E

Instruction Manual

PROFESSIONAL

DHG-35M
DHG-60M

ModelNo.

,DIGITAL S-LINE HEIGHT GAUGE
Precision Measuring Instrument■R①e reer c u er c ca ere
rer aee.

ea●e Scr e「L・・a er

①e
Scr e.

e Scr eaScre

① Post●⑦ Slide Clamp⑮ Data Output Port

⑪ Lock Button⑥ Slide⑫ 䣝PR/SET PR䣟䢢Btn.

②Slide②er

CR2032（⑩ [ON/0-ABS] Button · · · · · · · · Power ON/OFF; set measurement to zero.
⑪ Lock Button⑫ [PR/SET PR] Button · · · · · · Set and recall the Preset Value.
⑭ Scribe (w/ Carbide Tip) · · · · Measurement tip, or to scribe workpiece.

⑰ Jaw · · · · · · · · · · · · · · · · · · · · · · · · · Scribe mounts here.

Use and store only in environment meeting the following
conditions:● Dry and protected from rain, water, and oils.
● Protected from direct sunlight.
● Location that is not excessively hot.
● Not subject strong electric or magnetic fields.
● Protected from children and unauthorized users.
・Use in places contrary to the above may cause damage tothe product or affect accuracy resulting in accident or injury.

Please Observe

Denotes a requirement– You MUST do.

Remove Battery from gauge when it will not be used foran extended period.
・Battery may leak if unused for a long time.

Read the manual and follow the directions.
・Use of product other than as described in this manual
may cause accident.
Use only for measuring and scribe work.
・Use for any other purpose may damage or wear theinstrument. Improper use may also cause accident.
Handle with care.
・Do not drop or subject instrument to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage or poor accuracy.
・ Do not scratch instrument surfaces.
・ Before use, clean the surface plate, Base underside, andScribe Measurement Surface with a wipe or soft cloth.

Scribe tip is sharp; handle with care.
・It may cause injury.

u」
：

１

■ueea ure e① urer.
ereaeRc「LongSe e Scr ea ea ure e .

」：「「Long「LongeA dial Indicator (sold separately) can be mounted in place of theScribe using an optional Indicator Mounting Arm (also soldseparately.)ec䷷ Insert the Mounting Arm into the Scribe Clamp replace
onto the Jaw. Adjust the amount of overhang, andtighten the Scribe Clamp Screw to secure.

a eDial Indicator.
Re ea e⑦Se「

「Long⑥u Scree」⑬ Scribe Clamp Screwc

PR

Indicator Clamp：」：

⑰ Jaw

⑦

「Long」「※」e⑭① ueeuea . Surface

eee er f r aru
eS e ac Scre .
eeu②e Scr e ear e e reeeeuc Scre .
eeu
ee Scre③ ura uee Scr e.

ceSSurfaceae⑦■arae① urer.

② ueSeS e③⑯e③ See「 .

⑤⑥●Remove battery if instrument will not be used for a long periodof time.

SPECIFICATIONS
Model No.
Meas. Range
Resolution
Max. Permissible Error
Display
Mass
Op. Temp. Range
PowerAccessories

DHG-35M
0−350mm

DHG-60M
0−600mm0.01mm±0.04mm±0.05mm
5 digit LCD~7㎏e

Data Buttoneaee er u aeuSue Refere ce Surface areee ea ure ee Scr eea ure」.
eData Transfer Cableea ure e「」.

ecr urface

f r aruc Scre .

~8㎏

10〜40℃
CR2032（lithium battery）
・Instruction Manual ・CR2032
・Battery Tool（lithium battery）※ for test

● Measuring and marking workpieces, ideal for tool and die work.
● Handwheel for smooth operation.
● Digital display for easy reading.
● Zero point can be set at any position.
● Preset Value can be set and stored.
● Auto-off function (Settable to 8, 120 min., disabled)e● If Instrument is not working properly, or if you have anyquestions, please contact distributor or place of purchase.
● Please note, manufacturer is unable to respond to inquires
or provide service directly. Please contact distributor or placeof purchase.

Coverc・For communication details, please refer to the manual forthe『SK-LOG Measuring Instrument Communication
Software』, sold separately.

ea ure .

TROUBLESHOOTING
Error Condition

Corrective Action

傘Error備䢢is displayed on
the LCD.

The Slide was moved too fast;press the [ON/0-ABS] Button andretake the measurement.

To maintain instrument accuracy, periodic calibration is recommended.
Calibration interval necessary to maintain accuracy will vary depending on frequency
and conditions of use. Please follow your company's guidelines and calibrate regularly.
Outside Japan,Please contact distributor or place of purchase to inquire about calibration services.

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518⑭３４D●SERVICEea e f●After use, clean measuring surfaces, sliding surfaces, base,and exterior using a dry cloth and perform rust prevention
treatment.
For rust prevention, lightly apply anti-rust oil.

Ref. Surface

ea

AFTER USE 兟 STORAGEAPPLICATIONS兟FEATURES

Connecting the Data Transfer Cable
Using a small screwdriver, remove the Cover and attach thecable.

※era■ Data Output
Connect the optional Data Transfer Cable (DTC-M1) to output
data. To send data, connect the cable and press the Data Buttonon the cable.

⑲■.

」

Indicator Mounting Arme a e ecure
ur ae uS e aee .

：Set the.

.
」⑱ Scribe Clamp

⑥PR」

２

䷶ loosen the Scribe Clamp Screw and remove from theJaw.

ea ure a.

.

ea ure ee③「dir 1」：「dir -1」：

e.

②』

.

Scree①ȢȢ.

」：「Long䷵ On the Indicator Mounting Arm, loosen the Clamp andinsert the Indicator into the mounting hole and secureby tightening the Clamp.

eȡREL

※ru」：
ȡeea e.

.

■uereue Surfacee a eac acRSf r aruc Scre .

er aeraS「■ Mounting a Dial Indicatore er u aeeueeSe Scr eu

：

…

e『※」

●Do not clean main body with organic solvents.
・It may damage product and affect accuracy.

e「■■ uDo not write on instrument with electric engraver.
・Do not engrave instrument, such as an ID number, withengraver or electric pen.

③ea ure e…
reDo not disassemble or modify.
・Disassembly may damage product or cause accident.
For repair, please contact distributor or place of purchase.

② ueSeS eeu

S⑱ Scribe Clamp · · · · · · · · · · · · · · Secures the Scribe.

⑲ Base · · · · · · · · · · · · · · · · · · · · · · · Supports the Height Gauge.

「L⑱S⑯ Fine Adjust Feed · · · · · · · · · · Turn for fine adjustment.

Indicates risk of personal injuryR）

・・⑬ Scribe Clamp Screw · · · · · · Tighten to secure the Scribe.

⑲ Base「「L· · · · · · · · · · · · Disables button operation.

WARNING: or property damage if not followed.

RSee Scr ee Scr e au au
re acee a .
efer aaee Scr e aecure.

erfaca u
Scre⑭S⑨ LCD Display · · · · · · · · · · · · · · · Digital display of measurement.

⑰ JawDenotes a prohibition– You MUST NOT do.

●②⑧ Battery Cover · · · · · · · · · · · · · Open to access Battery.

To prevent harm to yourself or others, and to prevent damage to property,
always follow the procedures marked by the following symbols.

c（＋）⑦ Slide Clamp · · · · · · · · · · · · · · · Locks Slide position.

⑮ Data Output Port · · · · · · · · · · Data Transfer Cable connects here.

SAFETY NOTICES●a er .

⑰re「Longu⑥ Slide Handwheel · · · · · · · · · · For moving the Slide.

④ Base Ref. Surface⑭ Scribe（ w/ Carbide Tip ）

055
18R

⑤ Fine Adjust Lock Screw · · · Tighten when using Fine Adjust Feed.

⑧ Battery Cover⑱ Scribe
Clamp

⑭S⑱S

- REQUIRED step or necessary condition.

④ Base Reference Surface · · · Measurements are to this surface.

Handwheel⑬ Scribe
Clamp Screw「L③ Fine Adjust Carriage · · · · · For precise control of Slide position.

⑨ LCD Display⑩ 䣝ON/0-ABS䣟䢢Button「- RISK OF PERSONAL INJURY OR
PROPERTY DAMAGE if not followed.
- PROHIBITED action.

② Slide · · · · · · · · · · · · · · · · · · · · · · Moves up and down with Display.

⑤Fine Adj. Lock Scr.

e⑰

① Post · · · · · · · · · · · · · · · · · · · · · · Supports the Slide and Display.

③ Fine Adjust Car.

⑯ Fine Adjust
Feed Screwe

SAFETY NOTICES

use and follow the procedures described.
Please keep manual where it is accessible to user for future reference.
● Keep this manual with the instrument if transferred or leased to a third party.
● For inquiries about this product, please contact the dealer wherepurchased, or Niigata Seiki.

■PARTS ID & FUNCTION

re⑬

Thank you for purchasing the DIGITAL S-LINE HEIGHT GAUGE.
This precision measuring instrument is for use on a surface plate to measure the height of a feature, or to scribe a workpiece.
● For safe and proper use, please read this instruction manual before

aMAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

●
●

業務用

取扱説明書

精密測定器

DHG-35M
DHG-60M

品番

デジタルSラインハイトゲージ

この度は「デジタルSラインハイトゲージ」をお買上げ頂きありがとうございます。この製品は、定盤上や一定の基準面に置き、高さ測定やケガキ作業を行う精密測定器です。
●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。

この説明書には拡大損害が予想される事項には、禁止事項には、必ずお守りいただきたい
強制事項にはマークを付けて表示しています。

●本製品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称と機能

①本尺

①本尺

⑯微動送りねじ⑦クランプレバー⑮データ出力端子⑪ロックボタン

⑥スライダ
ハンドル

⑫PR/SET PRボタン②スライダ⑤微動止めねじ⑰ジョウ⑭スクライバ⑩ON/0-ABSボタン ……… 電源のON/OFF、数値のゼロリセットを行います。

⑫PR/SET PRボタン ……… プリセット値の設定、呼び出しをします。
⑭スクライバ（超硬チップ付）… 被測定物にあて測定やケガキをします。

………………… スクライバを取り付けます。

⑲ベース………………… ハイトゲージ全体を支えます。

使用方法⑱スクライバクランプ …… スクライバを固定します。

必ず実行していただく
内容『強制事項』です。

お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

⑱スクライバクランプ

・電池は必ずCR2032をお使いください。
・初めてお使いになる際は絶縁シートを剥がしてください。

……… データ転送ケーブルを接続します。

………… 回してスライダを微動送りします。

⑰ジョウ

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。

CR2032（リチウム電池）……… 各ボタンの操作を無効にします。

⑬スクライバ止めねじ …… スクライバを固定します。

安全上のご注意 必ずお守りください

取扱説明書をよく読み、指示に従う。
・取扱説明書に記載された内容以外での使用は、事故の原因となります。
測定作業、ケガキ作業にのみ使用する。
・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。
大切に取り扱う。
・ぶつける、落とすなどの衝撃を与えたり、重量物を上に乗せたりしないでください。精度不良や破損の原因となります。
・製品に傷をつけないでください。
・使用前はグラスペーパなどでスクライバの測定面、ベース基準面、定盤表面の汚れを除去する。

②スクライバの先端を外側にしてスクライバクランプに入れ、突き出し量調節を行った後、スクライバ止めねじを締めて固定します。

電池蓋……… スライダを固定します。

⑪ロックボタン⑲ベース

下記の条件を満たす環境で使用・保管する。
●雨や水、油などがかからない、乾燥した場所●直射日光の当たらない場所●高温にならない場所●強い磁気や高電圧のない場所●子どもや、使用者以外が近づかない場所
・上記に反する場所での使用は、精度不良や製品の破損、事故やけがの原因となります。
長期間使用しない時は、電池を本体から取り出す。
・長時間使用しないで放置すると、電池の液漏れの原因となります。
スクライバ先端は尖っているので、扱いに注意する。
・けがの原因となります。

突き出し量調節■各ボタンの使用方法

押す…押してすぐはなす（1秒以内）
長押し…押して保持する（1秒以上）●ON/0-ABS ボタン「押す」 ：比較測定モードに切り替え■電源の ON/OFF

LCD 表示部に「REL」が表示され数値が「0.00ｍｍ」となります。

［ON/0-ABS］ボタンを押す、またはスライダを動かすと電源がON になります。
［ON/0-ABS］と［PR-SET PR］ボタンを同時に長押しすると電源が OFF になります。

REL※画面は OFF になりますが、直前の状態は保持されたままです。
電源を ON にすると、OFF にした時の状態で復帰します。

「長押し」：絶対測定モードに切り替え（前回の基準点を保持）

プリセット機能で設定した数値を基準点とした測定値が表示されます。LCD 表示部には、測定値のみ表示されます。

約8分間無操作状態が続くと、自動的に電源がOFFになります。
電池残量が残りわずかになると『 』マークが表示されます。
電池を交換してください。

分解・改造を行わない。
・製品の破損や事故の原因となります。修理はお買い上げの販売店、または発売元までご連絡ください。

「押す」（電源 OFF 時）：電源 ON「長押し」（［PR-SET PR］ボタンと同時）：電源 OFF「長押し」（ロックボタン

電気ペンなどで書き込まない。
・電気ペンなどで本体などに、番号などの書き込みをしないでください。

LCD 表示部に「dir 1」と「dir -1」が表示されます。

「dir 1」：スライダを上に動かすと数値が増加し、下に動かすと数値が減少します。

１

スクライバを外し、ダイヤルインジケータホルダ（別売）を取り付けることで、ダイヤルインジゲータ（別売）を装着することができます。

●測定面・摺動面・ベース・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う③スライダハンドルと微動送り機能を使い、スクライバの測定面を定盤に密着させます。

①ダイヤルインジゲータホルダのホルダ止めねじをゆるめ、取付穴にダイヤルインジゲータの軸を差し込んでからホルダ止めねじを締めて固定します。

●長期間使用しない場合は電池を製品から取り外した状態で保管する④［PR/SET PR］スイッチを押して、LCD 表示部の数値を「0.00ｍｍ」にします。

②スクライバクランプのスクライバ止めねじをゆるめ、ジョウに通します。

⑤被測定物にスクライバの測定面を当てて測定します。

③ダイヤルインジゲータホルダをスクライバクランプに入れ、突き出し調節を行った後、スクライバ止めねじを締めて固定します。

②クランプレバーを手前に起こしてスライダハンドルのロックを解除し、微動止めねじをゆるめます。

●PR/SET PR ボタン

測定画面で呼び出すことができる、任意の数値を設定します。
（プリセット機能）※あらかじめ [PR/SET PR] ボタンを長押しし、プリセット値を「0.00mm」に設定してください。

測定画面で使用後のお手入れ・保管■ダイヤルインジゲータホルダについて①電源を入れます。

LCD 表示部にロックのマークが表示され、各ボタンの操作が無効になります。再び長押しすると解除されます。

「dir -1」：スライダを上に動かすと数値が減少し、下に動かすと数値が増加します。

２
■絶対測定

●ロックボタン

防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。

仕様

設定画面で品番測 定 範 囲
最小読取値
最大許容誤差表
示
質量
使用温度範囲
電源「長押し」：次の桁に移動

付「押す」 ：設定した数値を表示し、絶対測定モードに切り替え「長押し」：設定画面に切り替え

LCD 表示部に「PR」が表示され、「＋」が点滅します。

ダイヤルインジゲータ

ロック解除⑦クランプレバー⑬スクライバ止めねじ

ロック⑥スライダハンドルPR

ホルダ止めねじ「押す」 ：＋／−の切り替え、数字の増加
最後の桁で長押しすると決定し、数値が保存されます。

⑰ジョウ⑦クランプレバー※スライダハンドル使用時は、ベースを手でしっかりと押さえてご使用ください。

PR

設定途中で［ON/0-ABS］ボタンを押すと、設定を保存せず測定画面に戻ります。

⑲ベース※初期設定ではオートオフ時間は 8 分に設定されています。

測定面■微動送り機能①クランプレバーを手前に起こしてスライダハンドルのロックを解除し、微動止めねじをゆるめます。
②目的のおおよその位置にスクライバ先端を合わせ、微動止めねじを締めます。

盤③微動調整装置

約8㎏

10〜40℃
CR2032（リチウム電池）

・取扱説明書品 ・電池蓋レンチ

・CR2032（リチウム電池）
※テスト用

修理の際は

［DATA］ボタン基準となる被測定物との差が表示されます。

●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店、または発売元までご連絡ください。
●お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

カバー
・通信については別売の『測定器通信統合ソフトSK-LOG』の取扱説明書を参照してください。

トラブルと対応
トラブルの内容

測定精度を保つため、定期校正を実施することをおすすめします

性能の劣化は、使用の頻度・環境によって大きく異なります。
お客様の社内規格に準じた周期で、定期点検を行うことをおすすめ致します。
当社にて校正を受け付けております。
全国にある当社代理店、取扱店を経由してお申し込みください。

対応策

ディスプレイに「Error」 スライダの移動が早すぎます。と表示される［ON/0-ABS］ボタンを押し、再度測定を行ってください。

⑭スクライバ±0.05mm

用途・特長

マイナスドライバなどでデータ出力端子のカバーを外し、データ転送ケーブルを接続します。

①電源を入れます。

④比較する被測定物を測定します。

⑥スライダ
ハンドル

LCD5桁表示●金型・治工具の測定やケガキ作業に最適。
●ハンドル操作によるスムーズなスライドが可能。
●読み取りやすいデジタル表示。
●任意の位置でゼロ設定ができます。
●数値のプリセット機能付。
●オートオフ機能付（8分・120分・なし）。

データ転送ケーブル接続方法

データ転送ケーブル

画面には「REL」の文字が表示されます。

⑦クランプレバー■データ出力■比較測定③基準となる被測定物にスクライバの測定面を当て、［ON/0-ABS］ボタンを押して、LCD 表示部の数値を「0.00ｍｍ」にします。

⑤微動止めねじ

約7㎏0.01mm

基準面②クランプレバーを手前に起こしてスライダハンドルのロックを解除し、微動止めねじをゆるめます。

③微動送りねじを回し、微調整を行います。

⑯微動送りねじ定±0.04mm

DHG-60M
0−600mm

ダイヤルインジゲータホルダ

別売のデータ転送ケーブル（DTC-M1）を接続することで、データ出力が可能です。データ送信はデータ転送ケーブルを接続した状態でケーブルの［DATA］ボタンを押してください。

⑭スクライバLCD 表示部に「OFF」が表示されます。

［PR/SET PR］ボタンを押してオートオフ時間 8/120/-（なし）を選択し、［PR/SET PR］ボタンを長押しして決定します。

属

DHG-35M
0−350mm⑱スクライバクランプ⑥スライダハンドル「長押し」（ロックボタン と同時）：オートオフ時間の変更（8 分 /120 分 / なし）と同時）：測定方向の切替

測定方向の＋/−を切り替えます。

本体のお手入れに有機溶剤は使用しない。
・製品の精度不良や破損の原因となります。

「長押し」：各ボタンの操作無効 / 有効の切り替え⑭スクライバ

⑱スクライバクランプ………… 微動送り機能を使用する際に締めて固定します。

⑨LCD表示部 ……………… 測定数値をデジタル表示します。

⑰ジョウ

注意

②CR2032（リチウム電池）を挿入します。

………… スライダの移動を微調整します。

⑮データ出力端子

してはいけない内容『禁止事項』です。

⑰ジョウ………………… 開けて電池をセットします。

⑯微動送りねじ⑭スクライバ（超硬チップ付）⑬スクライバ
止めねじ………… 測定の基準面になります。

⑦クランプレバー⑧電池蓋④ベース基準面⑱スクライバクランプ①スクライバクランプのスクライバ止めねじをゆるめ、スクライバを取り外します。

電池蓋⑥スライダハンドル …… スライダを移動させます。

⑧電池蓋⑬スクライバ
止めねじ……………… 液晶ユニットが内蔵され上下に移動します。

④ベース基準面⑨LCD表示部⑩ON/0-ABSボタン…………………… デジタル液晶部のスライドを保持します。

③微動調整装置⑤微動止めねじ・開梱時、スクライバは先端部保護のため通常使用時とは反対方向に取り付けられています。
・スクライバ先端部分は鋭利になっているので、取り扱いにご注意ください。

①付属の電池蓋レンチなどを使用し、電池蓋を反時計方向に回して外します。

電池は必ず（＋）側を表にしてセットしてください。

②スライダ③微動調整装置■スクライバの取り付け

電池蓋レンチ

安全上のお知らせ●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。

電池のセット・交換〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

３４I436-K12111