Rated300W
Max.
420W
Peak600W
SNP-P30x-S Series

Medical & ITE
General Purpose

Features:
●
●
●
●
●
●
●
●
●
●

Design for BF application
With +5Vsb & +12V Fan output
Remote on/off & Remote senseWith Power good & Power fail signal
Fan speed controlled by load
High mechanical torque start-up-35oC to +70oC operating temperature
5,000m operation altitude
Convection cooling for rated load
Forced air for max. load

3.7" x 6.6" x 1.38"

General Specifications:
Input voltage ......................................... 90 VAC to 264 VAC
Input frequency ............................................. 47 Hz to 63 Hz
Standby power ...........................................< 0.2W at no load
(remote off status)< 0.5W at 5Vsb/40mA output
Inrush current ............................... < 30/60A at 115/230VAC
Efficiency ................................................................ 90% typ.
Hold up time ...........................................................18ms typ.
Over load/Short circuit protection ................... auto recovery
Over voltage protection ............................................ latch off
Operating temperature ..................................... -35oC to 70oC
derating: 2.5% / oC > 50oCMechanical Specifications:
167.0 (6.6")

95.0(3.7")

Cooling ..................................... 300W at convection cooling
420W at forced air cooling
Storage temperature ..................................... -40oC to +85oC
EMI .......................................... EN55022 "B", EN61000-3-3
Harmonics........................................... EN61000-3-2, class D
EMS..................................... EN61000-4-2,-3,-4,-5,-6,-8,-11
Safety ................................ UL/CSA/EN60950-1, 2nd edition
ANSI/AMMI/CSA/EN60601-1, 3.1 edition
Energy Saving .............................................. ENERGY STARfor computers version 6.0
for displays version 6.0
ErP regulation EC(No) 1275/2008
Notes:
1.
2.
3.

4.

5.

-Hui-

MAY 2015

Size:
3.7" x 6.6" x 1.38"
Mounting Hole:Side edge: 155 x 29 (mm)Bottom: 155 x 63 (mm)Connectors:
AC input: Using JST B5P-VH withdraw 2 pins or equivalent
DC output: Using JST B12P-VH or equivalent
Remote ON/OFF & Standby Output & Fan output& Remote Sense: 2 x 5 (10 pins) 0.1 pitch
Output Pin assignment (TB2):
Function Pin assignment (TB3):		 Pin 1 FAN GND Pin 2 FAN
1~67 ~ 12Vo

GND

3579RS+
RSGND
PG/PF46810GND+5Vsb
RemoteGND

Packing:
Net weight: 635 g approx. / unit
Gross weight: 15.5 kg approx. / carton, 20 units / carton
Carton size (mm): 453 (L) x 362 (W) x 279 (H)10 years Warranty (contact Skynet's Distributors for details)** This data sheet is only for models selection. For business, engineering specification by model must be used.

2-1

Rated300W
Max.
420W			Peak600W
SNP-P30x-S Series

Medical & ITE
General Purpose

Output Specifications:MODEL

OUTPUT			LOAD

INITIAL

STEP EFFICIENCYAVERAGE@ 20% LOAD @ 50% LOAD @ 100% LOADEFFICIENCY

RAIL

MIN. RATED		 MAX. PEAK

ACCURACY

+12V

0A

25A

35A 50A+11.9V~+12.1V

+5Vsb

0A

1A

1.5A		

+24V

0A

+5Vsb

0A

SNP-P30H-S+60V(ITE only)

+5Vsb

NO.

SNP-P307-S		

87%
90.5%
+4.5V~+5.5V			

91%

89%		

SNP-P309-S

12.5A 17.5A

25A+23.9V~+24.1V

1A

1.5A		

+4.5V~+5.5V

0A

5A

7A 10A+59.1V~+60.1V

0A

1A

1.5A		

+4.5V~+5.5V

87%

91%

91.5%

89%

87%

91%

92%

90%Note:
1.
2.
3.
4.

5.

6.
7.

Standby Power Cosumption with System:
For computers and displays, ENERGY STAR in U.S. and ErP regulation in Europe require the input power should be less than 0.5W at standby mode.
Output Load:
300W for convection cooling; 420W for forced air cooling.
Peak Load Duration:
Peak 600W can last for 5 sec.
High Torque Output Current:
For motor drive or audio amplifier, the output current can be up to 300% rated current within 1000ms, and output voltage can keep above 60% Voltage.
Example: SNP-P309-SIsolation Grade:Primary ←→ Ground : 1MOPP (1500Vac)
Primary ←→ Secondary : 2MOPP (4000Vac)
Secondary ←→ Ground		 : 1MOPP (1500Vac)
Leakage Current:
Earth leakage current < 300uA
Touch current < 100uA
Model Selection:
SNP-P307-S and SNP-P307-S are for ITE & medical applications.
SNP-P30H-S is only for ITE application.

-Hui-2-2** This data sheet is only for models selection. For business, engineering specification by model must be used.

MAY 2015