Rated60W
Peak84W
SNP-H06x-M Series

MedicalGeneral Purpose

Features:●

Design for BF application●High mechanical torque start-up●-20oC to +70oC operating temperature●

3,000m operation altitude●

Convection cooling for rated load●

Cost effective2" x 4" x 0.97"

General Specifications:
Input voltage ......................................... 90 VAC to 264 VAC
Input frequency ............................................. 47 Hz to 63 Hz
Inrush current ................................ < 30/60A at 115/230VAC
Hold up time ................................................................. 16ms
Over load/Short citcuit protection ................... auto recovery
Over voltage protection ............................................ latch off
Operating temperature ..................................... -20oC to 70oC
derating: 2.5% / oC > 50oC
Storage temperature ..................................... -40oC to +85oC

EMI .......................................... EN55011 "B", EN61000-3-3
Harmonics............................................ EN61000-3-2, class A
EMS...................................... EN61000-4-2,-3,-4,-5,-6,-8,-11
Safety ................. ANSI/AMMI/CSA/EN60601-1, 3rd edition
CB report, CE mark, RM report/file

Mechanical Specifications:Ø3.2 TYP. MOUNTING HOLES
x4Notes:101.6 (4")

TB1

2.

LN

44.5

1.

TRANSFORMER

TB2

50.8(2")

3.
4.

3.15

95.3

3.15

5.
24.6 max.
(0.97")

Size:
2" x 4" x 0.97"
Mounting Hole:
44.5 x 95.3 (mm)Connectors:
AC input: Molex 5277-02A or equivalent
DC output: Molex 5273-04A or equivalen
Output Pin assignment:
1234VoVo

GND

GND

Packing:
Net weight: 110 g approx. / unit
Gross weight: 14 kg approx. / carton, 100 units / carton
Carton size (mm): 420 (L) x 382 (W) x 277 (H)

2.4 max. (3.0 Max. for SMD components)-Hui-MAY 2015

10 years Warranty (contact Skynet's Distributors for details)** This data sheet is only for models selection. For business, engineering specification by model must be used.

4-1

Medical

General Purpose

Rated
60W
Peak84W
SNP-H06x-M Series

Output Specifications:
PRODUCTRAILVOLTAGE

LINE

LOAD

RATED 		 MAX.

PEAK

ACCURACYRIPPLEMIN.

SNP-H067 -M

+12V

0A

5.00A

6.50A

+11.9V~+12.1V

120mVpp±1%±1%

SNP-H068 -M+15V0A

4.00A

5.60A

+14.9V~+15.1V

100mVpp±1%±1%

SNP-H065 -M		

+18V0A

3.33A

4.67A

+17.9V~+18.1V

150mVpp±1%±1%

SNP-H069 -M+24V0A

2.50A

3.50A

+23.9V~+24.1V

150mVpp±1%±1%		
SNP-H06G-M+28V0A

2.14A

3.00A

+27.9V~+28.1V

150mVpp±1%±1%

SNP-H06J -M

+36V

0A

1.66A

2.21A

+35.8V~+36.2V

200mVpp±1%±1%

SNP-H06T-M

+48V

0A

1.25A

1.75A

+47.8V~+48.2V

250mVpp±1%±1%

SNP-H06H-M		

+60V0A

1.00A

1.40A

+59.6V~+60.4V

300mVpp±1%±1%NO.

OUTPUT		 LOAD

NOISE

REG.

REG.

Note:
1.
2.
3.

4.
5.
6.

Output Load:
60W for convection cooling; 72W for forced air cooling.
Peak Load Duration:
Peak load can last for 5 sec.
Isolation Grade:Primary ←→ Ground : 1MOPP (1500Vac)
Primary ←→ Secondary : 2MOPP (4000Vac)
Secondary ←→ Ground		 : 1MOPP (1500Vac)
Leakage Current:
Earth leakage current < 300uA
Touch current < 100uA
EMI Grounding:If there is a metal sheet under the power supply, connect the EMI ground to that metal sheet.
Model Selection:
SNP-HF6x-M is for medical application.

-Hui-4-2** This data sheet is only for models selection. For business, engineering specification by model must be used.

MAY 2015

Rated60W
Peak84W
SNP-H06x-M Series

MedicalGeneral Purpose

Performance for SNP-H067-M (input voltage is 115VAC, unless others specified):
1. Switching frequency ripple

2. Line frequency ripple+12V/5A+12V/5A

50mV/div, 5us/div

50mV/div, 5ms/div

3. Output turn on wave form

4. Output turn off wave form
+12V/5A+12V/5A

5.0V/div, 10ms/div

5.0V/div, 10ms/div

5. Hold-up time

6. Over voltage protection
Trip point around 16.5V(60% Rated load)

5.0V/div

Primary DC50V/div
10ms/div

5.0V/div, 10ms/div-Hui-MAY 2015** This data sheet is only for models selection. For business, engineering specification by model must be used.

4-3

MedicalGeneral Purpose

7. +12V step response

8. Start up time

+12V+12V

Rated60W
Peak84W
SNP-H06x-M Series

Ac input

200mV/div, 2ms/div

0.1s/div+12V step from 1A to 5A

9. FCC B

10. CISPR 22 B

11. Power Derating Curve-Hui-4-4** This data sheet is only for models selection. For business, engineering specification by model must be used.

MAY 2015