DAZZLERTM products are turn-key ultrafast pulse shaping systems
performing simultaneous and independent spectral phase andamplitude programming of ultrafast laser pulses.

DA ZZLERU LT R A FA S T P U L S E S H A P E R

With over 400 systems installed all over the world, the DAZZLERTM
systems constitute the premier choice for your pulse shapingapplications.

Data Sheet

Ultra-compact
deviceSingle calibration
point
User-friendly

Quantitativeshaping
No pixels:
No ghost pulses

Key benefits

Advanced software
functionalities

In-line beamgeometry

Simple opticalsetup

All DazzlerTM systems include:- A crystal module (50x100x20mm3) enabling to control, through an acousto-opticinteraction, optical signals in the hundreds of THz range with acoustic signals in
the tens of MHz range.
- A RF generator which generates user-defined RF waveforms and communicateswith the control PC via a USB link.
- A laptop computer with pre-installed user-friendly software, allowing easy
programming of RF waveforms to perform specific optical shaping.

Available options:- Wizzler feedback loop for high-dynamic pulse compression optimization.
- Streaming of over 100 000 pre-defined waveforms, in order to change the laser
pulse characteristics at every trigger event at repetition rates up to 1kHz.
- CEP analog modulation to control and stabilize the Carrier Enveloppe Phase withsub π/500 resolution at repetition rates up to 30kHz.
- Jitter-free electronics for sub 100 attoseconds optical jitter.
- 20W and 50W RF amplifiers for maximum diffraction efficiency.
- T4 fast communication protocol (1 waveform loaded in less than 3ms)

Phone:
Fax:
E-mail:+33 (0)4 8813 1751
+33 (0)4 9295 7690
info@fastlite.com

www.fastlite.com

FASTLITE

Bât D1 - Les collines de Sophia
1900 route des crêtes
06560 VALBONNEFRANCES TA N D A R D D A Z Z L E R P R O D U C T S
DA Z ZLER s ys tems for V IS and IR
3 crystal cuts for laser pulses between 470 and 3500nm- HR for optimum shaping capability- WB for optimum diffraction efficiency- WR intermediate cut (for visible laser pulses)

Spectrum Chart

Pulse shaping capability of VIS & IR Dazzlers
Dazzler systems for CPA and FCPA

Shaping capability ps16

Dazzler systems
yafter OPA and NOPA

1412WB25
WB45
HR25
HR45WR25108640.5

Dazzler systems
yfor OPCPA

1.0

1.5

2.0

2.5

3.0

3.5

Wavelength μm
Spectral resolution of VIS & IR Dazzlers

Custom wavelength ranges available on request
10.0

Input beam requirements for VIS & IR Dazzlers

UV-250-400

EnergyUV-255-280-HE

Wavelength tuning range

250 to 400nm

255 to 280nm

Instantaneous bandwidth

up to 150nm

up to 25nm

Mid - IR DA Z ZLER

Spatial properties

Max Φ=2,5mm,collimated

5.0Resolution nm

U V DA Z ZLER sMax 30μJfor <1ps pulses
on max beam diameter

0,1nm

Maximum programmable 4,5ps at 250nm
delay
3,5ps at 400nm

4ps

20% at 250nm
10% at 400nm>40%Max 10μJ

Max 100μJ

Diffraction efficiency*
Input beam requirements

Spectral resolution

Diffraction efficiency

15% on 100cm-1
Max 20μJ

WB25
WB45
HR25
HR45
WR25

0.2

1.0

1.5

2.0

2.5

3.0

3.5

Wavelength μ m

Diffraction efficiency of VIS & IR Dazzlers
HR25

Maximum programmable
4ps at 5000nm
delay

Input beam requirements* for 50fs pulses

7cm-1 at 5000nm

0.50.5

Wavelength tuning range 3500 to 7000nm
Spectral resolution

1.00.1UWB 3500-7000
0,1nm at 250nm
0,2nm at 400nm

2.0WR25

WB25

HR45

WB45

Standard50% on 50nm 60% on 50nm 50% on 100nm
25% on 100nm 30% on 100nm 25% on 200nm

50% on 100nm
25% on 200nm

50% on 200nm
25% on 400nm

With 20W option
With 50W option

40% on 100nm 40% on 100nm 35% on 200nm
50% on 200nm 50% on 200nm 40% on 400nm

40% on 200nm
50% on 400nm

40% on 400nm
50% on 800nm