MODEL 3022 ACCELEROMETERSPECIFICATIONSDIMENSIONS

Piezoresistive MEMS
DC Response
Circuit Board Mountable
Low Cost

The Model 3022 is a silicon MEMS accelerometerin a Wheatstone bridge configuration. The accelerometer ispackaged on a ceramic substrate with an epoxy sealed
ceramic cover and is designed for adhesive mounting. Theaccelerometer is offered in ranges from ±2g to ±200g range
and provides a flat frequency response to minimum 2000Hz.
The silicon MEMS sensor is gas damped and incorporates
over-range stops for high-g shock protection.
For a similar accelerometer designed for bolt
mounting, see the model 3028..

FEATURES

Adhesive Mounted±0.5% Non-linearity
Open Wheatstone Bridge
DC ResponseGas Damping
Built-in Overrange Stops
Low Power Consumption

APPLICATIONS





SENSOR SOLUTIONS /// Model 3022 Rev B

Vibration & Shock Monitoring
Motion Control
Impact & Shock Testing
Modal Analysis
Embedded Applications
Machinery

07/2017

Page 1

MODEL 3022 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 5Vdc excitation unless otherwise stated. Measurement Specialties reserves the right to update andchange these specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)1
Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)±28.0-20.00-150
700±0.5
30.7
2000

ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Input Resistance (Ω)
Output Resistance (Ω)
Insulation Resistance (MΩ)
Residual Noise (µV RMS)
Ground Isolation

Notes±201.5-3.00-600
1500±0.530.75000±500.6-1.50-1000
4000±0.5
30.7
5000±1000.3-0.60-1500
6000±0.5
30.7
5000±2000.15-0.30-2000
8000
±0.5
30.6
5000

±25±25±25
2 to 102 to 102 to 102500-6500 2500-6500 2500-6500
2500-6500 2500-6500 2500-6500>100>100>100101010
Isolated from Mounting Surface±25
2 to 102500-6500
2500-6500>10010±25
2 to 102500-6500
2500-6500>10010±25
2 to 102500-6500
2500-6500>10010±25
2 to 102500-6500
2500-6500>10010Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Compensated Temperature (°C)
Storage Temperature (°C)

-0.09
-0.09
-0.15
-0.15
-40 to +125
Not Compensated-40 to +125

-0.09
-0.15

-0.09
-0.15

-0.09
-0.15

-0.09
-0.15

TypicalTypicalPHYSICAL
Case Material
Weight (grams)
Mounting

Ceramic
3.1
Adhesive or solder

12±56.0-15.00-250
800±0.530.7
2000±103.0-6.00-400
1000±0.5
30.7
5000

-0.09
-0.15

@5Vdc Exc.
±5%@50Vdc
MaximumSee Note 2

Output is ratiometric to excitation voltage
Order model 3022-XXX-10254 for temperature compensation resistor values included in the calibration certificate.

Optional accessories:

121
140

Three Channel DC Differential Amplifier
Auto-Zero Inline Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 3022 Rev B

07/2017

Page 2

MODEL 3022 ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range+Electrical Connection

3022-GGG-Pl ll l_____Electrical Connection (P=pins, N=solder pads)
l___________Range (010 is 10g)

Example: 3022-010-P
Model 3022, 10g, Pins

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 3022 Rev B

07/2017

Page 3