Rated
200WMax.
300W
Peak400W
SNP-P20x-S Series

Medical & ITE
General Purpose

2015 Q4Features:
●
●
●
●
●
●
●
●
●
●
●

3" x 6" x 1"

Design for BF application
Safety Class II
With +5Vsb & +12V Fan output
Remote on/off & Remote senseWith Power good & Power fail signal
Fan speed controlled by load
High mechanical torque start-up-30oC to +70oC operating temperature
5,000m operation altitude
Convection cooling for rated load
Forced air for max. load

General Specifications:
Input voltage ......................................... 90 VAC to 264 VAC
Input frequency ............................................. 47 Hz to 63 Hz
Standby power ...........................................< 0.2W at no load
(remote off status)< 0.5W at 5Vsb/40mA output
Inrush current ............................... < 30/60A at 115/230VAC
Efficiency ................................................................ 90% typ.
Hold up time ...........................................................18ms typ.
Over load/Short circuit protection ................... auto recovery
Over voltage protection ............................................ latch off
Operating temperature ..................................... -30oC to 70oC
derating: 2.5% / oC > 50oC

Mechanical Specifications:

Cooling ..................................... 200W at convection cooling
300W at forced air cooling
Storage temperature ..................................... -40oC to +85oC
EMI .......................................... EN55022 "B", EN61000-3-3
Harmonics........................................... EN61000-3-2, class D
EMS..................................... EN61000-4-2,-3,-4,-5,-6,-8,-11
Safety ................................ UL/CSA/EN60950-1, 2nd edition
ANSI/AMMI/CSA/EN60601-1, 3rd edition
Energy Saving .............................................. ENERGY STARfor computers version 6.0
for displays version 6.0
ErP regulation EC(No) 1275/2008
Notes:
1.

152.4 (6")

76.2(3")Size:
3" x 6" x 1"		2. Mounting Hole:
TBD
3. Connectors:		
AC input: Molex 5277-02A or equivalent
DC output : Terminal Blocks (default for SNP-P207) or			 Molex 5273-08 (default for others) or equivalent
Remote ON/OFF & Standby Output & Fan output &Remote Sense: Molex 5272-03 or equivalent
4. Output Pin assignment:		
Function Pin assignment:				
TB3 TB4 TB5 TB6Pin No.

12345678SNP-P207-S +Vo +Vo GND GND

5.

-Hui-JUL. 2015

OTHER+Vo +Vo +Vo +Vo GND GND GND GNDMODELS

TB7

Function FAN Standby Remote Remote		
P.G/P.F.		
Output Output Sense ControlPin
1GND

2+12V +5Vsb Sense+

GND Sense- GND

Packing:
Net weight: TBD g approx. / unit
Gross weight: TBD kg approx. / carton, TBD units / carton
Carton size (mm): TBD (L) x TBD (W) x TBD (H)+GND+

10 years Warranty (contact Skynet's Distributors for details)** This data sheet is only for models selection. For business, engineering specification by model must be used.

2-1

Rated200W
Max.
300W			Peak400W
SNP-P20x-S Series

Medical & ITE
General Purpose

Output Specifications:
MODELNO.

SNP-P207-S

OUTPUT			LOAD

INITIAL

STEP EFFICIENCYAVERAGE@ 20% LOAD @ 50% LOAD @ 100% LOADEFFICIENCYRAIL

MIN. RATED		 MAX. PEAK

ACCURACY+12V

0A+11.9V~+12.1V

+5Vsb

0A

16.66A

25A 33.3A

1A			

85%
90%
+4.5V~+5.5V			

90%

88%90%88%						SNP-P209-S

+24V

0A

8.33A 12.25A 16.66A+23.9V~+24.1V

+5Vsb

0A

1A			

+4.5V~+5.5V

85%

90%

Note:1.
2.
3.
4.

5.
6.
7.

Standby Power Cosumption with System:
For computers and displays, ENERGY STAR in U.S. and ErP regulation in Europe require the input power should be less than 0.5W at standby mode.
Output Load:
200W for convection cooling; 300W for forced air cooling.
Peak Load Duration:
Peak 400W can last for 5 sec.
Isolation Grade:Primary ←→ Ground : 1MOPP (1500Vac)
Primary ←→ Secondary : 2MOPP (4000Vac)
Secondary ←→ Ground		 : 1MOPP (1500Vac)
Leakage Current:
Earth leakage current < 300uA
Touch current < 100uA
EMI Grounding:If there is a metal sheet under the power supply, connect the EMI ground to that metal sheet.
Model Selection:
SNP-P20x-S is for ITE & medical applications.

-Hui-2-2** This data sheet is only for models selection. For business, engineering specification by model must be used.

JUL. 2015