MODEL 808 ACCELEROMETERSPECIFICATIONSDIMENSIONS

TO-8 Series Accelerometer
2-Wire IEPE Interface
Low-Cost, Embedded Applications
Wide Bandwidth to 6kHz

The Model 808 is a miniature adhesive mount IEPE
accelerometer built on a 3-pin TO-8 header. It is available
in ±10g and ±50g dynamic ranges and offers a flat
frequency response to 6kHz. The accelerometer features ahermetic construction in a TO-8 header configuration.
The model 808 incorporates a stable piezoceramic crystal and an integrated charge converter amplifier
in a 100% shielded housing suitable for many embedded
OEM applications.The accelerometer is offered in twoconfigurations; one for adhesive mounting and one for studmounting.

FEATURES

±10g and ±50g Range
Wide bandwidth up to 6kHz
TO-8 Configuration
Hermetically Sealed-40°C to +100°C Operating Range
Case Grounded Design

APPLICATIONSSENSOR SOLUTIONS /// Model 808 Rev A

Machine Monitoring
Low-Cost OEM Installation
Permanent Structural Study
Embedded Sensing Solution

07/2017

Page 1

MODEL 808 ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 4mA excitation unless otherwise stated. Measurement Specialties reserves the right to update
and change these specifications without notice. Standard product parameters are described in PSC-1001 for Embedded ACAccelerometers.
Parameters
DYNAMIC
Range (g)Sensitivity (mV/g)
Frequency Response (Hz)Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Shock Limit (g)Residual Noise (g RMS)

±105001-30000.3-5000
18000±155000
0.0008ELECTRICAL
Compliance Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Bias Voltage (Vdc)
Output Impedance (Ω)
Full Scale Output Voltage (V)Grounding

18 to 30
2 to 108 to 126 to 13<100±5
Case grounded to mounting surface

ENVIRONMENTAL
Temperature Response (%)
Operating Temperature (°C)
Storage Temperature (°C)

±10-40 to +100
-54 to +125

PHYSICAL
Sensing Element
Case Material
Weight (grams)
Mounting

Piezo-Ceramic, Shear Mode
Stainless Steel
20 (for 10g range), 16 (for 50g range)
Adhesive or Circuit Board (M5 Stud Mounting for -01 Option)

Notes±501002-40000.3-6000
24000±155000
0.001±10% (see Note 1 for ±5% option)
±1dB
±3dB

Broadband 1Hz to 10kHz+125°C option, see Note 2

1

Contact factory for model number to order with ±5% sensitivity tolerance

2

Contact factory for model number to order with +125°C operating temperature

Calibration supplied:

CS-SENS-0100

Wiring color code:

See schematic

Room Temperature-40 to +100°C

NIST Traceable Amplitude Calibration at 80Hz

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. Measurement Specialties, Inc. reserves theright to make changes without further notice to any product herein. Measurement Specialties, Inc. makes no warranty, representation or guarantee regarding thesuitability of its product for any particular purpose, nor does Measurement Specialties, Inc. assume any liability arising out of the application or use of any product or
circuit and specifically disclaims any and all liability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different
applications. All operating parameters must be validated for each customer application by customer’s technical experts. Measurement Specialties, Inc. does notconvey any license under its patent rights nor the rights of others.

SENSOR SOLUTIONS /// Model 808 Rev A

07/2017

Page 2

MODEL 808 ACCELEROMETER

ORDERING INFORMATIONPART NUMBERING

Model Number+Range

808-GGGG -XX
ll_____ Optional Stud Dash Number
l_____ Range (0010 is 10g)

Dash Number-01

Stud Options
M5 x 0.8

Example: 808-0010
Model 808, 10g

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s StandardTerms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// Model 808 Rev A

07/2017

Page 3