参考

物品仕様書

CV ケーブル活線劣化診断装置
DISC-20-AS(標準仕様

50Hz，60Hz 兼用)

―

目次

―

1

総則 ..................................................................................................................................... - 1 -1.1

適用範囲............................................................................................................................... - 1 -1.2

使用条件............................................................................................................................... - 1 -1.3

表示....................................................................................................................................... - 1 -1.4

物品名称............................................................................................................................... - 1 -1.5

関連規格............................................................................................................................... - 1 -1.6

保守・点検........................................................................................................................... - 1 -1.7

部品供給............................................................................................................................... - 1 -2構造 ..................................................................................................................................... - 2 -2.1

外箱....................................................................................................................................... - 2 -2.2

構成....................................................................................................................................... - 2 -3

機能 ..................................................................................................................................... - 3 -3.1

各部の機能........................................................................................................................... - 3 -3.1.1

測定装置 ........................................................................................................................ - 3 -

3.1.2

重畳装置 ........................................................................................................................ - 6 -3.1.3

電源アダプタ................................................................................................................. - 7 -3.2

測定フローチャート ........................................................................................................... - 9 -4

各部の名称 ....................................................................................................................... - 10 -4.1

測定装置............................................................................................................................. - 10 -4.2

重畳装置............................................................................................................................. - 11 -4.3

電源アダプタ（AC アダプタ） ...................................................................................... - 12 -4.4

電源アダプタ（DC アダプタ） ...................................................................................... - 13 -5

測定手順 ........................................................................................................................... - 14 5.1.1

準備 .............................................................................................................................. - 15 -

5.1.2

測定 .............................................................................................................................. - 18 -5.1.3

片づけ .......................................................................................................................... - 18 -6

操作方法 ........................................................................................................................... - 19 -6.1

キーの説明......................................................................................................................... - 19 -

6.2

システム異常処理 ............................................................................................................. - 20 -6.2.1

プリンタ異常画面 1.................................................................................................... - 20 -6.2.2

プリンタ異常画面 2.................................................................................................... - 20 -6.2.3

電源異常画面(DC アダプタ使用時).......................................................................... - 20 -6.2.4

測定用リード線断線異常画面................................................................................... - 21 -

6.2.5

重畳装置過電流異常画面........................................................................................... - 21 -

6.2.6

重畳装置過電圧異常画面........................................................................................... - 21 -

6.2.7

測定装置過電流異常画面........................................................................................... - 22 -6.2.8

保護回路設定異常画面............................................................................................... - 22 -6.2.9

重畳部過電流異常画面............................................................................................... - 22 -6.2.10

重畳部過電圧異常画面............................................................................................. - 23 -6.2.11

重畳部温度異常画面 ................................................................................................. - 23 -6.2.12

重畳電圧の検出異常画面......................................................................................... - 23 -6.2.13電磁接触器の動作異常画面..................................................................................... - 24 -6.2.14 USB メモリ異常画面 ................................................................................................. - 24 6.2.15 USB メモリ空き容量不足画面 ................................................................................. - 24 -6.2.16

システム異常画面..................................................................................................... - 25 -6.2.17

処理中メッセ－ジ 1.................................................................................................. - 25 -6.2.18

処理中メッセ－ジ 2.................................................................................................. - 26 -6.2.19

自己チェック異常画面............................................................................................. - 26 -6.2.20

非常停止 .................................................................................................................... - 26 -6.3

プリンタ用紙および注意事項 ......................................................................................... - 28 -6.4

USB メモリの取り扱い .................................................................................................... - 29 -6.4.1 USB メモリ仕様 ........................................................................................................... - 29 6.4.2

操作方法 ...................................................................................................................... - 29 -6.4.3

注意事項 ...................................................................................................................... - 29 -6.4.4

データファイルの内容............................................................................................... - 29 -7

特性 ........................................................................................................................................31

7.1

外観・構造..............................................................................................................................317.2

基本機能試験 ..........................................................................................................................31

7.2.1

各種機能 ...........................................................................................................................31

7.2.2

交流重畳出力電圧............................................................................................................31

7.2.3

交流重畳出力電圧歪み....................................................................................................31

7.2.4

充電機能（DC アダプタ） .............................................................................................317.3

動作試験..................................................................................................................................31

7.3.1

交流重畳電流（1Hz） .....................................................................................................31

7.3.2

シース絶縁抵抗................................................................................................................317.3.3

過電流保護機能................................................................................................................31

7.3.4

過電圧保護機能................................................................................................................31

7.3.5

測定用リード線の断線チェック機能 ............................................................................317.4

強度試験..................................................................................................................................327.4.1

絶縁抵抗 ...........................................................................................................................32

7.4.2

耐電圧 ...............................................................................................................................32

8

試験方法 ................................................................................................................................338.1

外観・構造..............................................................................................................................33

8.2

基本機能試験 ..........................................................................................................................33

8.2.1

基本機能 ...........................................................................................................................33

8.2.2

交流重畳出力電圧............................................................................................................33

8.2.3

交流重畳出力電圧歪み....................................................................................................33

8.2.4

充電機能（DC アダプタ） .............................................................................................33

8.3

動作試験..................................................................................................................................338.3.1

交流重畳電流(1Hz) ..........................................................................................................33

8.3.2

シース絶縁抵抗................................................................................................................338.3.3

過電流保護機能................................................................................................................33

8.3.4

過電圧保護機能................................................................................................................33

8.3.5

測定用リード線の断線チェック機能 ............................................................................33

8.4

強度試験..................................................................................................................................348.4.1

絶縁抵抗 ...........................................................................................................................34

8.4.2

耐電圧 ...............................................................................................................................34

9

仕様一覧 ................................................................................................................................35

9.1

測定装置..................................................................................................................................35

9.2

重畳装置..................................................................................................................................359.3

電源アダプタ：AC アダプタ ...............................................................................................369.4

電源アダプタ：DC アダプタ ...............................................................................................36

1

総則
1.1 適用範囲
本仕様書は交流重畳法を測定原理とし、高圧CVケーブル（6kV級）を活線状態で絶縁劣化診断する「CVケーブル活線劣化診断装置」に適用します。

1.2 使用条件
CVケーブル活線劣化診断装置は下記の条件で正常な機能を有します。
(1)

周囲温度

0～40℃

(2)

湿度

90%RH 以下（但し、結露しないこと）

(3)

使用場所

屋外（但し、雨・雪・霧等が本装置にかからないようにすること）

(4)

耐衝撃

衝撃を与えないこと

(5)

運搬時

装置を横にしないこと

1.3 表示CVケーブル活線劣化診断装置の銘板に次の事項を記します。
(1)

品名

(2)

形式

(3)

製造番号

(4)

製造年月

1.4 物品名称
物品名称は「CVケーブル活線劣化診断装置（DISC-20-AS）」とします。

1.5 関連規格
本仕様書に関連する規格はJIC（日本工業規格），JEC（電気学会

電気規格調査会標準規格）です。

1.6 保守・点検
CVケーブル活線劣化診断装置は使用頻度や使用環境等により、劣化が進行し、測定精度の低下、故障の原因となる場合があります。長期的に使用するためには、定期点検、および修理が必要となります。
定期点検、および修理については最寄りの矢崎エナジーシステム（株）各支店の電線営業部にお問合せ下さい。連絡先は、本仕様書の裏表紙に記載しておりますのでご参照ください。

1.7 部品供給
CVケーブル活線劣化診断装置はお客様にお断りなく製造を終了する場合があります。
製造終了した場合の部品供給期間は、製造中止後5年とします。修理対応期間は、製造中止後5年または販売後10年のいずれか早い方としますが、修理対応期間を経過した場合も当社が対応できる限り対応致します。但し、修理対応期間であっても修理が困難な場合は代替装置を提案させて頂くことがあります。

-1-

2

構造
2.1 外箱(1)

外箱はアルミとポリエチレンの複合板製のキャリーケースとします。

(2)

色はアルミ色（つや消し）とします。

(3)

上蓋は約120°の開度で固定できる構造とします。

(4)上蓋および両側面に取手を取付けます。

(5)

両側面にショルダーベルト用のフックを取付けます（ベルトは取外し可）。

(6)

表面パネルにはポリカーボネート板に印刷を施し、着色したものを使用します。

2.2 構成CV ケーブル活線劣化診断装置の構成を以下に示します。
(1)

測定装置

1台

(2)

重畳装置

1台

(3)

電源アダプタ・ACアダプタ

1台

・DCアダプタ

0～2台

注）※DCアダプタの数による

(4)

附属品

a) 電源ケーブル

2m

1～3本

b) 信号ケーブル

0.6m

1本

c) 測定用リード線
・8mm2×2心5m

1本・8mm ×2心 15m

1本2d) ケーブル収納袋
・測定用リード線5m用

1袋

・測定用リード線15m用

1袋e) ショルダーベルト

3～5本f) 感熱ロール紙

2箱（10本/箱）

g) USBメモリ

1個（USB2.0準拠）

h) USBコネクタキャップ

1個i) 温湿度計

1台j) 取扱説明書

2部k) CD-ROM(解析ソフト)1枚l) 試験成績書

1部m) 保証書

1部※DCアダプタの数による

注）DCアダプタは別売です。
注）出荷時、USB コネクタには防塵用のコネクタキャップを嵌めています。

-2-

3

機能
CVケーブル活線劣化診断装置の機能ブロック図を 図 3.1に示します。以下、各部の主な機能を示します。

図 3.1

CV ケーブル活線劣化診断装置の機能ブロック図

3.1 各部の機能
3.1.1

測定装置

a) 測定機能CVケーブル活線劣化診断装置の測定項目および測定範囲を表 3.1

測定項目および測定範囲に示します。
表 3.1

測定項目および測定範囲

測定項目

略号

測定範囲

交流重畳電流Isa

1nA～100μA＊

備考

＊＊

－

シース絶縁抵抗Rs

100kΩ～100MΩ

迷走電流

Idc

1nA～100μA（参考値）

不平衡充電電流Iac

1mA～850mA

（参考値）

－

※最大誤差は±10%以内。但し、50nA以下では±5nA以内とします。

※※100kΩ ～ 100MΩで最大誤差は±10%以内とします。
CVケーブル活線劣化診断装置では、重畳装置内の交流発生器より所定の交流電圧(101/121Hz、50V)を出力し、交流重畳電流値：Isa（1Hz近傍の微小電流）を測定します。

-3-b) 交流重畳電流
CVケーブル活線劣化診断装置では、101/121Hz、50Vの交流電圧を重畳した時に接地線に流れる0～2.0Hzの電流を、0.05Hzの周波数分解能でFFT処理を行っています。このデー
タをもとに、対象周波数範囲を基本的には0.8～1.2Hzとして交流重畳電流値（Isa）を算出
しています。
c) 劣化判定機能
CVケーブル活線劣化診断装置では、交流重畳電流値（Isa）を使って劣化判定を行い、その結果を出力します。シース絶縁抵抗値（Rs）が250kΩ未満では正確な判定ができなくなることがある為、判定不能とします。
なお、ケーブルが「劣化」と判断されるレベルは、ユーザ殿の考え、運用方法によって異なります。装置で「劣化」と判定されるレベルは、概ね直流漏れ電流法で-16kV印加した時に漏れ電流値が0.1μAを超える水トリーのレベルです（100%正確な劣化判定はできません）。
※Isa の値は、ノイズ推定量を差し引いています。この為、マイナス値となる場合があります。この時、Isa に対するノイズ推定量が大きく、ノイズの影響が大きい事が考えられる為、「ノイズ影響大」と印字されます。
d) シース絶縁抵抗測定機能
測定回路に－5Vの電圧を印加し、測定される直流成分電流Ir から下記の式で求めます。

Rs 

5
()
Ir

……………(1)e) 測定結果の印字機能プリンタでの出力表示例を 図 3.2に示します。なお、測定したデータはUSBメモリに記録でき、記録したデータはUSBメモリを介してパソコンでの結果表示とプリントアウトが可能です。なお、結果表示用の解析ソフト（CD-ROM）が同梱されています。

-4-

＊USB メモリは、付属品を使用してください。
また、パソコンは USB2.0 に対応した機種を使用
してください。

判定結果
交流重畳電流シース絶縁抵抗
迷走電流
不平衡充電電流

※1

※2

図 3.2

出力表示例（プリンタ印字）※1「測定電流波形グラフ」
測定電流波形グラフより、歪みの少ない 1Hz の電流波形であれば劣化信号であることを確認することができます。歪みの多い電流波形であれば健全ケーブル、またはノイズが多く含まれていることが確認できます。
※2「FFT 結果グラフ」
FFT 結果グラフより、1Hz の電流値が他の周波数に比べて突出しているときに劣化信号であることを確認できます。1Hz 以外の周波数が大きければ健全ケーブル、またはノイズが多く含まれていることが確認できます。

-5-

重畳装置

3.1.2

本重畳装置は、交流重畳電圧を発生する重畳部と異常時に人身およびCVケーブル活線
劣化診断装置を保護する保安回路で構成されています。

a) 重畳部
本重畳部の主な仕様を 表 3.2に示します。

表 3.2 重畳部の主な仕様
項

目

内

容

備

考

入力電圧

DC24V

－

入力容量

約 50VA

－

定格容量

100VA

－

出力電圧

50V（正弦波）

精度±5% 以内

最大電流

1A

ピーク電流 2.5A

周 波 数

101/121Hz

精度±0.01 Hz 以内

タ イ プ

リニアアンプ方式－b) 保安回路
本保安回路の基本構成を 図 3.3に、保護機能を 表 3.3に示します。保護機能は安全性を万全にするため、バックアップシステムを設置しております。被測定ケーブルが地絡事故を生じた際等の異常時に生じる過電圧注）、過電流をOC1あるいはOC2が検出し、電磁接触器を制御（測定系を短絡保護）します。
注）交流重畳法は、回路中の非線形素子により発生した劣化信号を検出します。このため、過電圧の検出はＰＴを用いず、アレスタ1あるいはアレスタ2の動作をOC1あるいはOC2で検出します。

重畳装置

既設
接地線

測定用
リード線

アレスタ１

電磁接触器

ＯＣ１

ＯＣ２

過電圧、過電流検出

アレスタ２

測定装置

過電圧、過電流検出

図 3.3 保安回路の基本構成
例えば、過電圧が入力した場合、まずアレスタ1が動作します。これにより発生した放電電流をOC1の過電流検出回路で検知し、電磁接触器で入力を短絡保護（60ms以内）します。万一、アレスタ1が不動作でもアレスタ2が動作し、この動作をOC2の過電流検出回
路で検知します。また、過電流が入力した場合には、OC1あるいはOC2の過電流検出回路
がこれを検知し、電磁接触器を動作させます。なお、測定中に被測定ケーブルで地絡事故-6-

が生じ、本重畳装置の保安回路が動作した場合、CVケーブル活線劣化診断装置全体の点検を受けてから使用して下さい。

表 3.3
項

保護機能

目

動 作 開 始 範 囲

過電圧検出回路AC 70V 以上 100V 以下

過電流検出回路AC 850mA ± 10%

耐

量

アレスタ1

AC 10A（1回限り）

アレスタ2

AC 5A、1秒

電磁接触器

AC 1kA、1秒

アレスタ1

DC230V± 20%

アレスタ2

DC90V ± 20%c）測定用リード線の断線チェック
フィールドでの測定では、測定用リード線を含め、測定回路全体で安全性を確認する必要があります。そこで、液晶表示画面に従い、測定用リード線を既設接地線に接続後、エンターキーを押すことにより、測定用リード線の断線チェックを自動的に行います。 図

3.4に確認回路を示します。なお、この際に、電磁接触器の動作確認もあわせてチェック
します。
接続金具
接続金具
測定用
測定用
リード線(赤)

1kΩ
1k
Ω
200kΩ
200k
Ω
(断線チェック用検出抵抗)(断線 チ ェ ッ ク 用 検 出 抵 抗)

既
設接地線
既設接地線

接続金具
接続金具

図 3.4

3.1.3

150μF

50μF

1kΩ(アンプ保護)

5V

測定用
測定用
リード線(緑)
ﾘｰﾄﾞ線( 緑)

測定用リード線の断線チェック確認回路

電源アダプタCVケーブル活線劣化診断装置の駆動には、ACアダプタあるいはDCアダプタを用います。
・ACアダプタ
商用電源をDC24V に変換し、測定装置、重畳装置に直流電圧を供給します。ACアダプ
タの主な仕様を表 3.4に示します。

-7-

表 3.4 AC アダプタの主な仕様
項

目

内

容

備

考

入力電圧

AC100V(90～110V)※

50/60Hz

入力周波数

47.5Hz～63Hz―

入力サージ電流30A

Ta=25℃※交流電源は、商用電源、もしくは商用電源と同等の電源品質のものをご使用ください。
・DCアダプタ
内蔵バッテリー（DC24V）から測定装置、重畳装置に直流電圧を供給します。なお、
本DCアダプタには、フル充電（新品）の状態でCVケーブル活線劣化診断装置を連続3時間以上、駆動させることができます。内蔵バッテリーの主な仕様を表 3.5に示します。また、充電回路の主な仕様を表 3.6に示します。

表 3.5 内蔵バッテリーの仕様
項

目

内

電

圧

DC24V

DC12Vの電池を2直列で使用

容

量

7.2Ah

―

鉛蓄電池

密閉形

タイプ

容

備

考

表 3.6 充電回路の主な仕様
項

目

内

容

備

考

入力電圧

AC100V

50/60Hz

充電方式

定電圧・定電流充電方式―

定電圧レベル

DC28V

サイクルユース用

定電流レベル

1.0A（平均）

―

充電時間

8時間以内

―

-8-

3.2 測定フローチャート
測定フローチャートを以下に示します。
測

定

開

始

不平衡充電電流（Iac）

迷走電流（Idc）

シース絶縁抵抗（Rs）

交流電圧の重畳（101/121Hz）

重畳電圧印加交流電流波形抽出処理

交流重畳電流の算出（Isa）

劣

化

判

定

測定条件印字出力
測定結果印字出力
測定電流波形グラフ印字出力
FFT結果グラフ印字出力
測定データ保存（選択）

測

定

終

-9-

了

4

各部の名称
4.1 測定装置

1-5

1-6
1-3

1-2

1-1

図 4.1
1-1

1-4

測定装置操作パネル図

キー入力
操作に必要な設定は、このキーで行います。

1-2

信号入力
重畳装置の信号出力端子から信号ケーブルにて接続します。

1-3

ブザー
測定終了時、キー操作時、異常発生時等に音を発します。

1-4

プリンタ
測定条件、結果を印字出力するための小型サーマルプリンタです。

1-5

液晶表示器
対話形式による操作の手順、および測定の経過等を示します。

1-6

USBコネクタUSBメモリを挿入して測定結果を保存します。

- 10 -

4.2 重畳装置

2-1

2-6

2-5

図 4.2
2-1

2-2

2-4

2-3

重畳装置操作パネル図

測定用リード線入力端子
被測定ケーブルへ接続します。
電磁接触器が動作時は、〔閉〕（緑色）のランプが点灯します。
電磁接触器が解放時は、〔開〕（赤色）のランプが点滅します。

2-2

電磁接触器
電磁接触器の開閉状態をLEDランプにて表示します。

2-3

信号出力
測定装置の信号入力端子と信号ケーブルで接続します。

2-4

電磁接触器テストスイッチ
内蔵の電磁接触器の手動による動作試験用です。
〔開〕スイッチを押すと電磁接触器が開放されます。
〔閉〕スイッチを押すと電磁接触器が動作し短絡状態になります。
通常は、測定装置が自動的に動作試験を行います。

2-5

筐体アース
重畳装置の筺体アースです。

2-6

電源入力
電源アダプタ(DCアダプタあるいはACアダプタ)に直結されている電源ケーブ
ルで接続します。

- 11 -

4.3 電源アダプタ（AC アダプタ）

図 4.3
3-1

AC アダプタ操作パネル図

電源スイッチ
このボタンを押すと照光式の LED が点灯し、CV ケーブル活線劣化診断装置が駆動します。

3-2

電源出力
電源ケーブルを重畳装置の電源入力へ接続します。

3-3

AC100V
電源コードを用いて商用電源（AC100V 50/60 Hz）に接続します。

3-4

ヒューズ
過電流（AC100V）によってヒューズが動作した場合、電源回路を遮断します。

- 12 -

4.4 電源アダプタ（DC アダプタ）

図 4.4
4-1

DC アダプタ操作パネル図

電源出力
電源ケーブルを重畳装置の電源入力へ接続します。

4-2

電源スイッチ
このボタンを押すと照光式の LED が点灯し、CV ケーブル活線劣化診断装置が駆動します。

4-3

AC100V
内蔵のバッテリーを充電する際に使用します。電源コードを用いて商用電源（AC100V 50/60Hz）に接続します。接続した後に充電が始まります。

4-4

ヒューズ
過電流（AC100V）によってヒューズが動作した場合、電源回路を遮断し
ます。

4-5

充電
バッテリーを充電中に状態を表示します。充電中は赤の点灯、終了時は緑の点灯となります。

4-6

バッテリーチェック右側のボタンを押すと、バッテリーの充電状態が表示されます。

- 13 -

5

測定手順
測定時の基本構成を 図 5.1に示します。測定は三相一括で実施してください。被測定ケ
ーブルの接地は、測定時、片端接地にする必要があります。

図 5.1 測定時の基本構成- 14 -

準備

5.1.1a) 保安施設の設置
一般の人が測定域内に入らないように、必ず保安施設を設置してください。
b) CVケーブル活線劣化診断装置の設置
平坦な場所に設置してください。
c) 測定前の点検
各装置（測定装置、重畳装置および電源アダプタ［ACアダプタあるいはDCアダプタ］）、
測定用リード線、信号ケーブル、電源ケーブル等に有害な損傷のないことを確認します。
d) CVケーブル活線劣化診断装置のセット①

測定装置、重畳装置および電源アダプタの蓋を開けます。

②

使用する電源アダプタの〔電源〕スイッチが押されていないことを確認します。

③

重畳装置のアース端子には、被測定ケーブルの接地とは別の接地に接続してください。別の接地が得られない場合は、何も接続しないでください。

④

信号ケーブルで測定装置と重畳装置を接続します。

⑤

ACアダプタで駆動させる場合は電源コードを接続し、商用電源につなぎます。電源ケーブルで重畳装置と接続します。
DCアダプタで駆動させる場合は、バッテリーの充電状態を確認してください。電源ケーブルで重畳装置と接続します。DCアダプタは、スイッチON後、約3秒後に電源（DC 24V）が供給されます。

⑥

電源アダプタの〔電源〕スイッチを押して、電源をONにします。

⑦

測定装置の自己チェック機能が動作しますので、液晶表示画面で異常メッセージの有無を確認します。

⑧

測定装置を操作して、各種設定をキー入力します（測定装置の操作方法については後述します）。

⑨

液晶表示画面でコネクタ接続確認の指示画面になった後、測定用リード線を重畳装置に接続します。

⑩

重畳装置のパネル面、電磁接触器〔閉〕（緑色）のランプが点灯していることを確
認します。

⑪

対象のケーブルが目的の被測定ケーブルであることを確認します。

⑫

測定用リード線が重畳装置に接続されていることを確認した後、被測定ケーブルの接地線に、先端の接続金具（ 図 5.2）を用いて測定用リード線を接続します。接続
金具の配列はグリップ部の表示に従い（ 図 5.3）、測定中に外れないように確実に固定して下さい。専用の端子台が設置されている場合は、測定用リード線先端の圧着端子を使って接続してください（ 図 5.4）。

⑬

・ グリップ部 赤色

：

ケーブル側

・ グリップ部 緑色：

接地側

既設接地線に測定用リード線が接続されているのを確認し、エンターキーを押します。測定用リード線断線チェックが開始されます。

⑭

測定用リード線断線チェック終了後、液晶表示画面の指示により、接続金具間の既設接地線を開放します。専用の端子台が設置されている場合は、短絡バーを外します（図 5.4）。

- 15 -

図 5.2

図 5.3

接続金具

接続金具の配列- 16 -

接地

被測定ケーブル

接地

被測定ケーブル

短絡バー
測定用リード線(接地)

測定用リード線(信号)

測定用リード線(接地)（1）入出力を短絡

図 5.4

測定用リード線(信号)（2）入出力を解放
端子台での接続例- 17 -

5.1.2

測定①

エンターキーを押して測定を開始します。

②

測定が開始されると、重畳装置の電磁接触器〔閉〕（緑色）のランプが消灯し、〔開〕（赤色）のランプが点滅します。

③

測定中の進行状況の目安が、測定装置の液晶表示器に表示されます。交流電圧を重畳しているときは、‘＋交流重畳中＋’のメッセージが表示されます。

④

測定が終了するとブザーが鳴り、重畳装置の電磁接触器〔開〕（赤色）のランプが消灯し、〔閉〕（緑色）のランプが点灯します。

※

測定時、被測定ケーブルで地絡事故が生じた場合、自動的に重畳装置内の電磁接触器が動作し測定を中断します。以降の測定を中止し、CVケーブル活線劣化診断装置の点検を弊社にご依頼ください。

5.1.3
①

片づけ
重畳装置の電磁接触器〔閉〕（緑色）のランプが点灯していることを確認した後、既設接地線の復旧を行います。専用の端子台を使用した場合は、短絡バーを取り付けます。測定用リード線を既設接地線から取り外します。

②

測定用リード線を重畳装置から外します。この際、測定用リード線が被測定ケーブ
ルの接地線から外れていることを確認します。

③

電源アダプタの〔電源〕スイッチを押して、電源をOFFします。

④

その後、測定用リード線、信号ケーブル、電源ケーブル、電源コード（ACアダプ
タ使用時）を取り外し、片づけをします。

⑤

最後に各装置を撤去します。

- 18 -

6

操作方法
CVケーブル活線劣化診断装置の操作は、液晶表示器との対話形式によるキー入力で、簡単に行うことができます。

6.1 キーの説明a） キーの構成

7

キーの構成を図 6.1に示します。

8

9
戻る

b） キーの機能

4

各キーの機能を表 6.1に示します。

5

6

F
フィード

1

2

3

M
モード

0

C

R

E

クリア

リセット

エンター

図 6.1

キ－構成図

表 6.1 キー機能表
キー

機

能

説

明

設定値の入力、項目の選択等に［0］～［9］
［

］

(戻る)

数値入力
前画面への
戻り

使用します。
前の入力画面に戻ります。

リピート

クリック音

可

有

不可

有

－

無

不可

有

可

有

不可

有

可

有

押している間、プリンタ用紙を［F］

プリンタ紙送り(フィード)

送り出します。
(注)印字中はできません。
測定条件の設定をスキップし、

［M］

設定先送り

(モード)

復帰

先に進めます。
エラーメッセージの表示を通常状態に復帰します。

［E］(エンター)

入力確定

［R］

システム

(リセット)

リセット

［C］

入力クリア

(クリア)

設定の確定等に使用します。
システムをセットし直すために設定の最初の画面に戻ります。
測定中は測定を中断します。
数値入力の訂正、前設定の表示、設定値の初期化に使用します｡- 19 -

6.2 システム異常処理
CV ケーブル活線劣化診断装置は異常検出時、自動的に復旧するように設計していますが、人の判断、操作を必要とする場合は、LCD 画面にその内容を表示し、操作の要求を促します。また、異常に対する復旧が出来ないときは、装置全体を初期化して、起動状態に戻します。この時、設定されていたデータは、全て工場出荷時の設定になります。

6.2.1

プリンタ異常画面 1＊＊＊＊

プリンタ異常＊＊＊＊プリンタ接続部の異常を検出しました。取扱説明書を確認下さい。

＜異常内容＞
プリンタの異常です。
＜対応＞
プリンタをチェックして、異常がなければ［Ｅ］を押してください。

6.2.2

プリンタ異常画面 2＊＊＊

プリンタ用紙切れ＊＊＊

プリンタ用紙がなくなりました。
新しい用紙をセットして下さい。

＜異常内容＞
プリンタの用紙切れです。
＜対応＞
プリンタに新しい用紙をセットして、［Ｅ］を押してください。

6.2.3

電源異常画面(DC アダプタ使用時)
＊＊＊＊＊

ＢＡＴＴ

＊＊＊＊＊

バッテリーがありません。
バッテリーを充電して下さい。

＜異常内容＞
電源のバッテリー残量が残り少なくなっています。表示中は 1 秒おきに“ピッ”とブザ－音がします。
＜対応＞
測定中バッテリー残量が少ない場合、自動的に測定を終了します。重畳装置の電磁接触器[閉]（緑色）のランプが点灯していることを確認してください。電源を切って充電済みの DC アダプタと交換してください。

- 20 -

6.2.4

測定用リード線断線異常画面
＊＊＊＊＊

ＳＴＯＰ

＊＊＊＊＊

リード線断線
リード線の端子の接続状態を確認して下さい。
＜異常内容＞
測定用リード線の断線が検出されました。
＜対応＞
測定用リード線の接続状態を確認し、断線状態を解消してください。その後、［M］
キーを押し、画面を復帰させます。

6.2.5

重畳装置過電流異常画面
＊＊＊＊＊

ＳＴＯＰ

重畳装置

＊＊＊＊＊

過電流

測定回路の過電流を検出しました。
取扱説明書を確認下さい。
＜異常内容＞
測定中に、重畳装置で過電流が検出されました。
＜対応＞
各機器の接続状態、測定状態等を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。

6.2.6

重畳装置過電圧異常画面
＊＊＊＊＊

ＳＴＯＰ

重畳装置

＊＊＊＊＊

過電圧

リード線端子間の過電圧を検出しました。取扱説明書を確認下さい。
＜異常内容＞
測定中に、重畳装置で過電圧が検出されました。
＜対応＞
各機器の接続状態、測定状態等を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。

- 21 -

6.2.7

測定装置過電流異常画面
＊＊＊＊＊

ＳＴＯＰ

測定装置

＊＊＊＊＊

過電流

測定回路の過電流を検出しました。
取扱説明書を確認下さい。
＜異常内容＞
測定中に、測定装置で過電流が検出されました。
＜対応＞
各機器の接続状態、測定状態等を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。

6.2.8

保護回路設定異常画面
＊＊＊＊＊

ＳＴＯＰ

＊＊＊＊＊

保護回路設定異常
保護回路設定以上の電流を検出しました。取扱説明書を確認下さい。
＜異常内容＞
測定中に、測定装置で設定以上の過電流が検出されました。
＜対応＞
保護回路の設定および各機器の接続状態、測定状態等を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。その後、保護回路の設定を変更し、再度測定します。

6.2.9

重畳部過電流異常画面
＊＊＊＊＊

ＳＴＯＰ

重畳部

＊＊＊＊＊

過電流

重畳電源部で過電流を検出しました。取扱説明書を確認下さい。
＜異常内容＞
測定中に、重畳電源部で過電流が検出されました。
＜対応＞
各機器の接続状態、測定状態等を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。

- 22 -

6.2.10 重畳部過電圧異常画面
＊＊＊＊＊

ＳＴＯＰ

重畳部

＊＊＊＊＊

過電圧

重畳電源部で過電圧を検出しました。取扱説明書を確認下さい。
＜異常内容＞
測定中に、重畳電源部で過電圧が検出されました。
＜対応＞
各機器の接続状態、測定状態等を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。

6.2.11 重畳部温度異常画面
＊＊＊＊＊

ＳＴＯＰ

重畳部

＊＊＊＊＊

温度異常

重畳電源部の温度異常を検出しました。時間をおいて測定して下さい。
＜異常内容＞
測定中に、重畳電源部で異常な温度が検出されました。
＜対応＞［M］キーを押すことで画面が復帰します。重畳電源部の温度を下げるため、時間をおいてから、再度測定をして下さい。

6.2.12 重畳電圧の検出異常画面
＊＊＊＊＊

ＳＴＯＰ

＊＊＊＊＊

重畳電圧出力異常
重畳電圧出力の異常を検出しました。
取扱説明書を確認下さい。
＜異常内容＞
重畳すべき電圧値に対して過不足が検出されました。
＜対応＞
各機器の接続状態、測定状態等を確認してください。DC アダプタを使用している
場合、バッテリー電圧が低下している可能性があります。バッテリーチェックスイ
ッチを押し、バッテリー電圧を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。

- 23 -

6.2.13 電磁接触器の動作異常画面
＊＊＊＊＊

ＳＴＯＰ

＊＊＊＊＊

電磁接触器が動作しました
電磁接触器の異常動作を検出しました。
取扱説明書を確認下さい。
＜異常内容＞
電磁接触器が動作をしました。なお、システムの保護回路が動作した場合は、このメッセージは表示されません。
＜対応＞
各機器の接続状態、測定状態等を確認してください。異常がなければ［M］キーを押すことで画面が復帰します。

6.2.14 USB メモリ異常画面＊＊＊

ＵＳＢメモリ異常＊＊＊ＵＳＢメモリにデータを保存できま
せん。ＵＳＢメモリが正しく挿入されているか確認下さい。
＜異常内容＞USB メモリが正常に接続されていないか、USB メモリに異常があるため、データの書き込みができません。USB メモリがフォーマットされていない場合も、この画面が表示されます。
＜対応＞
USB メモリを確認して、異常が無ければ［Ｅ］キーを押してください。

6.2.15 USB メモリ空き容量不足画面＊＊

ＵＳＢメモリ容量不足＊＊

ＵＳＢメモリの容量が不足しています。保存データを確認して下さい。

＜異常内容＞USB メモリの空き容量が足りないため、データの書き込みができません。
＜対応＞
空き容量のある USB メモリと交換して、［Ｅ］キーを押してください。

- 24 -

6.2.16 システム異常画面＊＊＊＊

システム異常Ｎｏ＝０００１＊＊＊＊

ＣＤ＝００００

取扱説明書を確認下さい。

＜異常内容＞
電源投入時、装置の自己診断中に異常を検出すると表示されます。”Ｎｏ．”および、“CD”の番号により異常内容を判別します（ 表 6.2参照）。

表 6.2 システム異常と内容
No.

CD

0001

0000

0002

0000

0003

0000

異

常

内

ROM メモリチェック異常

容装置に異常があります。
システムを停止します。

RAM データバックアップ

設定値デ－タを初期値に戻し

異常

処理は続けます。
LCD 装置に異常があります。

LCD チェック異常

処理は続けます。

対応
要修理
測定可
要修理

プリンタ内のレバーが、上が
0004

0001

プリンタセット異常っています。レバーを下げる

測定可

ことで処理は続けます。
プリンタに用紙がセット
0004

0002

プリンタ用紙なし検出

されていません。

測定可

処理は続けます。
0005

0000

ROM サムチェック異常

装置に異常があります。
システムを停止します。

要修理

6.2.17 処理中メッセ－ジ 1＃

プリンタ印字中です＃＜内容＞
プリンタ印字バッファが一杯になっています。このメッセ－ジ表示中は、入力を受け付けませんのでしばらくお待ちください。印字バッファに空きができると復帰します。

- 25 -

6.2.18 処理中メッセ－ジ 2
＃

測定中断

＃

しばらくお待ちください＜内容＞
測定を中断すると表示されます。
（［Ｒ］キ－による中断）中断処理が終了すると復帰します（約 10 秒以内）。

6.2.19 自己チェック異常画面
＊＊＊＊＊

ＳＴＯＰ

＊＊＊＊＊

リミットチェック異常取扱説明書を確認下さい。

＜異常内容＞
内部にある 1GΩ の抵抗を測定し、測定結果が設定範囲外にあります。
＜対応＞
装置の点検が必要です。点検を依頼してください。

6.2.20 非常停止
＊＊＊＊＊

ＳＴＯＰ

＊＊＊＊＊

非常停止しました。
回路を確認して下さい。

＜機能＞測定を中断させる場合、[Ｒ]キーを押します。
測定を中断し、電磁接触器が動作し閉状態となります。その後、下記の表示となります。この間に、“測定中断終了しました”と印字をします。

＜キ－操作＞
・[Ｒ]システム設定の最初の画面に戻ります。

終了する際は電源をＯＦＦにします。

表 6.3 にエラーメッセージの一覧表を示します。

- 26 -

表 6.3 エラーメッセージ- 27 -

一覧表

6.3 プリンタ用紙および注意事項a） プリンタ用紙の仕様
感熱ロール紙の仕様を、表 6.4 に示します。
CV ケーブル活線劣化診断装置のプリンタには、表 6.4 の感熱ロール紙あるいは同等品を使用してください。

表 6.4 感熱ロ－ル紙の仕様
項目内容
TP-201C/TP058-18C
型
式
メ － カ
小林クリエイト㈱
58mm
横
幅
38mm
直
径
8mm
芯 空 径

b） 感熱ロ－ル紙の取扱注意事項
感熱ロ－ル紙は、熱化学反応で発色する特殊紙です。未使用、記録済みを問わず、熱、湿気、光等を嫌いますので下記の注意事項に気をつけてください。
①
汗ばんだ手で触れると指紋が付いたり、記録がぼやけることがあります。
②③④

固いもので強く擦ると発色することがあります。
糊付け時には、一般に水性の糊をご使用ください。口紅タイプの糊が適当です。
粘着テ－プは感光紙を発色させることがあるので、裏面を両面テ－プ等で止めてく

⑤
⑥
⑦

ださい。
塩ビフィルムに長時間接触させると、記録部が変色しますので避けてください。
有機溶剤に接触させると発色することがあります。
保管場所は、なるべく乾燥した冷暗所に保管してください。

⑧

記録済みの感熱紙についてはコピ－（乾式）を行い、コピ－したものを保管するよ
うにしてください。

- 28 -

6.4 USB メモリの取り扱い
6.4.1 USB メモリ仕様a）記録メディアUSB メモリb）適合規格USB2.0c）データ保存ファイル名

MMDDHHMM.csv（測定開始月、日、時、分）d）データファイル容量

約 6kB／ファイルe）格納場所

ルートディレクトリ

操作方法

6.4.2

a）測定終了後、USB メモリにデータ保存をする／しないを選択するメッセージが表示されます。
「保存する」を選択すると、データ保存ファイル名を自動で取得し、USB メモリに測定データを保存します。ファイル名が重複する場合は、自動的に上書きされます。
b）USB メモリを抜き取った後は、防塵のため、付属の USB コネクタキャップを嵌めてく
ださい。

注意事項

6.4.3

a）USB メモリを挿入したまま測定装置の蓋を閉じた場合、USB メモリと蓋が干渉する場合があります。USB メモリ及び装置の破損、故障の原因となりますので、測定装置の蓋を閉じる際は必ず USB メモリは抜き取って下さい。
b）USB メモリは、フォーマット済みのものを使用してください。CV ケーブル活線劣化診断装置では、フォーマットできませんので、予めパソコンでフォーマットしてください。
c）USB メモリは、大量のデータを蓄積したままにすることは避けてください。静電気などでデータを破損・消失する可能性があります。保存したデータは、早めにパソコン等へ移すことをお勧めします。
d）USB メモリは、外部から静電気や衝撃等が加わると、故障および内部のデータを破損・消失する可能性がありますので、取り扱いには十分注意してください。特に以下の事項には注意してください。
・

データの保存中に、USB メモリを抜き取らないでください。

・

データの保存中に、装置の電源を切らないでください。

・

データの保存中に、外部より強い衝撃や振動を与えないでください。

・USB メモリを保管する場所は、静電気、水気、埃のある所は避けてください。

・USB メモリには寿命がありますので、長期間使用するとデータの保存、再生ができな
くなります。寿命は、使用する環境によって変化します。

e）使用可能なメモリ容量は 512MB～64GB です。
f）暗号化機能のついた USB メモリ、ウイルス駆除機能のついた USB メモリなど特殊機能が付加されたものは使用できません。また、メーカーが独自に付加した高速転送機能な
どは使えない他、場合によっては相性問題が発生する場合があります。

6.4.4

データファイルの内容保存されるデータファイルの内容を表 6.5 に示します。

- 29 -

表 6.5 保存データ
ブロッ ク
ケーブル番号
測定環境

行数
1
2

測定ケーブルの情報

3

測定区間

4

測定結果

測定電流波形データ

5
6
7
8
9
10
11
12
13～44

FFT結果グラフデータ

45～50

Isaの各値

51
52
53
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
72
73
74
75
76
77
78
79
80
81
82

測定値，
パラメータ

FFT変換値

項目
ケーブル番号
天候，
気温，
湿度
ケーブル種類，
ケーブルサイズ，
ケーブル長さ，布設年月
測定端機器種類（データはありません），
測定端機器Ｎｏ（データはありません），
縁端機器種類（データはありません），
縁端機器Ｎｏ（データはありません）
測定時刻
測定方法（データはありません）
判定結果
Isa値
Rs値
ノイズ
Idc値
Iac値
0～5s間のデータ（256ポイント）
8ポイント毎に改行マークを入れる
0.0～2.0Hzのデータ（41ポイント）
8ポイント毎に改行マークを入れる
Isa1max値，Isa1min値，Isa1ave値
Isa2max値，Isa2min値，Isa2ave値
Isa3min値
Isa4min値
Isa5min値
Isa6min値
Isap1値
Isap2値
peekfreq値
Isaj1値，Isaj11値，Isaj21値，Isaj31値
Isaj2値，Isaj12値，Isaj22値，Isaj32値
Isaj3値，Isaj13値，Isaj23値，Isaj33値
Isaj4値，Isaj14値，Isaj24値，Isaj34値
Isaj1pIdcY（V11値，V12値，V11-V12値）
IdcY値
Ixbas（V1値，V2値，V1-V2値）
Ixbas値
IdcBAS値
Rchosei値
Rs値，Rc値
Isa2値，Isa値
Idc値
Es値
Iac値
Vx1（V5値，V6値，検出抵抗の電圧値）
Vx2（V7値，V8値，検出抵抗の電圧値）
Ix1値（第1段），Ix1値（第2段）
Ix2値（第1段），Ix2値（第2段）
Isa値（第1段），Isa値（第2段）
Iac値
adcnt値（0.0～2.0Hzのデータ、周波数毎に改行マークを入れる）
7回分のデータ，7回分の平均，自己相関分×383～123
各値に係数を掛ける
Isa測定時の区間毎のデータ
fftno
S/N比判定指標

Isa測定時の入力波形の124～130 最大値，最小値，最大値－最小値，Isa波形の絶対値，測定順番（７区間分、区間毎に改行マークを入れる）
131
fftno
132
Isaj1pn値

・項目欄の（書式########(-#)
AAA
##.#
##.#
AAAA
####
####
YY/MM
YY/MM/DD hh:mm
AAAAAAAAAAAAA###.###／ＡＡＡＡＡＡ（測定異常時）###.###／ＡＡＡＡＡＡ（測定異常時）######.######.###／ＡＡＡＡＡＡ（測定異常時）###.###／ＡＡＡＡＡＡ（測定異常時）#####.###.#####，##.#####，##.#######.#####，##.#####，##.#######.#######.#######.#######.#######.#######.##########.####.#####，##.#####，##.#####，##.#######.#####，##.#####，##.#####，##.#######.#####，##.#####，##.#####，##.#######.#####，##.#####，##.#####，##.#######.##########.#，###.#，#####.######.#######.#，###.#，#####.######.#######.#######.######.###，######.#######.#，######.########.#########.#######.#######.#，###.#，#####.######.#，###.#，#####.#######.#，######.#######.#，######.#######.#，######.#######.###########.##，##.##，##.##，##.##，##.##，##.##，##.##，##.##，#.####，#.####，#.######.##，##.##，##.##，##.##，##########.####

・1 行あたりに複数のデータが入る場合は“、（カンマ）”で区切ります。
・書式欄の＃は 0 から 9 までの数値を表します。また、A は文字データを表します。
・各値の単位は固定です。
・Isa、Rs、Idc、Iac の各値は測定異常時に文字データが入ります。
注 1）999999.0 以上もしくは-999999.0 以下（リミットオーバー）："OVER_D"：”OVER_A”

ゲインオーバー：”OVER_G”
測定なし：ブランク（空白）- 30 -

文字データ－，
晴れ／曇り／雨（℃），（％）－，
CVT／CV／その他
（ｍ㎡），
（ｍ），
数値 又は "不明"
年月 又は "不明"
－，
－，
－，
－
－，
（nA）
（MΩ）
（nA）
（nA）
（mA）
（nA）

-

）内は補足説明です。

注 2）A/D オーバー

単位

（mV）
（nA）
（mV）
（nA）
（nA）
（MΩ）
（MΩ）
（nA）
（nA）
（mV）
（mA）
（mV）
（mV）

判定不能／良／劣化
注1）、注2）
注2）
注2）
注2）

7

特性
7.1 外観・構造
外観・構造等が本仕様書に適合するか否かを検査します。試験方法8.1を適用します。

7.2 基本機能試験
7.2.1

各種機能
各操作スイッチ、表示、プリンタ印字機能に異常がないことを確認します。試験方法8.2.1を適用します。

7.2.2

交流重畳出力電圧
被測定ケーブルに重畳される電圧が表 7.1に示す範囲であることを確認します。試験方法8.2.2を適用します。

表 7.1 交流重畳出力電圧

7.2.3

電圧

周波数

50 V ± 5 %101/121Hz ± 0.01 Hz

交流重畳出力電圧歪み
重畳電源ユニットから出力される電圧の波形歪みが、0.5%以下（無負荷時）であることを確認します。試験方法8.2.3を適用します。

7.2.4

充電機能（DC アダプタ）
DCアダプタの充電機能に異常がないことを確認します。試験方法8.2.4を適用します。

7.3 動作試験
7.3.1

交流重畳電流（1Hz）
100μA以下での最大誤差が±10%以内であることを確認します。但し、50nA以下では±5nA以内であることを確認します。試験方法8.3.1を適用します。

7.3.2

シース絶縁抵抗
100MΩ以下での最大誤差が±10%以内であることを確認します。試験方法8.3.2を適用し

ます。

7.3.3

過電流保護機能
設定値(850mA)の±10%以内で重畳装置内の電磁接触器が動作することを確認します。

試験方法8.3.3を適用します。

7.3.4

過電圧保護機能
AC70V以上、100V以下で重畳装置内の電磁接触器が動作することを確認します。試験

方法8.3.4を適用します。

7.3.5

測定用リード線の断線チェック機能
測定用リード線の断線チェック機能が正常に動作することを確認します。試験方法8.3.5を適用します。

- 31 -

7.4 強度試験
7.4.1

絶縁抵抗
下記個所の絶縁抵抗が100MΩ以上であることを確認します。試験方法8.4.1を適用します。
・電源コード先端の電源端子～電源アダプタ外箱間・測定用リード線先端の接続端子～重畳装置外箱間および測定用ケーブル入力端金具間・測定用リード線先端の接続端子～測定装置外箱間および信号入力端金具間

7.4.2

耐電圧
下記個所の耐電圧試験（AC1,000V、1分間）を行い、損傷がなく、7.2、7.3に示す特性を満足することを確認します。試験方法8.4.2を適用します。
・電源コード先端の電源端子～電源アダプタ外箱間・測定用リード線先端の接続端子～重畳装置外箱間および測定用ケーブル入力端金具間・測定用リード線先端の接続端子～測定装置外箱間および信号入力端金具間- 32 -

8

試験方法
8.1 外観・構造
外観、構造等が本仕様書に適合するか否かを目視、計量等で検査します。

8.2 基本機能試験
8.2.1

基本機能
図 5.1のように各部装置を接続し、この状態で測定を実施し、各操作スイッチ、表示、プリンタ印字機能等を検査します。

8.2.2

交流重畳出力電圧
図 5.1のように各装置を接続し、測定を開始します。そして、重畳電圧出力中に、接続金具間で交流重畳出力電圧（電圧、周波数）を測定します。

8.2.3

交流重畳出力電圧歪み
図 5.1のように各装置を接続し、測定を開始します。そして、重畳電圧出力中に、接続金具間で交流重畳出力電圧の歪みを測定します。

8.2.4

充電機能（DC アダプタ）
DCアダプタに専用の電源ケーブルでAC100Vを供給し、充電機能等を調査します。

8.3 動作試験
8.3.1

交流重畳電流(1Hz)
図 5.1のように各部装置を接続します。そして、交流電圧（1Hz、1.0Vp-p）と抵抗（10kΩ～100MΩ）によって構成される図 8.1に示した模擬回路に、測定用リード線先端の接続端子を接続します。この状態で、回路に流れる交流重畳電流値(Isa)(振幅値)を測定します。
測定を実施する交流重畳電流値(Isa)は、10nA 、100nA、1μA、10μA、100μAの5点とします。また、シース絶縁抵抗(Rs)の低下時の性能確認として、図 8.2に示した回路でも同様の試験を行います。

8.3.2

シース絶縁抵抗
図 5.1のように各部装置を接続します。そして、標準抵抗(100kΩ～100MΩ)によって構成される図 8.3に示した模擬回路に、測定用リード線先端の接続端子を接続します。この状態で、シース絶縁抵抗Rsを測定します。測定を実施する抵抗値は、100kΩ、1MΩ、10MΩ、100MΩの4点とします。
8.3.3

過電流保護機能
図 5.1のように各部装置を接続します。そして、測定用リード線先端の接続端子間に過電流を流し、重畳装置内の電磁接触器が動作する電流値を求めます。

8.3.4

過電圧保護機能
図 5.1のように各部装置を接続します。そして、測定用ケーブル先端の接続端子間に過電圧を印加し、重畳装置内の電磁接触器が動作する電圧値を求めます。

8.3.5

測定用リード線の断線チェック機能
図 5.1のように各部装置を接続します。そして、測定用リード線先端の接続端子間を開放します。測定用リード線の断線チェック動作後に液晶表示画面に****STOP****‘リード
線断線’が表示され、測定開始ができないようにプロテクトされていることを確認します。

- 33 -

分流阻止用コンデンサ（1[μF]）標準抵抗10[kΩ]～100[MΩ]

測定用リード線

CVケーブル活線劣化診断装置
DISC-20-AS

交流安定化電源
1[Hz] 1.0[Vp-p]

模擬ケーブル
コンデンサ（1[μF]）

模擬シース絶縁抵抗
250[kΩ],500[kΩ]

図 8.1 交流重畳電流の確認用模擬回路 1
標準抵抗10[kΩ]～100[MΩ]

CVケーブル活線劣化診断装置
DISC-20-AS

測定用リード線
交流安定化電源
1[Hz] 1.0[Vp-p]

模擬ケーブル
コンデンサ（1[μF]）

図 8.2 交流重畳電流の確認用模擬回路 2

標 準 抵 抗 ： １ ０kΩ ～ １ ０ ０MΩ
標準抵抗：100[kΩ]～100[MΩ]

ＣＶケーブル活線劣化診断装置
CVケーブル活線劣化診断装置
DISC-20-AS
ＤＩＳ
Ｃ－１０Ｇ－ＡＳ

測定用リード線
測定用リード線

図 8.3 シース絶縁抵抗の確認用模擬回路
8.4 強度試験
8.4.1

絶縁抵抗
図 5.1のように各部装置を接続します。そして、絶縁抵抗計（500V）で7.4.1に示した箇所の絶縁抵抗を測定します。

8.4.2

耐電圧
図 5.1のように各部装置を接続します。そして、適当な電圧発生装置を使用し、AC1,000

Vを発生させ、7.4.2に示した箇所に印加します。

- 34 -

9仕様一覧
9.1 測定装置(1) 測定機能a) 交流重畳電流（Isa）

1nA～100μAb) シース絶縁抵抗（Rs）

100kΩ～100MΩ

c) 迷走電流（Idc）

1nA～100μA（参考値）

d) 不平衡充電電流（Iac）

1mA～850mA（参考値）(2) 測定時間
約22分(3) 液晶表示
表示文字数

16 文字×4 行

表示文字サイズ7.4mm(W)×7.4mm(H)

表示サイズ

132mm (W) × 39.0mm (H)(4) サーマルプリンタ

24×24 ドットマトリックス

16 桁(5) USB ポート

USB2.0 対応(6) 寸法(最大)350mm(W) × 230mm(D) × 345mm(H)(7) 質量約 12kg9.2 重畳装置(1) 交流重畳電圧
電圧

50V

周波数

101/121Hz(2) アレスタ（フェイル・セイフ形）
アレスタ1
放電開始電圧DC 230V±20%

放電耐量

AC 10A（1回限り）

アレスタ2
放電開始電圧DC 90V±20%

放電耐量

AC 5A、1秒(3) 電磁接触器
接点定格

AC 500V、100A

耐量

AC 1kA、1秒

耐圧

AC 2,200V、1分間(4) 保護機能a) 過電流保護

AC 850mA以下b) 過電圧保護AC 70V以上100V以下(5) 寸法(最大)350mm(W) × 230mm(D) × 345mm(H)(6) 質量約 19kg- 35 -

9.3 電源アダプタ：AC アダプタ
(1) 電源AC 100V(50/60Hz)(2) 寸法(最大)175mm(W) × 230mm(D) × 345mm(H)(3) 質量約 9kg

9.4 電源アダプタ：DC アダプタ
(1) 電源DC24V(2) 容量DC12V 7.2Ah(2 直列にて使用)(3) 連続使用時間

3 時間以上(新品、フル充電時)(4) 充電用電源

AC 100V(50/60Hz)(5) 充電時間

8 時間以内(6) 寸法(最大)175mm(W) × 230mm(D) × 345mm(H)(7) 質量約 14kg＊内蔵バッテリーの寿命は、充放電回数 200 回あるいは 1 年以内の短い方- 36 -

本

社：〒 108-8333 東京都港区三田 1-4-28 三田国際ビル 17Ｆ
TEL 03(3455)5661

販

売

企

画

部：〒 410-8515 静岡県沼津市大岡 2771
TEL 055(924)2274

矢崎総業北海道販売㈱：〒 062-8532 札幌市豊平区福住 3 条 2-4-5
TEL 011(852)2912
東 北 電 線 営 業 部：〒 984-8510 仙台市若林区大和町 4-17-1
TEL 022(284)9112
関 東 電 線 第 二 営 業 部：〒 108-0075 港区港南 1-8-15 W ビル 6F
TEL 03(5782)2711
中 部 電 線 営 業 部：〒 465-0095 名古屋市名東区高社 2-252
TEL 052(769)1531
北 陸 電 線 営 業 部：〒 939-8211 富山市二口町 1 丁目 1-2
TEL 076(492)8606
関 西 電 線 営 業 部：〒 553-0003 大阪市福島区福島 3-1-46
TEL 06(6458)5231n
中 国 電 線 営 業 部：〒 732-0045 広島市東区曙 3-1-12
TEL 082(568)7801
矢 崎 総 業 四 国 販 売 ㈱：〒 760-0080 高松市木太町 1925-1
TEL 087(833)3334
九 州 電 線 営 業 部：〒 812-0042 福岡市博多区豊 1-10-68
TEL 092(411)4832- 37 -