USER’S MANUAL

DaqScan/2000 Series

Ethernet-Based Data Acquisition System Components

DaqScan/2001
DaqScan/2002
DaqScan/2004
DaqScan/2005

DaqScan/2000 Series
1126-0901 rev 2.2*327556B-01*
372556B-01

Measurement Computing
10 Commerce Way
Norton, MA 02766(508) 946-5100
Fax: (508) 946-9500
info@mccdaq.com

www.mccdaq.com

ii

Warranty Information

Contact Measurement Computing by phone, fax, or e-mail in regard to warranty-related issues:
Phone: (508) 946-5100, fax: (508) 946-9500, e-mail: info@mccdaq.com

Limitation of Liability

Measurement Computing cannot be held liable for any damages resulting from the use or misuse of this product.

Copyright, Trademark, and Licensing Notice

All Measurement Computing documentation, software, and hardware are copyright with all rights reserved. No part ofthis product may be copied, reproduced or transmitted by any mechanical, photographic, electronic, or other method
without Measurement Computing’s prior written consent. IOtech product names are trademarked; other product names, asapplicable, are trademarks of their respective holders. All supplied IOtech software (including miscellaneous support
files, drivers, and sample programs) may only be used on one installation. You may make archival backup copies.

CE Notice

Many Measurement Computing products carry the CE marker indicating they comply with the safety and emissions
standards of the European Community. When applicable these products have a Declaration of Conformity stating which
specifications and operating conditions apply. You can view the Declarations of Conformity at
www.mccdaq.com/legal.aspx (CE Information page).

Warnings, Cautions, Notes, and Tips

Refer all service to qualified personnel. This caution symbol warns of possible personal injury or equipment damage
under noted conditions. Follow all safety standards of professional practice and the recommendations in this manual.
Using this equipment in ways other than described in this manual can present serious safety hazards or cause equipment
damage.
This warning symbol is used in this manual or on the equipment to warn of possible injury or death from electricalshock under noted conditions.

This ESD caution symbol urges proper handling of equipment or components sensitive to damage from electrostatic
discharge. Proper handling guidelines include the use of grounded anti-static mats and wrist straps, ESD-protective bags
and cartons, and related procedures.

This symbol indicates the message is important, but is not of a Warning or Caution category. These notes can be of great
benefit to the user, and should be read.
In this manual, the book symbol always precedes the words “Reference Note.” This type of note identifies the locationof additional information that may prove helpful. References may be made to other chapters or other documentation.
Tips provide advice that may save time during a procedure, or help to clarify an issue. Tips may include additionalreference.

Specifications and Calibration
Specifications are subject to change without notice. Significant changes will be addressed in an addendum or revision to
the manual. As applicable, the hardware is calibrated to published specifications. Periodic hardware calibration is notcovered under the warranty and must be performed by qualified personnel as specified in this manual. Improper
calibration procedures may void the warranty.

CAUTION
Using this equipment in ways other than described in this manual can cause
personal injury or equipment damage. Before setting up and using yourequipment, you should read all documentation that covers your system.
Pay special attention to Warnings and Cautions.

Note:

During software installation, Adobe® PDF versions of user manuals will automaticallyinstall onto your hard drive as a part of product support. The default location is in the
Programs group, which can be accessed from the Windows Desktop. Initial navigation
is as follows:
Start [Desktop “Start” pull-down menu] Programs
 IOtech DaqX Software
You can also access the PDF documents directly from the data acquisition CD by using
the <View PDFs> button located on the opening screen.
Refer to the PDF documentation for details regarding both hardware and software.

DaqScan2000.pdf
PDF

Contains hardware-related material for DaqScan/2000 Series devices. This pdf fileplus the following constitute a complete set of documentation for these devices.

1126-0901

DaqView_DaqViewXL.pdf

Discusses how to install and use these “out-of-the-box” data acquisition programs.
PDF

457-0909

PostAcquisition Analysis.pdf
PDF

1086-0926
1086-0922

iv

This pdf consists of two documents. The first discusses eZ-PostView, a post dataacquisition analysis program. The application is included free as a part of productsupport. The second includes information regarding eZ-FrequencyView andeZ-TimeView. These two applications have more features than does eZ-PostView
and are available for purchase. They can; however, be used freely during a 30-day
trial period.

947091

DaqScan/2000 Series

DBK Options.pdf
PDF

457-0905

The DBK Option Cards and Modules User’s Manual discusses each DBK product.
In addition, the document includes chapters on signal management, system
connections, pinouts, DBK set up in software, and troubleshooting.

ProgrammersManual.pdf
PDF1008-0901

The Programmer’s Manual pertains to developing custom programs using Applications
Program Interface (API) commands.
Programmers should check the readme.file on the install CD-ROM for the location ofprogram examples included on the CD.

Manual Layout
Chapter 1 – Daq Systems
Chapter 2 – DaqScan/2000 Series, Device Overviews
Chapter 3 – Block Diagrams
Chapter 4 – Connectors, Indicators, Cables, & Pinouts
Chapter 5 – Software
Chapter 6 – Configuring the System for Operation
Chapter 7 – CE Compliance and Noise Considerations
Chapter 8 – Taking Measurements
Chapter 9 – Calibration
Chapter 10 – Troubleshooting and Customer Support
Chapter 11 – Specifications
Glossary
Reference Note:
A reference to additional documents of importance begins on page iv.

DaqScan/2000 Series

947091v

This page is intentionally blank.

vi

947091

DaqScan/2000 Series

Table of Contents
Quick Start
1 – Daq Systems
2 – DaqScan/2000 Series, Device Overviews
Overview …… 2-1
Product Features …… 2-2
System Applications …… 2-3

3 – Block Diagrams
4 – Connectors, Indicators, Cables, & Pinouts
DaqScan/2000 Series Hardware …… 4-1
Cables ……4-3
Pinouts ……4-5

5 – Software
6 – Configuring the System for Operation
7 – CE Compliance and Noise Considerations
8 – Taking Measurements
Introduction …… 8-1
Scanning …… 8-1
Triggering …… 8-2
Synchronous I/O Operations …… 8-3
Asynchronous I/O Operations …… 8-5
Synchronizing Multiple Units …… 8-6
Using DBK Cards and Modules for Signal Conditioning …… 8-8

9 – Calibration
10 – Troubleshooting and Customer Support
Electrostatic Discharge (ESD), Handling Notice…… 10-1
Product Care …… 10-1
ReadMe Files and the Install CD-ROM …… 10-2
Driver Support…… 10-2
DaqCOM Issues …… 10-2
Ethernet Problems …… 10-3
Frequently Asked Questions …… 10-4
Customer Support …… 10-5

11 - Specifications
Glossary

DaqScan/2000 Series

898095vii

This page is intentionally blank.

viii

898095

DaqScan/2000 Series

DaqScan/2000 Series

Quick Start

Ethernet-Based Data Acquisition Components
Before you get started
Verify that you have the following items.
•
•
•
•
•
•
•
DaqScan/2001

DaqScan/2000 Series Device
AC Power Cord
Ethernet Patch Cable
Ethernet jack [on PC or on a hub connected to the Ethernet]
Data Acquisition CD
Monitor: SVGA, 1024 x 768 screen resolution

Windows 2000 and Windows XP users:
PC with Intel™ Pentium, 1 GHz or equivalent;
512 MB memory; 10 GB disk space• Windows Vista users:
PC must be Windows Vista Premium Ready

Step 1 - Install Software
1.
2.

Close all running applications on the host PC.
Insert the Data Acquisition CD into your CD-ROM drive and wait for the CD to auto-run. An Opening
Screen will appear. As an alternative, you can download software from www.PFFGDT.com/VRIWZDUH

3.

Click the <ENTER SETUP> button.
Note: If you are downloading software from our website, follow instructions provided there.

4.

From the hardware selection screen [which follows a licensing agreement], select thedevice from the drop-down list and follow the on-screen instructions.

Step 2 - Connect the DaqScan to the Ethernet

In the figure, the DaqScan is connected directly to an Ethernet port on a host computer. Please consult your user’s
manual (located in PDF format on the CD) should you need information regarding network types other than this simple
direct connection.
1.

Connect the Ethernet cable to the Ethernet jack on the DaqScan rear panel.

2.

Connect the other end of the Ethernet cable to the Ethernet jack on the host computer or
network hub.

Step 3 - Connect the DaqScan to Power
1.

With the DaqScan’s power switch “OFF,” connect the power cord to the unit.

2.

Connect the cord’s plug to a standard AC outlet; then turn the DaqScan Power switch to “ON.”1126-0940, rev 1.0

324633A-01

Printed in Hungary

Step 4 - Configure Computer Network Settings
Note: We recommend that you discuss this procedure with your Network Administrator before proceeding.

1.
2.
3.
4.
5.
6.
7.

8.
9.

Open the Control Panel by navigating from the Windows Desktop as follows:
Start Menu ⇒ Settings ⇒ Control Panel.
Double-click the “Network Connections” icon.
Double-click the icon for the network that the 2000 Series device is connected to.
In the “Local Area Connection Status” box, click on the <Properties> button. The “Local Area Connection
Properties” box will appear.
Double-click the “Internet Protocol (TCP/IP)” component. The “Internet Protocol (TCP/IP) Properties” boxwill appear.
Select the “Use the following IP Address” radio
button. (See figure at left).
Set the IP address field to 10.0.0.x
where x is some number from 1 to 254.
Make sure that each computer and each device on
the dedicated network has a unique IP address.
Set the Subnet mask to 255.0.0.0. Note that theremaining fields can be left as is. Click <OK>.
Internet Protocol (TCP/IP) Properties
Click <OK> on follow-up screens to exit.
(Partial View)

Step 5 - Configure & Test the System
1.

Open the Daq Configuration Applet.
a.
b.

Navigate from the Windows’ Desktop: Start Menu ⇒ Settings ⇒ Control Panel
From the Control Panel, double-click the Daq Configuration icon.

2. Add the 2000 Series Device to the list of installed devices.
a.
b.
c.
d.
e.
f.
g.

Select the Computer image in the Device Inventory configuration tree.
Click the <Add Device> button. The “Select Device Type” box will appear.
Select the DaqScan/2000 Series device from the list of devices.
Click the <OK> button. The “Properties” box will appear for the selected device.
Enter the Serial Number of the device. The number is on the MAC label (rear panel, upper left). The
serial number is located just below a barcode.
Select the “Auto IP Setting” radio button. The IP Address of the device will automatically becalculated and displayed in the IP Address field
Click the <OK> button.

3. Test the system connections.
a.
b.
c.
d.
e.

Make sure the device has been properly installed and is powered-on.
Make sure all cables are properly and securely connected.
Click the “Test Hardware” tab.
Click the <TCP/IP Test> button. This tests the Transmission Control Protocol / Internet Protocol.
Upon completion of the TCP/IP test, click the <Resource Test> button.
When testing, if the unit does not respond within 30 seconds perform the following steps:
1) reboot the system, 2) upon power-up, re-open the Daq Configuration applet, 3)select anotherconfiguration setting, 4) reinitiate the test.

Step 6 - Connect Data Acquisition Signal Lines
Prior to making signal connections review the Specifications chapter of your user’s manual to ensure that yourintended signal inputs do not exceed the specified limits. The manual is included in PDF format on your CD.

Measurement Computing
10 Commerce Way
Norton, MA 02766(508) 946-5100
Fax: (508) 946-9500
info@mccdaq.com

www.mccdaq.com*324633A-01*
324633A-01

Printed in HungaryDaq Systems

1

Daq Systems, the Modular Concept …… 1-1
DaqBooks, DaqBoards, and Daq PC-Cards ….. 1-2
Using DBK Cards and Modules for Signal Conditioning …… 1-4
Daq Software …… 1-5

This chapter provides a brief overview of Daq devices.
Users interested in jumping straight to DaqScan information
should skip directly to chapter 2.

Daq Systems, the Modular Concept
Daq equipment and software form a modular, interrelated family of products that provide
great flexibility in data acquisition system design. This flexibility allows for the developmentof custom systems that are unique to the user, and which can be optimized for his or her
specific application needs. With the Daq product line, system expansion or redesign cantypically be accomplished with relative ease.

•

Primary Acquisition Device. This is the main data acquisition device, e.g., a
DaqBook, DaqBoard, DaqLab, DaqScan, or Daq PC-Card. These devices provide a vital
data conversion and communications link between the data source of transducers andsignal conditioners and the data processor of the host computer.

•

DBK Option Cards and Modules. Over 35 DBK cards and modules (the number isconstantly growing) provide various types of signal conditioning and system expansion.
Note that certain DBK modules exist for the purpose of supplying power to othermembers of the acquisition system. The DBK options are discussed in the DBK Option
Cards and User’s Manual (p/n 457-0905).
Note: Only passive DBKs, such as the DBK1 BNC module, the DBK11A screw terminal
card, and the DBK40 BNC analog interface, can be used with a Daq PC-Cards.
Reference Note:
DBK options are discussed in the DBK Option Cards and Modules User’s
Manual (p/n 457-0905). As a part of product support, this manual is
automatically loaded onto your hard drive during software installation. The
default location is the Programs directory, which can be accessed through theWindows Desktop.

•

Software. DaqView out-of-the-box software provides a graphical user interface witheasy to read spreadsheet formats for viewing channel data, as well as a choice of analog,
digital, and bar-graph meters. Waveform analysis can be performed, when applicable.
A product support option, included on the data acquisition CD, provides a means of
performing post data analysis. More information is included in the software-specific PDF
documents that are installed on your hard-drive as a part of product support.
In addition to the included out-of-the-box software, Daq products can be controlled via
user-written custom programs through Applications Program Interface (API). Several
languages are supported, e.g., C/C++, and VisualBASIC.
Note that third party icon-driven software, such as DASYLab and LabView, can be usedwith the Daq devices.

DaqScan/2000 Series

949595

Daq Systems

1-1

DaqView and DASYLab can only be used with one DaqScan/2000
Series device at a time.
LabView can be used with multiple Daq/2000 Series devices.
For multiple device applications (via custom programming) refer tothe Using Multiple Devices section of the Programmer’s Manual.
During software installation from the data acquisition CD(p/n 1022-0601), a PDF version of the Programmer’s Manual is
automatically loaded onto your hard drive as a part of productsupport. The default location is the Programs directory.

Reference Note:
Programming topics are covered in the Programmer’s User Manual(p/n 1008-0901). As a part of product support, this manual is automaticallyloaded onto your hard drive during software installation. The default locationis the Programs directory, which can be accessed through the Windows
Desktop.

DaqBooks, DaqBoards and Daq PC-Cards
Daq products typically connect to one or more DBKs on their signal input side and to a hostcomputer on their output side. The following table categorizes the various Daq devices andprovides very basic information.

1-2

Daq Systems

949595

DaqScan/2000 SeriesDaq Data Acquisition Devices
Category

DevicePrimary
Data
Acquisition
DeviceDaqLab/2001
DaqLab/2005

Ethernet-Based Laboratory Data Acquisition
16-bit, 200 kHz devices with Ethernet connectivity.
Each can serve as host to an internal analog and/or digital DBK
card. Each has a built-in AC-to-DC power circuit and can beplugged directly into a standard AC outlet.

DaqScan/2001
DaqScan/2002
DaqScan/2004
DaqScan/2005

Ethernet-Based Data Acquisition System Components
16-bit, 200 kHz devices with Ethernet connectivity.
Each has a built-in AC-to-DC power circuit and can be plugged
directly into a standard AC outlet.

DaqBook/2001
DaqBook/2005
DaqBook/2020

Portable Data Acquisition
16-bit, 200 kHz devices with Ethernet connectivity.
Make use of an external power supply.

DaqOEM/2001
DaqOEM/2005

Portable Data Acquisition
16-bit, 200 kHz boards with Ethernet connectivity.
These boards reside external to the host PC.

DaqBook/100
Series
DaqBook/200
SeriesPortable Data Acquisition
12-bit: DaqBook/100, /112, /120
16-bit: DaqBook/200, /216, /260

DaqBook/2000A
DaqBook/2000E
DaqBook/2000X

Portable Data Acquisition
16-bit, 200 kHz devices with Ethernet connectivity, though only
the DaqBook/2000E provides for a direct Ethernet connection.
These devices include a 100-pin P4 connector.
Each makes use of an external power supply.

DaqBoard/2000
SeriesPlug-In Boards for PCI Bus-Slots
16-bit , 200 kHz. Six boards identified as /2000 through /2005

DaqBoard/2000c
SeriesPlug-In Boards for Compact-PCI Bus-Slots
16-bit , 200 kHz. Six boards identified as /2000c through/2005cDaqBoard(ISA types)*

Plug-In Boards for ISA Bus-Slots
12-bit: DaqBoard/100A, /112A
16-bit: DaqBoard/200A, /216A, /2000

Daq PC-Card*

Plug-In PCMCI Card
12-bit: Daq/112B
16-bit: Daq/216B

Analog Signal
Conditioning

Cards and DBK modules used to condition Analog Signals
DBK/ 4, 7, 8, 9, 12, 13, 15, 17, 18, 19, 42, 43A, 44, 45, 50, 51,
52, 53, 54, 55, 65, 80, 81, 82, 83, 84, 85, 90, 100, 207,
207/CJC

Analog Output

Cards used to modify Analog Output Signals
DBK/ 2, 5

Digital I/O andControlCards and DBK modules used to condition Digital I/O

Expansion
Connections

Cards and DBK modules used to expand the acquisition
system.
DBK/ 1, 10, 11A, 35, 40, 41, 60, 200, 201, 202, 203, 204, 205,
206, 209

Power Supply

DBKs: 30A, 32A, 33, 34

Included Software

DaqView, Post Data Acquisition Analysis Program (actual
application not specified), Visual Basic extensions, Application
Programming Interface (API)

Optional Software

DaqViewXL, DASYLabDBK Option
Cards and
Modules*

SoftwareDescription

DBK/ 20, 21, 23, 24, 25, 208, 210* The list of DBK cards and modules continues to grow. For information regarding new and upcoming
products consult out website, your sales representative, or contact the factory directly.

DaqScan/2000 Series

949595

Daq Systems

1-3

Using DBK Cards and Modules for Signal Conditioning
The DBK signal-conditioning cards and modules are designed for use with DaqLab, DaqScan,
DaqBook, LogBook, and various data acquisition boards, i.e., ISA, PCI, and compact-PCI
types. The DBKs perform best when used with an acquisition device that can dynamicallyselect channel, gain, and range. DBK cards and modules with dynamic channel andgain/range selection allow for high channel-to-channel scan rates with a variety oftransducers.
Note: Only passive DBKs, such as the DBK1 BNC module, the DBK11A screw terminal card,and the DBK40 BNC analog interface, can be used with Daq PC-Cards.
DBK output signals can be bipolar, e.g., -5 to +5 V, or unipolar, e.g., 0 to 10 V. The user canselect a range of relevant values to correspond to the lowest signal (e.g., -5 or 0 V) and thehighest signal (e.g., 5 or 10 V) signal. This type of range selection guarantees the highest
resolution in 12-bit or 16-bit conversion.
DBK modules share the same footprint as the DaqBook and a typical notebook PCs; allowing
for convenient stacking. The majority of these modules have their own power supply;
however, several options exist for packaging and powering the DBKs.

Reference Note:
DBK options are detailed in the DBK Option Cards and Modules User’s Manual(p/n 457-0905). As a part of product support, this manual is automatically loaded
onto your hard drive during software installation. The default location is thePrograms directory, which can be accessed through the Windows Desktop.

1-4

Daq Systems

949595

DaqScan/2000 SeriesDaq Software
The Daq devices have software options capable of handling most applications. Three types ofsoftware are available:•

ready-to-use graphical programs, e.g., DaqView, DaqViewXL, and post acquisition
data analysis programs such as eZ-PostView•

drivers for third-party, icon-driven software such as DASYLab and LabView•

various language drivers to aid custom programming using API

Ready-to-use programs are convenient for fill-in-the-blank applications that do not require
programming for basic data acquisition and display:•

DaqView is a Windows-based program for basic set-up and data acquisition. DaqViewlets you select desired channels, gains, transducer types (including thermocouples),and a host of other parameters with a click of a PC’s mouse. DaqView lets you
stream data to disk and display data in numerical or graphical formats. PostView is apost-acquisition waveform-display program within DaqView.

•

DaqViewXL allows you to interface directly with Microsoft Excel to enhance data
handling and display. Within Excel you have a full-featured Daq control panel and all
the data display capabilities of Excel.

•

A post acquisition data analysis programs, e.g., eZ-PostView, typically allows you toview and edit post-acquisition data.

•

The Daq Configuration control panel allows for interface configuration, testing, andtroubleshooting.

Each Daq system comes with an Application Programming Interface (API) via a subroutine
library (DAQ) or an ActiveX/COM based library (DaqCOM). API-language drivers include:
C/C++, and Visual Basic. The latest software is a 32-bit version API.

Reference Notes:
¾

¾

PDF
Note:

The software document modules, DaqView, DaqViewXL, and Post Acquisition
Data Analysis User’s Guide, are not included as part of the hardcopy manual,
but are available in PDF version. See the PDF Note, below.
Programming topics are covered in the Programmer’s User Manual (10080901). As a part of product support, this manual is automatically loaded onto
your hard drive during software installation. The default location is thePrograms directory, which can be accessed through the Windows Desktop.

During software installation, Adobe® PDF versions of user manuals willautomatically install onto your hard drive as a part of product support. The
default location is in the Programs group, which can be accessed from theWindows Desktop. Refer to the PDF documentation for details regarding both
hardware and software.
A copy of the Adobe Acrobat Reader® is included on your CD. The Reader
providesa means of reading and printing the PDF documents. Note that hardcopy
versions of the manuals can be ordered from the factory.

DaqScan/2000 Series

949595

Daq Systems

1-5

1-6

Daq Systems

949595

DaqScan/2000 Series

DaqScan/2000 Series

2
Device Overviews

Overview …… 2-1
Product Features …… 2-2
System Applications …… 2-3

Overview
DaqScan/2000 Series devices are high-speed, multi-function, data acquisition devices for use
with Ethernet PCs and Ethernet networks. Each unit connects directly to a PC’s Ethernet port
or network hub. All I/O is accessed via female DB37 connectors located on the rear of theunit, making cabling easy from the DaqScan to the test device.
The DaqScan/2000 Series includes four models, ranging from the DaqScan/2002, which offers
digital/frequency I/O, to the full-featured DaqScan/2001 which includes 16 analog inputs, 4
analog outputs, 40 digital I/O, 4 counter inputs and 2 timer outputs. Each features a 16-bit,
200 kHz A/D converter and digital calibration. In addition, all four units include an internal
AC-to-DC power converter.

I/O Comparison Matrix
Model

Analog Inputs
SE / DIFF

DigitalI/OCounter
InputsTimer
Outputs

AnalogOutputs

DaqScan/2001

16/8

40

4

2

4

DaqScan/2002

0

40

4

2

0

DaqScan/2004

0

40

4

2

4

DaqScan/2005

16/8

40

4

2

0

Synchronous Operation Matrix
Operation

DaqScan/
2001

DaqScan/
2002

DaqScan/
2004

DaqScan/
2005

Yes
Yes
Yes
Yes
Yes

No
No
Yes
Yes
Yes

No
No
Yes
Yes
Yes

Yes
Yes
Yes
Yes
Yes

Yes

No

Yes

No

Yes
Yes
Yes

Yes
Yes
Yes

Yes
Yes
Yes

Yes
Yes
Yes

Yes

No

Yes

No

Synchronous Input
Analog Main Unit Inputs
Analog Expansion Input
Counter Inputs
Digital Main Unit Inputs
Digital Expansion Inputs

Synchronous Output
Analog D/A Waveform Output

Asynchronous IO
Main Unit Digital I/O
Expansion Digital I/OTimer Output(Pulse Generation)
Analog Output

DaqLab/2000 Series

898095

Device Overviews

2-1

Each DaqScan/2000 Series model is packaged in 1U high (1.75”), 19” wide full-rack package,and includes a rackmount kit that can attach to either the front or the rear of the enclosure.
Multiple DaqScan models can be combined in the same system. Up to 4 units can besynchronized [scan synchronous post-trigger] via rear panel SYNC connectors.
DaqScan/2000 Series devices include comprehensive drivers for popular Windows-based
environments, including Visual Basic, C/C++, ActiveX/COM, LabVIEW, MATLAB, and DASYLab.
The devices include DaqView, an interactive spreadsheet-style application that is ideal forverifying signal connections during system design.
Reference Notes:¾ Block diagrams are presented in chapter 3.
¾ Connectors and pinouts are discussed chapter 4.

Product Features• Can be used to add analog I/O, digital I/O and frequency I/O to Ethernet-based test
systems. Each DaqScan/2000 Series device has a different combination of features(see comparison matrix on page 2-1).
• All I/O can be synchronous, enabling precise timing between various I/O functions
Up to 4 units can be synchronized [scan synchronous post-trigger] via rear panel
SYNC connectors.

• 8 differential, or 16 single-ended inputs, expandable up to 256 voltage or 896 TC
channels using signal conditioning and expansion options• Up to 40 built-in TTL-level digital I/O, expandable up to 256 channels of isolated I/Ousing low-cost isolation modules• Configuration through software only. Device settings are made via software. There
are no hardware settings to be made, for example, there are no device DIP switches
to be set or jumpers to be positioned. However, note that DBK expansion options
may require s small degree of manual set up. Refer to the DBK Option Cards andModules User’s Manual (457-0905) for details. A PDF version of the document isincluded on the installation CD.
•

Includes support for Visual Basic®, C/C++, ActiveX/COM, LabVIEW®, MATLAB®, andDASYLab®

• Includes Out-of-the-Box™ DaqView™ software to facilitate signal and wiring
verification with no programming required• Convenient 1U high (1.75”), 19” wide rack mount package minimizes rack space intest systems•

Four 16-bit counters. Each can accept frequency inputs up to 10 MHz. The counters
can be cascaded into two 32-bit counters. Counter connection is via a removablescrew terminal block on the front panel.

•

Two 16-bit timer outputs. Each can generate square waves from 16 Hz to 1 MHz.
Connection to the timer outputs is via a removable screw terminal block on the front
panel• Scan Sequencing. Each DaqLab/2000 Series device has a scan sequencer that canselect up to 16384 channel/range combinations. Scan rates are programmable on a
per channel basis to 5 µs/channel, 10 µs/channel, or 1000 µs/channel.
Note:

2-2

Device Overviews

Inputs can be measured at the 5 µs/channel rate. However, some DBK
expansion options require more settling time and can be measured at theslower rate of 10 µs/channel or 1000 µs/channel. All types of measurements
can be mixed within the scan group.

898095

DaqLab/2000 Series

System Applications
In addition to basic applications that make use of a DaqScan model’s built-in I/O, a wide
variety of signal conditioning and expansion options make additional applications possible.
Each of the following sample systems are obtained by using a DaqScan along with one ormore DBK options. The applications, which are only briefly discussed, include:
•
•
•
•
•

High Channel Count Thermocouple Measurements
High-Isolation Voltage & Thermocouple Measurements
Isolated Digital I/OHigh Speed Voltage Measurement
Multifunction I/OHigh Channel Count Thermocouple Measurements
When combined with 56-channel DBK90 thermocouple input modules, DaqScan can measure
up to 896 channels of thermocouple (TC) input. Any TC type can be installed into any channel
using standard mini-TC connectors. DBK90’s built-in cold junction compensation (CJC)
coupled with the software’s TC conversion algorithms make temperature measurements easy.
Note that thermocouples are measured at a rate of 1 ms/channel in DBK90 based systems.
High-Isolation Voltage & Thermocouple Measurements
DaqScan units can be combined with DBK207/CJC™ options to create an isolated systemcapable of measuring up to 256 channels of voltage, thermocouple, RTD and strain gage
inputs. All input channels can be scanned up to 200 kHz, and are isolated by 500V from otherchannels and from system common. Any combination of input signals is possible by selecting
the appropriate 5B signal conditioning module for the DBK207/CJC.
Isolated Digital I/O System
When the DaqScan/2002 is combined with a DBK208 or DBK210, up to 256 channels of
isolated digital I/O are possible, capable of controlling AC or DC voltages up to 250V, orsensing the presence of AC or DC voltages. All channels are isolated from one another andfrom the system by up to 500V. Inputs can be scanned in 16-bit increments at speeds up to
200 Kreadings/s.
High-speed Voltage Measurement System
The DaqScan/2005 can be combined with DBK85 16-channel voltage input modules. This
creates a 5 µs/channel voltage measurement system with up to 256 channels. Each input canhave a range from 156 mV FS to 10V FS, which is programmable on a per-channel basis.
Multifunction I/O System
All of the aforementioned capabilities can be combined into one system, using one
DaqScan/2001 as the heart of the system. Such a DaqScan/2001 system could provide
56 non-isolated TC inputs, 16 isolated voltage inputs, 16 low-voltage inputs, 4 analog outputs,
4 frequency inputs, and 32 isolated discrete high-voltage outputs.
Reference Note:
For information regarding DBK options refer to the DBK Options User’s Manual(p/n 457-0905). A PDf version of the manual is included on your installation CD.

DaqLab/2000 Series

898095

Device Overviews

2-3

2-4

Device Overviews

898095

DaqLab/2000 SeriesBlock Diagrams

3

DaqScan/2001 Block Diagram

DaqScan/2000 Series

947495Block Diagrams

3-1

DaqScan/2002 Block Diagram

DaqScan/2004 Block Diagram

3-2

Block Diagram

947495

DaqScan/2000 Series

DaqScan/2005 Block Diagram

DaqScan/2000 Series

947495Block Diagrams

3-3

3-4

Block Diagram

947495

DaqScan/2000 Series

Connectors, Indicators, Cables, & Pinouts

4

DaqScan/2000 Series Hardware …… 4-1
Cables …… 4-3
Pinouts …… 4-5

DaqScan/2000 Series Hardware

DaqScan Front Panel
Front Panel
The DaqScan/2000 Series front panels contain a power indicator LED. The power switch andconnectors are on the rear panel.

DaqScan/2001, Rear Panel
Rear Panel
The DaqScan/2000 Series rear panels each contain a Power Switch, AC power cord connector,
2 Synchronization jacks, a MAC Address Label, Ethernet connector, 5 LEDs [ 2 of which are on
the Ethernet connector], and a set of DB37 connectors. All four DaqScan models have P2 andP3 connectors; but in addition, DaqScan/2001 and DaqScan/2005 have P1 connectors foranalog I/O.
P1 - DB37 Connector: P1 applies to DaqScan/2001 and DaqScan/2005 only. P1 is theanalog input port for 16 single-ended analog inputs or 8 differential, programmable on a perchannel basis as single-ended or differential and as uni-polar or bipolar. The 3 programmable
ranges are from ±10 V to ±156 mV full scale. P1 can be used to connect the DaqScan to an
optional analog DBK card.
P2 - DB37 Connector: P2 is a general purpose digital I/O port with 24 (3 x 8-bit) lines on P2,
or digital I/O expansion port controlling up to 272 external lines. Programmable as input or
output. P2 can be used to connect to an optional digital DBK card.
P3 - DB37 Connector: 16-bit digital I/O port with 4 counter inputs and 2 timer outputs. In
addition, DaqScan/2001 and DaqScan/2004 each have a P3 connection to 4 analog outputs(DAC0, DA1, DAC2, and DAC3).

DaqScan/2000 Series

898095Connectors, Indicators, Cables, & Pinouts

4-1

POWER Switch and Power LED: A rocker-type switch with a “0” label for Power Off, and a“1” for Power On. When the unit is on and has power, the POWER LED will light. An AC to DC
power converter resides within each DaqScan/2000 series device.
MAC Address Label: The Media Access Control (MAC) label shows the device serial number inthree formats: (1) Barcode, (2) base 10, and (3) hexadecimal. If prompted to enter a serial
number in software, use the base 10 number. Conversion to a hexadecimal number for use inaddressing will be automatic.
ETHERNET: The 10/100BaseT Ethernet jack can be used to connect to the Ethernet port of
the host PC, or to an Ethernet network. Either of two Ethernet patch cables may be used tomake the connection. CA-242 is a 1.5 foot cable. CA-242-7 is a 7-foot cable.
The Ethernet connector has two built in LEDs (Tx and Rx) that indicate traffic flow. These aredefined in the following LED section. Note that the Ethernet cable length must be <10m inorder for the system to be CE Compliant.
LEDs: There are 5 ETHERNET Status LEDS. Two rectangular LEDs, Tx
and Rx are built right into the frame of the Ethernet jack. The otherthree, located just to the right of the jack, are round and are labeled L, T,
and D.
Tx – “ON” indicates traffic is being transmitted (see figure atright).
Rx – “ON” indicates that the port is receiving traffic.
L (Link) “ON” indicates a link exists. “OFF” indicates no link.
T (Transmit) “ON” indicates 100 M-Bits, “OFF” indicates
10 M-Bits.
D (Duplex) “ON” indicates full duplex, which allows
simultaneous two-way data traffic. “OFF” indicates halfduplex, which only allows one-way data traffic at any
given time.

Tx and Rx LEDs

SYNC (Qty of 2) – Two “synchronization ports” provide a means of synchronizing up to four
DaqScan/2000 Series units in regard to post post-trigger scanning. The ports accept CA-74-1
and CA-74-5 RJ-11, 6 conductor type cables. The total combined length of the SYNC cables is
not to exceed 15 feet (4.57 m).
POWER IN: 90 to 250 VAC, 3-prong socket for power cord.

4-2

Connectors, Indicators, Cables, & Pinouts

898095

DaqScan/2000 Series

CablesShould you need help in regard to cables or have any other product inquiries
please contact the factory.

Cables for use with DaqScan/2000 Series Devices
Description

Part No.

1.

Molded expansion cable. Connects a
DBK module to a DaqScan, 2 in.

CA-255-2T2.

Molded expansion cable. Connects a
DBK module to a DaqScan, 4 in.

CA-255-4T

3.

DBK expansion cable. Connects a
DBK module to a DaqScan, 2.5 in.

CA-37-1T

4.

DBK expansion cable. Connects a
DBK module to a DaqScan, 4.5 in.

CA-37-3T

5.

Expansion-card cable. Connects DBK
series expansion card options. The“x” in the part no. indicates thenumber of cards that can beconnected via the cable.

CA-37-x

6.

Ethernet patch cable, shielded, 18 in.

CA-242

7.

Ethernet patch cable, shielded, 7 ft.

CA-242-7

8.

Sync cable, RJ-11, 6 conductor, 1 ft.

CA-74-1

9.

Sync cable, RJ-11, 6 conductor, 5 ft.

CA-74-5

The total combined length of the SYNC cables is not toexceed 15 feet (4.57 m).

DaqScan/2000 Series

898095Connectors, Indicators, Cables, & Pinouts

4-3

4-4

Connectors, Indicators, Cables, & Pinouts

898095

DaqScan/2000 Series

Pinouts
P1 – Analog I/O (Applies to DaqScan/2001 and DaqScan/2005 only) …… 4-6
P2 – Digital I/O …… 4-7
P3 – Pulse/Frequency/Digital I/O…… 4-8

CAUTION
Turn off power to all devices connected to the system before
connecting cables or setting configuration jumpers and switches.
Electrical shock or damage to equipment can result even under
low-voltage conditions.

CAUTION
The discharge of static electricity can damage some electronic
components. Semiconductor devices are especially susceptible to ESD
damage. You should always handle components carefully, and you
should never touch connector pins or circuit components unless you
are following ESD guidelines in an appropriate ESD controlled area.
Such guidelines include the use of properly grounded mats and wrist
straps, ESD bags and cartons, and related procedures.

Each of the four DaqScan/2000 Series devices has 37-pin connectors designated as P2 and P3.
In addition, DaqScan/2001 and DaqScan/2005 have a third DB37 connector, P1.
Signal assignment to connectors is as follows.

•

P1 is used for Analog I/O (DaqScan/2001 and DaqScan/2005 only)•

P2 is used for Digital I/O•

P3 is used for Pulse/Frequency and Digital I/O; in addition, for DaqScan/2001 and
DaqScan/2004, P3 includes active pins for 4 analog output (DAC) channels.

To connect a DBK option to a DaqScan connect an appropriate cable (see table page 4-2) tothe appropriate DB37 connector [P1, P2, or P3] on the DaqScan and connect the free end of
the cable to the DBK card or module. If you are not familiar with DBK options, or needrefreshed, refer to the DBK Options Manual, p/n 457-0905.

DaqScan/2000 Series

898095Connectors, Indicators, Cables, & Pinouts

4-5

P1 (Applies only to DaqSCan/2001 and DaqScan/2004)

P1 Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37

4-6

Signal Name
+5 Volts-15 Volts
Expansion 7
Expansion 9
Expansion 5
Expansion 6
DGND
Negative Reference
Positive Reference
Not Connected
CH 15 (SE), or CH 7 LO DIFF
CH 14 (SE), or CH 6 LO DIFF
CH 13 (SE), or CH 5 LO DIFF
CH 12 (SE), or CH 4 LO DIFF
CH 11 (SE), or CH 3 LO DIFF
CH 10 (SE), or CH 2 LO DIFF
CH 9 (SE), or CH 1 LO DIFF
CH 8 (SE), or CH 0 LO DIFF
Signal Ground (SGND)
A/I Clock+15 Volts
Expansion 8
Expansion 10
Not Connected
TTL Trigger
Expansion 11
Not Connected
AGND
AGND
CH 7 (SE), or CH 7 HI DIFF
CH 6 (SE), or CH 6 HI DIFF
CH 5 (SE), or CH 5 HI DIFF
CH 4 (SE), or CH 4 HI DIFF
CH 3 (SE), or CH 3 HI DIFF
CH 2 (SE), or CH 2 HI DIFF
CH 1 (SE), or CH 1 HI DIFF
CH 0 (SE), or CH 0 HI DIFF

Connectors, Indicators, Cables, & Pinouts

Analog I/O

Description
Expansion +5 V power
Expansion -15 VDC power
Digital OUT, external ADDRESS select bit 3
Digital OUT, external ADDRESS select bit 1
Digital OUT, external GAIN select bit 1
Digital OUT, external GAIN select bit 0
Digital Ground
Analog, -5 V reference
Analog +5V reference
N/A
Ch 15 HI IN (single-ended mode) / Ch 7 LO IN (differential mode)
Ch 14 HI IN (single-ended mode) / Ch 6 LO IN (differential mode)
Ch 13 HI IN (single-ended mode) / Ch 5 LO IN (differential mode)
Ch 12 HI IN (single-ended mode) / Ch 4 LO IN (differential mode)
Ch 11 HI IN (single-ended mode) / Ch 3 LO IN (differential mode)
Ch 10 HI IN (single-ended mode) / Ch 2 LO IN (differential mode)
Ch 9 HI IN (single-ended mode) / Ch 1 LO IN (differential mode)
Ch 8 HI IN (single-ended mode) / Ch 0 LO IN (differential mode)
Sense Common (SGND)
External ADC Pacer Clock Input / Internal ADC Pacer Clock Output
Expansion +15 V power
Digital OUT, external ADDRESS select bit 2
Digital OUT, external ADDRESS select bit 0
N/A
Digital IN, External TTL Trigger Input
Digital OUT, Simultaneous sample and hold (SSH)
N/A
Analog Ground
Analog GroundCh 7 IN (single-ended mode) / Ch 7 HI IN (differential mode)
Ch 6 IN (single-ended mode) / Ch 6 HI IN (differential mode)
Ch 5 IN (single-ended mode) / Ch 5 HI IN (differential mode)
Ch 4 IN (single-ended mode) / Ch 4 HI IN (differential mode)
Ch 3 IN (single-ended mode) / Ch 3 HI IN (differential mode)
Ch 2 IN (single-ended mode) / Ch 2 HI IN (differential mode)
Ch 1 IN (single-ended mode) / Ch 1 HI IN (differential mode)
Ch 0 IN (single-ended mode) / Ch 0 HI IN (differential mode)

898095

DaqScan/2000 Series

P2P2 Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37

Digital I/O

Signal Name
Not Connected
Not Connected
Port B - B7
Port B - B6
Port B - B5
Port B - B4
Port B - B3
Port B - B2
Port B - B1
Port B - B0
DGND
Not Connected
DGND
Not Connected
DGND
Not Connected
DGND+ 5 Volt Supply
DGND+ 5 Volt Supply
DGND
Port C - C7
Port C - C6
Port C - C5
Port C - C4
Port C - C3
Port C - C2
Port C - C1
Port C - C0
Port A - A7
Port A - A6
Port A - A5
Port A - A4
Port A - A3
Port A - A2
Port A - A1
Port A - A0

DaqScan/2000 Series

Description
N/A
N/A
Digital I/O: P2 Digital Port B, Bit 7; or, P2 Expansion Address Bit 0 Out
Digital I/O: P2 Digital Port B, Bit 6; or, P2 Expansion Address Bit 1 Out
Digital I/O: P2 Digital Port B, Bit 5; or, P2 Expansion Address Bit 2 Out
Digital I/O: P2 Digital Port B, Bit 4; or, P2 Expansion Address Bit 3 Out
Digital I/O: P2 Digital Port B, Bit 3; or, P2 Expansion Address Bit 4 Out
Digital I/O: P2 Digital Port B, Bit 2; or, P2 Expansion RESET Output
Digital I/O: P2 Digital Port B, Bit 1; or, P2 Expansion WRITE Output
Digital I/O: P2 Digital Port B, Bit 0; or, P2 Expansion READ Output
Digital Ground
N/A
Digital Ground
N/A
Digital Ground
N/A
Digital Ground
Expansion +5 Volt Power
Digital Ground
Expansion +5 Volt Power
Digital Ground
Digital I/O: P2 Digital Port C, Bit 7; or, P2 Expansion Data Bit 7
Digital I/O: P2 Digital Port C, Bit 6; or, P2 Expansion Data Bit 6
Digital I/O: P2 Digital Port C, Bit 5; or, P2 Expansion Data Bit 5
Digital I/O: P2 Digital Port C, Bit 4; or, P2 Expansion Data Bit 4
Digital I/O: P2 Digital Port C, Bit 3; or, P2 Expansion Data Bit 3
Digital I/O: P2 Digital Port C, Bit 2; or, P2 Expansion Data Bit 2
Digital I/O: P2 Digital Port C, Bit 1; or, P2 Expansion Data Bit 1
Digital I/O: P2 Digital Port C, Bit 0; or, P2 Expansion Data Bit 0
Digital I/O: P2 Digital Port A, Bit 7
Digital I/O: P2 Digital Port A, Bit 6
Digital I/O: P2 Digital Port A, Bit 5
Digital I/O: P2 Digital Port A, Bit 4
Digital I/O: P2 Digital Port A, Bit 3
Digital I/O: P2 Digital Port A, Bit 2
Digital I/O: P2 Digital Port A, Bit 1
Digital I/O: P2 Digital Port A, Bit 0898095Connectors, Indicators, Cables, & Pinouts

4-7

P3

Pulse/Frequency/Digital I/O

P3 Pin
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35
36
37

4-8

Signal Name
DGND
DGND
Digital 7
Digital 6
Digital 5
Digital 4
Digital 3
Digital 2
Digital 1
Digital 0
DGND
Expansion 2
Expansion 3
Expansion 4
Timer 0Timer 1Counter 2
Counter 0+ 15 Volts+ 5 Volt Supply
Analog Out Clock
Digital 15
Digital 14
Digital 13
Digital 12
Digital 11
Digital 10
Digital 9
Digital 8
AGND
Analog Out DAC 3
Analog Out DAC 2
Analog Out DAC 1
Analog Out DAC 0
Counter 3
Counter 1- 15 Volts

Description
Digital Ground
Digital GroundP3 Digital Port Bit 7
P3 Digital Port Bit 6
P3 Digital Port Bit 5
P3 Digital Port Bit 4
P3 Digital Port Bit 3
P3 Digital Port Bit 2
P3 Digital Port Bit 1
P3 Digital Port Bit 0
Digital GroundReserved for XC/DReserved for XWRReserved for XRDP3 Timer 0 Output
P3 Timer 1 Output
P3 Counter 2 Input
P3 Counter 0 Input
Expansion, +15 VDC
Expansion, +5 Volt Power
External DAC Pacer Clock Input/Internal DAC Pacer Clock Output
Digital I/O; P3 Digital Port Bit 15
Digital I/O; P3 Digital Port Bit 14
P3 Digital Port Bit 13
P3 Digital Port Bit 12
P3 Digital Port Bit 11
P3 Digital Port Bit 10
P3 Digital Port Bit 9
P3 Digital Port Bit 8
Analog Ground
Analog DAC 3 Output
Note: DAC0, DAC1, DAC2, and DAC3 apply to DaqScan/2001 andAnalog DAC 2 Output
DaqScan/2004. They do not apply to DaqScan/2002 or DaqScan/2005.
Analog DAC 1 Output
Analog DAC 0 Output
P3 Counter 3 Input
P3 Counter 1 Input
Expansion, - 15 VDC Power

Connectors, Indicators, Cables, & Pinouts

898095

DaqScan/2000 Series

Software

5

Overview …… 5-1
Out-of-the-BoxTM Software …… 5-1
Drivers for Third-party, Icon-driven Software …… 5-2
Language Drivers …… 5-2

Overview
The Daq devices have software options capable of handling most applications. Three typesof software are available:•

Out-of the-box graphical programs, e.g., DaqView, DaqViewXL, and post acquisition
data analysis programs such as eZ-PostView, eZ-TimeView, and eZ-FrequencyView.

•

drivers for third-party, icon-driven software such as DASYLab and LabVIEW.

•

various language drivers to aid custom programming using API; includes
DaqCOMTM ActiveX/COM support.

Out-of-theBoxTM Software
Out-of-the-Box programs are convenient for fill-in-the-blank applications that do not require
programming for basic data acquisition and display:

DaqScan/2000 Series•

DaqView is a Windows-based program for basic set-up and data acquisition.
DaqView lets you select desired channels, gains, transducer types (including
thermocouples), and a host of other parameters with a click of a PC’s mouse.
DaqView lets you stream data to disk and display data in numerical or graphicalformats.

•

DaqViewXL allows you to interface directly with Microsoft Excel to enhance data
handling and display. Within Excel you have a full-featured Daq control panel and all
the data display capabilities of Excel.

•

eZ-PostView provides a simple method of graphically viewing acquired waveforms
from the acquisition device. Up to 8 windows can be displayed on one screen, with up
to 16 channels overlaid on each window. eZ-PostView makes it simple to visuallyinspect acquired waveforms from multiple channels within seconds of acquiring thedata. Documentation, in Adobe PDF format, is provided on the CD. Refer to the
PostAcquisition Analysis.PDF.

•

eZ-TimeView & eZ-FrequencyView
eZ-TimeView and eZ-FrequencyView are optional post-acquisition analysis packages,
which are related to eZ-PostView, but include more features. eZ-TimeView istargeted at time-domain analysis, including min/max, peak-peak, mean, RMS, plus awide variety of plotting and waveform viewing capabilities. eZ-FrequencyView istargeted at post-acquisition frequency-domain analysis, including FFT’s, octave analysis, plus dozens of other analysis features. Documentation, in Adobe PDF
format, is provided on the CD. Refer to the PostAcquisition Analysis.PDF.

•

The Daq Configuration control panel allows for interface configuration, testing,and troubleshooting.

948895

Software5-1

Reference Notes:
¾

¾

PDF
Note:

The software documentation for DaqView, DaqViewXL, and Post
Acquisition Data Analysis are not included as part of the hardcopy
manual, but are available in PDF format. See the PDF Note, below.
Programming topics are covered in the Programmer’s User Manual(1008-0901). As a part of product support, that manual is automaticallyloaded onto your hard drive during software installation. The default
location is the Programs directory, which can be accessed through theWindows Desktop.

During software installation, Adobe® PDF versions of user manuals willautomatically install onto your hard drive as a part of product support. The
default location is in the Programs group, which can be accessed from theWindows Desktop. Refer to the PDF documentation for details regarding both
hardware and software.
Note that the PDF documents can be read directly from the CD by using the<View PDFs> button located on the opening install screen.
A copy of the Adobe Acrobat Reader® is included on your CD. The Reader
provides a means of reading and printing the PDF documents. Note that
hardcopy versions of the manuals can be ordered from the factory.

Drivers for Third-party, Icon-driven Software
LabVIEW® Support
We provide LabVIEW® support for expansion and signal conditioning modules. Our dataacquisition VIs for LabVIEW are more than just simple hardware access VIs, they are completewith engineering data conversion, data display and logging capabilities.
DASYLab®
If your application requirements go beyond the scope of DaqView, you may want to consider
DASYLab software, as the latter offers a greater degree of flexibility and customizing.

Language Drivers
Each Daq system comes with an Application Programming Interface (API). API-language
drivers include: C/C++, and Visual Basic. The latest software is a 32-bit version API.
DaqScan/2000 Series devices are supplied with DaqX Subroutine API Libraries. These provide
complete support for all of the functionality available on each data acquisition device in Visual
Basic®, and C/C++. In addition, DaqX is supported under all versions of Windows®. Over100 program examples are included with DaqX Subroutine API Libraries. API documentation
is provided in a Programmer’s Manual (p/n 1008-0901), which is included on the installationCD.
The DaqCOM™ suite of programming allows applications developers to rapidly develop anddeploy custom systems by leveraging COM (Component Object Model) technology. DaqCOM
does this by providing a powerful easy-to-use interface to most programming languages
including, Visual Basic®, VBA, C++, and J++. In addition, DaqCOM supports the new
Windows.NET architecture and includes examples for VisualBasic.NET and C++. Support for
VisualStudio.NET is accomplished via the COMInterop feature within VisualStudio.NET.

5-2

Software948895

DaqScan/2000 SeriesConfiguring the System for Operation

6

Step 1 - Install the Software ….. 6-2
Step 2 - Determine the type of Network Connection …… 6-2
Step 3 - Connect System Components …… 6-4
Step 4 - Power-up the System …… 6-5
Step 5 - Configure the Computer’s Network Settings …… 6-6
Step 6 - Configure Device Network Settings using DaqIPConfig …… 6-9
Step 7 - Configure and Test the System using the Daq Configuration …… 6-10

Connecting a DaqLab/2000 Series or DaqScan/2000 Series Device to the Ethernet
The DaqLab/2000 Series and DaqScan/2000 Series acquisition devices connect directly to an
Ethernet port on a PC or network hub via a built-in 10/100BaseT Ethernet interface. An Ethernet
patch cable CA-242 (1.5 foot) or CA-242-7 (7 foot) cable is used to make the connection.

CAUTION
Turn off power to the system devices and externally connected equipment before
connecting cables. Electric shock or damage to equipment can result even under lowvoltage conditions.
Take ESD precautions (packaging, proper handling, grounded wrist strap, etc.)

Reference Note: Adobe PDF versionsof user manuals will automatically install
onto your hard drive as a part ofproduct support. The default location isin the Programs group, which can beaccessed from the Windows Desktop.
You can also access documents directlyfrom the data acquisition CD via the<View PDFs> button located on the
CD’s opening screen.

Before you get started
Verify that you have the following items.
DaqLab/2000 or DaqScan/2000 Series Device
DBK option cards, as applicable
AC Power Cord
Ethernet Patch Cable
Ethernet jack [on PC or on a hub connected to the Ethernet]
Data Acquisition CD
Monitor: SVGA, 1024 x 768 screen resolution
Windows 2000 SP4 and Windows XP users:
PC with Intel™ Pentium, 1 GHz or equivalent;
512 MB memory; 10 GB disk space
Windows Vista users:
PC must be Windows Vista Premium Ready

DaqLab/2000 Series & DaqScan/2000 Series

947091

Configuring the System for Operation

6-1

STEP 1 –Install the Software
1. Remove previous version Daq drivers, if present. You can do this through Microsoft’s
Add/Remove Programs feature.
2. Place the Data Acquisition CD into the CD-ROM drive. Wait for PC to auto-run the CD. This
may take a few moments, depending on your PC. If the CD does not auto-run, use the
Desktop’s Start/Run/Browse feature to locate and run Setup.exe [from the applicable
CD-ROM drive].

3. After the intro-screen appears, follow the screen prompts.

STEP 2 – Determine the type of Network Connection
To properly connect and configure the device you must determine the type of network it will
become part of. This is because the type of network used has a direct bearing on the IP
address of the device.
Briefly, the four network types are as follows:
Dedicated Network - with a direct cable connection from the PC to thedeviceDedicated Network - making use of a network hub or switch
LAN with a DHCP server(Local Area Network with a Dynamic Host Configuration Protocol)
LAN without a DHCP server(Local Area Network with no Dynamic Host Configuration Protocol)
Brief descriptions and illustrations follow.

Dedicated Network - with a direct cable connection from the PC to the device
In this scenario a DaqLab/2000 Series device is connected directly to an Ethernet
jack on a host computer. The following figure could just as easily portray a
DaqScan/2000 Series device.

Dedicated Network using a Direct Cable Connection

6-2

Configuring the System for Operation

947091

DaqLab/2000 Series & DaqScan/2000 Series

Dedicated Network - making use of a network hub or switch
In the following figure a DaqLab/2000 Series device is connected to the Ethernet through a
network hub/switch. At least one computer is also connected to the hub. DaqScan/2000
series devices can be connected directly to the hub/switch in the same manner.

Dedicated Network using a Hub/Switch
Some network devices such as a wireless access point may act as a DHCP
server. If this is the case, follow the instructions for the LAN with a DHCP
server. For detailed information consult the documentation that is specificto your network device.

LAN with a DHCP Server(Local Area Network with a Dynamic Host Configuration Protocol server)

Many corporations use the LAN/Server with DHCP arrangement for their networks. In
this type of setup several computers are typically connected to a network that makes useof a DHCP server. In addition, the primary data acquisition device, e.g., DaqLab/2001, isconnected to the network hub/switch.

LAN with a DHCP Server

DaqLab/2000 Series & DaqScan/2000 Series

947091

Configuring the System for Operation

6-3

Notes: Using a DaqLab/2000 Series or DaqScan/2000 Series device on a typical LAN may
affect the speed of the network and internet data transfer. Because of this werecommend adding a network card to the computer and using one of the two
dedicated network configurations.
Contact your network administrator before connecting your device to a corporate
network.

LAN with no DHCP Server(Local Area Network with no Dynamic Host Configuration Protocol server)
This scenario looks the same as that shown in the previous illustration, except there is no
Dynamic Host Configuration Protocol (DHCP). In this type of setup, one or more computers
are connected to a network; and each computer has a static IP address.

STEP 3 – Connect the System Components
Reference Note:
For information regarding connections, refer to Chapter 4, Connectors, Indicators, Cables,& Pinouts.

What you will need to connect your DaqLab/2000 Series or DaqScan/2000 Series
device to the Ethernet:

An available connection to the Ethernet. The connection can be either- an Ethernet jack on a computer or- an Ethernet jack on a hub that is connected to the Ethernet.
An Ethernet patch cable, e.g., a CA-242 (1.5 foot cable) or a
CA-242-7 (7-foot cable).

1. Connect the Ethernet cable to the Ethernet jack on the Daq device.
2. Connect the other end of the Ethernet cable to the Ethernet jack on the hostcomputer or network hub.

6-4

Configuring the System for Operation

947091

DaqLab/2000 Series & DaqScan/2000 Series

STEP 4 – Power-up the System
What you will need:
A 90 to 240 VAC power supply.
Your unit’s power supply cord.

How to make the connection:
In the following steps we use the term “Daq Device” to mean a DaqLab/2000 Series device or
DaqScan/2000 Series device as applicable to your system.
1. Using the unit’s power switch, turn the Daq device “OFF.” The switch will be in the“0” position and the power LED will be unlit.
2. Connect the power cord to the Power In connector on the Daq device.
3. Connect the plug to a standard AC outlet, or to the outlet of another reliable 90 to
240 VAC power supply.
4. If your power supply has a switch or breaker, close it to turn power “ON.”5. Turn ON the Daq device by placing its power switch to the “1” position. The unit’s
power led will light up.

DaqLab/2000 Series & DaqScan/2000 Series

947091

Configuring the System for Operation

6-5

STEP 5 – Configure the Computer’s Network Settings [Applies to “dedicated networks” only]
If using a LAN (Local Area Network), which has a DHCP server, skip this section
and continue with STEP 7 - Configure and Test the System using the
Daq*Configuration Applet (page 6-10).
If using a LAN (Local Area Network), which has no DHCP server, skip thissection and continue with STEP 6 - Configure Device Network Settings using
DaqIPConfig (page 6-9).
1. Open the Control Panel by navigating from the Windows Desktop as follows:
Start Menu
Settings
Control Panel.
2. Double-click the “Network and Dial-up Connections” icon.
3. Double-click the “Network Connection” icon for the network that the Daq device isconnected to.

Local Area Connection Status
4.

6-6

In the “Local Area Connection Status” box (previous figure), click on the <Properties>
button. The “Local Area Connection Properties” box will appear (following figure).

Configuring the System for Operation

947091

DaqLab/2000 Series & DaqScan/2000 Series

Local Area Connection Properties

5. Double-click the “Internet Protocol (TCP/IP)” component (previous figure).
The “Internet Protocol (TCP/IP) Properties” box will appear (following figure).

Internet Protocol (TCP/IP) Properties

DaqLab/2000 Series & DaqScan/2000 Series

947091

Configuring the System for Operation

6-7

Configure the Computer’s TCP/IP settings as follows.

Make sure that each computer on the dedicated network has a unique
IP address. Make sure that the IP address of the acquisition device is also
unique. The computer and the device cannot use the same IP address.

Internet Protocol (TCP/IP) Properties

6. Select the “Use the following IP Address” radio button.
7. Set the IP address field to 10.0.0.x where x is some number from 1 to 255.
Make sure that each computer on the dedicated network has a unique
IP address!

8. Set the Subnet mask to 255.0.0.0. Note that the remaining fields can be left
unchanged.

6-8

Configuring the System for Operation

947091

DaqLab/2000 Series & DaqScan/2000 Series

STEP 6 – Configure Device Network Settings using DaqIPConfig
Applies only to a LAN (Local Area Network), which has a no DHCP server.

Never set the TCP/IP for the device to be the same as the host computer’s IP address!
Each computer and each device on a dedicated network must have a unique IP address!
If using a LAN (Local Area Network), which has a DHCP server, skip this section and continue
with STEP 7 - Configure and Test the System using the Daq*Configuration Applet (page 6-10).
The DaqIPConfig applet is designed for 32-bit Windows/2000/XP/Vista systems. You can use DaqIPConfig tochange the IP address of the device to be compatible with networks that require fixed IP addresses. As aprecaution, you should always consult with your IT administrator before using the applet to ensure that eachdevice and host computer connected to the network maintains a unique IP address. The applet is located in the
program group for the associated device and can be accessed from the Windows Desktop via the start menu.
1. Locate the DaqIPConfig Applet.
Locate the DaqIPConfig applet by navigating fromthe Windows’ Desktop as follows:
Programs
IOtech DaqX
Start Menu
Software
DaqIPConfig
2. Open the DaqIPConfig Applet.
Click on the DaqIPConfig selection to open theapplet.
3. Select the device that is to have the addresschange.
Note: In the above figure there is only one device
to select, i.e., DaqLab/2001.
4. Set the internet protocol (TCP/IP) settings to
be compatible with host computer.

(a) Select the radio button labeled “Use thefollowing IP address.”(b) Enter the new internet protocol settings. Ifneeded, consult your network administrator
for acceptable numbers.

Do not set the TCP/IP to the computer’s IP address!
(c) Click the <OK> button.
5. Reboot the device.
The new IP address will not take affect until the device has been powered-off, then poweredback on.
6.

Repeat steps 3, 4, and 5 for other devices in the system.
After configuring the network settings for all devices, proceed to Step 7.

DaqLab/2000 Series & DaqScan/2000 Series

947091

Configuring the System for Operation

6-9

STEP 7 – Configure and test the System using the Daq* Configuration Applet
The Daq* Configuration applet is located in the Windows Control Panel. It allows you to add
or remove a device and change configuration settings. The included test utility provides
feedback on the validity of current configuration settings and performance summaries.
1. Open the Daq* Configuration Applet.
a.

Open the Control Panel by navigating from the Windows’ Desktop as follows:
Start Menu
Settings
Control Panelb.

From the Control Panel, double-click the Daq* Configuration icon.

2. Add the first-level device to the list of installed devices.
The first-level device is the device that will be connected directly to the Ethernet, via a hostcomputer’s Ethernet jack or a jack on a network hub. DaqLab/2001 is an example of a firstlevel device.
a.

Select the Computer image in the Device Inventory configuration tree (following
figure).

b.

Click the <Add Device> button. The “Select Device Type” box will appear.

c.

Select the device from the list of devices. In our example we are selecting a
DaqLab/2001.

d.

Click the <OK> button. The “Properties” box will appear for the selected device.

Using Daq Configuration’s “Device Inventory” and”Select Device Type” to Add a Device

Note: Although the above figure indicates that a DaqLab/2001 was selected as thefirst level device, we could similarly have selected another device, forexample, a DaqLab/2005 or a DaqScan/2004.
3. Set the properties of the first-level device.
In this step you will set the device properties according to one of the following two methods,
depending on whether you have a “Dedicated Network” or a “LAN with DHCP Server Network.”
Illustrations of the network types are provided on page 6-3.

6-10

Configuring the System for Operation

947091

DaqLab/2000 Series & DaqScan/2000 Series

Users of Dedicated Networks follow these 2 steps.
a.

Enter the Serial Number of the first-level device, e.g., DaqLab/2001. In thefollowing screen shots the Serial Number is 800069.

b.

Select the “Auto IP Setting” radio button. Note that the IP Address of the
DaqLab/2001 will automatically be calculated and displayed in the IP Address field
as indicated in the following left-hand figure.

c.

Click the <OK> button.

For LAN with DHCP Server Networks

For DEDICATED Networks

Daq Configuration, Properties Dialog Boxes

Note: The above images are based on DaqLab/2001.

We could similarly have selected another device.

Users of LAN with DHCP Server Networks follow these 3 steps.
The DaqIPConfig applet provides the Serial Number and the IP Address of the device.
Users of LAN with DHCP Server Networks will need to enter both numbers in the Daq
Configuration, Properties dialog boxes (previous right-hand figure). Page 6-9 includes
instructions for accessing DaqIPConfig.
If needed, refer to the upper right-hand figure in regard to radio-button and data entry
locations.
a.

Enter the base 10 version of the Serial Number of the device, e.g., DaqLab/2001.

b.

Select the “Manual IP Setting” radio button.

c.

In the IP Address field, enter the IP address.

d.

Click the <OK> button.

 Provide your network administrator with the information located on
the device’s MAC label.

 Find out from the administrator if the IP Address will be changing. If

so, see if you can obtain a permanent IP Address that will be dedicatedspecifically to your device.

4. Test the system connections.
a. Make sure the device has been properly installed and is powered-on.
b.

Make sure all cables are properly and securely connected.

c.

Click the “Test Hardware” tab.

d.

Click the <TCP/IP Test> button. This tests the Transmission Control Protocol /Internet Protocol. The TCP/IP test has two components: Pinging Device and TCPConnection.
The Pinging Device test is an ICMP (Internet Control Message Protocol) ping test.
In addition to indicating either “Passed” or “Failed,” the test displays the round-trip

DaqLab/2000 Series & DaqScan/2000 Series

947091

Configuring the System for Operation

6-11

time of the ping, for example, <10 ms; and Packet Loss expressed as a percent ofdata lost.
A long ping round trip time, for example, >50 ms and/or any packet loss indicatesa slow network that is not optimized.
If the Pinging Device Test fails a <Details> button will appear just above the<Resource Test> button. If this is the case, click the <Details> button to
obtain possible causes and possible solutions.
The result of the TCP (Transmission Control Protocol) test will be either
passed or failed. If the TCP test failed a <Details> button will appear just abovethe <Resource Test> button. If this is the case, click the <Details> button to
obtain possible causes and possible solutions.
e.

Upon completion of the TCP/IP test, click the <Resource Test> button. The
Resource Test consists of two components: Resource Tests and Performance
Tests.
The Resource Test performs a quick communications check. If the host PC cancommunicate with the device a message that the test passed is displayed.
The Performance Tests obtain quantitative results using the current
configuration of the acquisition device. The test results represent the maximum
rates at which various operations can be performed, and will vary according toport hardware capabilities.
The ADC FIFO Input Speed part of the test results in a display of the maximum
rate at which data can be transferred from the tested device’s internal ADC FIFO
to computer memory. Results are given in samples/second, where a sample(2 bytes in length) represents a single A/D value.
When testing a DaqLab/2000 Series or a DaqScan/2000 Series device, if
the unit does not respond within 30 seconds perform the following steps:
1) reboot the system
2) upon power-up, re-open the Daq Configuration applet
3) select another configuration setting4) reinitiate the test

This completes the procedure for connecting a DaqLab/2000 Series or a DaqScan/2000 Series
device to the Ethernet.
Reference Notes:
Depending on your application and system setup, related information can be foundin the following sources:

The DBK Options Manual (p/n 457-0905).
The Programmer’s Manual, (p/n 1008-0901).

Note: You can access PDF documents directly from the opening screen of the data acquisition
CD via the <View PDFs> button.

6-12

Configuring the System for Operation

947091

DaqLab/2000 Series & DaqScan/2000 Series

CE Compliance & Noise Considerations

7

Overview ……7-1
CE Standards and Directives …… 7-1
Safety Conditions ……7-2
Emissions/Immunity Conditions ……7-2
CE Enhancements for DBKs .…… 7-3
CE Compliance for System Expansion …… 7-3
Noise Considerations …… 7-4

Overview
CE-compliant products bear the “CE” mark and include a Declaration of Conformity
stating the particular specifications and conditions that apply. The test records andsupporting documentation that validate the compliance are kept on file at thefactory.
The European Union established CE standards in 1985. The standards include specifications
for safety, EMI emissions, and immunity from electromagnetic interference. Products that are
intended for placement in the European Union must meet or exceed the standards and bearthe "CE" mark.
Although not required in the USA, meeting or exceeding the CE standards is considered goodengineering practice, since doing so enhances safety while reducing noise and ESD problems.
In contracted and in-house testing, most Daq products met the required specifications. In
many cases, products that were not originally in compliance were redesigned accordingly. In
noted instances, alternate product versions, shield plates, edge guards, special connectors, or
add-on kits are required to meet CE compliance.

CE Standards and Directives
The electromagnetic compatibility (EMC) directives specify two basic requirements:
1.
2.

The device must not interfere with radio or telecommunications.
The device must be immune from electromagnetic interference from RF transmitters,
etc.

The standards are published in the Official Journal of European Union under direction of
CENELEC (European Committee for Electrotechnical Standardization). The specific standards
relevant to Daq equipment are listed on the product’s Declaration of Conformity.
The safety standard that applies to Daq products is EN 61010-1:1993 (Safety Requirements
for Electrical Equipment for Measurement, Control, and Laboratory Use, Part 1: General
Requirements). Environmental conditions include the following:

•
•
•
•

indoor use•••

mains supply voltage fluctuations not to exceed ±10% of the nominal voltage•

pollution degree I or II in accordance with IEC 664

altitude up to 2000 m
temperature 5°C to 40°C (41°F to 104°F)
maximum relative humidity 80% for temperatures up to 31°C (87.8°F) decreasing
linearly to 50% relative humidity at 40°C (104°F)
other supply voltage fluctuations as stated by the manufacturertransient overvoltage according to installation categories (overvoltage categories) I, II
and III. For mains supply, the minimum and normal category is II.

DaqLab/2000 Series and DaqScan/2000 Series

959195

CE-Compliance & Noise Considerations

7-1

For clarification, terms used in some Declarations of Conformity include:•

pollution degree: any addition of foreign matter, solid, liquid or gaseous (ionizedgases) that may produce a reduction of dielectric strength or surface resistivity.
Pollution Degree I has no influence on safety and implies: the equipment is at
operating temperature with non-condensing humidity conditions; no conductive particles
are permitted in the atmosphere; warm-up time is sufficient to avert any condensation
or frost; no hazardous voltages are applied until completion of the warm-up period.
Pollution Degree II implies the expectation of occasional condensation.

•

overvoltage (installation) category: classification with limits for transient
overvoltage, dependent on the nominal line voltage to earth. Category I implies signals
without high transient values. Category II applies to typical mains power lines withsome transients.

Safety Conditions
Users must comply with all relevant safety conditions in the user’s manual and the
Declarations of Conformity. This manual and Daq hardware make use of the following
Warning and Caution symbols: If you see either of these symbols on a product, carefully readthe related information and be alert to the possibility of personal injury.
This warning symbol is used in this manual or on the equipment to warnof possible injury or death from electrical shock under noted conditions.
This warning/caution symbol is used to warn of possible personal injury
or equipment damage under noted conditions.
Daq products contain no user-serviceable parts; refer all service to qualified
personnel. The specific safety conditions for CE compliance vary by product; but
general safety conditions include:•

The operator must observe all safety cautions and operating conditions specified in the
documentation for all hardware used.

••

The host computer and all connected equipment must be CE compliant.

•

Isolation voltage ratings: do not exceed documented voltage limits for power and signal
inputs. All wire insulation and terminal blocks in the system must be rated for theisolation voltage in use. Voltages above 30 Vrms or ±60 VDC must not be applied if any
condensation has formed on the device.

•

Current and power use must not exceed specifications. Do not defeat fuses or otherover-current protection.

All power must be off to the device and externally connected equipment before internal
access to the device is permitted.

Emissions/Immunity Conditions
The specific immunity conditions for CE compliance vary by product; but general immunity
conditions include:

7-2•

Cables must be shielded, braid-type with metal-shelled connectors. Input terminal
connections are to be made with shielded wire. The shield should be connected to thechassis ground with the hardware provided.

••

The host computer must be properly grounded.
In low-level analog applications, some inaccuracy is to be expected when I/O leads are
exposed to RF fields or transients over 3 or 10 V/m as noted on the Declaration ofConformity.

CE-Compliance & Noise Considerations

959195

DaqLab/2000 Series and DaqScan/2000 Series

CE Enhancements for DBKs
The following CE enhancements are detailed in the DBK Cards and Modules User’s Manual(p/n 457-0905).

•••

DBK41/CE
Edge Guard (for DBK5, DBK8, and DBK44)
Applicable cables and connectors

CE Compliance for System Expansion
The Daq devices are CE compliant at the time they leave the factory, and will remain incompliance providing that the conditions stated on their respective Declaration of Conformity
cards continue to be met.
You may want to expand your data acquisition system while having it retain its “CE Compliant”
status. If so, refer to the most recent Declaration of Conformity for each device, prior to
making any efforts to expand the system.

Connecting a non-CE Compliant DBK module or card to yoursystem will make the system non-compliant.

In order to remain CE Compliant, DaqLab/2000 Series and DaqScan/2000 Series devices
require the use of ferrite inductive collars and a ground line, which are included in an Ethernet
CE Kit. Installation instructions for the kit follow this chapter.
General rules of thumb regarding expansion, while remaining CE Compliant, are:

•
•
•
•

Review the most recent Declaration of Conformity for each device.
Use short cables.
When possible, use shielded cables.
When assembling or disassembling components, take ESD precautions, including
the use of grounded wrist straps.

• Ensure that the host computer is CE Compliant.
• Ensure that expansion devices are CE Compliant• Ensure all system components are properly grounded.

DaqLab/2000 Series and DaqScan/2000 Series

959195

CE-Compliance & Noise Considerations

7-3

Noise Considerations
Controlling electrical noise is imperative because it can present problems even with the best
measurement equipment. Most laboratory and industrial environments suffer from multiple
sources of electrical noise. For example, AC power lines, heavy equipment (particularly ifturned on and off frequently), local radio stations, and electronic equipment can create noise
in a multitude of frequency ranges.
Local radio stations are a source of high frequency noise, while computers and other electronicequipment can create noise in all frequency ranges. Creating a completely noise-freeenvironment for test and measurement is seldom practical. Fortunately, simple techniquessuch as using shielded/twisted pair wires, filtering, and differential voltage measurement areavailable for controlling the noise in our measurements. Some techniques prevent noise fromentering the system; other techniques remove noise from the signal.
While many techniques for controlling noise in signals provide a means of removing the noisethat is already present, the preferred solution is to prevent the occurrence of noise in thesignal in the first place.
The following practices, some of which are required for CE compliance, should be employed tominimize noise.
•

Make a solid earth ground connection. Required for CE Compliance. Ensure that thechassis of the primary data acquisition device, e.g., DaqLab/2001 is connected to earth
ground. This practice: (a) keeps radiated emissions low by keeping the chassis
electrically quiet, (b) keeps potential common-mode voltages low, (c) improves user
safety, and (d) provides a safe path for Electrostatic Discharge energy back to earth
ground. Use a shielded Ethernet cable, e.g., CA-242, to connect the DaqLab or DaqScan
chassis to the host computer’s chassis.

•

Use short Ethernet cables.
shorter the cable the better.

•

Use shielded cables. Loose wires are effective antennae for radio frequency pickup andcan form loops for inductive pickup. The use of properly connected shields will greatly
reduce such noise.

•

Minimize ambient EMI. The lower the ambient EMI, the better. Sources ofelectromagnetic interference include solenoids, motors, computer equipment, high power
distribution wiring, etc.

•

Distance cables. Power supply switch transients can vary in strength and frequency.
Ethernet cables can radiate digital switching noise. For these reasons route the power
supply and Ethernet cables such that they are as far as possible from all analog lines. The
analog lines include those that connect to P1 on the front panel, as well as those thatconnect to the rear panel via an expansion port, or through the P4 connector.

•

Use ferrite inductive collars. A clamp-on ferrite collar can be secured at each end of
the Ethernet cable and at both ends of the power supply cable. The collars on the
Ethernet cable will reduce digital switching noise. Note that the collars will not reduce theintegrity of the Ethernet channel if the system is making use of shielded Ethernet cables,
part number CA-242. Ferrite collars on the power cable will reduce the effects of powertransients.

The use of short Ethernet cables will reduce noise. The

This partial view of the host PC shows the
Ethernet cable with a ferrite collar. The
PC’s connection to the ground-line is also
visible.

7-4

CE-Compliance & Noise Considerations

959195This partial view of the data acquisition deviceshows its connection to the ground-line. It also
shows two ferrite collars, the foremost of which isclamped to the power cable. The backgroundcollar is clamped onto the Ethernet cable.
DaqLab/2000 Series and DaqScan/2000 Series

Taking Measurements

8

Introduction …… 8-1
Scanning …… 8-1
Triggering …… 8-2
Synchronous I/O Operations…… 8-3
Asynchronous I/O Operations …… 8-5
Synchronizing Multiple Units …… 8-6
Using DBK Cards and Modules for Signal Conditioning ….. 8-8

Introduction
With signal lines connected and a system up and running you can begin acquiring data.
Specific connector details and pinouts are provided in chapter 4. You should refer to thesoftware documentation applicable to your data acquisition system, for example, DaqView.

Scanning
DaqScan/2000 Series devices have an on-board scan sequencer that permits the user to selectany combination of up to 16384 channel/range combinations. The sequencer scans allchannels contained in the sequence at the rate of 5 µs/channel, 10 µs/channel, or
1000 µs/channel, depending on the speed of the DBK expansion options; thereby minimizing
the time-skew from channel-to-channel. The user can also set the time between scan groups,
from 0 to 6 hours. In addition to scanning analog inputs, the sequencer can scan digital inputs
and counter inputs.
Digital and frequency inputs can be scanned using the same scan sequencer employed foranalog inputs, enabling the time correlation of acquired digital data to acquired analog data.
(See Note)

5µs, 10µs, or 1000µs(See Note)

Scanning Example
Note:

DaqScan/2000 Series

All inputs can be measured at the 5 µs/channel rate. Some DBK expansion options require more
settling time and can be measured at the slower rate of 10 µs/channel or 1000 µs/channel. All
types of measurements can be mixed within the scan group.

898095

Taking Measurements

8-1

Triggering
Triggering can be the most critical aspect of acquiring data. DaqScan/2000 Series devices
support a full complement of trigger modes to accommodate a wide variety of applications.
Hardware Analog Triggering. Many data acquisition products claim analog triggering, butrely on the PC to take readings and make a decision, which leads to uncertain and potentiallylong latencies. DaqScan/2000 Series devices use true analog triggering, whereby the trigger
level programmed by the user, sets an analog DAC, which is then compared in hardware to
the analog input level on the selected channel. The result is analog trigger latency that is
guaranteed to be less than 5 µs, significantly shorter than most data acquisition devices. Any
analog channel can be selected as the trigger channel. The user can program both the trigger
level, as well as the edge (rising or falling).
Digital Triggering (P1). Applies to DaqScan/2001 and DaqScan/2005 only. A separate
digital trigger input line is provided, allowing TTL-level triggering, again with latencies
guaranteed to be less than 5 µs. Both the logic levels (1 or 0), as well as the edge (rising or
falling), can be programmed for the discrete digital trigger input. The digital trigger input islabeled TTLTRG.
Digital Pattern Triggering (P2 and P3). The devices support digital pattern triggering,whereby the user can designate any of the digital input ports as the trigger port. The
programmed digital pattern, including the ability to mask or ignore specific bits, is thencompared to the actual input until a match is detected, after which the sequencer begins thescan sequence.
Counter Triggering. Triggering can be programmed to occur when one of the countersreaches, exceeds, or is within a programmed level. Any of the built-in counter/totalizer channels can be programmed as a trigger source.
Software-Based Triggering. Software-based triggering differs from the modes describedabove because the readings [analog, digital, or counter] are interrogated by the PC to detect
the trigger event [not in the hardware as described above]. The advantage of this mode isthat it permits triggering based on more complex situations. For example: to trigger on a
specific temperature, which was derived from the acquisition of at least two analog
measurements, plus the calculation of the measured temperature using linearization
algorithms.
Normally software-based triggering results in long latencies from the time that a trigger
condition is detected, until the actual capturing of data commences. However, circuitry
circumvents this undesirable phenomenon by use of pre-trigger data. Specifically, whensoftware-based triggering is employed, and the PC detects that a trigger condition has
occurred, (which may be thousands of readings later than the actual occurrence of the signal),
the Daq driver automatically looks back to the location in memory where the actual triggercausing measurement occurred. The acquired data that is presented to the user actually
begins at that point. The latency in this mode is equal to one scan cycle.

8-2

Taking Measurements

898095

DaqScan/2000 SeriesPre- and Post-Triggering Modes. Six modes of pre- and post-triggering are supported,
providing a wide variety of options to accommodate any measurement requirement. Whenusing pre-trigger, the user must use software-based triggering to initiate an acquisition.

•

No pre-trigger, post-trigger stop event. This, the simplest of modes, acquires
data upon receipt of the trigger, and stops acquiring upon receipt of the stoptrigger event.

•Fixed pre-trigger with post-trigger stop event. In this mode, the user
specifies the number of pre-trigger readings to be acquired, after which,acquisition continues until a stop-trigger event occurs.

•

No pre-trigger, infinite post-trigger. No pre-trigger data is acquired in thismode. Instead, data is acquired beginning with the trigger event, and is
terminated when the operator issues a command to halt the acquisition.

•

Fixed pre-trigger with infinite post-trigger. The user specifies the amount of
pre-trigger data to acquire, after which the system continues to acquire data until
the program issues a command to halt acquisition.

•

Variable pre-trigger with post trigger stop event (driver support only). Unlike
the previous pre-trigger modes, this mode does not have to satisfy the pre-trigger
number of readings before recognizing the trigger event. Thus the number of pretrigger readings acquired is variable and dependent on the time of the trigger
event relative to the start. In this mode, data continues to be acquired until thestop trigger event is detected.

•

Variable pre-trigger with infinite post trigger (driver support only). This issimilar to the mode just described, except that the acquisition is terminated uponreceipt of a command from the program.

•

Stop Trigger. Any of the software trigger modes described above can be used tostop an acquisition. Thus an acquisition can be programmed to begin on one
event, such as a temperature level, and then stopped on another, such as a digitalpattern event.

Synchronous I/O Operations
Synchronous Input Operations
The DaqScan/2000 Series devices allow synchronous scanning and acquisition of Analog Input[for those models with a P1 connector], Digital Input and Counter Input Data at up to 200 kHz
aggregate scanning rates. When applicable, the Analog Input data can be either main unit
(DaqScan/2001 or DaqScan/2005) or expansion modules from P1-compatible analog input
modules. The Digital Input data can be main unit 8-bit P2 (8255) digital input or from P2compatible DBK digital input expansion modules.
Analog Input Channels
P1 – Channels 0 through 15 (DaqScan/2001 and DaqScan/2005 only)
DaqScan/2001 and DaqScan/2005 allow analog input configuration of themselves, as well as
for any connected P1 compatible DBK analog input expansion modules.
Channel Selection and Mode Settings
DaqScan/2001 and DaqScan/2005 have P1 connectors that allow for expansion of up to 256
high-speed channels, or 896 TC channels. In regard to system expansion, refer to the DBK
Option Cards & Modules User’s Manual (p/n 457-0905).
Channel Range and Polarity
The analog input channels of DaqScan/2001 and DaqScan/2005 can be programmed for gain
settings of 1, 2, 4, 8, 16, 32, and 64.

DaqScan/2000 Series

898195

Taking Measurements

8-3

Channel Sampling Interval
In regard to analog input, DaqScan/2001 and DaqScan/2005 allow for programmablesampling intervals of 5 us, 10 us, or 1000 us on a per channel basis. This mode allows some
DBK expansion channels [that change slowly for a higher degree of accuracy] to be sampled ata longer interval, and channels that change more rapidly to be sampled at a shorter interval.
Each 5 µs, 10 µs, or 1000 µs interval reduces the maximum aggregate acquisition rate for theentire scan by that amount.

Digital I/O Channels
There are 40 Digital I/O channels available to DaqScan/2000 Series devices. The channels aregrouped as follows:
8 Channels, A0 through A7 from P2 Port A
8 Channels, B0 through B7 from P2 Port B
8 Channels, C0 through C7 from P2 Port C
16 Channels, Digital 0 through Digital 15 from P3
The devices allow for either synchronous scanning of digital input channels or asynchronous
I/O operations for all configured digital channels.
Counter Input Channels
Counter 0, 1, 2, 3 (CTR0, CTR1, CTR2, CTR3)
DaqScan/2000 Series devices allow synchronous scanning of four16-bit counter input
channels. The 4 channels can be cascaded into two 32-bit counter channels. For eithercascaded or non-cascaded counter channels each channel can be configured for:••Pulse Counting Mode – specifies that each counter should be cleared upon being
read and placed into the input scan.
Totalize Counting Mode – specifies that each counter is to free-run and not becleared during the input acquisition.

Synchronous Input Acquisition Clocking
A /I Clock
DaqScan/2000 Series devices allow clocking of the synchronized inputs either by an internal,
programmable pacer clock or by external clocking. These products use a sequencer toimplement a multiplexing approach to gathering the input data. This means that with either
internal or external clocking the entire channel scan (including the sampling time for eachchannel) may not exceed the maximum aggregate rate of 200 kHz.

Synchronous Output Operations
DaqScan/2001 and DaqScan/2004 allow synchronous output of any available D/A channels
at up to 100 kHz for each channel. All D/A channels may have output clocked out
synchronously.
Output Channel Configuration
Analog Output Channels (DaqScan/2001 and DaqScan/2004 only)
Each of the four D/A channels for DaqScan/2001 and DaqScan/2004 can be configured forwaveform output individually. If the D/A channel is not configured for waveform output itthen is available for asynchronous output operations.
Synchronous Output Clocking (DaqScan/2001 and DaqScan/2004 only)
A / O Clock
DaqScan/2001 and DaqScan/2004 allow clocking of the synchronized output by the acquisition
clock source, an internal, programmable pacer clock or by an external clock source. When theclock source generates a new clock signal all outputs are updated concurrently. Regardless ofthe clock source, the clock may not exceed the maximum update rate of 100 kHz.
8-4

Taking Measurements

898095

DaqScan/2000 Series

Synchronous Output Data Source (DaqScan/2001 and DaqScan/2004 only)
DaqScan/2001 and DaqScan/2004 allow the data source for synchronized DAC output
operations to be that of an internal memory-based buffer. The output data, for all channels, iscontained in the buffer.

Asynchronous I/O Operations
Each DaqScan/2000 Series device allows asynchronous input of any counter or digital channel
that is not currently configured for synchronous acquisition. The circuitry allows forasynchronous output to any D/A channels that are not currently configured for waveform
output. Likewise, the digital port can be used for both asynchronous input and output
operations. In addition, the timer outputs can be programmed at any time regardless of thecurrent state of synchronous or asynchronous operations on other channels.
Digital I/O Channels
Local 8255 Channels (P2: Port A, Port B, Port C)
The devices have an implemented Intel 8255 core in the digital I/O logic on the P2 port. With
the Intel 8255 there are three 8-bit wide ports available for I/O and one 8-bit wide port for
configuration purposes. The configuration port is used to configure the other three 8-bit ports
for either input or output operations.
Local 16-bit P3 PortThe 16-bit P3 digital port can be used as either an input or output port. With this port, noconfiguration is required. The port simply outputs when written to and inputs when read.
Expansion Digital I/O
The devices have the ability to expand their digital I/O capabilities via the P2 port and theconnection of applicable digital I/O expansion modules. These modules are discussed in the
DBK Option Cards & Modules User’s Manual (p/n 457-0905). When using the digital I/O
expansion modules the local P2 Intel 8255 digital I/O becomes inaccessible in lieu of the
expansion modules. These expansion modules provide additional Intel 8255 ports as well as
input isolation for applications that require the expanded capabilities.
Pulse Stream Output Using Timers(Timer0, Timer1)

Each device allows for the generation of output pulses based upon a programmable setting.
These output timers can be set at any time regardless of the state of any synchronous or
asynchronous operations which are currently taking place on other channels.
The timer outputs are through screw terminals on the front panel.
Analog Output Channels (DaqScan/2001 and DaqScan/2004 only)
DaqScan/2001 and DaqScan/2004 can output analog data to any of their four D/A channels.
Each D/A channel may be asynchronously updated by an application if the D/A channel is not
currently being used for waveform output operations. The four analog outputs are via P3 (seepinout in chapter 4).
Counter Input Channels (CTR0, CTR1, CTR2, CTR3)
DaqScan/2000 Series devices have counter input capabilities and have the ability to readcounter input providing that the counter channel is not configured for synchronous acquisition.
As in the case of synchronous operations the four 16-bit counter input channels can be used
individually or cascaded into two 32-bit counter channels. For either cascaded or noncascaded counter channels each channel can be configured for:

DaqScan/2000 Series•

Clear on Read Mode - specifies that each counter should be cleared (reset to 0)
upon being read.

•Continuous Totalize Mode – specifies that each counter is to free-run and not becleared during the read operation.

898195

Taking Measurements

8-5

Synchronizing Multiple Units
Up to four DaqScan/2000 Series devices can be synchronized to each other via their rear
panel SYNC connectors and associated cables. Each device has two identical SYNC ports;
either [or both] can be used to connect units. The DaqScan units can be scan-synchronized
and triggered from any other SYNC-connected unit. Note that cables CA-74-1 and CA-74-5
are 1-foot and 5-foot in length, respectively. The total combined cable length of the Sync
cables is not to exceed 15 feet (4.57 m).

Synchronization Concept Block Diagram
The preceding diagram shows how a DaqScan/2000 Series device can input or output
synchronization timing and trigger events on either SYNC connector. A more detailed block
diagram follows shortly.
Reference Notes:¾

The SYNC features are programmable via software. For detailed information
see the Using Multiple Devices and the daqAdcSetClockSource sections of theProgrammer’s Manual (p/n 1008-0901).

¾

During software installation, Adobe® PDF versions of user manuals
automatically install onto your hard drive as a part of product support. The
default location is in the Programs group, which can be accessed from theWindows Desktop. Refer to the PDF documentation for details regarding both
hardware and software.

Synchronization Model for DaqScan/2000 Series Devices (see note)
Note: DaqScan/2002 and DaqScan/2004 have no P1 connector; thus the pin 20
and pin 25 lines do not apply to those two devices.

8-6

Taking Measurements

898095

DaqScan/2000 SeriesIn regard to synchronization, the following applies:•

One of three signals can be used to drive the DaqScan/2000 Series acquisition scanrate. These are:− the internal acquisition pacer clock− an external acquisition pacer clock (for DaqScan/2001 and /2005 only)− the global sync (G-SYNC) input from the SYNC ports•

Both the SYNC connector input and the external clock input can be divided down.

•

When a DaqScan is in the Master Mode, both the trigger condition and the scan timing
are output on the SYNC port. The global trigger (G-TRIG) is selected instead of the
TTL Trigger Input.

Whether the DaqScan is using its internal scan clock, or external clock input, it can be
programmed to output the clock on the SYNC connector. In either case, the unit is behaving
like a synchronization master. Other DaqScan units [that are connected to the master, via
a SYNC port] should be programmed as synchronization slaves.
When a DaqScan is programmed as a synchronization slave, it will derive its scan period fromthe SYNC port. If the slave unit must be triggered at the same time as the master unit, thenthe slave unit should use TTL Trigger as its trigger source.

Sync Notes:•

When a DaqScan/2000 Series device is programmed as a synchronization slave, the TTL
trigger source is automatically derived from the SYNC port.

•

When a DaqScan/2000 Series device is programmed as a synchronization master, it will
output its trigger condition on the SYNC port.

•

When synchronizing two or more DaqScan/2000 Series devices, the slave units should haveat least 0.1 µs of dead time in the scan period. By “dead time,” we mean a duration inwhich no channels are sampled. This accommodates fundamental differences in the clocks[for each DaqScan].

•All DaqScans that are connected via SYNC cables can be scan-synchronized to within
0.1 µs of each other.

•

A maximum of 4 units can be synchronized; scan-synchronous (post-trigger).

•

Sync cables are not to exceed a total combined length of 15 feet (4.57m).

Multi-unit synchronization via the SYNC ports does not support pre-trigger
data. Only post-trigger data can be collected and time correlated between
the units.

DaqScan/2000 Series

898195

Taking Measurements

8-7

Using DBK Cards and Modules for Signal Conditioning
The DBK signal-conditioning cards and module are designed for use with DaqLabs, DaqScans,
DaqBooks, LogBooks, and various types of data acquisition boards, i.e., ISA, PCI, andcompact-PCI types. The DBKs perform best when used with an acquisition device that candynamically select channel, gain, and range. DBK cards and modules with dynamic channel
and gain/range selection allow for high channel-to-channel scan rates with a variety oftransducers.
Note: Only passive DBKs, such as the DBK1 BNC module, the DBK11A screw terminal card,and the DBK40 BNC analog interface, can be used with Daq PC-Cards.
DBK output signals can be bipolar, e.g., -5 to +5 V, or unipolar, e.g., 0 to 10 V. The user canselect a range of relevant values to correspond to the lowest signal (e.g., -5 or 0 V) and thehighest signal (e.g., 5 or 10 V) signal. This type of range selection guarantees the highest
resolution in 12-bit or 16-bit conversion.
The majority of these modules have their own power supply; however, several options exist
for packaging and powering the DBKs.

Reference Note:
DBK options are detailed in the DBK Option Cards and Modules User’s Manual(p/n 457-0905). As a part of product support, the manual is automatically loaded
onto your hard drive during software installation. The default location is thePrograms group, which can be accessed through the Windows Desktop. A PDF
version of the document can be read directly from the data acquisition CD via theintroduction screen’s <View PDFs> button.

8-8

Taking Measurements

898095

DaqScan/2000 Series

Calibration

9

Overview …… 9-1
Using DaqCal.exe …… 9-1

Overview
Daqscan/2000 Series devices are digitally calibrated at the factory. The digital method
involves storing a correction factor on the circuit board for each range at the time ofcalibration. Whenever a particular range is selected, the appropriate calibration constant is
automatically applied to a compensating DAC, thereby calibrating the specific range. The
result is that readings generated by the A/D are already calibrated, and do not require
additional processing.
This is significantly better than the calibration method in which readings are adjusted insoftware after they are transferred to the host PC. That method reduces the dynamic range of
the A/D, and can adversely affect the speed at which the PC obtains a calibrated reading.
Daqscan/2000 Series devices include a user-cal mode, whereby the user can adjust thecalibration of the unit’s board, without destroying the supplied factory calibration. This isaccomplished with two distinct calibration tables in the unit’s on-board EPROM, one whichcontains the factory calibration factors, and the other which is available for user calibration.

Using DaqCal.exe
When calibration adjustments are needed they should be completed in the following order:

1.
2.
3.
4.
5.
6.
7.
8.

PGA Input and Output Offset
Sample/Hold Offset
A/D Offset and Gain
VDC Voltage Reference
DAC0 Full-Scale1
DAC1 Full-Scale1
DAC2 Full-Scale1
DAC3 Full-Scale1
1

DAC Full-Scale only applies to Daqscan/2001 and DaqScan/2004.

A Windows-based program, DaqCal.exe, is used to calibrate Daq systems, including analog
expansion cards. DaqCal.exe is used in conjunction with:•••

a 4.5 digit, digital multi-meteran adjustable voltage calibrator
an ambient temperature meter

To use the calibration program:1.

Launch DaqCal.
Note:

2.

Daqscan/2000 Series

DaqCal is installed automatically from your data acquisition CD as a partof product support. This takes place during software installation. DaqCal’sdefault location is the IOtech DaqX Software folder, in the Programsgroup.

When DaqCal opens you will be prompted to select your device from a list. After
doing so, simply follow the illustrated on-screen instructions.

948695

Calibration

9-1

9-2

Calibration

948695

Daqscan/2000 SeriesTroubleshooting and Customer Support

10

Electrostatic Discharge (ESD), Handling Notice…… 10-1
Product Care …… 10-1
ReadMe Files and the Install CD-ROM ……10-2
DaqCOM Issues …… 10-2
Driver Support……10-2
Ethernet Problems …… 10-3
Frequently Asked Questions …… 10-4
Customer Support …… 10-5

Electrostatic Discharge (ESD), Handling Notice
The discharge of static electricity can damage some electronic components. Semiconductor
devices are especially susceptible to ESD damage. You should always handle components
carefully, and you should never touch connector pins or circuit components unless you arefollowing ESD guidelines in an appropriate ESD-controlled area. Such guidelines include theuse of properly grounded mats and wrist straps, ESD bags and cartons, and related
procedures.

Product Care
Daq products and DBK options are essentially maintenance free and require a minimal amountof care. They should be treated much like any other high-tech equipment. In general:• Keep them clear of harsh chemicals and abrasive elements.
• Avoid exposing the products to extreme heat, for example, avoid setting up a Daq systemnear a boiler or furnace.

• Avoid extreme shock and vibration.
• Avoid subjecting the units to liquids and extremely fine air particulate, such as silica dust.
• Never touch circuit boards without proper ESD (Electrostatic Discharge) measures inplace.

A “common-sense” approach to handling your data acquisition equipment will go a long way inprotecting it from inadvertent damage.

Note that you can use lint-free rags and Isopropyl Alcohol(Rubbing Alcohol) to clean the outer plate surfaces of Daq
devices and optional DBK modules.

DaqScan/2000 Series

898095

Troubleshooting & Customer Support

10-1

ReadMe Files and the Install CD-ROM
The Install CD-ROM includes ReadMe Files. These files often contain late-breaking informationthat may not appear in the user documentation. During installation you should review the
ReadMe files when prompted to by the program.
The Install CD-ROM includes:• DaqBook/2000 Series1 Windows NT/2000/XP driver• DaqX.DLL (32-bit API)• Microsoft C/C++ API support• Microsoft VB API support• Daq Configuration Control Panel Applet• User documentation in Adobe® PDF format• DaqView
Upon installation, the user needs to run the Daq Configuration Control Panel applet. The
applet is used to configure the Daq device for an appropriate Ethernet Port. This must be
performed in order for the application to access the device through the Windows NT/2000/XP
driver.
1

DaqBook/2000 Series drivers also apply to DaqScan/2000 Series devices.

Driver Support
The daqX API can be used with WaveBook, DaqBook, DaqBoard, DaqLab, DaqScan,
Daq PC Card and TempBook product lines. All daqX functions share the daq prefix.
The API support examples can be found in the installed directory:<Install> ProgrammingLanguageSupport\DAQ
Refer to the Programmer’s Manual, p/n 1008-0901.

DaqCOM Issues
DaqCOM support is contained within the Help file support, which is installed with DaqCOM. It
is located within the DaqCOM program group.
DaqCOM support will not be available unless it has been specifically installed with the product.
If the DaqCOM program group is not listed under programs (which can be accessed via the
Windows desktop), then you will need to install it from the data acquisition CD, or download itfrom our website.

10-2

Troubleshooting & Customer Support

898095

DaqScan/2000 SeriesEthernet Problems• Make sure the Ethernet network is not experiencing technical problems. Check with your
Network Administrator.

• Ensure that your system is properly configured in the Daq Configuration Applet [located in
the Windows’ Control Panel]. If a device is not configured correctly it will not be
accessible from an application.

• Verify that the correct MAC (Media Access Control) number is reflected by the software.
• If communications cannot be established with the Daq device or, if trying to connect
causes the system to hang or crash, try the following:-

Verify that the device power switch is set to ON and that the POWER ON indicator
is lit.

-Ensure that all physical device interface connections are correct. This includes
Ethernet connections from the Daq device to the host PC’s Ethernet jack, or
Ethernet Network hub.

-Run the Daq Configuration applet from the Windows Control Panel. Make surethe device is configured to the appropriate Ethernet port with the appropriate
protocol. Run the device test and if it fails try other protocols for the device. Be
sure to click on <Apply> to apply the settings, before re-running the test.

• If problems persist, notify your Network Administrator. If he or she can’t solve theproblem, contact the factory.

DaqScan/2000 Series

898095

Troubleshooting & Customer Support

10-3

Frequently Asked Questions(1) Topic: Environmental Factors
Question: What environments are the products intended for?
Answer:

Refer to the product specifications in regard to operating and storage temperature
ranges, and relative humidity. Note that system components are not intended to be
exposed to harsh environments and should always be protected from snow, rain,
extreme dust, and harsh sun.

(2) Topic: DaqX
Question: What is DaqX software?
Answer: DaqX software is low-level software support for application development. DaqX is
installed during the installation of the DaqView application. DaqX supports VisualC++ and Visual Basic programming environments.

Question: Where can I get the latest version of DaqX software?
Answer: You can download the latest versions from our website. This service is free.

(3) Topic: Calibration
Question: Should I calibrate my Daq device when I receive it?
Answer: No, because each Daq-related product is factory-calibrated to its rated accuracy
before it leaves the factory.

Question: When is calibration required?
Answer: Depending on your operating guidelines, your equipment will require periodic
calibration to ensure the original accuracy is maintained. We recommend that you
have your DaqScan system calibrated at least once a year.

(4) Topic: Computers for Data Acquisition

10-4

Question:

What type of computer do you recommend for use with DaqScan/2000 Series
devices?

Answer:

The PC should have built in Ethernet capabilities and use an operating system ofWindows NT, 2000, XP, or later.

Troubleshooting & Customer Support

898095

DaqScan/2000 Series

Customer Support
If you want to Expand or Enhance Your Daq System . . .
You can visit our internet site at ZZZPFFGDTFRP to find the latest accessories,
expansion cards and modules. You can also call our sales department directly at() -.

If you need to Report Problems or Request Product Support
Note:

Please do not return any equipment to the factory unless it has an RMA
number (Return Merchandise Authorization number). RMA numbers areissued by the factory.

To report problems or request support, contact our applications department or your service
representative. Contact information follows shortly. When you contact us, please have thefollowing information available, as applicable:•
Hardware model numbers and software version numbers.
•

Operating system, type of computer, and device information in the Windows
control panel, such as interrupts and address settings for our hardware andothers.

•Results of tests, such as the Daq Configuration control panel.

•

Hardware setup and software configuration.

•

Information on input signals, including voltage ranges, signal impedance ranges,
noise content, and common mode voltages.

•

The name of a contact person from your company who can discuss the problems
encountered.

••

Instructions for return shipping.

•

All troubleshooting notes and comments on tests performed, and all problemrelated conditions.
If your problem is Ethernet related, discuss the problem with your Network
Administrator prior to contacting the factory. Please provide us with youradministrator’s comments, if applicable.

Note: Before calling for assistance, take a few minutes to read all parts of the manual that
may be relevant to the problem. Also, please review the troubleshooting material.
You can reach 0HDVXUHPHQW&RPSXWLQJ by one of the following means:
Phone: () 
Fax:( 

E-mail (Product Information/Sales7HFKQLFDO6XSSRUW): LQIR@PFFGDT.com,QWHUQHWZZZPFFGDTFRP
Mail: 0HDVXUHPHQW&RPSXWLQJ&RUS. z &RPPHUFH:D\ z 1RUWRQ0$ 
All equipment returned to the manufacturer must have an RMA (Return Material Authorization)
number. You can obtain an RMA number by calling the Applications Department. When
returning the equipment, use the original shipping container (or equivalent) to prevent
damage. Put the RMA number on your shipping label to ensure that your shipment will behandled properly. After receiving your equipment, we will fax a confirmation form thatsummarizes the charges (if applicable) and expected return date.

DaqScan/2000 Series

898095

Troubleshooting & Customer Support

10-5

10-6

Troubleshooting & Customer Support

898095

DaqScan/2000 Series

Specifications

11

DaqScan/2000 Series

GeneralSupply Voltage Range: 90 to 250 VAC
Power Required: 15W, assuming no DBK options
Power Available for External Signal Conditioning & Expansion Options:
5V at 1A, ±15V at 500 mA, not to exceed 10W
Operating Temperature: 0°C to + 50°C (32° to 122°F)
Storage Temperature: -40°C to +80°C (-40° to 176°F)
Relative Humidity: 0 to 95% non-condensing
Vibration: MIL STD 810E, Categories 1 and 10
Signal I/O Connectors:
DaqScan/2001 and DaqScan/2005: DB37 male connectors: P1, P2, P3 on rear panel
DaqScan/2002 and DaqScan/2004: DB37 male connectors: P2 and P3 on rear panel
Dimensions: 425 mm W x 220 mm D x 45 mm H (16.75” x 8.5” x 1.75”)
Weight: 2.3 kg (5 lbs.)

A/D Specifications
Type: Successive approximation
Resolution: 16 bit
Conversion Time: 5 µs
Maximum Sample Rate: 200 kHz
Nonlinearity (Integral): ±1 LSB
Nonlinearity (Differential): ± 2 LSB
Missing Codes: None, over full operating temperature range

Analog Inputs (DaqScan/2001 and DaqScan/2005 only)
Channels:
16 single-ended or 8 differential; programmable on a per-channel basis as single-ended or differential and as
unipolar or bipolar
Connectors:
37-pin DSUB (P1) on rear panel
Expansion:
Up to 896 TC channels when used with DBK90 expansion option [1 µs/channel], or up to 256 channels when usedwith other expansion options [5 µs/channel]
Bandwidth: 500 kHz
Settling Time: 5 µs maximum to 1 LSB for full-scale step
Temperature Coefficient: ±(10ppm +0.3 LSB)/°C outside the range of 0° to 35°C
Input Impedance: 10 MΩ for single-ended; 20 MΩ for differential
Bias Current: <1nA at 0° to 35°C
Common Mode Rejection: 86 dB, DC to 60 Hz for gains < 8; >100 dB for gains >16
Maximum Input Voltage: +11V relative to analog common
Over-Voltage Protection: ±35V
Crosstalk: -100 dB DC to 60 Hz; -86 dB @ 10 kHz

DaqScan/2000 Series

898095Specifications

11-1

Specifications are subject to change without notice.

Ranges: Software programmable via sequencer on a per-channel basis.
Voltage Range(Note 1)

Accuracy (Note 2)
One Year, 0 to 35°C±(% reading + % range)

Input Noise (Note 3)(LSB rms)
10 Hz to 200 kHz

Absolute
0.015 + 0.005
0.015 + 0.005
0.015 + 0.005
0.015 + 0.008
0.015 + 0.008
0.015 + 0.008
0.015 + 0.005
0.015 + 0.005
0.015 + 0.005
0.015 + 0.005
0.015 + 0.008
0.015 + 0.008
0.020 + 0.008

Typical
1
1
1
2
2
2
1
1
1
2
2
3
3

0 to +10 V
0 to +5 V
0 to +2.5 V
0 to +1.25 V
0 to +0.625 V
0 to +0.3125 V-10 to +10 V-5 to +5 V
-2.5 to +2.5 V-1.25 to +1.25 V-0.625 to +0.625 V-0.3125 to +0.3125 V-0.156 to +0.156 V

Notes:
1) Specifications assume differential input single channel scan, 200 kHz scan rate, unfiltered.
2) Accuracy specification is exclusive of noise. Measurements were taken at P1.
3) Inputs shorted to Signal Ground (SGND). 8192 samples.

Input Sequencer
Analog, digital and counter inputs can be scanned synchronously based either on an internal programmable timer, oran external clock source. Analog and digital outputs can be synchronized to either of these clocks.
Scan Clock Sources: 2•
Internal, programmable from 5 µs to 5.96 hours maximum in 1 µs steps•
External, TTL level input up to 200 kHz maximum
Programmable Parameters per Scan: channel (random order), gain, unipolar or bipolar
Depth: 16,384 locations
On-board Channel-to-Channel Scan Rate: 5 or 10 µs per channel, programmable
Expansion Channel Scan Rate: 5 µs,10 µs, or 1000 µs per channel, programmable

External Acquisition Scan Clock Input
Maximum Rate: 200 kHz
Clock Signal Range: 0V to +5V
Minimum Pulse Width: 50 ns high, 50 ns low

External Sync Ports
Number of External Sync Ports: 2, on rear panel
Maximum Number of Units to be Synchronized: 4 units, scan-synchronous (post trigger)
Maximum Length of Sync Cables: 15 feet (4.57 m), total for all cables

Triggering
Trigger Sources: 6, individually selectable for starting and stopping an acquisition. Stop acquisition can occur ona different channel than start acquisition; stop acquisition can be triggered via modes 2, 4, 5, or 6 described
below. Pre-trigger is supported with fixed or variable pre-trigger periods.
1. Single-Channel Analog Hardware Trigger
Any analog input channel, including any of the analog expansion channels, can be software programmed as
the analog trigger channel.
Input Signal Range: -10V to +10V max
Trigger Level: Programmable, 11-bit resolution
Hysteresis: Programmable, 11-bit resolution
Latency: 5 µs max

11-2

Specifications

898095

DaqScan/2000 Series

Specifications are subject to change without notice.

2. Single-Channel Analog Software Trigger.
Any analog input channel, including any of the analog expansion channels, can be selected as the software
trigger channel. If the trigger channel involves a calculation, such as with temperature, then the driver
automatically compensates for the delay required to calculate the reading, resulting in a maximum latency of
one scan period.
Input Signal Range: Anywhere within the range of the selected trigger channel
Trigger Level: Programmable, 16-bit resolution, including “window triggering”
Hysteresis: Programmable, 16-bit resolution
Latency: One scan period max.
3. Single-Channel Digital Trigger: A separate digital input is provided for digital triggering.
Input Signal Range: -15V to +15V max.
Trigger Level: TTL
Minimum Pulse Width: 50 ns high, 50 ns low
Latency: 5 µs max.
4. Digital Pattern Triggering: 8 or 16-bit pattern triggering on any digital input. Programmable for trigger on
equal, above, below, or within or outside of a window. Individual bits can be masked for “don’t care”
condition.
Latency: One scan period maximum.
5. Counter/Totalizer Triggering: Counter/totalizer inputs can trigger an acquisition. User can select to trigger
on a frequency or on total counts that are equal, above, below, or within/outside of a window.
Latency: One scan period maximum.
6. Software Triggering: Trigger can be initiated under program control.

Analog Outputs (DaqScan/2001 and DaqScan/2004 only)
For DaqScan/2001 and DaqScan/2004, four analog output channels are updated synchronously relative toscanned inputs, and clocked from either an internal onboard clock, or an external clock source. Analog outputs
can also be updated asynchronously, independent of any other scanning in the system.
Channels: 4 DAC channels; numbered 0, 1, 2, and 3
Connectors: pins 31 (DAC3), 32 (DAC2), 33 (DAC1), and 34 (DAC0) on the P3 DB37 connector
Resolution: 16 bits
Data Buffer: 256 Ksample
Output voltage range: ±10 V
Output current: ±10 mA
Offset error: ±0.0045 V maximum
Gain error: ±0.01%
Digital Feedthru: 50 mV when updated
Update rate: 100 kHz maximum, 1.5 Hz minimum (no minimum with external clock)
Settling Time: 10 µs maximum to 1 LSB for full-scale step
Clock Sources: 4 programmable
1. Onboard D/A clock, independent of scanning input clock
2. Onboard scanning input clock
3. External D/A input clock, independent of external scanning input clock
4. External scanning input clock; available via screw-terminal on front panel removable block

DaqScan/2000 Series

898095Specifications

11-3

Specifications are subject to change without notice.

Digital I/O
Channels: 40, expandable up to 272 with external digital DBK options
Input Scanning Modes: 2 programmable
1. Asynchronous, under program control at any time relative to input scanning
2. Synchronous with input scanning
Ports: 3 x 8-bit (82C55 emulation), and 1 x 16-bit; each port is programmable as input or output.
Connectors:
37-pin DSUB (P2) on rear panel: for three 8-bit ports (24-bits)
37-pin DSUB (P3) on rear panel for DIGITAL I/O channels 0 through 15 (16-bit port)
Input Characteristics: See following Digital I/O figures
Input protection: ±8 kV ESD clamp diodes parallel
I/O levels: TTL
Sampling/Update rate: 200 kHz maximum
Output Characteristics: See following Digital I/O figures

Frequency/Pulse Counters
Counter inputs can be scanned synchronously along with analog and digital scanned inputs, based either oninternal programmable timer, or an external clock source. Bus mastering DMA provides CPU and systemindependent data transfers, ensuring data acquisition performance irrespective of other system activities.
Counters can be configured to clear when read, or to totalize and clear under program control.
Channels: 4 x 16-bit, cascadable as 2 x 32-bit
Frequency Measurement Rate: 10 MHz maximum
Connectors: 37-pin DSUB (P3) on rear panel; pin 18 (CTR0), pin 36 (CTR1), pin 17 (CTR2), pin 35 (CTR 3)
Input Signal Range: -15 V to +15 V
Trigger Level: TTL
Input Characteristics: 2.7 kΩ series in parallel with 20 pF to common and 10 kΩ to +5 V
Input protection: ±8 kV ESD clamp diodes parallel
Minimum pulse width: 50 ns high, 50 ns low

Timer Outputs
Channels: 2 x 16-bit
Connectors: 37-pin DSUB (P3) on rear panel; pin 15 (TMR0), pin 16 (TMR1)
Output Waveform: Square wave
Output rate: 1 MHz base rate divided by 1 to 65535 (programmable)
High level output voltage: 2.0 V minimum @ -3.75 mA, 3.0 V minimum @ -2.5 mA
Low level output voltage: 0.4 V maximum @ 2.5 mA

11-4

Specifications

898095

DaqScan/2000 Series

Specifications are subject to change without notice.

Accessories and Cables
Cables for use with DaqScan/2000 Series Devices
Description

Part No.

1.

Molded expansion cable, 2 in.

CA-255-2T2.

Molded expansion cable, 4 in.

CA-255-4T

3.

DBK expansion cable, not-molded, 2.5 in.

CA-37-1T

4.

DBK expansion cable, not-molded, 4.5 in.

CA-37-3T

5.

Expansion-card cable. Connects DBK series expansion card options.
The “x” in the part no. indicates the number of cards that can beconnected via the cable.

CA-37-x

6.

Ethernet patch cable, shielded, 18 in.

CA-242

7.

Ethernet patch cable, shielded, 7 ft.

CA-242-7

8.

SYNC cable, RJ-11, 6 conductor, 1 ft.

CA-74-1

9.

SYNC cable, RJ-11, 6 conductor, 5 ft.

CA-74-5

The total combined length of the SYNC cables is not to exceed 15 feet (4.57 m).

Manuals
Manual Set (Set #1126-0900). This is a hardcopy set of manuals that is available for purchase.
The set includes the following three manuals:
DaqScan/2000 Series User’s Manual (p/n 1126-0901)
Programmer’s Manual for developing custom programs using API commands.
(p/n 1008-0901)
DBK Option Cards & Modules User’s Manual. (p/n 457-0905)

Software
DaqViewXL for seamless execution of DaqView from within Microsoft Excel’s tool palette.
DASYLab icon-based data acquisition graphics, control, and analysis software.
eZ-TimeView post-acquisition time-domain analysis software
eZ-FrequencyView post-acquisition frequency domain analysis software

DaqScan/2000 Series

898095Specifications

11-5

Specifications are subject to change without notice.

This page is intentionally blank.

11-6

Specifications

898095

DaqScan/2000 SeriesGlossary

Acquisition

A collection of scans acquired at a specified rate as controlled by the sequencer.

AnalogA signal of varying voltage or current that communicates data.

Analog-to-Digital
Converter (ADC)

A circuit or device that converts analog values into digital values, such as binary bits, for use indigital computer processing.

API

Application Program Interface. The interface program within the Daq system’s driver that
includes function calls specific to Daq hardware and can be used with user-written programs(several languages supported).

BipolarA range of analog signals with positive and negative values (e.g., -5 to +5 V); see unipolar.

BufferBuffer refers to a circuit or device that allows a signal to pass through it, while providing
isolation, or another function, without altering the signal. Buffer usually refers to:(a)

A device or circuit that allows for the temporary storage of data during data transfers.
Such storage can compensate for differences in data flow rates. In a FIFO (First In - First
Out) buffer, the data that is stored first is also the first data to leave the buffer.

(b)

A follower stage used to drive a number of gates without overloading the preceding stage.

(c)

An amplifier which accepts high source impedance input and results in low source
impedance output (effectively, an impedance buffer).

Buffer Amplifier

An amplifier used primarily to match two different impedance points, and isolate one stage froma succeeding stage in order to prevent an undesirable interaction between the two stages.
(Also see, Buffer).

ChannelIn reference to Daq devices, channel simply refers to a single input, or output entity.
In a broader sense, an input channel is a signal path between the transducer at the point of
measurement and the data acquisition system. A channel can go through various stages(buffers, multiplexers, or signal conditioning amplifiers and filters). Input channels are
periodically sampled for readings.
An output channel from a device can be digital or analog. Outputs can vary in a programmed
way in response to an input channel signal.

Common mode

Common mode pertains to signals that are identical in amplitude and duration; also can be usedin reference to signal components.

Common mode
voltageCommon mode voltage refers to a voltage magnitude (referenced to a common point) that isshared by two or more signals. Example: referenced to common, Signal 1 is +5 VDC and
Signal 2 is +6 VDC. The common mode voltage for the two signals is +5.5 VDC [(5 + 6)/2].

CrosstalkAn undesired transfer of signals between systems or system components. Crosstalk causes
signal interference, more commonly referred to as noise.

DigitalA digital signal is one of discrete value, in contrast to a varying signal. Combinations of binary
digits (0s and 1s) represent digital data.

Digital-to-Analog
Converter (DAC)

A circuit or device that converts digital values (binary bits), into analog signals.

DIP switch

A DIP switch is a group of miniature switches in a small Dual In-line Package (DIP). Typically,users set these switches to configure their particular application.

Differential mode

The differential mode measures a voltage between 2 signal lines for a single channel. (Also see
single-ended mode).

Glossary

959395G-1

Differential mode
voltage

Differential mode voltage refers to a voltage difference between two signals that are referenced
to a common point. Example: Signal 1 is +5 VDC referenced to common. Signal 2 is +6 VDC
referenced to common.
If the +5 VDC signal is used as the reference, the differential mode voltage is +1 VDC(+ 6 VDC - +5 VDC = +1 VDC).
If the +6 VDC signal is used as the reference, the differential mode voltage is -1 VDC(+ 5 VDC - +6 VDC = -1 VDC).

ESDElectrostatic discharge (ESD) is the transfer of an electrostatic charge between bodies having
different electrostatic potentials. This transfer occurs during direct contact of the bodies, orwhen induced by an electrostatic field. ESD energy can damage an integrated circuit (IC).

Excitation

Some transducers [e.g. strain gages, thermistors, and resistance temperature detectors(RTDs)] require a known voltage or current. Typically, the variation of this signal through thetransducer corresponds to the condition measured.

Gain

The degree to which an input signal is amplified (or attenuated) to allow greater accuracy andresolution; can be expressed as ×n or ±dB.

Isolation

The arrangement or operation of a circuit so that signals from another circuit or device do notaffect the isolated circuit.
In reference to Daq devices, isolation usually refers to a separation of the direct link between
the signal source and the analog-to-digital converter (ADC). Isolation is necessary whenmeasuring high common-mode voltage.

Linearization

Some transducers produce a voltage in linear proportion to the condition measured. Other
transducers (e.g., thermocouples) have a nonlinear response. To convert nonlinear signals
into accurate readings requires software to calibrate several points in the range used and theninterpolate values between these points.

Multiplexer (MUX)

A device that collects signals from several inputs and outputs them on a single channel.

Sample (reading)

The value of a signal on a channel at an instant in time. When triggered, the ADC reads thechannel and converts the sampled value into a 12- or 16-bit value.

Scan

A series of measurements across a pre-selected sequence of channels.

Sequencer

A programmable device that manages channels and channel-specific settings.

Simultaneous
Sample-and-Hold

An operation that gathers samples from multiple channels at the same instant and holds thesevalues until all are sequentially converted to digital values.

Single-ended mode

The single-ended mode measures a voltage between a signal line and a common reference that
may be shared with other channels. (Also see differential mode).

Trigger

An event to start a scan or mark an instant during an acquisition. The event can be defined invarious ways; e.g., a TTL signal, a specified voltage level in a monitored channel, a button
manually or mechanically engaged, a software command, etc. Some applications may use
pre- and post-triggers to gather data around an instant or based on signal counts.

TTL

Transistor-Transistor Logic (TTL) is a circuit in which a multiple-emitter transistor has replaced
the multiple diode cluster (of the diode-transistor logic circuit); typically used to communicate
logic signals at 5 V.

Unipolar

A range of analog signals that is always zero or positive (e.g., 0 to 10 V). Evaluating a signal in
the right range (unipolar or bipolar) allows greater resolution by using the full-range of thecorresponding digital value. See bipolar.

G-2959395Glossary