Keep watch on ordinary
Transmit extraordinary

Water level monitoring/Web camera monitoring/com

Che// ckw
w
w it.pot ouekan t!
et!
.

POTEKA Series☆ the water level sensors are adopted as 3L sensors by MLIT

Web camera．．．

http:Water level sensor; Radio wave/Pressure/Ultra sonicAgainst Heavy Rain Disaster

POTEKAis supporting disaster monitoring in Terrestrial rain, etc as well as

weather observation. It monitors landslides and water level condition by optional water levelsensor and web camera. It is also useful for judgement whether the evacuation order is lifted○ Easy installation

Solar panel & 3G module is mounted and No electric wire arrangement required .

○ 3 type of water level sensor selectable

Pressure type, Radio wave type , Ultra sonic type can be chosen depending on installation point！

○Real time monitoring by rain gauge and camera
Rain fall data: every one minute. Picture: every 10 minutes○Check with Internet environment anytime anywhere
In office, out of office, No designated software required○Disseminate the alert by Email

Automatic notification according to hourly rain fall fain fall intensity

POTEKA NET
JMA certified

Compact Weather Station

POTEKAsensor

Observation items
Temperature
Pressure
Humidity
Solar radiation
Wind SP DI
Rain Detection

Screen3G

Artificial photoCloudServer

3G

Web Camera○○ SiteInternet
EmailMonitor
CameraWater level sensor;
Radio wave(Recommended)
Pressure(Recommended)
Ultra sonicObservation Station※No dedicatedsoftware required☞ System composition
POTEKA sensor
Monitor Camera

Observation items: Temperature, humidity, Pressure, Solar Radiation, Wind speed & direction, Rain detection

Water Level sensor

Pressure type, Radio wave type , Ultra sonic type ; Output 4-20mA

Static screen VGA 640 x 480 Minimum illuminance 0.6lx Size 100x 100x 74mm, Weight 350g
Tipping bucket, Diameter φ200mm, Resolution 0.5mm, JMA certified, sizeφ222 x 450mm, weight 1.7kgs

Rain gauge

Analysis; Rainfall intensity, one hour accumulation, Continuous rainfall, effective rainfall

Power Comm Box

3G module(rainfall data: every one minute, Static image: every 10 minutes, automatic transmission, Battery back up(3ayswithout sunshine

Solar Panel30w x 2panels(min) for Web camera and water level sensor, 30w x 1panel(min) POTEKA sensor,

Base & Pole
Operating Condition

Size 1.0 x 1.0 x 1.5m, Weight 70kg including each composition (150kg including weights)

Cloud server

Meisei Cloud server

Web page screen

General Browse (Map screen, Live screen)

Service to be available

Dissemination by Email according to hour rainfall, rain fall intensity

Temperature – 10 ～ 50 ℃ , Wind speed resistance 60 m

●For safe and correct usage, please read the "Operation Manual" prior to the use of the products.

●The specifications and appearances might be changed without prior notice, which please understand.

●The specifications shown in the catalog are of our standard products. We are pleased to customize it to meet customer's requirements. For the details, please contact us.
●Please understand in advance that our company cannot assume the responsibility of any claims made by the third party about any monetary damages or any loss of profits

arising out from the use of our products.
●The color of the product photography on catalog might be different from that of actual product because of printing.

MEISEI ELECTRIC CO., LTD.

1-1, Toyosu 3-chome, Koto-ku, Tokyo 135-8115, Japan
Tel: +81-3-6204-8254 Fax: +81-3-6204-8888
Meteorological & Disaster Prevention Systems Div..
https://www.meisei.co.jp/MSPA4-087 M2009

The specifications this catalog are current as of September 2020