MODEL 8032-01
ACCELEROMETERSPECIFICATIONS

dimensions

Low Cost Industrial Accelerometer
12kHz Bandwidth.
-55°C to +100°C Operating Range
Integral Cable, Molded Strain Relief

The Model 8032-01 is an internally shielded rugged IEPE
accelerometer designed for harsh environments. The
accelerometer is available in ±50 to ±500g dynamic ranges
and has a flat bandwidth up to greater than12kHz. The
model 8032-01 features an integral molded strain relief andis designed to operating in temperature ranges from -55°C
to +100°C. The unit features the popular stud mountinstallation.

FEATURES

±50g to ±500g Dynamic Range
Wide bandwidth up to 12kHz
Case Isolated, Internally ShieldedReverse Wiring Protection
Stable Temperature Response
Integral Cable, Molded Strain Relief

APPLICATIONSSENSOR SOLUTIONS /// Model 8032-01 Rev B

General Purpose
Machine Monitoring
Industrial Applications
Harsh Environments
Gearbox Monitoring

07/2017

Page 1

MODEL 8032-01 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 4mA excitation unless otherwise stated. Measurement Specialties reserves the right to updateand change these specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Shock Limit (g)Residual Noise (g RMS)
Spectral Noise, 10Hz (µg√Hz)
Spectral Noise, 100Hz (µg√Hz)
Spectral Noise, 1kHz (µg√Hz)±50100
1-80000.4-10000
47000±1<750000.0004127.5
3.5

ELECTRICAL
Compliance Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Bias Voltage (Vdc)
Output Impedance (Ω)
Insulation Resistance (MΩ)
Warm-up Time (sec)Grounding

18 to 30
2 to 108 to 126 to 14<100>100<2Case Isolated, Internally Shielded

ENVIRONMENTAL
Temperature Response (%)
Operating Temperature (°C)
Storage Temperature (°C)
Humidity±5-55 to +100-55 to +100
Environmentally Sealed, Molded, IP67

PHYSICAL
Sensing Element
Cable
Case Material
Weight (grams)
Mounting
Mounting TorqueCeramic
FEP Insulated Leads, Braided Shield, TPE Jacket
Stainless Steel
90¼-28 to ¼-28 Mounting Stud (included)
24 lb-in (2.7 N-m)

Notes
±500
101-8000
0.4-12000
47000±1<75000
0.000925158±10%±1dB±3dB

Broadband 1Hz to 10kHz

Room Temperature-55 to +100°C@100Vdc

Calibration supplied:

CS-SENS-0100

NIST Traceable Amplitude Calibration at 100Hz

Supplied accessories:

AC-A03663¼-28 to ¼-28 mounting stud

Optional accessories:

CS-FREQ-0100
AC-D03664
AC-D03665

NIST Traceable Amplitude Calibration from 20Hz to ±3dB Frequency Response Limit¼-28 to M5 mounting stud¼-28 to M6 mounting stud

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves the rightto make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding the suitabilityof its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or circuit andspecifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different applications.
All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does not convey any licenseunder its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 8032-01 Rev B

07/2017

Page 2

MODEL 8032-01 ACCELEROMETER

40

35

35

30

30

25

25

20

20

15

15

1010

5

50

0-5-5-10-10-15-15-20-20-25-25-30-30-35-35-40
101001000-40100000

10000

Frequency (Hz)%dB

ORDERING INFORMATIONPART NUMBERING

Model Number+Range+Cable Length8032-01-GGGG-ZZZlll
l_____Cable Length (120 is 120 inches)
l___________ Range (0050 is 50g)
Example: 8032-01-0050-120
Model 8032-01, 50g, 120inch (10ft) cable length

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 8032-01 Rev B

07/2017

Page 3

Sensitivity Deviation (dB)

Sensitivity Deviation (%)

Typical Frequency Response40