SeQual® eQuinox™

PORTABLE OXYGEN CONCENTRATOR

THE PREMIER PORTABLE
CONCENTRATOR

This quiet and advanced portable oxygen concentrator is requested bypatients because of its small size and countless features. The eQuinox™
will help drive your referrals and can treat many of your oxygen users at
various activity levels.

The Most O2 to Treat the Most Users

Producing up to 3.0 LPM continuous flow and with pulse settings 1-9
with a bolus size up to 192 mL the highest of any concentrator, theeQuinox™ can cover a wide variety of oxygen users at high levels ofambulation.

The Most User-Friendly Features

Users and providers alike will get added peace of mind with the uniquemulti-language voice module that verbally announces any alarms, lowbatteries, or changes in settings. Other features include an easy access
front-loading battery, a 24-cell battery for extended durations, and ahour:minute display of remaining battery life.

A True 24/7 Device

Robust enough to run 24/7, it can be used during sleep, at home, on thego, and on the road — an all-in-one oxygen solution.
Continuous flow options eliminate the worry of desaturation during
sleep, and the high pulse output allows patients to be active with aportable device and lengthy battery life.

Advanced Clinical Features

Featuring autoSAT® Technology that detects theuser’s breath rate and automatically adjusts themotor speed to ensure oxygen is always delivered
with every breath. This helps to maintain consistent
FiO2, even in high flow patients.
The eQuinox allows oxygen delivery to be tailoredto each individual user for the best therapy. Bothsensitivity and bolus rise time can be adjusted toensure proper treatment.

www.CAIREMedical.com** SEE PRODUCT WARRANTY STATEMENT FOR COMPLETE INFORMATION.
PLEASE CONSULT THE APPLICABLE PRODUCT INSTRUCTIONS FOR USE FOR PRODUCT INDICATIONS,CONTRAINDICATIONS, WARNINGS, PRECAUTIONS, AND DETAILED SAFETY INFORMATION.
WARNING: AS WITH ANY ELECTRICALLY POWERED DEVICE, THE USER MAY EXPERIENCE PERIODSOF NON-OPERATION AS A RESULT OF ELECTRICAL POWER INTERRUPTION, OR THE NEED TO HAVETHE EQUINOX SERVICED BY A QUALIFIED TECHNICIAN. THE EQUINOX IS NOT APPROPRIATE FORANY PATIENT WHO WOULD EXPERIENCE ADVERSE HEALTH CONSEQUENCES AS THE RESULT OF
SUCH TEMPORARY INTERRUPTION.
© 2017 CAIRE Inc. All Rights Reserved ML-CONC0088 Rev D

PROVIDER FEATURESPATIENT FEATURES• 24/7 capability and reliability• Run on AC, DC or battery power• Service door for easy
maintenance without
opening cabinet• Only 14 lbs (6.4 kg)• Bill both E1390 & E1392 codes• Up to 5.9 hour battery life
at 2 Pulse• Drive referrals and retail sales• Ideal for travel programs andhospital discharge programs• Diagnostic alarm codes foreasier service calls• Multi-language voice interface• Front-loading battery• HH:MM Battery Life Display• FAA approved• Robust, adjustable cart
SPECIFICATIONSContinuous Flow 0.5–3.0 LPM

Pulse Flow Settings 1-6 (16-96 mL)
Settings 7-9 (128 – 192 mL)

Weight w/12-Cell Battery 14.0 lbs (6.4 kg)

Dimensions 13.6 x 10.6 x 7.4 inches(34.5 x 26.9 x 18.8 cm)

O2 Concentration 90% (+5.5 / -3 %)Sound Level 45 dB(A) @ 2.0 LPM

Operating Temp. 50oF to 104oF (10oC to 40oC)

Operating Altitude -1254 to 13123 ft(-382 to 4000 m)

12-Cell Battery Weight 1.85 lbs (0.84 kg)

12-Cell Battery Duration 2.8 hours @ Pulse 2
1.2 hours @ 2LPM

24-Cell Battery Weight 3.6 lbs (1.6 kg)

24-Cell Battery Duration 5.9 hours @ Pulse 2
2.8 hours @ 2LPM

Battery Re-Charge 1.5 – 3.5 hours

AC Power 87-264 VAC, 47-63Hz

DC Power 12 V Nominal

Power Draw 110 W @ 3LPM

50 W Battery Charging

Warranty 3 Years eQuinox**

CLINICAL FEATURES• Proprietary autoSAT®
Technology• Adjustable sensitivity
and rise time
• Bolus sizes up to 192 mL• O2 monitor standard
on all units• Highest pulse dose of
any concentrator• Ideal for pulmonary
rehab patients
ORDERING INFORMATION
eQuinox™ Includes: Portable Concentrator,Battery, AC Power Supply & Cord, DC Power
Cord, Removable Cart, Accessory Bag, SpareIntake Filter
4807-SEQ USA, 12 cell batt
4837-SEQ USA, 24 cell batt
4807-08-SEQ UK, 12 cell batt
4837-08-SEQ UK, 24 cell batt
4807-C1-SEQ EU, 12 cell (EN, FR, DU, ES, IT, PT, NL)
4837-C1-SEQ EU, 24 cell (EN, FR, DU, ES, IT, PT, NL)
4807-C2-SEQ EU, 12 cell (SV, DA, NO, FI)
4837-C2-SEQ EU, 14 cell (SV, DA, NO, FI)*other countries’ power supplies may beavailable - contact customer service*

ACCESSORIES

20952900
20952906
4920-SEQ
4830-SEQ
4836-SEQ
4997-SEQ
4998-SEQ
4999-SEQ
4980-SEQ

12-Cell Battery
24-Cell Battery
Accessory Bag
DC Power Cable
AC Supply (no cord)
AC Cord (US)AC Cord (EU)AC Cord (UK)
Desktop Charger (US) (cord included)

1 Year Battery & Accessories

Maintenance Schedule Air Inlet, Compressor Intake,

HEPA Filters – 1 Yr Replacement

www.CAIREMedical.com** SEE PRODUCT WARRANTY STATEMENT FOR COMPLETE INFORMATION.
PLEASE CONSULT THE APPLICABLE PRODUCT INSTRUCTIONS FOR USE FOR PRODUCT INDICATIONS,CONTRAINDICATIONS, WARNINGS, PRECAUTIONS, AND DETAILED SAFETY INFORMATION.
WARNING: AS WITH ANY ELECTRICALLY POWERED DEVICE, THE USER MAY EXPERIENCE PERIODSOF NON-OPERATION AS A RESULT OF ELECTRICAL POWER INTERRUPTION, OR THE NEED TO HAVETHE EQUINOX SERVICED BY A QUALIFIED TECHNICIAN. THE EQUINOX IS NOT APPROPRIATE FORANY PATIENT WHO WOULD EXPERIENCE ADVERSE HEALTH CONSEQUENCES AS THE RESULT OFSUCH TEMPORARY INTERRUPTION.
© 2017 CAIRE Inc. All Rights Reserved ML-CONC0088 Rev D