FN7110
Double Range Load Cell
SPECIFICATIONS Double S-Beam Load Cell

 10/100 N to 1k/10k N (2/20 lbf to 200/2k lbf)
 High resolution

 For compression and traction uses

The FN7110 features high accuracy measurement channels intwo ranges in the same load cell.

FEATURESDouble Full Scale Range
Accuracy : 0.1% F.S. for each range
Integrated Mechanical Stops
Integrated Amplifier optional

APPLICATIONS

Process Control Equipment
Robotics and Effectors
Product Validation Testing
Laboratory and Research

SENSOR SOLUTIONS /// FN7110_en

The standard ratio between the ranges is 1 to 10, andstandard load range combinations go from 0-10 and 0-100 N
through 0-1000 and 0-10,000 N. The FN7110's percentage
accuracy is the same over each range in the load cell. In practice
one maintains accuracy of 0.1% over the high force range evenin the first 10% of the range.
During measurement, once the lower range sensing
element reaches its designed full scale limit, mechanical stops
protect it against overloads, in tension and compression, up to12 times the full scale limit of the higher range sensing element(which is equivalent to 12 times the full scale limit of the lowerrange). For high-level output, a model with an integrated
amplifier is available.
On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpful
tips.

06/2017

Page 1

FN7110
Double Range Load Cell

STANDARD RANGES
Ranges in N (FS)

10/10050/500100/1k

200/2k

500/5k

1k/10kRanges in lbf

2/20

10/10020/20040/400100/1k

200/2kPERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)PARAMETERS
Operating Temperature Range [OTR]-20 to 80° C [-4 to 176° F]

Compensated Temperature Range [CTR]

0 to 60° C [32 to 140° F]

Thermal Zero Shift in CTR<0.5% F.S. / 50º C [/100° F]

Thermal Sensitivity Shift in CTR<1% of reading / 50º C [/100° F]

Over-Range
1.2 x F.S. of the higher range

Without Damage
Accuracy±0.1% F.S. of each range

Linearity
Electrical Characteristics (for each channel)
Model

FN7110 1

FN7110-A1

FN7110-A2

Supply Voltage

2 to 12 Vdc

10 to 30Vdc±15Vdc (±12 to 18Vdc)

Sensitivity “FSO” 2±2mV/V±2V ±0.2V±5V ±0.2V

Zero Offset 2<±1mV

2.5V ±0.2V

0V ±0.2V

Input Impedance/Consumption

150 to 200Ω

<50mA

<50mA

Output Impedance

300 to 400Ω

1 kΩ 51 kΩ 5Insulation under 50Vdc≥100MΩ≥100MΩ≥100MΩ

Notes
1. Sensors are calibrated with 10Vdc power supply as standard.
2. Signal goes positive in tension with standard wiring configuration. Other signal output on request
3. Cable Termination: Connector output including mate, prewired, standard length 2 m
4. Material: Body in aluminum alloy or stainless steel depending on F.S.
5. Output impedance < 100Ω on request
6. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN7110_en

06/2017

Page 2

FN7110
Double Range Load CellDIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)

F.S. Ranges in N
[in lbf]F (Thread)

10/100[2/20]

50/500[10/100]

100/1k[20-200]M6

500/5k[100/1k]

1k/10k
[200/2k]

M10Material

SENSOR SOLUTIONS /// FN7110_en200/2k[40/400]

Aluminium

06/2017

Stainless Steel

Page 3

FN7110
Double Range Load CellOPTIONS
A1 : Amplified Tension output with unipolar power supply
A2 : Amplified Tension output with bipolar power supply
V00: Non-standard power supply calibration, replace “00” with value in Volt (standard 10Vdc, unamplified sensor only)

ORDERING INFO

FN7110 - A1 - 10/100N - /V04
Options (V05,...)
Range in Newton
Amplified version (none, A1 or A2)
Model

SUPPLIED ACCESSOIRESEFMX-7M : mating plug Jaeger 530-272-006 with clamp 530-371-006 standard

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s StandardTerms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN7110_en

All Rights Reserved.

06/2017

Page 4