PCI-DIO96H
Digital Input/Output
User's Guide

Document Revision 3A, May, 2009
© Copyright 2009, Measurement Computing Corporation

Trademark and Copyright Information
Measurement Computing Corporation, InstaCal, Universal Library, and the Measurement Computing logo areeither trademarks or registered trademarks of Measurement Computing Corporation. Refer to the Copyrights &Trademarks section on mccdaq.com/legal for more information about Measurement Computing trademarks.
Other product and company names mentioned herein are trademarks or trade names of their respective
companies.
© 2009 Measurement Computing Corporation. All rights reserved. No part of this publication may be
reproduced, stored in a retrieval system, or transmitted, in any form by any means, electronic, mechanical, by
photocopying, recording, or otherwise without the prior written permission of Measurement Computing
Corporation.
NoticeMeasurement Computing Corporation does not authorize any Measurement Computing Corporation product for
use in life support systems and/or devices without prior written consent from Measurement Computing
Corporation. Life support devices/systems are devices or systems that, a) are intended for surgical implantation
into the body, or b) support or sustain life and whose failure to perform can be reasonably expected to result ininjury. Measurement Computing Corporation products are not designed with the components required, and are
not subject to the testing required to ensure a level of reliability suitable for the treatment and diagnosis ofpeople.

HM PCI-DIO96H.doc

3

4

Table of Contents
Preface

About this User's Guide .......................................................................................................................6
What you will learn from this user's guide ......................................................................................................... 6
Conventions in this user's guide ......................................................................................................................... 6
Where to find more information ......................................................................................................................... 6
Chapter 1

Introducing the PCI-DIO96H .................................................................................................................7
Overview: PCI-DIO96H features ....................................................................................................................... 7
Software features ................................................................................................................................................ 7
PCI-DIO96H block diagram ............................................................................................................................... 8
Chapter 2

Installing the PCI-DIO96H .....................................................................................................................9
What comes with your PCI-DIO96H shipment? ................................................................................................ 9
Hardware .......................................................................................................................................................................... 9
Additional documentation ................................................................................................................................................. 9
Optional components ........................................................................................................................................................ 9

Unpacking the PCI-DIO96H ............................................................................................................................ 10
Installing the software ...................................................................................................................................... 10
Installing the hardware ..................................................................................................................................... 10
Connecting the board for I/O operations .......................................................................................................... 11
Connectors, cables – main I/O connector.........................................................................................................................11
Pin out – main I/O connector ...........................................................................................................................................12
Cabling.............................................................................................................................................................................13
Field wiring and signal termination accessories...............................................................................................................14
Chapter 3

Functional Details ...............................................................................................................................15
CIO-ERB24 and SSR-RACK24 daisy chain configuration .............................................................................. 15
82C55 emulation .............................................................................................................................................. 15
Pull-up and pull-down resistors .......................................................................................................................................16
Chapter 4

Specifications ......................................................................................................................................17
Digital input / output......................................................................................................................................... 17Power Consumption ......................................................................................................................................... 17
Environmental .................................................................................................................................................. 17
Mechanical ....................................................................................................................................................... 17
Main connector and pin out .............................................................................................................................. 18
Declaration of Conformity ..................................................................................................................20

5

Preface

About this User's Guide
What you will learn from this user's guideThis user's guide explains how to install, configure, and use the PCI-DIO96H so that you get the most out of its
digital I/O features.
This user's guide also refers you to related documents available on our web site, and to technical support
resources.

Conventions in this user's guide
For more information on …
Text presented in a box signifies additional information and helpful hints related to the subject matter you arereading.
Caution! Shaded caution statements present information to help you avoid injuring yourself and others,
damaging your hardware, or losing your data.
< : >Angle brackets that enclose numbers separated by a colon signify a range of numbers, such as those assigned
to registers, bit settings, etc.

bold text

Bold text is used for the names of objects on the screen, such as buttons, text boxes, and check boxes. For
example:1. Insert the disk or CD and click the OK button.

italic text

Italic text is used for the names of manuals and help topic titles, and to emphasize a word or phrase. For
example:The InstaCal® installation procedure is explained in the Quick Start Guide.
Never touch the exposed pins or circuit connections on the board.

Where to find more information
For additional information relevant to the operation of your hardware, refer to the Documents subdirectory
where you installed the MCC DAQ software (C:\Program Files\Measurement Computing\DAQ by default), or
search for your device on our website at www.mccdaq.com.
If you need to program at the register level in your application, you can find more information in the Register
Map for the PCI-DIO48H and PCI-DIO96H. This document is available atwww.mccdaq.com/registermaps/RegMapPCI-DIOxxH.pdf.

6

Chapter 1

Introducing the PCI-DIO96H
Overview: PCI-DIO96H features
This manual explains how to install and use the PCI-DIO96H board. The PCI-DIO96H is a high-density, logiclevel digital I/O board designed for the PCI-bus.
The PCI-DIO96H provides 96-bits of digital I/O. The I/O is organized into four 24-bit groups based on an
82C55 mode 0 emulation. Each 24-bit group is divided into three eight-bit ports labeled PORTA, PORTB and
PORTC. PORTC can be split into two four-bit nibbles — Port C-HI and Port C-LO. Each of these ports may be
individually programmed as input or output.
All digital inputs are LSTTL. The output signals are buffered high output drive TTL. The digital output drivers
are 74S244 chips that can sink 64 mA and source 15 mA. The input buffers are 74LS373 chips and havestandard high input impedance of the 74LS series devices.
On power up and reset, all I/O bits are set to input mode. If you are using the board to control items that must be
OFF on reset, install pull-down resistors. Each board is equipped with open locations where you can install SIP
resistor networks for either pull-up or pull-down.
The PCI-DIO96H board is completely plug-and-play, with no jumpers or switches to set. All board addresses
are set by the board's plug-and-play software. Board configuration is controlled by your system's BIOS.

Software features
For information on the features of InstaCal and the other software included with your PCI-DIO96H, refer to the
Quick Start Guide that shipped with your device. The Quick Start Guide is also available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.
Check www.mccdaq.com/download.htm for the latest software version or versions of the software supported
under less commonly used operating systems.

7

PCI-DIO96H User's Guide

Introducing the PCI-DIO96H

PCI-DIO96H block diagram
PCI-DIO96H functions are illustrated in the block diagram shown here.
HIGH DRIVE
FOURTHPORTFOURTHPORTA

FOURTHPORTB(7:0)

FOURTHPORTBFOURTHPORTCH(3:0)
FOURTHPORTCL(3:0)

FOURTHPORTCH
FOURTHPORTCL

PCI-DIO96H
Block Diagram

Control

FOURTHPORTA(7:0)

THIRDPORTA(7:0)

THIRDPORTA

THIRDPORTB(7:0)

THIRDPORTBTHIRDPORTCH(3:0)
THIRDPORTCL(3:0)

THIRDPORTCH
THIRDPORTCL

Control

HIGH DRIVE
THIRDPORTHIGH DRIVESECONDPORTSECONDPORTASECONDPORTB(7:0)

SECONDPORTBSECONDPORTCH(3:0)
SECONDPORTCL(3:0)

SECONDPORTCH
SECONDPORTCL

ControlBus

Control

SECONDPORTA(7:0)

Controller FPGA and Logic
Control
Registers

Decode/Status

HIGH DRIVE
FIRSTPORTFIRSTPORTAFIRSTPORTB(7:0)

FIRSTPORTBFIRSTPORTCH(3:0)
FIRSTPORTCL(3:0)

FIRSTPORTCH
FIRSTPORTCLBus
TimingControl

FIRSTPORTA(7:0)

LOCAL BUS

Boot
EEPROM

PCI
Controller

PCI BUS (5V, 32-BIT, 33MHZ)

8

BADR2

Chapter 2

Installing the PCI-DIO96H
What comes with your PCI-DIO96H shipment?
The following items are shipped with the PCI-DIO96H.

Hardware

PCI-DIO96H board

Additional documentation
In addition to this hardware user's guide, you should also receive the Quick Start Guide (available in PDF at
www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf). This booklet supplies a brief description of
the software you received with your MCC Hardware and information regarding installation of that software.
Please read this booklet completely before installing any software or hardware.

Optional components


C100FF-x cable

Signal termination and conditioning accessories
MCC provides signal termination products for use with the PCI-DIO96H. Refer to Field wiring, signal
termination and conditioning on page 14 for a complete list of compatible accessory products.

9

PCI-DIO96H User's Guide

Installing the PCI-DIO96H

Unpacking the PCI-DIO96H
As with any electronic device, you should take care while handling to avoid damage from static
electricity. Before removing the PCI-DIO96H from its packaging, ground yourself using a wrist strap or bysimply touching the computer chassis or other grounded object to eliminate any stored static charge.
If any components are missing or damaged, notify Measurement Computing Corporation immediately byphone, fax, or e-mail:

Phone: 508-946-5100 and follow the instructions for reaching Tech Support.
Fax: 508-946-9500 to the attention of Tech Support
Email: techsupport@mccdaq.com

Installing the software
Install the software included with your board before you install the hardware. Installing the software firstensures that the information required for proper board detection is installed and available at boot up.
Refer to the Quick Start Guide for instructions on installing the software on the Measurement Computing Data
Acquisition Software CD. This booklet is available in PDF at www.mccdaq.com/PDFmanuals/DAQ-SoftwareQuick-Start.pdf.

Installing the hardware
The PCI-DIO96H board is completely plug-and-play. There are no switches or jumpers to set on the board.
Configuration is controlled by your system's BIOS. To install your board, follow the steps below.
Install the MCC DAQ software before you install your board
The driver needed to run your board is installed with the MCC DAQ software. Therefore, you need to install the
MCC DAQ software before you install your board. Refer to the Quick Start Guide for instructions on installingthe software.
1.

Turn your computer off, open it up, and insert your board into an available PCI slot.

2.

Close your computer and turn it on.
If you are using an operating system with support for plug-and-play (such as Windows 2000 or Windows
XP), a dialog box pops up as the system loads indicating that new hardware has been detected. If theinformation file for this board is not already loaded onto your PC, you will be prompted for the disk
containing this file. The MCC DAQ software contains this file. If required, insert the Measurement
Computing Data Acquisition Software CD and click OK.

3.

To test your installation and configure your board, run the InstaCal utility installed in the previous section.
Refer to the Quick Start Guide that came with your board for information on how to initially set up andload InstaCal.

10

PCI-DIO96H User's Guide

Installing the PCI-DIO96H

Connecting the board for I/O operations
Connectors, cables – main I/O connector
The table below lists the board connectors, applicable cables and compatible accessory boards.
Board connectors, cables, accessory equipment
Connector typeCompatible cables
Compatible accessory products with
the C100FF-x cable

100-pin, high-density connector
C100FF-x (Figure 1)
CIO-MINI50*
CIO-SPADE50*
CIO-TERM100
SCB-50CIO-ERB24
CIO-SERB24/FD
CIO-ERB48
CIO-SERB48
SSR-RACK24
SSR-RACK48* two devices are required

The PCI-DIO96H board has a 100-pin, high-density Robinson-Nugent male connector. Connector pinouts arelisted on page 12. The C100FF-x cable can be used to split the 100 I/O lines into two, 50-pin cables.
Board connector pins 1 to 50 are mapped directly to pins 1 to 50 on the C100FF-x cable's first 50-pin connector.
Board connector pins 51 to 100 are mapped directly to pins 1 to 50 on the C100FF-x cable's second 50-pin
connector (pin 51 is mapped to pin 1, and pin 100 is mapped to pin 50.) A sample C100FF-x cableconfiguration is shown in Figure 2 on page 13.
Information on signal connections
General information regarding signal connection and configuration is available in the Guide to Signal
Connections. This document is available on our web site at www.mccdaq.com/signals/signals.pdf.
Caution! When connecting a cable to the board's I/O connector, make sure that the arrow indicating pin 1
on the board connector lines up with the arrow indicating pin 1 on the cable connector. Incorrectly
connected cables can damage the board and the I/O controller.

11

PCI-DIO96H User's Guide

Installing the PCI-DIO96H

Pin out – main I/O connector
Main I/O connector pin out
Signal name
GND
+5V
THIRDPORTC Bit 0
THIRDPORTC Bit 1
THIRDPORTC Bit 2
THIRDPORTC Bit 3
THIRDPORTC Bit 4
THIRDPORTC Bit 5
THIRDPORTC Bit 6
THIRDPORTC Bit 7
THIRDPORTB Bit 0
THIRDPORTB Bit 1
THIRDPORTB Bit 2
THIRDPORTB Bit 3
THIRDPORTB Bit 4
THIRDPORTB Bit 5
THIRDPORTB Bit 6
THIRDPORTB Bit 7
THIRDPORTA Bit 0
THIRDPORTA Bit 1
THIRDPORTA Bit 2
THIRDPORTA Bit 3
THIRDPORTA Bit 4
THIRDPORTA Bit 5
THIRDPORTA Bit 6
THIRDPORTA Bit 7
FOURTHPORTC Bit 0
FOURTHPORTC Bit 1
FOURTHPORTC Bit 2
FOURTHPORTC Bit 3
FOURTHPORTC Bit 4
FOURTHPORTC Bit 5
FOURTHPORTC Bit 6
FOURTHPORTC Bit 7
FOURTHPORTB Bit 0
FOURTHPORTB Bit 1
FOURTHPORTB Bit 2
FOURTHPORTB Bit 3
FOURTHPORTB Bit 4
FOURTHPORTB Bit 5
FOURTHPORTB Bit 6
FOURTHPORTB Bit 7
FOURTHPORTA Bit 0
FOURTHPORTA Bit 1
FOURTHPORTA Bit 2
FOURTHPORTA Bit 3
FOURTHPORTA Bit 4
FOURTHPORTA Bit 5
FOURTHPORTA Bit 6
FOURTHPORTA Bit 7

Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
79
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

PCI slot

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

↓

12

Signal name
GND
+5V
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
SECONDPORTC Bit 0
SECONDPORTC Bit 1
SECONDPORTC Bit 2
SECONDPORTC Bit 3
SECONDPORTC Bit 4
SECONDPORTC Bit 5
SECONDPORTC Bit 6
SECONDPORTC Bit 7
SECONDPORTB Bit 0
SECONDPORTB Bit 1
SECONDPORTB Bit 2
SECONDPORTB Bit 3
SECONDPORTB Bit 4
SECONDPORTB Bit 5
SECONDPORTB Bit 6
SECONDPORTB Bit 7
SECONDPORTA Bit 0
SECONDPORTA Bit 1
SECONDPORTA Bit 2
SECONDPORTA Bit 3
SECONDPORTA Bit 4
SECONDPORTA Bit 5
SECONDPORTA Bit 6
SECONDPORTA Bit 7

PCI-DIO96H User's Guide

Installing the PCI-DIO96H

Cabling
Cable is labeled“Pins 1-50”.

50

49

50

1

2

Key

100

The red stripe
identifies pin # 1

99 100

1

51

Key

51 52

Cable is labeled“Pins 51-100”.

Figure 1. C100FF-x cable

PCI-DIO96H
100-pin I/O connector

C100FF-x

Digital I/O
pins 1 to 50
IN
Digital signal conditioning or
50-pin screw terminal board
Digital I/O
pins 51 to 100
IN
Digital signal conditioning or
50-pin screw terminal board
Figure 2. C100FF-x cable configuration

13

The red stripe
identifies pin # 51

PCI-DIO96H User's Guide

Installing the PCI-DIO96H

Field wiring and signal termination accessories
You can use the following screw terminal boards to terminate field signals and route them into the PCI-DIO96H
using the C100FF-x cable.
CIO-MINI50 – 50-pin screw terminal board.
CIO-TERM100 – 100-pin screw terminal board (Two 50-pin IDC connectors).
CIO-SPADE50 — 16" X 4" termination panel which mates with both 37-pin and 50-pin connectors.
SCB-50 – 50 conductor, shielded signal connection/screw terminal box provides two independent 50-pin
connections.

Details on these products are available on our web site at www.mccdaq.com/products/screw_terminal_bnc.aspx.







CIO-ERB24 – 24 Form C relays, 6 Amp relay accessory board for digital signal conditioning.
CIO-SERB24/FD – 24 Form C relays, 10 Amp, fault detecting relay accessory board with socketed andfield-replaceable relays.
CIO-ERB48 – 48 Form C relays, 6 Amp, relay, 50-pin accessory board for digital signal conditioning.
CIO-SERB48 – 24 Form C relays, 10 Amp relay accessory board with socketed and field-replaceable
relays.
SSR-RACK24 – 24-channel, solid-state relay mounting rack for digital signal conditioning.
SSR-RACK48 – 48-channel, solid-state relay mounting rack with quad-format modules.

Details on these products are available on our web site at www.mccdaq.com/products/signal_conditioning.aspx.
For additional information about digital interfacing…
Detailed information regarding digital interfacing is contained in MCC's Guide to Signal Connections. This
document is available on our web site at www.measurementcomputing.com/signals/signals.pdf.

14

Chapter 3

Functional Details
CIO-ERB24 and SSR-RACK24 daisy chain configuration
Many relay and solid-state relay (SSR) racks provide only 24-bits of digital I/O. You can configure the CIOERB24 relay output board and SSR-RACK24 I/O module rack in a daisy chain configuration to use all of thedigital I/O bits provided by the PCI-DIO96H board. An example of the daisy chain configuration scheme foreach board is shown below.
The PCI-DIO96H board provides digital I/O in a group of 96 bits. Each of the C100FF-x cable's 50-pin
connectors provides 48 bits. To use all of the board's 96 digital I/O bits to control relays and/or SSRs, configure
the daisy chain as shown in Figure 3.

C100FF-x Cable

CIO-ERB24 IN
orOUT
SSR-Rack24

CIO-ERB24 IN
orOUT
SSR-Rack24
CIO-ERB24 IN
orOUT
SSR-Rack24

CIO-ERB24 IN
orOUT
SSR-Rack24

Figure 3. PCI-DIO96H to C100FF-x to relay rack daisy chain cabling

The 24 digital I/O bits on pins 1-24 control the first relay board on the chain. The 24 digital I/O bits on pins 2550 control the second relay/SSR board on the daisy chain and so on, for up to 100 pins.

82C55 emulation
The PCI-DIO96H board emulates the 82C55 chip. The 82C55 emulation initializes all ports as inputs on powerup and reset. A TTL input is a high impedance input. If you connect another TTL input device to the output, it
could be turned on or off every time the board is reset.
To establish a consistent TTL level at power-up, use resistors tied to either +5V (pull-up) or ground (pulldown). There are open locations for pull-up and pull-down resistor packs on the board.
Whenever an 82C55 emulation is powered on or reset, all pins are set to high-impedance input. Based onstandard TTL functionality, these inputs will typically float high, and may have enough drive current to turn on
external devices.
Consequently, if you have output devices such as solid state relays, they may be switched on whenever thecomputer is powered on or reset. To prevent unwanted switching, and to drive all outputs to a known state after
power on or reset, pull all pins either high or low through a 2.2 K resistor.

15

PCI-DIO96H User's Guide

Functional Details

Pull-up and pull-down resistors
The PCI-DIO96H board has open locations where you can install a 2.2 K , eight-resistor single inline package(SIP) resistor network for each port. The locations are marked PORT#A, PORT#B and PORT#C (RN10 throughRN21), and are adjacent to the I/O connector. PORT0A corresponds to FIRSTPORTA, PORT1A corresponds
to SECONDPORTA, etc, as shown on the pinout diagram.
The SIP is made up of eight 2.2 K resistors. One side of each resistor is connected to a single common point andbrought out to a pin. The common line is marked with a dot or line at one end of the SIP. The remaining resistor
ends are brought out to the other eight pins (refer to Figure 4).
2.2 KOhm SIP
Dot(LO or HI)

I/O Lines

Figure 4. Eight-Resistor SIP Schematic

The SIP may be installed as pull-up or pull-down. At each RN# location, there are 10 holes in a line. One end of
the line is +5V, the other end is GND. They are marked HI and LO respectively. The eight holes in the middle
are connected to the eight lines of a port.


For a pull-up function, mount the SIP with the common pin (marked with a dot or line) in the HI position.
For a pull-down function, mount the SIP with the common pin in the LO position.

When installing pull-up and pull-down resistor SIP packs, we recommend using a 2.2 K, eight-resistor SIP(MCC part number SP-K2.29C).
Unconnected inputs float
Unconnected inputs typically float high, but not reliably. If you are using a PCI-DIO96H board for input andhave unconnected inputs, ignore the data from those lines. You do not have to terminate input lines, andunconnected lines will not affect the performance of connected lines. Ensure that you mask out any
unconnected bits in software.

16

Chapter 4

Specifications
Typical for 25 °C unless otherwise specified.
Specifications in italic text are guaranteed by design.

Digital input / output
Table 1. Digital I/O specifications
Digital typeOutputInput
Configuration
Number of I/O
Output highOutput low
Input highInput low
Power-up / reset state
Pull-up/pull-down resistors

8255 emulation, Mode 0
74S24474LS373
8 banks of 8, 8 banks of 4, programmable by bank as input or output
96
2.4 volts min @ -15 mA
0.5 volts max @ 64 mA
2.0 volts min, 7 volts absolute max
0.8 volts max, -0.5 volts absolute min
Input mode (high impedance)
SIP resistor locations provided for pull-up or pull-down configuration.

Power Consumption
Table 2. Power consumption specifications+5V Operating

2.1 A typical, 3.4 A max

Environmental
Table 3. Environmental specifications
Operating temperature range
Storage temperature range
Humidity

0 to 50 °C-20 to 70 °C
0 to 90% non-condensing

Mechanical
Table 4. Mechanical specifications
Card dimensions

292.1 mm (L) x 106.6 mm (H) x 14.5 mm (W)17

PCI-DIO96H User's Guide

Specifications

Main connector and pin out
Table 5. Main connector specifications
Connector typeCompatible cables
Compatible accessory products

100-pin, high-density
C100FF- x
CIO-MINI50*
CIO-SPADE50*
CIO-TERM100
SCB-50CIO-ERB24
CIO-SERB24
CIO-ERB48
CIO-SERB48
SSR-RACK24
SSR-RACK48* two devices are required18

PCI-DIO96H User's Guide

Specifications

Table 6. Main connector pin out
Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
78
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

Signal name
GND
+5V
THIRDPORTC Bit 0
THIRDPORTC Bit 1
THIRDPORTC Bit 2
THIRDPORTC Bit 3
THIRDPORTC Bit 4
THIRDPORTC Bit 5
THIRDPORTC Bit 6
THIRDPORTC Bit 7
THIRDPORTB Bit 0
THIRDPORTB Bit 1
THIRDPORTB Bit 2
THIRDPORTB Bit 3
THIRDPORTB Bit 4
THIRDPORTB Bit 5
THIRDPORTB Bit 6
THIRDPORTB Bit 7
THIRDPORTA Bit 0
THIRDPORTA Bit 1
THIRDPORTA Bit 2
THIRDPORTA Bit 3
THIRDPORTA Bit 4
THIRDPORTA Bit 5
THIRDPORTA Bit 6
THIRDPORTA Bit 7
FOURTHPORTC Bit 0
FOURTHPORTC Bit 1
FOURTHPORTC Bit 2
FOURTHPORTC Bit 3
FOURTHPORTC Bit 4
FOURTHPORTC Bit 5
FOURTHPORTC Bit 6
FOURTHPORTC Bit 7
FOURTHPORTB Bit 0
FOURTHPORTB Bit 1
FOURTHPORTB Bit 2
FOURTHPORTB Bit 3
FOURTHPORTB Bit 4
FOURTHPORTB Bit 5
FOURTHPORTB Bit 6
FOURTHPORTB Bit 7
FOURTHPORTA Bit 0
FOURTHPORTA Bit 1
FOURTHPORTA Bit 2
FOURTHPORTA Bit 3
FOURTHPORTA Bit 4
FOURTHPORTA Bit 5
FOURTHPORTA Bit 6
FOURTHPORTA Bit 7

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

19

Signal name
GND
+5V
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
SECONDPORTC Bit 0
SECONDPORTC Bit 1
SECONDPORTC Bit 2
SECONDPORTC Bit 3
SECONDPORTC Bit 4
SECONDPORTC Bit 5
SECONDPORTC Bit 6
SECONDPORTC Bit 7
SECONDPORTB Bit 0
SECONDPORTB Bit 1
SECONDPORTB Bit 2
SECONDPORTB Bit 3
SECONDPORTB Bit 4
SECONDPORTB Bit 5
SECONDPORTB Bit 6
SECONDPORTB Bit 7
SECONDPORTA Bit 0
SECONDPORTA Bit 1
SECONDPORTA Bit 2
SECONDPORTA Bit 3
SECONDPORTA Bit 4
SECONDPORTA Bit 5
SECONDPORTA Bit 6
SECONDPORTA Bit 7

Declaration of Conformity
Manufacturer:
Address:

Category:

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, MA 02766
USAElectrical equipment for measurement, control and laboratory use.

Measurement Computing Corporation declares under sole responsibility that the product

PCI-DIO96H
to which this declaration relates is in conformity with the relevant provisions of the following standards or otherdocuments:
EC EMC Directive 2004/108/EC: General Requirements, EN 61326-1:2006 (IEC 61326-1:2005).
Emissions:EN 55011 (2007) / CISPR 11(2003): Radiated emissions: Group 1, Class A
EN 55011 (2007) / CISPR 11(2003): Conducted emissions: Group 1, Class A

Immunity: EN 61326-1:2006, Table 3.







IEC 61000-4-2 (2001): Electrostatic Discharge immunity.
IEC 61000-4-3 (2002): Radiated Electromagnetic Field immunity.
IEC 61000-4-4 (2004): Electric Fast Transient Burst Immunity.
IEC 61000-4-5 (2001): Surge Immunity.
IEC 61000-4-6 (2003): Radio Frequency Common Mode Immunity.
IEC 61000-4-11 (2004): Voltage Interrupts.

To maintain compliance to the standards of this declaration, the following conditions must be met.






The host computer, peripheral equipment, power sources, and expansion hardware must be CE
compliant.
All I/O cables must be shielded, with the shields connected to ground.
I/O cables must be less than 3 meters (9.75 feet) in length.
The host computer must be properly grounded.
Equipment must be operated in a controlled electromagnetic environment as defined by Standards EN
61326-1:2006, or IEC 61326-1:2005.

Declaration of Conformity based on tests conducted by Chomerics Test Services, Woburn, MA 01801, USA in
September, 2001. Test records are outlined in Chomerics Test Report #EMI3053.01. Further testing was
conducted by Chomerics Test Services, Woburn, MA. 01801, USA in January, 2009. Test records are outlined
in Chomerics Test report #EMI5243.09.
We hereby declare that the equipment specified conforms to the above Directives and Standards.

Carl Haapaoja, Director of Quality Assurance

Measurement Computing Corporation
16 Commerce Boulevard,
Middleboro, Massachusetts 02346(508) 946-5100
Fax: (508) 946-9500
E-mail: info@mccdaq.com
www.mccdaq.com