI,
TRGIN (external trigger input) signals together
with send/receive data2, Maximum count 999999(Resolution 1msec., 10msec., 100msec.)
2, Maximum count 999999
Maximum count for SD/RD 4294967295
4 sets of trigger conditions and actions can bespecified
External trigger (1 input, 2 outputs)
Communication error, communication data
string of up to 8 characters (don't care and bit
mask can be specified), idle time longer thanspecified time, matching timer/counter value,
logical state of control line and external input
Stop measurement/test (offset number can bespecified), enable trigger condition, timer
control, counter control, buzzer sound, filesave, specified character string transmission,pulse output to external terminal
Automatic setting of measurement conditions
such as speed (up to 460.8Kbps) and data
frame *3
Specified time RUN/STOP(Hourly/Daily/Monthly repeatable),Power-on RUN setting possible
Automatically save a specified number ofcommunication log files in external storage
with a specified file size.
Ring record, automatic stop at maximum size
can be specified
Measures and displays the interval of change
in the interface signal line.
(current/min/max/average, resolution: 0. 1ms)

Voltage measurement
function

RS-232C input range ±18V
TTL input range -1V to 6V, resolution 0.1V

Logic analyzer function

1KHz to 20MHz sampling, zoom display,
cursor time measurement, signal order swap!

SAFETYWARNING

Read the instruction manual provided with the product before use
and use the product as explained in that manual. Using the product inways not guaranteed in the manual, connecting it to systems outsideof the specified ranges and remodeling can all cause trouble anddamage. LINEEYE CO., LTD. will assume no responsibility
whatsoever for trouble or damage arising because of unauthorizedways of use.

Simulation function

Data can be sent to the test target while
checking the received data

Test data table

Transmit test data 160 types (16K data) can beregistered, BCC/LRC automatic calculation,parity error data can be registered

Test mode

4 modes (MANUAL, FLOW, ECHO,POLLING) selectable
Measures line quality such as bit error rate
and block error rate according to ITU-T
G.821 *4
Save, load and delete measurement data
and measurement conditions
PC link software (light version *5) included
Measurement data display, recording,text/CSV conversion
4.3 inch TFT color LCD (480x272dot)
With touch panel
Line state display, power/charge display
USB2.0 (Standard A /micro-B each 1),
SDHC card slot, Wi-Fi 802.11 b/g/n *6
SDHC card and USB memory *7 Up to32GB
USB bus power, built-in lithium-ion battery
Battery operating time: About 7 hours *8
Operating temperature: 0-40°C
Storage temperature: -10-50°C
190 (W) x 153 (D) x 38 (H) mm, About 550g
Monitor cable for DSUB 25-pin (LE-25M1),
Monitor cable for DSUB 9-pin (LE-009M2),
DSUB25pin-9pin conversion adapter, 5wires TTL prove cable (LE-5LS), micro USB
cable, USB charger (LE-P2USB), carryingbag (LEB-01), Utility CD, quick start guide,and warranty

Bit error rate test
File management
Remote control
LCD display
LED display
External interface
External storage

Power supply
Temperature range
Dimension, mass
Accessories*1: Raw data display only. *2: Communication data, idle time, time stamp, and line status consume
4-8 bytes of memory for each capture. *3: Automatic setting is available only for ASYNC, SYNC/BSC,
HDLC/SDLC. If the communication data amount is small or there are many errors, the automatic
setting cannot be performed correctly. *4: Only available in ASYNC and SYNC modes. *5: Somefunctions of the product version PC link software LE-PC300R are limited. *6: It can be shipped with
the Wi-Fi function disabled due to the radio law of each country. Please contact our sales department
for details. *7: Operation of SDHC cards other than LINEEYE option is not supported. *8: According toour measurement conditions.

OptionItem name
Expansion kit for CAN FD/CAN/CXPI
Expansion kit for CAN FD/CAN/LIN
Expansion board for RS-530
Expansion kit for current loop comm.
Expansion kit for TTL/I2C/SPI
5 wires TTL prove cable
X.21 monitor cable
RS-449 monitor cable
V.35 monitor cable
RS-530 cable
USB charger
PC link software
PC link software (For OP-SB7XC/7XL)
3.81mm pitch terminal block (5 POS.)

16GB SDHC card
32GB SDHC card

Model number
OP-SB7XC
OP-SB7XL
OP-SB10N
OP-SB1C
OP-SB5GL
LE-5LS
LE-25Y15
LE-25Y37
LE-25M34
LE-25S530
LE-P2USB
LE-PC300R
LE-PC7XCL
LA-5ETB45
SD-16GX
SD-32GX

Remarks*1*2*2*2*2*1*3*3*4*1: Same as the attached one.
*2: OP-SB10N required. *3: Full edition with nofunctional restrictions *4: Same as that set on the analyzer.
•All brand names and product names mentioned in this leaflet are trademarks or registeredtrademarks of their respective companies. •Specifications and designs of products listed inthis leaflet are as of July 2020, and are subject to change without notice for improvement.
•Colors of actual products may differ slightly from that listed due to printing condition. •Thisleaflet may not be reprinted or duplicated, in part or in whole.
©2020 by LINEEYE CO., LTD.。

LINEEYE CO., LTD.
Head Office/Sales Office
Marufuku Bldg 4F, 39-1 Karahashi Nishihiragaki-cho, Minami-ku, Kyoto, 601-8468
PHONE: 81-75-693-0161 FAX:81-75-693-0163

URL： https://lineeye.co.jp

E-mail： info@lineeye.co.jp
Printed in Japan

L-20071E/LE