Extreme highStability - using
ChoppedRadiation Method

Very highResolution 0.03°C

Fast Response
Times from 5 ms
on low
Temperatures

Wide Temperature
Ranges -50°C to3000°C

Features• Focus on
application – allspectral ranges
from freeze
drying to molt
• Stainless steel
cooling andprotecting
housing up to320°C• Fast target
measurement• Focus laser
marks field ofview• Very highlinearization
accuracy 0.02K• User interface
software forcompleteprogrammingandmeasurement• Wide range ofaccessories e.g.
• Vacuum tightlensesInfraredRadiation PyrometerNon Contact Temperature
Measurement in the mostAdvanced Technology

HEITRONICS
Infrarot Messtechnik

Infrared Radiation Pyrometer KT15 IIP Series at a Glance
Selection Guide
Application / Material

Model / Type

Temperature
Range / °C

Metal / Metal oxide / Ceramic
Metal / metal oxide / ceramic

KT15.01 IIP
KT15.02 IIP

250 - 2500
140 - 2500

Thin Plastic Film
… / oil / paintKT15.21 IIP
PE, PP, PVC, …
KT15.23 IIP
PET, PA, …
KT15.24 IIP
PTFE, PET, PVC, …
KT15.25 IIP
Natural material, Paint, Chemicals, Rubber …
Paper, textiles, ceramics,
asphalt, wood, electronic
components, building
materials, food, …
… better NETD

80 - 400
0 - 400
0 - 400
0 - 400

KT15.81 IIP

0 -1000

KT15.82 IIP-30 - 1000

KT15.83 IIP

0 - 500

Application / Material

Model / Type

Temperature
Range / °C

Incinerator
Burning gas temperature
KT15.69 IIP
Through flames and gas
KT15.41 IIPGlass / Quartz
Glass volume
KT15.01 IIPGlass volume
KT15.41 IIP
SurfaceKT15.42 IIPThin material / in furnace
KT15.43 IIP
Meteorological, biological measurement
Meteorology

0 - 1700
250 - 2500
250 - 2500
250 - 2500
100 - 2500
0 - 1400

KT15.85 IIP-25 - 200

Temperature of hot gases

KT15.6x IIP

400-2500

All materials

KT15.99 IIP-50 - 3000

Special applications

General Specifications
Temperature range
Temperature resolution (NETD)
Accuracy (uncertainty)
Long term stability
Field of view diameter
Field of view marking

Laser function
Spectral responses
Programmable Functions via serialinterface
Emissivity
Response time
Temperature unit
Analog output (Hardware)
Analog output (Functions)
Serial interface
Thermal switch
Power requirements
Permissible operating temperature
Storage temperature
Protective class, WeightHousing-50°C ... 3000°C, depends on model, see table above
Depends on model, measuring temperature and response time, typical value 0,06°C± 0.5°C plus 0.7% of the difference between target and housing temperature
Better than 0,01% of the absolute measured temperature per month
From Ø 0,7mm, depends on detector type and lens
Focus laser, built-in: aims the size of the field of view in focal distance
Pilot laser, built-in: aims the center of the field of view in any distanceLaser pointer as accessory for non transparent lenses
More mechanical pointers are available
Time out or permanent operation, while flashing or continuous marking
Depends on model
Emissivity, Environmental temperature, Analog output, Function of analog output, Response time,Temperature unit, Valley/ Peak-picker with decay function, Laser operation
0,100 to 1,000 in 0,001-steps
from 5 ms to 600 s (0.005, 0.01, 0.03, 0.1, 0.3, 1, 3, 10, 30, 60, 120, 240, 360, 480, 600 s)°C, °F or K
4 scalable output signals, temperature linear
0-10V, 0-1V, 0-20mA or 4-20mA, Zoom function for temperature span > 50 K
Actual value, Maximum value or Minimum value
RS232-Interface, bi directional, 9.600 to 115.000 bps, for programming and data transfer
Monitors the instrument temperature
22-30 VDC or 24 VAC ± 10%, 48-400 Hz< 150 mA @ 24 VDC-20°C ... 60°C
-20°C ... 70°C
IP65 (IEC), NEMA 4 equivalent, 1.3 kg
Stainless steel and aluminum

KT15IIP/V5/01.2004/HT/ e/© HEITRONICS, Wiesbaden

Housing Dimensions in mm [inches]

HEITRONICS Infrarot Messtechnik GmbH $ Kreuzberger Ring 40 $ D-65205 Wiesbaden $Tel.: +49 (0)611 973930 $ Fax +49 (0)611 9739326 $ email: Info@HEITRONICS.com $ www.HEITRONICS.com

HEITRONICS
Infrarot Messtechnik