MULTI PROTOCOL ANALYZER

10 Gigabit Ethernet Measurement

LE-8600X / LE-8600XR
1 Gigabit Ethernet / PoE++ Measurement

LE-8500X / LE-8500XRGigabit Ethernet supported, stand-alone LAN analyzer
With easy-to-use touch panel operation

MULTI PROTOCOL ANALYZER

NEW

LE-8600X / LE-8600XR
234(W) x 186(D) x 44(H)mm, about 990g

10Gbit supported

MULTI PROTOCOL ANALYZER

NEW

LE-8500X / LE-8500XR
234(W) x 186(D) x 44(H)mm, about 990g

1Gbit supportedMULTI PROTOCOL ANALYZER
Supports various measurement interfaces of Gbit LANThe LE-8600X/LE-8600XR have two measurement ports that support
SFP/SFP+ modules such as 10GBASE-T and 10GBASE-SR. As astand-alone unit, it supports high-speed LAN measurement tests thatuse single-mode or multi-mode of optical fiber cables.

LE-8600X / LE-8600XR

SFP module port

External input/output

GPS antenna connector

PPS signal connector[ 10GBASE-LR SFP+ module ]LE-8500X / LE-8500XR

LAN measurement port

External input/output

GPS antenna connector

PPS signal connector[ 10GBASE-T SFP+ module ]The LE-8500X/LE-8500XR have four measurement LAN ports capableof 1GbE measurement and can simultaneously measure two channels of
Ethernet packets flowing on a LAN cable that is branched and connected
between ports A and B and between ports C and D. As it also supports
simultaneous measurement of Ethernet packets and PoE power
between ports A and B, you can measure and record packets of 2
Ethernet channels (up to 1GbE) and 1 PoE channel at the same time.

Stand-alone measurement of high-speed Ethernet from 10 Gbit to 1 Gbit
LE-8600X/LE-8600XR can measure and record 10Gbit / 1Gbit
Ethernet, and LE-8500X / LE-8500XR can measure and record
1Gbit Ethernet communication independently without using a PC.
The packet measurement data can be monitored in real time anddisplayed on the 7-inch wide color LCD, and can be saved into a
500GB internal SSD (LE-8600X/LE-8600XR) or an external storage
such as a USB memory or SSD connected via USB3.0. In addition,a statistical analysis function that can check the tendency of[ Translation display ][ Statistical analysis graph display ]communication traffic for each time zone, and a delay time measurement function (LE-8500XR/LE-8500X) that canmeasure the reception time difference of the received packets of each port, are also available. It can be widely usedfrom network systems to development testing and maintenance of embedded communication equipment.

Frame counter type:
Total number of received frames, number of normal frames,
number of broadcasts, number of multicasts, number of pause
frames, number of each frame of 0-63 byte length / 64 byte
length / 65-127 byte length / 128-255 byte length /256-511 byte length / 512-1023 byte length / 1024-1518
byte length / 1519 bytes length, number of FCS error frames,
number of FCS error frames less than 64 byte length

Pass/fail judgment and long-time powerrecording for PoE/PoE+/PoE++
In addition to PoE and PoE+, it can measure up to 4.5 hours recording of power consumption, voltage, and current, andsupports appropriate judgment, graph display, and dump display of PoE++ (IEEE802.3bt) - a new standard that supports
high-power supply by 4 pairs of twisted paired cables. This unit is ideal for developing and analyzing the latest equipmentthat uses PoE++. Furthermore, for between ports A-B, it supports simultaneous measurement of Ethernet packets and PoEpower. It is useful when checking the transition of power changes of PoE devices such as an Web camera, in response to[ PoE measurement graph display ]LAN packet control commands. It can be used in a wide range of scenes where PoE measurement is required.

Timestamp which can be time-synchronized by GNSS (PPS) signal
The time synchronization function by GNSS/GPS realizes
more accurate timestamping compared to the timestamping

GPSGPSusing a general crystal oscillator. If you measure for a long

GPS antenna

GPS antenna

period of time at two points where communication failure

PPS signal

rarely occurs by using two analyzers at the same time, you
can compare and verify each measurement data from theanalyzers based on the timestamps.

LE-8500X[ GPS antenna connection image ]LE-8600X

LE-8500X[ Simultaneous measurement by using PPS signal ]Outputs test packets at the line-rate
The LE-8500X/LE-8500XR can output Ethernet packets of arbitrary test data from ports C or D, or both ports at the same time, and theLE-8600 can output it from SFP+ ports A/B. Test data can be set to 16 transmission data tables, and the transfer rate can be adjusted byspecifying the IFG (Inter Frame Gap) for each table. It can be used for network load testing.

LE-8600X / LE-8600XR
1 Gigabit Ethernet / PoE++ Measurement LE-8500X / LE-8500XR
10 Gigabit Ethernet Measurement

Automatically records communication logs to alarge-capacity external storage via USB 3.0
LAN port(for PC link)

While recording the measurement data in the capture memory, this unit canautomatically save it continuously for a long time into external storage such

USB3.0device port

as USB memory or SSD. As the communication log file can be divided into
multiple files of the specified size and saved, the communication log before

USB3.0host port

and after the failure can be narrowed down from the time stamp of the file.

SD cardslot［ External storage connection ］

Newly designed battery-powered lightweight
housing - ideal for on-site measurement
Uses a new battery-powered housing that weighs less than 1 kg, B5 size. The measurement datacan be displayed directly on the 7-inch wide color LCD, and all measurement tests can beperformed without using a PC. Not only can you execute setting operations and smooth data scroll
display by touching the screen or swiping, but also you can operate with key switches as with theconventional model so that you can operate even at the site where you need to wear gloves. This
product is ideal for on-site measurement at customer sites where the carry-on of a PC is restricted.

Save data as PcapNG files that can be analyzed by Wireshark
With the file management function, you can save, read, and delete measurement data,setting data, and a screenshot into external storage such as a USB memory or SDHC card.
It can also be saved as a PcapNG file that can be read by the analysis software, Wireshark.

Transfer measurement data to PC via Ethernet, USB, or Wi-Fi (XR only) connection
High-speed data transfer via G-bit Ethernet or USB 3.0 is supported for remote connection and data transfer to a PC. The Wi-Fi compatiblemodel LE-8600XR/LE-8500XR can connect to a PC via Wi-Fi.

Supports high-speed serial communication of upto 20 Mbps by using the expansion option
Supports RS-232C, RS-530, RS-422/485, and TTL serial communication measurement bychanging the standard measurement board to the optional SB-R2TS1 measurement board.
Communication data can be recorded without data loss even in the monitoring of a 20Mbps
high-speed SPI communication or a 10Mbps CC-Link test. It also supports legacy interfaces
such as V.35 and X.20/21 by using the RS-530 measurement port.

[ Expansion kit for serial communications “SB-R2TS1” ][ Options ]Model

Description

Applicable Model

LE-1G-BT-45

1000BASE-T SFP module

LE-8600X, LE-8600XR

LE-1G-LX-31

1000BASE-LX/LH SFP module

LE-8600X, LE-8600XR

LE-1G-SX-85
LE-10G-BT-45

LE-10G-SR-85
LE-10G-LR-31
SB-GE2

SB-R2TS1
LE-25Y15
LE-25Y37

LE-25M34

LE-25S530

EB-SL-AA170

LE-SMA-MM-2
P-26LW2

1000BASE-SX SFP module
10GBASE-T SFP+ module

10GBASE-SR SFP+ module
10GBASE-LR SFP+ module

Expansion kit for Gbit LAN communication

Expansion kit for RS-232C/530/422/485/TTL communication
X.21 monitor cable

RS-449 monitor cable
V.35 monitor cable
RS-530 cable

GPS active antenna

Coaxial cable for PPS synchronization
Lithium ion battery packLE-8600X, LE-8600XR
LE-8600X, LE-8600XR
LE-8600X, LE-8600XR
LE-8600X, LE-8600XR

LE-8600X, LE-8600XR, LE-8500X-RT, LE-8500XR-RT
LE-8600X, LE-8600XR, LE-8500X, LE-8500XR
SB-R2TS1, LE-8500X-RT, LE-8500XR-RT* Standard measurement board of LE-8500X/LE-8500XR* Standard measurement board of LE-8500X-RT/LE-8500XR-RT

SB-R2TS1, LE-8500X-RT, LE-8500XR-RT
SB-R2TS1, LE-8500X-RT, LE-8500XR-RT
SB-R2TS1, LE-8500X-RT, LE-8500XR-RT

LE-8600X, LE-8600XR, LE-8500X,LE-8500XR, LE-8500X-RT, LE-8500XR-RT
LE-8600X, LE-8600XR, LE-8500X, LE-8500XR

LE-8600X, LE-8600XR, LE-8500X, LE-8500XR, LE-8500X-RT, LE-8500XR-RT (Spare or replacement parts)

Specification
Model

LE-8600X / LE-8600XR

LE-8500X / LE-8500XR

SFP+ 2 ports. A and B（SFP/SFP+ modules to be connected）
10GBASE-T/SR/LR/ER、1000BASE-T/SX/LX/EX etc.

Interface

RJ-45 connectors. Port A, B, C, D：1000BASE-T/100BASE-TX/10BASE-T
Port A – B is failsafe tap*1

Capacity: 1Gbyte（96byte to 10560byte for 1 frame）2 divided use, accidental-erasure protection, and selection of ring buffer and fixed size buffer are available.

Capture Memory

Measurement conditions and part of the latest data (about 16 Mbytes) can be saved automatically*2

Backup Memory
Internal Storage

500Gbyte SSD−Online Monitor Function

Recording and real-time display of Ethernet frames flowing between ports A-B

Timestamps

The reception time is added as timestamp data for each received frame. Resolution: 8n seconds / 1u seconds / 10u seconds can be selected
Time synchronization by GNSS PPS signal or external PPS signal is available*4

Frame Size

Simultaneous recording and real-time display of 2 channels of Ethernet frames flowing between ports A-B and C-D*3

60 byte to 9Kbyte

Scroll display, 2-split comparison display, jump operation to specified screen, mark jump operation

Data Display/Control

Supported protocols: IPv4, ARP, ICMP, TCP, UDP, DHCP, EtherCATTranslation

Measurement data can be converted and saved to a PcapNG format file*5

PcapNG Conversion

Can monitor only the specific frames that match one or two specified conditions.

Filter Function

When a specific frame that matches the specified conditions is received or when the external TTL signal changes, it can automatically stop monitoring, count how mant times the conditions are matched, or output the external TTL signal.

Trigger Function

Layer 2 type (IPv4, ARP, NetBIOS, IPv6, EtherCAT, type number specified), Layer 2 source address, Layer 2 destination address, IPv4 protocol (ICMP, IGMP, TCP, UDP,protocol number specified), source port number / address, destination port number / address, subnet mask

Filter/Trigger Condition
Search Function
Delay Time Measurement Function

Statistical Analysis Function

PoE Measurement Function

From the measured data, it can search, cue and display, or count only the specific frames that match the specified conditions
Search conditions: Error frame, specified frame size, measurement port, and specific frame that can be specified in the same way as the trigger condition

Can measure the time difference between the two specified reception timings among PortA-B Tx, Port A-B Rx, Port C-D Tx, and Port C-D Rx in u seconds, and calculate the current
value, maximum value, minimum value, average value, and display them as delay time.

−Statistics of two frame counter values can be collected and displayed in a graph at specified intervals (1 to 240 minutes), and all frame counter values can be displayed in real time.

Measurement, recording, appropriateness judgment, graph display, and dump display for the power consumption,
voltage, current, power supply type (Alternative A / B, polarity) of PoE/PoE+/PoE++ (IEEE802.3af/at/bt) between
the devices connected to ports A and B. Simultaneous measurement with Ethernet frames is available.
Recording interval: 1 msec to 1 sec, maximum number of recordings: 16.77 million times, voltage: 0 to60 V (accuracy: ±1% F.S.), current: 0 to ±900mA (accuracy: ±1% F.S.)*6−Can output arbitrary packets at the wire rate from ports C and D (LE-8500X(R)) and SFP+ ports A and B (LE-8600X(R)).
Up to 16 types of packets set in the 16 transmission data tables (total 32k data) of each port can be transmitted sequentially from each port for a specified number of times or continuously.
Frame gap (interval until the next packet is transmitted) can be set for each transmission data table.

PG Function

Transmits a PING command from port C (LE-8500X (R)), SFP+ port A (LE-8600X (R)) to display the number of responses and response time (current value, maximum value, minimum value, average value).

Ping Function−Port Blink Function

Auto Save Function

Auto Save Mode
File Size

Max File Number

Time-specified Auto RUN/STOP Function
Power-on Auto RUN Function
File Management Function

File Type

File Control

DisplayLine Status LED

USB Device PortUSB Host PortSD Card SlotExternal I/O terminal

GPS Antenna Connector
PPS Signal Connector
Wi-Fi Connection*8

Blinks the link LED on the port of the hub connected to port C

The contents of the capture memory (monitored data) can be automatically saved as a communication log file in the storage of the built-in SSD (LE-8600X(R)), USB memory / SDHC, etc.*7
Re-recording mode, append mode, Max stop (records up to specified capacity and stop) mode can be selected
Capture memory size, 4M bytes, 16M bytes, 64M bytes, 256M bytes
2048

Measurement operation can be started and stopped at a specified time and at a specified repetition cycle (selectable from monthly, daily, and hourly).
Measurement operation can be started automatically after the power is turned on.

Measurement data and measurement conditions can be saved in an external storage such as USB memory / SDHC card in a format that can be read by a PC.
Measurement data (.DT), all measurement conditions (.SU), auto save data (#nnnnnnn.DT), screenshot (.PNG), PcapNG format file (.PCAPNG)
Normal file display, file display and sorting by specified type, save, load, delete, delete all files
7 inch TFT color LCD with capacitive touch panel

11 LEDs always display the connection status of measurement ports

Type-C connector, SuperSpeed transfer supported. For PC connection

Standard A connector, SuperSpeed transfer supported. For external storage (USB memory / SSD)
For standard size SD / SDHC memory cards, compliant with SD Association standard
4-pin connector for TTL level trigger input/output signals

SMA (female) connector for active GPS antenna connection
SMA (female) connector for TTL level PPS I/O signals

IEEE802.11b/g/n frequency range: 2412MHz to 2484MHz
Transmission power 802.11b: +18.5dBm, 802.11g: +18.0dBm, 802.11n: +17.0dBm
Attached AC adapter Input: AC100-240V 50/60Hz Output: DC9V 2A

PowerBattery
Ambient Temperatures
Ambient Humidity
Standard

Size and Weight
Accessories

Lithium-ion rechargeable battery (model number: P-26LW2) Battery drive time: 2 hours*9Charging time: Approximately 3.5 hours when using the AC adapter, approximately 4.5 hours when using the USB Type-C cable (when using a Type-C charger that can supply 2A or more)
In operation : 0 to 40 degree Celsius, In storage : -20 to 50 degree Celsius
20 to 85%RH (No condensation)
CE (Class A)

234(W) x 186(D) x 44(H)mm, about 990g

LAN cable (LE-8500X(R): 2 pcs, LE-8600X(R): 1 pcs), USB cable (standard A-Type-C), external signal input/output cable (LE-4TG), AC adapter (6A-181WP09), carryingbag (LEB-01), utility CD, quick start guide, warranty card

* 1 When the power of this unit is turned off, ports A and B are connected through in this unit, and the link between the devices connected between A and B is maintained. The link may be temporarily disconnected during the startup period andshutdown period of this unit. * 2 Uses a part of eMMC for saving measurement firmware * 3 As data can be recorded in the capture memory at a maximum of approximately 3Gbps, it is possible to record without packet loss when measuring
1000BASE-T on one channel. When measuring two channels of 1000BASE-T high traffic lines at the same time, some packet loss may occur. * 4 A GPS antenna (sold separately) is required for GNSS synchronization. External PPS synchronization
requires another unit to which a GPS antenna is connected and an SMA coaxial cable (sold separately). * 5 LINEEYE will provide a PC software that can convert the measurement data file of this unit to a PcapNG format file on a PC. * 6 Cannot beused to measure PoE lines in which a current exceeding 720mA flows continuously for 3 seconds or longer. * 7 Not all frames may be recorded in the storage depending on the high traffic line or the performance of the external storage. Themaximum recording speed is about 1Gbps (according to our measurement test using USB3.0 connected SSD) * 8 LE-8500XR/LE-8600XR only. For PC connection * 9 According to our test conditions assuming a normal usage situation

Sisterproduct

SAFETYWARNING

Read the instruction manual provided with the product before use and use theproduct as explained in that manual. Using the product in ways not guaranteed in
the manual, connecting it to systems outside of the specified ranges and remodelingcan all cause trouble and damage. LINEEYE CO. LTD. will assume no responsibility
whatsoever for trouble or damage arising because of unauthorized ways of use.

LE-8500X-RT / LE-8500XR-RT

Supports RS-232C, RS-530, RS-422 / 485, TTL Maximum measurement speed 20Mbps• All brand names and product names mentioned in this catalog are trademarks or registered trademarksof their respective companies.
• Specifications and designs of products listed in this catalog are as of April 2022, and are subject tochange without notice for improvement.
• Colors of actual products may differ slightly from that listed due to printing condition.
• This catalog may not be reprinted or duplicated, in part or in whole.
©2022 by LINEEYE CO., LTD.

To be released in October 2022

RS-232C/RS-530
Measurement por

TTL measurement port /External trigger terminal

RS-422/485
Measurement port

GPS antenna connector

LINEEYE CO., LTD.

Head Office/Sales Office
Marufuku Bldg 4F, 39-1 Karahashi Nishihiragaki-cho, Minami-ku, Kyoto, 601-8468
PHONE: 81-75-693-0161 FAX:81-75-693-0163●URL https://www.lineeye.com●E-mail : info@lineeye.co.jp* LINEEYE CO. LTD. is a venture company founded by electronicequipment development members of the former Sekisui Chemical Co.,
Ltd. with investment from the Sekisui Venture Fund. The electronicequipment business of Sekisui Electronic Co. Ltd. was transferred toLINEEYE CO. LTD. in October 2000.

Printed in Japan

L-22402E/LE G