ARF/XRF Agile RF Synthesizer + AOM Driver

The MOGLabs ARF/XRF agile RF synthesizer
provides two channels of agile RF frequency
synthesis with high-power output drivers.
Each channel spans a frequency range of 20
to 400 MHz with output power up to +36dBm(4 W). The two channels can be controlled via
front panel knobs to adjust frequency andpower, or via computer interface. Twoexternal analogue inputs are provided foreach channel, to allow FM/AM/PM at up to10 MHz bandwidth, and PID servo feedback isbuilt-in for laser noise-eating or frequency
locking.
The computer interface (10/100 Ethernet andUSB) allows full control of all parameters,advanced table sequence control, andmonitoring. Ultrafast digital outputs can be
individually controlled in synchronisation withthe table sequences.

Features Two RF channels, independent or synchronised
 High output power: up to +36dBm per channel
 Wide frequency range: 20-400MHz
 High modulation bandwidth up to 10 MHz (AM, FM, φ) RF power output monitoring and protection
 External digital inputs for fast on/off, trigger
 16 high-speed digital IO (table sequence control) Autonomous execution of complicatedfrequency/power/phase sequences External sync clock input
 Three analogue outputs
 Robust open- and short-circuit protection
 Ethernet and USB interfaces

Applications
 AOM driver Noise eater or laser frequency lock
 Diamond NV quantum control
 Laser cooling, trapping, spectroscopy Bose-Einstein condensation
 Quantum optics: squeezed light Electromagnetic transparency, slow light Time and frequency standards

Agile Frequency Synthesizer/AOM Driver
Specifications ARF/XRF
RF characteristics
RF output power

ARF421, XRF421: 0 to +36 dBm
ARF021, XRF021: 0 to +16 dBm

Frequency

20 to 400 MHz, 32-bit resolution (0.23Hz steps)

Frequency stability±1 ppm (0 to 50°C)

Phase

0 to 360°, 16-bit resolution

Absolute phase noise‒ 115dBc/Hz @ 10kHz, ‒ 113dBc/Hz @ 1kHz, ‒ 105dBc/Hz @ 100Hz

Signal to noise> 80dBc @ 30dBm

Intermodulation and spurious< ‒ 80dBc

Crosstalk between channels< ‒ 70dBc (off), < ‒ 50dBc (on)RF ‘off’ level< ‒ 70dBm

External clock

5 MHz to 1 GHz

14-bit resolutionAnalogue input/output
Number

2 inputs and 2 outputs per RF channel

Function

Modulation bandwidth

FM, AM, φ or analogue sampling for DSP applications± 1V, 7th order anti-alias filter
12-bit resolution, 65MHz sampling rate
10 MHz first parameter, 1 MHz second parameter

DAC

3 channels, ± 2.5 V, 14-bit, 1MHz bandwidth

Digital input/output
Temperature
RF on/off

Software control, front-panel buttons, hardwired TTL

Trigger input

Per channel, start/retrigger by edge or levelShutter output

Per channel TTL output

High speed digital IO

16 TTL input/output, user-controllable and via table mode (advanced)

Sensitivity

Computer interface
Ethernet

10/100 TP, RJ45

USB

USB2, plug type USB-A

Table mode

Up to 8k programming points per channel

Table timing resolution

ARF: 1 µs, XRF: 16 ns

Dimensions and power
Dimensions

250 x 79 x 292mm (WxHxD), 2kg

Power

95 – 264 Vac, 47 to 63Hz, 1A

株式会社 ルクスレイ

URL: http://www.LxRay.jp E-mail: info@LxRay.jp

関西本社 〒663-8113 兵庫県西宮市甲子園口3-28-22
東京事務所 〒113-0033 東京都文京区本郷2-11-6 第1谷口ビル1F

MOG Laboratories Pty Ltd
Stes 34-35, 49 University StCarlton VIC 3053
Australia
Tel: +61 3 9939 0677
info@moglabs.com

MOGLabs USA
419 14th St
Huntingdon, PA 16652
USA
Tel: +1 814 251 4363
info@moglabsusa.com

MOGLabs EuropeGoethepark 910627 Berlin
Germany
Tel: +49 30 21 960 959
christoph.p@moglabs.com

TEL: 0798-31-0500, FAX: 0798-31-0505
TEL: 03-3868-0200, FAX: 03-6912-6394© 2019 MOG Laboratories Pty Ltd
Product specifications and descriptionsin this document are subject to change
without notice