英語版(CE あり) 17-07
Revised: 25 November, 2019
(Revision 6)[Operating Instructions]
Identification Number: 227AFBZX00042000

1. Liner Application and Firing①In case where using Liner, paste type, agitate Liner in the container, scoop requisite amount on the pallet and then mix it until
it reaches a uniform consistency by a spatula.
②Apply Liner thinly and condense lightly then fire as above mentioned schedule.
2. Dentine, Enamel and Accent Application and Firing
Mix each Dentine, Enamel, Accent with distilled water, and build up to the crown form. Fire as above mentioned schedule after
condensation and water absorption.
3. Correction to shape and Glaze Firing①Correct the shape using a carborundum, diamond point or white point.
②After correction to shape, clean by an ultrasonic cleaner with diluted water or alcohol. Then fire for glaze.
③If necessary, build up by applying Stain or Supplement and fire.

Dental Material 2, Dental Crown Material
Controlled Medical Device – Ceramics materials (70802000)

ZEO CE LIGHT ZR[Contraindications and Prohibitions ]Do not use this product on patients who have a history of allergy such as rush or irritation to crown restorations with this product orsimilar products.

[Shapes, Composition and Principles ]This product consists of the following items and contains the ingredients listed below:[Technical Directions and Points for Attention]
1. Do not use this product in combination with other products.
2. Do not grind or polish this product in the mouth.

[Precautions]

ItemState

LinerPaste

Glass powder in the system SiO₂, pigment, polyethylene glycol, etc.

Ingredients

DentinePowderGlass powder in the system SiO₂, pigment, etc.

EnamelPowder

Glass powder in the system SiO₂, pigment, etc.

AccentPowderGlass powder in the system SiO₂, pigment, etc.

Supplement

PowderGlass powder in the system SiO₂, pigment, etc.

StainPaste

Glass powder in the system SiO₂, pigment, propylene glycol, etc.

1. Precautions

This product is used to produce restorations made from dental ceramics.

①Stop using this product and seek medical attention if any sign of allergy, such as rash, eczema, redness, ulcer, swelling, itching,numbness and etc. appears in patients.
②Operators who have a history of allergy such as irritation or rush to this product or organic solvent must wear plastic gloves toavoid direct skin contact. Operators must discontinue using and seek medical attention if any sign of allergy, such as rash, eczema,redness, ulcer, swelling, itching, numbness and etc.
③In case of eye contact, immediately flush with running water and consult a physician.
2. Important Basic Cautions①When grinding and polishing this product, use exhaust fan and anti-dust mask approved by your local public safety agency in order
to avoid inhaling dust. Wear safety goggles to protect the eyes.
②Provide adequate ventilation during operation.

[Product Specification ][Handling and Storage][Intended use and effect / efficacy ]Physical Property
Item
Flexural strength
Chemical solubility
Liner thermal expansion(Test method: ISO6872:2015)
Requirement
More than 50 MPa
Max. 100 micro-gram/
Shall not deviated by more than 0.5 × 10-6/K from the stated value
Stated value: Firing 2 times, 4 times
Liner: 10.0
Dentine, Enamel: 9.6

Glass transition temperature

Shall not deviated by more than 20 degrees C. Degree from the stated value.
Start Value: Firing 2 times, 4 times
Liner: 610 degrees C
Dentine, Enamel: 590 degrees C[ Firing Schedule ]LinerDentine
Self-Glaze
Supplement
Stain

Dry Time(min.)Start Temp.
（℃）

Heat Rate（℃/min.）Firing Temp.
（℃）

Hold Time

5-10
6-9
4
64

450-500
550-600
550-600
550-600
550-600

60
60
60
60
45-60

920-940
900-920880-900
800-850
850-870

11
1
1
1Reducing
Pressure(kPa)
9797
Air
97
97(NOTE)・In case where Stain is used on the surface of Zirconia, Heat Rate should be set 45℃/min. In case where Stain is used for Ceramics,
it should be set 60℃/min.
・Conditions, such as the above schedule is a guideline, not precise data. Conditions of firing are different depending on kinds orshapes of furnaces. Confirm the temperature and time by test firing before clinical case.

Keep out of reach of personnel other than dental staff.

[Expiry Date]
1. This product must be used before the expiry date printed on the package.
2. The expiry date printed on the package is based on our authentication.
3. The expiry date printed on the package refers to use expiry date.
*(ex.; YYYY-MM means the last day of YYYY (year)/MM (month) as the expiry date)[ZEO CE LIGHT ZR: Package]
Item

ShadeStateLiner

L-A(1,2,3,3.5,4), L-B(1,2,3,4), L-C(1,2,3,4),
L-D(2,3,4), L-G(1,2,3,4,5,Or), L-W(1,2,3,4)

Paste

DentineD-A(1,2,3,3.5,4), D-B(1,2,3,4), D-C(1,2,3,4), D-D(2,3,4), D-W(1,2,3,4)Powder

EnamelE-(1,2,3,4), OE(1,2,3,4,Amber)

PowderAccentOpal 1，Opal 2，Snow White，White，Vanilla，Milky, Yellow Ocher, Mamelon Vanilla，Mamelon Orange, Mamelon Yellow，Occlusal Brown，Occlusal Orange，Occlusal Enamel，T-Natural,
Powder
T-Glass, T-Clear, T-Opacity，T-Buff, T-Amber，T-Salmon，T-Gray，T-Violet，T-Blue，T-Baby Pink，
T-Pink，G(1,2,3,4,5,6,7,Or)

Supplement

SUP-(D,E,T)

PowderStain

Glaze，White，Milky，Light Orange，Orange，Brown，Dark Brown，Gray，Violet，Blue，Pink，Yellow，Black，Indigo，Dark Pink，Amber，Khaki，Buff, Fluorescent, Opal Glaze,
Shade A, Shade B, Shade C, Shade D

Paste