F388A

WAVEFORM MONITOR FOR SENSORSWITH ANALOGUE (VOLTAGE/CURRENT) OUTPUT

4000 times/sec. high-speed processing
Analog monitor output
Voltage output is proportionate to the input signal making
the recording on recorder convenient.
At voltage input: Approx. 0.6 V per 1 V
At current input: Approx. 0.15 V per 1 mA
Variety of interfaces
RS-232C DeviceNet

CC-Link

Ethernet

Saves Measurement
Data in SD Card
Measurement data and set values can be logged(recorded) in the SD Card where it can be retained
as a 100% recorded quality data or be used when setting up equipments
or when performing cause analysis or improvement of problems.
The data can be easily converted to CSV format and is therefore easily edited inMicrosoft Excel or its like.

3.5-inch color LCD module & touch panel
Operation can be effortlessly performed by a direct touch on
the touch panel.

Example data in CSV format

Excellent operability
F388A is right-down demanding on straightforwardness and istherefore made able to automatically mask non-required setting
items and also to display setting in the required sequence whenthat particular set item has specific setting sequence.
I/O Input: Plus common / Minus common shared
It can be connected to various types of external equipments suchas PLCs.

Waveform comparison function

Multi hold function

This function compares the actual measurement waveform against the setup
High/Low limit waveforms and will give out an NG judgment when any of thepoint exceeded the preset High/Low limit waveforms. As it compares the overallmeasurement waveform, accurate judgment can be made even applications
that are unable to narrow down its judgment points.

After the measuring range is segmented, judgment is carried out while the type ofhold (sample, peak, bottom, P-P, max, min, inflection point, End Displacement) isinterchanged as set. The multi hold function can specify the Hi/Lo limit value andtype of hold at each of the segmented range.

Pulse input as a standard equipment
It performs 2-dimensional waveform comparison & multi hold through its pulse
input. On X-axis, pulse input can be connected while on Y-axis, voltage current
output sensor can be connected.
When nothing is connected with X-axis, Waveform Comparison & Multi Holdby the time series can be done.

Judgment results display▲Waveform comparison screen
Hi and Lo limit comparison of overallmeasurement waveform can beperformed.

▲Setup Waveform Creation Screen

The comparison results of Waveform Comparison Function and Multi Hold
Function can be verified on the display. 【Result(List)】(An individual display) and【Result(Single)】(a list display) to selection is possible. (Latest 40 data)

The High/Low limit waveforms can beeasily created on the actual measurement
waveform or on the setup waveform
creation screen.

Result(List)

Result(Single)

SpecificationsSENSOR

Voltage Current input

INPUT

Signal input range-10 to +10 V or -20 to +20 mA
Voltage input: Input impedance 1 M or more
Current input: Input resistance Approx. 250ΩZero Gain Adjustable range
Automatic adjustment by digital operation
Equivalent input calibration range−10 to +10 V or -20 to +20 mA (0 is excluded.)
Equivalent input calibration error
Within 0.1% FS-10 to +10 V or -20 to +20 mA
Actual calibration range* In Approx. -0.02 to +0.02 V or Approx. -0.03 to +0.03 mA,a zero calibration point to calibration is impossible.
Accuracy
Nonlinearity: Within 0.02% FS±1 digit (at 10 V or 20 mA input)
Zero drift: Within 0.2 mV/℃ RTI or Within 0.4 A/℃ RTI
Gain drift: Within 0.01%/℃
Analog filter
Low-pass filter (-6 dB/oct.) Selectable from10, 30, 100, 300 Hz
A/Dconverter
Rate: 4000 times/sec. Resolution: 24 bit (binary)Effective resolution: Approx. 1/30000 to 10 V or 20 mA
Analog voltage output
Output level Approx. 0.6 V per 1 V input orApprox. 0.15 V per 1 mA input Load resistance 2 kΩ or more
Pulse input (open collector)
Maximum input frequency
Internal counting range
Adaptable sensor

DISPLAY

Display

50 kHzApprox. 1000000
Output: Incremental type 2-phase output (A/B signal output)
Also capable of single-phase output (A-phase input used.
All pulses are counted as in the plus direction.)
Output stage circuit specification: open collector(NPN type, Vceo = 30 V or more, Ic = 30 mA or more)

INTERFACE

RS-232C Communication interface232
ODN DeviceNet interface (Options)
CCL CC-Link interface (Options)
ETN Ethernet interface (Options)(Only one option can be installed)

OPTIONISC

Outside dimensions
WeightIndicated value

71(W) × 53(H) mm
320×240 dot

LoadDisplacement-9999 to +9999-9999 to +32000

Decimal point

The decimal place is to be input together with a value at the time of calibration.
0.000, 0.00, 0.0, 0
Number of display times
Fixed at 3 times/sec.

FCN series I/O connector (with cover) ×1, Jumper wire ×1, Operation manual ×1,
1 GByte SD card ×1, Analog I/O connector terminal block (Already mounted on the main unit) ×1,
DeviceNet connector (when DeviceNet option is selected) ×1,
CC-Link connector (when CC-Link option is selected) ×1

ACCESSORIESDTC1SD1GSD2G
CA81-232X
CN52
CN57
CN60
CN71
CN72
CN81
CND01
GMP96×96

Special case
1 GByte SD card
2 GByte SD card
miniDIN-D-Sub9p cross cable 1.5 m
FCN series I/O connector (with cover)
FCN series I/O connector (with diagonal cover)Circular DIN 8p connector for RS-232C
CC-Link connector
Double row connector for CC-LinkAnalog I/O connector terminal block (Same accessory as the attached one)
DeviceNet connector
Rubber packing* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,fluorescent display tubes and LCDs due to manufacturing process or production lots.

MEASUREMENT Multi Hold Mode 16 ch (Settings can be saved)
FUNCTION
The measurement section is divided, and Hold is switched arbitrarily and judged.
Sample, Peak, Bottom, P-P, Relative Maximum, Relative Minimum, Inflection Point,
Average, End Displacement

Structure of product code

Waveform Comparison Mode 16 ch (Settings can be saved)
The setting waveform of an upper and lower limit is compared with actual
measurement waveform. If the whole measurement waveform serves as a candidate
for upper and lower limit comparison and at least one and exceeds a setting
waveform, it will be judged NG.
EXTERNAL
SIGNALDC 24 V (±15%)
5 W typ.
2 A, 10 msec (at ordinary temperature, cold-start)
Operating temperature range: -10 to +40℃
Storage temperature range: -20 to +60℃
Humidity: 85% RH or less (non-condensing)96(W) × 96(H) × 117.3(D) mm (not including projections)
Approx. 1.0 kgATTACHMENTS

3.5-inch TFT color LCD module
Display area
Dot configuration

I/O Source board

Power supply voltage
GENERAL
SPECIFICATIONS Power consumption
Inrush current typ.
Operating conditions

F388A-SDC

Output signals (16 points)
Hold Result (Load/DPM)/ Overload/ Complete/ Wave Result/ Load OK/ DPM OK/ Run/ SD OK
Output type Sink type/source type selectable. (Source type is optional [ISC].)
Output transistor ON at signal ON. To connect an input unit like a PLC, connect plus commonfor sinktype, and minus common for source type.
Rated voltage 30 V
Rated current 30 mA
Isolation photocoupler

Interface

Standard unit
I/O output

SignStandard
ISCSignStandard

Output type
Sink type(NPN output)
Source type(PNP output)

Interface
RS-232C

in addition the standard interface.

ODN
CCL
ETN

Input signals (16 points)
Load Digital Zero/ DPM Positioning/ Start/ Stop/ Hold/ Reset/ Prohibit Touch Panel/ Backlight On/
Work
Input type

DeviceNet
CC-Link
Ethernet

Plus common/minus common shared.
To connect a transistor, connect NPN output type (sink type) for plus commonand PNP output type (source type) for minus common.
ON voltage
12 V or more
OFF voltage 3 V or less
At 24 V load approx. 5 mA
Isolation photocouplers

External dimension(Side View)(Rear View)

Panel cut-out

96

91

Panel thickness
1.6 to 3.2 mm

17.8

99.5

91

92

Unit : mm

DTC1: Special case

160
182

96

92

(Front View)

1201362m