Flue gas analyzer

HT-1200N/1200NT

7O2℃

0 〜 20.9 vol%0 〜 650℃

OptionMeasurement item ;1200N ; O2
1200NT ; O2・Temperature

Calculated items ; CO2 and more.

IR PrinterHT-1610Roll paper x 1
AAbattery x 4

Printer paper
HT-1636

５rollsAmbient airtemperature sensor
HT-13210 〜 100℃
For air temperature

Attach caseHT-1315

Made by aluminium
Size
350×460×155mm
2.9kg

Measurement
Software for PC
Online View 2000
HT-2084

RS232 transfer cable
USB cable
Recommended OS：
Windows 7/8.1/10

CO2 is calculated from O2 concentration based on
selectable fuel data.

Easy to measure air ratio

O2 Alarm

Battery charge forabroad
HT-1318

AC100 〜 240V

Selectable O2 alarm threshold value.

Integrated Data logger
Data storage up to 100 data.

Probe handle
HT-7201A

Sampling hose length
1800mmInc. condensate trap

Probe tube
HT-7230

L=300ｍｍ φ6mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7235

L=500ｍｍ φ6mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7232

L=750ｍｍ φ6mm
Flue gas temp.;
0 〜 650℃

Probe tube
HT-7233

L=750ｍｍ φ8mm
Flue gas temp.;
0 〜 1100℃8Speciﬁcation

Standard equipment ※Select one of probes

Measurement items
Measurement range
Accuracy

O2

Resolution
Response time
Measurement range

TemperatureＧ（K. thermocouple）

Accuracy
Resolution
Measurement range
Accuracy
Resolution

TemperatureＡ（Pt2000Ω）＊2

0 〜 20.9vol%±0.2vol%
0.1vol%<10 sec.
0 〜 650℃
0 〜 1100℃（depend on probe tube）＊1±1℃ or±1% reading（0 〜 650℃）±2% reading（650.1 〜 1100℃）
0.1℃（0 〜 999.9℃） 1℃（1000 〜 1100℃）
0 〜 100℃±1℃
0.1℃

Calculated items＊3
CO2
Air ratio
LossesEﬃciency

0 〜 CO2 Max.（calculation from O2 value）
1.0 〜 50.0
0 〜 100%（ only with Ｋthermocouple and Pt2000Ω）
0 〜 100%（ only with Ｋthermocouple and Pt2000Ω）

Fuel
13A、6C、LPG、Kerosine、light oil、heavy A、heavy C、Pellet

Electrochemical
K thermocouple
Pt2000Ω

Main unit
Temperature
DisplaySize
WeightPower

Pump, Interface for PC (RS232)
Data logger (possible to store 100 data), Interface for printerAccessory
Battery, Sampling probe, Condensate trap, Reﬂection plate, Carrying case, Softcase, Filter set
Spec. / remarks
Description
Art. No.
Sampling probe/HT-1200NHT-1229DS L=180mm φ5mm, with condensate trap, w/o temp.measurement
Sampling probe( with ﬂue gas temp. sensor)/HT-1200NT HT-1007 L=180mm φ5mm, with condensate trap, ﬂue gas temp.0 to 650℃

OptionProbesLong probe
HT-1235DS L=690mm φ6mm, with condensate trap, w/o temp.measurement
Ｌ-ProbeHT-1238DS L=120mm φ5mm, with condensate trap, w/o temp.measurement
Ｌ-ProbeHT-1006LS L=120mm φ5mm, with condensate trap, ﬂue gas temp.0 to 650℃
Probe handle
Probe handle
HT-7201AS with hose, condensate trap
Probe tube（for probe handle HT-7201AS）
Probe tube
HT-7230 L=300ｍｍ φ6mm, ﬂue gas temp.0 to 650℃
（With ﬂue gas temp. sensor）
HT-7235 L=500ｍｍ φ6mm, ﬂue gas temp.0 to 650℃
HT-7232 L=750ｍｍ φ6mm, ﬂue gas temp.0 to 650℃
HT-7233 L=750ｍｍ φ8mm, ﬂue gas temp.0 to 1100℃
Temp. probe
Kthermocouple HT-1252a φ1.5×130L, 0〜950℃,
for air/liquids
HT-1253a φ3×130L, 0〜400℃,
for air/liquids/food, centric top
HT-1254a 130L, 0〜400℃,
for surface /rifts/air/liquid
HT-1255a φ4×130L, 0〜650℃,Sensor
O2Temperature G
Temperature A

Integrated

Operating：0℃ 〜 ＋45℃
Storage：−20℃ 〜 ＋60℃
Dotmatrix 4 lines（W×H×D）80×150×35 mm
Approx.350g
Battery charger（AC100V 50/60Hz DC12V 100mA）
Integrated Nicd rechargeable battery（Max.8hours continuous operation）

for surface /air/liquid
HT-1256a 0〜450℃, magnetic probe for surface
HT-1257a 0〜180℃, Pliers probe for plates, tube
Air temp.probe
HT-1382 0〜100℃, for airPt2000Ω
Ambient air temp. sensor Pt2000Ω
HT-1321 0〜100℃
Battery charger
HT-1318 AC100〜240V
Hard caseHT-1315 Aluminum, dimension : 350×460×155,weight : 2.9kg
Measurement software
HT-2084 Online View 2000
（with RS232 cable・USB conversion cable）（Recommended OS：Windows 7/8.1/10）
Infrared printerHT-1610 Roll paper ×1 AA battery× 4
Roll paper for printerHT-1636 5 rolls
Food
HT-1376 for hot water supply device (gas/oil), stove＊1：Compensate the measured value when measure ﬂue gas include H2.
＊2：The higher value is applied as accuracy.
＊3：Ambient air temp. (optional items) sensor is needed.
＊Accuracy is based on cylinder gas.

Speciﬁcation might be changed without notice.

HT-1200N
Standard set

HT-1200NT
Standard set

・Main unit・Sampling probeHT-1229DS
・Battery charger
・Carrying case・Softcase・Filter set
・Operating manual

・Main unit・Sampling probeHT-1007
・Battery charger
・Carrying case・Softcase・Filter set
・Operating manualPrice 115,000yen

Price 129,000yen(tax excluded)

9(tax excluded)

Flue gas analyzer

HT-1200N/1200NT
Dimension

HT-1200N

Inlet port gas

Charge LED35

Battery charger port
Interface RS232
Ambient temp. sensor port
KIR printer InterfaceφdL

Gas inletGas sampling cable
Metal cone

Filter

φ7～22

Description
Sampling probe
Long probe

Art.No.
HT-1229DS
HT-1235DS

150

Condensate trap

80

L(mm) K(mm) φd(mm) Sampling hose length(mm) Flue gas temp.（℃）
0 〜 180 31551500−
0 〜 690 790
61500−

HT-1200NT

Inlet port gas

Charge LED

35

Battery charger port
Interface RS232
Ambient temp. sensor port

K

IR printer Interface

Lφd

Gas samping cable

Metal cone
φ7～22FilterCondensate
trap
150

Gas inlet
Built-in ﬂue gas temp.sensor

80

Description
Sampling probe

Art. No. L(mm) K(mm) φd(mm) Sampling hose length(mm) Flue gas temp.（℃）
HT-1007 0 〜 180 290
51800
0 〜 650

Flue gas temp. connector
Flue gas temp.compensation cable10