For Infrared thermometers

TMCXseries
Parameter setting unit
PWCX
Software for parameter setting PWSX/PWUX
Display

Easy to see, easy to use
Maximize the performance ofinfrared thermometers !

Challenge, Create, Customize

Use them to display and conﬁgure our infrared thermometers.
5 types of displays and setting units
Easy to adjust for the emissivity setting with the teaching function.
Easy to connect to a thermometer.
Displays and Setting Units

TMCXseries

TMCX-H/N
TMCX-H● Easy to look at the display with large LED charactrs.

(LED is brighter than the previous model)● RS485 equipped.

The temperature data can be sent digitally to external devices.

● Sighting light can be controlled by external devices via RS485.
● 100 to 240 V AC

TMCX-N● Reduce cost compared to the previous model.
● Analog output equipped.
● CE Mark Compliant

TMCX-TDE● Easy operation with the color touch screen.

● The measured temperatures can be displayed on a graph view.
● The measured data can be stored in a microSD card.
● Wall mount type.

● CE Mark Compliant

Parameter setting unit

PWCX
● Capability to conﬁgure multiple thermometers to provide cost savings.
● Operation time is doubled by increasing batteries (8 x AA size).
● Reduced cost compared to the previous model.

2/4Supporting thermometers for quick response of 0.1ms
Licensed software for parameter setting.
Cost-eﬀective
Parameter setting software

PWSX● Easy to retrieve data

● Data can be stored in CSV format in a PC.

● Automatic detection of port numbers and easy initialization.
● Supported OS : Windows 7/8/10

LicensedInstall

USB-RS232C Converter

PWUX● Digital signal of RS232C can be retrieved to a PC on USB.

USB

● Smaller size than the previous model.

PWUX
Thermometer

3/4

Speciﬁcations

e-CON: an 8 pin easily attachable connector.

Model

TMCX-HA/HD

TMCX-NDETMCX-TDE

PWCX

PWUX

Parameter setting
Dimensions○96mm□
HA：AC100〜240V
HD：DC13〜27V
e-CON with cable
○○
○
○
○
○

○48mm□○
98×48mm
DC22.8〜25.2V（DC24V±5%）
e-CON with cable○○48mm□○PC Screen

Batteries

USB○○
○

○

Supply voltage
Connection with a thermometer
Analog output1
Analog output2
Alarm output1
Alarm output2
Emissivity analogsettingRS485
Temperature chart
Data logging
Parameter logging

DC8〜27V
e-CON
○○○○

e-CON

○○○

○
○
○

Diagrams
Thermometer

Display

DCAnalog output

Alarm output
RS485

RS232C

Thermometer

Analog output

DCAC

Branch cable: TMBX-B01

DC

Analog output
PWCX

RS232C

Thermometer

RS232C

PWUX

DCPC
USB

PWSX

Connect only at measurement
Connect only at measurement● The contents described in this catalog are subject to change for

improvement without notice.

● If you have any questions, contact us at the address or links shown below.

http://www.japansensor.co.jp
info@japansensor.co.jp

4/4

TMCX-Cata-V201906P201906 1K

Speciﬁcations : Display / Parameter setting unit / USB-RS232C converter

TMCXseries

High-end

Connection with thermometer

Emissivity

Analog output1
Analog output2
Alarm outputs
Alarm output mode
Alarm output optional
Alarm output optional response time
RS485
Peak hold reset mode
Sample and holdSensor correction function
Display Options
Resolution
Temperature Units
Supply Voltage
Dimensions
WeightAcceptable panel thickness
Ambient temperature
Ambient humidity
Storage temperature

RoHSRoHS

TMCX-NDE *1
TMCX-HDRS232C(output swing ±5V) Maximum speed 5ms
ring tongue M3 terminale-CON
connecting with the TMBX-H (an accompanying cable)
guarantee range: 0.3 ~ 1.0
setting range: 0.050~1.000(setting resolution0.001)
correction function of reﬂection
analog setting(analog input 0 ~ 5V = emissivity 0 ~ 1.0)
response time: Max 2s + smoothing setting value
via a thermometer output.
Check the TMHX and the FTKX series' instruction manual for speciﬁcations.
conﬁgurable: 4 ~ 20mA, 0 ~ 20mA, 0 ~ 1V, mV/℃
scaling functionnone
response time: 250ms + averaging time (0~95%)
21hysteresis: 0 ~ 99℃, resolution1℃
opto-isolator: 24V DC, Max. 50mA
relay: 240V AC, Max. 2Anone
5ms (opto-isolator) 10ms(relay)
telecommunication: 4.8, 9.6, 19.2 kbps
none
check the TMCX telecommunications speciﬁcations for functional insulation and software
time: 0.01 ~ 10.00s(conﬁgurable) ,discharge: 0.01 ~ 10.00s, level0.20 ~ 1.00
External input: dry contact or open collectornone
Dry contact or open collector
none
span: 0.500 ~ 2.000, Zero: -50.0 ~ +50.0( ℃ or ℉ selectable)
Upper line: Temperature / Alarm H / Alarm L / Blank
Bottom line: Temperature unit / Emissivity / Alarm H / Alarm L / Blank
switchable 1℃ or 0.1℃ (above 1000℃ is to 1℃) Max. 3276℃℃ or ℉
AC100 ~ 240V 20W max.
DC8 ~ 27V 9.5W max.
DC13 ~ 27V 9.5W max.
±10％ 50／60Hz
96W×96H×69Dmm
48W×48H×66Dmm
230g
150g
1.0 ~ 3.5ｍｍ-10 ~ 50℃
≦90%RH (non-condensing)-20 ~ 65℃
TMCX-HA

ModelCommunications

Low-end

CE Mark Compliant (EMC EN61326-1: 2013, RoHS EN50581: 2012) *1: LVD (Low Voltage Directive) EN61010-1:2010
Speciﬁcations
Alarm output 1
Alarm output 2
spareOptional:TMCX-HA/HD-0-1
010

opto-isolator (standard)
relay
opto-isolator (standard)
relay(standard)

1/7

Speciﬁcations : Display / Parameter setting unit / USB-RS232C converter

TMCXseries

Model
Communications
Connection
Emissivity
Analog output
Alarm outputs
Alarm output
Alarm output optional response time
Peak hold reset mode
Sensor correction function
Display Options
Resolution
Temperature Units
Supply Voltage
DimensionsWeightInstallation typeOperation
Ambient temperature
Ambient humidity
Storage temperatureStandard attachment
Memory card

Touch screen type

RoHS
TMCX-TDE
RS232C (output swing ±5V) Maximum speed 5mse-CON
setting range: 0.050 ~ 1.000 (setting resolution0.001)
correction function of reﬂection
via a thermometer output. Check the TMHX and the FTKX series' instruction manual for speciﬁcations.
2
2 alarm outputs, hysteresis range: 0 ~ 99℃, resolution1℃
relay (24V DC, 1A / isolator 500V DC)
230ms（90％）
time: 0.01 ~ 10.00s (conﬁgurable) ,discharge: 0.01 ~ 10.00s, level 0.20 ~ 1.00
span: 0.500 ~ 2.000, Zero: -50.0 ~ +50.0( ℃ or ℉ selectable)temperature units ℃ or ℉ (selectable)graph options (displayed instantaneous temperature, hold)
The background turns bright red when an alarm is activated.
All the settings are adjustable via the built-in touch screen interface.
switchable 1℃ or 0.1℃ (above 1000℃ is to 1℃) Max. 3276℃℃ or ℉
DC22.8 ~ 25.2V（DC24V±5％）100mA max.
98W×64H×36Dmm
324ｇ
wall mount
touch screen type0 ~ 60℃≦95% (non-condensing)-20 ~ 80℃
lithium Coin Cell Battery (BR1225 3V)
microSD card, Max. 32GB (optional)

CE Mark Compliant (EMC EN61326-1: 2013, RoHS EN50581: 2012)

2/7

Speciﬁcations : Display / Parameter setting unit / USB-RS232C converter

PWCX/PWUX

ModelCommunications
Connection with thermometer
Emissivity
Analog output
Alarm outputs
Alarm output
Alarm output optional response time
Alarm output
Sample and holdSensor correction function
Display Options
Resolution
Temperature Units
Supply Voltage
Dimensions
WeightAmbient temperature
Ambient humidityStorage temperature
Attachment

Parameter setting unit

USB-RS232C converter

RoHSRoHSPWCXRS232C (output swing ±5V) Maximum speed 5ms

PWUX
RS232C (output swing ±12V)
e-CON
guaranteed range: 0.3 ~ 1.0
setting range: 0.050 ~ 1.000 (setting resolution0.001)
correction function of reﬂection

via a thermometer output. Check the TMHX and theFTKX series' instruction manual for speciﬁcations.
1hysteresis: 0 ~ 99℃, resolution1℃
opto-isolator: 24V DC, 50mA
5ms

none
none
none
nonetime: 0.01 ~ 10.00s (conﬁgurable) , discharge: 0.01 ~ 10.00s, level 0.20 ~ 1.00
by operating with a PC
by pressing the ENT key
span: 0.500 ~ 2.000, Zero: -50.0 ~ +50.0( ℃ or ℉ selectable)
Upper line: Temperature / Alarm H / Alarm L / Blank

Bottom line: Temperature unit / Emissivity /Alarm H / Alarm L / Blank
switchable 1℃ to 0.1℃(above 1000℃ is to 1℃) Max. 3276℃℃ or ℉
8 batteries
64W×137H×70.4Dmm
500g-10 ~ 50℃
≦90%RH (non-condensing)-20 ~ 65℃
8 batteries

on the PC screennoneUSB
41W×15H×80Dmm (except:nubs)
50g (except:USB cable)
0 ~ 50℃≦80%RH (non-condensing)-10 ~ 60℃
USB miniB cable 1.5ｍ

3/7

Speciﬁcations : Display / Parameter setting unit / USB-RS232C converter

Dimensions

TMCX-H

Panel cut-out size

9696

LKSH

MODE

ENT

25

13226

14

327

15

428

16529630

18731

19832

20

933

21

1034

92 +0.8−0117≧130PH

A288A192 +0.8
−022

1135

23

1236

24

≧13069

TMCX-NDE
Panel cut-out size+0.6MODE

PH SHLK4848

4/7561ENT+0.6

3245 −01

≧60

48

A1

44.7

45 −07866≧60

Speciﬁcations : Display / Parameter setting unit / USB-RS232C converter

Dimensions

TMCX-TDE

98

TMHXSensor

Mounting hole
2-φ4.2 counterbore
φ7.4, depth18

Power &Output

NO2
COM2
NC2

36

NO1
COM1
NC1

OPOP+
PWR+
PWR-

Red
Yellow
Black
Orange

64

White
Green
PurpleBlue

86

Relays
1&2

19-24

48

17 AF

2

PWUX

8.4

6262

USB
40.6

PWCX

DC15V

Connector

80.8

A1MODE
LK

14.6

SH

ENT

137

PH

365/7

Speciﬁcations : Display / Parameter setting unit / USB-RS232C converter

Wiring

TMCX-HDC input
TMCX-HD＋

DC13~27V

AC input
TMCX-HA
AC100~240V

113214315416517 17−

6−Alarm output1
Alarm output2

Analog output1
0 ~ 20mA/4 ~ 20mA
0 ~ 1V/mV/℃
switchable
Analog output2
0 ~ 20mA/4 ~ 20mA
0 ~ 1V/mV/℃
switchable
RS-485M3 Terminals＋

Load−＋

External
input timing＋

Emissivity
input (0 ~ 5V)−

Red1118 18

Yellow

22＋719 19Black33−

820 20Orange44＋921 21White55−

1022 22Green66＋1123 23Purple

77−1224 24Blue

88LoadAttach: in case of 0 ~ 5V or 0 ~ 10V
Resistor for 0 ~ 5V（TR-251N）
Resistor for 0 ~ 10V（TR-501N）
Unnecessary : in case of built-in resistor

0.3□

VoltageON

20mA
Alarmoutput/
Externalsight

DC+IN(4.7〜27V)

+1

A.GND

87+A.O
6ALM
5RxD
4GND
3TxD
2+12.8V
1−A.O

Connection cable

6/7

TxD

Alarm output/
External sight

GND

RxD

ALM
+A.O

20mA-A.O
A.GND

Voltage
ONM3 Terminals
e-CON

2−

DC+IN(4.7〜27V)

Connector

TMCX-NDE
Sensor head+12.8V

Connection cable

Relay cable
TMBX-H (standard)

Emissivity input(0 ~ 5V)

Sensor head

TMHX/FTKX

876543213

Power supply:
8~27V DC or 8 batteries+45678−Load＋−＋−

Alarm output1
Analog output1

Speciﬁcations : Display / Parameter setting unit / USB-RS232C converter

TMCX-TDE Wiring
Sensor head
TMHX

Connecting cable
Red

DC+IN

3.3V

Yellow

TXD

Black

GND

Alarm output

TMHX
SensorExternalsight input

OrangeRXD

White

Analogoutput

GreenRed
YellowBlackOrangeWhiteGreenPurple
BluePurple

Power &Output

A. GNDBlue

Analog output -Relays
1&2

Analog output +DC24 +
DC24 -

Alarm relay
output1

OPOP+PWR+
PWRNO1COM1
NC1NO2COM2
NC2

Alarm relay
output2
The TMCX-TDE is supplied with an e-CON, 8 pin easily attachable connector, for connection with the TMHX series.
Please refer to the wiring above when you need to connect the lead cable of the TMHX directly.

Accessories
Terminal Cover

5g

Branch cable

10g

TCX-H
TCX-N

For protectionof terminal

85g

TMBX-B01
For setting unit
when thermometeris used alone.
e-CON branch

Resistor:
0 ~ 5V / 0 ~ 10V

Capacitor :noise suppressor for analog

Ferrite Core2g2g

TR-251T
TR-501T

TC-105T

FC-2032

For Analog output
noiseConnect to Analog
signal receiver

For Power Noise.
Install to cables

Analog output
TR-251N for 0 ~ 5V
TR-501N for 0 ~ 10V
Convert current
0 ~ 20mA to voltage

23g● The contents described in this catalog are subject to change for

improvement without notice.

● If you have any questions, contact us at the address or links shown below.

http://www.japansensor.co.jp
info@japansensor.co.jp

TMCX-Spec-V201906P201906 1K

7/7