DBK206

Screw-Terminal Adapter BoardFeatures• Provides removable screw-terminalconnection• Includes P1, P2, and P3 connectors• Optional rack and DIN-rail mountable
The DBK206 screw-terminal board provides
convenient, removable screw-terminal
access to all P1, P2, and P3 ­signals. The
DBK206 features three horizontally
mounted male DB37 connectors for attaching to a DAQ system, or for analog
and digital channel expansion (P1 analog
I/O, P2 digital I/O, and P3 analog output,
digital and counter-timer I/O).

The DBK206 provides removable screw-terminal connectors for DaqBoard/2000 Series I/O

DBK206 Block Diagram

SIGNAL I/O
P4

The board mates with the DaqBoard/2000
Series boards via a 3 ft. CA-195 or 6 ft.
CA-195-6 cable, and each port (P1, P2, andP3) connects to DBK signal conditioning
and expansion options via optional CA-37-x
cable(s). The standard DBK206 can be panel
mounted, but it also is DIN-rail mountablewith option DIN1 and is 19” rack-­mountablewith optional Rack3.

P1 screw terminal
analog input
P2 screw terminal
digital I/O
P3 screw terminal
analog output, digital I/O, counter/timer

Specifications

Connectors
P4: 100-pin connector provides for connection toDaqBoard/2000 Series P4 connector via a CA-195
cable
P1: One P1 (DB37) connector provides for analogexpansion via a single CA-37-x cable or multiple
CA-37-1 cables or connection to a DAQ system
P2: One P2 (DB37) connector provides for digital
expansion via a single CA-37-x cable or multiple
CA-37-1 cables or connection to a DAQ system
P3: One P3 (DB37) connector provides for digitaland analog output accessScrew Terminals: 12 sets of removable 10-connector
blocks for I/O signals
Wire Size Range: 10-22 gage

Ordering Information
Description
Termination and expansion board
with removable screw terminals
for access to all DaqBoard/2000 Series
I/O with direct connection for DBK
signal conditioning and expansion
options (P1, P2 and P3 for analog
and digital I/O) via CA-37-1 cable(s);connects to DaqBoard/2000 Series
P4 connector via a CA-195
or CA-209 cable
Expansion board for direct connection
to DBK signal conditioning andexpansion options

Measurement Computing

P1On-board P1 connector for analog expansion
or connection to Daq system P1

P2

On-board P2 connector for digital I/O expansion
or connection to Daq system P2

P3

On-board P3 connector for analog output
and digital I/O access or connection to Daq system P3

Accessories

CablesDescription
Part No.
Rack-mount kit for DBK206, DBK207,
DBK208, DBK209, and DBK210 signal
termination and expansion boards
Rack3
DIN-rail adapter kit for DBK206, DBK207,
DBK208, and DBK210 signal terminationand expansion boards
DIN1
DIN-rail adapter kit for DBK209 signal
termination and expansion boardDIN2Description
Part No.
Ribbon cable for connecting DBK Series
expansion options; where x is thenumber of DBK devices attachedCA-37-x
100-conductor expansion cable, mateswith the DaqBoard/2000 Series,
DBK200, DBK202, DBK203A, DBK206,
DBK207, DBK207/CJC, DBK208,
DBK209, and DBK210		 3 ft. expansion cableCA-195		 6 ft. expansion cableCA-195-6
Same as CA-195 with
CE compliance; 3 ft.
CA-209Part No.

Product Compatibility✔ LogBook✔ DaqBook✔ DaqScan✔ DaqBoard/2000 Series

DBK206

Rack3 rack-mount kit for the DBK206

DBK209(508) 946-5100

1info@mccdaq.com

mccdaq.com