FC500-CCL
FC500-DAC
FC500-EIP
FC500-485
FC500-232

DIN RAIL MOUNT WEIGHING INDICATOR
ATOR

Suitable for many weighing applications
li tisuchh ashopper scale, packing scale, weight level meter, etc….

Lightweight & compact bodyfor 35 mm DIN rail mount.

FC500-CCL A weighing system can be configured easily using CC-Link network.
FC500-DAC Current output, corresponding to the indicated value, is available.
FC500-EIP
FC400-485
FC500-232

A weighing system can be configured easily using EtherNet/IP network.
Measured data and results can be collected using RS-485 interface.
Measured data and results can be collected using RS-232C interface.

High sampling rate & resolution

Application software for USB interface

High-Speed A/D conversion and powerful digitalprocessing capability of 1200 times/sec.
(Convertible to 300 times/sec.)
High display resolution of 1/100000 (max).

With communication through USB
interface, logging, graphdisplay, setting parameters, andcalibration can be done.

Measurement can be performed quickly andprecisely due to high speed A/D conversion.

Various interfaces
FC500-CCL Equipped with CC-Link, SI/F, USB interface
FC500-DAC Equipped with D/A converter, SI/F, USB interface
FC500-EIP Equipped with EtherNet/IP, SI/F, USB interface
FC500-485 Equipped with RS-485, SI/F, USB interface
FC500-232 Equipped with RS-232C, SI/F, USB interfaceex) CC-Link
EtherNet/IP
RS-485
RS-232C
D/A converter
SI/F

Sensorinput
SI/Fex) CC-Link
EtherNet/IP
RS-485

Display etc.

D/A
converter

RecorderRS-232C
CC-Link
EtherNet/IP
RS-485

Load cell
input

CC-Link
EtherNet/IP
RS-485PLCex) USB

FC500

USB
PLC

unlPULXE

FC500

FC500

3&www.unipulse.tokyo/en/

Input conversion value display

Latest 100 data of calibration value and error information withclock time are recorded and can be checked via USB interface.

The output signal level of load cell can be displayed in mV/V for
monitor purpose.
Malfunction indicator or faulty sensor can be differentiated easily.

Basic weighing process control function

Multipoint calibration (linearization)

Equipped with weighing sequence function to control
feeding/discharge gate.

FC500

Data memory function

Three additional points can be defined in the middle between
zero and span for better linearity. Eventhough the scale has poorlinearity, it can be corrected to be a highly accurate scale.

Memory for 32 weight settings
32 different weight settings can be saved in the memory and selected
through I/O or interface. Batch weighing can be performed easily.

Equivalent input calibration
Theoretical calibration can be performed easily by registering thecapacity and rated output of load cells.

Specifications
Analog

Display

External
signalInterface

Excitation voltage

DC 5 V±5% Output current: Within 90 mA
Ratiometric method(Up to 6 350 Ω load cells can be connected in parallel)-2.5 to +5.1 mV/V
Signal input range
Zero adjustment range Automatic adjustment by digital processing -0.5 to +2.0 mV/V
Span adjustment range Automatic adjustment by digital processing 0.005 to 3.2 mV/V
Linearization function Up to three point multi-point calibration is possibleusing linearization function
Min. input sensitivity
0.15 ȣV/countAccuracyNon-linearity: Within 0.01% FS
Zero drift:
0.0002% FS/Υ typ. * When calibrated by 3 mV/VGain drift:
1 ppm/Υ typ.
FilterDigital low-pass filter 0.1 to 300 Hz
Moving average filter OFF, 2 to 512 times
A/D converter
Speed:1200 times/sec. (Convertible to 300 times/sec.)
Resolution: 24 bit (binary)
Display unit
Character height 11 mm
Numerical display by liquid crystal display module
Display value
Up to 6 digits. Sign‫؟‬Minus display on the highest digit
Unit selectablet, g, kg, lb, N, None
Display frequency
Selectable from 1, 3, 6, 13, and 25 times/sec.
COMPL./SP1/SP2/SP3/HI/GO/LO/NZ/TARE/NET/HOLD/ZALM/STAB
Status display
RUN/SD/RD/ERR‫ق‬FC500-CCL‫ك‬MS/NS‫ق‬FC500-EIP‫ك‬
Selectable/ configurable
Input signal<No-voltage contact input>(5)
Input is ON when shorted to COM terminal by contact (relay,switch, etc.) or non contact (transistor, photocoupler, etc.).
<Voltage input plus common/ minus common shared>(specifies
at time of order)
Input is ON when a voltage is applied in between to input
terminal and COM terminal by contact (relay, switch, etc.) or noncontact (transistor, photocoupler, etc.).
Rated voltage: DC 27.6 V or less"ON": when the voltage is above DC 9 V (Load Current: approx.
10 mA at DC 24 V), "OFF: " below DC 3 V.
Selectable/ configurable
Output signal
PhotoMOS relay output (common for sink and source type)(5)
Vceo = 30 V, Ic = 50 mA
CC-Link interface (FC500-CCL)
D/A converter (Current output) (FC500-DAC)
EtherNet/IP interface (FC500-EIP)
RS-485 interface (Selectable from Modbus-RTU or UNI-Format) (FC500-485)
RS-232C interface (Selectable from Modbus-RTU or UNI-Format) (FC500-232)
SI/F 2-wire type serial interface
USB interface

GeneralPower supply voltage
specification Power consumption

DC 24 V (±15%)
6 W typ. (FC500-DAC, FC500-EIP)
5 W typ. (FC500-CCL)
4 W typ. (FC500-485, FC500-232)
Operating conditions Operation temperature: -10 to +50Υ
Storage temperature: -20 to +85Υ
Humidity: 85% RH or less (non-condensing)65(W) × 94(H) × 108(D) mm (Not including projections)
Dimension
Approx. 370g
WeightMini driver ....................................1
Attachments Operation manual ............................2.....................................2
Jumper wire
I/O connector ...............................1
Power connector ..............................1
Interface connector (FC500-CCL, FC500-DAC, FC500-485).......................1
Optional
CN74:
CC-Link connector (Same as the attachment)
accessories CN75:
CC-Link connector (Y type branch connector)
CN76:
CC-Link connector (Terminator connector)
CN86:
3 p connector D/A converter (Same as the attachment)
CN7B‫؟‬Power connector(Same as the attachment)
CN7D‫؟‬I/O connector(Same as the attachment)
CN7C‫؟‬RS-485connecter (Same as the attachment)
CE marking EMC directive EN61326-1
certification* Please note that there are possibilities of individual differences in a color tone on display devices suchas LEDs, fluorescent display tubes and LCDs due to manufacturing process or production lots.

Structure of product code

⋇Standard unit

FC500 ‫ٸ‬⋇⋈‫ٸ‬⋉⋈Interface
ModelCCLDAC
EIP485232

Interface
CC-Link, SI/F, USB
D/A converter (Current output), SI/F, USB
EtherNet/IP, SI/F, USB
RS-485 (Selectable from Modbus-RTU or UNI-Format), SI/F, USB
RS-232C (Selectable from Modbus-RTU or UNI-Format), SI/F, USB⋉Input signal
Sign
Input type
Standard No-voltage contact input
DCI
Voltage input

External dimension■ FC500-EIP10812165

108122

unlPULXE65■ FC500-232

9499■ FC500-485

1081216594991081219499

9499■ FC500-DAC

9499■ FC500-CCL

6510812165

Unit: mmwww.unipulse.tokyo/en/