PCI-DIO96
Digital I/O Board
User's Guide

Document Revision 6A, May, 2009
© Copyright 2009, Measurement Computing Corporation

Trademark and Copyright Information
Measurement Computing Corporation, InstaCal, Universal Library, and the Measurement Computing logo areeither trademarks or registered trademarks of Measurement Computing Corporation. Refer to the Copyrights &Trademarks section on mccdaq.com/legal for more information about Measurement Computing trademarks.
Other product and company names mentioned herein are trademarks or trade names of their respective
companies.
© 2009 Measurement Computing Corporation. All rights reserved. No part of this publication may be
reproduced, stored in a retrieval system, or transmitted, in any form by any means, electronic, mechanical, by
photocopying, recording, or otherwise without the prior written permission of Measurement Computing
Corporation.
NoticeMeasurement Computing Corporation does not authorize any Measurement Computing Corporation product for
use in life support systems and/or devices without prior written consent from Measurement Computing
Corporation. Life support devices/systems are devices or systems that, a) are intended for surgical implantation
into the body, or b) support or sustain life and whose failure to perform can be reasonably expected to result ininjury. Measurement Computing Corporation products are not designed with the components required, and are
not subject to the testing required to ensure a level of reliability suitable for the treatment and diagnosis ofpeople.

HM PCI-DIO96.doc

3

4

Table of Contents
Preface

About this User's Guide .......................................................................................................................6
What you will learn from this user's guide ......................................................................................................... 6
Conventions in this user's guide ........................................................................................................................................ 6

Where to find more information ......................................................................................................................... 6
Chapter 1

Introducing the PCI-DIO96 board ........................................................................................................7
Overview: PCI-DIO96 features .......................................................................................................................... 7
Software features ................................................................................................................................................ 7
Chapter 2

Installing the PCI-DIO96 .......................................................................................................................8
What comes with your PCI-DIO96 shipment? ................................................................................................... 8
Hardware .......................................................................................................................................................................... 8
Additional documentation ................................................................................................................................................. 8
Optional components ........................................................................................................................................................ 8

Unpacking the PCI-DIO96 ................................................................................................................................. 9
Installing the software ........................................................................................................................................ 9
Installing the PCI-DIO96 ................................................................................................................................... 9
Configuring the PCI-DIO96 ............................................................................................................................... 9Connecting the board for I/O operations .......................................................................................................... 10
Connectors, cables – main I/O connector ........................................................................................................................10
Pin out – main I/O connector ...........................................................................................................................................11
Field wiring and signal termination accessories ..............................................................................................................13
Chapter 3

Functional Details ...............................................................................................................................14
PCI-DIO96 block diagram ............................................................................................................................... 14
Signal level control ........................................................................................................................................... 15
CIO-ERB24 and SSR-RACK24 daisy chain configuration ............................................................................. 17
Chapter 4

Specifications ......................................................................................................................................18
Power consumption .......................................................................................................................................... 18
Digital input/output .......................................................................................................................................... 18
Counters............................................................................................................................................................ 18
Interrupts .......................................................................................................................................................... 19
Crystal oscillator ............................................................................................................................................... 19
Environmental .................................................................................................................................................. 19
Mechanical ....................................................................................................................................................... 19
Main connector and pin out .............................................................................................................................. 20
Declaration of Conformity ..................................................................................................................225

Preface

About this User's Guide
What you will learn from this user's guideThis user's guide explains how to install, configure, and use the PCI-DIO96 so that you get the most out of its
digital I/O features.
This user's guide also refers you to related documents available on our web site, and to technical support
resources.

Conventions in this user's guide
For more information on …
Text presented in a box signifies additional information and helpful hints related to the subject matter you arereading.
Caution! Shaded caution statements present information to help you avoid injuring yourself and others,
damaging your hardware, or losing your data.
< : >Angle brackets that enclose numbers separated by a colon signify a range of numbers, such as those assigned
to registers, bit settings, etc.

bold text

Bold text is used for the names of objects on the screen, such as buttons, text boxes, and check boxes. For
example:1. Insert the disk or CD and click the OK button.

italic text

Italic text is used for the names of manuals and help topic titles, and to emphasize a word or phrase. For
example:The InstaCal installation procedure is explained in the Quick Start Guide.
Never touch the exposed pins or circuit connections on the board.

Where to find more information
For additional information relevant to the operation of your hardware, refer to the Documents subdirectory
where you installed the MCC DAQ software (C:\Program Files\Measurement Computing\DAQ by default), or
search for your device on our website at www.mccdaq.com.
If you need to program at the register level in your application, refer to the Register Map for the PCI-DIO96.
This document is available at www.mccdaq.com/registermaps/RegMapPCI-DIO96.pdf.

6

Chapter 1

Introducing the PCI-DIO96 board
Overview: PCI-DIO96 features
This manual provides installation and programming information for the PCI-DIO96 digital I/O board.
The PCI-DIO96 is a 96-bit line digital I/O board. The board provides the 96 bits in 24-bit groups based on the
82C55 specification. Each 24-bit group is divided into three eight-bit ports.
The ports are labeled on the board as PORT #A, PORT #B, and PORT #C, where # is from 1 to 4. PORT #C
can be further split into two four-bit nibbles — PORT #C-HI and PORT #C -LO.
Each of these ports may be individually programmed as input or output.

Software features
For information on the features of InstaCal and the other software included with your PCI-DIO96, refer to the
Quick Start Guide that shipped with your device. The Quick Start Guide is also available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
Check www.mccdaq.com/download.htm for the latest software version.

7

Chapter 2

Installing the PCI-DIO96
What comes with your PCI-DIO96 shipment?
Hardware

PCI-DIO96

Additional documentation
In addition to this hardware user's guide, you should also receive the Quick Start Guide (available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf). This booklet supplies a brief description of
the software you received with your PCI-DIO96 and information regarding installation of that software. Please
read this booklet completely before installing any software or hardware.

Optional components


CablesC100FF-x



Signal termination and conditioning accessories
MCC provides signal termination and signal conditioning products for use with the PCI-DIO96. Refer toField wiring, signal termination and conditioning on page 13 for a list of compatible accessory products.
8

PCI-DIO96 User's Guide

Installing the PCI-DIO96

Unpacking the PCI-DIO96
As with any electronic device, you should take care while handling to avoid damage from static
electricity. Before removing the PCI-DIO96 from its packaging, ground yourself using a wrist strap or bysimply touching the computer chassis or other grounded object to eliminate any stored static charge.
If any components are missing or damaged, notify Measurement Computing Corporation immediately byphone, fax, or e-mail:

Phone: 508-946-5100 and follow the instructions for reaching Tech Support.
Fax: 508-946-9500 to the attention of Tech Support
Email: techsupport@mccdaq.com

Installing the software
Refer to the Quick Start Guide for instructions on installing the software on the Measurement Computing Data
Acquisition Software CD. This booklet is available in PDF at www.mccdaq.com/PDFmanuals/DAQ-SoftwareQuick-Start.pdf.

Installing the PCI-DIO96
The PCI-DIO96 board is completely plug-and-play, with no switches or jumpers to set. Configuration iscontrolled by your system's BIOS. To install your board, follow the steps below.
Install the MCC DAQ software before you install your board
The driver needed to run your board is installed with the MCC DAQ software. Therefore, you need to install
the MCC DAQ software before you install your board. Refer to the Quick Start Guide for instructions on
installing the software.
1.

Turn your computer off, open it up, and insert your board into an available PCI slot.

2.

Close your computer and turn it on.
If you are using an operating system with support for plug-and-play (such as Windows 2000 or
Windows XP), a dialog box opens as the system loads, indicating that new hardware has been detected.
The information file for this board should have already been loaded onto your PC when you installed the
Measurement Computing Data Acquisition Software CD supplied with your board, and should be detected
automatically by Windows. If you have not installed this software, cancel the dialog and install it now

3.

To test your installation and configure your board, run the InstaCal utility installed in the previous section.
Refer to the Quick Start Guide that came with your board for information on how to initially set up andload InstaCal.

Configuring the PCI-DIO96
All hardware configuration options on the PCI-DIO96 are software controlled. You can select some of theconfiguration options using InstaCal, such as digital channel configuration (input or output). Once selected, any
program that uses the Universal Library will initialize the hardware according to these selections.

9

PCI-DIO96 User's Guide

Installing the PCI-DIO96

Connecting the board for I/O operations
Connectors, cables – main I/O connector
The table below lists the board connectors, applicable cable and compatible accessory products for the PCIDIO96.
Board connectors, cables, and accessory equipment
Connector typeCompatibility
Compatible cables
Compatible accessory products

100 pin high-density Robinson-Nugent.
 Pinout identical to PCI-DIO96H
 Compatible with CIO-DIO96H using the C100FF-x cable
C100FF-x
SCB50
CIO-MINI50
CIO-TERM100
CIO-SPADE50
CIO-ERB24
CIO-ERB48
SSR-RACK24
SSR-RACK48

The PCI-DIO96 board has a 100-pin, high-density Robinson-Nugent male connector. Connector pinouts arelisted on page 11.
You can use the optional C100FF-x cable to split the 100 I/O lines into two, 50-pin cables. Board connector
pins 1 to 50 are mapped directly to pins 1 to 50 on the C100FF-x cable's first 50-pin connector. Board
connector pins 51 to 100 are mapped directly to pins 1 to 50 on the C100FF-x cable's second 50-pin connector(pin 51 is mapped to pin 1, and pin 100 is mapped to pin 50.) A sample C100FF-x cable configuration is shownin Figure 2 on page 12.
Information on signal connections
General information regarding signal connection and configuration is available in the Guide to Signal
Connections. This document is available on our web site at www.mccdaq.com/signals/signals.pdf.
Caution! When connecting a cable to the board's I/O connector, make sure that the arrow indicating pin 1
on the board connector lines up with the arrow indicating pin 1 on the cable connector. Incorrectly
connected cables can damage the board and the I/O controller.

10

PCI-DIO96 User's Guide

Installing the PCI-DIO96

Pin out – main I/O connector
Main connector pin out
Signal name
GND
+5V
THIRDPORTC Bit 0
THIRDPORTC Bit 1
THIRDPORTC Bit 2
THIRDPORTC Bit 3
THIRDPORTC Bit 4
THIRDPORTC Bit 5
THIRDPORTC Bit 6
THIRDPORTC Bit 7
THIRDPORTB Bit 0
THIRDPORTB Bit 1
THIRDPORTB Bit 2
THIRDPORTB Bit 3
THIRDPORTB Bit 4
THIRDPORTB Bit 5
THIRDPORTB Bit 6
THIRDPORTB Bit 7
THIRDPORTA Bit 0
THIRDPORTA Bit 1
THIRDPORTA Bit 2
THIRDPORTA Bit 3
THIRDPORTA Bit 4
THIRDPORTA Bit 5
THIRDPORTA Bit 6
THIRDPORTA Bit 7
FOURTHPORTC Bit 0
FOURTHPORTC Bit 1
FOURTHPORTC Bit 2
FOURTHPORTC Bit 3
FOURTHPORTC Bit 4
FOURTHPORTC Bit 5
FOURTHPORTC Bit 6
FOURTHPORTC Bit 7
FOURTHPORTB Bit 0
FOURTHPORTB Bit 1
FOURTHPORTB Bit 2
FOURTHPORTB Bit 3
FOURTHPORTB Bit 4
FOURTHPORTB Bit 5
FOURTHPORTB Bit 6
FOURTHPORTB Bit 7
FOURTHPORTA Bit 0
FOURTHPORTA Bit 1
FOURTHPORTA Bit 2
FOURTHPORTA Bit 3
FOURTHPORTA Bit 4
FOURTHPORTA Bit 5
FOURTHPORTA Bit 6
FOURTHPORTA Bit 7

Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

PCI slot

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

↓

11

Signal name
GND
+5V
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
SECONDPORTC Bit 0
SECONDPORTC Bit 1
SECONDPORTC Bit 2
SECONDPORTC Bit 3
SECONDPORTC Bit 4
SECONDPORTC Bit 5
SECONDPORTC Bit 6
SECONDPORTC Bit 7
SECONDPORTB Bit 0
SECONDPORTB Bit 1
SECONDPORTB Bit 2
SECONDPORTB Bit 3
SECONDPORTB Bit 4
SECONDPORTB Bit 5
SECONDPORTB Bit 6
SECONDPORTB Bit 7
SECONDPORTA Bit 0
SECONDPORTA Bit 1
SECONDPORTA Bit 2
SECONDPORTA Bit 3
SECONDPORTA Bit 4
SECONDPORTA Bit 5
SECONDPORTA Bit 6
SECONDPORTA Bit 7

PCI-DIO96 User's Guide

Installing the PCI-DIO96

Cable is labeled“Pins 1-50”.

50

49

50

1

2

Key

100

The red stripe
identifies pin # 1

99 100

1

51

Key

51 52

Cable is labeled“Pins 51-100”.

The red stripe
identifies pin # 51

Figure 1. C100FF-x cable

PCI-DIO96
100-pin I/O connector

C100FF-x

Analog I/O
pins 1 to 50
IN
Analog signal conditioning or
50-pin screw terminal board
Digital I/O
pins 51 to 100
IN
Digital signal conditioning or
50-pin screw terminal board
Figure 2. C100FF-x cable configuration

Details on these cables are available on our web site at www.mccdaq.com/products/accessories.aspx.

12

PCI-DIO96 User's Guide

Installing the PCI-DIO96

Field wiring and signal termination accessories
You can use the following screw terminal boards to terminate field signals and route them into the PCI-DIO96
using the C100FF-x cable.


SCB-50 – 50 conductor, shielded signal connection/screw terminal box provides two independent 50-pin
connections.
CIO-MINI50 – 50-pin screw terminal board.
CIO-TERM100 – 100-pin screw terminal board (daisy-chained 50-pin IDC connectors).
CIO-SPADE50 — 16" X 4" termination panel which mates with both 37-pin and 50-pin connectors.

Details on these products are available on our web site atwww.mccdaq.com/products/screw_terminal_bnc.aspx.
CIO-ERB24 – 24 Form C relays, 6 Amp relay accessory board for digital signal conditioning.
CIO-ERB48 – 48 Form C relays, 6 Amp, relay, 50-pin accessory board for digital signal conditioning.
SSR-RACK24 – 24-channel, solid-state relay mounting rack for digital signal conditioning.
SSR-RACK48 – 48-channel, solid-state relay mounting rack with quad-format modules.

Details on these products are available on our web site at www.mccdaq.com/products/signal_conditioning.aspx.
For additional information about digital interfacing…
Detailed information regarding digital interfacing is contained in MCC's Guide to Signal Connections. This
document is available on our web site at www.mccdaq.com/signals/signals.pdf.

13

Chapter 3

Functional Details
PCI-DIO96 block diagram
PCI-DIO96 functions are illustrated in the block diagram shown here.

FOURTHPORTA(7:0)

FOURTHPORTA

FOURTHPORTB(7:0)

FOURTHPORTBFOURTHPORTCH(3:0)
FOURTHPORTCL(3:0)

FOURTHPORTCH
FOURTHPORTCL
82C55

Control

DIO Group 3

THIRDPORTA(7:0)

THIRDPORTA

THIRDPORTB(7:0)

THIRDPORTBTHIRDPORTCH(3:0)
THIRDPORTCL(3:0)

THIRDPORTCH
THIRDPORTCL

Control

DIO Group 2

82C55

Controller FPGA and Logic
DIO Group 1
SECONDPORTASECONDPORTB(7:0)

SECONDPORTBSECONDPORTCH(3:0)
SECONDPORTCL(3:0)

SECONDPORTCH
SECONDPORTCL

ControlBus

Control

SECONDPORTA(7:0)

Control
Registers

Counters

Decode/Status

82C54

82C55

DIO Group 0

FIRSTPORTB(7:0)

FIRSTPORTBFIRSTPORTCH(3:0)
FIRSTPORTCL(3:0)

FIRSTPORTCH
FIRSTPORTCL

INT.
Control

FIRSTPORTA(7:0)

FIRSTPORTA

Local Bus

82C55

Boot
EEPROM

PCI
Controller

BADR3

PLX-9052

PCI BUS (5V, 32-bit, 33 MHz)

Figure 3. PCI-DIO96 functional block diagram

14

PCI-DIO96 User's Guide

Functional Details

Signal level control
All the digital inputs on the PCI-DIO96 are 8255 CMOS TTL. The PCI-DIO96 output signals are 8255 CMOS.
Note
The 82C55 digital I/O chip initializes all ports as inputs on power- up and reset. A TTL input is a highimpedance input. If you connect another TTL input device to the 82C55 it could be turned ON or OFF every
time the 82C55 is reset. Remember, the 82C55 is reset to the INPUT mode.
On power up and reset, all I/O bits are set to input mode. If you are using the board to control items that mustbe OFF on reset, install pull-down resistors. The PCI-DIO96 has open locations where you can install Single
Inline Packages (SIP) resistor networks in either pull-up or pull-down configurations (refer to Figure 4).

Figure 4. Open locations for SIP packs

You can install pull-up and pull-down resistor SIP packs at PORT #A, PORT #B, and PORT #C. These ports
are adjacent to the I/O connector. When installed, the SIP establishes either a high or low logic level at each of
the eight I/O lines on the port.

15

PCI-DIO96 User's Guide

Functional Details

To safeguard against unwanted signal levels, the devices being controlled by PCI-DIO96 series boards should
be tied low or high as required by a 2.2K resistor. In a 2.2K eight-resistor SIP pack, one side of all of theresistors is connected to a single common point and brought out to a pin. The common line, usually marked
with a dot or line, is at one end of the SIP. The remaining resistor ends are brought out to the other eight pins
(refer to Figure 5).
2.2KOhm SIP
Dot indicates thecommon line

(LO or HI)

I/O Lines

Figure 5. Eight-resistor SIP schematic

Figure 6 shows a schematic of an SIP installed in both the pull-up and pull-down positions. Each port provides
10 holes in a line. The end labeled HI connects to +5V. The end marked LO connects to GND. The eight holes
in the middle (n0 –n7) connect to the eight lines of the Port, A, B or C.
+ 5 VDC+ 5 VDC

HI

COM

n6

n6

User Connector
Digital I/O Lines

n7

n4
n3
n2

n = A, B, or C

HI

n7

n5

Digital
I/O
Port

Dot

n1

2.2 K SIP

User Connector
Digital I/O Lines

2.2 K SIP

n5

Digital
I/O
Port

n4
n3
n2

n = A, B, or C

n1

n0

n0

LO
(GND)

LO
(GND)

2.2 K SIP installed for pull-up

COM

Dot

2.2 K SIP installed for pull-downFigure 6. Pull-up and pull-down resistor SIPs schematic

To pull-up lines, orient the SIP with the common pin (dot) toward the HI end; to pull-down, install the resistor
with the common pin in the LO hole.
When installing pull-up and pull-down resistor SIP packs, we recommend using 2.2K, eight-resistor Single
Inline Packages (MCC part number SP-K2.29C). Use a different value only if necessary.
Information on signal connections
General information regarding signal connection and configuration is available in the Guide to Signal
Connections. This document is available on our web site at www.mccdaq.com/signals/signals.pdf.

16

PCI-DIO96 User's Guide

Functional Details

CIO-ERB24 and SSR-RACK24 daisy chain configuration
Many relay and solid-state relay (SSR) racks provide only 24-bits of digital I/O. You can configure the
CIO-ERB24 relay output board and SSR-RACK24 I/O module rack in a daisy chain configuration to use all of
the digital I/O bits provided by the PCI-DIO96 board. An example of the daisy chain configuration scheme isshown below.
The PCI-DIO96 board provides digital I/O in a group of 96 bits. Each of the C100FF-x cable's 50-pin
connectors provides 48-bits. To use all of the board's 96 digital I/O bits to control relays and/or SSRs, configure
the daisy chain as shown in Figure 7.

C100FF-x Cable

CIO-ERB24 IN
orOUT
SSR-Rack24

CIO-ERB24 IN
orOUT
SSR-Rack24
CIO-ERB24 IN
orOUT
SSR-Rack24

CIO-ERB24 IN
orOUT
SSR-Rack24

Figure 7. PCI-DIO96H to C100FF-x to relay rack daisy chain cabling

The PCI-DIO96 board's 24 digital I/O bits on pins 1-24 (base address +0 through +3) control the first relay
board on the chain. The 24 DIO bits on pins 25-50 control the second relay/SSR board on the daisy chain andso on, for up to 100 pins.

17

Chapter 4

Specifications
Typical for 25 °C unless otherwise specified.
Specifications in italic text are guaranteed by design.

Power consumption
Table 1. Power consumption specifications+5V quiescent

150 mA max

Digital input/output
Table 2. Digital input/output specifications
Digital type
Number of I/O
Configuration per 82C55

Output highOutput low
Input highInput low
Power-up / reset state
Pull-up/pull-down resistors

Four 82C55
96
 2 banks of 8 and 2 banks of 4 or
 3 banks of 8 or
 2 banks of 8 with handshake
3.0 volts min @ -2.5 mA
0.4 volts max @ 2.5 mA
2.0 volts min, 5.5 volts absolute max
0.8 volts max, -0.5 volts absolute min
Input mode (high impedance)
User installed. Dual footprint allows pull-up or pull-down configuration

Counters
Table 3. Counters specifications
Counter type
Configuration
Counter 1

Counter 2

Counter 3 - Not used

82C54
3 counters, 16 bits each
Source:2 MHz (xtal /8)
Gate:
Tied to +5V
Output:Selectable Interrupt source
Source:Counter 1 OUT
Gate:
Tied to +5V
Output:Selectable interrupt source
SourceGate
Output18

PCI-DIO96 User's Guide

Specifications

Interrupts
The interrupt control registers allow the four 82C55 devices and the 8254 counter timer to be used as interrupt
sources.
Table 4. Interrupt specifications
Interrupt

INTA# - mapped to IRQn via PCI BIOS at boot-time

PCI Interrupt enable

Programmable through PLX9052 INTCSR register (INTCSR 4Ch)

Interrupt polarity

High or low level. Programmable through PLX9052
Rising or falling edge. Programmable through PLX9052
82C55 in Mode 1 or Mode 2 interrupt configuration:
 First Port C0
 First Port C3
 Second Port C0
 Second Port C3
 Third Port C0
 Third Port C3
 Fourth Port C0
 Fourth Port C3
Note: Any interrupt source above may be individually enabled.

Interrupt sources

82C54 Counter Counter 1 OUT
 Counter 2 OUT
Note: Counter 1 and 2 interrupts are exclusive. Only one counter may beenabled as an interrupt source at one time.

Crystal oscillator
Table 5. Crystal oscillator specifications
Oscillator type
Frequency
Frequency stability

AT-cut crystal
16 MHz±100 ppm

Environmental
Table 6. Environmental specifications
Operating temperature range
Storage temperature range
Humidity

0 to 70 °C-40 to 70 °C
0 to 95% non-condensing

Mechanical
Table 7. Mechanical specifications
Card dimensions

PCI short card: 136.0 mm (L) x 100.6 mm (W) x11.00 mm (H)19

PCI-DIO96 User's Guide

Specifications

Main connector and pin out
Table 8. Board connectors, cables, and accessory equipment
Connector typeCompatibility
Compatible cables
Compatible accessory products

100 pin high-density Robinson-Nugent.
 Pinout identical to PCI-DIO96H.
 Compatible with CIO-DIO96H using C100FF-x.
C100FF-x
SCB50
CIO-MINI50
CIO-TERM100
CIO-SPADE50
CIO-ERB24
CIO-ERB48
SSR-RACK24
SSR-RACK48

20

PCI-DIO96 User's Guide

Specifications

Table 9. Main connector pin out
Signal name
GND
+5V
THIRDPORTC Bit 0
THIRDPORTC Bit 1
THIRDPORTC Bit 2
THIRDPORTC Bit 3
THIRDPORTC Bit 4
THIRDPORTC Bit 5
THIRDPORTC Bit 6
THIRDPORTC Bit 7
THIRDPORTB Bit 0
THIRDPORTB Bit 1
THIRDPORTB Bit 2
THIRDPORTB Bit 3
THIRDPORTB Bit 4
THIRDPORTB Bit 5
THIRDPORTB Bit 6
THIRDPORTB Bit 7
THIRDPORTA Bit 0
THIRDPORTA Bit 1
THIRDPORTA Bit 2
THIRDPORTA Bit 3
THIRDPORTA Bit 4
THIRDPORTA Bit 5
THIRDPORTA Bit 6
THIRDPORTA Bit 7
FOURTHPORTC Bit 0
FOURTHPORTC Bit 1
FOURTHPORTC Bit 2
FOURTHPORTC Bit 3
FOURTHPORTC Bit 4
FOURTHPORTC Bit 5
FOURTHPORTC Bit 6
FOURTHPORTC Bit 7
FOURTHPORTB Bit 0
FOURTHPORTB Bit 1
FOURTHPORTB Bit 2
FOURTHPORTB Bit 3
FOURTHPORTB Bit 4
FOURTHPORTB Bit 5
FOURTHPORTB Bit 6
FOURTHPORTB Bit 7
FOURTHPORTA Bit 0
FOURTHPORTA Bit 1
FOURTHPORTA Bit 2
FOURTHPORTA Bit 3
FOURTHPORTA Bit 4
FOURTHPORTA Bit 5
FOURTHPORTA Bit 6
FOURTHPORTA Bit 7

Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

PCI slot ↓21

Signal name
GND
+5V
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
SECONDPORTC Bit 0
SECONDPORTC Bit 1
SECONDPORTC Bit 2
SECONDPORTC Bit 3
SECONDPORTC Bit 4
SECONDPORTC Bit 5
SECONDPORTC Bit 6
SECONDPORTC Bit 7
SECONDPORTB Bit 0
SECONDPORTB Bit 1
SECONDPORTB Bit 2
SECONDPORTB Bit 3
SECONDPORTB Bit 4
SECONDPORTB Bit 5
SECONDPORTB Bit 6
SECONDPORTB Bit 7
SECONDPORTA Bit 0
SECONDPORTA Bit 1
SECONDPORTA Bit 2
SECONDPORTA Bit 3
SECONDPORTA Bit 4
SECONDPORTA Bit 5
SECONDPORTA Bit 6
SECONDPORTA Bit 7

Declaration of Conformity
Manufacturer:
Address:

Category:

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, MA 02766
USAElectrical equipment for measurement, control and laboratory use.

Measurement Computing Corporation declares under sole responsibility that the product

PCI-DIO96
to which this declaration relates is in conformity with the relevant provisions of the following standards orother documents:
EC EMC Directive 2004/108/EC: General Requirements, EN 61326-1:2006 (IEC 61326-1:2005).
Emissions:EN 55011 (2007) / CISPR 11(2003): Radiated emissions: Group 1, Class A
EN 55011 (2007) / CISPR 11(2003): Conducted emissions: Group 1, Class A

Immunity: EN 61326-1:2006, Table 3.







IEC 61000-4-2 (2001): Electrostatic Discharge immunity.
IEC 61000-4-3 (2002): Radiated Electromagnetic Field immunity.
IEC 61000-4-4 (2004): Electric Fast Transient Burst Immunity.
IEC 61000-4-5 (2001): Surge Immunity.
IEC 61000-4-6 (2003): Radio Frequency Common Mode Immunity.
IEC 61000-4-11 (2004): Voltage Interrupts.

To maintain compliance to the standards of this declaration, the following conditions must be met.






The host computer, peripheral equipment, power sources, and expansion hardware must be CE
compliant.
All I/O cables must be shielded, with the shields connected to ground.
I/O cables must be less than 3 meters (9.75 feet) in length.
The host computer must be properly grounded.
Equipment must be operated in a controlled electromagnetic environment as defined by Standards EN
61326-1:2006, or IEC 61326-1:2005.

Declaration of Conformity based on tests conducted by Chomerics Test Services, Woburn, MA 01801, USA in
September, 2001. Test records are outlined in Chomerics Test Report #EMI3053.01. Further testing was
conducted by Chomerics Test Services, Woburn, MA. 01801, USA in January, 2009. Test records are outlined
in Chomerics Test report #EMI5243.09.
We hereby declare that the equipment specified conforms to the above Directives and Standards.

Carl Haapaoja, Director of Quality Assurance

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, Massachusetts 02766(508) 946-5100
Fax: (508) 946-9500
E-mail: info@mccdaq.com
www.mccdaq.com