QUICK START

LogBook/360

Stand-alone, Intelligent PC-based,
Data Acquisition System

LogBook/360
461-0941 rev 3.0*324587C-01*
324587C-01

Measurement Computing
10 Commerce Way
Norton, MA 02766(508) 946-5100
Fax: (508) 946-9500
info@mccdaq.com

www.mccdaq.com

LogBook/360

Quick Start Guide

Software Installation …… QS360-1
Setting Up the Card Drawer …… QS360-3
Connecting the LogBook/360 to the PC,
External DBKs, and Power …… QS360-6
Hardware Configuration …… QS360-7
LogBook/360 Device Configuration… QS360-8
Testing the Hardware …… QS360-10
Acquisition Configuration …… QS360-10
A Note About Calibration …… QS360-11

Before you get started

verify that you have the following items.
LogBook/300
TR-40U Power Supply
Serial/Parallel Patch Cable, or PC Card Reader
PC with ECP (Enhanced Capabilities Port)

Data Acquisition CD
Monitor: SVGA, 1024 x 768 screen resolution
Windows XP users:
PC with Intel™ Pentium [or equivalent], 1 GHz;
512 MB memory; 10 GB disk space
Windows Vista users:
PC must be Windows Vista Premium Ready

IMPORTANT! Software must be installed first!

Software Installation
1.

Close all running applications on the host PC.

2.

Remove previous version drivers, if present. You can do this through Microsoft’s Add/Remove Programs
feature.

3.

Insert the Data Acquisition CD (8.1.2 or higher) into your CD-ROM drive and wait for the PC to auto-run. An
Opening Screen will appear. If AutoRun is disabled: (a) right click Windows Start menu, (b) select Explore; and(c) double-click MasterSetup.exe. As an alternative to using the CD, you can download software from:www.mccdaq.com/software.
Click the <ENTER SETUP> button.
Note: If you are downloading software from our website, follow instructions provided there.

4.
5.

From the hardware selection screen, select your data acquisition device then follow the on-screen instructions.

Note: To enable all LogView features:(a) From LogView’s main window, open the File pull-down menu,(b) select Authorization,(c) enter ED7B55484273 in the dialog box and apply the code.

LogBook/360 combines the features and capabilities of LogBook/300 with a DBK60 expansion chassis. The lower
portion of the front panel has three male DB37 connectors (P1, P2, and P3) for system expansion, and a post forconnecting to CHASSIS ground. The upper section is nearly identical to LogBook/300.

LogBook/360, Front Panel

LogBook/360 Quick Start Guide

QS360-1

Reference Notes:
Depending on the nature of your LogBook system, you may find one or more of the following references to the
LogBook User’s Manual PDF helpful. The PDF manual is included on the Data Acquisition CD and is alsoavailable on our web-site.
ooooFor block diagrams and operational overviews, refer to Chapter 1 of the LogBook User’s Manual PDF.
For system expansion and calculating system power, refer to Ch. 4 and to the DBK Basics section.
For information regarding LBK options, refer to Chapter 5.
For specific DBK card information, refer to the DBK Options Manual PDF.
Note: The LogBook/360 rear panel supports three termination panels. Many different combinations are
possible.

WARNING
Electric shock hazard. Turn off power to all system-connected devices prior to connecting or
disconnecting cables, or setting hardware configurations. Failure to do so could result in electric shockor death, and equipment damage, even under low-voltage conditions.

CAUTIONUse ESD tools, containers, and procedures during setup of DBK cards. Electrostatic
discharge can damage some components. To prevent pin damage, align DBK cards with the
backplane DB37 connectors, then gently press them together.
When using LogBook in attached mode, the PC-Card [in LogBook] must already have the filelogbook.sys. Otherwise, LogView cannot communicate with LogBook, and LogBook willappear dead.

QS360-2

LogBook/360 Quick Start Guide

Setting Up the Card Drawer
LogBook/360 can house three DBK cards internally, and make use of various termination panels. For user convenience, a card
drawer can be slid free of the device. The following steps should be used when adding, removing, or changing cards. Refer tothe following figure as needed.
1 – Turn off system power and disconnect LogBook/360.
Turn power off to the LogBook/360 and all connected devices. Disconnect LogBook/360 from the system.
2 – Remove top cover.
If you need to make any change on the LogBook motherboard, you will need to remove the top cover. Otherwise, the cover
can remain in place. To remove the top cover, simply remove the two top cover screws and slide the cover free of thedevice.
3 – Remove card drawer.
A.
Remove the two screws that hold the card drawer to the chassis.
B.
Loosen the three termination panel thumbscrews.
C.
Carefully pull the card drawer free of the chassis.
4 – Remove termination panels.
For each termination panel, remove the two screws that mount it to the card drawer, then remove the termination panel.
5 – Determine power requirements.
Depending on the power needs of your system’s DBK cards, you may need to add a power card.
Refer to the DBK Basics section of the LogBook User’s Manual PDF in regard to calculating yoursystem’s power requirement.
If the required power is more than the available power your system will require auxiliary power. One of two power
supply cards can be used with LogBook/360:
DBK32A – This DBK supplies ±15 V for use with a LogBook, DaqBook, or DaqBoard.
DBK33 – This DBK supplies +5 V and ±15 Vfor use with Log Book, DaqBook, DaqBoard,or Daq PC-Card.

Leave jumpers JP1 and JP2 in place unless a DBK33 is being used. If a DBK33 is being
used configure the P1 interconnect board according to the table presented in step 6.

LogBook/360 Quick Start Guide

QS360-3

6 - FOR DBK33 USERS ONLY! Configure the P1 interconnect board for power source.
This step only applies when a DBK33 is being used with the system.
If you are not using a DBK33 skip to step 7.
The schematic to the right shows the LogBook/360 jumpers, JP1 andJP2, which are used to distribute the LogBook’s +5V power supply.
DBK cards that attach to the LogBook’s P1 bus need +5V, +15V, and-15V power sources to operate. Typically, the LogBook’s power supply
meets this requirement, but if additional power is needed, a DBK32A or
DBK33 must be added.
The LogBook’s power supply employs diode protection on its +/-15V
supplies so that no conflict will take place when a DBK32A or DBK33 isadded to the P1 bus. However, adding a DBK33, with its +5V source,
necessitates the proper positioning of the JP1 and JP2 jumpers.
Adding a DBK33 to your system with the jumpers set incorrectly candamage your LogBook, DBK33, and other system components. Refer tothe following table for important information regarding configuration.

CAUTION
Adding a DBK33, with its +5V source, necessitates the proper positioning of the JP1 and JP2
jumpers on the LogBook/360 P1 interconnect board. Adding a DBK33 to your system with thejumpers set incorrectly can damage the LogBook, DBK33, and other system components.
WHEN JP1 AND JP2 JUMPERS ARE IN PLACE A DBK33 CANNOT BE USED IN THE
SYSTEM!

P1 Interconnect Board Configurations for DBK33 Applications
LogBook +5V Power Source
Internal P1
External P1
Bus
Bus

Notes Pertaining to DBK33

Out

Yes

No

When the JP1 jumper is in place and the JP2
jumper is out, if the system is using external
DBKs the DBK33 will need to be external.
Internal DBKs will be powered from
LogBook’s internal P1 bus.

Out

In

No

No

When the JP1 jumper is out and the JP2
jumper is in place, if DBKs are usedinternally or externally, the DBK33 can beinternal or external and will power bothinternal and external DBK cards.

Out

Out

No

No

When the JP1 and JP2 jumpers are both out,
the following apply:

JP1

JP2

In

P1 Interconnect
Board*o If using internal DBKs, you will need touse an internal DBK33.
o If using external DBKs, you will need touse an external DBK33.
* Refer to the LogBook/360 Hardware Setup figure on page QS360-3 for the P1 interconnect board’s location.
Leave jumpers JP1 and JP2 in place, unless a DBK33 is being used. If a DBK33 is being
used configure the P1 interconnect board according to the above table.

QS360-4

LogBook/360 Quick Start Guide7 – Install power card if necessary.
If you determined in step 5 that additional power was needed, add a DBK32A or DBK33 power card to thechassis. The DBK Option Cards & Modules User’s Manual includes detailed information regarding thesepower-related cards.
A. Carefully align the power card’s DB37 connector with a DB37 connector on the interconnect board andgently press them together.
B. Mount the power card with two screws into the standoffs on the card drawer.
8 – Configure DBK cards.
Configure unique channel addresses with the jumpers on the DBK cards. Some cards have other jumpers and/or
DIP switches. Refer to the specific DBK sections of the DBK Options Manual (p/n 457-0905) as needed.
9 – Install DBK cards.
You must use all analog DBK cards in the LogBook/360; unless you have a factory modification that allows theuse of all digital cards. You can not use both analog and digital cards at the same time.
A. Carefully align the DBK card’s DB37 connector with a DB37 connector on the interconnect board andgently press them together (see figure).
B. Mount the DBK card with two screws into the standoffs on the card drawer (see previous figure).
C. Continue installation of any remaining DBK cards.
10 – Connect internal signals.
Connect signal inputs from DBK cards to termination panels. DBK cards connect to the termination panels invarious ways (see figure and particular DBK sections in the DBK Option Cards and Modules User’s Manual):
Single-ended connections use analog common.
Differential connections require the proper polarity, typically red-to-red for high (+)
and black-to-black for low (-).
For thermocouples, red is generally the low side. Always make sure the T/C connector and wire typematch the T/C type used.

11 – Install termination panels.
Mount the termination panels to the card drawer with two screws for each panel.
12 – Install card drawer.
The card drawer slides into the bottom track of the chassis.
A. Hold the card drawer by its handle and tilt it up slightly. Place it on the bottom track of the chassis.
B. Carefully slide the card drawer into the chassis. When it engages the bottom track, level the card drawer andcontinue inserting it until it engages with the P1 interconnect board.
C. Tighten the three captive thumbscrews holding the termination panels to the chassis.
D. Install the two screws holding the card drawer to the chassis.
13 – Connect external signals.
Connect signal inputs from sensors to termination panels.
14 – Install top cover.
If the top cover was removed, slide it back into place and secure with two screws.

LogBook/360 Quick Start Guide

QS360-5

Connecting the LogBook/360 to the PC, External DBKs, and to PowerThe following hardware-connection figure and procedure are generic; details vary with system complexity. For“unattached mode,” a pre-configured PC-Card is inserted in the PC-Card slot, and no connection to a PC is made.
The following figure illustrates the “attached” mode.
Note: Connecting LogBook/360 to a PC applies to the “attached mode” only. Many applications will make use of
three internal DBK cards only, having no need to attach external DBK cards or modules as discussed in thefollowing text.
Step 4

Connect the LogBook’s DIN5 power in connector to the AC adapter
and connect the adapter to a suitable power outlet. Note that otherpower options are discussed in the DBK Basics section of thismanual.

LogBook/360 System, “Attached Mode,” Basic Connections
Note: Rear panel connections may be made via terminal blocks, as discussed in the previous section,
Card Drawer Setup.

After verifying that all equipment power is off, hardware connection typically proceeds as follows.
Refer to the above figure as needed.
1.

QS360-6

Connect the LogBook/360 to the host PC. There are four ways for LogBook/360 to communicate
with the PC. These are: parallel port, serial port to serial port, serial port to USB, and manualtransportation of the PC-Card. Note that the parallel port method is represented in the previous
figure.
a)Parallel port to parallel port – If using the parallel port, connect a DB25male to DB25female parallel
cable [e.g., CA-35-2 or CA-35-6 (2ft. or 6ft. respectively)] to PARALLEL PORT on LogBook/360, and tothe corresponding parallel port on the host PC. When this method is used, the PC must be set to the ECP
mode. See ECP Parallel Port, page QS360-9 for additional information.

b)

Serial port to serial port – If using the serial port, use a null modem serial cable (CA-47) to connect the
DB9 SERIAL PORT on LogBook/360 to the corresponding DB9 serial port on the host PC.

LogBook/360 Quick Start Guidec)

Serial port to USB port - Requires an adapter kit. To connect the LogBook to a USB port you will need a
RS-232 to USB Interface Adapter Kit p/n CA-232-USB-KIT (available from MCC). Refer to theassociated Quick Start document (QS RS-232_to_USB_Interface.pdf) for a list of the kit’s contents and for
RS-232 to USB conversion instructions. The pdf is included on Data Acquisition CD (rev 8.1.2 andhigher).

d)

PC-Card – With PC-Card communication, LogBook/360 does not require a connection to the computer, as
communication is accomplished via the PC-Card. To provide the PC-Card with the correct configuration
file, it must be configured from the PC, through LogView. After the PC-Card is configured, it is inserted
into LogBook’s PC-Card slot, located behind the front panel door.
Reference Note:
Information pertaining to PC-Cards can be found in chapter 1 and in the LogView section of the
LogBook User’s Manual PDF.

2.

Connect LogBook/360 to the DBK cards and modules. For connecting internal DBK cards, refer tothe earlier section entitled, Setting Up the Card Drawer.
Most analog DBKs connect to P1; digital DBKs generally connect to P2. Refer to the DBK Options
Manual (457-0905) in regard to your particular DBKs and for general DBK installation details.
The CA-37-x cable can daisy-chain several DBKs including the DBK41, which has a built-in
P1 bus connection for 10 DBK cards. The x in the cable part number refers to the number of devices
that can be connected (a CA-37-1 actually has two DB-37 connectors).
Note: Chapter 4 of the LogBook User’s Manual PDF includes pinouts for LogBook/360’s P1, P2,
and P3 connectors.

CAUTION
For analog signal inputs via P1, do not exceed -35 VDC or +45 VDC.
Exceeding these limits could result in equipment damage.
3.

Connect DBK(s) to transducer(s). Follow instructions for particular DBK as described in the
DBK Option Cards & Modules User’s Manual and for the particular transducer. Some DBKs canaccommodate both BNC and screw-terminal connections.

4.

Connect LogBook/360 to a suitable power source, such as
the included AC-to-DC adapter or optional DBK34A
UPS / Battery Module. DC power sources such as a carbatteries must supply 10 to 45 VDC and use the correct
DIN5 pinout (see figure). A locking DIN5 connector
assures a secure power connection for applications subject
to vibration and thermal stress.

DIN5 Power Input Connector(As seen on LogBook/360 Front Panel)

Hardware Configuration
Reference Notes:
Refer to the device-specific sections of the LBK options chapter [in the user’s manual PDF]
and the DBK Option Cards & Modules User’s Manual PDF for information regarding thesedevices. Note that certain DBK options require manual configuration.
LogBook/360's top cover does not need to be removed, except to add or remove an LBK option, or to
replace the fuse.
Most LogBook/360 configuration is done via software as described in section, LogBook/360 Device
Configuration. LogBook/360 configuration does not require the setting of jumpers or switches, unlessthe RS-485 communication option is being used.

LogBook/360 Quick Start Guide

QS360-7

LogBook/360 Device Configuration
A configuration utility is supplied via a control panel applet. The LogBook Configuration applet allows
you to add a device, remove a device, or change existing configuration settings. From this same window,
you can also access a built-in utility to test the connected device for current setup and performance.
LogBook Configuration can be found in the Windows Control Panel. The Control Panel can be navigatedto from Window’s desktop Start button as follows:
Start

Settings

Control Panel

You can enter LogBook Configuration during driver installation or whenever you wish to add, remove orchange device configuration settings. The following description applies to either method.
The first configuration window will display configured devices in the Device Inventory field based on theport they’re connected to. Devices are represented by an icon, and text, e.g., LogBook (LPT1), as can beseen in the following figure. If no devices are configured, the device inventory field will remain blank. The
figure shows the first and second configuration windows overlapped.

LogBook Configuration Windows
The four buttons across the bottom of the first configuration window (previous figure) are used as follows:
Properties. Configuration settings for a device can be changed or modified from the corresponding properties
window. To do so, double-click the device icon or single-click the device and then single-click the Properties
button. The second configuration window will appear for the selected device as shown in the previous figure.
Add Device. The Add Device button is used to add a device configuration whenever a new device is added tothe system. LogView cannot recognize a device unless listed in the configuration window.
Remove. The Remove button is used to remove a device from the configuration. A device may be removed if it
is no longer installed or if the device’s configuration no longer applies.
Close. The Close button may be used at any time to exit the LogBook Configuration applet.

The second configuration window displays the properties for the selected LogBook. Fields include:
Device Name is displayed with the default name, numbered successively as configured. This field can bechanged to any descriptive name as desired.
Connection Type can be serial or parallel port.
Device Connection specifies the port name.
Protocol is used to set the parallel port protocol (ECP only) or serial protocol (RS-232 or RS-485).
Device Timeout specifies the number of seconds LogView will be wait for a LogBook response before
displaying an error condition.
QS360-8

LogBook/360 Quick Start Guide

ECP Parallel Port
To use parallel port communication with an attached LogBook/360, your PC must
support the ECP protocol AND be set in the ECP mode.
PCs made since 1994 probably support the Enhanced Computer Port protocol (ECP). If your parallel port
does not support ECP, you can communicate with the LogBook/360 via the RS-232 serial port, or you can
add an ECP-compatible ISA board or PC-Card parallel port. Setting the PC to ECP mode varies withdifferent computers. On some computers, you can enter the BIOS Setup utility from Windows Settings orduring startup by pressing the F1 function key. The Parallel Port Mode property can be found under the
Peripheral Configuration group menu item. If necessary, consult your PC’s documentation or your PC’s
manufacturer.
To ensure ECP compatibility after proper setup, use the Test Hardware utility(described on page QS360-10). Before testing, make sure LogBook/360 is properly
connected, powered on, and that the Parallel Port Mode is set to ECP (in BIOS Setup).

CAUTION
Making errors in BIOS Setup can disrupt your system’s operation. If test hardware
indicates a problem and you have inadequate experience with the BIOS Setup utility,
consult your System Administrator or other qualified individual.

Serial Port
If the selected device is connected to a serial port the properties window will include the fields shown in the
figure at right. Baud rate can be set from 1200 to 115200 bits per second (default 9600). When all fields
have been changed to the desired settings, you can click on one of the following options:
Apply to store the device configuration. Parameters are not locked in until you click the Apply
button. If you make changes and don’t click Apply, clicking the Test button in Test Hardware will
yield unexpected errors.
OK to store the configuration and exit the current property screen.
Cancel to exit the current screen without storing any changes.
Test Hardware to test the current device.

LogBook Properties Tab

LogBook/360 Quick Start Guide

Test Hardware Tab

QS360-9Testing the Hardware
Before testing LogBook/360:a) Verify the device has been properly installed.
b) Make sure the communication cable (serial or parallel) is firmly in place to the proper ports.
c) Verify the device is powered-on.
Testing the LogBook/360 device might cause the system to hang. If test results are notdisplayed within 30 seconds, or if the system does not respond properly do the following:
Reboot the system.
Upon power-up, re-enter the LogBook Configuration.
Ensure the configuration settings are correct. Change the settings as applicable.
To begin the test, click the Test button. Test results should be displayed within a few seconds.
Test results indicate if the device is online (properly connected, powered on and ready to transfer the data)
or offline. If the device is online, Performance Test will display Download and Upload speed rates. These
rates represent the maximum speed at which downloading and uploading files can be performed. Actual
transfer time will depend on channel configuration and the size of the transfer.

Acquisition Configuration
An acquisition is configured using LogView on a PC and then stored as an acquisition setup file on a
PC-Card. The PC-Card may be in an attached LogBook/360 or in the PC to be later manually transferred toan unattached LogBook/360. The system’s DBK cards are listed; the scan sequence is defined; the trigger
conditions are specified, etc.
Reference Note:
Configuring the acquisition is described in the LogView section of the user’s manual PDF.

QS360-10

LogBook/360 Quick Start Guide

A Note about Calibration
Calibration is typically performed automatically through LogView software; however, some DBKs may
require manual calibration. LogView’s 2-point calibration fine-tunes the reading’s slope and offset error(mx+b). DBKs working with non-linear sensors typically condition/convert the reading to a linear form.
Otherwise, a non-linear analog input signal is difficult to read accurately. Careful use of the calculatedchannels may yield usable approximations in simple, limited-range conditions.

Reference Notes:
An example of 2-point calibration is provided under the Procedures heading in the
LogView section of the user’s manual PDF.
For information on calibrating the DBK16 and the DBK43A, refer to the
DBK Option Cards and Modules User’s Manual PDF.

Measurement Computing
10 Commerce Way
Norton, MA 02766(508) 946-5100
Fax: (508) 946-9500
info@mccdaq.com

www.mccdaq.com

LogBook/360 Quick Start Guide

QS360-11