®

Hispec HI2200/2300
High Temperature Transmitter• High operating andd ambientammbi
temperatures of up to 20200ºC• Pressure ranges to 1,51,500 bar• High accuracyoption• Temperature compensatedccompatibility for a• Good chemicalchemrange of applications• A range
ran of electrical and processconnections available

Vers. 20/1/Eng

Hispec ® HI2200/2300 High Temperature Transmitter
Technical Data
Description
HISPEC - HI2200/2300 series of high temperaturepressure transducers with state-of-the-art Siliconon-Sapphire sensor technology offer levels ofaccuracy and performance previously unobtainableor prohibitively expensive. It is capable of operating
at a constant 200 °C both media and ambient.
The advanced sensor design consists of a piezoresistive
silicon strain gauge circuit, which is epitaxially grown onto thesurface of a sapphire diaphragm to form a single crystalline
structure. The sapphire sensor element is then molecularly
bonded to a titanium alloy sub-diaphragm. This enables thesensor to endure higher over- pressures and provides superb
corrosion resistance.The sensor exhibits virtually no hysteresisand excellent long-term stability. With outstanding insulation
properties, the sapphire substrate protects the strain gauge
circuit and allows the sensor to operate over a very wide
temperature range without loss of performance.

With pressure ranges from 0-1 bar to 0- 1,500 bar, the hightemperature HISPEC transducer is available in two options;model HI2200 offers a non-compensated and un-rationalised
signal output of between 10 mV/V and 20 mV/V, whilst model
HI2300 is fully temperature compensated with a rationalised 10mV/V signal output. All models are available with either PTFEcable outlet or military bayonet style plug to MIL-C-26482,
both of which are rated for use at 200 °C. This means that
not only does the transducer perform effectively at high media
temperatures but uniquely can be used in environments where
there are elevated ambient temperatures of up to 200 °C. For
instance this device may be mounted inside an oven or thermalchamber.

Dimensions(in mm)15
ELECTRICAL CONNECTION
MIL-C-26482
Pin.
ABCDEF

Designation+supply+output-output-supply
N/C
N/C

Ø25.4
~68

ELECTRICAL CONNECTION
CABLE OUTLETWIRE COLOUR
REDGREEN
YELLOWBLUE

Designation+supply+output-output-supply-output to case

19 A/FHEX12

Hispec ® HI2200/2300 High Temperature Transmitter
Technical Data
TypeSensor Technology:

Output Signal:

Supply Voltage:

HI2200/HI2210

HI2300/HI2310

Silicon-on-Sapphire (SoS)
10 to 20 mV/V (Un-rationalised and uncompensated)

10 mV/V (Rationalised and compensated)

10 VDC (5 – 15V)

Pressure Reference:

Gauge

Standard Pressure Ranges (bar):0 – 1 bar Vac; 0 – 1 bar; 0 – 10 bar; 0 – 25 bar; 0 – 100 bar; 0 – 250 bar; 0 – 700 bar; 0 – 1,500 bar (otherranges available)

Standard Pressure Ranges (psi):

0-30 in Hg; 0-15 psi; 0-150 psi; 0-300 psi; 0-1,500 psi; 0-3,000 psi; 0-10,000 psi; 0-20,000 psi (otherranges available)

Overpressure Safety:

2x for ranges 1 bar Vac to 600 bar; 1.5x for 1,000 bar range; 1.1x for 1,500 bar range

Load Driving Capability:

Accuracy NLHR:

Zero Offset and Span Tolerance:n/a≤ ±0.1 % of span BFSL

Zero offset: ±1 mV/V; Span Tolerance: 10-20 mV/V

Zero offset: ±1 mV; Span Tolerance: ±1% FS

Operating Ambient Temperature:-40 °C to +200 °C (-40 °F to +392 °F)

Operating Media Temperature:-50 °C to +200 °C (-58 °F to +392 °F)

Storage Temperature:

Temperature Effects:

Electromagnetic Compatibility:+5 °C to +40 °C (+41 °F to +104°F) Recommended Best Practice

Typical thermal zero and span coefficientscompensated ±0.05 %FS/ °C±2.0 %FS Total error band -20 °C to +150 °C, typicalthermal zero and span coefficients ±0.005 %FS/ °C

Emissions: EN61000-6-4; Immunity: EN61000-6-2; Certification: CE Marked

Insulation Resistance:> 100 MΩ @ 50 VDC

Response time 10-90 %:1 mS

Wetted Parts:

Titanium alloy

Pressure Media:

All fluids compatible with Titanium alloy

Pressure Connection:

1/4” BSP male (G1/4) or 1/4” NPT male (others options available)

Electrical Connection:

HI2x00: PTFE insulated flying lead, conductor size 7/0.1 mm. HI2x10: MIL-C-26482 6 pin bayonet
connector (Accessory not included: mating connector type MS3116F10-6S).

Net. Weight (Kg):

0.1 Kg

Hispec ® HI2200/2300 High Temperature Transmitter
Order Matrix

Output10-20 mV/V
10 mV/V

Electrical Connection

Wires

Type1m PTFE insulated flying lead

4HI2200

MIL-C-26482 6 pin bayonet

4HI22101m PTFE insulated flying lead

4HI2300

MIL-C-26482 6 pin bayonet

4

HI2310

Options

PressureRange

Process Connection

Options
No special options-Pressure Range in bar
0-1 barVac

V001

0-1 bar0001

0-10 bar0010

0-25 bar0025

0-100 bar

0100

0-250 bar0250

0-700 bar

0700

0-1,500 bar

1500

Process Connection
1/4” BSP male (G1/4)

AB

1/4” NPT maleAMOrder Number Example

HI2200-0700AB

For options not listed please contact the sales teamDISCLAIMER : ESI Technology Ltd operates a policy of continuous product development. We reserve the right to change specification without
prior notice.All products manufactured by ESI Technology Ltd are calibrated using precision calibration equipment, traceable to national
measurement standards.

t. +44(0)1978 262 255
e. sales@esi-tec.com

www.esi-tec.com