Rotem Meter View ｿﾌﾄｳｴｱ (RMV)
Version 2.05

産業科学株式会社

Rotem Meter View Software

PRIR42X9.DOC

Page 1

目次序章....................................................................................................................................................... 3使用環境............................................................................................................................................... 3インストール....................................................................................................................................... 3
起動方法............................................................................................................................................... 4セットアップ....................................................................................................................................... 4
5.1
PC 用 COM ポートの設定 .......................................................................................................... 4
5.2
Excel レポートフォームの設定................................................................................................. 5
パラメータの設定....................................................................................................................... 5
5.3
閾値パラメータの設定....................................................................................................... 5
5.3.1
5.3.2 Lost Contact Interval の設定 .................................................................................................... 6
5.3.3 Audio Alarm の設定 ................................................................................................................. 6
5.3.4 Auto Arrange Form の設定 ...................................................................................................... 6
オンラインインターバルの設定............................................................................................... 65.4
6 PC へのデータダウンロード ............................................................................................................. 6
オフラインモード....................................................................................................................... 66.1
オンラインモード....................................................................................................................... 86.2
7 モニター用グラフとリスト............................................................................................................... 8
グラフ画面................................................................................................................................... 87.1
ズーム........................................................................................................................................... 97.2
パラメータ................................................................................................................................... 97.3
オールデータ............................................................................................................................... 97.4
リスト........................................................................................................................................... 97.5
7.6
Online/offline .............................................................................................................................. 108 機能..................................................................................................................................................... 108.1
Excel ファイルへの保存........................................................................................................... 10
時刻の設定................................................................................................................................. 11
8.2
キャリブレーションデータの取り込み................................................................................. 11
8.3
9 Data コマンド..................................................................................................................................... 119.1
Start Online ................................................................................................................................. 11
9.2
Get offline data............................................................................................................................ 119.3
Get Legend .................................................................................................................................. 11
9.4
Get Calibration Data ................................................................................................................... 11
10 Setting コマンド................................................................................................................................. 1110.1 Online Interval............................................................................................................................. 1110.2 Communication Port ................................................................................................................... 1110.3 Parameters ................................................................................................................................... 11
11 Utility コマンド.................................................................................................................................. 1211.1 Save ............................................................................................................................................. 1211.2 Send Time ................................................................................................................................... 1211.3 Graph........................................................................................................................................... 1211.4 Test .............................................................................................................................................. 12
トラブルシューティング............................................................................................................. 121212345

Rotem Meter View Software

PRIR42X9.DOC

Page 2

1 序章
Rotem Meter View Software (RMV) は Windows 95/98 をベースとしたプログラムで
ROTEM サーベイメータから PC へデータをダウンロードするのに使用します。
この使用できる測定器類は RAM ION、 TelePole、 RAM DA-2000 になります。
測定器と PC 間のコミュニケーションには２種類のモードがあります：
オンライン – ユーザーによって設定されたインターバル時間で、測定器から PC にデータをダウンロードする。
オフライン – データは測定器内部に保存され (347 ﾃﾞｰﾀﾚｺｰﾄﾞ)、後に PC にダウンロードする。

2

使用環境
WINDOWS 95/98 対応 PC、 RS-232 コミュニケーションポート付
ROTEM 社より提供されたコミュニケーションケーブル
ROTEM 社製ポータブルサーベイメータ
ROTEM 社によって供給された RMV ソフトウエア
Excel ソフトウエア(必要な場合のみ)

3

インストールCD ドライブに CDR を挿入します。setup.exe ファイルを選択し、ダブルクリックします。
画面の指示に従ってインストールを行います。
注）インストール中に数種の ファイルがすでにシステムに存在します というメッセージがでる場合があります。この場合は既存のファイルを残すため、 はい(Yes) を選択して下さい。
注）インストール中に C:\windows\system\EXPSRV.DLL についてのエラー がでる場合があります。この場合はすでにシステムにファイルが存在することを意味しますので無視(ignore) を選択してインストールを進めて下さい。

Rotem Meter View Software

PRIR42X9.DOC

Page 3

4

起動方法
1. コミュニケーションケーブルを使用して ROTEM 測定器の一つを（RS232C ポートを経由）PC に接続します。
2. Start/Programs/Rotem Instruments/Rmv をクリックして RMV ソフトを起動させます。

5

セットアップ
5.1 PC 用 COM ポートの設定
コミュニケーションケーブルを使用してコンピュータの COM ポートに測定器を接続します。４つのホットボタンの内、左側の３つのホットボタンが使用可能状態にあることチェックします。
ホットボタン:

Stop／／Start Online collection(ｽﾄｯﾌﾟ/ｽﾀｰﾄｵﾝﾗｲﾝﾃﾞｰﾀ取り込み)
Mute Alarm(ﾐｭｰﾄｱﾗｰﾑ)
Get Offline Data from Instrument(測定器からｵﾌﾗｲﾝﾃﾞｰﾀの取り込み)
Display Graph(ｸﾞﾗﾌ表示)
使用可能になっていない場合は、以下のように COM ポートを設定します：
Setting/Communication Port をクリックすると以下の画面が現れます.

正しい COM ポートを選択し、OK をクリックします。ホットボタンがまだ使用不可の場合には測定器の電源を一度切り、再度電源を入れます。

Rotem Meter View Software

PRIR42X9.DOC

Page 4

5.2 Excel レポートフォームの設定
Data/Get legend を選択して表題パラメータを入力します。

Excel レポートを開いたとき、上部にこの表題パラメータが表示されます:

5.3 パラメータの設定
5.3.1 閾値パラメータの設定
Setting/Parameters を選択すると以下の画面が表示されます：

Threshold1 には第１閾値の設定を、Threshold2 には第２閾値の設定をします。受け取ったオンラインデータと、この画面上で設定された閾値間の関係に従い、アラームが鳴って（ユーザーが設定可能）ディスプレイのバックグランドカラーが変化します。
バックグランドカラー

緑 – 受け取ったデータが第１閾値を越えた場合
黄 −受け取ったデータが第１閾値を越え第２閾値を下回る場合
赤−受け取ったデータが第２閾値を越えた場合

閾値設定を使用しない時は、両方のブランクに"999999"を入力します。

Rotem Meter View Software

PRIR42X9.DOC

Page 5

5.3.2 Lost Contact Interval の設定
Setting/Parameters をクリックして、ロストコンタクトインターバルタイムを設定します。このインターバルタイムは、PC からの問いかけに対して反応しなかった時、アラームを出すまでのタイムリミットを設定するもので、バックグランド表示がライトブルーになり、アラームが鳴ります。

5.3.3 Audio Alarm の設定
Setting/Parameters をクリックして、アラーム音の ON/OFF を設定します。

5.3.4 Auto Arrange Form の設定
Setting/Parameters をクリックして "Auto Arrange Form”ボタンを選択します。
これを選択した場合、主画面、グラフ、リスト等の画面表示がみやすく並べ替えられて表示されます。

5.4 オンラインインターバルの設定
Setting/Online interval をクリックして、以下の画面を表示します：

この画面は RMV ソフトが測定器に指示を送り、オンラインでデータを取り込む時間間隔を設定します。

6

PC へのデータダウンロード
6.1 オフラインモード
1.

B.CODE 押しボタンを短く押して測定器に場所 ID を保存します。
（測定器にバーコードリーダがついていない場合は不要）

2.

データ保存用押しボタン（例：RAMDA-2000 の場合は LIGHT ボタン）を短く押して測定器に日付と測定値を保存します。

3.

"Get Offline Data from Instrument"のホットボタンを選択して測定器のデータをPC にダウンロードすると 取り込みに成功しました というメッセージがでてデータリストが表示されます。

Rotem Meter View Software

PRIR42X9.DOC

Page 6

4.

次にデータを取り込むときには以下の画面が表示されます："Yes"をクリックしたとき、先に読み込んだデータリストの最後の行に、これからダウンロードするデータを追加保存します。
"No"をクリックしたとき、先に読み込んだデータリストがクリアされ、新しいデータリストが作られます。
注：表示されるデータリストはグラフを作成したり、グラフのリストを表示するための表（最大３００データ表示）です。データリストは、一度データが Excelファイルに保存されると自動的にクリアされます。

表示されたデータリストの Records は取り込んだデータ数で、Instrument No.は測定器の番号を表します（例：RAM DA-2000 の場合は 2000 と表示）。 Detector は検出器の種類が表示されます。Status は閾値を越えた場合か不良アラームを感知した場合に数値が示されます。通常時は OK が表示されます。
status に表示される数値は以下の内容を示します：
32168ｵｰﾊﾞｰﾌﾛｰ ｵｰﾊﾞｰﾌﾛｰ 線量率閾
線量値ｵｰﾊﾞｰ
線量率42線量閾値 検出器ｵｰﾊﾞｰ不良1電池不足例）
Status = 1 – 電池不足
Status = 8 – 線量率閾値オーバー
Status = 9 – 電池不足及び線量率閾値オーバー

Rotem Meter View Software

PRIR42X9.DOC

Page 7

6.2 オンラインモード
1.Setting/Online Interval を選択し、インターバル時間を設定します。

2."Start/Stop Online collection" ホットボタンをクリックすると、ユーザーによって設定されたインターバルで測定器が PC にデータをダウンロードします。
3.再度"Start/Stop Online collection"ホットボタンをクリックするとオンラインモードから抜けることができます。

モニター用グラフとリスト

7

Rotem Meter View ソフトは８から３０００データまでをグラフにして表示することが可能です。

7.1

グラフ画面
1. Utility/Graph または Graph ホットボタンを押すとデータリストに取り込まれたデータをグラフに表示します。
2. ユーザーがグラフの特定のポイントをクリックするとポイントの詳細(13-Aug-00
12:30:43 PM, 32 cpm) が上のブランクに表示されます。キーボードの左右の矢印キーを押すによってポイント間を移動できます。
3. 画面のサイズは角をドラッグすることで修正可能です。
4. MARK – 選択するとインターバルポイントが表示されます。

Rotem Meter View Software

PRIR42X9.DOC

Page 8

7.2

ズーム
拡大したいグラフのデータ部分をドラッグすると、グラフを部分的にズームアップすることができます。 元のグラフに戻すときは All Data をクリックします。

7.3

パラメータ
Parameters を選択するとグラフのＹ軸スケールを設定できます。
Y 軸のスケール設定：Auto – Y 軸スケールはデータの最大、最小値レベルによって自動的に設定されます。
0-10 – Y 軸スケールを 0−10 に設定します。
0-100 – Y 軸スケールを 0−100 に設定します。
0-1000 – Y 軸スケールを 0−1000 に設定します。

7.4

オールデータAll Data を選択するとズーム後に、データリストにある全データを使ってグラフを表示します。

7.5

リスト
List を選択すると現在グラフに表示している全データを表示（大元のデータリストは300 データのみ表示）します。 Online でグラフを表示している時は List ボタンを押して、リストを表示したときまでに取り込んだデータのみを表示します。

Rotem Meter View Software

PRIR42X9.DOC

Page 9

7.6

Online/offline“Online”と表示されているときは offline でデータ取り込みを行い、”offline”と表示されているときは online でデータ取り込みを行います。

8機能
8.1 Excel ファイルへの保存
1. Excel がコンピュータにインストールされていることを確認して下さい。
2. Data/Get Legend をクリックして表題パラメータを入力します。
3. データを保存するために Utility/Save をクリックします：

Save in Excel File: Excel フォーマットで保存する場合にはこのオプションを使用します。
Open Excel after save: Save in Excel File で保存したファイルで、保存後直ちにデータを見るときは: Save in Excel File に加えてこのオプションを使用します。
File Name: RMV ソフトがデフォルトでファイル名を表示しますが変更も可能です。
Save: ここをクリックするとファイルを保存して現在表示しているデータリストをクリアします。（データリスト (RMVTEMP.LST)が作成される）RMV ソフトと同じディレクトリにファイルが保存されます。
Run Macro: RMV directory に以下と同じフォーマットが作られます。マクロを修正することも可能です。この機能はマクロを修正したり作成したりすることができる人のための機能です。

Rotem Meter View Software

PRIR42X9.DOCPage 10

8.2

時刻の設定
測定器の内部時計が合っていないときは、Utility/Send Time をクリックしてコンピュータの日付／時間情報を測定器に送信し、合わせることができます。

8.3

キャリブレーションデータの取り込み
Data/Get calibration data をクリックして測定器のキャリブレーションパラメータを取り込むことができます。この機能は RAM ION と TelePole のみ使用可能です。

9 Data コマンド
9.1

Start Online
オンラインで測定器からデータの受け取り開始と終了に使用します。

9.2

Get offline data
測定器に保存されているデータをダウンロードするのに使用します。

9.3

Get Legend
Excel ファイル用に表題パラメータを設定するのに使用します。

9.4

Get Calibration Data
RMV ソフトに接続した測定器(RAM ION または TelePole)のキャリブレーションデータを表示するのに使用します。

10 Setting コマンド
10.1 Online Interval
測定器からオンラインでデータを受け取るインターバル時間を設定します。

10.2 Communication PortPC の COM ポートを設定します。

10.3 Parameters閾値レベル、ロストコンタクトインターバル、アラームのオン／オフ、自動アレンジフォームの設定をします。

Rotem Meter View Software

PRIR42X9.DOC

Page 11

11 Utility コマンド
11.1 Save
一時的な"RMVTemp.lst"から Excel フォーマットファイルに保存するときに使用します。

11.2 Send TimePC のコントロールパネルから測定器の時間スタンプを書き換えるのに使用します。

11.3 Graph
オンラインまたはオフラインでグラフを表示します。

11.4 Test
PC と測定器間のコミュニケーションテストをするのに使用します。接続されたときには"Start/Stop Online communication" ホットボタンが使用可能状態になります。

12 トラブルシューティング
1. データを受け取り続けるがグラフを表示しない− ダウンロードしたデータの時間スタンプが順序良く並んでいないことが考えられます。測定器の時間スタンプがコンピュータの時間と合っていないとグラフは表示されません。Utility/Send Time をクリックして測定器の時間スタンプを正しく設定します。

Rotem Meter View Software

PRIR42X9.DOC

Page 12