6K-ERB08
8-relay accessory board
for use with PCI-DAS6000 Series and
PCI-DAC6700 Series boards
User's Guide

Document Revision 2A, July, 2006
© Copyright 2006, Measurement Computing Corporation

Trademark and Copyright Information
Measurement Computing Corporation, InstaCal, Universal Library, and the Measurement Computing logo areeither trademarks or registered trademarks of Measurement Computing Corporation. Refer to the Copyrights &Trademarks section on mccdaq.com/legal for more information about Measurement Computing trademarks.
Other product and company names mentioned herein are trademarks or trade names of their respective
companies.
© 2006 Measurement Computing Corporation. All rights reserved. No part of this publication may be
reproduced, stored in a retrieval system, or transmitted, in any form by any means, electronic, mechanical, by
photocopying, recording, or otherwise without the prior written permission of Measurement Computing
Corporation.
NoticeMeasurement Computing Corporation does not authorize any Measurement Computing Corporation product for
use in life support systems and/or devices without prior written consent from Measurement Computing
Corporation. Life support devices/systems are devices or systems that, a) are intended for surgical implantation
into the body, or b) support or sustain life and whose failure to perform can be reasonably expected to result ininjury. Measurement Computing Corporation products are not designed with the components required, and are
not subject to the testing required to ensure a level of reliability suitable for the treatment and diagnosis ofpeople.

HM 6K-ERB08.doc

ii

iii

Table of Contents
Preface

About this User’s Guide ......................................................................................................................vi
What you will learn from this user’s guide........................................................................................................vi
Conventions in this user's guide ........................................................................................................................vi
Where to find more information ........................................................................................................................vi
Chapter 1

Introducing the 6K-ERB08................................................................................................................ 1-1
Overview: 6K-ERB08 features....................................................................................................................... 1-1
Software features ............................................................................................................................................ 1-1
Chapter 2

Installing the 6K-ERB08.................................................................................................................... 2-1
What comes with your 6K-ERB08 shipment?................................................................................................ 2-1
Hardware ....................................................................................................................................................................... 2-1
Additional documentation.............................................................................................................................................. 2-2
Optional components ..................................................................................................................................................... 2-2

Unpacking the 6K-ERB08.............................................................................................................................. 2-2
Connecting the 6K-ERB08 for I/O operations................................................................................................ 2-2

Connectors, cables – interface I/O connectors ............................................................................................................... 2-3
Pin out – interface connectors........................................................................................................................................ 2-4Signal cables .................................................................................................................................................................. 2-7

Connecting to your 6000 Series control board ............................................................................................... 2-8
Configuring your 6K-ERB08 board ............................................................................................................... 2-8

Power source.................................................................................................................................................................. 2-9
Relay logic ................................................................................................................................................................... 2-10

Chapter 3

Functional Details ............................................................................................................................. 3-1
6K-ERB08 block diagram .............................................................................................................................. 3-1
Components.................................................................................................................................................... 3-2
Screw terminal connections............................................................................................................................ 3-3
Wire gauge..................................................................................................................................................................... 3-3

Power up conditions ....................................................................................................................................... 3-3
Relay contact protection circuit for inductive loads ....................................................................................... 3-4
6K-ERB08 assembly diagram ........................................................................................................................ 3-5
Chapter 4

Specifications.................................................................................................................................... 4-1
Power consumption ........................................................................................................................................ 4-1
Output specification........................................................................................................................................ 4-1
Environmental ................................................................................................................................................ 4-1
Mechanical ..................................................................................................................................................... 4-1
Relay screw terminals..................................................................................................................................... 4-2
Relay logic jumper (JP2) ................................................................................................................................ 4-2
Power-in jumper (JP1).................................................................................................................................... 4-2
Relays pull-up/down option............................................................................................................................ 4-2
Compatible products....................................................................................................................................... 4-3
Main connectors and pin out........................................................................................................................... 4-3

iv

6K-ERB08 User's Guide

User connector P20 ........................................................................................................................................................ 4-3
User connector P21 ........................................................................................................................................................ 4-4
User connector P5.......................................................................................................................................................... 4-5
User connector P6.......................................................................................................................................................... 4-6v

Preface

About this User’s Guide
What you will learn from this user’s guideThis user’s guide explains how to install, configure, and use the 6K-ERB08 so that you get the most out of its
features.
This user’s guide also refers you to related documents available on our web site, and to technical support
resources.

Conventions in this user's guide
For more information on …
Text presented in a box signifies additional information and helpful hints related to the subject matter you arereading.
Caution! Shaded caution statements present information to help you avoid injuring yourself and others,
damaging your hardware, or losing your data.
<#:#>

Angle brackets that enclose numbers separated by a colon signify a range of numbers, such as those assigned
to registers, bit settings, etc.

bold text

Bold text is used for the names of objects on the screen, such as buttons, text boxes, and check boxes. For
example:1. Insert the disk or CD and click the OK button.

italic text

Italic text is used for the names of manuals and help topic titles, and to emphasize a word or phrase. For
example:The InstaCal installation procedure is explained in the Quick Start Guide.
Never touch the exposed pins or circuit connections on the board.

Where to find more information
The following electronic documents provide information that is relevant to the operation of the 6K-ERB08.







MCC's Specifications: 6K-ERB08 (the PDF version of the Specifications chapter in this guide) is available
on our web site at www.mccdaq.com/pdfs/manuals/6K-ERB08.pdf.
MCC's Quick Start Guide is available on our web site atwww.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
MCC's Guide to Signal Connections is available on our web site atwww.mccdaq.com/signals/signals.pdf.
MCC's Universal Library User's Guide is available on our web site atwww.mccdaq.com/PDFmanuals/sm-ul-user-guide.pdf.
MCC's Universal Library Function Reference is available on our web site atwww.mccdaq.com/PDFmanuals/sm-ul-functions.pdf.
MCC's Universal Library for LabVIEW™ User’s Guide is available on our web site atwww.mccdaq.com/PDFmanuals/SM-UL-LabVIEW.pdf.

6K-ERB08 User's Guide (this document) is also available on our web site atwww.mccdaq.com/PDFmanuals/6K-ERB08.pdf.

vi6K-ERB08 User's Guide

About this User’s Guide

Links to the hardware manuals for each board supported by the 6K-ERB08 are listed below:
PCI-DAS6000 Series board

Online User’s Guide

PCI-DAS6013, PCI-DAS6014
PCI-DAS6023, PCI-DAS6025
PCI-DAS6030, PCI-DAS6032
PCI-DAS6031, PCI-DAS6033
PCI-DAS6034, PCI-DAS6035, PCI-DAS6036
PCI-DAS6040
PCI-DAS6052
PCI-DAS6070
PCI-DAS6071

www.mccdaq.com/PDFmanuals/pci-das6013-6014.pdf
www.mccdaq.com/PDFmanuals/pci-das6025-23.pdf
www.mccdaq.com/PDFmanuals/pci-das6030-32.pdf
www.mccdaq.com/PDFmanuals/pci-das6031-33.pdf
www.mccdaq.com/PDFmanuals/pci-das603x.pdf
www.mccdaq.com/PDFmanuals/pci-das6040.pdf
www.mccdaq.com/PDFmanuals/pci-das6052.pdf
www.mccdaq.com/PDFmanuals/pci-das6070.pdf
www.mccdaq.com/PDFmanuals/pci-das6071.pdf

PCI-DAC6700 Series boards
PCI-DAC6702, PCI-DAC6703

www.mccdaq.com/PDFmanuals/pci-dac670x.pdf

viiChapter 1

Introducing the 6K-ERB08
Overview: 6K-ERB08 features
This manual explains how to install and connect the 6K-ERB08 board with supported Measurement Computing
hardware. You can use the 6K-ERB08 in your control applications to switch on and off a variety of devices,such as fans, blowers, pumps, etc.
The 6K-ERB08 has eight single-pole double-throw (SPDT) Form C electromechanical relays that are controlled
by eight digital channels connected to them. The AUXPORT digital I/O lines (DIO0 – DIO7) on yourPCI-DAS6000 Series board or PCI-DAC6700 Series board directly control the relays on your 6K-ERB08.
The PCI-DAS6000 Series board and PCI-DAC6700 Series board are referred to as the "6000 Series control
board" for the remainder of this guide.
The 6K-ERB08 has two 100-pin and two 50-pin interface connectors. Use one of these connectors to connect toyour 6000 Series control board. You can use the additional connectors to daisy chain the 6K-ERB08 to different
accessory boards.
A 10 k pull-down resistor is installed at each relay control line to pull the digital relay control line low when
disconnected from the 6000 Series control board, or when the digital lines on the 6000 Series control board arein high impedance (or input) mode. You set the relay logic for active high or low with an on-board jumper.
The 6K-ERB08 has 24 screw terminals for connecting your field wiring to the relay contacts. Each relay
provides three connections – normally open (NO), normally closed (NC) and common (C). The board hascomponent locations where you can install a pull-up or pull-down resistor at the NO and NC connections.
You can power the 6K-ERB08 from your PCI-DAS6000 Series board, from your computer's power connectors,
or from the AC adapter included with your 6K-ERB08. You set the power source with an on-board jumper.

Software features
For information on the features of InstaCal and the other software included with your 6K-ERB08, refer to the
Quick Start Guide that shipped with your device. The Quick Start Guide is also available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
Check www.mccdaq.com/download.htm for the latest software version or versions of the software supported
under less commonly used operating systems.

1-1

Chapter 2

Installing the 6K-ERB08
What comes with your 6K-ERB08 shipment?
The following items are shipped with the 6K-ERB08.

Hardware

6K-ERB08 (shown with standoffs attached)C-PCPOWER-10 cable — 10 foot cable that you can connect to your computer’s +5 V power supply
connectors or to an external +5 V power supply.



CB-PWR-9 AC adapter — 9 volt, 1 amp DC power supply, 110 VAC input.

Six (6) standoffs with screws for attaching to the 6K-ERB08 board

2-1

6K-ERB08 User's Guide

Installing the 6K-ERB08

Additional documentation
In addition to this hardware user's guide, you should also receive the Quick Start Guide (available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf). This booklet supplies a brief description of
the software you received with your 6K-ERB08 and information regarding installation of that software. Please
read this booklet completely before installing any software or hardware.

Optional components


Cables

C100MMS-x

C100HD50-x

C50FF-x

Unpacking the 6K-ERB08
As with any electronic device, you should take care while handling to avoid damage from static
electricity. Before removing the 6K-ERB08 from its packaging, ground yourself using a wrist strap or by simply
touching the computer chassis or other grounded object to eliminate any stored static charge.
If any components are missing or damaged, notify Measurement Computing Corporation immediately byphone, fax, or e-mail:

Phone: 508-946-5100 and follow the instructions for reaching Tech Support.
Fax: 508-946-9500 to the attention of Tech Support
Email: techsupport@mccdaq.com

Connecting the 6K-ERB08 for I/O operations
Follow the guidelines below when connecting the 6K-ERB08 to your computer and to your MCC hardware.


Make sure your connections to the 6K-ERB08 do not exceed any of the specified maximum ratings, or you
could damage the 6K-ERB08, your computer, and your 6000 Series control board. The maximum ratingsfor connections to the 6K-ERB08 are listed in Chapter 4 "Specifications ."Turn off your PC and any 6K-ERB08 power sources before you connect or disconnect the cable from the
6K-ERB08 to the 6000 Series control board.
Connect one of the interface connectors to the 6000 Series control board’s interface cable. Refer to Table
2-1 for appropriate cable and connector combinations.
The remaining interface connectors are provided for connections to other accessory boards, such as the
CIO-MINI50 or CIO-TERM100 screw terminal boards.

External power is required when using the C100HD50-x cable
The C100HD50-x cable does not provide power to the 6K-ERB08. If you use the C100HD50-x cable, power
the 6K-ERB08 with an external power source.
Caution!

Connect the 6K-ERB08 to a PCI-DAS6000 Series board or to a PCI-DAC6700 Series board only.
Connecting to a board that is not supported by the 6K-ERB08 can damage both boards.

2-2

6K-ERB08 User's Guide

Installing the 6K-ERB08

Connectors, cables – interface I/O connectors
The 6K-ERB08 board’s eight relays are controlled by your 6000 Series control board’s digital channels through
AUXPORT. Digital channels DIO<0:7> correspond to relays 1 to 8. Connect your signal cable from the6000 Series control board’s interface connector to one of the interface connectors on the relay board.
The 6K-ERB08 has two 100-pin interface connectors labeled P21 and P20, and two 50-pin interface connectors
labeled P6 and P5. Signals pass through each connector. Only connect one of the four interface connectors toyour 6000 Series control board.
Table 2-1 lists the board connectors, applicable cables, and compatible MCC hardware for the 6K-ERB08.
Table 2-1. Board connectors, cables, and compatible hardware
Connector typeCompatible cable(connect to P21 or P20)
Compatible cable(connect to P6 or P5)

Compatible MCC hardware P21 and P20: Shielded, SCSI 100-pin D-type P6 and P5: Unshielded 50 pin ribbon connector
C100MMS-x shielded round cable (Figure 2-1). x = 1, 2, or 3 meters C100HD50-x shielded round cable (Figure 2-2). x = 3 or 6 feet
 C50FF-x unshielded ribbon cable (Figure 2-3)
x = 1, 2, 3, 4, 5, 10, 15, 20, 25 or 50 feet
Note: These cables do not provide power to the 6K-ERB08. External power is required.
PCI-DAS6000 Series boards:
 PCI-DAS6013
 PCI-DAS6036
 PCI-DAS6014
 PCI-DAS6052
 PCI-DAS6030
 PCI-DAS6023
 PCI-DAS6031
 PCI-DAS6025
 PCI-DAS6032
 PCI-DAS6040
 PCI-DAS6033
 PCI-DAS6070
 PCI-DAS6034
 PCI-DAS6071
 PCI-DAS6035
PCI-DAC6700 Series analog output boards:
 PCI-DAC6702
 PCI-DAC6703

Daisy chaining to other 6K-ERB08 boards
Do not daisy chain additional 6K-ERB08 boards to the extra interface connectors, since any additional boards
would be controlled with the same DIO lines.

2-3

6K-ERB08 User's Guide

Installing the 6K-ERB08

Pin out – interface connectors
Table 2-2. Connector P20 pin out
Signal NameGND
P21 Pass Through 99
P21 Pass Through 98
P21 Pass Through 97
P21 Pass Through 96
P21 Pass Through 95
P21 Pass Through 94
P21 Pass Through 93
DIO7
DIO6
DIO5
DIO4
DIO3
DIO2
DIO1
DIO0
P21 Pass Through 84
P21 Pass Through 83
P21 Pass Through 82
P21 Pass Through 81
P21 Pass Through 80
P21 Pass Through 79
P21 Pass Through 78
P21 Pass Through 77
P21 Pass Through 76
P21 Pass Through 75
P21 Pass Through 74
P21 Pass Through 73
P21 Pass Through 72
P21 Pass Through 71
P21 Pass Through 70
P21 Pass Through 69
P21 Pass Through 68
P21 Pass Through 67
P21 Pass Through 66
P21 Pass Through 65
P21 Pass Through 64
P21 Pass Through 63
P21 Pass Through 62
P21 Pass Through 61
P21 Pass Through 60
P21 Pass Through 59
P21 Pass Through 58
P21 Pass Through 57
P21 Pass Through 56
P21 Pass Through 55
P21 Pass Through 54
P21 Pass Through 53
P21 Pass Through 52
P21 Pass Through 51

Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

2-4

Pin

Signal Name

50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

GND
P21 Pass Through 49
P21 Pass Through 48
P21 Pass Through 47
P21 Pass Through 46
P21 Pass Through 45
P21 Pass Through 44
P21 Pass Through 43
P21 Pass Through 42
P21 Pass Through 41
P21 Pass Through 40
PC +5VP21 Pass Through 38
P21 Pass Through 37
P21 Pass Through 36
P21 Pass Through 35
P21 Pass Through 34
P21 Pass Through 33
P21 Pass Through 32
P21 Pass Through 31
P21 Pass Through 30
P21 Pass Through 29
P21 Pass Through 28
P21 Pass Through 27
P21 Pass Through 26
P21 Pass Through 25
P21 Pass Through 24
P21 Pass Through 23
P21 Pass Through 22
P21 Pass Through 21
P21 Pass Through 20
P21 Pass Through 19
P21 Pass Through 18
P21 Pass Through 17
P21 Pass Through 16
P21 Pass Through 15
P21 Pass Through 14
P21 Pass Through 13
P21 Pass Through 12
P21 Pass Through 11
P21 Pass Through 10
P21 Pass Through 9
P21 Pass Through 8
P21 Pass Through 7
P21 Pass Through 6
P21 Pass Through 5
P21 Pass Through 4
P21 Pass Through 3
P21 Pass Through 2
P21 Pass Through 1

6K-ERB08 User's Guide

Installing the 6K-ERB08

Table 2-3. Connector P21 pin out
Signal Name
GND
P20 Pass Through 99
P20 Pass Through 98
P20 Pass Through 97
P20 Pass Through 96
P20 Pass Through 95
P20 Pass Through 94
P20 Pass Through 93
DIO7
DIO6
DIO5
DIO4
DIO3
DIO2
DIO1
DIO0
P20 Pass Through 84
P20 Pass Through 83
P20 Pass Through 82
P20 Pass Through 81
P20 Pass Through 80
P20 Pass Through 79
P20 Pass Through 78
P20 Pass Through 77
P20 Pass Through 76
P20 Pass Through 75
P20 Pass Through 74
P20 Pass Through 73
P20 Pass Through 72
P20 Pass Through 71
P20 Pass Through 70
P20 Pass Through 69
P20 Pass Through 68
P20 Pass Through 67
P20 Pass Through 66
P20 Pass Through 65
P20 Pass Through 64
P20 Pass Through 63
P20 Pass Through 62
P20 Pass Through 61
P20 Pass Through 60
P20 Pass Through 59
P20 Pass Through 58
P20 Pass Through 57
P20 Pass Through 56
P20 Pass Through 55
P20 Pass Through 54
P20 Pass Through 53
P20 Pass Through 52
P20 Pass Through 51

Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

2-5

Pin

Signal Name

50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

GND
P20 Pass Through 49
P20 Pass Through 48
P20 Pass Through 47
P20 Pass Through 46
P20 Pass Through 45
P20 Pass Through 44
P20 Pass Through 43
P20 Pass Through 42
P20 Pass Through 41
P20 Pass Through 40
PC +5VP20 Pass Through 38
P20 Pass Through 37
P20 Pass Through 36
P20 Pass Through 35
P20 Pass Through 34
P20 Pass Through 33
P20 Pass Through 32
P20 Pass Through 31
P20 Pass Through 30
P20 Pass Through 29
P20 Pass Through 28
P20 Pass Through 27
P20 Pass Through 26
P20 Pass Through 25
P20 Pass Through 24
P20 Pass Through 23
P20 Pass Through 22
P20 Pass Through 21
P20 Pass Through 20
P20 Pass Through 19
P20 Pass Through 18
P20 Pass Through 17
P20 Pass Through 16
P20 Pass Through 15
P20 Pass Through 14
P20 Pass Through 13
P20 Pass Through 12
P20 Pass Through 11
P20 Pass Through 10
P20 Pass Through 9
P20 Pass Through 8
P20 Pass Through 7
P20 Pass Through 6
P20 Pass Through 5
P20 Pass Through 4
P20 Pass Through 3
P20 Pass Through 2
P20 Pass Through 1

6K-ERB08 User's Guide

Installing the 6K-ERB08

Table 2-4. Connector P5 pin out
Signal Name

Pin

GND
P6 Pass Through 98
P6 Pass Through 96
P6 Pass Through 94
DIO7
DIO5
DIO3
DIO1
P6 Pass Through 84
P6 Pass Through 82
P6 Pass Through 80
P6 Pass Through 78
P6 Pass Through 76
P6 Pass Through 74
P6 Pass Through 72
P6 Pass Through 70
P6 Pass Through 68
P6 Pass Through 66
P6 Pass Through 64
P6 Pass Through 62
P6 Pass Through 60
P6 Pass Through 58
P6 Pass Through 56
P6 Pass Through 54
P6 Pass Through 52

50
48
46
44
42
40
38
36
34
32
30
28
26
24
22
20
18
16
14
12
10
8
6
4
2

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

Pin

Signal Name

49
47
45
43
41
39
37
35
33
31
29
27
25
23
21
19
17
15
13
11
9
7
5
3
1

P6 Pass Through 99
P6 Pass Through 97
P6 Pass Through 95
P6 Pass Through 93
DIO6
DIO4
DIO2
DIO0
P6 Pass Through 83
P6 Pass Through 81
P6 Pass Through 79
P6 Pass Through 77
P6 Pass Through 75
P6 Pass Through 73
P6 Pass Through 71
P6 Pass Through 69
P6 Pass Through 67
P6 Pass Through 65
P6 Pass Through 63
P6 Pass Through 61
P6 Pass Through 59
P6 Pass Through 57
P6 Pass Through 55
P6 Pass Through 53
P6 Pass Through 51

Table 2-5. Connector P6 pin out
Signal Name
GND
P5 Pass Through 98
P5 Pass Through 96
P5 Pass Through 94
DIO7
DIO5
DIO3
DIO1
P5 Pass Through 84
P5 Pass Through 82
P5 Pass Through 80
P5 Pass Through 78
P5 Pass Through 76
P5 Pass Through 74
P5 Pass Through 72
P5 Pass Through 70
P5 Pass Through 68
P5 Pass Through 66
P5 Pass Through 64
P5 Pass Through 62
P5 Pass Through 60
P5 Pass Through 58
P5 Pass Through 56
P5 Pass Through 54
P5 Pass Through 52

Pin
50
48
46
44
42
40
38
36
34
32
30
28
26
24
22
20
18
16
14
12
10
8
6
4
2

••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••
••

Pin

Signal Name

49
47
45
43
41
39
37
35
33
31
29
27
25
23
21
19
17
15
13
11
9
7
5
3
1

P5 Pass Through 99
P5 Pass Through 97
P5 Pass Through 95
P5 Pass Through 93
DIO6
DIO4
DIO2
DIO0
P5 Pass Through 83
P5 Pass Through 81
P5 Pass Through 79
P5 Pass Through 77
P5 Pass Through 75
P5 Pass Through 73
P5 Pass Through 71
P5 Pass Through 69
P5 Pass Through 67
P5 Pass Through 65
P5 Pass Through 63
P5 Pass Through 61
P5 Pass Through 59
P5 Pass Through 57
P5 Pass Through 55
P5 Pass Through 53
P5 Pass Through 51

Information on signal connections
For general information about signal connection and configuration, refer to the Guide to Signal Connections on
our web site at www.mccdaq.com/signals/signals.pdf.

2-6

6K-ERB08 User's Guide

Installing the 6K-ERB08

Signal cables
50

100

50

100

1

51

1

51

Figure 2-1. C100MMS-x cable

Details on the C100MMS-x cable are available on our web site at www.mccdaq.com.
.
Pins 1-50 are on the long sideof the “D” connector.

50

Strain relief isstamped “Pins 1-50”.

49

50

1

2

100

Key

The red stripe
identifies pin # 1
99 100

1

51

Key

Pins 51-100 are on
the short side of
the “D” connector.

51 52

Strain relief is
Stamped “Pins 51-100”.

The red stripe
identifies pin # 51

Figure 2-2. C100HD50-x cable

Details on the C100HD50-x cable are available on our web site at www.mccdaq.com.
.
2

1

50

49

The red stripe
identifies pin # 1

Figure 2-3. C50FF-x cable

2-7

2

1

50

49

6K-ERB08 User's Guide

Installing the 6K-ERB08

Details on the C50FF-x cable are available on our web site atConnecting to your 6000 Series control board
The following procedure provides the general steps to follow to connect your 6K-ERB08 to a PCI-DAS6000
Series board or to a PCI-DAC6700 Series board.
1.

Turn off your computer and open the cover.

2.

Install your PCI-DAS6000 Series or PCI-DAC6700 Series board in the computer. Refer to your board’s
hardware User Guide for installation instructions.

3.

Connect the digital signal cable from one of the interface connectors on the 6K-ERB08 to the 100-pin
interface connector on your 6000 Series control board. Your signal cable determines which interface
connector to connect to.
ooC100MMS-x cable: Connect one end of the cable to either 100-pin interface connector (P21 or P20)

on the 6K-ERB08, and connect the other end of the cable to the 100-pin interface connector on the6000 Series control board.
C100HD50-x cable: Connect the 50-pin cable end stamped Pins 51 to 100 to either 50-pin connector(P6 or P5) on the 6K-ERB08, and connect the 100-pin cable end to the 100-pin interface connector on
the 6000 Series control board.

External power is required when using the C100HD50-x cable
The C100HD50-x cable does not provide power to the 6K-ERB08. If you use the C100HD50-x cable, power
the 6K-ERB08 with an external power source.
4.

To connect other accessory boards or devices to your 6K-ERB08, use the remaining 100-pin interface
connector and/or the two 50-pin connectors for the connection(s). The following typical configurations are
possible:ooWhen using a C100MMS-x cable, you can use a second C100MMS-x cable to daisy chain to an SCB100 accessory board, OR you could use a C50FF-x cable to daisy chain to a 50-pin accessory board.
When using a C100HD50-x cable, you can use a C50FF-x cable to daisy chain to a 50-pin accessory
board.

Configuring your 6K-ERB08 board
Configure the power source jumper and relay logic jumper before using the 6K-ERB08. The factory-configured
default settings are listed in Table 2-6.
Table 2-6. Default hardware configuration
Jumper label

Jumper description

Default setting

POWER IN
JP1

Sets the power source to one of three options: +5BD: +5V power from your 6000 Series control board.
 +5PC: +5V power from the computer’s power supply.
 +9V EXT: power from an AC adapter.
Sets the relay logic for active high or active low.

+5BD

INVERT / NON-INVERTJP2

Figure 2-4 shows the location of the jumpers and external power connectors.

2-8

NON-INVERT(active high)6K-ERB08 User's Guide

Installing the 6K-ERB08

Figure 2-4. 6K-ERB08 board jumpers and external power connectors

Power source
Set the power source with jumper JP1 (labeled POWER IN). You can power the 6K-ERB08 using one of thefollowing sources.


The 6000 Series control board’s internal +5 V power supply (default)Your computer’s +5 V power supply, or external +5 V supply (power cable supplied)
AC adapter (supplied)Figure 2-5 shows the POWER IN jumper setting for each power source.
9V EXT+5BD+5PC

Internal powerfrom the MCC board

External power fromthe computer’s power supply

External power froman AC adapter

Figure 2-5. Power source jumper JP1configurations

External +5 V PC power connections
You can power the 6K-ERB08 with your computer’s +5 V power supply using the C-PCPOWER-10 cord thatwas shipped with the board. Each end of the cable has a keyed MOLEX type connector. To connect to yourcomputer’s internal power connectors, do the following:
1.

Turn your computer off, remove the cover, and insert your board into an available PCI slot.

2.

Close your computer and turn it on.

3.

Turn off power to the computer and remove the cover.

4.

Connect one end of the C-PCPOWER10 power cord to one of your computer’s internal power connectors.
If necessary, refer to your computer's user manual for the location of the power supply connectors.

5.

Run the power cable out the back of the computer through an expansion slot or other opening, and replace
the cover on the computer.
2-9

6K-ERB08 User's Guide

Caution!

Installing the 6K-ERB08

Be careful not to pinch the cable when you replace the cover — if this cable is cut, the resulting
short circuit can damage the computer.

6.

Connect the other end of the power cord to the connector labeled PC POWER connector (P18) on the 6KERB08.

7.

Set the POWER IN jumper (JP1) for +5PC.

External +9V AC power adapter connections
You can power the 6K-ERB08 using the AC power adapter (MCC part number CB-PWR-9). This adapter
provides 9 volt, 1 amp DC power, 110 VAC power. To connect the AC adapter, do the following.
1.

Turn off power to the computer.

2.

Connect the CB-PWR-9 cable to the 9V EXTERNAL POWER connector (P19).

3.

Plug the AC adapter into a power outlet.

4.

Configure the POWER IN jumper (JP1) for 9V EXT.

Relay logic
Use jumper JP2 to select inverting or non-inverting logic to control the relays. Refer to Figure 2-4 for thelocation of this jumper. The relay logic jumper is shown here configured for NON-INVERT (digital high). Thisis the default position.
JP2
INVERT

NON-INVERT

1 2 3
Figure 2-6. Relay logic jumper JP2 (default position)



When set to non-invert, the relay activates when the DIO signal is high (1), and becomes inactive(non-energized) when the DIO signal is low (0).
When set to invert, the relay activates when the DIO signal is low (0), and becomes inactive(non-energized) when the DIO signal is high (1).

Table 2-7 summarizes the relay state in conjunction with the relay logic jumper setting.
Table 2-7. Relay state and jumper position
Relay state
Active (energized)
Inactive (not-energized)

JP2 position
INVERT

NON-INVERT

LO
HI

HI
LO

2-10

Chapter 3

Functional Details
6K-ERB08 block diagram
6K-ERB08 functions are illustrated in the block diagram shown here.

5V
regulator

Screw terminal block

+5V

CH2

Channel 3 relay

CH3

NO C NC

Channel 4 relay

CH4

NO C NC
NO C NC

NO C NC

Channel 2 relay

4

Channel 5 relay

CH5

Channel 6 relay

CH6

Channel 7 relay

CH7

Channel 8 relay

CH8

4

Relay modules 5-8

Figure 3-1. 6K-ERB08 functional block diagram

3-1

NO C NC

Relay
driverCH1

NO C NC

Relay control bits
DIO (4:7)

Relay
driverChannel 1 relay

NO C NC

Relay control bits
DIO (0:3)

NO C NC

Relay modules 1-4

Input power

100-pin I/O connector

PCI-DAS6000 Series or PCI-DAC6700 Series board

+5V

Load

6K-ERB08 User's Guide

Functional Details

Components
Major components on the 6K-ERB08 are shown in Figure 3-2.

Figure 3-2. 6K-ERB08 component locations

Table 3-8 describes the board’s major components.
Table 3-8. 6K-ERB08 component descriptions
Callout

Board
label

Description

1

P21, P20

2

P6, P5

3

RN1

4
5

JP2
JP1

6
7
8

P19
P18
K1 to K8

9

1 to 8100-pin interface connectors for connecting to the 6000 Series control board or different
accessory boards.
50-pin interface connectors for connecting to the 6000 Series control board or different accessory
boards.
Pull-down resistor to control the state of the relay control lines when not driven by the control
board.
Jumper to configure the relay logic for active high or low.
Main +5 V power jumper to select the power source — internal power from the 6000 Series
control board, external power from the computer, or external power from the AC adapter.
External power connector. Connect to the AC adapter.
External power connector. Connect to the computer's +5 V supply.
Relays <1:8>. These relays are controlled digitally by the 6000 Series control board's digital bits
DIO<0:7>.
Relay screw terminals <1:8>. The screw terminals are for the common (C), normally open (NO)and normally closed (NC) contacts for relays 1 through 8.

3-2

6K-ERB08 User's Guide

Functional Details

Screw terminal connections
Connect external devices to the relay contacts using the 6K-ERB08 screw terminals. Each relay has a common(C), normally closed (NC), and normally open (NO) contact. Figure 3-3 shows the screw terminals on a typical
relay channel.

NO C NC
Figure 3-3. Typical relay channel

The relay configuration is illustrated in Figure 3-4. You can install a pull-up or pull-down resistor at the NO andNC terminals on each relay.
User-installed
pull-up / pull-down resistor
+5
GND
+5
GND

InvertLogic
jumperNon-Invert

Buffer/driverDigital output
from theMCC board

NO

C

NC
Screw
terminals(3 perrelay)

10 k
pull-down
resistor

Figure 3-4. Relay configuration

Caution!

Before connecting signal wires to the relay screw terminals, turn off the power to the 6K-ERB08,
and make sure that the signal wires do not contain live voltages.

Wire gauge
Use 12 AWG to 20 AWG wire to connect field devices. Properly insulate the wires to avoid any short circuit tothe other connections, ground, or other points on the board.

Power up conditions
The state of the relay modules at power up depends on both the state of the digital signals controlling them andthe relay logic jumper setting.
When you set the relay logic jumper to NON-INVERT, the C and NC contacts are connected when the 6KERB08 is disconnected from the 6000 Series control board or when the digital lines on the 6000 Series
control board are in high impedance (or input) mode.
When you set the relay logic jumper to INVERT, the C and NO contacts are connected when the 6K-ERB08
is disconnected from the 6000 Series control board, or when the digital lines on the 6000 Series control
board are in high impedance (or input) mode.

3-3

6K-ERB08 User's Guide

Functional Details

Relay contact protection circuit for inductive loads
When you connect an inductive load to a relay, energy stored in the inductive load can induce a large voltage
surge when you switch the relay. This voltage can severely damage the relay contacts. To limit the voltage
surge across the inductive load, install a kickback diode across the DC load. Refer to the contact protection
circuit in Figure 3-5. For AC loads, install a metal oxide varistor (MOV).

Relay
NO

C

NC

+
V
-

Inductive
Load

Kickback
Diode

Figure 3-5. Relay contact protection circuit

3-4

6K-ERB08 User's Guide

Functional Details

6K-ERB08 assembly diagram

Figure 3-6. 6K-ERB08 assembly diagram

3-5Chapter 4

Specifications
Typical for 25 °C unless otherwise specified.
Specifications in italic text are guaranteed by design.

Power consumption
Table 1. Power consumption specifications
5 V PC auxiliary power / PCI bus power /external regulated
External 7.5 V to 15 V unregulated supply

All relays off
All relays on
All relays off
All relays on

15 mA typical, 20 mA max.
480 mA typical, 550 mA max.
20 mA typical, 25 mA max.
500 mA typical, 580 mA max.

Output specification
Table 2. Output specifications
NumberContact configuration

8
8 Form C (SPDT) NO, NC and Common available at screw terminals

Contact rating
Contact resistanceOperate time
Release time
Vibration
Shock
Dielectric isolation (between open contact)
Life expectancy
Power on state (no connection to DIO)

6 A @ 120 VAC or 28 VDC resistive
100 milliohms max (initial value)
10 milliseconds max
5 milliseconds max
10 to 55 Hz (amplitude 1.5 mm)
10 G (11 milliseconds)
300 VAC, 50/60 Hz (1 minute)10 million mechanical operations, min
Not energized. NC in contact to Common. (JP2 in NON-INVERT position)

Environmental
Table 3. Environmental specifications
Operating temperature range
Storage temperature range
Humidity

0 to 70 °C-40 to 100 °C
0 to 95% non-condensing

Mechanical
Table 4. Mechanical specifications
Card dimensions (without standoff)
Weight229 mm (L) x 102 mm (W) x 20 mm (H)
9" (L) x 4" (W) x 0.8" (H)
0.6 lb.

4-1

6K-ERB08 User's Guide

Specifications

Relay screw terminals
Table 5. Relay screw terminal specifications
Wire gauge range

12 AWG to 20 AWG
Table 6. Screw terminal pin out

Pin

Signal Name

1-NO

Relay 1 Normally Open contact

1-C

Relay 1 Common contact

1-NC

Relay 1 Normally Closed contact

2-NO

Relay 2 Normally Open contact

2-C

Relay 2 Common contact

2-NC

Relay 2 Normally Closed contact

3-NO

Relay 3 Normally Open contact

3-C

Relay 3 Common contact

3-NC

Relay 3 Normally Closed contact

4-NO

Relay 4 Normally Open contact

4-C

Relay 4 Common contact

4-NC

Relay 4 Normally Closed contact

5-NO

Relay 5 Normally Open contact

5-C

Relay 5 Common contact

5-NC

Relay 5 Normally Closed contact

6-NO

Relay 6 Normally Open contact

6-C

Relay 6 Common contact

6-NC

Relay 6 Normally Closed contact

7-NO

Relay 7 Normally Open contact

7-C

Relay 7 Common contact

7-NC

Relay 7 Normally Closed contact

8-NO

Relay 8 Normally Open contact

8-C

Relay 8 Common contact

8-NC

Relay 8 Normally Closed contact

Relay logic jumper (JP2)
Table 7. JP2 specifications
Invert (1-2)
Non-invert (2-3) (default)

Relay activates when DIO is LOW
Relay activates when DIO is HIGH

Power-in jumper (JP1)
Table 8. JP1 specifications+5PC+9V EXT+5BD (default)

Use cable C-PCPOWER-10
Use Adapter CB-PWR-9
Powered from 100-pin connector

Relays pull-up/down option
Table 9. Relay pull-up/down specifications
R21,23,25,27,29,31,33,35
R20,22,24,26,28,30,32,34

Relays NC pin pull-up/down
Relays NO pin pull-up/down

4-2

6K-ERB08 User's Guide

Specifications

Compatible products
Table 10. Compatible product specifications
Analog input boards

 PCI-DAS6013
 PCI-DAS6014
 PCI-DAS6030
 PCI-DAS6031
 PCI-DAS6032
 PCI-DAS6033
 PCI-DAS6034
 PCI-DAS6035
 PCI-DAS6036
 PCI-DAS6052
 PCI-DAS6023
 PCI-DAS6025
 PCI-DAS6040
 PCI-DAS6070
 PCI-DAS6071
 PCI-DAC6702
 PCI-DAC6703

Analog output boards

Note 1: The 6K-ERB08 requires external power (for all products above) when used with C100HD50(pins 51-100) ribbon cable.

Main connectors and pin out
User connector P20
Table 11. Connector P20 specifications
Connector typeCompatible cables

Shielded SCSI 100 D-type
C100MMS-x, shielded round cable. x = 1, 2 or 3 meters
Table 12. P20 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28

Signal name
P21 Pass Through 1
P21 Pass Through 2
P21 Pass Through 3
P21 Pass Through 4
P21 Pass Through 5
P21 Pass Through 6
P21 Pass Through 7
P21 Pass Through 8
P21 Pass Through 9
P21 Pass Through 10
P21 Pass Through 11
P21 Pass Through 12
P21 Pass Through 13
P21 Pass Through 14
P21 Pass Through 15
P21 Pass Through 16
P21 Pass Through 17
P21 Pass Through 18
P21 Pass Through 19
P21 Pass Through 20
P21 Pass Through 21
P21 Pass Through 22
P21 Pass Through 23
P21 Pass Through 24
P21 Pass Through 25
P21 Pass Through 26
P21 Pass Through 27
P21 Pass Through 28

Pin
51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78

4-3

Signal name
P21 Pass Through 51
P21 Pass Through 52
P21 Pass Through 53
P21 Pass Through 54
P21 Pass Through 55
P21 Pass Through 56
P21 Pass Through 57
P21 Pass Through 58
P21 Pass Through 59
P21 Pass Through 60
P21 Pass Through 61
P21 Pass Through 62
P21 Pass Through 63
P21 Pass Through 64
P21 Pass Through 65
P21 Pass Through 66
P21 Pass Through 67
P21 Pass Through 68
P21 Pass Through 69
P21 Pass Through 70
P21 Pass Through 71
P21 Pass Through 72
P21 Pass Through 73
P21 Pass Through 74
P21 Pass Through 75
P21 Pass Through 76
P21 Pass Through 77
P21 Pass Through 78

6K-ERB08 User's Guide

Pin
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

Specifications

Signal name
P21 Pass Through 29
P21 Pass Through 30
P21 Pass Through 31
P21 Pass Through 32
P21 Pass Through 33
P21 Pass Through 34
P21 Pass Through 35
P21 Pass Through 36
P21 Pass Through 37
P21 Pass Through 38
PC +5VP21 Pass Through 40
P21 Pass Through 41
P21 Pass Through 42
P21 Pass Through 43
P21 Pass Through 44
P21 Pass Through 45
P21 Pass Through 46
P21 Pass Through 47
P21 Pass Through 48
P21 Pass Through 49
GND

Pin
79
80
81
82
83
84
85
86
87
88
89
90
91
92
93
94
95
96
97
98
99
100

Signal name
P21 Pass Through 79
P21 Pass Through 80
P21 Pass Through 81
P21 Pass Through 82
P21 Pass Through 83
P21 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P21 Pass Through 93
P21 Pass Through 94
P21 Pass Through 95
P21 Pass Through 96
P21 Pass Through 97
P21 Pass Through 98
P21 Pass Through 99
GND

User connector P21
Table 13. Connector P21 specifications
Connector typeCompatible cables

Shielded SCSI 100 D-type
C100MMS-x, shielded round cable. x =1, 2 or 3 meters
Table 14. P21 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36

Signal name
P20 Pass Through 1
P20 Pass Through 2
P20 Pass Through 3
P20 Pass Through 4
P20 Pass Through 5
P20 Pass Through 6
P20 Pass Through 7
P20 Pass Through 8
P20 Pass Through 9
P20 Pass Through 10
P20 Pass Through 11
P20 Pass Through 12
P20 Pass Through 13
P20 Pass Through 14
P20 Pass Through 15
P20 Pass Through 16
P20 Pass Through 17
P20 Pass Through 18
P20 Pass Through 19
P20 Pass Through 20
P20 Pass Through 21
P20 Pass Through 22
P20 Pass Through 23
P20 Pass Through 24
P20 Pass Through 25
P20 Pass Through 26
P20 Pass Through 27
P20 Pass Through 28
P20 Pass Through 29
P20 Pass Through 30
P20 Pass Through 31
P20 Pass Through 32
P20 Pass Through 33
P20 Pass Through 34
P20 Pass Through 35
P20 Pass Through 36

Pin
51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79
80
81
82
83
84
85
86

4-4

Signal name
P20 Pass Through 51
P20 Pass Through 52
P20 Pass Through 53
P20 Pass Through 54
P20 Pass Through 55
P20 Pass Through 56
P20 Pass Through 57
P20 Pass Through 58
P20 Pass Through 59
P20 Pass Through 60
P20 Pass Through 61
P20 Pass Through 62
P20 Pass Through 63
P20 Pass Through 64
P20 Pass Through 65
P20 Pass Through 66
P20 Pass Through 67
P20 Pass Through 68
P20 Pass Through 69
P20 Pass Through 70
P20 Pass Through 71
P20 Pass Through 72
P20 Pass Through 73
P20 Pass Through 74
P20 Pass Through 75
P20 Pass Through 76
P20 Pass Through 77
P20 Pass Through 78
P20 Pass Through 79
P20 Pass Through 80
P20 Pass Through 81
P20 Pass Through 82
P20 Pass Through 83
P20 Pass Through 84
DIO0
DIO1

6K-ERB08 User's Guide

Pin
37
38
39
40
41
42
43
44
45
46
47
48
49
50

Specifications

Signal name
P20 Pass Through 37
P20 Pass Through 38
PC +5VP20 Pass Through 40
P20 Pass Through 41
P20 Pass Through 42
P20 Pass Through 43
P20 Pass Through 44
P20 Pass Through 45
P20 Pass Through 46
P20 Pass Through 47
P20 Pass Through 48
P20 Pass Through 49
GND

Pin
87
88
89
90
91
92
93
94
95
96
97
98
99
100

Signal name
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P20 Pass Through 93
P20 Pass Through 94
P20 Pass Through 95
P20 Pass Through 96
P20 Pass Through 97
P20 Pass Through 98
P20 Pass Through 99
GND

User connector P5
Table 15. Connector P5 specifications
Connector typeCompatible cables

Unshielded 50-pin ribbon connector - male
C100HD50-x, C50FF-x, unshielded ribbon cable. x = 3 or 6 feet
Table 16. P5 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

Signal name
P6 Pass Through 51
P6 Pass Through 52
P6 Pass Through 53
P6 Pass Through 54
P6 Pass Through 55
P6 Pass Through 56
P6 Pass Through 57
P6 Pass Through 58
P6 Pass Through 59
P6 Pass Through 60
P6 Pass Through 61
P6 Pass Through 62
P6 Pass Through 63
P6 Pass Through 64
P6 Pass Through 65
P6 Pass Through 66
P6 Pass Through 67
P6 Pass Through 68
P6 Pass Through 69
P6 Pass Through 70
P6 Pass Through 71
P6 Pass Through 72
P6 Pass Through 73
P6 Pass Through 74
P6 Pass Through 75

Pin
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

4-5

Signal name
P6 Pass Through 76
P6 Pass Through 77
P6 Pass Through 78
P6 Pass Through 79
P6 Pass Through 80
P6 Pass Through 81
P6 Pass Through 82
P6 Pass Through 83
P6 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P6 Pass Through 93
P6 Pass Through 94
P6 Pass Through 95
P6 Pass Through 96
P6 Pass Through 97
P6 Pass Through 98
P6 Pass Through 99
GND

6K-ERB08 User's Guide

Specifications

User connector P6
Table 17. Connector P6 specifications
Connector typeCompatible cables

Unshielded 50 pin ribbon connector - male
C100HD50-x, C50FF-x, unshielded ribbon cable. x = 3 or 6 feet
Table 18. P6 pin out

Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

Signal name
P5 Pass Through 51
P5 Pass Through 52
P5 Pass Through 53
P5 Pass Through 54
P5 Pass Through 55
P5 Pass Through 56
P5 Pass Through 57
P5 Pass Through 58
P5 Pass Through 59
P5 Pass Through 60
P5 Pass Through 61
P5 Pass Through 62
P5 Pass Through 63
P5 Pass Through 64
P5 Pass Through 65
P5 Pass Through 66
P5 Pass Through 67
P5 Pass Through 68
P5 Pass Through 69
P5 Pass Through 70
P5 Pass Through 71
P5 Pass Through 72
P5 Pass Through 73
P5 Pass Through 74
P5 Pass Through 75

Pin
26
27
28
29
30
31
32
33
34
35
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50

4-6

Signal name
P5 Pass Through 76
P5 Pass Through 77
P5 Pass Through 78
P5 Pass Through 79
P5 Pass Through 80
P5 Pass Through 81
P5 Pass Through 82
P5 Pass Through 83
P5 Pass Through 84
DIO0
DIO1
DIO2
DIO3
DIO4
DIO5
DIO6
DIO7
P5 Pass Through 93
P5 Pass Through 94
P5 Pass Through 95
P5 Pass Through 96
P5 Pass Through 97
P5 Pass Through 98
P5 Pass Through 99
GND

Declaration of Conformity
Manufacturer:
Address:

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, MA 02766
USA

Measurement Computing Corporation declares under sole responsibility that the product

6K-ERB08
to which this declaration relates is in conformity with the relevant provisions of the following standards or otherdocuments:
EU EMC Directive 89/336/EEC: Electromagnetic Compatibility, EN 61326 (1997) Amendment 1 (1998)
Emissions: Group 1, Class A
EN 55011 (1990)/CISPR 11: Radiated and Conducted emissions.

Immunity: EN61326, Annex A








IEC 1000-4-2 (1995): Electrostatic Discharge immunity, Criteria C.
IEC 1000-4-3 (1995): Radiated Electromagnetic Field immunity Criteria A.
IEC 1000-4-4 (1995): Electric Fast Transient Burst immunity Criteria B.
IEC 1000-4-5 (1995): Surge immunity Criteria A.
IEC 1000-4-6 (1996): Radio Frequency Common Mode immunity Criteria A.
IEC 1000-4-8 (1994): Magnetic Field immunity Criteria A.
IEC 1000-4-11 (1994): Voltage Dip and Interrupt immunity Criteria A.

Declaration of Conformity based on tests conducted by Chomerics Test Services, Woburn, MA 01801, USA in
April, 2005. Test records are outlined in Chomerics Test Report #EMI3931.04.
We hereby declare that the equipment specified conforms to the above Directives and Standards.

Carl Haapaoja, Director of Quality Assurance

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, Massachusetts 02766(508) 946-5100
Fax: (508) 946-9500
E-mail: info@mccdaq.com
www.mccdaq.com