FN1010
Load Pin
SPECIFICATIONS Range from 10 kN to 2000 kN (2 klbf to 400 klbf)
 Standard and custom load in designs
 Tension and compression uses Other Ranges on Request

 Bidirectional versions available
TE CONNECTIVITY’s load pins, model FN1010, are designed tofit in the place of the regular mounting unit.

FEATURESAccuracy 1% FS
High Coefficient of Security
Watertight optional
Level Output Model with Integrated
Amplifier

APPLICATIONS

Crane Monitoring
Building Machine Monitoring
Load-Limited DeviceOffshore

The implantation is facilitated by the possibility ofmodifying a certain number of dimensions. The FN1010 issuitable for numerous applications on lifting motors and handlingequipment. The load pins can be used to measure forces on
rotating components (pulleys, sheaves, etc.) and can be directly
mounted on shackles.
The sensing element is fitted with thin film strain gages
in a Wheatstone bridge circuit. All FN1010 Load Pins incorporatea keyed anti-rotation slot. Optionally, the load pins may be madewatertight for certain applications while resting insensitive tohydrostatic pressure effects. Additionally, the FN1010 isavailable with an integrated high-level analogue output.
With a long standing experience as a designer andmanufacturer of sensors, TE CONNECTIVITY often works withcustomers to design or customize sensors for specific uses andtesting environments.
On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpful
tips.

SENSOR SOLUTIONS /// FN1010_en06/2017

Page 1

FN10110
Load Pin

STANDARD RANGES
Ranges in N (FS)

10k

20k

50k

100k

200k

500k

1 000k2 000kRanges in lbf

2k

4k

10k

20k

40k

100k

200k

400k

PERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)
PARAMETERS
Operating Temperature Range (OTR)-20 to 80° C [-4 to 176° F]

Compensated Temperature Range (CTR)

0 to 60° C [32 to 140° F]

Thermal Zero Shift in CTR<0.5% F.S. / 50° C [/100° F]

Thermal Sensitivity Shift in CTR<1% of reading / 50° C [/100° F]

Over-Range
Without Damage

1.5 x F.S.

Without Destruction

5 x F.S.

Accuracy±1% F.S.

Combined Non-Linearity & Hysteresis
Electrical Characteristics
Model

FN1010 1

FN1010-A1

FN1010-A2

FN1010-A3 (2 wires)

Supply Voltage

1 to 10 Vdc

10 to 30Vdc±15Vdc (±12 to ±18Vdc)

12-36Vdc

Sensitivity “FSO” 4±1.5mV/V±2V ±0.2V±5V ±0.2V

16 (or ±8) ±0.4mA 5

Zero Offset 4<±1mV

2.5V ±0.2V

0V ±0.2V

4 (or 12) ±0.4 mA 5

Input Impedance/Consumption

350 to 700Ω

<50mA

<50mA-

Output Impedance

350 to 700Ω

1 kΩ 61 kΩ 6-Insulation under 50Vdc≥100MΩ≥100MΩ≥100MΩ≥100MΩ

Notes
1. Sensors are calibrated with 10Vdc power supply as standard.
2. Electrical Termination: Shielded cable; standard length 2m [6.5ft]
3. Materials: Body in stainless steel
4. Other output signal on request
5. Signal is 4-20mA. Depending of application calibration it could beTensiont & compression: 12mA at 0N and ±8mA for ±FS
Compression or tension only: 4mA at 0N and 20mA at FS
6. Output impedance < 100 Ω on request
7. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN1010_en

06/2017

Page 2

FN10110
Load Pin

DIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)
Wiring Schematic

All dimensions correspond to a standard. They can be modified, if
necessary, for mounting. Please consult us for details.
In order to simplify the use of load pins and limit the mechanicalmodifications associated with their implantation, all dimensions aregiven between two limits within which performances andcharacteristics can be maintained without increasing financial cost tothe user.
Note: The delimiter groove can be placed on the output cable side.

Dimensions in mm [inch]
Ranges in N
In [lbf]
AB

C

10k
[2k]

20k
[4k]

50k
[10k]

100k
[20k]

200k
[40k]

500k
[100k]

1000k[200k]

2000k[400k]

Min.
Nominal
Max
Min.

22

[0.87]

27[1.06]

30[1.18]

42[1.65]

54[2.13]

82[3.23]

110[4.33]

150[5.91]

25
30[0.98][1.18]

30
35[1.18][1.38]

35
40[1.38][1.57]

45
50[1.77][1.97]

60
65[2.36][2.56]

90100[3.54][3.94]

120
130[4.72][5.12]

160
170[6.30][6.69]

25[0.98]

25[0.98]

30[1.18]

40[1.57]

50[1.97]

65[2.56]

80[3.15]

120[4.72]

NominalMax.

30[1.18]

30[1.18]

40[1.57]

50[1.97]

70[2.76]

90[3.54]

110[4.33]

160[6.30]

35[1.38]

35[1.38]

50[1.97]

65[2.56]

90[3.54]

115[4.53]

140[5.51]

200[7.87]

Min.
Nominal10[0.39]

10[0.39]

15[0.59]

20[0.79]

25[0.98]

30[1.18]

40[1.57]

60[2.36]

15[0.59]

15[0.59]

20[0.79]

25

[0.98

30

[1.18]

40[1.57]

55[2.17]80[3.15]

Max.

20[0.79]

20[0.79]

25[0.98]

30[1.18]

35[1.38]

50[1.97]

70[2.76]

100[3.94]

DD

3[0.12]

3[0.12]

4[0.16]

5[0.20]5[0.20]5[0.20]5[0.20]5[0.20]EE

5[0.20]5[0.20]5[0.20]

10[0.39]

10[0.39]

10[0.39]

15[0.59]

15[0.59]SENSOR SOLUTIONS /// FN1010_en06/2017

Page 3

FN10110
Load PinOPTIONS
A1 : Amplified Tension output with unipolar power supply
A2 : Amplified Tension output with bipolar power supply
A3 : Current output (2 wires)
V00 : Non-standard power supply calibration, replace “00” with value in Volt (standard 10Vdc)
L00M : special cable length, replace “00” with total length in meters

ORDERING INFORMATIONFN1010 - A1 - A00/B00/C00 - 100KN - /SC
Options (L00M, ..)Range in Newton
Dimensions (see drawings)
A XX number will be
provided with order
Amplified version (none,A1 or A2)
ModelTE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s StandardTerms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN1010_en

All Rights Reserved.

06/2017

Page 4