Challenge, Create, Customize

Modular design with a wide range of customer choices

Fiber type infrared thermometers

This series is a built-to-orderproduct line to meetyour speciﬁcations and environment

This series is a built-to-order product line

Fiber type infrared thermometers theThis FTKX series is a ﬁber type infrared thermometer that can meet customer applications such astemperature ranges, measurement distances or spot sizes.
This series does not have a circuitry on the sensor head and so is appropriate for adverse environments
or small areas.

You can choose a sensor head,a ﬁber cable and a temperature
converter to meet your requirements.
Temperature converter

You can customize
and select from morethan 140 patterns.
● Temperature range／ 100～2000℃ (9 types)*1● Measuring range／ 25mm～1000mm● Spot size／ Minimum φ0.15mm● Response speed／ 1ms (0.001s)*2*1 : Max.3200℃(special order)*2 : 0.1ms(special order)

Fiber cable

Sensor head

Usable with small targets
Spot size ; Minimum φ0.15mm

Using the quartz permeable
wavelength, it allows
measurement through glass.

The world’s fastest
response time
The world’s fastest response time of 1ms (0.001sec)
allows detection of rapid temperature changes. Thesensor head never misses sudden temperature changes.
Response time

2000

temperature
（℃）

1700

1ms

1500

Easy adjustment
The FTKX series is easy to position. This series ofproducts uses the optical coaxial LED sights. It canmeasure the target precisely.

1900
1800
1600

Measurable through glass[The optical coaxial LED sight]

1400

Measuring rangeof temperature

1300
1200
1100
1000
01

CE Mark Compliant

2

23time（ms）

4

5

The LED sightSensor head

The optical coaxial
LED sights can indicate
most of the target area,
so that it is easy toadjust positions
precisely.

Ruggedized to work in an adverse
environment with strong body

Analog output andRS232 output

By using an aluminium body, enclosure strength,
heat resistance and chemical resistance areimproved. The enclosure is rated IP67.

Analog output; 4～20mA, 0～20mA, 0～1V, mV/℃

Immune to electromagnetic ﬁelds

Supporting small areas

The sensor head does not have a circuit and not being

The sensor head is small and light, so that it can be

It is appropriate for adverse environments.

the ﬁber cable are removable.

aﬀected by electromagnetic ﬁelds.

Heat resistance up to 150℃
The sensor head and the ﬁber cable are heat

resistance up to 150℃ without being aﬀected by noise.

switchable

used in small or unstable areas. The sensor head andEasy adjustment andaccurate measurement
This series used a green LED for the sight. Green LEDs arebrighter than red LEDs. So they are much more visible.

Improved optics
Enhanced optical components are used, so that it ispossible to measure much smaller points.

Special accessories

Diﬀerent types ofsensor heads
ex; the sensor head of the optical sight

CCD Camera

Laser pointerOur laser pointer can accurately
measure the distance when the sightis unclear or long ranges from thetarget.

Please check the speciﬁcations sheet for combinations and speciﬁcations of our items.
We can accommodate your custom speciﬁcations. Please feel free to contact us with any requests.

3How to select the infrared thermometer

Selection guide for infrared thermometers
The following four factors should be considered before making your selection.

13

Temperature range
What is the desired temperature range ofoperation?

Size of the spot
What is the target size? The spot size must be
less than 80% of the diameter of the target.

24

Material of targets
You need to identify the wavelength ofmeasurement according to the material of thetarget (ﬁlm, glass or semiconductor).

Measurement of distance
Distance between the thermometer and thetarget. Generally, the longer the distance ofmeasurement, the larger the spot size.

Recommended cases

We test the board temperaturefluctuation with vapor deposited to10mm sample chips in a vacuum
tank. The temperature changes
quickly. The vacuum tank has aquartz window.
Temperature range：350℃ ～ 1600℃
Measurement distance: 500mm
Professor

Spot size: ≦φ5.0mm

Temperature range：300℃ ～ 2000℃
Measurement distance: 500mm
Spot size: φ3.0mm

Temperature range: 120℃ ～ 600℃
Measurement distance: 25mm
Electronics
parts maker

Spot size: ≦φ1.0mm

We manufacture steel pipes that
we bend by high-frequency
heating. We have to measure a gapof 10mm between the surrounding
coils. The air in the workplace is not
good, because we splash coolant
water, however we have instrument
air blowing equipment.
Temperature range: 900℃ ～ 1200℃
SteelmakerMeasurement distance: 300mm
Spot size: ≦φ4.0mm

High-speed response time

We recommend the FTKX-PNE0300-0500B201-000.
This thermometer has a wavelength that goes through quartz,
so that it is able to measure the target through glass. The
response time is extremely fast, with a speed of 1ms (0.001s),
it never misses a sudden temperature change. It is possible tosecure the measurement because the spot size is smaller than
Φ5.0mm.

Temperature range：100℃ ～ 1500℃

We have to measure soldering
points by YAG laser. We need tomeasure tiny points because thetarget area is very small.

Through glass

Measurement distance: 25mm

Tiny spot

Spot size: φ0.6mm

We recommend the FTKX-TNE0100-0025B601-000.
This is able to measure the temperature of a minute point (φ0.6mm) and suitable for measuring a soldering point or a small
point of devices. This is able to measure regardless of thewavelength of the YAG laser. The optical coaxial LED sight canindicate most of the target area, so that it is easy to adjust thetarget in the same way as human eyes.

Temperature range：600℃ ～ 2000℃
Measurement distance: 300mm
Spot size: φ4mm

Adverse environment

We recommend the FTKX-ANE0600-0300S201-000.
This is able to measure a stable target because the sensor
head has no eﬀect on radio frequency. You can use an air
purge hood for protection of lens from stain if the target is inan adverse environment. By putting the temperature converter
into the electronic control panel cabinet, and putting anextension cable on the ﬁber cable, it is able to measure astable target without being aﬀected by noise.

4Application examples

Laser soldering equipment
・You need to monitor the temperatureof soldering

・You need to check the point of laser
welding.

Microwave burning furnace
・You need to monitor the temperatureduring (a) food processing, (b) tires
and rubber processing, (c) powder ofphosphor, plastic or chemical material,(d) heating and drying-out (e) heating,welding or drying-out of ceramic,(f) glass or hollow ﬁber membrane.

Inside of vacuum tank・You need to measure the target's
temperature inside of vacuum tankthrough glass window.

Temperature control
immediately after seam welding
・You need to monitor the temperature
at seam welding used for products
such as steel cans.

Temperature of ﬁlament orelectrode in electric bulb.
・Measuring the temperature of ﬁlament
in order to prolong electric bulb.

Product on conveyor
・You need to measure the temperatureof products on conveyor belt.

Temperature control of
Si/SiC/GaN single crystal
production equipment
・You need to monitor the temperatureof growing a silicon ingot.

Please contact us if you do not ﬁnd your applications in the examples here.

5Measuring the temperature
before press-ﬁtting of insert
nut・You need to make sure that theinsert nut has been precisely heatedat the right temperature.

Measurement of fusing
・You need to measure thetemperature during coils orinductors fusing.

High-frequency heating・You need to measure and monitor
the temperature of the target
between a gap of coils or top andbottom of coils.

Hot metal
・You need to monitor thetemperature of the metal plates
during its rolling to keep a constantquality.

Metal forging
・You need to measure and monitor
the temperature during metalforging.

Metal drawing
・You need to monitor thetemperature during metal drawing.

Thermode tip
・You need to monitor the temperatureduring thermode tip implanting.

Selections of Displays and Setting Units for your applications
Unit to be built into customer control panel

Small type

TMCX-N

Large type

TMCX-H

Analog output
Alarm output
Emissivity input

Analog output
Alarm output
Emissivity input
RS485

Touch screen type to check on site
Color LCD, The graph view shows
the history of the measured temperature.

Portable type

Touch screen tyape

TMCX-TDE-110

Analog output
Alarm output
microSD card

Parameter setting unit

Conﬁgurable parameters for diﬀerent
types of thermometers

Monitoring on PC

PWCX※Analog outputs via optional
branch cable(Model : TMBX-B01)

Batteries type

Parameter setting kit
PC software

Parameters and the graph view ofthe measured temperature
can be checked on your PC

Licensed : PWSX
USB-RS232C converter : PWUX
※Analog outputs via optional
branch cable(Model : TMBX-B01)

Accessories
Mounting bracket
for round head

Mounting bracket
for square head

Tripod mounting bracket
for square head

Mounting bracket
for temperature
converter

Protective window

Air purge hood toprotect lens

FTX9-3

FTX9-1FTX9-2

TMAX-A

TMDX-06S、TMDX-15S
TMDX-25S

TMPX-06

Air purge hood toprotect lensAirless dust hood

Right-angled mirror(90degrees)

Fine adjustment stage

Extension cable

Branch cable

FTP9-6、FTP9-15

TMNX-06、TMNX-15
TMNX-25

TMLX-06S、TMLX-15S
TMLX-25S

FTT9-3

TMBX-E05

TMBX-B01

JAPAN SENSOR CORPORATION
http://www.japansensor.co.jp/■ If you have any questions, contact us at the addressor links shown below.

info@japansensor.co.jp

The contents described in this catalog are subject to change for improvement without notice.

FTKX-Cata-V201906P201906 1K