F650-CK

IN-MOTION CHECK WEIGHING INDICATOR
Specifications
AnalogExcitation voltage:
Signal input range:
Zero adjustment range:
Gain adjustment range:
Accuracy

A/D converter

DC 5 V ±5% Output current: Within 90 mA Ratiometric method(Up to 6 350 Ω load cells can be connected in parallel.)
㸫0.3 to +3.0 mV/V
㸫0.2 to +3.0 mV/V Automatic adjustment by digital computation
Automatic adjustment by digital computation
Non-linearity: Within 0.01% FS±1 digit (when 3.0 mV/V is input)
Zero drift: 0.025 ȣV/ΥRTI typ.
Gain drift: 1 ppm/Υ typ.

Theoretical calibration

Conversion rate: 1000 times/sec. Resolution: 24 bit (binary)
Equivalent input calibration:accuracy when theoretical calibration is performed: 1/1000Filter

Digital filter
Low-pass filter

Moving average (common for all modes): OFF, 2 to 999 times
Variable: 2.0 to 10.0 Hz

Display

Display unit

Total
function

External
signalTFT color LCD module 3.5 inch (320 × 240 dot)
Display area: 71(W) × 53(H) mm
Weight display
5-digits (signs: minus sign on the highest numerical digit)
Unit
NONE, kg, t, g, N, lb
Decimal place0, 0.0, 0.00, 0.000
Status display
BUSY, GO, EMP, SMP, COMP, NZ, STAB, RANK1 to 11
Histogram display: Numbers of data for each 9 weight ranges are displayed. Two set of data outof the range are displayed as well.
Weighing results: Results are displayed for each code.
Statistics: Display statistics data stored on F650-CK.Display average weight, maximum weight,minimum weight, number of data, population standard deviation, sample standard deviation,difference between maximum and minimum, latest data
External output (10 points): Transistor open collector output. (Emitter = COM terminal)
The output turns to LOW when the transistor turns ON.

OVER or RANK1 or RANK20/ GO or RANK2 or RANK21/ UNDER or RANK3 or RANK2 2 /NONE or RANK4 or RANK23/ NONE or RANK5 or STROBE/ OUTPUT SEL. 0/ OUTPUT SEL. 1/OUTPUT SEL. 2/ OUTPUT SEL. 3/ OUTPUT SEL. 4External input (10 points): ON when shorted with COM terminals by contact (relay, switch, etc.)
or non-contact (transistor, TTL open-collector output, etc.)‫ ە‬With the high-performance filter to attenuate noise and vibration,stable and accurate weight measurement can be achieved.
‫ ە‬Check if the weight is over or under the target weight and/or
sort products into preset grades or classes (sorting into 11
grades or classes at the most).
‫ ە‬Control signal is sent out for controlling conveyors.In-motion
check weigher can be designed with only F650-CK.
‫ ە‬Useful weighing modes to improve efficiency and accuracy.
Auto sorting mode is useful when conveyor speed, size, andweight are not constant.
Reduction of processing capacity can be prevented as zeroadjustment can be performed during operation.
‫ ە‬Detect the situation where two cartons are on the scale.

Even if two cartons are on the scale,weight of each individual
carton will be weighed and judged (OK/NG).

Load cells

Weight signal‫ ڦ‬Waveform display

Waveform can be constantly monitored.

CODE0/ CODE1/ CODE2 or KEY LOCK/ Graph drawing/ D/Z ON/ TARE ON/ TARE OFF/
Accumulation command/ Measurement start/ Measurement reset

Interface

Option

SIF: 2-wire serial interface
BSC: BCD parallel data output interface (Source type) (option)232: RS-232C communication interface DAV: D/A converter (voltage output) (option)
CCL: CC-Link interface (option)
DAI: D/A converter (current output) (option)
ODN: DeviceNet interface (option)
485: RS-485 interface (option)
BCO: BCD parallel data output interface (Sink type) (option)* Only one option can be installed.
ISC: I/O Source Board

GeneralOperating voltage࣭Power consumption: AC 100 to 240 V (㸩10㸫15%) (50/60 Hz) 4 W typ.
performance Operating conditions: Operation temperature: 㸫10 to 㸩40Υ Storage temperature: 㸫20 to 㸩60Υ
Humidity: 80% RH or less (non-condensing)
Dimension࣭Weight: 96(W) × 96(H) × 138(D) mm (Not including projections) Approx. 1.0 kg
Attachments AC power cable (voltage resistance: 125 V) (3 m)×1, Jumper wire×2, Operation manual×1,
FCN series I/O connector (with cover)×1, Analog I/O connector terminal block (Already mounted
on the main unit)×1, Mini driver (when D/A converter option is selected) ×1,
BCD output connector (when BCO option is selected)×1,DeviceNet connector (when ODN option is selected)×1,
CC-Link connector (when CCL option is selected)×1
Optional
CA372-I/O:
Cable with FCN connector at one-end 3 maccessories CA600-BCDCNV:FCN connector 32p-57࣭36p cabtyre cable 0.3 m
CA81-232X:
miniDIN-D-Sub9p cross cable 1.5 m
CAAC2P-B3:
AC input cord 3 m (Same as the attachment)
CAAC3P-B3:
AC input cord 3 m
CA325AC3P-B3:
AC input cord 3 m
CAAC3P-CEE7/7-B2:
AC power cable (Voltage resistance: 250 V) 2 m
CAA325AC3P-CEE7/7-B2: AC input cord (Voltage resistance: 250 V) 2 m
CN50:
FCN series I/O connector (with cover)(Same as the attachment)
CN51:
BCD output connector
CN55:
FCN series I/O connector (with diagonal cover)
CN60:
Circular DIN 8p connector for RS-232C
CN71:
CC-Link connector
CN72:
Double row connector for CC-Link
CN80:
Analog I/O connector terminal block (Same as the attachment)* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,
fluorescent display tubes and LCDs due to manufacturing process or production lots.

‫ ڦ‬Real-time statistic
In order to process statistics in real time,you can constantly check and monitor thevariation and distribution of data.

Structure of product code

F650-CKձ‫ڧ‬‫ڧ‬ղճճ Interface
SignStandardInterface
SI/F,RS-232C

CCL
ODN
BCO
BSC
DAV
DAI
485

CC-LinkDeviceNet
BCD output (Sink type)BCD output (Source type)
D/A converter (Voltage output)
D/A converter (Current output)
RS-485 (Modbus-RTU /UNI-format)↓ One optional interface can be added
in addition to the standard interface.

ձ Standard unit
ղ I/O output
Display waveform before and after thefilter setting is changed.

Histogram display

SignStandard
ISC

Output type
Sink type (NPN output)
Source type (PNP output)

External dimension(Front)(Side)(Rear)

Panel cutout size
Panel thickness
1.6 to 3.2 mm

Unit: mm

unlPULXE

www.unipulse.tokyo/en/