CES100 SERIESEXTENDED AREA SOURCES
The CES100 Series are the industry standard for extended
area infrared radiation sources with a maximum absolutetemperature of 100 / C. The fast slew rates, high stability,
uniformity and emissivity of the CES100 Series sources makethem the ideal choice for most IR imaging system testing
requirements. The stability and uniformity of these sources issurpassed only by EOI’s LES100 Series sources. If your test
requirements exceed 100/C, refer to EOI’s CES200 Series forsource temperatures up to 200/C.
Fast slew rates and high stability are the result of EOI’s
35 years of blackbody/controller design experience. The
temperature controller utilizes a proprietary Proportional Integral
Derivative (PID) control loop to provide fast settling times, zeroset point drift and excellent temperature stability.
High Accuracy in the displayed temperature reading isassured with the use of extremely stable probes and highresolution, precision processing electronics.
Emitting surface sizes range from 2 × 2 inch (51 × 51 mm)
to 24 × 24 inch (610 × 610 mm). Custom sizes are alsoavailable upon request.

The CES100 Series Sources are
suitable for laboratory or field use andfeature:—

Fast Slew Rates

Both IEEE-488 and RS232 Serial Interfaces areincluded as standard equipment with the CES100 Seriescontroller, not offered as an additional cost option, allowing
greater flexibility for integration into automated test systems.

—

High Stability and Accuracy—

Many Standard Sizes—

Two Computer Interfaces

A user-friendly menu system, accessed with a keypad on
the front panel of the controller, allows for easy configurationand operation of the CES100 Series system.

—Certification—

2 Year Warranty

CES100 Series extended-area sources areavailable in stand-alone configurations or
integrated into any of EOI's Target Projectors,
Simulators and Automatic Test Stations.
Available Options— Remote controlled shutter— Extended ambient range operation— Extended low end

Calibration of a CES100 Series source is a quick and easy
procedure using the front panel keypad and is traceable to theNational Institute of Standards and Technology (NIST). The
calibration procedure may be easily automated using EOI’s
Calsof99 calibration software, a Model LT100 thermometer anda computer with a GPIB interface. The recommended calibration
cycle for the CES100 Series is six months.

— V-groove or micro-cavity surface

859 Ward Drive — Santa Barbara, CA 93111-2960 — Phone 805-964-6701 — Fax 805-967-8590
E-mail eoi@electro-optical.com

www.electro-optical.com

SOURCE DIMENSION IN INCHES (mm)Depth to Emitting Surface

Emitting Surface
B

Case
WidthCCase
Depth
DCase
HeightEA
CES100-02

0.4(10)2.0(51)

3.6(91)5.8(147)

4.6(117)

2.7(69)

CES100-04

1.1(28)4.0(102)8.1(206)

6.8(173)

7.6(193)

4.0(102)

CES100-06

1.0(25)6.0(152)

10.9(277)

6.8(173)

12.7(323)

7.2(183)

CES100-08

1.1(28)8.0(203)12.1(307)8.7(221)11.1(282)

6.0(152)

CES100-101.0(25)10.0(254)

13.1(333)

7.2(183)

13.6(345)

7.0(178)

CES100-121.2(31)12.0(305)

14.9(378)

8.7(221)15.1(384)

7.8(198)

CES100-141.2(31)14.0(356)

16.1(409)

8.0(203)16.6(422)

8.5(216)

CES100-24

1.2(31)24.0(610)

29.0(737)

6.3(160)29.0(737)14.5(368)

Model

OpticalCenterline
F

SPECIFICATIONS
Temperature Range

0 / to 100 / C

Temperature Accuracy± 0 . 0 3 / C from 0 / to 100 / C

Stability± 0.003 / C from 0 / to 75 / C± 0.005 / C from 75 / to 100 / C

Uniformity± 0 . 0 1 / C within ± 5 / C of ambient temperature±0.3% thereafter

Slew Rate including
Settling Time to 0.01 / C< 7 0 s e c f o r a 1 0 / C temp change

Emissivity

0.97 ± 0.02, 2 : to 20 :Set Point Resolution

0.01/C

Display Resolution

0.01/ or 0.001/C

Cable length

8 feet standard, others available

Controller Size*

5.25" H x 19" W x 18" D (rack mount)Line Voltage*

100, 120, 220 or 240 VAC ± 10%(switch selectable), 50/60 Hz.

*Larger systems require 220V or 240V and may require a 7 inch high controller or a power
amplifier in addition to the temperature controller.
Copyright © 2000 Electro Optical Industries, Inc.
eo709CES100, OCT 2000

Specifications subject to change without notice.

859 Ward Drive — Santa Barbara, CA 93111-2960 — Phone 805-964-6701 — Fax 805-967-8590
E-mail eoi@electro-optical.com

www.electro-optical.com