FC400-CCL
FC400-DAC
FC400-EIP
FC400-ECT
FC400-232

DIN RAIL MOUNT WEIGHING INDICATOR

Suitable for many weighing applications such ashopper scale, packing scale, weight level meter, etc….
FC400-CCL
FC400-DAC
FC400-EIP
FC400-ECT
FC400-232

A weighing system can be configured easily using CC-Link network.
Either voltage or current output, corresponding to the indicated value, is available.
A weighing system can be configured easily using EtherNet/IP network.
A weighing system can be configured easily using EtherCAT network.
Measured data and results can be collected using RS-232C interface.

High sampling rate & resolution

Various interfaces

High-Speed A/D conversion and powerful digital processing
capability of 1200 times/sec.
High display resolution of 1/100000 (max).

FC400-CCL Equipped with CC-Link, SI/F, USB interface
FC400-DAC Equipped with D/A converter, RS-485, USB interface
FC400-EIP Equipped with EtherNet/IP and USB interface
FC400-ECT Equipped with EtherCAT and USB interface
FC400-232 Equipped with RS-232C, SI/F, USB interfaceex) CC-Link
EtherNet/IP
EtherCATRS-485

Measurement can be performed quickly andprecisely due to high speed A/D conversion.

Application software for USB interface

CC-Link
EtherNet/IP
EtherCAT
RS-485

Load cellinput

With communication through USB
interface, logging, graphdisplay, setting parameters, andcalibration can be done.

FC400

High performance filter & auto filter adjustment
With combination of low pass filter (0.1 to 300 Hz) whichcorresponds to various vibration and moving average filter (OFF,2 to 512) that is effective for periodic vibration, automatic
searching for optimal value with accuracy and stability
can be achieved. (You can manually adjust the settings too)

Thereadingsfluctuate….

By simply
pressing
the autoadjustment
key….

PLCex) CC-Link
EtherNet/IP
EtherCAT
RS-485
RS-232C
D/A converter
SI/F

Sensor
input
SI/F

Display etc.

D/A
FRQYHUWHU Recorder

Stabilized!

RS-232C

Compact size

FC400

It can be fit into a limited space!
Light weight & compact body for 35 mm DIN rail mount.

CC-Link
EtherNet/IP
EtherCAT
RS-485PLCex) USB
USB
FC400

unlPULXE

PC

www.unipulse.tokyo/en/Equivalent input calibration

Equipped with weighing sequence function to control
feeding/discharge gate.

Theoretical calibration can be performed easily by registering thecapacity and rated output of load cells.

Input conversion value display

Multipoint calibration (linearization)

The output signal level of load cell can be displayed in mV/V for
monitor purpose.
Malfunction indicator or faulty sensor can be differentiated easily.

Three additional points can be defined in the middle between
zero and span for better linearity. Eventhough the scale has poor
linearity, it can be corrected to be a highly accurate scale.

FC400

Basic weighing process control function

6-digit display
24 bit A/D converter enables 6-digit display

Specifications
Analog

Display

ExternalsignalInterface

Power supply voltage
Generalspecification Power consumption

DC 24 V (±15%)
3 W typ. (FC400-DAC, FC400-232)
4 W typ. (FC400-CCL, FC400-EIP, FC400-ECT)
Operating conditions Operation temperature: -10 to +50Υ
Storage temperature: -20 to +85Υ
Humidity: 85% RH or less (non-condensing)
Dimension
34(W) × 88(H) × 91(D) mm (Not including projections)
WeightApprox. 210 g (FC400-CCL, FC400-DAC)
Approx. 230 g (FC400-EIP, FC400-ECT)
Approx. 220 g (FC400-232)
Attachments Quick manual…2, Jumper wire…2, Mini driver…1, CC-Link connector…1 (FC400-CCL)
Various I/O connector (FC400-CCL, FC400-EIP, FC400-ECT)…2,(FC400-DAC, FC400-232)…3
CA81-USB: miniUSB-computer USB cable 1.8 m
Optional
CC-Link connector (Same as the attachment)
accessories CN74:
CN75:
CC-Link connector (Y type branch connector)
CN76:
CC-Link connector (Terminator connector)
CN85:
13 p connector for power source/sensor/RS-485(Same as the attachment)
CN86:
3 p connector D/A converter (Common with RS-232C connector)(Same as the attachment)
CN87:
10 p connector for external I/O (Same as the attachment)

DC 5 V±5% Output current: Within 90 mA
DC 2.5 V±5% Output current: Within 45 mA(depending on settings)
Ratiometric method(Up to 6 350 Ω load cells can be connected in parallel)-2.5 to +5.1 mV/V
Signal input range
Zero adjustment range Automatic adjustment by digital processing -0.5 to +2.0 mV/V
Span adjustment range Automatic adjustment by digital processing 0.01 to 3.0 mV/V
Linearization function Up to three point multi-point calibration is possibleusing linearization function
Min. input sensitivity
0.15 ȣV/countAccuracyNon-linearity: Within 0.01% FS
Zero drift:
0.0002% FS/Υ typ. * When calibrated by 3 mV/V with 5 Vof the excitation voltage
Gain drift:
1 ppm/Υ typ.
Digital low-pass filter 0.1 to 300 Hz
FilterMoving average filter OFF, 2 to 512 times
Speed:
1200 times/sec.
A/D converter
Resolution: 24 bit (binary)Character height 8 mm
Display unit
Numerical display by 7-segment green LED
Up to 6 digits.
Display value
Selectable from 1, 3, 6, 13, and 25 times/sec.
Display frequency
RUN, SD, RD, ERR (FC400-CCL)
Status display
MS, NS (FC400-EIP)
RUN, ERR (FC400-ECT)
Output signal
Transistor’s open collector output(5)
Vceo = 30 V, Ic = 50 mA
Input signal
Selectable/configurable(3)
Input is ON when shorted to COM terminal by contact (relayswitch, etc.) or non contact (transistor, open collector, etc.).
* DC 24 V external power supply is required.
CC-Link interface (FC400-CCL)
D/A converter (Voltage and current output) (FC400-DAC)
EtherNet/IP interface (FC400-EIP)
EtherCAT interface (FC400-ECT)
RS-232C interface (Selectable from Modbus-RTU or UNI-Format) (FC400-232)
RS-485 interface (Selectable from Modbus-RTU or UNI-Format) (FC400-DAC)
SI/F 2-wire type serial interface (FC400-CCL, FC400-232)
USB interface
Excitation voltage

CE marking
certification

EMC directive EN61326-1* FC400-ECT is Free Run mode.
* Please note that there are possibilities of individual differences in a color tone on display devices suchas LEDs, fluorescent display tubes and LCDs due to manufacturing process or production lots.

Structure of product codeModelFC400-CCL
FC400-DAC
FC400-EIP
FC400-ECT
FC400-232

Interface
CC-Link, SI/F, USB
D/A converter (Voltage and current output), RS-485, USB
EtherNet/IP, USB
EtherCAT, USB
RS-232C, SI/F, USB

External dimension

DIN rail mountadapter‫ ڦ‬FC400-ECT

unlPULXE

EIP-FA

EIP

88
101
91

34

DIN rail mountadapter

3491

DIN rail mountadapter‫ ڦ‬FC400-232

88
101

88
101
34

DAC-FA

88
101

DAC

CCL-FA

91

34‫ ڦ‬FC400-EIP‫ ڦ‬FC400-DAC

88
101

CCL‫ ڦ‬FC400-CCL

91

DIN rail mountadapter

34

91

DIN rail mountadapter

Unit: mmwww.unipulse.tokyo/en/