AVAILABLE
with Signature
Calibration

SEE PG. IN-5Mid Range
Wet/Wet Differential
Pressure Transducers
Models Z and A-5

50 TO 750 PSID

1-888-282-9891

ACCURACY TO 0.25%
HIGH LINEAMPLIFIED OUTPUT AVAILABLE

www.honeywell.com/sensing

Sensotec Sensors

Models Z and A-5 Mid Range Wet/Wet Differentials are high accuracy, bonded foil strain gagetransducers designed to accept fluid in both ports and measure differential pressure ranges of
50 to 750 psid. Standard features such as overload stops and stainless steel construction provide unit durability in rugged industrial environments. Each series is bi-directional and achieves
accuracies of .25-.5% full scale. The models are available with extended temperature ranges.
Typical applications include flow measurement, depth sensing, pressure equalization, and liquidlevel.

Dimensions
Model Z (Order Code AD122)
Model A-5 (Order Code AD123)
Available Ranges*50; 75; 100; 150; 200; 300; 500; 750 psid
* Stocked ranges are in bold faced print. Model A-5 is stocked complete and semi-complete. Model Z
is stocked semi-complete (3-6 weeks) only.

Model Z or A-5Options (See Appendix)
Temperature compensated 1b, 1c, 1d, 1e, 1f; Internal amps 2b, 2c, 2n or 2N intrinsically safe amp, see pageAP-6; 2j; Amp enhancements 3b, 3d; Electrical termination 6e, 6f, 6g, 6h; Int. shunt cal 8a; Signature calibration 53e
Premium Options: 1g, 1i; 2q; 3a; 5c; 6b, 6c, 6i; 10a; 26a
Accessories: Mating connectors and connector/cable assemblies; Pressure port adapters

DP-6

1-888-282-9891
Model Z

Model A-5

Order Code AD122

Order Code AD123

Pressure Ranges..........................
Accuracy (min) .............................
Non-Linearity (max)......................
Hysteresis (max) ..........................
Non-Repeatability (max)...............
Output (standard) .........................
Resolution ....................................

50 to 750 psid
±0.25% F.S.
±0.15% F.S.
±0.10% F.S.
±0.05% F.S
2mV/V
Infinite

50 to 750 psid
±0.5% F.S.
±0.25% F.S.
±0.13% F.S.
±0.07% F.S
2mV/V
Infinite

ENVIRONMENTAL

Temperature, Operating ...............
Temperature, Compensated ........
Temperature Effect– Zero (max).............................
– Span (max .............................

-65° F to 250° F
60° F to 160° F-65° F to 250° F
60° F to 160° F.5% F.S./100° F.5% Rdg./100° F.75% F.S./100° F
1% Rdg./100° F

Bonded foil
10VDC
Up to 10VDC or AC
5000 megohm @ 50VDC
350 ohm
Included#2 (See Pg. AP-8)
PTIH-10-6P or equiv.
(Hermetic stainless)
PT06A-10-6S or equiv.

Bonded foil
10VDC
Up to 15VDC or AC
5000 megohm @ 50VDC
350 ohm
Included#2 (See Pg. AP-8)
PTIH-10-6P or equiv.

ELECTRICAL

Strain Gage Type .........................
Excitation (calibration) ..................
Excitation (acceptable) .................
Insulation Resistance ...................
Bridge Resistance ........................
Shunt Calibration Data .................
Wiring Code (std) .........................
Electrical Termination (std) ..........
Mating Connector (not incl.) .........

MECHANICALMedia............................................
Maximum Line Pressure ..............
Maximum Overload –safe (either side) ......................
Pressure Port ...............................
Tapped Mounting Hole (2ea) .......
Drain Port .....................................
Wetted Parts Material...................
Weight ..........................................
O-Ring Seals ................................
Dead Volume................................
Case Material ...............................

Gas, Liquid1500 psi

Gas, Liquid1500 psi

1500 psi
1/8-27 NPT
1/4-28 UNF x 5/16 DP
1/8-27 NPT
17-4 PH Stainless
5.0 lbs.
viton
0.25 cu. in.
17-4 Stainless

1500 psi
1/8-27 NPT
1/4-28 UNF x 5/16 DP
1/8-27 NPT
17-4 PH Stainless
5.0 lbs.
viton
0.25 cu. in.
17-4 StainlessOutputs Available .........................
Additional Length..........................

±5VDC, 4-20mA
2"±5VDC, 4-20mA2"MID RANGEINTERNALLY AMPLIFIEDUNITS (Optional)

PT06A-10-6S or equiv.

DIFFERENTIALPERFORMANCE

NOTE: Unless otherwise specified on order, amplified units with 4-20mA output will provide 4mA at
0 psid and 20mA at positive full scale and the unit will not operate in the negative direction. An
available alternative is to specify 4mA at negative full scale and 20mA at positive full scale.
Intrinsically safe amp (Option 2n or 2N, see page AP-6) adds 2" to amplifier housing.

General Information
How to order (See Pg. AP-19)
Differential pressure selection flow chart (See Pg. DP-1)●お問い合わせは下記へ ̶DP-7