Temperature measurementfrom - 50° C up to + 3000° C

Up to 90° Scan angle
Example: 5 m scan length
at 2.5 m distance

Application
specific solutions

Dual mode - stand alone
and remote control

We keep your temperature under control

Line-Scanner
Temperature monitoring inpaper finishing

LS12
Precise thermal control inglass production

Accurate temperaturemonitoring in environmental
applications

Precise Temperature
Distribution
Measurement

The versatile Line-Scanner for temperature
measurement from -50° C to +3000° C
M The Line-Scanner
Ideal for temperature measurement ofmoving targets. Applications range fromfrozen food or plastic foil to sheet rock or
glass materials.
Adaptable: Various combinations ofscanner and radiation pyrometer allow
us to offer unique solutions for every
customer application. This custom
solution is specific to each application.
Service: All materials will be tested in
our application laboratory. We will find asolution which fits each customer’srequirements.
Compact: The Line-Scanner integrates
into production without problems. The
metal housing is rugged and fulfills all
relevant CE requirements.

The Line-Scanner measurestemperature along a line. Up to 250
measuring points can be processed on
the chosen measuring length. EFlexible: The Line-Scanner operates in
two modes.
• Stand alone - using a downloaded
configuration, the system runs on itsown.
• Remote control in combination with aPC; control and signal processing are
performed online via interface.

Our ScanPerfect software is Windows based (Windows 95/98/NT4.0), with the PC
and line scanner connected via serial interface. The scan length (scan angle), thescanning speed and the number of measuring points are all programmable.
The software displays and stores the data. The true target temperature is displayed
thanks to the table of emissivity versus measuring position utilized by ScanPerfect.
M The ScanPerfect Software
Versatile: the ScanPerfect softwarearchitecture allows for optional
modification of the algorithm andhardware outputs.
Secure, thanks to online watchdog
monitoring: ScanPerfect reports errors
and disruptions.
The measurement data may be exported
as an ASCII-file. H

E
The results will be
continuously
displayed in realtime.

F The Radiation Pyrometer
KT15D is part of the LineScanner which is available in twoversions. The standard version isavailable for environmental
temperatures up to 60° C
and a heavy duty version
is available for harshenvironments up to 220° C.
The false color display shows an overview of the temperature
distribution. HG The measurement result will be
individually processed and displayed
depending on the programmed options.
Signal processing and signal compression areoptionally available.
F The Line-Scanner can be programmed via PC and themeasuring data can be stored as an ASCII file.

M Many applications
M The Output
Graphics: In remote control mode
the PC handles the control and datastorage. The measuring values aredisplayed in the form of line diagram,
3-D graphics or other display modes.
Variable: In stand alone mode two
signals are provided for position andtemperature values. Processing of thesesignals is performed by the end user.

EUse in processcontrol.

The modular construction of the LineScanner enables the system to run itself
in a variety of applications.
For example:

E
Precise temperature
measurement infood production.

F
Reliable temperature
control in metal
processing…

E
Accurate point
measurement inpainting
applications.

F… and in plastic foil
production.

Analog: In both remote and stand-alone
modes the Line Scanner provides theanalog output of the temperature values.

Line-Scanner LS12-Series
Selction Guide
Model

Spectral Response µmT min …Tmax ° C

Application/Material

LS12.01LS12.21LS12.23
LS12.24
LS12.25
LS12.41
LS12.42LS12.43
LS12.69
LS12.82

2.00 ... 2.70
3.43 ± 0.15
6.80 ± 0.15
7.93 ± 0.15
8.05 ± 0.15
3.90 ± 0.10
4.90 ... 5.50
7.50 ... 8.20
x ... y8 ... 14300… 2400
80… 350
0… 4000… 4000… 400
250… 2500
100… 2500
0… 2500
400… 2500-50… 1000

metals, metal oxides, ceramics, glass volume
plastic film with CH-band, coating materials (oil, paints)
thin film plastics, e.g. PE, PP, PVC
thin film plastics, e.g. PET, PA, fluor carbon
thin film plastics, e.g. PTFE, PET, PVC
glass volume, measurements through hot gases and flames
glass (processing), quartz
glass (thin plates), quartz, ceramics
hot gases in incinerators and fossil fuel fired utility boilers, rotary kiln
paper, textiles, rubber, wood, ceramics, thicker plastics (>1mm),
painted or coated surfaces, asphalt, building materials,
electronic components, food, liquids
meteorological, biological, agricultural studies

LS12.85
9.6 ... 11.5
-25… 200
LS12.99 other spectral and temperatures ranges are available

General Specifications
Temperature range
Temperature resolution (NETD)*

Depends on model, minimum and maximum measuring temperature, see table above
Depends on model, measuring temperature and response time;
typical value ± 0.2° C
Accuracy± 0.9° C ± 0.9% of the difference between target temperature and housing temperature
Long-term stability
Better than 0.01% of the absolute measured temperature in Kelvin/month
Scanning angle
90° maximum scanning angle is freely programmable
Angle resolution
0.057°
Scanning speed
up to 90°/s
Measuring points
250 measuring points per programmed angle range
Aiming options*
Several optical and mechanical options are available, e.g. laser pointerEmissivity setting
Adjustable from 0.1 to 1.0, programmable in 0.001 increments.
Individual programmable for every measuring point
Analog output
4 scalable output signals; linear voltage or current
0...1 V or 0...10 V or 0...20 mA or 4...20 mA (programmable)
Serial interfaceRS-232 interface, bi-directional, baud rate 9,600 up to 57,600,for free programming and data transfer
Power requirements
22...30 VDC or 24 VAC ± 10%, 48...400 Hz; # 250 mA with 24 VDC
Permissible ambient temperature
0° C up to 60° C with protective and coolable housing 220° C (water cooling with 20° C and waterpressure 1 l/min with max. 6 bar)
Storage temperature- 20° C ... 70° C
Air purge
Airflow rate 3...5 m 3/h at 0.3...0.5 bar (industrial air)
Housing protection, weight
IP 65 (DIN 4005) NEMA 4 equivalent, approx. 2 kg*) Please ask for further data of Infrared Radiation Pyrometer KT15D-Series (temperature ranges, temperature resolution, field of view):"Technical Data“, "Field of View Diagrams“, "Options and Accessories“.

05.2000/030/e/D&F/ Wiehr- Design

Housing dimensions in mm (inches)

HEITRONICS Infrarot Messtechnik GmbH • Kreuzberger Ring 40 • D-65205 Wiesbaden • Tel. ++49 (0)611 973 930
Fax ++49 (0)611 973 93 26 • e-Mail HEITRONICS@T-Online.de • http://www.heitronics.com