Communication analyzer with touch panel and color LCD

Supports RS-232C, RS-422/485, TTL (UART, I2C, SPI)

Multi-protocol analyzerLE-3500XR

NEW

Arbitrary speed up to 2 Mbps
100MB buffer memory
Up to 32GB auto save function
PROGRAM simulation
Wi-Fi remote control
The screen can be scrolled byswiping. The key operation is alsoavailable.

MODBUS standard compatible, CAN/CAN FD/CXPI/LIN optional.
● Multi interface / Multi-Protocol supported
Equipped with RS-232C, RS-422/485, TTL (1.8/2.5/3.3/5V)
measurement ports.

Standard support:
ASYNC (start-stop synchronization), SYNC/BSC,HDLC/SDLC, PPP(ASYNC), I2C, SPI, BURST, MODBUS
Optional support:
CAN FD/CAN, CXPI,LIN, X.20/21, RS449, V.35, RS-530,
Current loop,
High speed CC-LINK● Safe and reliable Auto-Save function
The measurement log file can be continuously recorded for along time into a SDHC card or a USB flash drive. Even if thebattery runs out suddenly, the log file being written is safely
closed and measurement is completed, so valuable records canbe reliably saved.
DUTspeed(bps)9600
115.2K1M

Estimated continuous recording time *100MB Mainmemory only

32GB SDHC card
Using SD-32GX

Approx. 6hrs.
Approx. 28min.
Approx. 200sec.

Approx. 80days
Approx. 6.5days
Approx. 20hrs.

*: When 1 kilobyte data is repeatedly transmitted by full duplex
with intervals of 1m second idle time for each.

● Convenient Statistical analysis function andlogic analyzer function
Expandable by replacing the board
or adding a dedicated cable● Monitor, simulation, BERT available
Many test functions and modes are available that are suitable foranalysis purposes and test scenes.

● Utilize measurement data on a PC
Includes PC link software (light version). Wi-Fi connection
between analyzer and PC is also available.

LE-3500XR

specifications

Standard
measurement port
Expansion
measurement port
Standard protocol
Expansion protocol
Capture memory
Measurable speed
Line signaling mode
Data display code
Measurement
condition setting
Online monitor function

Error check function
Idle time measurement

Time stamp record
Line status record
Interval timer
General counter
Data counter
Trigger function
Trigger condition

Trigger action

Monitor conditions autosetting

Auto run/stop functionAuto save function

Delay time function
Voltage measurement
function
Statistical analysis
function
Logic analyzer function!

SAFETYWARNING

RS-232C, RS-422/485,
TTL (1.8V/2.5V/3.3V/5V level)
CAN, CAN FD, CXPI, LIN, X.20/21,
RS-449, V.35, RS-530, current loop
ASYNC, SYNC/BSC, HDLC/SDLC,PPP(ASYNC), I2C, SPI, MODBUS, BURST
CAN, DeviceNet*1, CAN FD, CXPI, LIN,
CC-LINK *1
100 MB *2
Can be split into 250bps ~ 2.048Mbps(Available with 4 significant figures)
NRZ, NRZI, FM0, FM1
ASCII, EBCDIC, JIS, Baudot, Transcode,IPARS, EBCD, EBCDIK, HEX
Protocol, data/parity/stop bit, sync clock, frame
end, address filter, bit transmission order, bitpolarity, etc. can be specified according to theDUT
Raw data display, protocol translation display,error display, bit shift display possible
Parity, Fleming, Break, BCC, Abort,
Short frame, Non-ACK (I2C)
Resolution 1 msec., 10 msec., 100 msec., andOFF can be specified up to 999.9 sec
Minimum 10 msec. (day:minute:second.10msec.), 6 time units and OFF can be specified
Record RTS, CTS, DTR, DSR, DCD, RI,TRGIN (external trigger input) signals together
with send/receive data
2, Maximum count 999999(Resolution 1msec., 10msec., 100msec.)
2, Maximum count 999999
Maximum count for SD/RD is 4294967295
4 sets of trigger conditions and actions can bespecified
External trigger (1 input, 2 outputs)
Communication error, communication data
string of up to 8 characters (don't care and bit
mask can be specified), idle time longer thanspecified time, matching timer/counter value,
logical state of control line and external input
Stop measurement/test (offset number can bespecified), enable trigger condition, timercontrol, counter control, buzzer sound, filesave, specified character string transmission,pulse output to external terminal
Automatic setting of measurement conditions
such as speed (up to 460.8Kbps) and data
frame *3
Specified time RUN/STOP(Hourly/Daily/Monthly repeatable),Power-on RUN setting possible
Automatically save a specified number ofcommunication log files in external storage witha specified file size.
Ring record, automatic stop at maximum size
can be specified
Measures and displays the interval of change
in the interface signal line.
(current/min/max/average, resolution: 0. 1ms)

RS-232C input range ±18V
TTL input range -1V to 6V, resolution 0.1V

Simulation function

Data can be sent to the test target while
checking the received data

Test data table

Transmit test data 160 types (16K data) can beregistered, BCC/LRC automatic calculation,
parity error data can be registered

Test mode

6 modes (MANUAL, FLOW, ECHO,POLLING, BUFFER, PROGRAM)
selectable
Measures line quality such as bit error rateand block error rate according to ITU-T
G.821 *4
Save, load and delete measurement data
and measurement conditions
PC link software (light version *5) included
Measurement data display, recording,text/CSV conversion
4.3 inch TFT color LCD (480x272dot)
With touch panel
Line state display, power/charge display
USB2.0 (Standard A /micro-B each 1),
SDHC card slot, Wi-Fi 802.11 b/g/n *6
SDHC card and USB memory *7 Up to32GB
USB bus power, built-in lithium-ion battery
Battery operating time: About 7 hours *8
Operating temperature: 0-40°C
Storage temperature: -10-50°C
190 (W) x 153 (D) x 38 (H) mm, About 550g
Monitor cable for DSUB 25-pin (LE-25M1),
Monitor cable for DSUB 9-pin (LE-009M2),
DSUB25pin-9pin conversion adapter,5 wires TTL prove cable (LE-5LS), micro
USB cable, USB charger (LE-P2USB),
carrying bag (LEB-01), Utility CD,
quick start guide, and warranty

Bit error rate test
File management
Remote control
LCD display
LED display
External interface
External storage

Power supply
Temperature range
Dimension, massAccessories*1: Raw data display only. The optional OP-FW10XR is required for high-speed CC-LINK. *2:Communication data, idle time, time stamp, and line status consume 4-8 bytes of memory for eachcapture. *3: Automatic setting is available only for ASYNC, SYNC/BSC, HDLC/SDLC. If thecommunication data amount is small or there are many errors, the automatic setting cannot beperformed correctly. *4: Only available in ASYNC and SYNC modes. *5: Some functions of the productversion PC link software LE-PC300R are limited. *6: It can be shipped with the Wi-Fi function disableddue to the radio law of each country. Please contact our sales department for details. *7: Operation ofSDHC cards other than LINEEYE option is not supported. *8: According to our measurement
conditions.

OptionItem name
Expansion kit for CAN FD/CAN/CXPI
Expansion kit for CAN FD/CAN/LIN
HDLC/SPI Firmware to speed up
Expansion board for RS-530
Expansion kit for current loop comm.
Expansion kit for TTL/I2C/SPI
5 wires TTL prove cable
X.21 monitor cable
RS-449 monitor cable
V.35 monitor cable
RS-530 cable
USB charger
PC link software
PC link software (For OP-SB7XC/7XL)
3.81mm pitch terminal block (5 POS.)

Model number
OP-SB7XC
OP-SB7XL
OP-FW10XR
OP-SB10N
OP-SB1C
OP-SB5GL
LE-5LS
LE-25Y15
LE-25Y37
LE-25M34
LE-25S530
LE-P2USB
LE-PC300R
LE-PC7XCL
LA-5ETB45
SD-16GX
SD-32GX

Remarks*1*2*3*3*3*3*2*4*4*5

Communication data, frame, and trigger match
count are aggregated for each specified period(1 to 24 minutes) and displayed as a graph
1KHz to 20MHz sampling, zoom display,
cursor time measurement, signal order swap

*1: Speed up to 10 Mbps
*2: Same as the attached one.
*3: OP-SB10N required.
*4: Full edition with no functional restrictions*5: Same as the one set to the analyzer.

Read the instruction manual provided with the product before
use and use the product as explained in that manual. Using
the product in ways not guaranteed in the manual, connecting
it to systems outside of the specified ranges and remodelingcan all cause trouble and damage. LINEEYE CO., LTD. willassume no responsibility whatsoever for trouble or damage
arising because of unauthorized ways of use.

•All brand names and product names mentioned in this leaflet are trademarks or registeredtrademarks of their respective companies. •Specifications and designs of products listed inthis leaflet are as of July 2020, and are subject to change without notice for improvement.
•Colors of actual products may differ slightly from that listed due to printing condition. •This
leaflet may not be reprinted or duplicated, in part or in whole.
©2020 by LINEEYE CO., LTD.。

16GB SDHC card
32GB SDHC card

LINEEYE CO., LTD.
Head Office/Sales OfficeMarufuku Bldg 4F, 39-1 Karahashi Nishihiragaki-cho, Minami-ku, Kyoto, 601-8468
PHONE: 81-75-693-0161 FAX:81-75-693-0163

URL： https://lineeye.co.jp

E-mail： info@lineeye.co.jp
Printed in Japan

L-20072E/LE