A simple and efficient MRI injector
with reliable performances

The OptiStar® Elite is a dual-head
contrast delivery system,
meeting the needs of healthcare
professionals in the MRI room.

Thanks to its ultrasonic motor,
the OptiStar® Elite is nonmagnetic
and is compatible withup to 3 Tesla scanners.

Pressure Limit:
200PSI

Up to 3 Tesla
Compatibility

Inject Delay:
0 to 300seconds

EFFICIENT & RELIABLE

SIMPLE TO USE

PFS compatible

Auto retract rams
for quick turnaround

Fully Prepare syringes
outside the MRI room

Powerhead keys:
Control at your fingertips& full access from inside
the MRI room

Battery-free: eliminate
associated time & cost

THE 3 CONFIGURATIONS:For your preferences and needs:

SIMPLE, EFFICIENT, RELIABLE…
the OptiStar® Elite will allow youto perform MRI exams in an optimized way
with specifically designed features.

MobilePedestal
Ceiling mountDesigned to be simple and easy to use, the console allows you
to manage the injection with a total remote control

START & STOPAN INJECTIONREMOTELYUP TO 6
PHASESIN PROTOCOLPRESSURE
CONTROL

Handswitch for maximum flexibility
CAN STOREUP TO 40
PROTOCOLS
IN MEMORYINJECTION FEATURES:

Patency Check®:
test the patient’s veinprior the injection
Timing Bolus®:
test to determine
the optimal scan delay
Drip mode:
maintaining
the venous accessColour touchscreen, uncluttered display
Console Enable(minimize delays & maximize control)

Clear indicators & status line

GUERBET SERVICE + SUPPORTOur digital solution to collect, archive, analyseand share patient injection data and injector activity
TRACEABILITY
Facilitate traceability
and saveadministration time

The 360° Technical support you can rely on

THE 3 SERVICE PACKAGES:PLATINUM

INFORMEDDECISIONSA complete viewon all relevant
patient information
including injection
historyGOLDSILVER

ANALYTICS
Improve your injection
practices and optimize
contrast mediautilization

For more information ask your Guerbet Representative

UNIK is a flexible solution, tailored to your specific needs
and challenges, designed to add value at each step of the patient
healthcare journey, interconnecting our comprehensive portfolio
of products, solutions and services in diagnostic imaging.

SYRINGE KITTUBINGS

MULTIPAK-OPTISTAR

Single check valve

Dual MRI 60ml Syringe withsingle check valve Y-tubingand filling spikes

Y-tubing 150cm

Y-tubing 230cm

Pressure Limit:
200 PSI

Pressure Limit:
350 PSI

Single Use
Pressure limit:
300 PSISINGLE SYRINGE KITCONTRAST MEDIA ADAPTERS

Single MRI 60ml Syringewith filling spike

10,15,20 ml

Single Use
Pressure limit:
300 PSI

DotaClip

StarClip

For Contrast Agent
Glass Syringe

For Contrast Agent
Plastic Syringe

| Medical Device

OptiStar® EliteINJECTOR PEDESTAL
Injector in pedestal version
Reference
M5200

214727

QTY/Box1

M5600** 234842**

OptiStar® Elite
CEILING-MOUNT

Injector in ceiling mount options:- long column- ceiling track mount

Reference
M5400

230943

QTY/Box

1M5700** 234844**

OptiStar® Elite
MOBILE-MOUNT

Injector in floor mount for mobile/ ambulatory unit*

Reference
M5300

225877

QTY/BoxInjection Management Solution
Ask your Guerbet Representative1

M5800** 234866**

TECHNICAL SPECIFICATIONSADAPTER STARCLIP

ADAPTER DOTACLIP

MULTIPAK-OPTISTAR

Reference

Reference

Adapters for prefilled syringes
10/15/20mL*

Adapters for prefilled syringes
10/15/20mL*

814735S 236900

802930S 230945

Reference

QTY/Box1

SYRINGE

1 syringe + 1 spikeReference

801801B 234159

QTY/Box

1Y-TUBING

QTY/Box50Single check valve Y-tubing 230cm
Reference

801107B 234157

QTY/Box100

2 syringe + Y-tubing 1 valve+ 2 spikes
801800B 234158

QTY/Box50

Y-TUBING

Single check valve Y-tubing 150cm
Reference

801106 223374*Presentations may differ from country to country. For complete information please refer to the local Summary of Product Characteristics**Injector version with new power supply

QTY/Box100

Flow RateA Sidew 60 mL syringes: 0.1–10.0 mL/secw 10, 15, 20, 30 mL syringes: 0.1–8.0 mL/sec
B Sidew 60 mL single-use disposable syringes: 0.1–8.0 mL/sec
w 125 mL prefilled saline syringes*: 0.1–8.0 mL/sec
VolumeA Side
w 10, 15, 20 mL syringes: 1 mL to 20 mL
w 60 mL syringes: 1 mL to installed syringe size
B Sidew 60 mL syringes: 1 mL to installed syringe size
w 125 mL prefilled saline syringes*: 1 mL to installed syringe size*
Number of Phases
w 6 including Timing Bolus® featurePhase Delayw 0-600 seconds + pause

Number of protocols
w 40Inject History
w Last 48 completed injections
Pressure Limit
w 60 mL empty syringes: 150 psi
w 10, 15, 20 mL prefilled syringes: 200 psi*
w 125 mL prefilled saline syringes: 150 psi*
Syringe Sizes
w 60 mL empty syringes
w 10, 15, 20 mL prefilled syringes*
w 125 mL prefilled saline syringes*
Scan Delayw 0-60 seconds
Inject Delayw 0-300 seconds
MR Compatibility
w Up to 1.5 & 3 Tesla

Optistar Elite Contrast Delivery System is a medical device intended for use by qualified healthcare professionals. It is
intended to inject MR contrast media and flushing solutions into a patient’s vascular system to obtain diagnostic images
when used in conjunction with Magnetic Resonance (MR) Imaging equipment. For complete information about precautions
and optimal usage conditions, we recommend consulting the instruction for use supplied with the device or by your local
Guerbet representative(s). For use only in countries with applicable health authority registrations.
Class IIb/CE. TÜV SÜD 0123. Manufacturer: Liebel-Flarsheim Company LLC EC Rep: Guerbet

P20000190 - June 2020 - SMM Srl - The graphical layout and all its parts are property of SMM and may not be reproduced or copied without written consent of SMM

OptiStar® Elite Injector consumables are medical devices intended for use by qualified healthcare professionals and designed to be used in the LF contrast injection systems. They contain the contrast and provide the delivery path for the contrast
to the patient. The injection systems are designed to inject contrast media into a patient’s vascular system to obtain diagnostic images when used with computed tomography (CT), MR, Angiographic or x-ray equipment. For complete information
about precautions and optimal usage conditions, we recommend consulting the instruction for use supplied with the device
or by your local Guerbet representative(s). For use only in countries with applicable health authority registrations.
Class IIa/CE
Manufacturer: Liebel-Flarsheim Company LLC. EC Rep: Guerbet. - TÜV SÜD 0123
Manufacturer: Shenzhen Boon Medical Supply Co., Ltd. EC Rep: Wellkang Ltd. - TÜV RHEINLAND 0197
Contrast&Care® is a medical device intended for use by healthcare professionals only. It allows imaging centers to collect,archive, view and share patients’ injection data, including data concerning contrast products, adverse events, injector activity,
data on the estimated eGFR and other pre-exam alerts, such as previously reported allergies. Contrast&Care® also provides
options to review protocols, create protocol libraries, and visualize analytics data and trends relating to injection activity andcontrast product usage. For complete information about precautions and optimal usage conditions, we recommend consulting the instruction for use supplied with the device or by your local Guerbet representative(s). For use only in countries withapplicable health authority registrations.
Class I/CE.
Manufacturer: MEDEXFrance Distributor: Guerbet, (Roissy CdG cedex, France) tel: +33 (0)1 45 91 50 00 www.guerbet.com - www.contrastncare.com
OptiStar® Elite, Contrast&Care®, Patency Check®, Timing Bolus® & OptiProtect™ are trademarks of Guerbet.
Issued patents. © Copyright 2015 Guerbet - All rights reserved.

GUERBET - BP 57400
95943 Roissy Charles de Gaulle Cedex - FranceTel.: +33 (0) 1 45 91 50 00 - Fax: +33 (0) 1 45 91 51 99

www.guerbet.com