Sprint & Stroller

Technical Service ManualI Preface
Sprint & Stroller

SERVICE MANUALSPRINTTM
STROLLER®
CAIRE Inc.TM
2200 Airport Industrial Drive
Suite 500
Ball Ground, GA 30107
www.caireinc.com
Customer/Technical Service:
United States		Phone:

800 482 2473

Asia, Australia, Pacific Rim
Phone:+61 297 494333

Fax:

888 932 2473

EuropePhone:

Fax:+44(0) 118 979 9245

+44 (0) 1189 367060

Abbreviations
FCV Flow Control Valve
LED Light Emitting Diode
LOX Liquid Oxygen
LPM Liters Per Minute
NER Normal Evaporation Rate
POI Patient Operating Instructions
N2 Nitrogen Gas
TF Top Fill
DF Dual Fill

Fax :

PRV
QDV
RMA
RP
R/R
SRV
O2
SF
PTFE

888 932 2473

Primary Relief Valve
Quick Disconnect Valve
Return Materials Authorization
Repair Procedure
Removal and Replacement
Secondary Relief Valve
Oxygen Gas
Side Fill
Polytetrafluoroethylene (Teflon®)

Definition of Terms
WARNING Description of a condition that can result in personal injury or death.
CAUTION Description of a condition that can result in equipment or component damage.
NOTE
A statement containing information important enough to emphasize or repeat.
(ITEM)
Item numbers used throughout this manual are shown on the illustrations beginning on page 34.
Disclaimer
This manual is intended for use by experienced personnel only. No attempt should be made to fill or maintain this equipment until both this manual and the Patient Operating Instruction booklet have been read and fully understood.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

1

II Table of Contents
II. Table of Contents. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2
III. Equipment Description. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3

Sprint & Stroller

I. Preface . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 1

IV. Specifications. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
V. Safety. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5
VI. Theory of Operation. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10VII. Unpacking and Setup . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12
VIII. Operation. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12
IX. Maintenance. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
X. Troubleshooting/Repair Procedures (Table of Contents) . . . . . . . 15
Troubleshooting Chart. . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17–21
Repair Procedures. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22
Service Tools/Equipment/Supplies. . . . . . . . . . . . . . . . . . . . . . 33
XI. Parts Price List. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 34
Ordering Information. . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 35
XII. Restocking Policy . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 36

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

12

III Equipment Description
CAIRE Inc.’s Sprint or Stroller is the portable component of the
Liberator/Portable supplementary oxygen system. Enclosed in acontoured light-weight case, the Sprint or Stroller incoporates astainless steel cryogenic container with the valves, plumbing, andassociated hardware required to deliver gaseous oxygen to the patient at near ambient temperature.
The Sprint and Stroller are comprised of five major assemblies,
grouped according to function.

Sprint & Stroller

FIGURE 1: Sprint / Stroller Components

1. Cryogenic Container – This assembly is a double walled,
vacuum insulated dewar for storing liquid oxygen at approximately -180°C (-300° F).
2. Breathing Circuit – This circuit consists of the manifold assembly, fixed orifice rotary flow control valve (FCV) andvaporizing coil. It withdraws liquid oxygen from the cryogenic container, warms it to near ambient temperature, andregulates the flow of oxygen gas to the patient. Water thatcondenses on the cold coils is collected and absorbed by thecondensate pad.
3. Case Assembly – The hard case assembly consists of a top
cover, side cover, and bottom cover. These cases house andprotect the cryogenic container, the breathing circuit, andthe liquid level meter. The Stroller soft case is a one pieceassembly.
4. Liquid Level Meter – This system uses a capacitance probe andan electronic (LED) readout to measure and display the LOXlevel by pressing the onboard operate button. Alternatively,
the Sprint / Stroller are also available in a Spring Scale version.
5. Filling Circuit – This manual covers both the side fill Sprint/
Stroller with the rotary quick disconnect valve and the top fill
Sprint/Stroller with the push quick disconnect valve for filling.
* Gen 4 meter with hardcase shown

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

3

IV
1 Specifications (Nominal Values)
PEDIATRICSTROLLER®

SPRINTTMPEDIATRICSPRINTTM

3.0 (1,36)
1,25
1026

3.0 (1,36)
1,25
1026

1.5 (0,68)
0,63
513

1.5 (0,68)
0,63
513

Off 0,25 0,50
0,75 1,0 1,52,0 2,5 3,04,0 5,0 6,0Off 0,03 0,06
0,13 0,25 0,38
0,50 0,63 0,75
1,0 1,5 2,0

Off 0,25 0,50
0,75 1,0 1,52,0 2,5 3,04,0 5,0 6,0Off 0,03 0,06
0,13 0,25 0,38
0,50 0,63 0,75
1,0 1,5 2,0± 0.1 liter per minuteor ±10% of flow
setting, whicheveris greater		Refer toTable 1 on
page 9± 0.1 liter per minuteor ±10% of flow
setting, whicheveris greater

Refer toTable 1 on
page 9

Capacity
LOX lbs. (kg)Liquid Liters
Gaseous Liters
Selectable Flow Rates
Liters Per Minute (LPM):

Flow Rate Accuracy

Normal Evaporation Rate lbs./day (kg/day)

1.3 (0,57)

1.3 (0,57)

1.3 (0,57)

1.3 (0,57)

Operating Pressure PSIG (Bar)20 (1,4)

20 (1,4)

20 (1,4)

20 (1,4)

19.7-21.5 (1,35-

19.7-21.5 (1,35-

19.7-21.5 (1,35-1,48)

27-33 (1,86-2,27)

27-33 (1,86-2,27)

27-33 (1,86-2,27)

27-33 (1,86-2,27)

2:00
1:00

2:00
1:00

1:30
0:45

1:30
0:45

Height in (cm)13.5 (34,3)

13.5 (34,3)

11.75 (29,8)

11.75 (29,8)

Hard Case
Width in (cm)Oval 6.5 x 5.0(16,5 x 12,7)

Oval 6.5 x 5.0(16,5 x 12,7)

Oval 6.5 x 4.0(16,5 x 10,2)

Oval 6.5 x 4.0(16,5 x 10,2)

Soft case
Width in (cm)Oval 7.25 x 5.0(18,4 x 12,7)

Oval 7.25 x 5.0(18,4 x 12,7)

5.0 (2,3)
8.0 (3,6)

5.0 (2,3)
8.0 (3,6)

4.5 (2,0)
6.0 (2,7)

4.5 (2,0)
6.0 (2,7)

Secondary Relief Valve Setting PSIG (Bar)
Filling Time
Warm minutes:seconds
Cold minutes:seconds

WeightEmpty lbs. (kg)Full lbs. (kg)Fill Connector Types

Sprint & Stroller

STROLLER®

Male side mounted rotary coupling, female bottom mounted push-on coupling,or female bottom mounted rotary coupling.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4

V Safety

These hazards require certain safety precautions to be taken whenworking with or around gaseous and/or liquid oxygen:
1. Never permit combustible substances such as greases, oils,
solvents, or other compounds not oxygen compatible tocontact any component of the unit exposed to higher-thanatmospheric concentrations of gaseous or liquid oxygen. This
especially applies to tubing, fittings, and valves.
2. Keep oxygen equipment away from open flames or electricalappliances such as heaters, stoves, toasters, and other devices
with heating elements.
3. Never permit smoking in an area where oxygen equipment isrepaired, filled, or used.
4. Always wear goggles, a face shield, and insulated gloves whenworking with or around liquid oxygen.
While CAIRE Inc. equipment is designed and built to the mostrigid standards, no piece of mechanical equipment can ever bemade 100% foolproof. Strict compliance with proper safety practices is necessary when using any Sprint/Stroller unit. We recommend that our distributors emphasize safety and safe handlingpractices to their employees and customers. While safety features
have been designed into the unit and safe operations are anticipated, it is necessary that all distributor personnel carefully readand fully understand WARNINGS, CAUTIONS, and NOTES
throughout the manual. Periodic review of this information is recommended.

WARNING: Excess accumulation of oxygen creates an oxygenenriched atmosphere (defined by the Compressed Gas Association
as an oxygen concentration above 23%). In an oxygen-enrichedatmosphere, flammable items may burn vigorously and may explode. Certain items considered non-combustible in air may burnrapidly in such an environment. Keep all organic materials andother flammable substances away from possible contact with oxygen; particularly oil, grease, kerosene, cloth, wood, paint, tar, coaldust, and dirt which may contain oil or grease. DO NOT permit
smoking or open flame in any area where oxygen is stored, handled, or used. Failure to comply with this warning may result inserious personal injury.

Sprint & Stroller

Oxygen, as it exists at standard atmospheric pressure and temperature, is a colorless, odorless, and tasteless gas. Oxygen constitutes
21% of the atmosphere, by volume. Aside from its well-documented ability to sustain life, oxygen also supports combustion,even though it is nonflammable. Many substances which will burnin air, burn at a faster rate and at a higher temperature in an oxygen enriched atmosphere. Other materials that do not burn in air
will burn as oxygen concentration increases. Additionally, many
greases and liquid solvents become extremely hazardous materials when placed in an oxygen-enriched environment. In its liquid
form, oxygen is still odorless and tasteless, but is pale blue in color.
At an operating pressure of 20 psig (1,4 bar), the temperature ofliquid oxygen is about -173°C (-280° F). Skin exposed to such a lowtemperature can become severely frostbitten.

WARNING: In the event a unit is dropped, tipped over, or unreasonably abused; immediately, but cautiously, raise the container
to its normal vertical position. If substantial container damage
has occurred, remove the liquid oxygen from the vessel in a safemanner (RP14). Purge the unit with an inert gas (nitrogen) andpromptly return it to CAIRE, Inc. for inspection. The container
should be prominently marked “CONTAINER DROPPED,INSPECT FOR DAMAGE.” Failure to comply with these procedures may result in personal injury and can seriously damage thecontainer.
WARNING: Personnel must remove liquid oxygen and depressurize the unit before removing parts or loosening fittings froma unit. Failure to do so may result in personal injury from theextreme cold of liquid oxygen and/or the pressure in the vessel.
WARNING: During transfer of liquid oxygen, components will
become extremely cold. Care should be used to avoid any contact
with these components, as serious frostbite may result.
WARNING: Keep filled unit upright at all times. Tip over of filledunit may result in liquid oxygen leakage and/or an oxygen-enriched atmosphere.
CAUTION: Only use replacement equipment which is compatible
with liquid oxygen and has been cleaned for oxygen use. Do not use
regulators, fittings, hoses, etc. which have been previously used in
non-oxygen service.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 5

V
1 Safety
WARNING: The use of Accessories, transducers, and cables otherthan those specified by the manufacturer may result in increased
Emissions or decreased immunity of the Sprint/Stroller.

WARNING: Portable and mobile RF communications equipment
can affect Medical Electrical Equipment.

WARNING: The Sprint/Stroller should not be used adjacent to or
stacked with other equipment, and that if adjacent or stacked use isnecessary, the Sprint/Stroller should be observed to verify normal
operation in the configuration in which it will be used.

Sprint & Stroller

WARNING: Medical electrical Equipment needs special precautions regarding EMC and needs to be installed and put into service
according to the EMC information provided in this manual.

Table 1
Guidance and Manufacturer’s declaration—electromagnetic emissions
The Sprint/Stroller is intended for use in the electromagnetic environment specified below. The customer or the user of the Sprint/
Stroller should assure that it is used in such an environment.
Emissions test

Compliance

RF emissions

Group 1CISPR 11		

Electromagnetic environment—guidance
The Sprint/Stroller uses RF energy only for internal function.
Therefore, its RF emissions are very low and are not likely tocause any interference in nearby electronic equipment.

RF emissions CISPR 11

Class B

Harmonic emissions
IEC 61000-3-2

The Sprint/Stroller is suitable for use in all establishments,
Not applicable

Voltage fluctuations/		
flicker emissions

including domestic establishments and those directly connected tothe public low-voltage power supply network that supplies 		

Not applicablebuildings used fordomestic purposes.

IEC 61000-3-3

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 6

V
1 Safety
Sprint & Stroller

Table 2
Guidance and manufacturers declaration—electromagnetic immunity
The Sprint/Stroller is intended for use in the electromagnetic environment specified below. The customer or the user of the Sprint/
Stroller should assure that it is used in such an environment.
Immunity test

IEC 60601 test level

Compliance level

Electrostatic±6 kV contact±6 kV contact±8 kV air

±8 kV airdischarge (ESD)
IEC 61000-4-2			

Electromagnetic environment—guidance
Floors should be wood, concrete or ceramic tile. If floors
are covered with synthetic material, the relative
humidity should be at least 30%.*

Electrical fast

±2 kV for powerNot applicabletransient/burst

supply lines

DC powered device

IEC 610004-4±1 kV forNot applicable		Not applicable

input/output lines No data input/output lines±1 kV line(s)SurgeIEC 61000-4-5

to line(s)Not Applicable±2 kV line(s)DC powered device

Not Applicableto earth
Voltage dips,<5% UT (>95% dip		short interruptionsin UT) for 0,5 cycle		and voltage

40% UT (60% dip		variations on

in UT) for 5 cycles

Not Applicable

power supply

70% UT (30% dip

DC powered device

input lines

in UT) for 25 cycles		IEC 61000-4-11<5% UT (>95% dip

Not Applicablein UT) for 5 sec
Power frequency

3 A/m

3 A/m

Power frequency magnetic fields should be at levels(50/60 Hz)

characteristic of a typical location in a typicalmagnetic fieldcommercial or hospital environment.

IEC 61000-4-8
Note: UT is the a.c. mains voltage prior to application of the test level.

* This statement indicates that the required testing was performed in a controlled environment and the Sprint/Strollers are found to becompliant with regulations.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

47

V
1 Safety
Sprint & Stroller

Table 4*
Guidance and manufacturers declaration—electromagnetic immunity
The Sprint/Stroller is intended for use in the electromagnetic environment specified below. The customer or the user of the Sprint/
Stroller should assure that it is used in such an environment.
Immunity test
Conducted RF
IEC 61000-4-6

Radiated RF
IEC 61000-4-3

IEC 60601 test level
3Vrms
150kHz to 80 MHz

3 V/m
80 MHz to 2,5 GHz		

Compliance level
Not Applicable
Battery powered
device3 V/mElectromagnetic environment—guidancePortable and mobile RF communications equipment should
be used no closer to any part of the Sprint/Stroller,
including cables, than the recommended separation
distance calculated from the equation applicable to thefrequency of the transmitter.
Recommended separation distanced = 1.2 √P
d = 1.2 √P 80 MHz to 800 MHz
d = 2.3 √P 800 MHz to 2,5 GHz
where P is the maximum output power rating of thetransmitter in watts (W) according to the transmitter
manufacturer and d is the recommended separation
distance in meters (m).
Field strengths from fixed RF transmitters, as determinedby an electromagnetic site surverya, should be less thanthe compliance level in each frequency rangeb.
Interference may occur in the vicinity of equipment marked
with the following symbol:

NOTE 1 At 80 MHz and 800 MHz, the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection fromstructures, objects and people.
Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur
radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To asses the electromagnetic
environment due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in thelocation in which the Sprint/Stroller is used exceeds the applicable RF compliance level above, the Sprint/Stroller should be observed toverify normal operation. If abnormal performance is observed, additional measures may be necessary, such as reorienting or relocating
the Sprint/Stroller.
bOver the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.
a

* This table is included as a standard requirement for equipment which has been tested to specific test levels and over specific frquency
ranges and been found compliant with regulations.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R48

V
1 Safety
Sprint & Stroller

Table 6*
Recommended separation distances between portable and mobile
RF communications equipment and the Sprint/Stroller
The Sprint/Stroller is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The customer or the user of the Sprint/Stroller can help prevent electromagnetic interference by maintaining a minimum distance between
portable and mobile RF communications equipment (transmitters) and the Sprint/Stroller as recommended below, according to the
maximum output power of the communications equipment.
Rated maximum output		Separation distance according to frequency of transmitter
power of transmitterm
150 kHz to 80 MHz

80 MHz and 800 MHz

800 MHz to 2,5 GHz

d=1.2√P

d=1.2 √P

d=2.3 √P

0,01

0.12 m0.12 m

0.23 m0,10.38 m0.38 m0.73 m

1

1.2 m

1.2 m

2.3 m

10

3.8 m

3.8 m

7.3 m

W

100
12 m
12 m
23 m
For transmitters rated at a maximum output power not listed above, the recommended separation distance (d) in meters (m) can be
estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer.
NOTE 1 at 80 MHz and 800 MHz, the separation distance for the higher frequency range applies.
NOTE 2 These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection fromstructures, objects and people.
* This table is included as a standard requirement for equipment which has been tested to specific test levels and over specific frquency
ranges and been found compliant with regulations.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 9

VI Theory of Operation
Oxygen Withdrawl

1. Method

With oxygen in the unit, and the vent valve closed, the pressurein the inner vessel will remain at or near the primary relief valvepressure of 19.7 - 20.6 psig (1,35 - 1,42 bar).

The Sprint and Stroller are portable units designed to be filled by
the patient from a Liberator, Low Loss Reservoir, or other compatible unit.
The unit is filled by coupling the quick disconnect valve of theSprint/Stroller with the quick disconnect valve on the Reservoir
and opening the vent valve on the portable. Liquid oxygen isforced up the Reservoir fill tube, through the coupled quick connectors, and into the inner vessel of the Sprint/Stroller.
There will be some oxygen vaporized during filling. This gas isdischarged through the vent valve. When the unit is full, liquid
oxygen is expelled through the vent valve. Closing the vent valve
and separating the units terminates the fill process.
2. Saturation Pressure

At operating pressure and with the flow control valve at any setting other than Off, pressure forces liquid oxygen up the liquid
withdrawl tube and into the breathing coil. In the breathing coil,liquid oxygen absorbs heat and vaporizes, warming to almost ambient temperature by the time gas is dispensed by the flow control
valve to the patient.

Sprint & Stroller

Filling

FIGURE 3: Oxygen Withdrawl

PRV

The liquid Oxygen saturation pressure can seriously affect theoverall efficiency and operation of the Sprint or Stroller.

FCV
Gas to Patienta. If the saturation pressure of the liquid in the fill source isgreater than 30 psig (2,1 bar), high filling losses, flow rates
out of tolerance, and/or frozen open relief devices may result.

SRVb. If the saturation pressure of the liquid in the fill source is below 18 psig (1,2 bar), below tolerance flow rates may result.

Gas

FIGURE 2: Filling

PRV

Gas
Vent Valve

FCV

SRV
LiquidFill Connection(QDV)

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 10

VI Theory of Operation
1. LED
Sprints and Strollers are equipped with a unique liquid level measurement system. This system measures the level of liquid oxygen
inside the unit with a capacitance probe and displays that liquid
level on the level meter’s LEDs.
The liquid level probe consists of two concentric stainless steel
cylinders extending inside the inner vessel. As the liquid oxygen
level rises, the capacitance of this assembly goes up. The level
meter then displays the liquid level in the cylinder based on acalibration relating capacitance to level. The higher the liquid
level in the dewar, the more LEDs are activated, beginning at theleftmost LED.
Electrical connection between the G4 meter and the probe ismade via a single conductor JST connector. The male plug isattached to its female counterpart extending from the probe,
creating a watertight connection. A single ground wire is connected from the meter to a male spade terminal on the mounting bracket.
FIGURE 4: Liquid Level Meter
Meter Package

The meter is powered by an internal battery, offering an estimated battery life of 5 years or more at 30 actuations per day.
The meter has a low battery (LOW BATT) indicator which signals the need for battery replacement. The meter battery is covered under a 2-year limited warranty. If battery failure occurs
within 2 years of Sprint/Stroller shipment date, contact CAIRE
Inc. customer service for a replacement meter. If the meter
battery is no longer under warranty, the CR2032 coin cell battery can be replaced (RP2). The battery can be found at mosthardware stores, or they can be ordered through CAIRE Inc.
customer service.

Sprint & Stroller

Liquid Level Measurement

The new level meter improves upon the previous meter by integrating all components within its casing, simplifying removal
and replacement (RP2). Even more importantly from a technical service point of view, there is a much improved calibration
procedure requiring no additional tools and a range of error
reporting codes which can be read directly from the LEDs toreport calibration errors. These calibration codes can be foundin the calibration procedure (RP6).
2. Spring Scale
Sprints or Strollers may also be equipped with a spring scaleliquid level measurement system. This system measures thelevel of liquid oxygen inside the unit with a simple spring scale
mechanism, which is incorporated into the strap assembly. The
spring scale display is calibrated to read empty when only theweight of the portable unit is suspended from the strap. As liquid oxygen is placed into the portable, the spring scale displays
the increase in weight. The only calibration is to set the scale tozero, or empty point.

Plug Connector

Male Quick
Connect Valve

Probe Outer
CylinderLiquid Fill Tube
(Probe Inner Cylinder)

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 11

VII Unpacking and Setup
FIGURE 5: Flow Control Knob

1. Inspect the carton for shipping damage. Report any damage tothe freight company before signing the bill of lading.
2. Check the description on the carton against your order.
3. Unpack the unit, including the Patient Operating Instructions(POI).

Sprint & Stroller

Unpacking4. Set aside the packing material in case the unit must be returned to the factory.
Setup
If desired, the flow control knob (Item 20) can be adjusted so it will
not exceed the maximum prescribed flow rate.
1. To remove the flow control knob, firmly grip and pull straight
up away from the portable.
2. Remove the flow rate decal number disc (Item 21).
3. Remove the 2 Phillips head screws (Item 22) from the flow
lock plate (Item 26) and remove the plate.
4. Remove the locking pin (Item 23) from its storage position on
the flow lock plate and place in the underside of the hole corresponding to the maximum allowable flow rate.
5. Replace the flow lock plate (Item 26) and tighten screws (Item22). Replace number disk (Item 21) and push-on knob (Item20). Verify the flow lock is at the correct position.

VIII Operation
Refer to Patient Operating Instrucions (POI) Manual.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 12

IX Maintenance (Schedule A, 30-Month)

Schedule A – 30-Month
A. Introduction
Routine maintenance is a series of steps used to assure that equipment is functioning properly.
1. If a unit fails a given test, one of two things may be done:
a. Refer to Troubleshooting section (Section XI) of this manual.
b. Return the unit to CAIRE, Inc. for repair.
2. Schedule A – Maximum of 30 Months between routine
main-tenance testing. Unit should be tested whenever a
problem is suspected.
B. Procedure
Follow the steps in order listed. If the unit fails any step,refer to the Troubleshooting section (Section XI) of this manual.
1. Visual Inspection:
a. Look for damaged or missing parts.
b. Dry condensation pad. Replace if soiled.
2. Full Unit:
a. Fill, vent full, of LOX.
b. Check for audible or visual leaks in QDV and vent valve.
c. Verify that the meter reads full (8 LED’s or “Full” on spring
scale).
3. Check Efficiency of Unit:
a. Allow the unit to stabilize after filling (10–15 minutes).
b. Inspect the bottle for cold sweaty condition, and for excessive venting from relief valve (some venting is normal).
c. If either condition is observed, conduct NER test (see RP29
in this manual).
4. Flow Test (Table 1):

Sprint & Stroller

There are two schedules for routine maintenance which the homehealth care distributor may follow. These schedules allow the distributor maximum flexibility while assuring that the equipment is
operating properly.
CAIRE, Inc. recommends returning the unit to a CAIRE service
facility every 10 years for vacuum check and re-evacuation
if necessary.

TABLE 1: Flow Test Acceptable Ranges
Unit TypeStandard
Standard
Standard
Standard
Standard
Standard
Standard
Standard
Standard
Standard
Standard
Standard
Standard
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric
Pediatric

FCV Setting
OFF
0,25
0,50
0,75
1,00
1,50
2,00
2,50
3,00
3,50
4,00
5,00
6,00
OFF
0.05
0.10
0.20
0.30
0.40
0,50
0.75
1.00
1.25
1,50
2,00

LPM
0
0.08 to 0.42
0.33 to 0.67
0.58 to 0.92
0.83 to 1.17
1.18 to 1.82
1.61 to 2.43
2.08 to 2.97
2.55 to 3.51
N/A
3.43 to 4.62
4.33 to 5.77
5.14 to 6.92
00.03 to 0.09
0.05 to 0.20
0.07 to 0.39
0.18 to 0.52
0.20 to 0.56
0.33 to 0.67
0.58 to 0.92
0.83 to 1.17
0.98 to1.55
1.18 to 1.82
1.61 to 2.43

NOTE: SLPM is corrected to normal temperature and pressure(70˚F & 14.7 psig or 21˚C and 1.01 Bar). The above acceptableflow rates are intended for field testing only of flow control
valves to account for variation in elevation and temperature.
This table reflects tolerance requirements of ISO 10524. This
does not affect CAIRE’s factory compliance with the +/-10%
tolerance requirement of ISO 18777.
a. Set FCV to the highest flow setting, run for at least 20 minutes.
b. Check all flow settings to chart above and check pressure to
be at least 18 psig (1,2 bar).
5. Prepare for Use:a. Empty contents by setting FCV at the highest flow setting
and running unit until dry and warm to room temperature(approximately two hours).
b. Verify that meter reads empty (1 LED) and that the low battery LED is not lit. For spring scale units, verify that scale
reads empty (red line).
c. Clean case with household glass cleaner and lint free cloth(do not get in any valves).
Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 13

IX Maintenance (Schedule B, Continuous)
A. Introduction
Continuous maintenance is a set of tests and inspections performed periodically to ensure the equipment is functioning properly. It can be done with equipment in service by drivers or otherpersonnel.
1. If a unit fails a given test, it should be taken out of service.
Refer to the Troubleshooting section (Section XI) or contact
Technical Service.
2. Schedule B – Checks are made when the driver visits patients
and when equipment is transferred between patients.

Sprint & Stroller

Schedule B – Continuous

4. Flow Test (Table 1):
a. Set flow control knob to highest flow setting, run for atleast 20 minutes.
b. Check all flow settings with chart below and check pressure to at least 18 psig (1,2 bar).
5. Prepare for Use:a. Empty contents by turning flow selection knob to highest
flow setting and running unit until dry and warm to room
temperature (approximately two hours).
b. Verify that the meter or scale reads empty.
c. Clean case per RP28.

B. Procedure
These inspections are to be performed at least once per year by
the driver or other personnel when the Sprint/Stroller is in use by
the patient (LOX in the unit).
1. Visual Inspection:a. Case not damaged or dirty. Flow control knob turns easily
with firm detent feel.
b. Condensation pad is not soaking wet or excessively dirty.
c. QDV pin not bent (side fill only).
2. Verify that the meter or scale reads correct liquid level.
3. Check flow rate. Erie liter meter (± 0.25 LPM) can be used.
These inspections/tests are to be done between patients.
1. Visual Inspection:a. Broken case.
b. Condensation pad (replace if soiled).
c. QDV pin not bent (side fill only).
d. Inspect interior of unit for dirt or contaminants.
2. Verify that liquid level meter or scale reads empty.
3. Fill Unit:
a. Verify that liquid level gauge or scale reads full.
b. Check for audible/visual leaks in QDV and vent valves.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 14

X Troubleshooting/Repair Procedures
Introduction.............................................................................................................16
Troubleshooting Charts...................................................................................................17—21
Repair Procedures
RP1
General...............................................................................................22
RP2
Battery RR . .......................................................................................22
RP3
Top Cover Assembly (Hard case) RR............................................23
RP4
Bottom and Side Cover (Hard case) RR........................................23
RP5Soft case RR.......................................................................................23
RP6
G4 Empty Capacitance Meter Calibration....................................24
RP7
G4 Meter Span Setting Adjustment...............................................25
RP8
Manifold Capacitance Test..............................................................25
RP9
Gen4 Meter RR.................................................................................26
RP10
Manifold Assembly RR....................................................................26
RP11
Resolder Feed-Thru Wire................................................................26
RP12
Clean/Dry Probe Dewar..................................................................27
RP13
LOX Fill..............................................................................................27
RP14
Liberator Lip Seal RR (Side Fill).....................................................27
RP15
Empty and Warm.............................................................................27
RP16
QDV RR (Side Fill)...........................................................................27
RP17
QDV RR (Top Fill)...........................................................................27
RP18
Pressure Retention Test ..................................................................28
RP19
Plumbing Leak Test..........................................................................28
RP20
Vent Valve RR . ................................................................................29
RP21
FCV RR..............................................................................................29
RP22
SRV RR...............................................................................................29
RP23
PRV RR .............................................................................................29
RP24A
Breathing Coil RR (Side Fill)...........................................................30
RP24B
Breathing Coil RR (Top Fill)...........................................................30
Flow Rate Testing.............................................................................31
Operating Pressure Test..................................................................31
Flowmeter Verification....................................................................31
Case Cleaning....................................................................................32
Normal Evaporation Rate Test.......................................................32
Spring Scale Liquid Level Calibration............................................32
Relief Valves test..............................................................................32
Service Tools/Equipment/Supplies .......................................................................................33

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

Sprint & Stroller

Table of Contents

4 15

X Troubleshooting/Repair Procedures
1. These procedures are designed to be performed only by qualified personnel with proper equipment.
2. Any failure during routine maintenance checks will refer you to this section. See troubleshooting chart for appropriate
dure.

proce-

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

Sprint & Stroller

Introduction

4 16

X Troubleshooting Chart
1)

Probable Cause

Unable to start fill ora) QDV not properly engaged on
excessivly long fill times
the reservoir

Sprint & Stroller

Symptom

Corrective Action
Make sure the QDV on the portable and reservoir are properlyaligned. Ensure that a downward force is being applied to theportable for top fill. Ensure the portable is completely engaged
for side fill.

b) Reservoir is empty

Swap or re-fill the reservoirc) Vent valve not open

Ensure that the vent valve lever is fully in the open position.
The lever must be open to begin a fill.

d) FCV is open

Be sure that the FCV knob is in the off (“0”) position. If thevalve is open, fill times can increase.

e) Reservoir saturation pressure is
too low

Swap reservoirs or allow the reservoir time to stabilize and buildpressuref) Vent valve is obstructed

Inspect the vent tubes for blockages. Clean by blowing out withcompressed gas or replace parts if necessary.

g) Leak in the system

Check the portable for leaks (RP19) and repair if needed.

h) QDV damaged or faulty

Inspect the QDV and be sure the poppet opens properly andsmoothly. If necessary, replace the QDV (RP16 or RP17)i) Faulty vent valve

Replace the vent valve (RP20)

2)

Liquid leaks from thecoupled QDVs during
the filla) Worn or damaged lip seal

Replace the QDV lip seal if top fill (RP17). See reservoir technical manual for replacement of side fill lip seal on reservoir.

3)

Unable to disconnect
the poratble from thereservoira) Pop-off assembly not being utilized (Top fill only)

Ensure that the pop-off assembly on the reservoir is being used
(top fill only). Do not use force to separate the QDVs on top fillor side fill.

b) QDVs are frozen together

Leave the units coupled with the vent valve closed and let themsit until they warm up enough to disconnect. Always ensure thatmale and female QDV’s are cleaned and dried prior to each fill.

a) Ice crystal preveting the QDVfrom closing properly.

Engage and disengage the portable onto the reservoir several
times to dislodge the ice crystal. Always be sure that the maleand female QDVs are wiped clean and dry before filling.

b) Dirty or damaged QDV poppet

Replace the QDV (RP16 or RP17)a) Vent valve is not fully closed

Ensure that the vent valve lever is fully in the closed position.

b) The portable has been transported or laid in an improper
operating position

Return the portable to an upright or acceptable operating position and allow several minutes for stabilization.

c) Vent valve is frozen open

Allow the portable to warm until the vent valve can close. Afterthe warm up, allow up to 60 minutes for the portable to stabilize
and build pressure before operating.

d) Faulty vent valve

Replace the vent valve (RP20)

4)

5)Liquid leaks from the
QDV after filling

Liquid leaks from thevent valve tube/outlet

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 17

X Troubleshooting Chart
6)

7)

Probable Cause

Corrective Action

Excessive venting from a) The portable has been transrelief valves (hissing
ported or laid in an impropersound)
operating position

Return the portable to an upright or acceptable operating position and allow several minutes for stabilization.

b) Saturation pressure too high.

Inspect the saturation pressure of the reservoir used for filling.
Allow at least 30 minutes at no flow for the portable to saturate
properly.

c) Relief valve frozen open

Allow the portable to warm and thaw. Attempt to re-fill the
portable.

d) Faulty relief valve

Test the operating pressure (RP26) and replace the relief valve ifnecessary (RP23)e) Partial or complete loss of
vacuumConduct the NER test (RP29) and return the unit to CAIRE,
Inc. if necessary.

a) Portable is empty

Check the contents indicator/level gauge and fill the portable ifneeded.

b) Flow control valve turned off

Ensure the flow control knob is not in the off (“0”) position.

No Flow

Sprint & Stroller

Symptomc) Nasal cannula kinked or discon- Ensure proper nasal cannula functionality and positioning
nectedd) Saturation pressure is too low

Inspect the saturation pressure of the reservoir used for filling.
Allow at least 30 minutes at no flow for the portable to saturate
properly.

e) Leak in the system

Preform a leak check on the plumbing (RP19). Repair leaks asnecessary.

f) Relief valve is open

Ensure that there is no venting from the relief valves. If there isrefer to the corrective actions for “Excessive venting from relief
valves (hissing sound)”g) Vent valve is open

Ensure that there is no venting from the vent valve outlet/tube.
If there is refer to the corrective actions for “Liquid leaks fromthe vent valve tube/outlet”h) Blockage in the liquid withdrawal Check the warming coils and withdrawal tubes for blockages.
circuitReplace if necessary.
i) FCV Faulty
8)

Low flow at all LPM
settings

Replace the FCV (RP21)a) Nasal cannula kinked or leaking Inspect the functionality of the nasal cannula.
b) Saturation pressure is too low

Inspect the saturation pressure of the reservoir used for filling.
Allow at least 30 minutes at no flow for the portable to saturate
properly.

c) Leak in the system

Preform a leak check on the plumbing (RP19). Repair leaks asnecessary.

d) PRV faulty

Test the operating pressure (RP26) and replace the relief valve ifnecessary (RP23).

e) Blockage in the liquid withdrawal Check the warming coils and withdrawal tubes for blockages.
circuitReplace if necessary.
f) FCV faulty

Replace the FCV (RP21)

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 18

X
1 Troubleshooting Chart
9)

Increased NER10) Excessive Frost
NOTE: Minimal frost on thecase and on the plubming isnormal, especially at higher
flow rates. This symptom
applies to frost that is muchgreater than what is normallyobserved.

11) Unit will not maintainacceptable pressure
when in use

Probable Cause

Corrective Actiona) Saturation Pressure is too highInspect the saturation pressure of the reservoir used for filling.
Allow at least 30 minutes at no flow for the portable to saturate
properly.

b) Leak in the system

Preform a leak check on the plumbing (RP19). Repair leaks asnecessary.

c) Relief valve open

Ensure that there is no venting from the relief valves. If there isrefer to the corrective actions for “Excessive venting from relief
valves (hissing sound)”d) Partial or complete loss of
vacuumConduct the NER test (RP29) and return the unit to CAIRE,
Inc. if necessary.

a) Frost is acceptable

Some frost on the outer case and on the plumbing is acceptable,
especially at high flow rates. This is due to the evaporation of
LOX to gas and the temperature difference between the LOX
and room temperature.

b) High humidity level

High humidity levels can increase frost accumulation.

c) Saturation pressure is too highInspect the saturation pressure of the reservoir used for filling.
Allow at least 30 minutes at no flow for the portable to saturate
properly.

d) Leak in the system

Preform a leak check on the plumbing (RP19). Repair leaks asnecessary.

e) Relief valve open

Ensure that there is no venting from the relief valves. If there isrefer to the corrective actions for “Excessive venting from relief
valves (hissing sound)”f) Partial or complete loss of
vacuumConduct the NER test (RP29) and return the unit to CAIRE,
Inc. if necessary.

a) Saturation pressure is unacceptable

Inspect the saturation pressure of the reservoir used for filling.
Allow at least 30 minutes at no flow for the portable to saturate
properly.

b) Leak in the system

Preform a leak check on the plumbing (RP19). Repair leaks asnecessary.

c) PRV faulty

Test the operating pressure (RP26) and replace the relief valve ifnecessary (RP23).

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

Sprint & Stroller

Symptom

4 19

X
1 Troubleshooting Chart
Good

Low
Battery
LED

Off/
Spring ScaleQDV PinSpring
Scale
Shows
Empty

OK

Yes
Fill

QDV
Leaks
During Fill

No

QDV
Leaks
After Fill

No

Vent
Leaks

Bad

On

Bad

No

Yes

Yes

Yes

Replace
Defective Part

Replace*
Battery (RP 2)

Replace QDV(RP 16, 17)

Calibrate
Spring Scale(RP 30)Replace Lip
Seal (RP 17)

Empty &
Warm Dewar(RP 15)

Empty &
Warm Dewar(RP 15)Fill

Replace Vent
Valve (RP 20)

No

Spring
Scale
Reads
Full

No

No

Yes

Calibrate
Spring Scale(RP 30)OK

Meter 1
LED onNo(0 LED)

Yes

No(2–8 LED)

No
QDV
Leaks
After Fill

Replace*
Battery (RP 2)

Sprint & Stroller

Case
VisualCalibrate
Meter

Yes

Meter
Indicates
8 LEDsYes

Empty &
Warm Dewar(RP 15)
Meter 1
LED OnYes
No(0 LED)No
Replace QDV(RP 16, 17)
Capacitance
Test (RP 8)
High

Empty &
Warm Dewar(RP 15)Good

Low
RemoveManifold(RP 10)Clean/Dry
Probe &
Dewar (RP 12)

CalibrateMeter(RP 6, 7)No(1–7 LED)

Good
Wire
BrokenNo

Replace FeedThru (RP 11)

Yes

Resolder Wire
(RP 11)

Capacitance
Test (RP 8)

Meter 1
LED OnBad

No

Replace
Manifold(RP 10)Replace Meter(RP 9)

Yes

Empty &
Warm Dewar(RP 15)

CalibrateMeter(RP 6, 7)

Bad

Capacitance
Test (RP 8)
Good

Reinstall
Manifold(RP 10)*see NOTE in RP 2

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 20

X Troubleshooting Chart
Sprint & Stroller

Suspected
Poor NERNo

Yes
Let Sit 16–18
Hours

Test Flows(RP 25)
Fail

Test
OperatingPressure (RP
26)

Pass

Meter
Indicates
4–8 LEDs

Spring Scale
Shows
0.5–FULL
Yes

Yes

Pass

Empty &
Warm Dewar(RP 15)Fail Low

No
Empty &
Warm Dewar(RP 15)No
Empty &
Warm Dewar(RP 15)

Calibrate
Spring Scale(RP 20)Clean Case
(RP 28)

Test
Flowmeter(RP 27)

Empty &
Warm Dewar(RP 15)

Pass

Plumbing
Leak Test(RP 19)Calibrate
Meter (RP 5,
6)

Fail

RefillRepair Leak

Replace PRV(RP 23)Empty &
Warm Dewar(RP 15)

Empty &
Warm Dewar(RP 15)

Replace PRV(RP 23)Replace FCV(RP 21)

Replace
Flowmeter (RP
27)

Pass

Return Unit toService

Let Sit 16–18
Hours

Fail

Let Sit 16–18
Hours

Pressure
Retention
Test (RP 18)

Spring Scale
Shows
0.5–FULLNo

Meter
Indicates
4–8 LEDs

Fail

File Records

RefillYes

Fail High

Pass
Yes

No
RefillReplace Dewar

Wait 4 Hours
Pressure
Retention Test
(RP 18)

Pass

Fail

Plumbing Leak
Test (RP 19)

Pass

Fail

Repair Leak

To use the Troubleshooting Chart:• Start at the upper left corner.
• Unless otherwise noted by the arrows, the flow through thechart is down or to the right.
Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 21

X Repair Procedures
RP2 – Liquid Level Meter Battery RR

The following procedures have been carefully prepared to allow
proper removal and replacement of defective components andshould be used in conjunction with the Troubleshooting Chart andthe tests listed in this section.

NOTE: Tampering with meter battery housing will void the meter’s 2-year warranty. If meter battery failure occurs while underwarranty, contact customer service to obtain a replacement meter. If the meter battery is no longer under warranty, the CR2032
coin cell battery may be replaced following the steps below.

WARNING: Make sure the unit is empty and vent valve is open
before replacing any component, except shroud assembly components or lip seals.
WARNING: The technician’s hands, tools, and clothing should befree of all oils and greases.
WARNING: Parts that are welded in place must not be replacedin the field. Should these parts fail, return complete assembly or
sub-assembly to factory for repair. DO NOT use solder or silver
solder to repair broken welds.
WARNING: The manufacturer of fluorolubricant warns users not
to allow fluorolubricant to contaminate tobacco products. Wash
fluorolubricant from hands before smoking.
WARNING: Do not use glue type thread locking compounds orunapproved sealants on any repairs.
CAUTION: When replacing components, make sure the new part isoriented exactly the same as the original part prior to installation.
CAUTION: Some components require a specific amount of torquewhen assembling. Follow torque requirements where specified.

a. Remove Gen 4 meter cover assembly (Item 1) by inserting a flathead screwdriver into the slot and lightly pushing over and up.
For soft case Stroller, unzip and peel back top cover to expose
meter.

Sprint & Stroller

RP1 – Generalb. Remove 2 Phillips head screws and turn meter over.
c. Disconnect gauge plug and remove gauge cover.
d. Using a pair of pliers, grasp the battery cover on the back of themeter. Twist and pull the cover away from the meter.
e. Slide the tip of a flat head jeweler’s screwdriver beside the battery
and carefully pry out the battery.
f. Insert new battery with the positive terminal facing up.
g. Apply a thin coating of fluorlubricant to the o-ring.
h. Replace battery cover by pushing it into the back of the meter’scase, ensuring the o-ring seals evenly to restore the air-tight seal.
FIGURE 6: Liquid Level Meter Battery

NOTE: All replacement parts must be factory approved, cleaned
for oxygen service, and stored in sealed plastic bags. The repair
area must be clean and separate from other areas. Room air should
be filtered, and free from dust, soot, and other contaminants.
NOTE: When replacing components with pipe threads, use PTFE
tape thread sealant. Apply two rounds of PTFE tape to threads
near end of component, avoiding first thread.
NOTE: When assembling new compression fittings, tighten 1/8”,
1/4” and 1/2” nuts eight flats past finger tight and 3/16” nuts fiveflats past finger tight. When reassembling previously used compression fittings, tighten nuts one to two flats past finger tight.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 22

X Repair Procedures
FIGURE 7: Top Cover Assembly

a. Remove Gen 4 gauge cover assembly as described in RP2.
b. Unscrew 3 screws (Item 8), unsnap and remove strap, and lift
top cover up.
c. Remove the QDV alignment sleeve (Item 11) on side fill units
by lifting up and out to disengage vent line; or the plug onbottom fill units, by lifting it straight up.

Sprint & Stroller

RP3 – Top Cover Assembly (Hard case) RR

d. Lift top cover (Item 10) straight up to remove.
e. To replace, reverse above procedure.
RP4 – Bottom and Side Cover (Hard case) RR
a. If removing the entire unit case, remove the top cover (RP3) first.
b. Invert Stroller. Remove condensation cup (Item 18) and condensate pad (Item 17).

FIGURE 8: Bottom and Side Coverc Remove the 2 screws (Item 15) retaining the bottom cover toStroller dewar.
d. Remove bottom/side cover (Item 13) by pulling straight down.
e. To replace, reverse above procedure. Be sure vaporizer coil iscentered in case.
CAUTION: Be careful not to pinch wires!
RP5 – Soft Case RR
a. Unzip fron and back zippers.
b. Carefully lift up and peel back top cover.
c. Open unziped case and carefully lift Stroller up and out of case.
Do not bend or force coils, as they are easily damaged.
d. To replace, carefully lower Stroller into open case and secure both
front and back zippers.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 23

X Repair Procedures
Sprint & Stroller

FIGURE 9: G4 Meter

RP6 – G4 Meter Empty Capacitance Calibration (Figure 9)a. Completely empty dewar.
NOTE: The meter must first be calibrated empty and then the
span must be set for your particular model. The unit must be
empty and dry for empty calibration, but does not have to be forthe span setting adjustment.

LED 8

LED 1

b. Allow dewar to warm to room temperature.
c. Enter Calibration Mode
1. Press and hold the hidden calibrate button located on thecenter of the meter face.

Calibrate Button(Hidden)Operate Button

2. While holding the calibrate button, press and hold thegreen operate button.
3. Continue holding both buttons until LED 1 and LED 8 alternately flash. The meter is now in calibration mode.
4. Release both buttons.
d. Press and hold the hidden calibrate button for 3 seconds. LED
NOTE: Step d must be performed within 45 seconds or calibration mode will exit.
1 will flash for 3 seconds followed by a short pause.

TABLE 2: Calibration Error Codes
Error CodeLED 1 & LED 4

Solution

Empty to full span is Ensure tank is empty.
too low.
Recalibrate.

LED 2 & LED 3

Open Circuit

Check meter connections
and wire integrity.

LED 6 & LED 7

Empty capacitanceis too high.

Remove moisturefrom inner vessel.
Decontaminate probe.

LED 5 & LED 8

Empty to full span is Ensure tank is empty.
too high.
Recalibrate.

LED 2, LED 4 &
LED 6High Capacitance

Remove moisturefrom inner vessel.
Decontaminate probe.

LED 1, LED 3,
LED 5 & LED 7

Calibrated full value
higher than empty
value.

Dry harness assy.
Decontaminate.

e. If successful, LED 1 will flash again for 3 seconds and exit
calibration mode storing the new empty value.
f. If calibration is unsuccessful, multiple LEDs will flash. The
LEDs indicate what error has occurred (see Table 2).

Reason for Fault

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 24

X Repair Procedures
Sprint & Stroller

RP7– G4 Meter Span Setting Adjustment (Figure 9)
a. Enter Calibration Mode

FIGURE 9: Manifold Capacitance Test Setup
Ground

1. Press and hold the hidden calibrate button located on thecenter of the meter face.
2. While holding the calibrate button, press and hold the green
operate button.
3. Continue holding both buttons until LED 1 and LED 8 alternately flash. The meter is now in calibration mode.
4. Release both buttons.
NOTE: Step d must be performed within 45 seconds or calibration mode will exit and the meter will NOT be calibrated properly. The calibration procedure will then need to be repeated fromthe beginning.
b. Press the green operate button 3 times within a 5 second period.
c. One of the meter LED’s will light continuously, signifying oneof the Stroller/Sprint models per Table 3 below.
d. Press the hidden calibrate button to scroll through the LED
until the LED which matches the Stroller/Sprint being calibrated is continuously lit. See Table 4 below for the correct
LED setting for your unit.
e. Once the correct LED for your unit is lit, press the green operate button to store the setting and exit calibration mode.

TABLE 3: Preset Capacitance Span LED Settings
Model
Stroller / High Flow

LED Setting
LED 8

SprintLED 3

RP8 – Manifold Capacitance Test (Figure 10)
a. Capacitance Meter Setup:
1. Connect wire adapter with jumper wire to capacitance meter,
following color code.
2. Turn on capacitance meter and select 200 pF range.
3. Move zero adjustment on front of meter until display reads
zero.
b. Capacitance Test:WARNING: Make sure the unit is empty and at room temperature
before testing.
1. Remove top cover (RP 3).
2. Connect wire adapter to manifold harness connection.
3. Connect alligator clip to bracket or any other properly
grounded surface.
4. Read manifold capacitance and compare to these acceptable ranges. If it does not meet specifications, call TechnicalSupport.
Stroller®: 145–180 pF
SprintTM: 125–155 pF

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 25

X Repair Procedures
RP11 – Resolder Feed-thru Wire (Figure 11)

a. Remove the G4 meter cover assembly (RP 2).

a Remove manifold assembly (See RP 10).

b. Remove 2 meter mounting screws.

b. Strip approximately 1/8” (0,3 cm) of insulation from feed-thru
wire if necessary.

c. Remove meter assembly from top cover.
d. Unplug the wiring connection.
e. Disconnect ground wire spade terminals.

c. Apply small amount of Stay-Clean flux to tinned area of probe
using a cotton swab.

f. To replace G4 meter, reverse above procedures.

d. Resolder feed-thru wire to tinned area of probe. Add small
amount of lead-free solid wire solder if necessary.

CAUTION: Be careful not to pinch wires!

e. Clean flux residue with distilled water and cotton swab. Dry
thoroughly.

RP10 – Manifold Assembly RR (Figure 11)

f. Replace manifold following listed procedure (RP10).

Sprint & Stroller

RP9 – Gen 4 Meter RR

FIGURE 11: Feed-thru WireWARNING: The unit must be empty, warm and vented before
starting procedure.

29

a. Remove meter, top cover, and bottom.side cover (RP 2, 3, 4, or
5).

28

b. Disengage FCV (Item 24) from bracket.
c. Remove (3) bracket screws (Item 29) and lift bracket (Item 28)
from manifold.
d. Loosen and disconnect breathing coil nut (Item 38) from withdrawl fitting (Item 41) and FCV (Item 24). Be careful not tobend or kink Teflon® tubing.
e. Loosen and disconnect primary relief valve from manifold fitting.
For pediatric units, also loosen and disconnect secondary relief
valve from maifold fitting.
f. For top fill units, loosen and remove QDV from manifold fitting.
g. Use a hex wrench to loosen and remove the (4) manifold retaining screws (Item 31).
h. Remove manifold assembly by lifting straight up.
NOTE: If substantial force is required, closely examine wire andprobe for damage.

31

24
38

52

50

41

45
44

N

46
45
38
48
49

i. To replace manifold, reverse above procedure.
NOTE: Ensure proper seating of O-ring in dewar. Apply small
amount of fluorolubricant to O-ring before assembly. Torquemani-fold mounting screws (Item 31) to 21–23 in-lbs (237–260 Ncm) us-ing cross-tightening method.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 26

X Repair Procedures
RP17 – QDV and Lip Seal RR (Top Fill)

Procedure 1:

WARNING: The unit must be empty and vented before starting procedure.

a. Follow steps A through E (RP 18).
b. Open vent valve (Item 50) and supply valves.

a. Remove bottom/side covers (RP 4, or 5).

c. Loosen feed-thru nut (Item 52) and remove harness.

b. Remove two retaining nuts (Item 54) and lift bottom fill tube
out of coil.

d. After vent valve (Item 50) is thawed, if all moisture is not removed, continue with procedure #2.
Procedure 2:
a. Remove manifold assembly (See RP 10).
b. Blow off probe assembly with clean, dry nitrogen gas.
c. Blow out inside of dewar with clean, dry nitrogen gas
until inside is clean and dry.

Sprint & Stroller

RP12 – Clean/Dry Probe and Dewar (Figure 11)c. Replace QDV by substituting old QDV (Item 62) with the newone, tightening Item 57 to 140–150 inch-lbs (1550–1700 N-cm)
and Item 58 to 300 inch-lbs (3000 N-cm).
d. Replace lipseal by removing Item 65 and Item 64. Apply a thinfilm of fluorolubricant to new Item 64 and insert into QDV.
Tighten Item 65 to 140–150 inch-lbs (1550–1700 N-cm).
FIGURE 12: Relief Valve Test Set-up

d. Replace manifold assembly (See RP 10).

37

RP13 – LOX FillA

Refer to POI for proper fill procedure.
NOTE: Filling source must contain a minimum of 5 liters ofproperly saturated LOX.

36

RP14 – Liberator Lip Seal RR (Side Fill)
Refer to Liberator Service Manual.
RP15 – Empty and Warm

FIGURE 13: Primary Relief Valvea. Turn FCV knob to highest flow setting.

53

b. Allow unit to sit for 24 hours before proceeding.

54

RP16 – QDV RR (Side Fill) (Figure 12)
WARNING: The unit must be empty and vented before starting procedure.
a. Remove top cover (RP 3, or 5).
b. Loosen retaining nut and remove QDV from manifold (Item A).
c. To replace QDV, reverse above procedure, aligning pin in the
upright position. Apply thin film of fluorolubricant to O-ring(Item 36). Torque retaining nut to 140–150 inch-lbs (1550–1700N-cm).
d. Replace top covers.

57
58
64
65

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 27

X Repair Proceduresb. Assemble pressure gauge (Item J) and adapter assembly. Threadhose barb onto pressure gauge (use Teflon® tape). Push tubing
onto hose bard and attach with clamp.

FIGURE 14: Plumbing Tests Set-up

49
24

J

c. Connect gauge assembly (Item J) to hose barb on FCV outlet.
Open FCV (Item 24) to highest flow setting.
d. Connect the proper pneumatic adapter to QDV (Item 37) forside fill units or (Item 62) for top fill.
e. Increase pressure to 19 psig (1,3 bar).

Sprint & Stroller

RP18 – Pressure Retention Test
a. Assemble oxygen regulator, pneumatic adapter and hose.
Connect assembly to oxygen gas source.

37

43

I

f. Turn FCV valve (Item 24) to Off setting.
g. Allow unit to sit undisturbed for 60 minutes.
h. Turn FCV valve (Item 24) to highest flow setting.

Fill Connection(QDV)

i. If pressure gauge (Item J) is at or above 10 psig (0,7 bar), unit
passes test.
RP19 – Plumbing Leak Test (Figure 14)
WARNING: The unit must be empty and vented before starting procedure.
a. Remove bottom, side and top covers (RP 2, 3, 4, or 5)b. Assemble oxygen regulator, pneumatic adapter and hose.
Connect assembly to oxygen gas source.
c. Assemble pressure gauge (Item J) and adapter assembly. Threadhose barb onto pressure gauge (use Teflon® tape). Push tubing
onto hose bard and attach with clamp.
d. Connect gauge assembly (Item J) to hose barb on FCV outlet.
Open FCV (Item 24) to highest flow setting.
e. Connect the proper pneumatic adapter to QDV (Item 37) forside fill units or (Item 62) for top fill.
f. Increase pressure to 19 psig (1,3 bar).

FIGURE 15: Coil and Vent Assemblies

g. Leak test all connections, joints, and valves with leak test solution.
NOTE: PRV (Item 49) and SRV (Item 43) may leak slowly. Repair
all other leaks first and retest for pressure retention before changing
relief valves.
h. Close FCV (Item 24) by turning to Off position. Remove pressuregauge assembly from humidifier adapter (Item 26).
i. Disconnect pneumatic adapter from QDV (Item 37).
j. Leak test QDV poppet and FCV outlet.
k. Repair all leaks by following appropriate repair procedures.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 28

X Repair Procedures
RP20 – Vent Valve RR (Figure 15)
WARNING: The unit must be empty and vented before starting procedure.
a. Remove bottom, side and top cover (RP 2, 3, 4 or 5).
b. Open the vent valve.
b. Remove vent valve (Item 16) by turning counterclockwise.

Sprint & Stroller

FIGURE 16: G4 Meter Top Fill (Push-on) QDV Assemblyc. To replace vent valve, reverse above procedure. Torque nut to90-100 inch-lbs (1000–1100 N-cm).
RP21 – FCV RR (FIgure 16, 17)
WARNING: The unit must be empty and vented before starting procedure.
a. Remove bottom, side and top cover (RP 2, 3, 4 or 5).
b. Disconnect coils from FCV inlet by loosening compression fitting (Item 44), and removing coils.
c. Remove FCV knob (Item 20) and flow decal (Item 21).
d. Disconnect and remove the FCV from bracket (Item 28) by removing two screws (Item 22).
e. Remove SRV (Item 43) from FCV.
FIGURE 17: Spring Scale Top Fill (Push-on) QDV Assembly

f. To replace FCV, reverse above procedure. Tighten screws(Item 22) to 4–6 inch-lbs. (45–70 N-cm). Record FCV serial
number.
RP22 – SRV RR
G4 meter units (Figure 16)
a. Remove bottom, side and top cover (RP 2, 3, 4 or 5).
b. Disconnect SRV (Item 43) from FCV (Item 24).
c. Replace SRV in reverse order. When reassembling SRV,
tighten approximately 10 to 20 degrees after the SRV contacts
the FCV body (20-30 inch-lbs, 225-350 N-cm minimum).
Spring scale units (Figure 17)
a. Remove bottom, side and top cover (RP 2, 3, 4 or 5).
b. Disconnect 30 psig (2,1 bar) SRV (Item 43) from tube adapter.
c. Reassemble in the reverse order using Teflon® tape on the
SRV threads.
RP23 – PRV RR (Figure 18)
a. Remove bottom, side and top cover (RP 2, 3, 4 or 5).
b. Remove PRV (Item 49) while holding adapter (Item 48).
c. Replace the PRV using two rounds of Teflon ® tape on thethreads.
d. Reassemble in the reverse order.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 29

X Repair Procedures
FIGURE 18: Side Fill (Push-On) QDV Assembly

WARNING: The unit must be empty and vented before starting procedure.
a. Remove bottom, side and top covers (RP 2, 3, 4 or 5).
b. Loosen the two compression fitting nuts (Items 38 and 44)
from liquid withdrawl fitting (Item 41) and FCV (Item 24).
Disconnect breathing coil (Item 67), being careful not to kinkbreathing coil or internal Teflon® tube.

Sprint & Stroller

RP24A – Breathing Coil RR (Side Fill) (Figure 18)

c. Carefully remove breathing coil outlet from around top of dewar.
d. To replace breathing coil, reverse above procedure.
CAUTION: The vaporizer coil is fragile. Do not drop coil. Do not
hit or mar coil with wrenches.

RP24B – Breathing Coil RR (Top Fill) (Figure 19)
WARNING: The unit must be empty and vented before starting procedure.
a. Remove bottom, side and top covers (RP 2, 3, 4 or 5).
b. Loosen the two compression fitting nuts (Items 38 and 44)
from liquid withdrawl fitting (Item 41) and FCV (Item 24).
Disconnect breathing coil (Item 67), being careful not to kinkbreathing coil or internal Teflon® tube.

FIGURE 19: Top Fill (Push-On) QDV Assemblyc. Loosen QDV tube (Item 56) nut.
d. Loosen QDV jam nut (Item 58). Remove QDV (Item 62) fromfill tube.
e. Cut two cable ties holding coil in place. Remove breathing coilby sliding straight down from fill line. Be careful to properly
remove breathing coil outlet from around top of dewar.
f. To replace coil, reverse above procedure. Tighten QDV jamnuts to 140–150 inch-lbs. (1550–1700 N-cm).

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 30

X Repair ProceduresRP25– Flow Rate Testing (Figure 20)

M

a. Unit must be at least 1/2 full with correctly saturated LOX.

24

N

b. Set FCV to 2 LPM setting and allow unit to operate for 20–60
minutes.

Jc. Connect FCV outlet (Item N) to flowmeter inlet (Item M)
with respiratory tubing. Make sure flowmeter outlet is open,
unobstructed, and properly positioned.

Sprint & Stroller

FIGURE 20: Top Fill (Push-On) QDV Assemblyd. Test flow rate at each FCV position. Record all flow rates.
e. Flow rates must be nominal valus within tolerances listed in
specification section of this manual or unit fails flow rate test.
NOTE: Be careful to allow for accuracy tolerances of flowmeter.
RP26 – Operating Pressure Test (Figure 20)
a. Assemble pressure gauge and adapter assembly (Item J).
Thread hose barb onto pressure gauge (use Teflon ® tape).
Push tubing onto hose barb and attach with clamp.
b. Connect gauge assembly (Item J) to hose barb on FCV outlet.
Open FCV (Item 24) to highest flow setting.
c. Read operating pressure on pressure gauge (Item J).
NOTE: If testing operating pressure because of improper flow
rates, test immediately after flow rate test.
d. Operating pressure must be 18-22 psig (1.2-1.5 bar) or unit
fails test.
RP27 – Flow Meter Verificationa. Flow meter accuracy is best verified by a calibration laboratory. Equipment should indicate liter per minute oxygen gas
at atmospheric pressure and 21°C (70° F).
b. Flow meter accuracy may also be tested by comparison to one
or more new, unused, calibrated flow meters. This method
will increase confidence in accuracy of readings, but not necessarily verify accuracy.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 31

X Repair Procedures
Sprint & Stroller

RP28 – Case Cleaning

RP31 – Relief Valve Test
FIGURE 21: Relief Valve Tests Set-up

NOTE: Clean case only after unit is empty and warm. Do notclean in oxygen enriched atmosphere.

49
24

a. Clean using household glass cleaner and lint-free cloth. Do
not get glass cleaner inside case or onto any valves.
37ab. Allow unit to dry thoroughly before filling.
Note: Only use disinfectants or cleaning agents approved for use
with this equipment by CAIRE such as Sporicidin®, HydroPure®,
Hydroklean®, or others as specified by CAIRE.

J

43

I

Fill Connection(QDV)

RP29 – Normal Evaporation Rate Testa. Fill unit following procedure found in the Patient Operating
Instructions (POI).
37bb. Push button and check that all 8 LEDs are on, or verify thatthe spring scale reads full.
c. Check for leaks in QDV and vent valve.
d. Let unit sit 16–18 hours.
e. Make sure FCV and vent valves are closed.

WARNING: The unit must be empty, warm and vented before
starting procedure.
a. Remove case (RP3 and RP4).

f. Do not move unit during this time.

b. Assemble pressure gauge (Item J) and adapter assembly. Threadhose barb onto pressure gauge. Push tubing onto hose barb andattach with clamp.

g. If the Stroller is at least half full (4 LEDs or 0.5 on spring scale),or any liquid remains in the unit, the unit passes. Maximum
loss rate prior to dewar servicing is 1.75 lbs./day (0,8 kg/day).

c. Assemble oxygen regulator, hose and pneumatic adapter.
Connect assembly to oxygen gas source. Connect pneumatic
adapter to unit QDV (Item 37a or 37b).

RP30 – Spring Scale Liquid Level Calibration
NOTE: Spring scale adjustments will only set the placement of theindicator and not the span (empty to full movement). The span
adjustment is fixed by the design.
a. Empty and warm portable unit (RP 15).
b. Lift the portable by the strap (directly above the spring
scale) and note the amount of misadjustment in theindicator.
c. Use a Phillips head screwdriver to adjust the internal
spring scale calibration screw, which can be accessed at thebase of the spring scale.
d. Adjust the spring scale calibration screw until the empty
red stripe indication is shown.
e. Fill unit with liquid oxygen. Verify FULL green stripe
indica-tion is shown.

d. Slowly begin increasing pressure to between 1,35-1,48
bar/19,7-21,5 psig. PRV (Item 49) should begin venting(audible noise will be heard or bubbling will be seen if leaktesting).
e. Slowly begin decreasing the pressure until the PRV (Item 49)
closes, meaning the audible noise is no longer heard or the
bubbling is no longer seen if leak testing. The pressure at which
the audible noise or bubbling is no longer seen should begreater than 1.24 bar/18psig.
f. Hold PRV (Item 49) closed (with finger) and increase pressure to 1,9-2,3 bar (27-33 psig). SRV (Item 43) must begin
open (audible venting and/or bubbling of leak test solution).
g. Slowly begin decreasing the pressure until the SRV (Item 43)
closes, meaning the audible noise is no longer heard or the
bubbling is no longer seen if leak testing. The pressure at which
the audible noise or bubbling is no longer seen should begreater than 1,68 bar/24,3 psig.
h. Slowly begin decreasing the pressure in order to release thefinger from the PRV.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 32

X Service Tools / Equipment / Supplies
1. Hex Wrenches (various sizes)

Tools and Accessories available from CAIRE Inc.

2. Flat Blade ScrewdriverPart No

Description

3. 5/16” Nut Driver

CA200071

240 AC Fluorolubricant

4. Open End Wrenches (1/2” to 1-1/8”)

CA200072”Snoop” Leak Detection Fluid

5. Side Cutters

CA400004

Replacement Filter/Male Transfer Line Adapter

6. Phillips Screwdriver

CA406308

7. Pliers

CA406310

10.3 bar (150 psi) Relief Valve Assembly
Teflon® Tape

CA406398

10.3 bar (150 psi) Relief Valve only

10662631

Service Manual

13214825

Airlift Backpack

10018138

C1000 Backpack

97112026

Female Side Fill Pneumatic Test Adapter

97200076

Erie “Liter Meter”

97217007Pressure Gauge Adapter

97212021

Male Side Fill Pneumatic Test Adapter

97403015

Capacitance Meter

97403016

Jeweler’s Screwdriver

97405279

Pneumatic Hose with DISS Fittings

97403574

Dewar Cap

97403577

0-4.1 bar (0-60 psig) Pressure Gauge

7. 02 Gas Source (High Pressure bottle)

12917730

Stroller Cart

8. 02 Liquid Source

97405147

0–45 psig Oxygen Regulator

9. N2 Gas or Clean, Dry Compressed Air Source

97405177

Portable Carrying Handle

10. Tubing (02 compatible)

97405431

Liquid Oxygen Transfer Line – 2 m (6 ft)

Required Supplies

97404564

Transfer Line Swivel Connector

1. Stay-Clean Flux

97405590

Lip Seal Service Tool

2. Cotton Swabs

97406471

Tandem Tee Kit

97406555

Super Flex Liquid Oxygen Transfer Line – 6’

97406630

Dual Fill Head Tee

13329091

G4 Capacitance Meter Adaptor Kit

8. Torque Driver/Wrenches:
12-17 N-cm (10-15 in-lbs)
23-29 N-cm (20-25 in-lbs)
69-92 N-cm (60-80 in-lbs)
104-115 N-cm (90-100 in-lbs)
6.2-6.9 N-m (45-50 ft-lbs)
9. Jeweler’s Screwdriver
Required Fixtures/Equipment
1. Capacitance Meter
2. Soldering Iron
3. Oxygen Regulator/Hose Kit
4. Pressure Gauge
5. Pressure Gauge Adapter
6. Flowmeter

3. Lead-free Solder
4. Distilled Water
5. Household Glass Cleaner
6. Lint-Free Cloth
7. Teflon® Tape

Sprint & Stroller

Required Tools

8. Fluorolubricant
9. Leak Detection Fluid

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 33

XI
1 Parts Price List
Sprint & Stroller

Contact Customer Service or visit
www.caireinc.com
to obtain your parts list.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 34

XI Ordering Information
The following steps should be used when ordering a new Sprint/
Stroller, or replacement parts for an existing unit:
1. Compile a list of all equipment and replacement parts to beordered. An exploded view of assemblies for easy parts identification, along with parts/price lists can be found on www.cairemedical.com.
For US/Canada Part Numbers, please call 770 721 7759.
For European Part Numbers, please call +44 (0) 1189 367060 . For
Asian, Australia, New Zealand Part Numbers, please call +61
297 494333.
2. Fill out a purchase order containing the following information:a. Purchase order number.
b. Name and address of billing location.
c. Name and address of shipping location.
d. Quantity, part number, description, and unit costfor each item ordered.

3. Telephone or fax CAIRE Inc. at one of the numbers listed below to begin immediate processing of the order:
USA
Toll Free Phone:

800 48 CAIRE
			(800 482 2473)
Toll Free Fax:
888 WE CAIRE(To place an order):		(888 932 2473)
Phone:		770 721 7759
Fax: 		770 721 7758

Sprint & Stroller

Ordering Information

Asia, Australia, Pacific Rim
Phone:+61 297 494333
Fax:
888-932-2473
EuropePhone:+44 (0) 1189 367060
Fax:+44(0) 118 979 9245
4. Mail or fax the completed purchase order for
confirmation to:
CAIRE Inc.
2200 Airport Industrial Drive
Suite 500
Ball Ground, GA 30107
CAIRE Medical Ltd.
Unit 6, Ashville Way
Wokingham, Berkshire, RG41 2PL
United Kingdom
All new equipment will be shipped either “prepaid”, F.O.B.
Canton, Georgia, F.O.B. Bracknell, UK, or collect via your specified carrier. All replacement parts will be sent by UPS “prepaid”,
and the shipping charges for equipment and parts will be added
to the final invoice. Payment for replacement parts are locatedon CAIRE, Inc.’s invoice with payment date indicated. All shipments will originate from Canton, Georgia or Bracknell, UK. If aparticular carrier or method of shipment is desired, specify whenplacing order.

For additional ordering and contact information, visit www.caireinc.com
Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 35

XII Return & Restocking Policy
If a problem with the unit should be encountered, reference
should be made to Troubleshooting Charts on pages 17-21. If theseprocedures do not provide a solution for the problem, the following steps should be taken:
1. Call CAIRE Inc. Customer Service. State the problem withthe unit. If it is determined that the problem cannot be solvedby the distributor, a Return Material Authorization (RMA)
number will be assigned to the unit or part(s). If a Purchase
Order Number is to be referenced, please give this number tothe Customer Service Representative at that time.
2. Carefully package the parts, or repack the unit in its original
shipping container, precisely as shipped.
3. Write the Return Material Authorization Number on the topof the shipping container.
4. Return the unit or parts by professional carrier to:
CAIRE Inc.
2000 Airport Drive
Ball Ground, GA 30107
CAIRE Medical Ltd.
Unit 6, Ashville Way
Wokingham, Berkshire, RG41 2PL
United Kingdom
All equipment returned to CAIRE Inc. must be shipped “prepaid”.
When the defective item(s) is received at CAIRE Inc. it will be serviced and returned to the distributor as soon as possible. A copyof the “Repair Cost Sheet” will be enclosed giving a detailed listingof any maintenance performed.

Restocking Policy
If it becomes necessary to cancel an order with CAIRE Inc. after the shipment has been received, use the following “Restock
Policy” procedure:
1. Notify the Customer Service Department at CAIRE Inc. using
the toll-free number. When contacting Customer Service personnel, it will be necessary to relay the following information:

Sprint & Stroller

When a Sprint/Stroller is received, it should be inspected immediately, as outlined in Section VII, Unpacking and Setup
Instructions.

a. State the quantity and description of equipment to be returned.
b. Give the Serial Number of each unit to be returned.
c. State the equipment purchase date.
2. An RMA number will be issued in the name of the distributor
by CAIRE Inc. for the equipment to be returned. When theequipment is shipped to the factory, the RMA number must
appear on the packing slip.
3. All equipment must be returned “prepaid” to the original
shipping location:
CAIRE Inc.
2000 Airport Drive
Ball Ground, GA 30107
CAIRE Medical Ltd.
Unit 6, Ashville Way
Wokingham, Berkshire, RG41 2PL
United Kingdom
4. Finally, a “Credit Memo”, minus a 15% restocking fee, will
be issued to the distributor when all equipment has been received, inspected, and restocked by CAIRE Inc.
Return of Unused Non-Defective Merchandise
CAIRE Inc., at its discretion, charges a 15% restocking fee for unused non-defective merchandise that is returned. A RMA number must be obtained from CAIRE Inc. Customer Service prior to
return of any goods. Merchandise cannot be returned for credit
after sixty (60) days. Customer to pay all freight charges. Tracking
capability and insurance on all returned goods is advised. CAIRE
Inc. will not be responsible for misdirected shipments.

Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4 36

CAIRE Inc.
2200 Airport Industrial Dr., Ste. 500 Ball Ground, GA 30107
U.S.A.

CAIRE Medical Ltd
Unit 6 Ashville Way
Wokingham
Berkshire, RG41 2PL, United Kingdom

CAIRE and CAIRE Inc. are registered trademarks of CAIRE Inc. Please visit our website below for a full listing of trademarks. Trademarks:
www.caireinc.com/corporate/trademarks/.

© Copyright 2021 CAIRE Inc. All Rights Reserved.
CAIRE Inc. reserves the right to discontinue its products, or change the prices, materials,
equipment, quality, descriptions, specifications and/or processes to its products at any time without prior notice and with no further obligation orconsequence. All rights not expressly stated herein are reserved by us, as applicable.

www.caireinc.com?????

2021 Sprint & Stroller Technical Service Manual • PN 10662631 Rev R

4