CLAMP ON POWER LOGGER
PW3360Handy and Easy to Use– Power Management Support

Now with

QUICK SET

Convenience
Harmonic Measurement Model

PW3360-21

Reliable measurements start withproper wiring.
The QUICK SET function
guides you in making theright connections.

n See demand and trend graphs on siten Supports single to three-phase, 4-wire circuits- Simultaneously measure up to three single-phase,2-wire circuits (in the same power system).

n Measure up to 780V with a 1000V display range■ Broadly applicable for many jobs,
including leakage current measurement- An optional clamp-on leakage sensor supports
measurements as low as 50 mA.

■ Store months of data on SD cards

2

Begin with QUICK SET Convenience
Select your Wiring Type, Clamp and Destination, and Connect

Select wiring type (example: 3P4W) and connect
1

Connect the leads
to the PW3360-20.

Make proper connections simply by observing the colors ofthe displayed leads.

2 Connect the voltage clips.

▼

PASS

Double checks yourvoltage input and phase

Proceed to the next
step when PASS appears.

Check wire connection
status and judgment indicators.

3 Connect the clamp sensors.

FAIL
If FAIL appears, move the cursor
to the indicator and press the [ENTER] key.
Corrective action
tips appear

Select thecurrent range

Correct
Orientation

Miswiring Example (Clamp Orientation)
Power supply side

Wiring Screen Display Examples

FAIL

The I vector’s phase direction is
opposite the determination area.

PASS

The I vector’s phase direction iswithin the determination area.

►

Load side

Point the arrow toward the load side
Affected measurement values:

Changed I3 Clamp

Examples: P (Power) displayed value is too low P: 6.5kW

P: 20.6kW

3

Reveal Power Consumption State! Graph Display Functions■ Demand Graph Display

Shows the demand value transitions useful for managing power consumption. Check maximum demand values and times
while recording.

Evaluate Photovoltaic Generation Capabilities
Power Purchased

P dem+(kW)

Read values at cursor

Active power demand value (consumption) P dem+

Maximum
DemandValuesOne-day graph showing 48 thirty-minute intervals

Automatically refresheswith latest values

Time

Switched Display▼

P demkW■ Trend Graph Display

TimePower Sold

Active power demand value (regeneration) P dem-

Capture and record all fluctuations

From all measurement items, select one for display.
Check states such as power fluctuations of devices in
on-site operating conditions.
* Except for demand and harmonics

Read values at cursor

To conveniently record fluctuations even over long periods, select “All”
saving items to record maximum, minimum and average values withineach recording interval.
Continuous calculation at 200 ms intervals without gaps

Of the interval time
Maximum Value

Average Value

Minimum Value

Data interval (1s to 60min)

Maximum data

Maximum data
Average
data

Graph Display

Measured value(one selected item)

Graph showing intervals of up to 200 points

Automatically refresheswith latest values

Average
data
Minimum
data

Minimum data
Record 3 data points
per interval

Time

Record 3 data points
per interval

Create a Graph to Clearly Grasp Power Consumption
Record power consumption on an
SD Card* at specific intervals. Loadthe data into the PC.

Record to
SD cardUse Excel graph processing for
before and after comparisons.
Before and After Air Conditioning
and Power Consumption
Measurements

Before Power consumption (W)After Power consumption (W)Before Power conservation (Wh)After Power conservation (Wh)Time

* Store up to one year’s data acquired at one minute intervals. Performance cannot be guaranteed on storage media other than Hioki-specified SD card options.

4

Accommodates All Worksites■ Tight spaces

48 mm

Compact(1.89")

100 mm

(3.94")

Easy to loop around,even in confined spaces

180 mm (7.09")

In dim environments
Easy-to-see color LCD

AC FLEXIBLE CURRENT SENSOR
CT9667-01, -02 (Sold separately)■ Where no AC power is available
Battery installation

Battery* power provides about eight hours of continuous operation. In addition, a Voltage Line Power Adapter* is available topower the PW3360-20 from the measurement lines.
* Battery Set PW9002 and Voltage Line Power Adapter PW9003 options
are sold separately.

Battery Set PW9002

8-Houroperation►

Voltage LinePower Adapter
Power Source
Obtains power from the measurement lines■ In severe temperature environments
The operating temperature range extends from –10°C (14°F)
to 50°C (122°F).
Even under battery operation, measurements can be
performed from 0 °C (32°F) to 40°C (104°F)(0°C (32°F) to 50°C (122 °F) when using LAN communication).

50˚C

-10˚C

■ Magnetic voltage adapters forhard-to-clip terminals
Magnetic voltage adapters convertible with the Voltage
Cords L9438-53 let you accurately detect voltage when thecircuit terminals are too shallow for alligator clips to latch on.
* Magnetic Adapter 9804 option sold separately.

9804-01 Magnetic Adapter (red) usage example
Magnet in the tip

Generally compatible with M6 pan screws

Loaded with More Useful Functions
Pulse Input

Simultaneous Measurements

The pulse input function can be used to record power data andproduction volume counts simultaneously. The power data andpulse volume (production volume) information are useful for unit
cost production management.

Simultaneously measures three single-phase2-wire circuits in the same system.

Load

Load

Power Source

LoadPower Source

Load

t

Pulse Input

Pulse Output
Use the Pulse Output function to acquire temperature and pulse (electrical energy) data simultaneously with a data logger.
Evaluate the relationship between air conditioner temperature control settings and power consumption.

LoadPower Source

Application Example

Temperature
Measurement

Data Logger LR8450

The pulse signal is output whenever
power consumption exceeds a set
output rate (from 1 Wh to 100 kWh).

Voltage Input
Temperature (thermocouple) Input
Pulse Input

Pulse Output

The LR8450 cannot perform measurementon their own. One or more plug-in unitsare
required (sold separately).

►

Leakage Current Measurement
With the optional leakage current clamp on sensors, turn the instrument into a 3-channel leakage current logger to help
identify trouble spots.
As a 3-channel Leakage Current Logger

OptionsLoad

φ30 mm(1.18 in)

Load

Power Source

Leak Clamp on Sensor
φ40 mm(1.57 in)

Load

9675

9657-10

Ideal for quick investigation of intermittent leakage by continuous calculation processing every 200 ms.
(Select to save the average, maximum and/or minimum at every interval.)

5

6

Harmonic Measurement Model

PW3360-21

Analyze voltage and current harmonics on a 50/60 Hz power linefrom the fundamental waveform to the 40th order.
• Displays the RMS value, content, and phase angle (numerical list or graph display)for each harmonic order.
• Vector display of power phase angle(vector display)

Harmonic graph screen

Harmonic power phase angle graph screen

Maximum, average, and minimum values
can be saved in binary format to SD card ateach interval.
Power Logger Viewer SF1001 is required to display the data on a PC.

SF1001 Display Example

Harmonic Time Series Display
Select and display a time series graph offundamental, third- and fifth-order current
harmonics.

PC

Processing

Power Logger Viewer SF1001 (option, sold separately)

Data saved to an SD card or internal memory can be loaded into a PC for expanded display, aggregation and analysis.
Supported models: PW3360, PW3365, 3169-20

On the same time axis, view measured power consumption and equipment operating status at specific intervals,
along with equipment characteristics and management
details.
Simultaneously measure and record separate
loads using three PW3360s
Air Conditioner A

Power Source
Air Conditioner B
Power Source
Air Conditioner C● Trend graph display function ● Summary display function ● Waveform display● Harmonic display ● Copy function ● Print function ● Report printing

Stacked Graph Display Example
Maximum Demand Values

Power Source

7

Get results from the job site in real-time

Present data from multiple sources as a graph or list together in real-time
PC Application Program

GENNECT One

Simultaneously monitor all data in real-timeConnect measuring instruments to PC with LAN cable

Download the GENNECT One SF4000 softwarefrom the Hioki website to your PC

Operation guaranteed for up to 15 units. Please contact your nearest Hioki distributor
for connections exceeding 15.

HUB

Software automatically recognizes LAN-connected measuring
instrument

Connect each measuring
instrument to PC with LAN cable

Display acquired data as graphs in real-time

The measured value (present value) displayed by the measuring instrument is obtained
at a certain interval (minimum 1s interval) according to the timer on the PC.

Operate measuring instruments connected via LAN from a PC
Automatically transfer files saved on a LAN-connected measuring
instrument to a PC
Power Quality Analyzer
PQ3100, PQ3198

Memory HiLogger
LR8450(Firmware V1.20 or later)

Clamp On Power Logger
PW3360, PW3365

Entire screen

1. Monitor

Manage and save results with software
List MAX, MIN and AVG values (Display time of MAX & MIN data)

2. Graph

3. ListPW3360 Voltage value

PW3360 Current value

12

LR8400 Temperature value

3

PW3360 Power factor value
LR8400 Voltage value

1. Monitor display (Max 512 items)

Display each measured data in real-time

2. Graph display (Max 32 items)

logging time (display up to 1024 points)

Display selected data as graphs

3. List display (Max 32 items)
Display selected data in list

LAN remote control function

LAN automatic file download function

The application displays
a virtual instrument andallows you to control it
directly with the mouse.
You can also easilychange instrument
settings and control
the instrument, forexample to start and stopmeasurement.

This function lets you acquire data in real time on a PC, including data
created when the instrument’s trigger is activated and measurement files
that are automatically generated on a daily basis. Example uses include
capturing abnormal phenomena with an instrument installed in the field andautomatically acquiring daily power consumption data on a PC.
Automatically transfer measurement files to a PC.
Review measured
values on a PC
anytime.

Downloading GENNECT One SF4000
HIOKI website > Search

Model No. (Order code)

SF4000

SearchEnter the model number in the search field to download the software to get started!
Compatible instruments
POWER QUALITY ANALYZER PQ3100, PQ3198
CLAMP ON POWER LOGGER PW3360, PW3365
POWER ANALYZER PW3390, PW6001
MEMORY HiLOGGER LR8450, LR8450-01
WIRELESS LOGGING STATION LR8410
MEMORY HiCORDER MR6000

Available items to monitor and save on PC Number of items that can be saved
Voltage
Current
Power

Temperature
Analog Input

Instantaneous value of eachinterval; MAX, MIN, AVG valueof each interval

Instantaneous valueof each interval

Save up to 512 items*Maximum 32 items whensimultaneously displaying graphs

Recording time

When memory size of acquired data reaches to
64MB, data will be separated automatically[Continuous measurement]
When storage capacity falls below 512MB,
measurement will stop

8

n PW3360-20, PW3360-21 Specifications

Specifications in orange available in Model PW3360-21 only

(Accuracy guaranteed for 1 year, Post-adjustment accuracy guaranteed for 1 year)

Input specifications
Measurement
line typeMeasurement
line Frequency
Number of input
channels
Voltage range

Single-phase 2-wire, single-phase 3-wire, three-phase 3-wire,
three-phase 4-wireMeasurement items
Voltage

RMS value, fundamental wave value,waveform peak (absolutevalue), fundamental wave phase angle, frequency (1)RMS value, fundamental wave value,waveform peak (absolutevalue), fundamental wave phase angle
Active power, reactive power (with lag/lead display), apparent
power, power factor, (with lag/lead display) or displacement
power factor (with lag/lead display), active energy (consumption,
regeneration, regeneration), reactive energy(lag, lead)
Energy cost display (per-kWh price × power consumption)
Active power demand value (consumption, regeneration),reactive power demand value (lag, lead),
active power demand quantity *(consumption, regeneration),reactive power demand quantity *(lag, lead),
power factor demand value, pulse input* Only data output to SD card
Harmonic voltage, current, power level, content , phase angle
Total harmonic distortion factor (THD-F or THD-R)

Current

50/ 60 Hz

Voltage: 3 channels U1 to U3Current: 3 channels I1 to I3
600 V AC
Total display area: 5V to 1000 V (less than 5 V displays as 0 V)
When RMS voltage is zero, zero is displayed for all orders ofharmonic voltage.
Effective measurement range: 90 V to 780 V, peak: ±1400V[OVER] indicates over-range warning
Current ranges
Load current
CLAMP ON SENSOR 9694
: 500 m/1/5/10/50 A
CLAMP ON SENSOR 9695-02: 500 m/1/5/10/50 A
CLAMP ON SENSOR 9660
: 5/10/50/100 A
CLAMP ON SENSOR 9695-03: 5/10/50/100 A
CLAMP ON SENSOR 9661
: 5/10/50/100/500 A
CLAMP ON SENSOR 9669
: 100/200/1 k A
AC FLEXIBLE CURRENT SENSOR CT9667-01 : 50/100 /500/1 k/5 kA
AC FLEXIBLE CURRENT SENSOR CT9667-02 : 50/100 /500/1 k/5 kA
AC FLEXIBLE CURRENT SENSOR CT9667-03 : 50/100 /500/1 k/5 kA
Leakage current
LEAK CLAMP ON SENSOR 9657-10 : 50 m/100 m/500 m/1/5 A
LEAK CLAMP ON SENSOR 9675
: 50 m/100 m/500 m/1/5 A
Total display range: Within 0.4 to 130% of the range(zero is suppressed for less than 0.4%)
When RMS current is zero, zero is displayed for all orders ofharmonic current.
Effective measurement range: Within 5 to 110% of the range
peak: ±400% of range, however, maximum range is 200%.
[OVER] indicates over-range warning
300.00 W to 9.0000 MW
Power ranges
Depends on voltage/current combination and measured line
type (see Measurement Range Configuration Tables)
Total display range: Within 0 to 130% of the range(“0W” display indicates zero rms voltage and/or current)
When RMS voltage and current are zero, zero is displayed
for all orders of harmonic active power and harmonic reactive
power.
Effective measurement area: Within 5 to 110% of the range
Any (0.01 to 9999.99)
VT ratio settings
Selections (1/60/100/200/300/600/700/1000/2000/2500/5000)
Any (0.01 to 9999.99)
CT ratio settings
Selections (1/40/60/80/120/160/200/240/300/400/600/800/1200)
Voltage: Insolated inputs (except between U1, U2, U3 and N)
Input methods
Current: Isolated input using a clamp-on sensor
Voltage input part: 3 MΩ ±20% (50/ 60 Hz)
Input resistance
Maximum rated voltage Voltage input section: 1000 VAC, 1400 Vpeak
Current input section: 1.7 VAC, 2.4 Vpeak
between terminals
Voltage input section: 600V Measurement Category III
Maximum rated
300V Measurement Category IV
voltage to earth
Current input section: Depends on clamp sensor in use.

Pulse input

Input specifications No-voltage contact input (counts when shorted terminals open)
Voltage input (Hi: 2 V to 45 V, Lo: 0 V to 0.5 V, counts at Lo to Hi)
Maximum rated input between terminals: 45 V DC
Maximum rated input to ground: not isolated (GND is equipment common)
Measurement range 0 to 9999 (maximum pulse count per save interval)
Filter On (for mechanical contacts) 25 Hz or less, and at least 20
Filterms Hi and Lo pulse width
Filter Off (for solid-state contacts) 5 kHz or less, and at least 100µs Hi and Lo pulse width
Displays product of pulse count and scaling factor setting
Scaling
Setting ranges: 0.001 to 1.000, and 1.000 to 100.00

Power

DemandHarmonic

Measurement screen
List

Voltage RMS value, current RMS value, frequency, total active power,
total reactive power, apparent power, power factor or displacement
power factor, active energy (consumption), elapsed time
Voltage RMS value, voltage fundamental wave value,voltage
waveform peak, voltage fundamental wave phase angle,
current RMS value, current fundamental wave value, current waveform peak, current fundamental wave phase angle
Per-channel and total active power, apparent power, reactive
power,power factor or displacement power factor
Active energy (consumption, regeneration), reactiv energy (lag,lead),recording start time, recording stop time, elapsed time, energy cost
Active power demand value (consumption, regeneration), reactive power demand value (lag, lead),
power factor demand value, or pulse input
Displays the maximum active power demand value and the timeat which it occurred (this information is not saved).
(data from up to 48 intervals is internally stored, then refreshed oldest-first).
Graph (voltage, current and power levels, content percentage and phase angle)
List (voltage, current and power levels, content percentage and phase angle)
Displays voltage and current waveform, voltage and current
RMS values, and frequency.
With a 3P3W3M connection, displays the phase voltage waveform from the virtual neutral point.
Enlarged view of 4 user-selected parameters
For one selected measurement item (except demand and harmonics), displays maximum, average and minimum values, with cursor
calculations available (Note: with Trend display, there is no poweroff backup function).

U/I

Power
Integ
DemandHarmonic
Waveform

Zoom

Trend

External interfaces Specifications

SD card Interface Settings data, measurement data, screen data, waveform data
100BASE-TX IEEE802.3 Compliance
LAN interface- HTTP server function- FTP server function
USB Ver 2.0, Windows 10 (32/64bit)/ Windows 8 (32/64bit)/
USB interface
Windows 7 (32/64bit) / Vista (32bit) /XP- When connected to a computer, the SD Card and internal
memory are recognized as removable storage devices.

Pulse output
Function

Pulse rate
Pulse width
Output signal

Output pulse rate is proportional to active power consumption(WP+) when measuring integral power consumption
OFF/ 1 Wh/ 10 Wh/ 100 Wh/ 1 kWh/ 10 kWh/ 100 kWh/ 1000 kWh(Default: 1 kWh)approx. 100 ms
Open-collector 30 V, 5 mA max (photocoupler isolated)
Active Low

WIRE SPECIFICATIONS

Pulse input terminals

Pulse output terminals

Pulse output device

Pulse counter

Electric wires that conform with:single line: φ0.65 mm (AWG22)
twisted wire: 0.32 mm2 (AWG22)
strand diameter: φ0.12 mm or more
Supported electric wires:single line: φ0.32 mm to φ0.65 mm (AWG28 to AWG22)
twisted wire: 0.08 mm2 to 0.32 mm2 (AWG28 to AWG22)
strand diameter: φ0.12 mm or more
exposed wire length: 8 mm

9Specifications in orange available in Model PW3360-21 only
General Specifications

Display device

Operatingenvironment
Operatingtemperature andhumidity(no condensation)
Storagetemperature andhumidity(no condensation)
Dielectric strength
Applicable standards
Power supply
Charge function
Maximum rated
power
Continuous
battery
operation time
Backup battery life
Dimensions
Mass
Accessories

3.5 inch TFT color LCD (320 × 240 pixel)
Japanese, English, Chinese, Korean, German, Italian, French,
Spanish, Turkish
Backlight auto-off function (after 2 minutes)
When AUTO OFF is active, the Power LED blinks
Indoors, Pollution degree 2, altitude up to 2000 m (6562-ft.)-10°C to 50°C (14°F to 122°F), 80% RH or lessDuring LAN communication: 0°C to 50°C (32°F to 122°F), 80% RH or lessDuring battery operation: 0°C to 40°C (32°F to 104°F), 80% RH or lessDuring battery charging: 10°C to 40°C (50°F to 104°F), 80% RH or less
-20°C to 60°C (-4°F to 140°F), 80% RH or lessHowever, the batteryُs storage temperature range is -20°C to
30°C (-4°F to 86°F), 80% RH or less4.29 kVrms AC (1 mA sense current) between voltage input terminals and external terminals, 50/ 60 Hz for 60 sec.
Safety: EN61010, EMC: EN61326, EN61000-3-2, EN61000-3-3•Z1006 AC Adapter (12 V, 1.25 A), Rated supply voltage 100 VAC
to 240 VAC, Rated power supply frequency 50/60 Hz•Model 9459 Battery Pack (Ni-MH DC7.2 V 2700 mAh)
Charges the battery regardless of whether the instrument is on or off.
Charge time: Max. 6 hr. 10 min. (reference value at 23°C)•When the Z1006 AC Adapter is used: 40 VA (including AC adapter),
13 VA (PW3360-20 instrument only)•When the 9459 Battery Pack is used: 3 VA
Approx. 8 hr. (Continuous, backlight off)
(when using the battery pack)
Clock and settings (Lithium battery), Approx. 10 years @23°C (@73.4°F)
Approx. 180W(7.09") × 100H(3.94") × 48D (1.89") mm (without PW9002)
Approx. 180W(7.09") × 100H(3.94") × 68D (2.68") mm (with PW9002)
Approx. 550g (19.4 oz) (without PW9002), Approx. 830g (29.3 oz) (with PW9002)
Voltage Cord L9438-53(1 set), AC Adapter Z1006 (1),USB cable(1), instruction manual (1), measurement guide (1),Color clip ×1 set: red, yellow, blue, white/two each, for color-coding clamp
sensors, Spiral tubes for grouping clamp sensor cords ×5

Harmonic Specifications (PW3360-21 only)

IEC61000-4-7:2002 compliant, but without interharmonics
Standard
10 cycles at 50 Hz, and 12 cycles at 60 Hz (with interpolation)
Window width
Rectangular, 2048 points
Points per window
Up to the 40th order
Analysis orders
THD calculation selection THD-F/THD-R
Harmonic level: Voltage, current and power levels for each harmonic
Analysis items(U12 and I12 obtained by calculation of the third channel in 3P3W2M
wiring are not displayed. Phase voltage is used for 3P3W3M wiring.)
Harmonic content: Voltage, current and power contents for each harmonic
Harmonic phase angle: Voltage, current and power phase angles for eachharmonic
Total harmonic distortion factor: Voltage and current (THD-F or THD-R)
Measurement
Harmonic level
1st to 15th orders: ±5% rdg. ±0.2% f.s.
accuracy
16th to 20th orders : ±10% rdg. ±0.2% f.s.
21st to 40th orders : ±20% rdg. ±0.3% f.s.
For voltage and current, add accuracy of clamp sensor.
Harmonic power phase angle
1st to 3rd orders: ±3°+clamp sensor accuracy
4th to 40th orders : ±0.1°×k±3°+clamp sensor accuracy
For each harmonic order at 6 V, harmonic current level isregulated at 1% f.s.
Total harmonic distortion factor: Accuracy unspecified

n POWER LOGGER VIEWER SF1001 Specifications
General Specifications
Supported models

PW3360-20, PW3360-21, PW3365, 3169-20, 3169-21
LR5000 series; Data previously loaded by the LR5000 Utility (.hrp2 format) using a PC

Supported computer Windows 8/8.1 (32/64bit), Windows 7 SP1 or later (32/64bit)
Windows Vista SP2 or later (32bit), Windows XP SP3 or later (32bit)
operating systems(Accuracy guaranteed for 1 year, Post-adjustment accuracy guaranteed for 1 year)

Measurement Specifications
Connection

Single-phase 2-wire (1P2W, 1P2W 5 2 circuits, 1P2W 5 3 circuits)
Single-phase 3-wire (1P3W, 1P3W+I, 1P3W1U, 1P3W1U+I)
Three-phase 3-wire (3P3W2M, 3P3W2M+I, 3P3W3M)
Three-phase 4-wire (3P4W), Current only: 1 to 3 channels

Simultaneous
1P3W+I: 1 power circuit and 1 current channel
power/current
measurement modes 3P3W2M+I: 1 power circuit and 1 current channel
Power factor, reactive and apparent power: rms calculation/ fundaCalculation
mental wave calculation
selection

Measurement
accuracy

Voltage: ±0.3% rdg. ±0.1% f.s.
Current: ±0.3% rdg. ±0.1% f.s. + clamp sensor accuracy(50/ 60Hz,
Active power: ±0.3% rdg. ±0.1% f.s. +clamp sensor accuracy
power factor = 1) Clamp-On Sensor 9661 accuracy: ±0.3% rdg. ±0.01% f.s.

(Accuracy depends on clamp sensor. See page 10 for the accuracy ofeach model, and page 11 for combined accuracy of Model PW3360-20
and each clamp sensor.)
Approx. 0.5 sec (except when accessing SD card or internal memory,
Display update rate or during LAN/USB communication)
However, approx. 1 s for power-related values
Digital sampling and zero cross synchronization calculation method
Measurement
Sampling: 10.24 kHz (2048 points)
methodCalculation processing
50 Hz: Continuous, gapless measurement at 10 cycles
60 Hz: Continuous, gapless measurement at 12 cycles
A/D converter resolution 16bit

Functions Specifications

Trend graphdisplay function

Summary display
function

Waveform display

Recording Specifications

Save destination SD Card, internal memory (capacity: approx. 320 KB)
Save interval time 1/2/5/10/15/30 seconds, 1/2/5/10/15/20/30/60 minutes* Available storage time is displayed on PW3360-20's setting screen
Save items
Measurement save: Average only / all (average, maximum, minimum)
Harmonic data save: Binary format (average, maximum andminimum)
Screen save: ON/OFF Saves the displayed screen as a BMP at a
fixed interval. (The minimum interval time for saving screen copies is 5 min. If the setting is less than 5 min., screen copies will besaved every 5 min.)Waveform save: Stores binary waveform data (with shortest interval 1 minute). When set to less than 1 minute, waveforms are saved
once every minute
Recording start methods Interval time, manual, specified time, repeat: Record period(00:00 to 24:00) ·Segment folder(off/day/week/month)
Recording stop methods Manual, specified time, timer, repeat(up to one year)

Harmonic display
Copy function

Print function

Report printing

Display items: Voltage, current, active power, reactive power,
apparent power, power factor, frequency, integrated active
power, integrated reactive power, demand volume, demandvalue, voltage disequilibrium factor, pulse, harmonics (level,
content, phase angle, total value, THD)
Stacked bar graph display: Up to 16 types of data series canbe displayed in an overlay graph
Cursor measurements: Measurement values can be displayed by the cursor
Displayed items are the same as for the trend Graph Display
Daily, weekly and monthly report displays: Accumulates anddisplays daily, weekly and monthly reports over specified period.
Load factor calculation display: Calculates and displays load factor
and demand factor results with daily, weekly and monthly reports
Time span aggregation: Aggregates data into up to four
specified time spans
CO2 equivalent display: Uses the specified conversion rate
to display CO2 equivalent values (reference values).
Displays waveform data at specified date and time
List display: Displays a list of harmonic data at specified date and time
Graph display: Displays a bar graph of harmonic data at
specified date and time
Cursor calculation: Calculates measurement data at cursors
in waveform and graph displays
Captures any display image to the clipboard
Preview and print content shown on the trend graph, report,
harmonic graph and settings displays.
Comment entry (Text comments can be entered in any printout)
Header/Footer settings: Sets the header and footer for each printout
Printing support: Any color or monochrome printing supported by theoperating system
Print (static) contents over a specific time period
Output contents: Standard or selected output items
Available output items: Trend graph, summary, daily report,
harmonic list, harmonic graph, waveform
Report creation method: Standard print
Report output settings: Save/load report output settings

10n CLAMP SENSOR Specifications
CLAMP ON SENSOR

9694

9660

9661

9669

9695-02

Insulated conductor

Not CE marked

Not CE marked

Appearance

Measurable conductor
diameterPrimary current rating
Accuracy

CONNECTION CORD 9219
Cord length: 3 m(9.84ft)

Cord length: 3 m(9.84ft)

Cord length: 3 m(9.84ft)

φ15 mm (0.59")

φ15 mm (0.59")

φ46 mm (0.81")

5 A AC±0.3% rdg.
±0.02% f.s.
Within ±2˚

100 A AC
±0.3% rdg.
±0.02% f.s.
Within ±1˚

500 A AC
±0.3% rdg.
±0.01% f.s.
Within ±0.5˚

Amplitude (45 to 66 Hz)
Phase (45 Hz to 5 kHz)Frequency characteristic
Effect of external magnetic field(with a magnetic field of 400 A/ m AC)

Cord length: 3 m(9.84ft)

Equivalent to 0.1 A or lessWithin ±1.5%
CAT III 600 Vrms
1000 A continuous

Within ±0.5%
CAT III 300 Vrms
60 A continuous 130 A continuous

5 21D (0.83") mm

5 21D (0.83") mm

5 42D (1.65") mm

99.5W (3.92") 5 188H (7.40")
5 42D (1.65") mm

50.5W (2.28") 5 58H (2.28")
5 18.7D (0.74") mm

230 g (8.1 oz)

230 g (8.1 oz)

380 g (13.4 oz)

590 g (20.8 oz)

50 g (1.8 oz)

φ100 mm(3.94")

Phase

Frequency characteristic

CT9667-03

Effect of external magnetic field

1.5% / f.s. or less.

(with a magnetic field of 400 A/ m AC)

Effect of conductor position
Maximum rated voltage to earth
Maximum input
Circuit box

Sensor cable diameter

MassPower supply

φ180 mm
φ254 mm(7.09")(10.00")500 A AC / 5000 A AC
±2.0% rdg. ±0.3% f.s.
Within ±1˚Within ±3 dB

10Hz to 20kHz (deviation from accuracy)

Dimensions

100 A AC
±0.3% rdg.
±0.02% f.s.
Within ±1˚Equivalent to 1 A or less

CLAMP ON LEAK SENSOR (Leakage Current Measurement Only)
9657-109675

Within ±3.0%
CAT III 1000 Vrms, CAT IV 600 Vrms

Cord length: 3 m(9.84ft)

Cord length: 3 m(9.84ft)(deviation from accuracy)

Effect of external magnetic field

35W (1.38") × 120H (4.74") × 34D (1.34") mm
φ13 mm (0.51”)

280 g (9.9 oz.)
470 g (16.6 oz.)
LR06 alkaline battery × 2 (continuous operation max. 7 days)or AC ADAPTER 9445-02/9445-03 (optional)

Available Recording Time

Insulated conductor

φ40 mm (1.57")
Measurable conductor diameter
10 A AC*Primary current rating
Accuracy Amplitude (45 to 66 Hz) ±1.0% rdg. ±0.05% f.s.
Phase angle (@50 or 60 Hz)
Within ±3˚Frequency characteristic
Within ±5%40 Hz to 5 kHz

10000 A continuous
φ7.4 mm (0.29”)

Insulated conductor

Appearance

Cord length : Sensor - circuit: 2 m (6.56ft)
Circuit - connector: 1 m (3.28ft)(45 to 66Hz)

50 A AC±0.3% rdg.
±0.02% f.s.
Within ±2˚Equivalent to 0.1 A or lessAppearance(45 to 66Hz)

φ15 mm (0.59")

Within ±1.0%

AC FLEXIBLE CURRENT SENSOR
CT9667-01 CT9667-02

Primary current rating
Amplitude
Accuracy

Cord length: 3 m (9.84ft)

φ15 mm (0.59")

Within ±2.0%

Within ±0.5%
Effect of conductor position
CAT III 300 Vrms
CAT III 600 Vrms
Maximum rated voltage to earth CAT III 300 Vrms
130 A continuous
550 A continuous
Maximum input (45 to 66Hz) 50 A continuous
46W (1.81") 5 135H (5.31") 46W (1.81") 5 135H (5.31") 77W (3.03") 5 151H (5.94")
Dimensions

Measurable conductor diameter

φ55 mm (2.17"),80 (3.15")×20 (0.79") mm
1000 A AC±1.0% rdg.
±0.01% f.s.
Within ±1˚Connect with the9695-02/-03,
Output BNC terminal

Within ±1.0%

40Hz to 5kHz
(deviation from accuracy)

Mass9695-03

Insulated conductor(with a magnetic field of 400 A/ m AC)

Effect of conductor position
Measurable conductor
Maximum input(45 to 66Hz)

Dimensions
Mass
Notes

φ30 mm (1.18")
10 A AC*±1.0% rdg. ±0.005% f.s.
Within ±5˚Within ±5%

7.5 mA max.

7.5 mA max.

Within ±0.1%
Insulated conductor

Within ±0.1%
Insulated conductor

30 A continuous

10 A continuous

74W (2.91") × 145H (5.71") 60W (2.36") × 112.5H (4.43")
× 42D (1.65")
× 23.6D (0.95")

380 g (13.4 oz)
160 g (5.6 oz)
Not used for power measurements* Maximum AC measurement range with PW3360-20 is 5 A.

PW3360-20 and PW3360-21 with Z4001 2-GB SD card, measuring 3P3W2M wiring

Saved Items: ALL data (Saves all data: average, maximum, and minimum values)
Screen save: OFF Waveform save: OFF

Save Time
PW3360-20
PW3360-21
PW3360-21
Interval time
(Saving of harmonic (Saving of harmonic
data: OFF)
data: ON)
1 seconds
15.9 days
24.7 hours
2 seconds
31.9 days
2.1 days
5 seconds
79.7 days
5.1 days
10 seconds
159 days
10.3 days
15 seconds
242 days
15.4 days

Save Time
PW3360-20
PW3360-21
Interval time
PW3360-21(Saving of harmonic (Saving of harmonic
data: OFF)
data: ON)30s
1 year30.8 days
1 minutes
1 year
61.7 days
2 minutes
1 year123 days
5 minutes
1 year308 daysMore than
1 year
1 year10 minites

The maximum recording time based on the settings can be confirmed right on the Settings
screen.

In any case, the maximum file size for measurement data is about 200 MB. When thisis exceeded, a new file is created and saving
continues.
<NOTE>
Regardless of the settings, the maximum savetime of the PW3360-20, PW3360-21 is one year.

11

n Measurement Range Configurations
Current

CLAMP ON SENSOR 9694 (CAT III 300 V) *1
CLAMP ON SENSOR 9695-02 (CAT III 300 V)
500.00 mA
1.0000 A
5.0000 A
10.000 A
50.000 A
300.00 W
600.00 W
3.0000 kW
6.0000 kW
30.000 kW

Voltage

Connection
1P2W
1P3W
1P3W1U
600.00 V
3P3W2M
3P3W3M
3P4W

600.00 W

1.2000 kW

6.0000 kW

12.000 kW

60.000 kW

900.00 W

1.8000 kW

9.0000 kW

18.000 kW

90.000 kW

Total display range

Voltage is displayed from 5 V to 1000 V, with less than
5 V displayed as 0 V.
Current is displayed from 0.4% to 130% of the selected
range, with less than 0.4% displayed as 0 A
Power is displayed from 0 to 130% of full scale, with
0 W displayed when voltage or current is zero.
The range configurations for apparent power (S) andreactive power (Q) are the same, with units of [VA] and[var], respectively.

*1. For the 9694 sensor, the range of guaranteed accuracy is from 500 mA to 5 A, and for the 9695-02, from 500 mA to 50 A.

Current

CLAMP ON SENSOR 9660, 9695-03 (CAT III 300 V) *2
CLAMP ON SENSOR 9661
5.0000 A
10.000 A
50.000 A
100.00 A
3.0000 kW
6.0000 kW 30.000 kW
60.000 kW

Voltage

When VT and CT ratios are set, the range configurationis the product (VT ratio × CT ratio).

Connection
500.00 A
300.00 kW
1P2W
1P3W
1P3W1U
6.0000 kW
12.000 kW 60.000 kW
120.00 kW
600.00 kW
600.00 V
3P3W2M
3P3W3M
9.0000 kW
18.000 kW 90.000 kW
180.00 kW
900.00 kW
3P4W*2. For the 9660 and 9695-03 sensors, the range of guaranteed accuracy is from 5 A to 100 A, and for the 9661, from 5 A to 500 A.
Current
CLAMP ON SENSOR 9669
Voltage

Connection
1P2W
1P3W
1P3W1U
600.00 V
3P3W2M
3P3W3M
3P4W

100.00 A
60.000 kW
120.00 kW

200.00 A
120.00 kW
240.00 kW

1.0000 kA
600.00 kW
1.2000 MW

180.00 kW

360.00 kW

1.8000 MW

AC FLEXIBLE CURRENT SENSOR CT9667-01, -02, -03
500/5000 A
500 A range
5000 A range
range
Voltage Connection 50.000 A 100.00 A 500.00 A 1.0000 kA 5.0000 kA
30.000 kW 60.000 kW 300.00 kW 600.00 kW 3.0000 MW
1P2W

Current
Range

1P3W1U

3P3W2M
3P3W3M
3P4W

Display range
650 A
Effective measurement range
2A
25 A
550 A
130 A
Effective measurement range
5A
110 A
65 A
Effective measurement range

100 A
0.4 A

50 A

60.000 kW 120.00 kW 600.00 kW 1.2000 MW 6.0000 MW
90.000 kW 180.00 kW 900.00 kW 1.8000 MW 9.0000 MW

0.2 A

2.5 A

0A

1P3W

600.00V

Range

50.000 mA/100.00 mA/500.00 mA/1.0000 A/5.0000 A

Combined accuracy of PW3360-20 + clamp sensors
Range
9694
9695-02―±0.6% rdg. ±0.12% f.s.
50.000 A―±0.6% rdg. ±0.2% f.s.
10.000 A±0.6% rdg. ±0.12% f.s.
±0.6% rdg. ±0.3% f.s.
5.0000 A±0.6% rdg. ±0.2% f.s.
±0.6% rdg. ±1.1% f.s.
1.0000 A±0.6% rdg. ±0.3% f.s.
±0.6% rdg. ±2.1% f.s.
500.00 mA
Range

500.00 A
100.00 A
50.000 A
10.000 A
5.0000 A

9660, 9695-03―±0.6% rdg. ±0.12% f.s.
±0.6% rdg. ±0.14% f.s.
±0.6% rdg. ±0.3% f.s.
±0.6% rdg. ±0.5% f.s.

Range

9669
±1.3% rdg. ±0.11% f.s.
±1.3% rdg. ±0.15% f.s.
±1.3% rdg. ±0.2% f.s.

1.0000 kA
200.00 A
100.00 A

Range

5.0000kA
1.0000kA
500.00A

100.00A
50.000A

9661±0.6% rdg. ±0.11% f.s.
±0.6% rdg. ±0.15% f.s.
±0.6% rdg. ±0.2% f.s.
±0.6% rdg. ±0.6% f.s.
±0.6% rdg. ±1.1% f.s.

-01

CT9667-02-03 5000A range±2.3% rdg. ±0.4% f.s.
±2.3% rdg. ±1.6% f.s.
±2.3% rdg. ±3.1% f.s.
─
─

-01

CT9667 -02
500A range-03──
±2.3% rdg. ±0.4% f.s.
±2.3% rdg. ±1.6% f.s.
±2.3% rdg. ±3.1% f.s.

10 A

100 A

1000 A
13 A

10 A

0.04 A

0.5 A

5A

Effective measurement range

Effective measurement range

0A

100 mA

6.5 A

11 A

5.5 A

0.25 A

1A

10 A

100 A

Input Current

n Measurement accuracy±0.3% rdg. ±0.1% f.s.
Voltage±0.3% rdg. ±0.1% f.s. + clamp sensor accuracy
Current
Active power ±0.3% rdg. ±0.1% f.s. + clamp sensor accuracy (power factor = 1)

55 A

1A

0.02 A

Leak current: CLAMP ON LEAK SENSOR 9657-10, 9675

For voltage, 90 to 780 V, with max. 1400 V peak.
For current, 5% to 110% of the selected range withpeak ±400% of range, but maximum range is ±200%.
For power, 5% to 110% of the selected range.
For frequency, 45 to 66 Hz.

n Current Display and Effective Measurement Ranges (typical)

500 A

Current

Effective measurement range

Conditions of guaranteed accuracy After 30 minute warm-up, with 50/60 Hz sine wave input

Temperature and humidity 23˚C ±5˚C (73 ± 9°F), 80%RH or lessfor guaranteed accuracy (applies to all specifications unless otherwise noted)
Display area
Effective measurement rangeof guaranteed accuracy

Within ±0.3 sec/day (at power ON, 0°C to 50 °C)
Within ±0.5 sec/day (at power ON, -10°C to 0 °C)
Temperature characteristic Within ±0.1% f.s./ ˚C (except 23 ±5°C)
Within ±0.2% f.s.
Effect of common mode
(600 V AC, 50/60 Hz, between voltage input terminal and case)voltage
Effect of external magnetic field Within ±1.5% f.s. (in a magnetic field of 400 A/m rms AC, 50/60 Hz)
Effect of phase
Phase accuracy ±1.3° equivalent (with 50/60 Hz f.s. input)
Real-time clock accuracy

Apparent power
Reactive power±1 dgt. for the calculation obtained from each measurement value
Fundamental waveform calculations±0.3% rdg. ±0.1% f.s. + clamp-on sensor accuracy (w/power factor = 1)
Rms calculations
From each measurement applied to calculation ±1 dgt.
Active and reactive power measurement accuracies ±1 dgt.
EnergyFrom each measurement applied to calculation ±1 dgt.
Power factor±0.5% rdg. (with 90 to 780 V sine wave input)
Frequency
Active and reactive power measurement accuracies ±1 dgt.
Demand value
Active and reactive power measurement accuracies ±1 dgt.
Demand quantity±1 dgt. for the calculation obtained from each measurement value
Pulse input
Frequency characteristic At 50/60 Hz fundamental waveform frequency,
up to 1 kHz, ±3% rdg. ±0.2% f.s.
up to 3kHz, ±10% rdg. ±0.2% f.s.
For current and active power, add clamp-on sensor accuracy.
Note: only for 3P3W3M wiring, add ±0.5% rdg.

Current sensors : Sold separately

Model : CLAMP ON POWER LOGGER PW3360

Model No. (Order Code) (Note)(English model, main unit only)
PW3360-20(English model, with harmonic analysis function)
PW3360-21
Accessories: Voltage cord L9438-53 ×1 set, AC adapter Z1006 ×1, USB cable ×1, Instruction manual
×1, Measurement guide ×1, Color clip ×1 set: red, yellow, blue, white/two each, for color-coding
clamp sensors, Spiral tubes for grouping clamp sensor cords ×5

Bundled Accessories
AC ADAPTER Z1006

cord length: 3m (9.84 ft)
1 cord each of black, red yellow,
and blue, and five spiral tubes forbundling cords

Note: At least one optional current sensor is necessary to measure current or power parameters. To store measurement data, use only the guaranteed SD cards sold by HIOKI.

100 to 240 V ACn Options

CLAMP ON SENSOR (for load current measurement)

CLAMP ON ADAPTER

CLAMP ON SENSOR 9694 (5 A AC)
CLAMP ON SENSOR 9660 (100 A AC)
CLAMP ON SENSOR 9661 (500 A AC)
CLAMP ON SENSOR 9669 (1000 A AC)
AC FLEXIBLE CURRENT SENSOR CT9667-01 (5000 A AC)
AC FLEXIBLE CURRENT SENSOR CT9667-02 (5000 A AC)
AC FLEXIBLE CURRENT SENSOR CT9667-03 (5000 A AC)
CLAMP ON SENSOR (Not CE marked) 9695-02 (50 A AC)
CLAMP ON SENSOR (Not CE marked) 9695-03 (100 A AC)
CONNECTION CORD 9219 (for connection to 9695-02, 9695-03)

9290-10

Secondary side
100 A

CAT III 600 V
Cord length: 3 m (9.84 ft)
Measurable conductor diameter
φ55 mm (2.17 in)
Bus bar: ■ 80 mm (3.46in) 5 20 mm (0.79 in)CT ratio: 10:1

CLAMP ON LEAK SENSOR(for leakage current measurement)

CLAMP ON LEAK SENSOR 9657-10
CLAMP ON LEAK SENSOR 9675

Storage media

SD MEMORY CARD 8GB
Z4003

CARRYING CASE

Rated voltage: 240 V AC
Operating temperature and humidity range: -10 to 50°C, 80% RH or lessPower Source

SD Card Precaution

CAT III 300V

MAGNET ADAPTER

C1005

PATCH CORD
L1021-01

Banana branch-banana, Red: 1,
Cable length: 0.5 m, For branchingfrom the L9438-50 or L1000, CAT
IV 600 V, CAT III 1000 V

L1021-02

Banana branch-banana, Black: 1,
Cable length: 0.5 m, For branchingfrom the L9438-50 or L1000, CAT
IV 600 V, CAT III 1000 V

VOLTAGE LINE POWER ADAPTER

PW9003(supplies power frommeasurement lines)

Stores up to one year’s data when acquired at one
minute intervals.
Use only SD Cards sold by HIOKI. Compatibility and performance are notguaranteed for SD cards made by other manufacturers. You may be unable
to read from or save data to such cards.

MAX. 1500 A AC (continuous: 1000 A)
Primary side1000 A

When purchasing the 9695-02 and 9695-03, we recommend also purchasing the separately sold 9219 Connection Cord.

SD MEMORY CARD 2GB
Z4001

VOLTAGE CORD L9438-53

9804-01 Red

BATTERY SET

Battery Case and Battery Pack Set
PW9002BATTERY PACK 9459

NiMH, Charges while installedin the main unit

POWER LOGGER VIEWER
SF1001LAN CABLE
9642

9804-02 Black
φ11mm (0.43 in)(generally compatible with M6 pan screws)
Magnetic tip for use with the standard
VOLTAGE CORD L9438-53
Red and black adapters sold separately.
Purchase the quantity and color appropriate for yourapplication.
(Example: 3P3W-3 adapters, 3P4W-4 adapters)

Straight Ethernet cable, supplied withstraight to cross conversion adapter, 5m (16.41 ft) length

Dimension:
Approx. 390W (15.4")×275H (10.8")×110D (4.3") mm

Note: Company names and product names appearing in this catalog are trademarks or registered trademarks of various companies.

DISTRIBUTED BY

HEADQUARTERS
81 Koizumi,
Ueda, Nagano 386-1192 Japan
https://www.hioki.com/

All information correct as of Mar. 10, 2020. All specifications are subject to change without notice.

PW3360E17-03B

Printed in Japan