Prospenser andProspenser Plus
Superior Bottle-topDispensers

Prospenser

Prospenser Plus

Prospenser and Prospenser Plus
Dispensing Made Easy
Prospenser and Prospenser Plus are high quality bottle-top dispensers with excellent chemical resistance.
They are safe, convenient, and customizable, making them perfect for a wide range of dispensing needs.
Superior Chemical Resistance Both the ceramic piston and all parts in the liquid pathway
are highly chemical resistant, guaranteeing versatility for awide range of applications
 High quality materials guarantee a long lifetime

Easy to Use Easy and comfortable dispensing for variousapplications
 Smooth and light to operate

High Accuracy and Precision The incremental volume setup ensures accurate andrepeatable results
 High quality parts and assembly ensure smooth
piston movement

Prospenser and Prospenser Plus Features
Chemical Resistance
Prospenser and Prospenser Plus offer excellent
chemical resistance thanks to their high quality
materials. The ceramic piston is chemically stable andcompatible with an extremely wide range of liquids.
Light to Use
The operating handle is designed to be comfortable tohold while aspirating and light to press while dispensing.
The piston moves smoothly, guaranteeing excellent
results in everyday use.
Easily Replaceable Valves
Valves can be easily replaced by the user so there isno need to send the unit to be serviced.
Easy to Attach Aspiration Tube
The aspiration tube is easy to attach with a secure
threaded connection. This prevents air entering thesystem and ensures consistent results.

 ind out moreF
For more information, please visit
www.sartorius.com/en/products/pipetting/bottle-top-dispensers

Volume Setting
Prospenser and Prospenser Plus have an incremental
volume setting enabling you to set the exact volume
every time.
Versatile Accessories
There is a wide range of accessories available including
bottle adapters, dispensing heads, and aspiration tubes.
Fully Autoclavable
Prospenser and Prospenser Plus are easy to disassemble
for cleaning and are fully autoclavable.

Prospenser Plus Advanced Features

Media Recirculation System
Prospenser Plus is equipped with a
media recirculation system that allows
unused media to be recovered fromthe device and returned to the bottle.
This ensures easy priming andminimizes media loss.

Smart 360° RotatingDispensing Head
The dispensing head can be turnedto the required position without
needing to turn the bottle. Thisensures the bottle label is always
visible to the user.

Connector for Drying Tubeand Filters
Prospenser Plus has a luer connec­
tion at the back of the unit for drying
tubes and filters. Drying tubes arerecommended for humidity sensitiveliquids.

Sartorius Minisart® Syringe Filters
Sartorius Minisart® filters fit Prospenser and Prospenser Plus dispensers perfectly. Minisart® filters offer
superior flow rate, high total throughput volumes, and exceptional quality.

Prospenser and Prospenser Plus can beequipped with Minisart® filters for media
filtration. Dispensing heads with luer lockconnection are available separately.

For filtering replacement air, Prospenser
Plus can be equipped with a Minisart® filter.

Sartorius Pipettes, Pipette Tips, and Services
Our leading range of superior electronic and mechanical pipettes are the perfect solution for your liquidhandling needs. Sartorius pipettes and tips fit seamlessly together maximizing pipetting performance andensuring accuracy and precision. For more information, visit www.sartorius.com.
Sartorius also provides pipette maintenance, repair, and calibration services globally. Contact your nearest
Sartorius service center or office for more information.

Technical DetailsFeatures

Prospenser

Prospenser Plus

Volume ranges

0.2–1 mL
0.4–2 mL
1–5 mL
2–10 mL
5–30 mL
10–60 mL

0.2–1 mL
0.4–2 mL
1–5 mL
2–10 mL
5–30 mL
10–60 mL

360° rotating dispensing headNoYesMedia recirculation system

NoYes

Connector for drying tube and filters

NoYes

Threaded bottle fitting

A32

A45Bottle adapters included

A28, S40, A45

A32, A38, S40

Extremely high chemical andthermal resistance

Ceramic piston, glass cylinder, andhighly resistant fluoroplastic parts

Ceramic piston, glass cylinder, andhighly resistant fluoroplastic parts

Repeatable volume setting

Yes

Yes

Operating force

Low

Low

Aspiration tube fitting

ThreadedThreaded

Delivered with quality certificate

Yes

Yes

Easily repleacable valves

Yes

Yes

Easy to dismantle for cleaning

Yes

Yes

Fully autoclavable

121 °C, 1 bar

121 °C, 1 bar

Ordering Information
Bottle-Top Dispensers
Prospenser

Prospenser Plus

Order Code

Item

Volume Range

Order CodeItem

Volume Range

LH-723060

Prospenser

0.2–1 mL

LH-723070

Prospenser Plus

0.2–1 mL

LH-723061

Prospenser

0.4–2 mL

LH-723071

Prospenser Plus

0.4–2 mL

LH-723062

Prospenser

1–5 mL

LH-723072

Prospenser Plus

1–5 mL

LH-723063

Prospenser

2–10 mL

LH-723073

Prospenser Plus

2–10 mL

LH-723064

Prospenser

5–30 mL

LH-723074

Prospenser Plus

5–30 mL

LH-723065

Prospenser

10–60 mL

LH-723075

Prospenser Plus

10–60 mL

Note: Bottle is not included.

2ISO9001
13485
14001
Sartorius
Year
WarrantyCertified

2ISO9001
13485
14001
Sartorius
Year
Warranty
Certified

2ISO9001
13485
14001
Sartorius
Year
Warranty
CertifiedAccessories
Order CodeItem

Material

LH-721647

Dispensing head, standard

FEP

LH-721655

Dispensing head, standard

FEP

LH-721656

Dispensing head, standard

FEP

LH-721648

Dispensing head, 1.5 mm spiral

FEP

LH-721649

Dispensing head, 3.0 mm spiral

FEP

LH-721657

Dispensing head, spiral

FEP

LH-721658

Dispensing head, spiral

FEP

LH-721653

Dispensing head, luer lock

FEP/PP

LH-721654

Dispensing head, luer lock

FEP/PFA

LH-721659

Dispensing head, luer lock

FEP/PP

LH-721660

Dispensing head, luer lock

FEP/PFA

LH-721678

Aspiration tube, 310 mm

FEP

LH-721680

Aspiration tube, 310 mm

FEP

LH-721679

Aspiration tube, 400 mm

FEP

LH-721681

Aspiration tube, 400 mm

FEP

LH-721682

Recirculation tube, 70 mm

FEP

LH-721736

Bottle adapter, A25

PP

LH-721737

Bottle adapter, A28

PP

LH-721689

Bottle adapter, A28

ETFE

LH-721684

Bottle adapter, A28

PP

LH-721683

Bottle adapter, A32

PP

LH-721688

Bottle adapter, A32

ETFE

LH-721686

Bottle adapter, A38

PP

LH-721733

Bottle adapter, A38

ETFE

LH-721738

Bottle adapter, A38

PP

LH-721685

Bottle adapter, A38/430

PP

LH-721732

Bottle adapter, A38/430

ETFE

LH-721741

Bottle adapter, A45

ETFE

LH-721739

Bottle adapter, A45

PP

LH-721687

Bottle adapter, S40

PP

LH-721734

Bottle adapter, S40

ETFE

LH-721740

Bottle adapter, S40

PP

LH-721742

Bottle adapter, S40

ETFE

LH-721735

Bottle adapter, NS29/32Silicone

LH-721743

Bottle support, for bottles 75–120 mm

Silicone

 ind out moreFFor more information, please visit
www.sartorius.com/en/products/pipetting/bottle-top-dispensers

Prospenser Plus

Prospenser

Prospenser

All models

1, 2, 5 and 10 mL

30 and 60 mL

Germany
Sartorius Lab Instruments GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0

For additional information,
visit www.sartorius.com

Specifications subject to change without notice.
Copyright Sartorius Lab Instruments GmbH & Co. KG.
Status: 04 | 2022

USASartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906