Laboratory Accreditation Programmes
Schedule to

CERTIFICATE OF ACCREDITATION
Laboratory

Temprecord International LimitedAddress

PO Box 58430, Greenmount, Auckland, 2141
Unit D, 239 Burswood Drive, Burswood, Auckland, 2013

Telephone

09 274-9825

Fax

09 273-4020

URL

www.temprecord.com

Authorised Representative

Mr Greg Hall
Laboratory Manager

Client No.

6964

Programme

Metrology & Calibration Laboratory

Accreditation Number

814

Initial Accreditation Date

4 March 2002

Conformance Standard

NZS ISO/IEC 17025:2005
General requirements for the competence of testing and calibrationlaboratories

Testing Services
Summary5.355.61

Hygrometry
Temperature Measuring Equipment

Mrs Joanne Chen
Mr Greg Hall
Mrs Liz Li
Ms RosalieMontederamos

5.35, 5.61
5.35, 5.61
5.35, 5.61
5.61

Signatories

Authorised:
General Manager

Issue 13

Date: 02/03/12

International Accreditation New Zealand - Private Bag 28 908 - Remuera - Auckland
Telephone 09-525 6655 - Facsimile 09-525 2266
www.ianz.govt.nz

Page 1 of 3

Laboratory Accreditation Programmes
Schedule to

CERTIFICATE OF ACCREDITATION
Temprecord International Limited
Metrology & Calibration Laboratory
SCOPE OF ACCREDITATION

Accreditation No 814Calibration temperature 20 ºC.
All measurement uncertainties are based on a level of confidence of at least 95 %.
Unless stated elsewhere in this schedule, calibrations are performed at the premises of theaccredited laboratory.

5.35

Hygrometry(a)

Humidity measuring devices

Two point calibration of relative humidity loggers
Range 12 % RH to 85 % RH

Least uncertainty of measurement± 1.0 % RH (at a dry bulb temperature range of20 ˚C to 30 ˚C and a dew point range of -6 ˚C to23˚ C)

5.61

Temperature Measuring Equipment(including temperature calibration of electronic thermometers)(e)Thermistors and other semi-conductor thermometers

Calibration of thermistor based temperature loggers
Three point calibration
Calibration temperature range
For loggers without probes(internal sensor)

Least uncertainty of measurement-15 ˚C to 40 ˚C± 0.061 ˚C (routinely 0.2 ˚C from -15˚C to 40 ˚Cin the F3 machine

For loggers with probes (external sensor)-38 ˚C to 130 ˚C± 0.056 ˚C (routinely 0.2 ˚C from -20 ˚C to 50 ˚C)in the dry block calibrator± 0.02 ˚C (routinely 0.2 ˚C from -20˚C to 50 ˚C)in a stirred liquid bath

Authorised:
General ManagerIssue 13

Date: 02/03/12

International Accreditation New Zealand - Private Bag 28 908 - Remuera - Auckland
Telephone 09-525 6655 - Facsimile 09-525 2266
www.ianz.govt.nz

Page 2 of 3

Laboratory Accreditation Programmes
Schedule toCERTIFICATE OF ACCREDITATION
Temprecord International Limited
Metrology & Calibration Laboratory
SCOPE OF ACCREDITATION

Accreditation No 814

Single point calibration

For loggers without probes (internal sensor)-24 ˚C to 80 C

± 0.061 ˚C (in a stirred liquid bath with equalisingblock)

For loggers with probes (external sensor)-24 ˚C to 130 ˚C-24 ˚C to 130 ˚C

Authorised:General Manager± 0.056 ˚C (in the dry block calibrator)± 0.036 ˚C (in a stirred liquid bath)

Issue 13

Date: 02/03/12

International Accreditation New Zealand - Private Bag 28 908 - Remuera - Auckland
Telephone 09-525 6655 - Facsimile 09-525 2266
www.ianz.govt.nz

Page 3 of 3