Product Datasheet

Arium® Pro
Ultrapure
Water Systems
Application-orientated
and flexible to meetthe highest demands---

Advantages
Modular – System selection specifically for your application
Flexible – Perfect integration into any laboratory
Easy to use – Display with touch function and intuitive menu
Fast – Favorites function with direct accessfor recurring volumes

Product Description
As a reliable source of ultrapure water, the Arium® Pro series
offers a flexible and modular system which, compared toconventional devices, demonstrates excellent added value.

All systems produce consistent ASTM Type 1 ultrapure water
quality and provide the best reproducible results.
The ­ultrapure water can be dispensed at up to 2 liters per­minute with a conductivity of 0.055 µS/cm (18.2 MΩ × cm).
When using an Arium® Sterile Plus final filter, the ultrapure
water is virtually free of microorganisms.
The patented Sartorius technology, the SD card slot, the longservice life and low maintenance requirement distinguish
the Arium® Pro systems as an easy-to-use, economical andreliable Type 1 ultrapure water system.

Modular

Display with touch function

Various device configurations are specifically tailoredto your application. Arium® Pro delivers any desired
ultrapure water quality for general, analytical and life
science ­applications.

Simply navigate intuitively in the easy-to-use and clear
menu by lightly touching the display – even with gloves.
Even the opening of the dispensing valve can be controlled
by the unique touch display.

“Favorites” function

FlexibleWith the new favorites function it is possible to saverecurring volumes and retrieve them as required by directaccess.

The space-saving installation of the device on, under, orabove your workstation integrates it perfectly into any
laboratory. The positioning of the display and the water
dispensing point is very flexible.

Simplify your daily routine by using the new function to savetime and work more efficiently in the laboratory.

Feed Water Quality
Treated water by reverse osmosis, distillation or deionization.1

Technical Specifications
Dimensions:width × height × depth

35.0 × 49.2 × 45.1 cm

Empty weight

17 – 19 kg, depending on the device typeOperating weight

27 – 29 kg, depending on the device type

Power supply

100 – 240 VAC (±10%);50 – 60 Hz, 130 VA (max.)

Operating temperature 2 °C – 35 °C at max. 80% relative humidity
Storage temperature

5 °C – 45 °C at max. 80% relative humidity

Data output

SD card slot2, RS-232 interface

Input pressure3

0 – 6.9 bar, recommended > 2 bar

Temperature2 – 30 °C

Specific conductivity< 100 µS/cm compensated to 25 °C

TOC< 50 ppb

Turbidity< 1 NTU

pH value4 – 10

1 With the Universal Kit, Arium® Pro can be directly fed with untreated drinking water to produce
ultrapure water.
The appropriate Sartorius application specialists should be consulted to check the feed water
specifications.
2 Does not apply to Arium® Pro³ Dynamic pressure/flow pressure 100 L/h

UV lamp (185|254 nm)

Cleaning port

Conductivity
measurement(input water)

Ultrafilter
Pump

Feedwater(inlet)Cartridge 1

Cartridge 2

TOC
Flushing wateroutletConductivity measurement(product water)
Final filter
0.2 µmProduct water

Flow diagram for Arium® Pro VF TOC
2

Water Applications
Pro
Water Quality
Type 1 Water

Lab Water System by Daily Water Consumption
Type 1 ultrapure water approx. 40 – 100 Liter/day

General Laboratory Application
Buffer, media and pH solutions
Histology
ELISA (Enzyme-Linked Immunosorbent Assay)
AAS (Atomic Absorption Spectroscopy)
Solutions for chemical analysis and synthesis
GF-AAS (Graphite Furnace Atomic Absorption Spectrometry)
Preparation of reagents
Photometry-----

Pro DI-----Pro UV-----

Molocular Biology | Lifescience Application
Electrophoresis
Northern BlotSouthern Blot
Western Blot
Endotoxin analysis
Immunocytochemistry
Production of mono­clonal antibodies
PCR (Polymerase Chain Reaction)
DNA Sequenzing
Nutrient media for cell culture (Mammalia & plant)
Chromatography

Analytical Application
SPE (Solid phase extraction)
Trace metal analysis
IC (Ion chromatography)
ICP-MS (Inductively Coupled Plasma Mass Spectrometry)
LC-MS (Liquid Chromatography with Mass Spectrometry)
GC-MS (Gas Chromatography–Mass Spectrometry)
HPLC (High-Performance Liquid Chromatography)
TOC analysis
All displayed applicable systems starting with the minimal requested water quality criteria-----

Pro UFPro VF------------

-----

-----

----3

Arium® Pro DI and Arium® Pro
Description

15:32
0.055 µS/cm c

The Arium® Pro DI produces ASTM Type 1 ultrapure waterand is used for standard daily laboratory applications.

Menu

10.01.20

Care

The water is treated using the specially developed Elemental
Kit, which reliably removes organic and inorganic impurities
from the feed water using a mixture of activated carbon andion exchange resins.
To protect against contamination by particles and bacteria,
an Arium® Sterile Plus can additionally be connected at theconsumer endpoint.

A more cost-effective alternative is the Arium® Pro.
Reduced to the most important functions, it produces­ultrapure water without compromising quality.

Product Water Quality
Water purification method

Adsorption by means of spherical activated carbon,
deionization, optional end-position particle and sterile filtration

Water type

ASTM Type 1 ultrapure water

Output120 L/hWater dispensing flow rate4

0.1 – 2 L/min, adjustable

Volume-controlled dispensing

0.05 in 0.05 L step, 0.1 – 2.0 L in 0.1 L steps,
2.0 – 20 L in 1 L steps, 20 – 50 L in 5 L steps

Volume accuracy5

3% between 0.25 L and 50 L

Conductivity1

0.055 µS/cm compensated to 25 °C

Resistivity1

18.2 MΩ × cm compensated to 25 °C

TOC3< 5 ppb

Bacteria2< 0.001 CFU/mL

Particle content2

No particles > 0.2 μm

1 Measured value output adjustable to 25 °C, compensatedor uncompensated2 When using an Arium® Sterile Plus final filter
3 Feedwater < 50 ppb TOC
4 At a dynamic flow pressure of 2 bar, depending on theconnected accessory or final filter
5 Under constant operating conditions

Ordering Information
Arium® Pro DI and Pro systems, for the production of ASTM Type 1 ultrapure water
Scope of supply: 1 Arium® Pro or Pro DI, water guard and connection set
Order number

Description

H2Opro-DI-T

Arium® Pro DI benchtop device

H2Opro-DI-B

Arium® Pro DI wall-mounted device

H2Obasic-T

Arium® Pro benchtop device

H2Obasic-B

Arium® Pro wall-mounted device

For under-bench installation of the Arium® Pro DI devices please order a comparable bench-top device, as well as the conversion kit
described under the accessories (H2O-ACK-D).

4

Arium® Pro UV
Description

15:32
0.055 µS/cm c
1.0 ppb TOC

The Arium® Pro UV produces ASTM Type 1 ultrapure
water and is used for critical analytical applications in thel­ a­boratory.

Menu

10.01.20

Care

In addition to the removal of organic and inorganic
impurities via the Analytical Kit, by means of a mixture ofactivated carbon and ion exchange resins, the Arium® Pro
UV also ­purifies the water via a UV lamp.

Current TOC values can be continuously measured by
the optionally integrated TOC monitor and shown on the­display.

The UV lamp (185|254 nm) reduces organic components
to a minimum (TOC ≤ 2 ppb) and thus ensures reliable andprecise analytical results.

An Arium® Sterile Plus can also be connected to the­consumer endpoint to protect against particulate and­bacterial contamination.

Product Water Quality
Water purification method

Adsorption by means of spherical activated carbon,
deionization, UV irradiation, optional end-position particle
and sterile filtration

Water type

ASTM Type 1 ultrapure water

Output120 L/hWater dispensing flow rate4

0.1 – 2 L/min, adjustable

Volume-controlled dispensing

0.05 in 0.05 L step, 0.1 – 2.0 L in 0.1 L steps,
2.0 – 20 L in 1 L steps, 20 – 50 L in 5 L steps

Volume accuracy5

3% between 0.25 L and 50 L

Conductivity1

0.055 µS/cm compensated to 25 °C

Resistivity1

18.2 MΩ × cm compensated to 25 °C

TOC3≤ 2 ppb

Bacteria2< 0.001 CFU/mL

Particle content2

No particles > 0.2 μm

1 Measured value output adjustable to 25 °C, compensatedor uncompensated2 When using an Arium® Sterile Plus final filter
3 Feedwater < 50 ppb TOC
4 At a dynamic flow pressure of 2 bar, depending on theconnected accessory or final filter
5 Under constant operating conditions

Ordering Information
Arium® Pro systems, for the production of ASTM Type 1 ultrapure water
Scope of supply: 1 Arium® Pro with UV lamp (185 | 254 nm), water guard and connection set
Order number

Description

H2Opro-UV-T

Arium® Pro UV benchtop device, including UV lamp

H2Opro-UV-B

Arium® Pro UV wall-mounted device, including UV lamp

H2Opro-UV-T-TOC

Arium® Pro UV benchtop device, including UV lamp and TOC monitor

H2Opro-UV-B-TOC

Arium® Pro UV wall-mounted device, including UV lamp and TOC monitor

For under-bench installation of the Arium® Pro UV devices please order a comparable bench-top device, as well as the conversion kit
described under the accessories (H2O-ACK-D).

5

Arium® Pro UF
Description

15:32
0.055 µS/cm c

The Arium® Pro UF produces ASTM Type 1 ultrapure
water and is used for critical biological applications in thelaboratory.

Menu

10.01.20

Care

The water is first treated with the Elemental Kit, using a­mixture of activated carbon and ion exchange resins to­remove organic and inorganic impurities.
Subsequently, purification takes place via an ultrafilter using
cross-flow technology. The ultrafilter reliably removes­endotoxins, DNases and RNases, making the system ideal
for use in cell culture or critical biological applications.

To protect against contamination by particles and bacteria,
an Arium® Sterile Plus can additionally be connected at theconsumer endpoint.

Product Water Quality
Water purification method

Adsorption by means of spherical activated carbon, deionization,
ultrafiltration, optional end-position particle and sterile filtration

Water type

ASTM Type 1 ultrapure water

Output102 L/hWater dispensing flow rate4

0.1 – 1.7 L/min, adjustable

Volume-controlled dispensing

0.05 in 0.05 L step, 0.1 – 2.0 L in 0.1 L steps,
2.0 – 20 L in 1 L steps, 20 – 50 L in 5 L steps

Volume accuracy5

3% between 0.25 L and 50 L

Conductivity1

0.055 µS/cm compensated to 25 °C

Resistivity1

18.2 MΩ × cm compensated to 25 °C

TOC3< 5 ppb

Bacteria2< 0.001 CFU/mL

Particle content2

No particles > 0.2 μm

Endotoxin< 0.001 EU/mL

RNase concentration< 1 pg/mL

DNase concentration< 5 pg/mL

1 Measured value output adjustable to 25 °C, compensatedor uncompensated2 When using an Arium® Sterile Plus final filter
3 Feedwater < 50 ppb TOC
4 At a dynamic flow pressure of 2 bar, depending on theconnected accessory or final filter
5 Under constant operating conditions

Ordering Information
Arium® Pro UF systems, for the production of ASTM Type 1 ultrapure water
Scope of supply: 1 Arium® Pro with ultrafilter, water guard and connection set
Order number

Description

H2Opro-UF-T

Arium® Pro UF benchtop device, including ultrafilter

H2Opro-UF-B

Arium® Pro UF wall-mounted device, including ultrafilter

For under-bench installation of the Arium® Pro UF devices please order a comparable bench-top device, as well as the conversion kit
described under the accessories (H2O-ACK-D).

6

Arium® Pro VF
Description

15:32
0.055 µS/cm c
1.0 ppb TOC

The Arium® Pro VF is the high-end instrument and­produces ASTM Type 1 ultrapure water for both critical­biological and critical analytical applications.

Menu

10.01.20

Care

In addition to water treatment via the Analytical Kit, using
activated carbon and ion exchange resins, the Arium® Pro
VF combines the advantages of the Arium® Pro UV and
Arium® Pro UF in one system.
The combination of integrated UV lamp (185|254) and­ultrafilter thus not only provides a reduction of organic­contamination to a minimum (TOC ≤ 2 ppb), but also
simultaneously removes endotoxins, as well as DNases and
RNases. This makes the Arium® Pro VF ideal for a variety ofdifferent critical applications in the laboratory.

Current TOC values can be continuously measured by
the optionally integrated TOC monitor and shown on the­display.
An Arium® Sterile Plus can also be connected to the­consumer endpoint to protect against particulate and­bacterial contamination.

Product Water Quality
Water purification method

Adsorption by means of spherical activated carbon, deionization,
ultrafiltration, UV irradiation, optional end-position particle andsterile filtration

Water type

ASTM Type 1 ultrapure water

Output102 L/hWater dispensing flow rate4

0.1 – 1.7 L/min, adjustable

Volume-controlled dispensing

0.05 in 0.05 L step, 0.1 – 2.0 L in 0.1 L steps, 2.0 – 20 L
in 1 L steps, 20 – 50 L in 5 L steps

Volume accuracy5

3% between 0.25 L and 50 L

Conductivity1

0.055 µS/cm compensated to 25 °C

Resistivity1

18.2 MΩ × cm compensated to 25 °C

TOC3≤ 2 ppb

Bacteria2< 0.001 CFU/mL

Particle content2

No particles > 0.2 μm

Endotoxin< 0.001 EU/ml

RNase concentration< 1 pg/mL

DNase concentration< 5 pg/mL

1 Measured value output adjustable to 25 °C, compensatedor uncompensated2 When using an Arium® Sterile Plus final filter
3 Feedwater < 50 ppb TOC
4 At a dynamic flow pressure of 2 bar, depending on theconnected accessory or final filter
5 Under constant operating conditions

7Ordering Information
Arium® Pro VF systems, for the production of ASTM Type 1 ultrapure water
Scope of supply: 1 Arium® Pro VF with UV lamp (185 | 254 nm). ultrafilter, water guard and connection set
Order number

Description

H2Opro-VF-T

Arium® Pro VF benchtop device, including UV lamp and ultrafilter

H2Opro-VF-B

Arium® Pro VF wall-mounted device, including UV lamp and ultrafilter

H2Opro-VF-T-TOC

Arium® Pro VF benchtop device, including UV lamp, ultrafilter and TOC monitor

H2Opro-VF-B-TOC

Arium® Pro VF wall-mounted device, including UV lamp, ultrafilter and TOC monitor

For under-bench installation of the Arium® Pro VF devices please order a comparable bench-top device, as well as the conversion kit
described under the accessories (H2O-ACK-D).

Additional product water specifications when connected to an Arium® Smart Station¹ final filter
Particle content²

No particles > 0.2 µm

Bacteria²< 0.001 CFU/mL

Endotoxins³< 0.001 EU/mL

RNase concentration³< 1 pg/mL

DNase concentration³< 5 pg/mL

Water dispensing flow rate⁴

Up to 2 L/min

Volume-controlled removal

0.05 – 50 L in 50 mL steps

1 Connected to an Arium® Pro DI, UV, UF and VF² When using an Arium® Sterile Plus (Sartopore® 2 150 final filter)³ When using an Arium® Cell Plus final filter
⁴ Depending on the connected Arium® Pro, hydrostatic pressure, connected accessories or end filter

Accessories and final filters for the Arium® Smart Station can be found in the Arium® Smart Station data sheet.

8

Accessories
Arium® Conversion Kit--

Flexibly placeable, simple and space-saving integration
Optimal integration into your laboratory furniture
Space-saving arrangement of the system through variable
wall installation of the display | dispenser unit
Full operation directly on the display | dispenser unit

Description
In conjunction with an Arium® bench top system, the Arium®
Conversion Kit also enables the installation of the device asa built-in version.
By extending the tube routing as well as the display |dispenser unit, the system can be ideally integrated into
your laboratory furniture.
This version creates more space on and above thelabora­tory bench, as the control unit with display and water
dispenser can be mounted on the wall in various ways.

Technical Specifications |Ordering Information
Dimensions
Tubing

1/4“Tube length

3.4 m

Cable length

3.0 m

Order number

Description

H2O-ACK-D

Arium® Conversion Kit, including wall
mounting kit for the display | dispenserunit** The Arium® Conversion Kit can only be used in conjunction with an Arium® bench-top device.
Conversion of the system should only be carried out by Sartorius Service specialists.

-

Intended Use
Device type:
Arium® Pro DI, Pro UF, Pro UV and Pro VF

9Arium® Smart Station---

Remote dispensing at high flexibility
Compact: Save space integrating in your lab
Intuitive: Touch-activated color display with direct accessto all important functions
Flexible: Stepless height adjustment to fill different size
containers
Accurate: Precise volume dispense for reliable bufferand sample preparation

Description
The Arium® Smart Station is designed for flexible remote
dispensing of ultrapure water directly at the point of use.
While dispensing water into a broad range of different sized
containers, the Smart Station offers constant control ofevery important quality parameters, at all times. The
ergonomic design supports left-and right-hand operationand can be easily adjusted to your need.
Additionally, for maximum flexibility, you can connect up
to three Smart Stations to the Arium® Pro. By using theavailable hose extension set, the distance between eachdispense device, can be extended up to 4 meters.
Furthermore, different point-of-use filters for different
applications can be added to the individual Smart
Stations, as required.

Arium® Smart Station Ultrapure:Supplies ultrapure water from Arium® Pro

10

Technical Specifications |Ordering Information
Dimensions Smart Station Bench-Top

Order number

Description

Control box with stand(w × d × h)

213 × 213 × 598 mm(8.4 × 8.4 × 8.2”)

H2O-ARST-UP-T

Arium® Smart Station Ultrapure
for benchtop installation

Operating range fixed
dispense arm (d × w × h)

428 × 476 × 835 mm(16.9 × 18.7 × 32.9”)

H2O-ARST-UP-B

Arium® Smart Station Ultrapure forwall-mounted installation

Tubing Length:
Distance to water system

2 Meter

Operating range flexible hand
held

0.7 Meter

Benchtop and wall-mounted edition can be assembeled forleft or right hand side, without additional equipment required.

WeightApprox. 4.9 kg (10.8 Ibs)

Dimensions Smart Station Wall-Mounted
Control box (w × d × h)

172 × 157 × 343 mm(6.8 × 6.2 × 13.5”)

Operating range
dispense arm (d × w × h)

242 × 90 × 300 mm(9.5 × 3.5 × 11.8”)

Tubing Length:
Distance to water system

2 Meter

Operating range flexible hand
held

0.7 Meter

WeightApprox. 2.4 kg (5.3 Ibs)

Accessories and final filters for the Arium® Smart Station canbe found in the Arium® Smart Station data sheet.

-

Intended Use
Device type:
Arium® Pro DI, Pro UF, Pro UV and Pro VF

General Specifications
Volume-controlled dispensing

0.05 - 50 L in 50 mL steps

Volume accuracy±5% between

Power supply

100 – 240 VAC; 50 and 60 Hz,
2.5 A (max.) 2 °C – 40 °C

Power cord (IEC 60320-1 /C14)Country specific11Arium® Water Guard-----

Early detection of leakages protects the laboratory
Highly sensitive optical sensor
Audiovisual alarm signals
Automatic water stop in the case of leakage
High-quality material, no corrosion
Easy to install
Integrated wall mounting bracket for solenoid valve

Description
Only the early detection of water leakages provides optimal
protection against water damage in the laboratory. Leak­ages are registered by the highly sensitive optical sensor.
In contrast to conventional sensors, this sensor functions
independently of conductivity measurement values asthese are so low in the ultrapure water that the activation of
the guard would not be guaranteed. Once a leak isdetected, the water guard automatically locks the feedwater inlet line. An acoustic warning is triggered
immediately and the system status can be constantly
controlled using the integrated LED display. With its
sensitive optical sensors and high-quality materials, theArium® Water Guard is perfect for all ultrapure water
systems.

Technical Specifications |Ordering Information
Sensor dimensions
Diameter

5 cm

Height2.5 cmCable length2mTubing connections
Input

⅜” Plug-in connector

Output⅜” Plug-in connector

Power supply

100 – 240 VAC | 50 – 60 Hz

12

Order number

Description

610AWG1

Arium® Water Guard, 1 pc-

Intended Use
Device type:
Arium® Pro, Pro DI, Pro UF, Pro UV and Pro VF

Arium® Foot Switch--

Greater convenience during ultrapure water dispensing
Water dispensing at a press of the footFacilitates work in the clean room and minimizes
the risk of contamination
Low installation height enables Comfortable,
fatigue-free switching

Description
Easy-to-connect foot switch to start and stop the water
extraction dispense be performed with both hands, e.g. forswitching vessels, and minimizes the risk of contaminationin the clean room.

Technical Specifications |Ordering Information
Material

Nylon, glass fiber-reinforced

Order number

Description

Dimensions [W × H × D]

14.0 × 4.5 (max.) × 10.6 cm

H2O-AFS1

Arium® Foot Switch, 1 pc

Cable length2mPower supply

100 – 240 VAC | 50 – 60 Hz

Connection

Phoenix plug, 2-pin-

Intended Use
Device type:
Arium® Pro DI, Pro UF, Pro UV, and Pro VF

13

Arium® Level Sensor--

Practical filling of separate tanks
Flexible water transport to any location
Any tank system can be filled

Description
The level sensor makes it easy to connect an external
water storage tank and subsequently fill a tank withultrapure water.

Technical Specifications |Ordering Information
Level sensor length

88 mm

Order number

Description

Connection diameter2.03 cm (max.)

H2O-ALS1

Arium® Level Sensor, 1 pc

Drill hole

1.65 cm

Cable length

3m

14

-

Intended Use
Device type:
Arium® Pro DI, Pro UF, Pro UV, and Pro VF

Arium® Printer---

GMP data documentation made easy
Acquisition and documentation of current
measurement data
High printing speed
Compact and robust design
Thermal transfer printing process (for durable prints
in regulated areas)
Direct thermal printing method possible (for lessstringent requirements in standard use)

Description
To assist with qualification and documentation tasks,
current measured values are output via an RS-232 interface
to the printer.

Technical Specifications |Ordering Information
Dimensions [W × H × D]

241.3 × 139.9 × 177.4 mm

Order number

Description

Interface

RS-232 (max 115,200 bps) – USB 2.0(full speed)

YDP30

Printer, 1 pc

SB-12-01-0250

Connection cable Arium® (required),
1 pc

69Y03285

Set of standard paper and ink ribbon
for thermal transfer printing(GMP-compliant)

69Y03287

Standard paper for direct thermal
printing

Power supply--

External universal switching
power supply
Input: 100 – 240 V~
Output: 24 V–; 2.5 A-

Intended Use
Device type:
Arium® Pro, Pro DI, Pro UF, Pro UV and Pro VF

15

Consumables
Arium® Pro Cartridge Sets---

Pretreatment and post-treatment cartridge using
top-down technology
High performance capacity thanks to efficient
ion exchange resins
Effective adsorption of impurities throughhigh-grade activated carbon
Optimized flow pattern, prevents separationof the mixed resin bed
Patented connection process simplifies the replacementof consumables

Description
The cartridge sets are optimized for the removal of bothorganic and inorganic constituents. Every set has been
designed specifically to match the unit and delivers
ultrapure water that even exceeds the ASTM type 1 quality
standard. This consistent level of high-quality water ensures
optimal reproducibility of your results.
Optimized purification materials such as highly effective
activated carbon coupled with a efficient ion-exchange

resins deliver long-lasting performance and thereby ensure
long maintenance intervals.
The top-down flow technology produces ideal purification
kinetics and prevents any mixing of cleaning media. The
cartridge was designed with the applicable standards forflow rate in the cross section and contact time with themedium in mind.

Technical Specifications |Ordering Information
Materials
HousingHigh-purity polypropylene

Fixing screws

Stainless steel

Cleaning media

Spherical, catalytic activated carbons
Ultrapure mixed bed ion exchange resin

Feed water requirements

see “Technical Specifications” page 2* With the Universal Kit, Arium® Pro can be directly fed with untreated drinking waterto produce ultrapure water. The appropriate Sartorius application specialists should be
consulted to check the feed water specifications.

16

Order number

Description

H2O-A-PACK

Analytical Kit, Arium® Pro cartridge set
for biological, analyticaland standard ultrapure water applications, 1 pc

H2O-B-PACK

Biological Kit, Arium® Pro cartridge set
for biological ultrapure water applications, 1 pc

H2O-E-PACK

Elemental Kit, Arium® Pro cartridge
set for standard ultrapure water applications, 1 pc

H2O-U-PACK*

Universal Kit, Arium® Pro cartridge set
for untreated feed water*, 1 pc---

Intended Use
Device type:
H2O-A-PACK Arium® Pro VF and Pro UV
H2O-B-PACK Arium® Pro UF
H2O-E-PACK Arium® Pro and Pro DI
H2O-U-PACK*	Arium® Pro, Pro DI, Pro UF, Pro UV
and Pro VF

Arium® Sterile Plus------

Sterile and particle-free water dispensing
Excellent service life and flow rates
Integrity tested
Validated according to HIMA and ASTM F-838-05
Meets WFI quality standards pursuant to USP
incl. USP plastic class VI test
Production in accordance with DIN ISO 9001
Easy to install
Automatic venting
Certified quality

Description
The Arium® Sterile Plus (Sartopore® 2 150) is a sterile,
ready-to-use membrane filter capsule suitable for themost stringent requirements. Arium® Sterile Plus membranefilter capsules contain a hydrophilic, heterogeneouspolyethersulfone double membrane. It enables an excellent
service life and flow rates. The capsule is attached in theend position by means of a quick connector and reliably
removes all particles and micro­organisms in the last water
purification step. A hydrophobic PTFE membrane at thefarthest point “upstream” allows for easy and cleanventilation of the capsule.

All pleated Arium® Sterile Plus membrane filter units arevalidated as sterile filters for biopharmaceutical applicationaccording to the HIMA and ASTM F-838-05 guidelines(documentation available). During the manufacturing
process, every capsule is integrity-tested to meet thehighest quality standards and safety regulations.

Technical Specifications |Ordering Information
Materials

Typical Specifications

Membranes

Asym. Polyethersulfone

Bacteria< 0.001 CFU/mL

Bell assembly

Polycarbonate

Particle content

No particles > 0.2 μm

Other plastics

Polypropylene
Order number

Description

5441307H4--CE

Arium® Sterile Plus (Sartopore® 2
150 Capsule), 0.2 µm pore size, 1 pc

General Specifications
Pore size

0.45 µm × 0.2 µm

Filtration area

0.015 m2

Input and Output¼” Plug-in connector

Sterilization (max. 3 cycles)

Autoclaving at 134 °C, 2 bar, 30 min.

Max. diffusion

1 mL/min @ 2.5 bar

Min. bubble point

3.2 bar--

Intended Use
Device type:
Arium® Smart Station Ultrapure
Arium® Pro, Pro DI, Pro UF, Pro UV and Pro VF

17

Arium® UV Lamp
(185 | 254 nm)---

Ultrapure water, free of TOC
Horizontal installation, optimized temperature gradient
Effectively removes organic compounds
Easy replacement

Description
The horizontally arranged UV lamp delivers especiallyreliable results. Unlike vertical units, the temperature
gradient is less pronounced and does not affect the activityof UV waves.
The two different wavelengths reliably removes organic
substances down to a TOC (total organic carbon) ≤  2 ppb*

Technical Specifications |Ordering Information
Typical Specifications
TOC value for product water*≤ 2 ppb

Order number

Description

611CEL1

Arium® UV Lamp (185 | 254 nm), 1 pc-

Intended Use
Device type:
Arium® Pro UV and Pro VF* Feed water < 50 ppb TOC

18

Arium® Ultrafilter---

Ultrapure water, free from endotoxins, DNases
and RNases
High flow rates
Integrity tested
Long service lives
Certified quality

Description
The hollow-fiber ultrafilter utilizes crossflow technology
to reliably remove bacterial endotoxins, micro-organisms
and particles, as well as DNases and RNases from theultrapure water.
The filters are developed and produced in accordancewith a DIN EN ISO 9001|DIN EN 46 001 certified quality
assurance system which fulfills the prerequisites of the
Quality System Regulation (Regulation on Quality Systemsof the FDA) 21 CFR Part 820. During the manufacturing
process, every unit is integrity tested to meet the highest
quality standards and safety regulations.

Technical Specifications |Ordering Information
Materials
Membrane

Polysulfone

Composite material

Polyurethane (PUR)

Housing, caps

Polycarbonate (PC)

Plugs

Polypropylene (PP)

Efficient membrane surface

2.1 m2Max. operating pressure

3 bar at room temperature

Sterilization

200 ppm sodium hypochlorite,
45 min, max. 1 ×/week

Tank volume
Lumen

152 mlFiltrate side

306 mlReduction of nucleases
RNase concentration< 1 pg/ml

DNase concentration< 5 pg/ml

Retention of bacteria and endotoxins
Brev. diminuta

LRV 7 – 10

E. coli O55:B5 endotoxin

LRV > 3,5

Natural endotoxins

LRV > 3,0Dimensions of the fibers
Interior diameter

215 μmWall thickness

50 μm

Molecular weight cut-off(MWCO)

5,000 (= 5 kD)

Order number

Description

611CDU5

Arium® Ultrafilter, 1 pc-

Intended Use
Device type:
Arium® Pro VF and Pro UF

19

Arium® Cleaning Solution---

Effective cleaning for a long lifetime
Highly effective cleaning
Free of organic components
Surfactant-freeGentle on the materials

Description
With this cleaning agent, the regular removal of impurities
that develop during water purification is easy and effective,
this results in higher flow rates and longer service lives.
This highly effective agent is already filled in 50 mL syringes
and ready for direct connection.

Technical Specifications |Ordering Information
Ingredients

Order number

Description

Sodium hypochlorite

611CDS1

Arium® Cleaning Solution,
50 mL cleaning solution
filled into syringes, 1 pc-

Intended Use
Device type:
Arium® Pro UF and Pro VF

20

Sartorius Service
We Ensure the Quality of Your Results

Our Services at a Glance:

At Sartorius, quality products go hand in hand with professional service. With our wide service offering, we will
help guarantee the safe, reliable and optimal operation
ofyour Arium® systems. Just ask us and we will even cover
the entire life cycle of your laboratory water system – fromcommissioning to qualification to regular maintenance.
Together with you, we will ensure the consistently highquality of your laboratory water purification.

Installation and Commissioning
Your advantage: Your system will operate reliably at peak
performance from day one
Equipment Qualification (IQ | OQ)Your advantage: You will meet all regulatory requirements(GMP | GLP)
Regular Preventative Maintenance, Including Calibration,
inspection and testing of your system and exchange ofconsumables
Your advantages: Optimal operation of your system; reliable
results; prevention of downtime or even equipment failure
Get more information now at:www.sartorius.com/service

21Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Strasse 20
37079 Goettingen
Phone +49 551 308 0F or further information, visit
www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 01 | 2024

USA
Sartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906