MODEL 606M1 ACCELEROMETERSPECIFICATIONSDIMENSIONS

Seat Pad Accelerometer
MEMS, Triaxial Sensors
DC Response
Accurate Temp Compensation
ISO 10326-1 Configuration

The Model 606M1 is a MEMS triaxial seat pad
accelerometer with both static and dynamic responses
designed specially for characterizing whole body vibrationin accordance with ISO 2631-1 and ISO 8041. The DC
response of the silicon MEMS sensors is the key to yieldaccurate velocity and displacement results from the raw
acceleration data.
The 606M1 incorporates integral temperaturecompensation that provides a stable output over a wide
operating range. The on-board voltage regulation circuit
works with power supply from 8 to 32Vdc.

FEATURES

Three Independent Circuits
Low Current Consumption
Ranges: ±25g
Gas Damped, DC ResponseHigh Over-Range Protection
Low Transverse Sensitivity

APPLICATIONS





SENSOR SOLUTIONS /// Model 606M1 Rev C

Whole Body Vibration Study
Vibration/Shock Monitoring
Helicopter Flight Testing
Heavy Equipment Testing
Biodynamic Study

07/2017

Page 1

MODEL 606M1 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 12Vdc excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change the specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)±25800-8000-1000
4000±1.0
<30.7
5000

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Output Impedance (Ω)
Insulation Resistance (MΩ)
Turn On Time (msec)
Residual Noise (µV RMS)
Ground Isolation±100
8 to 36<152.5<100>100<100800
Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO)
Thermal Sensitivity Shift (%)
Operating Temperature (°C)
Compensated Temperature (°C)
Storage Temperature (°C)

±3±3.5-20 to 85-20 to 85-20 to 85

TypicalTypicalPHYSICAL
Case Material (Seat Pad)
Cable
Weight (grams)

Nitrile Rubber
6x #28 AWG Conductors, PFA Insulated, Braided Shield, TPE Jacket
380Notes

Calibration supplied:

CS-FREQ-0100

Optional accessories:

121±5%±1dB@100Vdc
Passband

NIST Traceable Amplitude Calibration from 20Hz to ±5% Frequency Response Limit

Three Channel DC Signal Conditioner Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 606M1 Rev C

07/2017

Page 2

MODEL 606M1 ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number

606M1

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 606M1 Rev C

07/2017

Page 3