PCIe-DIO96H
High-Density, Logic-Level, Digital I/O

User's Guide

Document Revision 2A
September 2014
© Copyright 2014

Trademark and Copyright Information
Measurement Computing Corporation, InstaCal, Universal Library, and the Measurement Computing logo areeither trademarks or registered trademarks of Measurement Computing Corporation. Refer to the Copyrights &Trademarks section on mccdaq.com/legal for more information about Measurement Computing trademarks.
Other product and company names mentioned herein are trademarks or trade names of their respective
companies.
© 2014 Measurement Computing Corporation. All rights reserved. No part of this publication may be
reproduced, stored in a retrieval system, or transmitted, in any form by any means, electronic, mechanical, by
photocopying, recording, or otherwise without the prior written permission of Measurement Computing
Corporation.
NoticeMeasurement Computing Corporation does not authorize any Measurement Computing Corporation product for
use in life support systems and/or devices without prior written consent from Measurement Computing
Corporation. Life support devices/systems are devices or systems that, a) are intended for surgical implantation
into the body, or b) support or sustain life and whose failure to perform can be reasonably expected to result ininjury. Measurement Computing Corporation products are not designed with the components required, and are
not subject to the testing required to ensure a level of reliability suitable for the treatment and diagnosis ofpeople.

HM PCIe-DIO96H.docx

Table of Contents
Preface

About this User's Guide ....................................................................................................................... 4
Conventions in this user's guide ......................................................................................................................... 4
Where to find more information ......................................................................................................................... 4
Chapter 1

Introducing the PCIe-DIO96H ............................................................................................................... 5
Overview: PCIe-DIO96H features...................................................................................................................... 5
Functional block diagram ................................................................................................................................... 6
Chapter 2

Installing the PCIe-DIO96H ................................................................................................................... 7
What comes with your shipment? ....................................................................................................................... 7
Hardware .......................................................................................................................................................................... 7
Software ............................................................................................................................................................................ 7
Documentation .................................................................................................................................................................. 7
Optional components ........................................................................................................................................................ 7

Unpacking the PCIe-DIO96H............................................................................................................................. 7
Installing the software ........................................................................................................................................ 7
Installing the hardware ....................................................................................................................................... 8Signal connections .............................................................................................................................................. 9Connector pinout..............................................................................................................................................................10
Cabling.............................................................................................................................................................................11
Field wiring and signal termination .................................................................................................................................12
Chapter 3

Functional Details ...............................................................................................................................13
CIO-ERB24 and SSR-RACK24 daisy chain configuration .............................................................................. 13
82C55 emulation (mode 0) ............................................................................................................................... 13
Replacing a fuse................................................................................................................................................ 14
Fuse specifications ...........................................................................................................................................................14
Chapter 4

Specifications ......................................................................................................................................15
Digital input/output........................................................................................................................................... 15Power consumption .......................................................................................................................................... 15
Environmental .................................................................................................................................................. 15
Mechanical ....................................................................................................................................................... 15
Bus .................................................................................................................................................................... 16
Signal connector ............................................................................................................................................... 16
Declaration of Conformity ..................................................................................................................18

3

Preface

About this User's Guide
This user's guide describes the Measurement Computing PCIe-DIO96H data acquisition device and lists device
specifications.

Conventions in this user's guide
For more information
Text presented in a box signifies additional information related to the subject matter.
Caution! Shaded caution statements present information to help you avoid injuring yourself and others,
damaging your hardware, or losing your data.
bold text

Bold text is used for the names of objects on a screen, such as buttons, text boxes, and check boxes.

italic text

Italic text is used for the names of manuals and help topic titles, and to emphasize a word or phrase.

Where to find more information
Additional information about PCIe-DIO96H hardware is available on our website at www.mccdaq.com. You
can also contact Measurement Computing Corporation with specific questions.


Knowledgebase: kb.mccdaq.com
Tech support form: www.mccdaq.com/support/support_form.aspx
Email: techsupport@mccdaq.com
Phone: 508-946-5100 and follow the instructions for reaching Tech Support

If you need to program at the register level in your application, refer to the Register Map for the PCIe-DIO96H.
This document is available on our website at www.mccdaq.com/registermaps/RegMapPCIe-DIO96H.pdf.

4Chapter 1

Introducing the PCIe-DIO96H
Overview: PCIe-DIO96H features
This manual explains how to install and use the PCIe-DIO96H board. The PCIe-DIO96H is a high-density,
logic-level digital I/O board designed for PCI Express (PCIe) bus.
The PCIe-DIO96H is fully compatible with software written for the PCI-DIO96H and USB-DIO96H.
The PCIe-DIO96H provides 96 digital I/O lines in four independent port groups. Each digital port group isdivided into two 8-bit ports and two 4-bit ports. The PCIe-DIO96H emulates an 82C55 chip in Mode zero. You
can configure each port independently for either input or output.
The PCIe-DIO96H outputs are high-drive TTL that can source 15 mA and sink 64 mA. Additional buffering istypically not required to drive external devices. The board has two individual slow blow fuses rated at 1 ampthat provide protection for the +5 V User Outputs. A spare fuse is provided.
Each digital port has an associated 10 k resistor network. The board is shipped with each port configured in thepull-up state. You can change the port configuration with software. On power up and reset the configuration ofeach port is read from EEPROM.
Digital I/O lines are accessed through a 100-pin high-density connector.
The PCIe-DIO96H board is completely plug-and-play, with no jumpers or switches to set. All board addresses
are set by the board's plug-and-play software. Board configuration is controlled by your system's BIOS.
Power is provided by the computer’s internal 5 V Molex connector.

5

PCIe-DIO96H User's Guide

Introducing the PCIe-DIO96H

Functional block diagram
PCIe-DIO96H functions are illustrated in the block diagram shown here.
5V Power
Molex
Connector+5V (VDD)

+5V

Fuse
Protection

+5V

Fuse
Protection

To +5V components(Note: Outputs to CPLD
are 3.3V with 5V inputs)

1.5V
Regulator

FOURTHPORTA
24

ControlHigh Drive FOURTHPORTFOURTHPORTBFOURTHPORTCL In
FOURTHPORTCH Out

3.3V

3.3V Board Power

3.3V

BP1
8
8

THIRDPORTB
24

THIRDPORTCL
THIRDPORTCH

Control

THIRDPORTA

PCIe-to-PCI
Bridge

BP2

In
Out

8

5V Detect
Software Sensing

8

SECONDPORTA
24

ControlHigh Drive SECONDPORTSECONDPORT B
SECONDPORT CL In
SECONDPORT CH Out

Control
Bus
BP3
8
8

High Drive FIRSTPORTFIRSTPORTAFIRSTPORTB
24

10 k W
Pull Resistor
Network

FIRSTPORTCLFIRSTPORTCH

12

Note: Port CL and CH
are tied together on
the pull resistors.

EEPROM
SwitchControl

100-pin I/O Connector

High Drive THIRDPORT

In
OutComplex,
Programmable
Logic Device(CPLD)
Logic, Control,Interface
Decode/Status

BP4

PCI-to-Local
Bus

96
8

LAD(0:7)
8

5V (VDD)
8

GND

Figure 1. PCIe-DIO96H functional block diagram

6

PCI Bus

3.3V, 32-bit, 33 MHz

Local
Address&
Data Bus

BP(1:4)8

8

Bus
Timing

BADR2
+
BADR3

Boot
EEPROM

x1 Link

PCI Express Edge Connector

Spare
Fuse

Chapter 2

Installing the PCIe-DIO96H
What comes with your shipment?
The following items are shipped with the PCIe-DIO96H:

Hardware

PCIe-DIO96H

Software

MCC DAQ CD

Documentation

MCC DAQ Quick Start
This booklet provides an overview of the MCC DAQ software you received with the device, and includes
information about installing the software. Please read this booklet completely before installing any softwareor hardware.

Optional components


C100FF-x cable
Signal termination and conditioning accessories
Measurement Computing provides signal termination products for use with the PCIe-DIO96H. Refer to thesection "Field wiring and signal termination" on page 12 for compatible accessory products.

Unpacking the PCIe-DIO96H
As with any electronic device, you should take care while handling to avoid damage from static
electricity. Before removing the PCIe-DIO96H from its packaging, ground yourself using a wrist strap or bysimply touching the computer chassis or other grounded object to eliminate any stored static charge.
If any components are missing or damaged, contact us immediately using one of the following methods:

Knowledgebase: kb.mccdaq.com
Tech support form: www.mccdaq.com/support/support_form.aspx
Email: techsupport@mccdaq.com
Phone: 508-946-5100 and follow the instructions for reaching Tech Support

For international customers, contact your local distributor. Refer to the International Distributors section on our
website at www.mccdaq.com/International.

Installing the software
Refer to the MCC DAQ Quick Start for instructions on installing the software on the MCC DAQ CD. Thisbooklet is available in PDF at www.mccdaq.com/PDFmanuals/DAQ-Software-Quick-Start.pdf.

7

PCIe-DIO96H User's Guide

Installing the PCIe-DIO96H

Installing the hardware
The PCIe-DIO96H is completely plug-and-play. There are no switches or jumpers to set on the board.
Configuration is controlled by your system's BIOS. Perform the steps to install your board..
Install the MCC DAQ software before you install your board
The driver needed to run your board is installed with the MCC DAQ software. Therefore, you need to install the
MCC DAQ software before you install your board. Refer to the Quick Start Guide for instructions on installingthe software.
1.

Power off and unplug the computer, and remove the cover to expose the expansion slots.

2.

Touch any metal part of the computer to discharge static electricity that may be present. Static electricity
can damage the board.

3.

Insert the PCIe-DIO96H into an unused x1 PCIe expansion slot.
The PCIe-DIO96H is designed to install into an x1 slot. However, you can also install the board into anunused x4, x8, or x16 PCIe slot.

Caution! Ensure that you install the board into a PCI Express slot. Installing the PCIe-DIO96H into a
non-PCIe slot can damage both the board and the computer’s motherboard.
4.

Connect the board’s external power connector to one of the computer’s four-pin Molex power connectors.
The board’s Molex power connections are shown here.

No Connect

PC Ground

PC GroundPC +5V

Caution! Using an external power supply is not recommended. If external power is used, ensure that thepower supply ground or common is at the same ground potential as the computer power supply.
5.

Close your computer and turn it on.
A dialog box opens as the system loads, indicating that new hardware has been detected. The information
file for this board should have already been loaded onto your PC when you installed the Measurement
Computing Data Acquisition Software CD supplied with your board, and should be detected automaticallyby Windows. If you have not installed this software, cancel the dialog and install it now.

6.

Run InstaCal to test your installation and to configure the pull direction of the digital port resistors.
Refer to the Quick Start Guide that came with your board for information on how to initially set up
InstaCal.

8

PCIe-DIO96H User's Guide

Installing the PCIe-DIO96H

Signal connections
The table below lists the board I/O connector, applicable cables and compatible accessory boards.
Board connectors, cables, accessory equipment
Connector typeCompatible cableCompatible accessory products with
the C100FF-x cable

100-pin, high-density keyed connector
C100FF-x (Figure 2)
CIO-MINI50*
CIO-ERB24
CIO-SPADE50*
CIO-SERB24/FD
CIO-TERM100
CIO-ERB48
SCB-50
CIO-SERB48* two devices are required
SSR-RACK24
SSR-RACK48

The PCIe-DIO96H has a 100-pin, high-density Robinson-Nugent male connector. You can use the C100FF-x
cable to split the 100 I/O lines into two 50-pin cables (see Figure 2 on page 11).
Board connector pins 1 to 50 are mapped directly to pins 1 to 50 on the C100FF-x cable's first 50-pin connector.
Pins 51 to 100 are mapped directly to pins 1 to 50 on the second 50-pin connector (pin 51 is mapped to pin 1,
and pin 100 is mapped to pin 50.) A sample C100FF-x cable configuration is shown in Figure 3 on page 11.
Information on signal connections
General information regarding signal connection and configuration is available in the Guide to DAQ Signal
Connections. This document is available on our website at www.mccdaq.com/signals/signals.pdf.
Caution! When connecting a cable to the board's I/O connector, make sure that the arrow indicating pin 1
on the board connector lines up with the arrow indicating pin 1 on the cable connector. Incorrectly
connected cables can damage the board and the I/O controller.

9

PCIe-DIO96H User's Guide

Installing the PCIe-DIO96H

Connector pinout
Main I/O connector pinout
Signal name

Pin

GND
100
+5V User Output*
99
THIRDPORTC Bit 0
98
THIRDPORTC Bit 1
97
THIRDPORTC Bit 2
96
THIRDPORTC Bit 3
95
THIRDPORTC Bit 4
94
THIRDPORTC Bit 5
93
THIRDPORTC Bit 6
92
THIRDPORTC Bit 7
91
THIRDPORTB Bit 0
90
THIRDPORTB Bit 1
89
THIRDPORTB Bit 2
88
THIRDPORTB Bit 3
87
THIRDPORTB Bit 4
86
THIRDPORTB Bit 5
85
THIRDPORTB Bit 6
84
THIRDPORTB Bit 7
83
THIRDPORTA Bit 0
82
THIRDPORTA Bit 1
81
THIRDPORTA Bit 2
80
THIRDPORTA Bit 3
79
THIRDPORTA Bit 4
78
THIRDPORTA Bit 5
77
THIRDPORTA Bit 6
76
THIRDPORTA Bit 7
75
FOURTHPORTC Bit 0
74
FOURTHPORTC Bit 1
73
FOURTHPORTC Bit 2
72
FOURTHPORTC Bit 3
71
FOURTHPORTC Bit 4
70
FOURTHPORTC Bit 5
69
FOURTHPORTC Bit 6
68
FOURTHPORTC Bit 7
67
FOURTHPORTB Bit 0
66
FOURTHPORTB Bit 1
65
FOURTHPORTB Bit 2
64
FOURTHPORTB Bit 3
63
FOURTHPORTB Bit 4
62
FOURTHPORTB Bit 5
61
FOURTHPORTB Bit 6
60
FOURTHPORTB Bit 7
59
FOURTHPORTA Bit 0
58
FOURTHPORTA Bit 1
57
FOURTHPORTA Bit 2
56
FOURTHPORTA Bit 3
55
FOURTHPORTA Bit 4
54
FOURTHPORTA Bit 5
53
FOURTHPORTA Bit 6
52
FOURTHPORTA Bit 7
51
PCIe slot ↓




















































Pin

Signal name

50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

GND
+5V User Output*
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
SECONDPORTC Bit 0
SECONDPORTC Bit 1
SECONDPORTC Bit 2
SECONDPORTC Bit 3
SECONDPORTC Bit 4
SECONDPORTC Bit 5
SECONDPORTC Bit 6
SECONDPORTC Bit 7
SECONDPORTB Bit 0
SECONDPORTB Bit 1
SECONDPORTB Bit 2
SECONDPORTB Bit 3
SECONDPORTB Bit 4
SECONDPORTB Bit 5
SECONDPORTB Bit 6
SECONDPORTB Bit 7
SECONDPORTA Bit 0
SECONDPORTA Bit 1
SECONDPORTA Bit 2
SECONDPORTA Bit 3
SECONDPORTA Bit 4
SECONDPORTA Bit 5
SECONDPORTA Bit 6
SECONDPORTA Bit 7* The board has two individual slow blow fuses rated at 1 A. One fuse protects the +5V User Output at pin 49,
and one fuse protects the +5V User Output at pin 99.

10

PCIe-DIO96H User's Guide

Installing the PCIe-DIO96H

Cabling
Cable is labeled“Pins 1-50”.

50

49

50

1

2

Key

100

The red stripe
identifies pin # 1

99 100

1

51

Key

51 52

Cable is labeled“Pins 51-100”.

The red stripe
identifies pin # 51

Figure 2. C100FF-x cable

Caution! When connecting the cable to the board's I/O connector, make sure that the arrow indicating pin 1
on the board connector lines up with the arrow indicating pin 1 on the cable connector. Incorrectly
connected cables can damage the board and the I/O controller.
100-pin
I/O connector

C100FF-x cable

Digital I/O
pins 1 to 50
IN

Digital signal conditioning or
50-pin screw terminl board
Digital I/O
pins 51 to 100
IN
Digital signal conditioning or
50-pin screw terminl board
Figure 3. C100FF-x cable configuration

11

PCIe-DIO96H User's Guide

Installing the PCIe-DIO96H

Field wiring and signal termination
You can use the following screw terminal boards to terminate field signals and route them into the
PCIe-DIO96H using the C100FF-x cable.











CIO-MINI50 – 50-pin screw terminal board.
CIO-TERM100 – 100-pin screw terminal board (two 50-pin IDC connectors).
CIO-SPADE50 — 16" × 4" termination panel which mates with both 37-pin and 50-pin connectors.
SCB-50 – 50 conductor, shielded signal connection/screw terminal box provides two independent 50-pin
connections.
CIO-ERB24 – 24 Form C relays, 6 A relay accessory board for digital signal conditioning.
CIO-SERB24/FD – 24 Form C relays, 10 A, fault detecting relay accessory board with socketed and fieldreplaceable relays.
CIO-ERB48 – 48 Form C relays, 6 A, relay, 50-pin accessory board for digital signal conditioning.
CIO-SERB48 – 48 Form C relays, 10 A relay accessory board with socketed and field-replaceable relays.
SSR-RACK24 – 24-channel, solid-state relay mounting rack for digital signal conditioning.
SSR-RACK48 – 48-channel, solid-state relay mounting rack with quad-format modules.

Details on these products are available on our website.
For additional information about digital interfacing
Detailed information regarding digital interfacing is contained in MCC's Guide to DAQ Signal Connections.
This document is available on our website at www.measurementcomputing.com/signals/signals.pdf.

12Chapter 3

Functional Details
CIO-ERB24 and SSR-RACK24 daisy chain configuration
Many relay and solid-state relay (SSR) racks provide only 24-bits of digital I/O. You can configure the CIOERB24 relay output board and SSR-RACK24 I/O module rack in a daisy chain configuration to use all of thedigital I/O bits provided by the PCIe-DIO96H board. An example of the daisy chain configuration scheme foreach board is shown below.

PCIe-DIO96H

The PCIe-DIO96H board provides digital I/O in a group of 96 bits. Each of the C100FF-x cable's 50-pin
connectors provides 48 bits. To use all of the board's 96 digital I/O bits to control relays and/or SSRs, configure
the daisy chain as shown in Figure 4.

C100FF-x Cable

CIO-ERB24 IN
orOUT
SSR-Rack24

CIO-ERB24 IN
orOUT
SSR-Rack24
CIO-ERB24 IN
orOUT
SSR-Rack24

CIO-ERB24 IN
orOUT
SSR-Rack24

Figure 4. PCIe-DIO96H to C100FF-x to relay rack daisy chain cabling

The 24 digital I/O bits on pins 1-24 control the first relay board on the chain. The 24 digital I/O bits on pins
25-50 control the second relay/SSR board on the daisy chain and so on, for up to 100 pins.

82C55 emulation (mode 0)
The PCIe-DIO96H emulates the 82C55 chip (mode 0). The 82C55 emulation initializes all ports as inputs on
power-up and reset. A TTL input is a high impedance input. If you connect another TTL input device to the
output, it could be turned on or off every time the board is reset.
To establish a consistent TTL level at power-up, configure each port resistor with InstaCal for either pull-up or
pull-down.
Whenever an 82C55 emulation is powered on or reset, all pins are set to high-impedance input. Based onstandard TTL functionality, these inputs typically float high, and may have enough drive current to turn on
external devices. Consequently, if you have output devices such as solid state relays, they may be switched onwhenever the computer is powered on or reset. To prevent unwanted switching, and to drive all outputs to a
known state after power on or reset, configure each port resistor with InstaCal.
Unconnected inputs are forced to the pull direction
Unconnected inputs will float in the pull direction that is configured for the port with InstaCal (either up/high or
down/low).

13

PCIe-DIO96H User's Guide

Functional Details

Replacing a fuse
The PCIe-DIO96H has two individual 1 amp slow blow fuses. One fuse is connected to the +5V User Output at
pin 49, and is labeled F1 on the board. The second fuse is connected to the +5V User Output at pin 99, and islabeled F2 on the board. A spare fuse is installed on the board at location F3. All fuses are secured to the board
with clips for convenient replacement.
A fuse will blow during operation if amperage exceeds 1 amp. If you need to replace a fuse, perform thefollowing procedure.
1.

Pry the center of the fuse from the fuse holder clip.

2.

Insert the replacement fuse into the fuse holder clip.

Fuse specifications
Refer to the information below to purchase additional fuses, (or an equivalent), if required:






Manufacturer: LittelFuse®
Series: 452 Slo-Blo®
Part number: 0452001. (Include the period as part of the item number.)
1 amp, 125 volts, 0.225 Ω
http://www.littelfuse.com/part/0452001..html

14Chapter 4

Specifications
All specifications are subject to change without notice.
Typical for 25 °C unless otherwise specified.
Specifications in italic text are guaranteed by design.

Digital input/output
Table 1. Digital I/O specifications
Parameter

Specification

Digital typeOutputInput
Configuration
Number of I/O
Output highOutput low
Input highInput low
Power-up / reset state8255 emulation, Mode 0
74ABT244
74LV373A
8 banks of 8, 8 banks of 4, programmable by bank as input or output
96
2.0 volts min @ –15 mA
0.55 volts max @ 64 mA
2.0 volts min, 5.5 volts absolute max
0.8 volts max, –0.5 volts absolute min

Pull-up/pull-down resistors

Input mode (10 kW impedance from pull-up or pull-down)
EEPROM stored; software programmable driven by 74ACT244 through10 kWbussed resistor networks (shipped in the pull-up state)

Power consumption
Table 2. Power consumption specifications
Parameter

Specification+3.3 V operating (bus)+5 V operating (Molex)+5 V user output (Molex)+5 V user output Fuse

484 mA max (405 mA typ)1.74 A max (54 mA typ no load)
1 A max per +5 V user output (pins 49 and 99, protected with 1 A slow blow fuse)
1 A Littelfuse ® Slo-Blo ® Fuse P/N: 0452001 or equivalent

Environmental
Table 3. Environmental specifications
Parameter

Specification

Operating temperature range
Storage temperature range
Humidity

0 °C to 50 °C–35°C to 80 °C
0% to 90% non-condensing

Mechanical
Table 4. Mechanical specifications
Parameter

Specification

Dimensions (L × W × H)

167.6 × 106.6 × 14.48 mm (6.60 × 4.20 × 0.57 in.)15

PCIe-DIO96H User's Guide

Specifications

Bus
Table 5. Bus specifications
Parameter

Specification

Bus type
Bus width

PCI Express 1.0ax1 lane PCI Express

Signal connector
Table 6. Signal connector specifications
Parameter

Specification

Connector typeCompatible cables
Compatible accessory products

100-pin, high-density
C100FF-x
CIO-MINI50 (two devices are required)
CIO-TERM100
SCB-50CIO-ERB24
CIO-SERB24

16

CIO-ERB48
CIO-SERB48
SSR-RACK24
SSR-RACK48

PCIe-DIO96H User's Guide

Specifications

Table 7. Signal connector pinout
Pin
100
99
98
97
96
95
94
93
92
91
90
89
88
87
86
85
84
83
82
81
80
78
78
77
76
75
74
73
72
71
70
69
68
67
66
65
64
63
62
61
60
59
58
57
56
55
54
53
52
51

Signal name
GND
+5V User Output (Note 1)
THIRDPORTC Bit 0
THIRDPORTC Bit 1
THIRDPORTC Bit 2
THIRDPORTC Bit 3
THIRDPORTC Bit 4
THIRDPORTC Bit 5
THIRDPORTC Bit 6
THIRDPORTC Bit 7
THIRDPORTB Bit 0
THIRDPORTB Bit 1
THIRDPORTB Bit 2
THIRDPORTB Bit 3
THIRDPORTB Bit 4
THIRDPORTB Bit 5
THIRDPORTB Bit 6
THIRDPORTB Bit 7
THIRDPORTA Bit 0
THIRDPORTA Bit 1
THIRDPORTA Bit 2
THIRDPORTA Bit 3
THIRDPORTA Bit 4
THIRDPORTA Bit 5
THIRDPORTA Bit 6
THIRDPORTA Bit 7
FOURTHPORTC Bit 0
FOURTHPORTC Bit 1
FOURTHPORTC Bit 2
FOURTHPORTC Bit 3
FOURTHPORTC Bit 4
FOURTHPORTC Bit 5
FOURTHPORTC Bit 6
FOURTHPORTC Bit 7
FOURTHPORTB Bit 0
FOURTHPORTB Bit 1
FOURTHPORTB Bit 2
FOURTHPORTB Bit 3
FOURTHPORTB Bit 4
FOURTHPORTB Bit 5
FOURTHPORTB Bit 6
FOURTHPORTB Bit 7
FOURTHPORTA Bit 0
FOURTHPORTA Bit 1
FOURTHPORTA Bit 2
FOURTHPORTA Bit 3
FOURTHPORTA Bit 4
FOURTHPORTA Bit 5
FOURTHPORTA Bit 6
FOURTHPORTA Bit 7

Pin
50
49
48
47
46
45
44
43
42
41
40
39
38
37
36
35
34
33
32
31
30
29
28
27
26
25
24
23
22
21
20
19
18
17
16
15
14
13
12
11
10
9
8
7
6
5
4
3
2
1

Note 1: Protected by individual slow blow fuses rated at 1 A.

17

Signal name
GND
+5V User Output (Note 1)
FIRSTPORTC Bit 0
FIRSTPORTC Bit 1
FIRSTPORTC Bit 2
FIRSTPORTC Bit 3
FIRSTPORTC Bit 4
FIRSTPORTC Bit 5
FIRSTPORTC Bit 6
FIRSTPORTC Bit 7
FIRSTPORTB Bit 0
FIRSTPORTB Bit 1
FIRSTPORTB Bit 2
FIRSTPORTB Bit 3
FIRSTPORTB Bit 4
FIRSTPORTB Bit 5
FIRSTPORTB Bit 6
FIRSTPORTB Bit 7
FIRSTPORTA Bit 0
FIRSTPORTA Bit 1
FIRSTPORTA Bit 2
FIRSTPORTA Bit 3
FIRSTPORTA Bit 4
FIRSTPORTA Bit 5
FIRSTPORTA Bit 6
FIRSTPORTA Bit 7
SECONDPORTC Bit 0
SECONDPORTC Bit 1
SECONDPORTC Bit 2
SECONDPORTC Bit 3
SECONDPORTC Bit 4
SECONDPORTC Bit 5
SECONDPORTC Bit 6
SECONDPORTC Bit 7
SECONDPORTB Bit 0
SECONDPORTB Bit 1
SECONDPORTB Bit 2
SECONDPORTB Bit 3
SECONDPORTB Bit 4
SECONDPORTB Bit 5
SECONDPORTB Bit 6
SECONDPORTB Bit 7
SECONDPORTA Bit 0
SECONDPORTA Bit 1
SECONDPORTA Bit 2
SECONDPORTA Bit 3
SECONDPORTA Bit 4
SECONDPORTA Bit 5
SECONDPORTA Bit 6
SECONDPORTA Bit 7

Declaration of Conformity
According to ISO/IEC 17050-1:2010

Manufacturer:
Address:

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, MA 02766
USA

Product Category:

Electrical equipment for measurement, control and laboratory use.

Measurement Computing Corporation declares under sole responsibility that the product

PCIe-DIO96H
Complies with the essential requirements of the following applicable European Directives:
Electromagnetic Compatibility (EMC) Directive 2004/108/EC
Low Voltage Directive 2006/95/EC
RoHS Directive 2011/65/EU
Conformity is assessed in accordance to the following standards:
EMC:
Emissions: EN 61326-1:2006 (IEC 61326-1:2005), Class A
 EN 55011: 2007 (IEC CISPR 11:2003), Group 1, Class A
Immunity: EN 61326-1:2006 (IEC 61326-1:2005), Controlled EM Environments EN 61000-4-2:2001 (IEC 61000-4-2:2001)
 EN 61000-4-3 :2002 (IEC61000-4-3:2002)
 EN 61000-4-4 :2004 (IEC61000-4-4:2004)
 EN 61000-4-5 :2001 (IEC61000-4-5:2001)
 EN 61000-4-6 :2007 (IEC61000-4-6:2003)
 EN 61000-4-11:2004 (IEC61000-4-11:2004)
Safety: EN610101-1 (IEC61010-1)

Environmental Affairs:
Articles manufactured on or after the Date of Issue of this Declaration of Conformity do notcontain any of the restricted substances in concentrations/applications not permitted by the RoHS
Directive.

Carl Haapaoja, Director of Quality Assurance

Measurement Computing Corporation
10 Commerce Way
Suite 1008
Norton, Massachusetts 02766(508) 946-5100
Fax: (508) 946-9500
E-mail: info@mccdaq.com
www.mccdaq.com