SNAP Pro* Analyzer
Quick Start Guide
Guide de démarrage rapide
Guía de inicio rápido
Kurzanleitung
Guida rapida di avvio
クイックスタートガイド
快速入门指南Contacting IDEXX
U.S./Canada: 1-800-248-2483
France: 33 (0) 810 433 999
Germany: 49 (0)69 153 253 290
Italy: 39 02 87 10 36 76
Spain: 34 932 672 660 / 34 916 376 317
United Kingdom: 44 (0)20 3788 7508
Australia: 1300 44 33 99
New Zealand: 0800 83 85 22
Brazil: 0800-777-7027
Latin America: tecnico-latam@idexx.com
日本: 0120-71 4921
中国 400-678-6682

Distributors: distributors-eusupport@idexx.com

Contents
English..................................................................................................... 4Français................................................................................................. 12
Español.................................................................................................. 20
Deutsch................................................................................................. 28
Italiano................................................................................................... 36
日本語.................................................................................................... 44
简体中文................................................................................................ 54

EnglishSetting up the analyzer

12Open the battery compartment
on the back of the analyzer
and then place two batteries(provided), negative end first,
into the compartment.

51

 onnect the appropriate adapter for yourC
region to the power cord using these steps:4a. Remove the protective insert
while sliding the locking key
toward the cord.

North America
Japan

Australiab. Insert the adapter top-side
first and then snap into place.

United Kingdom

EuropeSouth America

33Connect the power cord to theanalyzer and an appropriate
electrical outlet. The status lightshould appear steady yellow.
Note: It will take up to 8 hours
to fully charge the analyzer (thestatus light will turn green when
fully charged). The analyzer may
be used while charging.

5
6

If you are planning to connect the analyzer wirelessly to an IDEXX VetLab* Station, proceed to step 5.
Note: IDEXX VetLab* wireless router required.
OR
If you are connecting the analyzer to an IDEXX VetLab Station using an IDEXX VetLab* wired router,connect the analyzer to a numbered port on the router using a CAT5 cable (provided).
Press the gray Power button at the top of the back of the SNAP Pro Analyzer to power it on. Afterthe startup screen, follow the on-screen instructions.
When prompted, do the following:a. Select your location and then tap the
Next
icon.
b. If you are connecting the analyzer to an
IDEXX VetLab Station, tap Yes, select the
IDEXX network from the list, and then tap the
Nexticon. If the IDEXX network is not listed,
tap the Refreshicon.
OR
If you are not connecting the analyzer to an
IDEXX VetLab Station, tap No and then tap
the Next
icon. When prompted, confirm thatyou are not connecting the analyzer to an IDEXX
VetLab Station.
c. Select the desired date format, enter today’s
date using the numeric keypad, and then tap
the Next
icon.

d. Select the desired time format, enter the current
time using the numeric keypad, and then tap the
Next
icon.
e. If you want the analyzer to beep when there isan error, warning, or other information displayed,
select the Soundicon, select the check boxes
next to the desired options, and then select if you
would like the sound to be Soft or Loud. Thentap the Check Markicon.
OR
If you do not want the analyzer to beep, selectthe Mute
icon and then tap the Check Mark
icon.
Note: Critical errors will always beep.
f. When prompted, tap OK to restart the analyzer.

Need help? Visit idexx.com/prohelp
for helpful installation videos andsupport documentation.

English4

Running a SNAP test
With an IDEXX VetLab* Station
1. Tap Analyze Sample on the IDEXX VetLab Station
Home screen to enter the patient information andstart a run.
OR
If the IDEXX VetLab Station is integrated with apractice management system, proceed to step 2 or
select the patient from the Pending or Census list on
the IDEXX VetLab Station Home screen.
Note: All patients that appear in the Pending list on
the IDEXX VetLab Station also appear on the
SNAP Pro* Home screen.
OR
Tap New Patient on the SNAP Pro Home screen toenter the patient’s name and client ID. When you havefinished entering the information, tap the CheckMark
icon and proceed to step 3.
2. Once the patient’s name appears in the list on the
SNAP Pro Home screen, tap the patient’s name andthen tap the Check Markicon.

3. Prepare the SNAP* test, but
do not activate the test.
4. Gently slide the SNAP test into
the port at the bottom of theanalyzer (well end first) until itclicks into position.
When analysis is complete,results are shown on the results
screen and are sent to the
IDEXX VetLab Station
automatically.
Tip: The color of thestatus light indicates theresults of the run:
Green—all results areGreennegative
Blue—at least one
Blueresult is positive
Yellow—visual
Yellowconfirmationneeded (use theinstructions on thenext page)When you need to confirm results

1. Tap New Patient on the SNAP Pro Home
screen to enter the patient’s name and client ID.
When you have finished entering the information,
tap the Check Markicon.

If there are very light spots or unexpected color
development (such as debris or a blue tint) in the
results window, you may be prompted to confirm
some or all of a test’s results. Results that need to be
confirmed are highlighted yellow.

2. Prepare the SNAP test,
but do not activate thetest.
3. Gently slide the SNAP test
into the port at the bottomof the analyzer (well end
first) until it clicks into
position (see illustration on left).
When analysis is complete, results are shown on
the results screen and are saved on the analyzer
automatically.
Tip: The color of the status light indicates theresults of the run:
Green—all results are negative
Green
Blue—at least one result is positiveBlue
Yellow—visual confirmation needed (use the
Yellow
instructions on the right)

1. If you believe the highlighted result is positive, tap
the plus icon.
OR
If you believe the highlighted result to be negative,
tap the minus icon.
2. When you have finished confirming the results, tap
the Check Markicon. The results will then be
saved on the analyzer and sent to the IDEXX VetLab
Station (if applicable).
Note: To cancel the test and return to the Home
screen, tap the close icon .

EnglishWithout an IDEXX VetLab Station

Cleaning the analyzer
To clean the case and screen (perform as needed)
Clean the case and screen using a lint-free cloth dampened with 70% isopropyl alcohol, mild liquid soap and water,or a 10% bleach solution. DO NOT wipe the vents underneath the power button or the USB/power/Ethernet ports.

To clean the internal components (perform weekly or as needed)

1. Power off the analyzer

2. Use a flat-tip screw

and disconnect thepower and/or Ethernet
cord.

driver or coin to turnthe 2 silver fasteners
counterclockwise aquarter turn.

3. Remove the base fromthe analyzer.

4. Clean the entire baseusing a lint-free wipe or
cotton swab dampened
with 70% isopropyl
alcohol, mild liquid soap
and water, or a 10%
bleach solution.

EnglishActivator
Axle
Lens
Reflectors

5. Clean the 3 whitereflectors, the axle, andthe activator using a newlint-free wipe or cottonswab dampened with
70% isopropyl alcohol,
mild liquid soap andwater, or a 10% bleachsolution. DO NOT clean
the lens.

6. After the base has
dried, reattach it to theanalyzer and ensure it
is locked into place.

7. Turn the silver fasteners
clockwise a quarter
turn to lock the baseinto place.

8. Ensure the baseand silver fasteners
are flush with theanalyzer and locked
into place. Not
locking the base andfasteners properly
can impact analyzer
performance.

Interpreting SNAP Pro icons
SNAP Pro* Icons
Icon

Description
Wireless signal status (5 gray bars is thebest signal; 5 red bars indicates no signal)
Connected and communicating with
the IDEXX VetLab* Station via a wired
connection (a red icon indicates theanalyzer is offline)
Battery at 100% power
Battery at 75% power
Battery at 50% power
Battery at 25% power
Battery is critically low
Battery is empty

Icon

Description
Battery is charging
A software upgrade is available(SmartService* Solutions
customers only)
Warning notification (a gray icon
indicates the warning has been viewed
and is unresolved)
Error notification (a gray icon indicatesthe error has been viewed andis unresolved)
Lost communication withIDEXX VetLab Station

External USB power supply adapter
Input: 100–240 V AC, 47–63 Hz, 0.3 A
Output: 5 V DC, 1 A

Consumption
Operating: Maximum 5 W

Batteries
Main batteriesPart number: 79-91922-01
Output: 3.7 V DC, 2,600 mAh
Watch battery
Part number: BR1225
Output: 3 V DC, 48 mAh

Operating conditions
Operating temperature: 15°C–30°C (59°F–86°F)

Safety precautions• The SNAP Pro* Analyzer contains nouser-serviceable components. Do not disassemble.
Call IDEXX Technical Support for any service issues.
•D o not insert anything in the port except SNAP* tests
approved for use with the SNAP Pro Analyzer.
•P rotect the analyzer from sources of heat or flames.
• Use only the supplied power adapter.
• Use only IDEXX-supplied batteries.

• Dispose of used batteries only in accordance withyour local ordinances. These batteries containlithium. Recycling is required in the European Unionand is strongly encouraged in all other countries.
• If the equipment is used in a manner other thanspecified, the protection provided by the analyzer
may be impaired.
• The SNAP Pro Analyzer should be placed ona flat surface during operation.
Notice: To satisfy Industry Canada (IC) radio frequency(RF) exposure requirements for mobile and basestation transmission analyzers, a separation distanceof 20 cm or more should be maintained between theantenna of this analyzer and persons during operation.
To ensure compliance, operation at closer than thisdistance is not recommended. The antenna(s) used forthis transmitter must not be colocated or operating in
conjunction with any other antenna or transmitter.

EnglishTechnical specifications and safety information

Français

Mise en place de l’analyseur

1
2

Ouvrez le compartiment situé
à l’arrière de l’analyseur
SNAP Pro* et insérez-y lesdeux batteries rechargeables(fournies), en commençant parla borne négative.

5

1

Sur le cordon d’alimentation, branchez
l’adaptateur approprié pour votre région ensuivant les instructions suivantes :4
a. Retirez le cache de protectionen faisant glisser le poussoirvers le cordon.

Amérique du Nord
Japon

Australie3b. Insérez l’adaptateur encommençant par le bord
supérieur, puis fixez-le en place.

Royaume-Uni

EuropeAmérique du Sud

3

Branchez le cordon d’alimentation sur
l’analyseur et sur une prise électriqueadaptée. Le témoin d’état jaune doits’allumer sans clignoter.
Remarque : le cycle de charge complet
de l’analyseur peut durer jusqu’à
8 heures (le témoin passe au vert
lorsque l’analyseur est complètement
chargé). Vous pouvez toutefois utiliser
l’analyseur pendant qu’il est en charge.

4

Si vous prévoyez utiliser une connexion sans fil entre le SNAP Pro et une IDEXX VetLab* Station, passez
directement à l’étape 5.
Remarque : un routeur sans fil IDEXX VetLab* est requis.

5
6

Appuyez sur le bouton d’alimentation gris situé sur la partie haute, à l’arrière du SNAP Pro* pourle mettre sous tension. Lorsque l’écran de démarrage apparaît, suivez les instructions qui s’affichent.
Lorsque vous y êtes invité, appliquez la procédure
suivante :a. Sélectionnez votre zone géographique etappuyez sur l’icône Suivant .
b. Si vous avez choisi de connecter le SNAP Pro
à une IDEXX VetLab Station, appuyez sur Yes(Oui), sélectionnez le réseau IDEXX dans la liste,puis appuyez sur l’icône Suivant . Si le réseauIDEXX souhaité ne figure pas dans la liste,
appuyez sur l’icône Actualiser .
OU
Si vous ne connectez pas le SNAP Pro à une
IDEXX VetLab Station, appuyez sur No (Non)puis sur l’icône Suivant . Lorsque vous y êtesinvité, confirmez que l’analyseur ne sera pas
connecté à une IDEXX VetLab Station.
c. Sélectionnez le format de date souhaité, indiquez
la date du jour à l’aide du pavé numérique, puis
appuyez sur l’icône Suivant .
d. Sélectionnez le format horaire souhaité, réglez
l’heure à l’aide du pavé numérique, puis appuyez
sur l’icône Suivant .

e. Si vous souhaitez que l’analyseur émette
un bip sonore en cas d’affichage d’erreur,d’avertissement ou autres informations,sélectionnez l’icône Sound (Son) , cochez
les options souhaitées, puis indiquez si voussouhaitez que le volume sonore soit Soft (Bas)ou Loud (Fort). Appuyez ensuite sur l’icône
Valider .
OU
Si vous ne souhaitez pas que l’analyseur émettedes bips sonores, sélectionnez l’icône Muet
et appuyez sur l’icône Valider .
Remarque : Les erreurs critiques génèrent
toujours un bip sonore.
f. Lorsque vous yêtes invité, appuyez sur OK pourredémarrer l’analyseur.

Vous avez besoin d’aide ? Veuillez vous
rendre sur le site idexx.com/prohelp(en anglais) pour regarder des vidéos
d’installation utiles et lire des documentscomplémentaires.

FrançaisOU
Si vous souhaitez connecter le SNAP Pro à une IDEXX VetLab Station à l’aide d’un routeur filaire IDEXX
VetLab*, branchez l’analyseur à l’un des ports numérotés du routeur à l’aide d’un câble CAT5 (fourni).

Réalisation d’un test SNAP
Avec une IDEXX VetLab* Station
1. Appuyez sur Analyze Sample (Analyse del’échantillon) sur l’écran d’Accueil de la IDEXX
VetLab* Station, afin de saisir les informations
relatives au patient et effectuer une analyse.
		 OU
Si la IDEXX VetLab Station est intégrée à un logiciel
de gestion de pratique, passez à l’étape 2 ou
sélectionnez le patient dans la liste Pending (Enattente) ou Census (Enregistré) de l’écran d’Accueilde la IDEXX VetLab Station.
Remarque : Tous les patients figurant dans la liste
En attente de la IDEXX VetLab Station apparaissentégalement dans l’écran d’Accueil de l’analyseur
SNAP Pro.
OU
Appuyez sur New Patient (Nouveau Patient)dans l’écran d’Accueil de l’analyseur SNAP Pro*
pour saisir le nom du patient et le code du client.
Lorsque vous avez terminé la saisie, appuyez surl’icône Valider
et passez à l’étape 3.
2. Lorsque le nom du patient s’affiche sur l’écran
d’Accueil du SNAP Pro, sélectionnez-le et appuyez
sur l’icône Valider .

3. Préparez le test SNAP*, mais
ne l’activez pas.
4. Faites glisser délicatement letest SNAP dans l’ouverturesituée à la base de l’analyseur(côté puits en premier) jusqu’à
ce qu’il soit en position (un léger
claquement se fait entendre).
Lorsqu’une analyse est terminée,les résultats s’affichent à l’écran
de résultats et sont envoyés
automatiquement au poste
IDEXX VetLab.
Conseil : La couleur duvoyant d’état indique lesrésultats des analyses :Vert : tous les résultats
sont négatifs.
Bleu : au moins unrésultat est positif.
Jaune : uneconfirmation visuelle
est nécessaire (utilisez
les instructions données à la page suivante).

Si vous devez confirmer les résultats

1. Appuyez sur New patient (Nouveau Patient)
sur l’écran d’Accueil de l’analyseur SNAP Pro,
pour saisir le nom du patient et le code du client.
Lorsque vous avez terminé la saisie, appuyez
sur l’icône Valider .

Si vous observez des pastilles très pâles ou undéveloppement inattendu de la couleur (par exemple,
des débris ou une teinte bleutée) dans la fenêtre desrésultats, il vous sera peut-être demandé de confirmer
certains ou l’ensemble des résultats du test. Les
résultats qui doivent être confirmés sont indiqués parune couleur jaune.

2. Préparez le test SNAP, mais
ne l’activez pas.
3. Faites glisser délicatement
le test SNAP dansl’ouverture située à la basede l’analyseur (côté puits
en premier) jusqu’à ce qu’ilsoit en position (un léger claquement se fait
entendre) (voir illustration à gauche).
Lorsque l’analyse est terminée, les résultats
s’affichent à l’écran de résultats et sontautomatiquement enregistrés dans l’analyseur.
Conseil : La couleur du voyant d’état indiqueles résultats des analyses :Vert : tous les résultats sont négatifs.
Bleu : au moins un résultat est positif.
Jaune : une confirmation visuelle est nécessaire(utilisez les instructions figurant à droite).

1. Si vous pensez que le résultat mis en relief est
positif, appuyez sur l’icône Plus.
OU
Si vous pensez que le résultat mis en relief est
négatif, appuyez sur l’icône Moins.
2. Après avoir fini de confirmer les résultats, appuyez
sur l’icône Coche . Les résultats seront alors
enregistrés dans l’analyseur et envoyés au poste
IDEXX VetLab (le cas échéant).
Remarque : Pour annuler le test et retournerà l’écran d’accueil, appuyez sur l’icône Fermer.

Français

Sans la IDEXX VetLab* Station

Nettoyage de l’analyseur
Pour nettoyer le boîtier et l’écran (effectuer cette opération au besoin)
Nettoyez le boîtier et l’écran en utilisant un chiffon non pelucheux humidifié avec 70 % d’alcool isopropylique, dusavon liquide doux et de l’eau ou une solution d’hypochlorite de sodium à 10 %. NE NETTOYEZ PAS les évents
situés en dessous du bouton d’alimentation ou les ports USB/alimentation/Ethernet.

Pour nettoyer les composants internes (effectuez cette opération chaque semaine ou au besoin)

1. Mettez l’analyseur
hors tension, puisdébranchez le cordon
d’alimentation et/ou
le câble de connexion
Ethernet.

2. À l’aide d’un tournevis
à bout plat ou d’une
pièce de monnaie,
tournez d’un quart detour les deux éléments
de fixation argentés
dans le sens inverse
des aiguilles d’une
montre.

3. Retirez le support
de l’analyseur.

4. Nettoyez l’ensemble
du support en utilisantune serviette ou
un coton-tige nonpelucheux humidifié
avec 70 % d’alcool
isopropylique, du savon
liquide doux et del’eau ou une solution
d’hypochlorite desodium à 10 %.

Français

Activateur
Axe
Lentille
Réflecteurs

5. Nettoyez les3 réflecteurs blancs,
l’axe et l’activateur enutilisant une servietteou un coton-tige nonpelucheux humidifié
avec 70 % d’alcool
isopropylique, du savon
liquide doux et del’eau ou une solution
d’hypochlorite de sodiumà 10 %. NE NETTOYEZ
PAS la lentille.

6. Lorsque le support
est sec, fixez-le denouveau sur l’analyseur
et vérifiez qu’il est
verrouillé.

7. Tournez d’un quartde tour les éléments
de fixation dans lesens des aiguilles
d’une montre afin deverrouiller le support.

8. Vérifiez que le supportet les éléments defixation soient à niveau
avec l’analyseur et bien
verrouillés. Le fait dene pas verrouiller
correctement lesupport et leséléments de fixation
peut avoir une
incidence sur laperformance del’analyseur.

Interprétation des icônes de SNAP Pro
Icône

Description

Icône

Description

État du signal sans fil (5 barres grises
indiquent un signal excellent et 5 barres
rouges une absence de signal)

Une mise à jour logicielle est
disponible (clients des solutions IDEXX
SmartService* uniquement)

Appareil connecté et en communication
avec la IDEXX VetLab* Station via uneconnexion filaire (une icône rouge
indique que l’analyseur est hors ligne)

Notification d’avertissement (une icône
grise indique que l’avertissement a été
consulté mais non résolu)

Batterie à 100 % de charge
Batterie à 75 % de charge
Batterie à 50 % de charge
Batterie à 25 % de charge
Niveau de batterie critiqueBatterie vide
Batterie en charge

Notification d’erreur (une icône grise
indique que l’erreur a été consultée mais
non résolue)
Perte de communication avec la
IDEXX VetLab Station

Adaptateur d’alimentation USB externe
Entrée : 100-240 V CA, 47-63 Hz, 0,3 A
Sortie : 5 V CC, 1 A

Consommation

En fonctionnement : maximum 5 W

Batteries

Batteries principales
Référence : 79-91922-01
Sortie : 3,7 V CC, 2 600 mAh
Pile de montre
Référence : BR1225
Sortie : 3 V CC, 48 mAh

Conditions de fonctionnement

Température de fonctionnement : 15 °C-30 °C

Consignes de sécurité• L’analyseur SNAP Pro contient des composants ne
pouvant être changés ou manipulés par l’utilisateur.
Ne le démontez jamais. Contactez le Support
technique IDEXX pour tout problème relatif àl’entretien.
•N ’insérez jamais d’objet dans l’ouverturede l’analyseur, à l’exception des tests SNAP*
spécialement adaptés à l’analyseur SNAP Pro.
• Protégez l’analyseur contre toute source de chaleur
ou de flammes.
• Utilisez uniquement l’adaptateur d’alimentation fourni.

• Utilisez uniquement les batteries fournies par IDEXX.
• La mise au rebut des batteries doit être effectuée
conformément aux réglementations locales envigueur. Ces batteries contiennent du lithium. Leur
recyclage est obligatoire en Union européenne etvivement recommandée dans les autres pays.
• La protection de l’analyseur peut s’altérer
si vous ne l’utilisez pas conformément aux
spécifications fournies.
• L’analyseur SNAP Pro doit être installé sur une
surface plane lors de son utilisation.
Notification : Pour satisfaire aux exigencesd’exposition aux radiofréquences (RF) établies
par Industrie Canada (IC) pour les appareils detransmission fixes et mobiles, une distance d’aumoins 20 cm doit être maintenue entre l’antenne
de l’analyseur et toute personne présente, durant
son fonctionnement. Pour garantir le respect decette exigence, toute utilisation avec une distancemoindre est déconseillée. La ou les antennes utiliséespour cet émetteur ne doivent pas être situées à
proximité d’autres antennes ou émetteurs, ou utiliséesconjointement avec ceux-ci.

Français

Spécifications techniques et consignes de sécurité

Español

Instalación del analizador

12Abra el compartimento parapilas situado en la parteposterior del analizador SNAPPro* e introduzca las dos pilas
incluidas, con el polo negativodelante, en el compartimento.

5

1

Conecte el adaptador adecuado para su región alcable de alimentación siguiendo estos pasos:4a. Retire la tapa protectora
mientras desliza la pestaña
de bloqueo hacia el cable.

Norteamérica
Japón

Australiab. Introduzca el adaptador,
con la parte superior delante,
hasta que quede encajado.

Reino Unido

EuropaSudamérica

3

3

Conecte el cable de alimentación alanalizador y a una toma de corriente. El
indicador luminoso de estado debe iluminarse
permanentemente de color amarillo.
Nota: El analizador puede tardar hasta
8 horas en cargarse por completo (el indicador
luminoso de estado cambiará a color verde
cuando se complete la carga). Es posible
utilizar el analizador mientras se está cargando.

5
6

Si va a conectar el analizador inalámbricamente a una IDEXX VetLab* Station, vaya al paso 5.
Nota: Se necesita un router inalámbrico IDEXX VetLab*.
O
Si va a conectar el analizador a una IDEXX VetLab Station utilizando un router IDEXX VetLab* con cable,
conecte el analizador a uno de los puertos numerados del router empleando un cable Cat. 5 (incluido).
Pulse el botón de encendido gris, situado en la parte superior trasera del analizador SNAP Pro, paraencenderlo. Una vez pasada la pantalla de inicio, siga las instrucciones que aparecen en pantalla.
Cuando se le indique, haga lo siguiente:a. Seleccione su ubicación y pulse el icono
Siguiente .
b. Si va a conectar el analizador a una IDEXX VetLab
Station, pulse Yes (Sí), seleccione la red IDEXX enla lista y pulse el icono Siguiente . Si la red IDEXX
no aparece en la lista, pulse el icono Actualizar .
O
Si no va a conectar el analizador a una IDEXX
VetLab Station, pulse No (No) y, a continuación,
pulse el icono Siguiente . Cuando se le pregunte,confirme que no va a conectar el analizador a una
IDEXX VetLab Station.
c. Seleccione el formato de fecha que prefiera,
introduzca la fecha actual utilizando el teclado
numérico y, a continuación, pulse el icono
Siguiente .
d. Seleccione el formato de hora que prefiera,
introduzca la hora actual utilizando el teclado
numérico y, a continuación, pulse el icono
Siguiente .

e. Si desea que el analizador emita un sonido de aviso
cuando se produzca un error o una advertencia ose muestre otra información, seleccione el icono
Sonido , marque las casillas de verificación
situadas junto a las opciones que desee y, porúltimo, elija si desea que el sonido sea Soft (Bajo)o Loud (Alto). A continuación, pulse el icono de
Marca de verificación .

Español

4O
Si no desea que el analizador emita ningún sonido,
seleccione el icono Silencio y, a continuación,
pulse el icono de Marca de verificación .
Nota: Siempre se emite un sonido de aviso en casode error grave.
f. Cuando se le indique, pulse OK para reiniciar
el analizador.

¿Necesita ayuda? Visite idexx.com/prohelp(en inglés) para consultar documentación desoporte y vídeos de instalación útiles.

Realización de un test SNAP
Con IDEXX VetLab* Station
1. Pulse Analyze Sample (Analizar muestra) en lapantalla de Inicio de la IDEXX VetLab Station para
introducir la información del paciente e iniciar elanálisis.
O
Si la IDEXX VetLab Station está integrada en unsistema de gestión de clínica, vaya al paso 2 oseleccione el paciente de la lista Pendientes o Census
en la pantalla de Inicio de la IDEXX VetLab Station.
Nota: Todos los pacientes que aparecen en la lista
Pendientes de la IDEXX VetLab Station también
aparecen en la pantalla de Inicio de SNAP Pro*.
O
Pulse New Patient (Nuevo paciente) en lapantalla de Inicio de SNAP Pro para introducir elnombre del paciente y el ID del cliente. Cuando
haya terminado, pulse el icono de Marca deverificación
y vaya al paso 3.
2. Cuando el nombre del paciente aparezca en la listade la pantalla de Inicio de SNAP Pro, pulse el nombredel paciente y, a continuación, el icono de Marca deverificación .
3. Prepare el test SNAP*, pero no lo active.

4. Deslice con cuidado el test
SNAP en el puerto situado enla parte inferior del analizador(con el pocillo de muestra
delante) hasta que se oiga
un “clic”.
Cuando se completa el análisis,
se muestran los resultados
en la pantalla de resultados y se
envían a la IDEXX VetLab Station
automáticamente.
Consejo: El color de la luz deestado indica los resultadosdel análisis:
Verde: Todos los resultadosVerdeson negativos
Azul: Al menos un resultado
Azul
es positivo
Amarillo Se necesita
Amarillo:
confirmación
visual (utilice las
instrucciones de lapágina siguiente).

Sin IDEXX VetLab Station

Cuándo necesita confirmar los resultados

1. Pulse New Patient (Nuevo paciente) en lapantalla de Inicio de SNAP Pro para introducir elnombre del paciente y el ID del cliente. Cuando
haya terminado, pulse el icono de Marca deverificación .

Si hay puntos muy claros o se desarrollan colores
imprevistos (como restos o tinte azul) en la ventana
de resultados, es posible que le pidan que confirme
algunos o todos los resultados del test. Los resultados
que hay que confirmar se resaltan en amarillo.

3. Deslice con cuidado eltest SNAP en la abertura
situada en la parte inferiordel analizador (con el
pocillo de muestra delante) hasta que se oiga
un “clic” (véase la figura de la izquierda).
Cuando se completa el análisis, se muestranlos resultados en la pantalla de resultados y se
guardan automáticamente en el analizador .
Consejo: El color de la luz de estado indica losresultados del análisis:
Verde: Todos los resultados son negativos
Verde
Azul: Al menos un resultado es positivo
Azul
Amarillo: Se necesita confirmación visual (utilice
Amarillo
las instrucciones que aparecen a la derecha).

1. Si cree que el resultado resaltado es positivo, pulse
el icono del signo más.
O BIENSi cree que el resultado resaltado es negativo, pulse
el icono del signo menos.
2. Cuando haya terminado de confirmar los resultados,
pulse el icono de Marca de verificación . Los
resultados se guardarán a continuación en elanalizador y se enviarán a la IDEXX VetLab Station(si corresponde).
Nota: Para cancelar el test y volver a la pantalla deInicio, pulse el icono de cerrar .

Español

2. Prepare el test SNAP, pero
no lo active.

Limpieza del analizador
Para limpiar la carcasa y la pantalla (realice según sea necesario)
Limpie la carcasa y la pantalla utilizando un paño que no genere pelusas humedecido con alcohol isopropílico al
70%, jabón líquido suave y agua, o una solución de lejía al 10%. NO limpie los orificios de ventilación bajo el botónde encendido o los puertos USB/de alimentación/Ethernet.

Para limpiar los componentes internos (realice semanalmente o según sea necesario)

1. Apague el analizador

2. Utilice un destornillador

3. Retire la basey desconecte el cablede alimentación y/o deEthernet.

de punta plana o una
moneda para girar los2 tornillos plateados ensentido antihorario un
cuarto de giro.

del analizador.

4. Limpie toda la baseutilizando un paño que
no genere pelusas oun hisopo de algodón
humedecido con
alcohol isopropílico
al 70%, jabón líquido
suave y agua, o una
solución de lejía al 10%.

Activador
Eje
Lente
Reflectores

5. Limpie los 3 reflectores
blancos, el eje y elactivador utilizando
un nuevo paño que
no genere pelusas oun hisopo de algodón
humedecido con alcohol
isopropílico al 70%,
jabón líquido suave
y agua, o una soluciónde lejía al 10%.
NO limpie la lente.

6. Cuando la base sehaya secado, vuelva
a fijarla al analizador yasegúrese de que estésujeta en su sitio.

7. Gire los tornillos
plateados en sentido
horario un cuarto degiro para asegurarla base en su sitio.

8. Asegúrese de quelos tornillos plateados
estén al ras con elanalizador y sujetos
en su sitio. No
asegurar la basey los tornillos
adecuadamente
puede afectar
al rendimientodel analizador.

Interpretación de los iconos del SNAP Pro
Iconos del SNAP Pro*
Icono

Descripción

Icono

Descripción

Estado de la señal inalámbrica (5 barras
grises indican que la señal es excelente;
5 barras rojas indican que no hay señal)

Hay una actualización de software
disponible (sólo para clientes de lassoluciones IDEXX SmartService*)

Conectado y comunicándose con
la IDEXX VetLab* Station a través deuna conexión de cable (si aparece encolor rojo indica que el analizador noestá conectado)

Advertencia (si aparece en color grisindica que la advertencia ha sido
visualizada pero no se ha resuelto)

Pila al 100% de capacidad
Pila al 75% de capacidad
Pila al 50% de capacidad
Pila al 25% de capacidad
Pila casi agotada
Pila agotada
Pila en proceso de carga

Error (si aparece en color gris indica que
el error ha sido visualizado pero no se
ha resuelto)
Se ha perdido la comunicación con la
IDEXX VetLab Station

Adaptador de alimentación externo USB
Entrada: 100–240 V CA, 47/63 Hz, 0,3 A
Salida: 5 V CC, 1 A

Consumo

En funcionamiento: máximo 5 W

Pilas

Pilas principales
Referencia: 79-91922-01
Salida: 3,7 V CC, 2.600 mAh
Pila de reloj
Referencia: BR1225
Salida: 3 V CC, 48 mAh

Condiciones de funcionamiento

Temperatura de funcionamiento: 15-30 ºC

Precauciones de seguridad• El analizador SNAP Pro* no contiene ningún
componente reparable por el usuario. No lo
desmonte. Llame al Servicio Técnico de IDEXX para
cualquier posible problema.
•N o introduzca nada en el puerto excepto los tests
SNAP* autorizados para su uso con el analizador
SNAP Pro.
• Proteja el analizador de fuentes de calor o llamas.
• U tilice únicamente el adaptador dealimentación incluido.

• Utilice únicamente las pilas facilitadas por IDEXX.
• Deseche las pilas únicamente del modo dispuesto
en la normativa local. Estas pilas contienen litio.
Reciclarlas es obligatorio en la Unión Europea y muy
recomendable en todos los demás países.
• Si el analizador se utiliza de algún modo distinto al
especificado, la protección proporcionada por éstepuede verse reducida.
• El analizador SNAP Pro debe colocarse sobre una
superficie plana durante su uso.
Aviso: Con el fin de satisfacer los requisitos deexposición a radiofrecuencia (RF) de Industry
Canada (IC) para analizadors de transmisión
móviles y estaciones base, debe mantenerse
una distancia de separación de 20 cm o más
entre la antena de este analizador y cualquierpersona cuando el analizador se esté utilizando.
Para asegurar el cumplimento de esta norma sedesaconseja su utilización a una distancia inferiora la indicada. La(s) antena(s) utilizadas para estetransmisor no deben estar coubicadas ni funcionar
junto con ninguna otra antena o transmisor.

Español

Especificaciones técnicas e información de seguridad

Deutsch

Analysegerät einrichten

12Öffnen Sie das Akkufach
auf der Rückseite desAnalysegerätes SNAP Pro*und setzen Sie zwei Akkus(mitgeliefert) mit dem Minuspol
voran in das Fach ein.

5

1

2 Schließen Sie den für Ihre Region geeigneten
Adapter an das Stromkabel an, indem Sie diese
Schritte befolgen:4
a. Entfernen Sie den
Schutzeinsatz, während Sie
die Sperrtaste in Richtung
des Kabels schieben.

Nordamerika
Japan

Australienb. Setzen Sie den Adapter
mit dem oberen Teil voran
ein, bis er einrastet.

UK

EuropaSüdamerika

3

3Schließen Sie das Stromkabel an
das Analysegerät und eine geeignete
Steckdose an. Die Status-LED muss nun
dauerhaft gelb leuchten.
Hinweis: Das vollständige Aufladen desAnalysegerätes dauert bis zu 8 Stunden(die Status-LED wechselt zu grün, sobald
das Analysegerät vollständig aufgeladenist). Das Analysegerät kann während desAufladens verwendet werden.

5
6

ODER
Wenn Sie das Analysegerät über einen kabelgebundenen IDEXX VetLab*-Router mit der IDEXX VetLab Station verbinden
möchten, schließen Sie es mit einem CAT5-Kabel (mitgeliefert) an einen nummerierten Anschluss am Router an.
Drücken Sie die graue Ein-/Aus-Taste oben an der Rückseite des mobilen Analysegerätes SNAP Pro*,um es einzuschalten. Befolgen Sie nach dem Initialisierungsbildschirm die Anweisungen am Bildschirm.
Führen Sie die folgenden Schritte aus, sobald Sie
dazu aufgefordert werden:
a. Wählen Sie Ihren Standort aus und tippen Sie auf
das Symbol „Weiter“ .
b. Wenn Sie Ihr Analysegerät an eine IDEXX VetLab
Station anschließen, tippen Sie auf „Yes“, “,
wählen Sie IDEXX-Netzwerk aus der Liste aus
und tippen Sie dann auf das Symbol „Weiter“. Wenn das IDEXX-Netzwerk nicht aufgelistet ist,tippen Sie auf das Symbol „Aktualisieren“ .
ODERWenn Sie Ihr Analysegerät nicht an eine IDEXX
VetLab Station anschließen, tippen Sie auf„Nein“ und dann auf das Symbol „Weiter“. Bestätigen Sie, sobald Sie dazu aufgefordert
werden, dass Sie das Analysegerät nicht an eine
IDEXX VetLab Station anschließen.
c. Wählen Sie das gewünschte Datumsformat aus,
geben Sie mit den Zifferntasten das heutige
Datum ein und tippen Sie dann auf das
Symbol „Weiter“ .

d. Wählen Sie das gewünschte Zeitformat aus,
geben Sie mit den Zifferntasten die aktuelle
Uhrzeit ein und tippen Sie dann auf das Symbol„Weiter“ .
e. Wenn Sie möchten, dass das Analysegerät einen
Piepton ausgibt, wenn ein Fehler, eine Warnung
oder eine andere Information angezeigt wird,
wählen Sie das Symbol „Ton“ , aus, markieren
Sie die Kontrollkästchen neben den gewünschten
Optionen und wählen Sie dann aus, ob der Ton
leise oder laut sein soll. Tippen Sie dann auf das
Häkchen-Symbol .
ODER

Deutsch4

Falls Sie vorhaben, das Analysegerät drahtlos mit der IDEXX VetLab* Station zu verbinden, fahren Sie mit Schritt 5 fort.
Hinweis: Ein IDEXX VetLab*-WLAN-Router ist erforderlich.

Wenn Sie keinen Piepton wünschen, wählen Sie
das Symbol „Ton“
aus und tippen Sie dannauf das Häkchen-Symbol .
Hinweis: Bei kritischen Fehlern wird immer ein
Piepton ausgegeben.
f. Tippen Sie auf OK, sobald Sie dazu aufgefordert
werden, um das Analysegerät neu zu starten.

Benötigen Sie Unterstützung? Gehen Sie auf idexx.com/prohelp(in englischer Sprache) für hilfreiche Installationsvideos und Supportdokumentation.

Einen SNAP-Test durchführen
Mit einer IDEXX VetLab* Station
1. Tippen Sie im Hauptmenü der IDEXX VetLab* Station
auf die Schaltfläche „Analyze Sample“, um die
Patienteninformation einzugeben und eine Analyse
zu starten.
ODER
Wenn die IDEXX VetLab Station in ein
Praxismanagement-System integriert ist, fahren Sie
mit Schritt 2 fort oder wählen Sie den Patienten aus
der Liste „Ausstehend“ oder der Liste „Zählung“ imHauptmenü der IDEXX VetLab Station aus.
Hinweis: Alle Patienten, die in der IDEXX VetLab Stationin der Liste „Ausstehend“ aufgeführt sind, werden
auch im Hauptmenü des Analysegerätes SNAP Pro
angezeigt.
ODER
Tippen Sie im Hauptmenü des SNAP Pro* auf New
Patient um den Patientennamen und die Kunden-ID
einzugeben. Wenn Sie die Eingabe der Information
beendet haben, tippen Sie auf das Häkchen-Symbol
und fahren Sie mit Schritt 3 fort.
2. Sobald der Patientenname in der Liste auf dem
Hauptmenü des SNAP Pro erscheint, tippen Sie auf
den Patientennamen und dann auf das HäkchenSymbol .

3. Bereiten Sie den SNAP* Testvor, ohne jedoch den Test zu
aktivieren.
4. Schieben Sie den SNAPTest vorsichtig in die
Analysekammer im unteren Teildes Analysegerätes (mit derProbenvertiefung voran) ein, bis
er einrastet.
Nach dem Abschluss der Analyse werden die Befunde
auf dem Bildschirm angezeigt undautomatisch an die IDEXX VetLab*Station gesendet.
Tipp: Die Farbe derStatusleuchte zeigt die
Ergebnisse des Durchlaufs
an:Grün—alle Ergebnisse sindGrünnegativ
Blau—mindestens ein
Blau
Ergebnis ist positiv
Gelb—visuelle
Gelb
Bestätigung erforderlich(gehen Sie gemäß den
Anweisungen auf dernächsten Seite vor).

Ohne IDEXX VetLab Station

Wenn Sie die Ergebnisse bestätigen müssen

1. Tippen Sie im Hauptmenü des SNAP Pro*auf New Patient, um den Patientennamen
und die Kunden-ID einzugeben. Wenn Sie die
Eingabe beendet haben, tippen Sie auf das
Häkchen-Symbol .

Wenn sehr helle Punkte oder eine unerwartete
Farbentwicklung (wie beispielsweise Verschmutzungoder eine blaue Färbung) im Ergebnisfenster auftreten,
werden Sie möglicherweise aufgefordert einige oderalle Testergebnisse bestätigen. Ergebnisse, die derBestätigung bedürfen, sind gelb markiert.

3. Schieben Sie den SNAP
Test vorsichtig in die
Analysekammer im unteren
Teil des Analysegerätes(mit der Probenvertiefung voran) ein, bis er
einrastet (siehe Abbildung links).
Nach Abschluss der Analyse werden die
Befunde auf dem Bildschirm angezeigt undautomatisch im Analysegerät gespeichert.
Tipp: Die Farbe der Statusleuchte zeigt die
Ergebnisses des Durchlaufs an:
Grün—alle Ergebnisse sind negativ
Grün
Blau—mindestens ein Ergebnis ist positiv
Blau
Gelb—visuelle Bestätigung erforderlich (gehen
Gelb
Sie gemäß den Anweisungen rechts vor)

1. Wenn Sie glauben, dass das markierte Ergebnis
positiv ist, tippen Sie auf das „Plus“-Symbol.
oder
Wenn Sie glauben, dass das markierte Ergebnis
negativ ist, tippen Sie auf das „Minus“-Symbol.
2. Wenn Sie die Bestätigung der Ergebnisse
abgeschlossen haben, tippen Sie auf das Symbol
Markierung prüfen. Die Befunde werden im
Analysegerät gespeichert und an die IDEXX VetLab
Station gesendet (sofern zutreffend).
Hinweis: Um den Test abzubrechen und zumStartbildschirm zurückzugehen, tippen Sie auf das
Symbol Schließen. .

Deutsch

2. Bereiten Sie den SNAP Testvor, ohne jedoch den Testzu aktivieren.

Reinigung des Analysegeräts
Reinigung des Gehäuses und Bildschirms (Durchführung nach Bedarf)
Reinigen Sie das Gehäuse und den Bildschirm mit einem fusselfreien, entweder in einer 70%igen
Isopropyllösung, in Seifenwasser oder einer 10%igen Bleichlösung angefeuchteten Tuch. WISCHEN SIE
NICHT die Lüftungsöffnungen unterhalb der Ein/Aus-Taste oder an den USB/Netz/Ethernet-Anschlüssen.

Reinigung der internen Komponenten (Durchführung wöchentlich oder nach Bedarf)

1. Schalten Sie das
Analysegerät ab undunterbrechen Sie die
Stromzufuhr und/oder ziehen Sie das
Ethernetkabel.

2. Nehmen Sie einen
Schlitzschraubendreher
oder eine Münze, um
die beiden silbernen
Verbindungselementeum eine Vierteldrehung
gegen denUhrzeigersinn zu
bewegen.

3. Nehmen Sie die Basisvom Analysegerät ab.

4. Reinigen Sie diegesamte Basis mit
einem fusselfreien,
entweder ineiner 70%igen
Isopropyllösung, inSeifenwasser oder einer
10%igen Bleichlösung
angefeuchteten Tupferoder Wattestäbchen.

Auslöser
Welle
LinsenReflektoren

5. Reinigen Sie die 3

weißen Reflektoren
und den Auslöser
mit einem neuen,
fusselfreien, entweder
in einer 70%igen
Isopropyllösung, inSeifenwasser oder einer
10%igen Bleichlösung
angefeuchteten Tupferoder Wattestäbchen.
REINIGEN SIE NICHT
die Linsen.

6. Nachdem die Basisgetrocknet ist, bringen
Sie diese wieder am
Analysegerät an undachten Sie darauf, dasssie richtig eingerastet ist.

7. Drehen Sie
die silbernen
Verbindungselementeum eine Vierteldrehung
im Uhrzeigersinn, damit
die Basis einrastet.

8. Achten Sie darauf,

dass die silbernen
Verbindungselemente
mit dem Analysegerät
bündig und richtig
eingerastet sind. Ein
unvollständiges
Einrasten vonBasis und
Verbindungselementen
kann die Leistungdes Analysegerätes
beeinträchtigen.

Interpretation der SNAP Pro-Symbole
SNAP Pro*-Symbole
SymbolBeschreibung

SymbolBeschreibung

Status der drahtlosen Signalübertragung(5 graue Balken bedeuten das beste Signal,
5 rote Balken bedeuten kein Signal)

Eine Software-Aktualisierung ist
verfügbar (nur für Kunden des IDEXX
SmartService*)

Verbunden mit der IDEXX VetLab* Station.
Kommunikation über eine kabelgebundene
Verbindung (ein rotes Symbol zeigt an,
dass das Analysegerät offline ist)

Warnmeldung (ein graues Symbol zeigt
an, dass die Warnung angezeigt undnicht behoben wurde)Ladezustand des Akkus bei 100 %Ladezustand des Akkus bei 75 %Ladezustand des Akkus bei 50 %Ladezustand des Akkus bei 25 %Ladezustand des Akkus sehr niedrig
Akku erschöpft
Akku wird aufgeladen

Fehlermeldung (ein graues Symbol zeigt
an, dass der Fehler angezeigt und nicht
behoben wurde)
Verbindung mit der IDEXX VetLab Station
unterbrochen

Technische Daten und Sicherheitsinformationen
Eingang: 100-240 V AC, 47-63 Hz, 0,3 A
Ausgang: 5 V DC, 1 A

Verbrauch

Betrieb: maximal 5 W

Akkus

Hauptakkus
Bestellnummer: 79-91922-01
Ausgang: 3,7 V DC, 2600 mAh
Uhrenbatterie
Bestellnummer: BR1225
Ausgang: 3 V DC, 48 mAh

Betriebsbedingungen

Betriebstemperatur: 15-30 ºC

Sicherheitsvorkehrungen• Das Analysegerät SNAP Pro* enthält keine
Komponenten, die vom Benutzer gewartet werden
können. Nehmen Sie das Analysegerät nicht
auseinander. Bei etwaigen Fragen bezüglich desService wenden Sie sich bitte an den Technischen
Kundendienst von IDEXX.
• Führen Sie außer den zur Verwendung mit dem
Analysegerät SNAP Pro zugelassenen SNAP* Tests
nichts in die Analysekammer ein.
• Schützen Sie das Analysegerät vor Wärmequellenoder Flammen.

• Verwenden Sie nur den im Lieferumfang enthaltenen
Stromadapter.
• Verwenden Sie nur die von IDEXX mitgelieferten Akkus.
• Entsorgen Sie die gebrauchten Akkus inÜbereinstimmung mit den örtlichen Vorschriften.
Diese Akkus enthalten Lithium. Das Recycling ist in
der Europäischen Union erforderlich und wird in allen
anderen Ländern nachdrücklich empfohlen.
• Eine nicht bestimmungsgemäße Verwendung desAnalysegerätes kann dessen technische Sicherheit
beeinträchtigen.
• Das mobile Analysegerät SNAP Pro sollte während
des Betriebs auf einer ebenen Unterlage stehen.
Hinweis: In Erfüllung der
Hochfrequenzexpositionsanforderungenvon Industry Canada (IC) für mobile undBasisstationsübertragungsvorrichtungen muss
während des Betriebs zwischen der Antenne diesem
Analysegerät und Personen ein Abstand von 20 cm
oder mehr eingehalten werden. Um die Einhaltung
dieser Anforderung zu gewährleisten, raten wir vongeringeren Abständen während des Betriebs ab.
Die für diese Übertragungsvorrichtung verwendete/n
Antenne/n darf/dürfen nicht zusammen mit anderen
Antennen oder Übertragungsvorrichtungen aufgestellt
bzw. betrieben werden.

DeutschExterner USB-Stromadapter

Italiano

Configurazione dell’analizzatore

12

Aprire il vano batteria ubicato
sul retro dell’analizzatore e
inserire due batterie (fornite indotazione) facendo entrare leestremità con il polo negativo.

5

1Collegare al cavo di alimentazione l’adattatore
appropriato per la propria regione geografica
attenendosi alle istruzioni riportate di seguito.

4a. Rimuovere l’inserto protettivo facendo
scorrere contemporaneamentel’analizzatore di bloccaggio verso ilcavo di alimentazione.

Nord America
Giappone

Australiab. Inserire per prima la partesuperiore dell’adattatore efarla scattare in posizione.

Regno Unito

EuropaSud America

3

3Collegare il cavo di alimentazione
all’analizzatore e a una presa elettrica
appropriata. L’indicatore di stato dovrebbeora essere giallo e rimanere acceso.
Nota - La completa ricarica
dell’analizzatore (l’indicatore di stato
diventerà verde) potrebbe richiedere finoa 8 ore. È possibile usare l’analizzatore
durante il caricamento.

5
6

Per il collegamento dell’analizzatorere a una IDEXX VetLab* Station in modalità wireless, passare al punto 5.
Nota - È necessario un router wireless IDEXX VetLab*.
OPPUREPer la connessione dell’analizzatore a una IDEXX VetLab Station mediante un router cablato IDEXX VetLab*,
collegare l’analizzatore a una porta numerata del router usando un cavo CAT5 (fornito in dotazione).
Premere il tasto grigio di Accensione nella parte superiore del retro dell’analizzatore SNAP Pro per
accenderlo. Quando appare la schermata di avvio, seguire le istruzioni riportate sulla schermata.
Quando richiesto, procedere nel modo descritto
di seguito.
a. Selezionare la propria posizione e premere
l’icona Next (Avanti) .
b. Se si collega l’analizzatore a una IDEXX VetLab
Station, premere Yes (Sì), selezionare la rete
IDEXX dall’elenco e premere quindi l’icona
Next (Avanti) . Se la rete IDEXX non compare
nella lista, premere l’icona Refresh (Rigenera) .
OPPURE
Se non si collega l’analizzatore a una IDEXX
VetLab Station, premere No e quindi l’icona
Next (Avanti) . Quando richiesto, confermare
che non si intende collegare l’analizzatore a una
IDEXX VetLab Station.
c. Selezionare il formato della data desiderato,
inserire la data corrente utilizzando il tastierino
numerico e premere l’icona Next (Avanti) .

e. Se si desidera che l’analizzatore emetta una
segnalazione acustica in caso di errore,
avvertenza o visualizzazione di altre informazioni,
selezionare l’icona Sound (Audio)
e quindi lecaselle di controllo accanto alle opzioni richieste
e indicare se si desidera che l’audio sia Soft
(Basso) oppure Loud (Alto). Selezionare quindi
il segno di spunta .
OPPURE
Se non si desidera che l’analizzatore emetta
una segnalazione acustica, selezionare l’icona
No audioe selezionare quindi il segno dispunta .
Nota - In caso di errori critici verrà sempre
emessa una segnalazione acustica.
f. Quando richiesto, premere OK per riavviare
l’analizzatore.

d. Selezionare il formato dell’ora desiderato, inserire Serve aiuto? Visitare il sito idexx.com/prohelp
l’ora corrente utilizzando il tastierino numerico e (in inglese) per visionare la documentazione dipremere l’icona Next (Avanti) .
supporto e alcuni video utili per l’installazione.

Italiano4Esecuzione di un test SNAP
Con IDEXX VetLab* Station
1. Premere Analyze Sample (Analizza campione)
nella schermata iniziale della IDEXX VetLab Station
per inserire le informazioni sul paziente
e avviare l’analisi.
OPPURESe la IDEXX VetLab Station è integrata in un sistema
di gestione dell’ambulatorio, passare al punto 2 oselezionare il paziente dalla dalla lista Pending (Inattesa) o Census (Censimento) sulla schermata inizialedella IDEXX VetLab Station.
Nota - Tutti i pazienti che appaiono nella lista Pending(In attesa) della IDEXX VetLab Station appaiono anche
sulla schermata iniziale di SNAP Pro*.
OPPURE
Premere New Patient (Nuovo paziente) sulla
schermata iniziale di SNAP Pro per inserire il nomedel paziente e l’ID del cliente. Dopo aver inserito
tutte le informazioni, selezionare il segno di spunta
e passare al punto 3.
2. Una volta visualizzato il nome del paziente nella listadella schermata iniziale di SNAP Pro, selezionare ilnome del paziente e selezionare quindi il segno dispunta .

3. Preparare il test SNAP*,
ma non procedere
all’attivazione.
4. Inserire delicatamente il test
SNAP nella porta ubicata nella
parte inferiore dell’analizzatore(inserendo prima la parte dei
pozzetti campione) sino aquando non scatta in posizione.
Una volta completata l’analisi,
i risultati verranno visualizzati
nella relativa schermata e inviati
automaticamente alla IDEXX
VetLab Station.
Suggerimento: il coloredella spia di stato indica
i risultati dell’analisi:
Verde — tutti i risultati sono
negativi
Blu — almeno un
risultato è positivo
Giallo — è necessaria
una conferma visiva(seguire le istruzioni
fornite nella pagina seguente).

Senza IDEXX VetLab Station

Necessità di conferma dei risultati

1. Premere New Patient (Nuovo paziente) sulla
schermata iniziale di SNAP Pro per inserire ilnome del paziente e l’ID del cliente. Dopo averinserito tutte le informazioni, selezionare quindi
il segno di spunta .

Laddove nella finestra dei risultati fossero presenti
punti di un colore molto chiaro o imprevisto (es.
presenza di residui o tonalità bluastra), potrebbe venire
richiesta la conferma di alcuni o tutti i risultati del test. I
risultati da confermare saranno evidenziati in giallo.

3. Inserire delicatamente
il test SNAP nella porta
ubicata nella parte inferioredell’analizzatore (inserendo
prima la parte dei pozzetti campione) sino
a quando non scatta in posizione (vedere
l’illustrazione a sinistra).
Una volta completata l’analisi, i risultati
verranno visualizzati nella relativa schermata
e salvati automaticamente nell’analizzatore.
Suggerimento: il colore della spia di stato
indica i risultati dell’analisi:
Verde — tutti i risultati sono negativi
Blu — almeno un risultato è positivo
Giallo — è necessaria una conferma visiva(seguire le istruzioni fornite sulla destra).

1. Qualora il risultato evidenziato sia da ritenersi
positivo, selezionare l’icona con il segno più (+).
OPPURE
Qualora il risultato evidenziato sia da ritenersi
negativo, selezionare l’icona con il segno meno (–).
2. Una volta confermati tutti i risultati, selezionare ilsegno di spunta . I risultati saranno memorizzati
nell’analizzatore e inviati alla IDEXX VetLab Station(se applicabile).
Nota: per annullare il test e tornare alla schermata
iniziale, selezionare l’icona di chiusura.

Italiano

2. Preparare il test SNAP,
ma non procedere
all’attivazione.

Pulizia dell’analizzatore
Pulizia dell’esterno e dello schermo (eseguire secondo necessità)
Pulire l’esterno e lo schermo utilizzando un panno che non lascia residui, inumidito con alcool isopropilico al 70%,
sapone liquido delicato e acqua, oppure una soluzione di candeggina al 10%. NON pulire le ventole poste sotto il
pulsante di accensione o le porte USB/Ethernet/di alimentazione.

Pulizia dei componenti interni (eseguire settimanalmente o secondo necessità)

1. Spegnere l’analizzatore
e scollegare il cavodi alimentazione e/o
Ethernet.

2. Utilizzare un cacciavite
a punta piatta o
una moneta per far
compiere un quarto di
giro in senso antiorario
ai due (2) fermi di color
argento.

3. Rimuovere la basedall’analizzatore.

4. Pulire l’intera baseutilizzando un pannoche non lascia residui,
inumidito con alcool
isopropilico al 70%,
sapone liquido delicato
e acqua, oppure
una soluzione dicandeggina al 10%.

Attivatore
Asse
Lente
Riflettori

5. Pulire i tre (3) riflettori,
l’asse e l’attivatore
utilizzando un pannoche non lascia residui,
inumidito con alcool
isopropilico al 70%,
sapone liquido delicato
e acqua, oppure una
soluzione di candeggina
al 10%. NON pulire
la lente.

6. Dopo che la basesi sarà asciugata,rimontarla
sull’analizzatore
facendo attenzione a
fissarla saldamente
nella sua sede.

7. Ruotare i fermi di color
argento di un quarto di
giro in senso orario per
fissare in sede la base.

8. Verificare che labase e i fermi dicolor argento siano
bloccati in posizionee non sporgano
dall’analizzatore.
Un bloccaggio
non corretto dellabase e dei due
fermi potrebbe
compromettere
le prestazioni
dell’analizzatore.

Interpretazione delle icone di SNAP Pro
Icone di SNAP Pro*
Icona

Descrizione

Icona

Descrizione

Stato del segnale wireless (5 barre grigie
indicano il segnale migliore, 5 barre rosse
indicano l’assenza di segnale)È disponibile un aggiornamento software(solo per i clienti delle soluzioni IDEXX
SmartService*)

Analizzatore collegato e comunicantecon la IDEXX VetLab* Station attraverso
una connessione cablata (un’icona rossa
indica che l’analizzatore non è in linea)

Notifica di avviso (un’icona grigia indica
che l’avvertenza è stata letta ma che
il problema non è ancora stato risolto)

Batteria carica al 100%
Batteria carica al 75%
Batteria carica al 50%
Batteria carica al 25%
Batteria criticamente bassa
Batteria scarica
Batteria in carica

Notifica di errore (un’icona grigia indica
che l’errore è stato letto ma non è ancora
stato risolto)
Interruzione della comunicazione con
la IDEXX VetLab Station

Specifiche tecniche e informazioni per la sicurezza
Ingresso: 100–240 V CA, 47-63 Hz, 0,3 A
Uscita: 5 V CC, 1 A

ConsumoFunzionamento: massimo 5 W

Batterie

Batterie principali
Codice prodotto: 79-91922-01
Uscita: 3,7 V CC, 2.600 mAh
Batteria dell’orologio
Codice prodotto: BR1225
Uscita: 3 V CC, 48 mAh

Condizioni operative

Temperatura di funzionamento: 15-30 °C

Precauzioni di sicurezza• L’analizzatore SNAP Pro* non contiene componenti
riparabili dall’utente. Non smontare. Per qualsiasi
problema, contattare l’Assistenza tecnica IDEXX.
• Inserire nella porta solo i test SNAP* approvati perl’uso con l’analizzatore SNAP Pro.
• Proteggere l’analizzatore da fonti di calore ofiamme.
• Utilizzare esclusivamente l’adattatore dialimentazione fornito in dotazione.
• Utilizzare esclusivamente le batterie fornite da IDEXX.

• Smaltire le batterie usate solo in conformità alla
normativa locale. Queste batterie contengono litio.
Il loro riciclaggio è richiesto nell’Unione europea eviene vivamente incoraggiato in tutti gli altri paesi.
• L’utilizzo dello strumento diversamente da quantospecificato potrebbe compromettere la protezione
fornita dall’analizzatore.
• Collocare l’analizzatore SNAP Pro su una superficie
piatta durante il funzionamento.
Avviso - Per soddisfare i requisiti di esposizione alle
radiofrequenze (RF) stabiliti da Industry Canada (IC)relativi agli analizzatori mobili e di trasmissione dallastazione base, è necessario mantenere una distanza
di almeno 20 cm tra l’antenna dell’analizzatore e lepersone durante il suo funzionamento. Per assicurare
la conformità, è sconsigliato il funzionamento a una
distanza più ravvicinata. Qualsiasi antenna usata
per questo trasmettitore non deve collocata o fatta
funzionare con altre antenne o trasmettitori.

ItalianoTrasformatore esterno USB

英語

IDEXX スナップ Pro*のセットアップ

1

2

IDEXX スナップ Proの背面にあるバッテリーコンパートメ
ントを開け、マイナス極を先にして、コンパートメントにバッテリー2個（付属）を入
れます。

5

1

 域に適したアダプターを、次のステップに地沿って電源コードに接続：

4
a. ロックキーをコード側にスライドし保護インサートを取り
外します。

北アメリカ
日本

オーストラリアb. アダプターを上面から先に差し込み、その後カチッと音がするまで嵌め込みます。

英国

ヨーロッパ
南アメリカ

3

3

電源コードをIDEXX スナップ Pro
および適切な電源出力に接続します。ステータスランプが黄色で点灯します。
注意：IDEXX スナップ Proを完全に充電するには8時間必要です（完全に充電されるとステータス
ランプは、緑になります）。充電
中でもIDEXX スナップ Proを使用することが可能です。

4

IDEXX ベットラボ*ステーションにIDEXX スナップProを無線で接続する場合は、ステップ5に進みます。
注意：IDEXX 無線ルーターが必要です。
またはIDEXX ベットラボステーションにIDEXX スナップProを接続する場合は、通信ケーブル（付属）を使用してIDEXX有線ルーター上の番号のついたポートにIDEXX スナップ Proを接続します。

5
6

IDEXX スナップ Proの背面上部の 電源ボタンを押し、電源をオンにします。起動画面の表示後、 画面上の指示に従ってください。
a. 地域を選択し、次へす。

アイコンをタップしまb. IDEXX スナップ ProをIDEXX ベットラボステー
ションに接続する場合は、はいをタップし、一覧からIDEXXネットワーク (IDEXXw1) を選択し、次へアイコンをタップします。IDEXXネットワーク （IDEXXw1) が一覧に表示されない
場合は、更新 アイコンをタップします。
またはIDEXX スナップ ProをIDEXXベットラボステーシ
ョンに接続しない場合は、 いいえをタップし、次へアイコンをタップします。指示に従って、IDEXX スナップ ProがIDEXX ベットラボス
テーションに接続されていないことを確認します。
c. 希望の日付の書式を選択し、数字キーパッドを使用して当日の日付を入力し、次へ アイコンをタップします。

d. 希望の時間書式を選択し、数字キーパッドを使用して現在の時刻を入力し、次へアイコンを
タップします。
e. エラー、警告があった際などに、ビープ音を鳴らしたい場合は、サウンド
アイコンを選
択し、各オプションのチェックボックスを選択
し、ソフトまたは大きくを選択します。チェッ
クマーク
アイコンをタップします
または
IDEXX スナップ Proにビープ音を鳴らさないよ
うにしたい場合は、ミュート
アイコンを選択し、チェックマーク
アイコンをタップします。
注意：重大なエラーの場合は必ずビープ音がなります。
f. 画面の指示に従い、OKをタップしてIDEXX スナ
ップ Proを再起動します。

日本語

画面上の指示に従って、以下の操作を行います：

スナップ検査方法

IDEXX ベットラボ*ステーションを使用する場合
1. IDEXX ベットラボステーション初期画面上の検査をタップして、患者情報を入力し、検査を開始します。
または
IDEXX ベットラボステーションが顧客情報管理システムと連携している場合、ステップ2に進む
か、IDEXX ベットラボステーション初期画面上の「検査待ちの患者」または「受付済みの患者」リス
トから患者を選択します。
注意：IDEXX ベットラボステーション上の検査待ちの患者リストに表示された全患者は、IDEXX スナッ
プ Pro*初期画面上にも表示されます。
またはIDEXX スナップ Pro初期画面上の新規患者をタップし、患者名とオーナーIDを入力します。情報入力が終わったら、チェックマーク
アイコンをタップ
し、ステップ3に進みます。
2. IDEXX スナップ Pro初期画面上のリストに患者名が表示されたら、患者名をタップし、チェックマーク
アイコンをタップします。

3. スナップ*検査を準備します。
アクティベーターを手で押さないでください。
4. スナップを、IDEXX スナッ
プ Proのポートに（検体注入口の端を先にして）カチッと音がするまでゆっくり挿入します。

検査が終了すると結果が結果画面に表示され、IDEXX ベットラボステーションステーションに自動送信されます。
ヒント：ステータスランプの色が検査結果を示します：
緑-結果が全て陰性又は正常
青-1つ以上の結果が陽性又は高値
黄-目視確認が必要（次ページの指示に従います）

IDEXX ベットラボステーションを使用しない場合
1. IDEXX スナップ Pro初期画面上の新規患者をタップし、患者名とオーナーIDを入力します。
情報の入力が終わったら、チェックマークアイコンをタップします。
2. スナップを準備します。
アクティベーターを手で押さないでください。

3. スナップ検査を、IDEXX ス
ナップ Proの下のポートに（検体注入口の端を先にして）カチッと音がするまでゆっくり挿入します。

検査が完了したら、結果が結果画面に表示され、自動的にIDEXX スナップ Proに保存されます。

結果の目視確認が必要な場合

結果ウィンドウに非常に明るい点また予測していない色（破片または淡い青）がある場合、検査結果の一部
またはすべての確認を促される場合があります。確認が必要な結果は黄色くハイライトされます。
1. ハイライトされた結果が陽性だと考える場合は、
プラスアイコンをタップします。
またはハイライトされた結果が陰性だと考える場合は、マイナスアイコンをタップします。

2. 結果の確認が完了したら、チェックマーク
アイ
コンをタップします 結果は、検査器に保存され、（IDEXX ベットラボステーションを使用した場合）
IDEXX ベットラボステーションに送られます。
注意：検査をキャンセルし、初期画面に戻るには、
閉じるアイコンをタップします。

日本語

ヒント：ステータスランプの色が検査結果を示
します：
緑-結果が全て陰性
青-1つ以上の結果が陽性
黄-目視確認が必要（右側の指示に従ってください。）

IDEXX スナップ Proのクリーニング
外部と画面のクリーニング（必要に応じて実施）

糸くずの出ない布に70％のイソプロピルアルコール、中性洗剤、水、または10％漂白剤溶液を少し含ませて外部とタッチスクリーンをクリーニングします。電源ボタンまたはUSB／電源／イーサネットポート下の通気口は拭かない でください。

内部のクリーニング（週1回または必要に応じて実施）

1. IDEXX スナップ Proの電源を切り、電源コー
ドおよびイーサネット
コードの接続を解除してください。

2. マイナスドライバーなどを使用し、2つの銀色の留め具を反時計回りに4分の1回転します。

3. IDEXX スナップ Proか
ら底面を外します。

4. 糸くずの出ない布または綿棒に70％のイソプ
ロピルアルコール、中性洗剤および水、または10％漂白剤溶液を少し含ませて底面全体を清掃します。

アクティベーター
軸
レンズ
リフレクター

5. 糸くずの出ない布または綿棒に70％のイソプ
ロピルアルコール、中性洗剤および水、または10％漂白剤溶液を少し含ませて3つの白色リフレクター、軸およびアクティベーターを清掃します。レンズは清掃しないでください

6. 底面を乾燥させたあと、IDEXX スナップ
Proにとりつけ、正しい場所にしっかりはま
ったことを確認します。

7. 銀色の留め具を時計回りに4分の1回転させ、底面をロックします。

8. 底面と銀色の留め具がしっかりと取り付けられ、ロックされたことを確認します。底面と留め具が正しくロックされないと、IDEXX ス
ナップ Proの性能に影響がでることがあります。

IDEXX スナップ Proアイコンの説明
IDEXX スナップ Pro*アイコン
アイコン

説明

ワイヤレス信号の状態（棒が5本 = 最適な信号状態；棒が0本 = 信号が来ていな
い状態）
IDEXX ベットラボステーションに有線接続中（赤色アイコンは、IDEXX スナ
ップ Proがオフラインであることを示します）
バッテリー残量100％
バッテリー残量75％
バッテリー残量50％
バッテリー残量25％

バッテリー残量がわずかです。
バッテリー残量がありません。

アイコン

説明

バッテリー充電中です。

ソフトウェアのアップグレードが可能です（IDEXXスマートサービス*に接続中の場合）
警告通知（灰色アイコンは、警告確認
済みで解決されていないことを示します）
エラー通知（灰色アイコンは、エラー確認済みで、問題が未だ解決されていないことを示します）
IDEXX ベットラボステーションとの通信が遮断されました

技術仕様および安全性情報
外部USB電源アダプター

入力：100–240 V AC、47–63 Hz、0.3 A
出力：5 V DC、1 A

電力消費量

動作条件：最大5W

バッテリー

メインバッテリー
品番：79-91922-01
出力：3.7 V DC、2,600 mAh
待機バッテリー
品番：BR1225
出力：3 V DC、48 mAh•弊 社が供給したバッテリーのみ使用してください。
•使 用済みバッテリーは、自治体の指示に従って廃棄してください。これらのバッテリーはリチウムが含有されています。リサイクルは欧州連合で必要とされており、その他全ての国で強く奨励されています。
• IDEXX スナップ Proを指定の設置条件・使用方法以外で使用した場合、機器の保護機能が無効になることがあります。
• IDEXX スナップ Proは、操作中平らな表面に設置してください。

動作条件

動作温度：15°C～30°C（59°F～86°F）• IDEXX スナップ Pro*には、ユーザーによる保守が可能な部品は含まれていません。IDEXX スナップ
Proを分解、改造しないこと。修理が必要な場合
は、IDEXX テクニカルサポートまでご連絡ください。
• IDEXX スナップ Proでの使用が認められたスナップ*
以外をスナップ検査ポートに入れないこと。
•熱 源や火気から離れた場所にIDEXX スナップ Proを設置し保護してください。
•付
 属のAC電源アダプターのみを使用してください。

日本語

安全上の注意使用上の注意

1. 本装置を分解、改造しないこと。

2. 本装置の取扱いには十分注意すること。
本体を落としたり、強く振動させたりすると内部のメカニズム、電気回路、
メモリーや光学系に障害を起こす恐れがあるので、衝撃を与えないように注意すること。
3. 化学薬品の保管場所やガスの発生する場所に設置しないこと。
4. 湿度の高い場所、直射日光が当たる場所、ほこりの多い場所などには設置しないこと。
5. 電気的ノイズを発生するものや、強い磁界を発生する装置の近くに設置しないこと。
6. 必ず熱源や火気から離れた場所に設置すること。

7. 流しや洗面台など水周りの場所を避け、平らな場所に設置すること。

8. 設置の際は、本装置の換気口を塞がないように注意すること。

9. 付属の電源コードおよびバッテリーを使用すること。
また、電源の周波数と電圧を十分に確認し、コードの接続が正確かつ完全であることを確認すること。

10. クリーニングの前には、電源を切ること。

11. スナップポートの近くでは、
エアゾール、
スプレー溶
液、
あるいは溶剤を使用しないこと。
また、
スナップ
ポートには、専用のスナップ以外の物を挿入しないこと。

12. 取扱説明書に従い、正しい操作をすること。

简体中文设置分析仪

1打开分析仪背面的电池盒，放入两节电池（随附），负极先
入电池盒。

5

1

2

 使用以下步骤将适合您所在地区的适配器请连接到电源线：4
a. 在 将锁定键向电源线方向滑动的同时，取下保护插件。

北美洲
日本

澳大利亚

b. 先 插入适配器的顶部，然后
卡入到位。

英国

欧洲
南美洲

3

3

分析仪接上电源线并插入适当的电源插座。状态指示灯应常
亮黄色。
注意：分析仪充满电最多需要
8小时（满电后状态指示灯将变为绿色）。在充电时可使用分
析仪。

5
6

如果您打算将分析仪通过无线方式连接到IDEXX VetLab* Station，请继续执行步骤5。
注意：需要使用IDEXX VetLab*无线路由器。
或如果要使用IDEXX VetLab*有线路由器将分析仪连接到IDEXX VetLab Station，请使用CAT5电缆（随附）将分
析仪连接到路由器上带有编号的端口。
按下SNAP Pro分析仪背面顶部的灰色电源
操作。

出现提示时，请执行以下操作：a. 选择您的位置，然后点击下一步按钮以开机。显示启动界面后，请遵循屏幕上的指示进行图标。

b.	如果您要将分析仪连接到IDEXX VetLab Station，
请点击是，从列表中选择IDEXX网络，然后点击下一步图标。如果IDEXX网络未列出，请点击
刷新
图标。
或

如果您不打算将分析仪连接到I D E X X V e t L a b
Station，请点击否，然后点击下一步图标。出现提示时，确认您不打算将分析仪连接到IDEXX
VetLab Station。

c.	选择所需的日期格式，使用数字键盘输入当天的日期，然后点击下一步图标。

d.	选择所需的时间格式，使用数字键盘输入当前时间，然后点击下一步图标。

e.	如果您希望在显示错误、警告或其他信息时让分析仪发出哔哔声，请选择声音
图标，选择所需选项旁边的复选框，然后选择希望声音为轻柔或大声。然后点击复选标记图标。
或如果您不希望分析仪发出哔哔声，请选择静音
图标，然后点击复选标记图标。

注意：对于严重错误，分析仪将始终发出哔哔声。

f. 出现提示时，点击确定以重新启动分析仪。

如需帮助，请访问idexx.com/prohelp
观看安装指导视频或查看帮助文档。

简体中文

4

进行SNAP 检测

装配IDEXX VetLab* Station时

1.	点击IDEXX VetLab Station主屏幕上的分析样本，输入宠物信息并开始运行。
或如果IDEXX VetLab Station与诊室管理系统整合在一起，请继续执行步骤2，或从IDEXX VetLab Station主屏
幕的“待测样本”或“所有样本”列表中选择宠物。
注意：出现在IDEXX VetLab Station“待测样本”列表中的所有宠物也会显示在SNAP Pro*主屏幕上。
或点击SNAP Pro主屏幕上的新的宠物，输入宠物名和宠主ID。当您完成信息输入时，请点击复选标记图标并继续执行步骤3。

2.	一 旦宠物名出现在SNAP Pro主屏幕的列表中，请点
击宠物名，然后点击复选标记图标。

3. 请准备SNAP*检测，但不要激活检测。

4. 轻轻将S N A P 检测板滑入分析
仪底部的端口（先插入有孔端），直到其卡入到位。
当分析完成时，结果会显示在结果屏幕上，并自动发送到IDEXX VetLab Station。
提示：状态指示灯的颜色表示
检测结果：绿色—所有结果均为阴性绿色蓝色—至少一项结果为蓝色阳性
黄色—需要肉眼确认
黄色（使用下一页的
说明）

未装配IDEXX VetLab Station时

需要确认结果时

2. 请准备SNAP*检测，但不要激活检测。

1. 如果您认为突出显示的结果为阳性，请点击加号图标。

3. 轻轻将SNAP检测板滑入分
析仪底部的端口（先插入有孔端），直到其卡入到位（参见左侧的示意图）。

当分析完成时，结果会显示
在结果屏幕上，并自动保存在分析仪上。
提示：状态指示灯的颜色表示检测结果：
绿色—所有结果均为阴性绿色蓝色—至少一项结果为阳性蓝色
黄色—需要肉眼确认（使用右侧的说明）
黄色

如果结果窗口中出现极浅点或非预期的显色（例如碎片或蓝色调），则可能会提示您对部分或全部检测结果进
行确认。需要确认的结果以黄色突出显示。
或如果您认为突出显示的结果为阴性，请点击减号图标。

2. 完成结果确认时，请点击复选标记 图标。然后，结
果会保存在分析仪上，并发送到IDEXX VetLab Station（如果适用）。
注意：如要取消检测并返回到主屏幕，请点击关闭图
标 。

简体中文

1. 点击SNAP Pro主屏幕上的新的宠物，输入宠物
名和宠主ID。当您完成信息输入时，请点击复选标记图标。

清洁分析仪

清洁外壳和屏幕（根据需要执行）

使用无绒布蘸70%异丙醇、中性肥皂水或10%漂白剂溶液，清洁外壳和屏幕。请勿擦拭电源按钮下方的通风孔或USB/电源/网线端口。

清洁内部组件（每周或根据需要执行）

1. 关闭分析仪并断开电源
和/或网线连接。

2. 使用平头螺丝刀或硬币将两个银色紧固件逆时
针旋转四分之一圈。

3. 取下分析仪的底座。

4. 使用无绒抹布或棉签蘸
70%异丙醇、中性肥皂
水或10%漂白剂溶液，清洁整个底座。

激活器
轴
镜片
反射器

签蘸70%异丙醇、中性
肥皂水或10%漂白剂溶
液，清洁3 个白色反射器、轴和激活器。请勿
清洁镜片。

6. 底座干燥后，将其重新
装回分析仪，确保锁定到位。

7. 沿 着 顺 时 针 方 向 将 银色紧固件旋转四分之一圈，以将底座锁定
到位。

8. 确保底座和银色紧固件
与分析仪平齐并锁定到位。未正确锁定底座和紧固件会影响分析仪的
性能。

简体中文

5. 使用新的无绒抹布或棉解读SNAP Pro图标
SNAP Pro*图标图标说明无线信号状态（5个灰色条表示最佳
信号；5个红色条表示无信号）

通过有线连接IDEXX VetLab* Station并与其通讯（红色图标表示分析仪离线）
电池电量为100%
电池电量为75%
电池电量为50%
电池电量为25%
电池电量极低
电池没电

图标

说明

电池正在充电

软件可以升级
（仅限SmartService*解决方案客户）

警告通知（灰色图标表示已查看警告，但未解决）错误通知（灰色图标表示已查看错误，但未解决）
与IDEXX VetLab Station的通讯中断技术规格和安全信息
外部USB电源适配器

输入：100–240 V AC，47–63 Hz，0.3 A
输出：5 V DC，1 A

功耗

工作中：最大5 W电池主电池
部件号：79-91922-01
输出：3.7 V DC，2600 mAh
手表电池
部件号：BR1225
输出：3 V DC，48 mAh

操作条件

工作温度15°C-30°C (59°F-86°F)•仅 根据您当地的法规条例处理使用过的电池。这些电
池含有锂。欧盟要求将其回收，在所有其他国家/地区都强烈建议回收。
•如
 果以非指定方式使用设备，可能会损坏分析仪自有的保护。
•在 操作过程中，应将SNAP Pro分析仪放在平坦表
面上。

注意：为符合移动和基站传输分析仪的加拿大工业(IC)射频(RF)暴露量要求，在操作过程中，此分析仪的天线与人员之间的距离应保持在20 cm或以上。为确保合规性，建议在低于该距离要求时不要进行操作。不得将此发射器使用的天线与其他任何天线或发射器放置在一起
或一起使用。

安全注意事项

• SNAP Pro*分析仪不包含用户可维修的组件。请勿拆
卸。如有任何维修问题，请致电IDEXX技术支持部。
•除 批准用于SNAP Pro分析仪的SNAP*检测板外，请
勿在端口中插入任何其他物品。
•仅 使用提供的电源适配器。
•仅 使用IDEXX提供的电池。

简体中文

•分
 析仪应远离热源或火焰，以保护分析仪。

Authorized representative in the European Community
Représentant agréé dans la Communauté européenne
Representante autorizado para la Comunidad Europea
Autorisierte EG-Vertretung
Rappresentante autorizzato nella Comunità Europea
欧州共同体の正式代表者
欧洲共同体授权代表

Temperature limitation
Limite de température
Limitación de temperatura
Zulässiger Temperaturbereich
Limite di temperatura
温度制限
温度限制

Manufacturer’s declaration that product meets applicable European
Community directives
Déclaration du fabricant stipulant que le produit est conforme aux
directives de la Communauté européenne en vigueur
Declaración del fabricante de que el producto cumple las directivas de la
Comunidad Europea aplicables
Erklärung des Herstellers bezüglich der Erfüllung der geltenden
Richtlinien der Europäischen Gemeinschaft
Dichiarazione di conformità del produttore alle direttive vigenti nella
Comunità Europea
本製品は該当する欧州共同体の指令に適合しているという製造業者による宣言
制造商声明产品符合适用的欧洲共同体指令

Consult instructions for use
Consulter les instructions d’utilisation
Consulte las instrucciones de uso
Gebrauchsanleitung beachten
Consultare le istruzioni per l’uso
使用方法を参照参阅使用说明书

Product complies with applicable Electromagnetic Compatibility (EMC)
standard for Australia
Produit conforme à la norme australienne en matière de compatibilitéélectromagnétique (EMC)
El producto cumple la norma de Compatibilidad Electromagnética (CEM)
aplicable en Australia
Das Produkt erfüllt die Normen für elektromagnetische Verträglichkeit(EMV) in Australien
Il prodotto è conforme allo standard sulla compatibilità elettromagnetica(EMC) applicabile in Australia
本製品はオーストラリアで適用される電磁両立性（EMC）規格に準拠しています产品符合适用的澳大利亚电磁兼容性标准

Recyclable
Recyclable
Reciclable
Wiederverwendbar
Riciclabile
リサイクル可能可回收利用This symbol is placed in accordance with the European Union Directive 2002/96 on the Waste Electrical and Electronic Equipment (the WEEE Directive). If disposed of within the European
Union, this product should be treated and recycled in accordance with the laws of your jurisdiction implementing the WEEE Directive. For additional information, visit idexx.com/weee.
Ce symbole est apposé conformément à la Directive 2002/96 de l’Union européenne relative aux déchets d’équipements électriques et électroniques (DEEE). S’il est mis au rebut au sein de l’Union
européenne, ce produit doit être traité et recyclé conformément aux lois de la juridiction applicable, dans le respect de la Directive DEEE. Pour plus d’informations, visitez le site idexx.com/weee.
Este símbolo se utiliza conforme a la directiva 2002/96 de la Unión Europea sobre residuos de aparatos eléctricos y electrónicos (directiva WEEE, por sus siglas en inglés). En caso de sereliminado dentro de la Unión Europea, este producto deberá ser tratado y reciclado de conformidad con la legislación de la jurisdicción competente que aplique la directiva WEEE. Paraobtener más información, consulte idexx.com/weee..
Dieses Symbol weist auf die Erfüllung der Richtlinie 2002/96/EG über Elektro- und Elektronik-Altgeräte (WEEE-Richtlinie) hin. Bei der Entsorgung innerhalb der Europäischen Union muss
dieses Produkt in Übereinstimmung mit den lokalen Gesetzen, welche die WEEE-Richtlinie umsetzen, behandelt und recycelt werden. Für ausführlichere Informationen besuchen Sie bitte
unsere Website idexx.com/weee.
Questo simbolo è apposto in conformità alla direttiva dell’Unione europea 2002/96 sui rifiuti di apparecchiature elettriche ed elettroniche (direttiva RAEE). Per lo smaltimento nel territorio
dell’Unione europea, questo prodotto deve essere trattato e riciclato in conformità alle leggi vigenti nello stato dell’utente che ha recepito la direttiva RAEE. Per ulteriori informazioni,
visitare il sito idexx.com/weee.

この記号は、廃電気・電子製品に関する欧州連合指令2002/96（WEEE指令）に従って取り付けられています。欧州連合内で廃棄する場合、本製品はWEEE指令を実施する管轄の法律に従って取り扱い、リサイクルされなければなりません。詳細情報については、idexx.com/weeeをご参照ください。

此符号表示符合电气电子设备废弃物欧盟指令2002/96（WEEE指令）的规定。如果在欧盟内处置，则应根据您实施WEEE指令所在辖区的法律对产品进行处理和回收。如需了解更多信
息，请访问idexx.com/weee。

IDEXX スナップ Pro 動物用一般医療機器｜免疫反応測定装置© 2023 IDEXX Laboratories, Inc. All rights reserved. • 06-29290-09*SNAP Pro, IDEXX VetLab, SNAP, and SmartService are trademarks or registered trademarks of IDEXX Laboratories, Inc.
or its affiliates in the United States and/or other countries.