ELHSMiniature Load Cell
SPECIFICATIONS Ranges from 1kN to 50kN (200 lbf to 10,000 lbf)

 Semi-conductor strain gages bridge

Low deflection, low mass

The ELHS load cell is a compact package able to fit into many
applications where others cannot.
The low noise Wheatstone bridge consists ofpiezoresistive strain gages which provide 200 mV of typical fullscale output with a compact design, an extreme stiffness andsuperior resolution. The ELHS is provided with either SAE ormetric threads for tension and compression applications.

FEATURES

Care should be exercised to isolate your ELHS from offaxis loads. The ELHS is a Poisson column design providing low
full scale input deflection and superior resolution. Designed
specifically to provide high resolution and low deflection.

 Low mass compression style ortension compression with studs
 Sensitivity “FSO”: 20mV/V
 High stiffness
 High reliability

The ELHS can be configured with a variety of options tofine tune the instrument to your application: select from several
standard compensated temperature ranges, input voltages, leadlengths or specify entirely unique combinations of these options.

APPLICATIONS Assembly forces Applications requiring ultra fastresponse Tool forces Thrust measurements
 Product validation testing

SENSOR SOLUTIONS /// ELHS_en

On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpful
tips.

06/2017

Page 1ELHS
Load Cell

STANDARD RANGES
RangesSensitivity "FSO" 1

Body style
Overrange

Compression

Tension & Compression(typical at 23±3°C)

1.5 x FS

ELHS-B1

ELHS-T1

20 mV/V500

1.5 x FS

ELHS-B1

ELHS-T1

20 mV/V

5K

1K

1.5 x FS

ELHS-B1ELHS-T120 mV/V

10K

2K

1.5 x FS

ELHS-B4

ELHS-T420 mV/V

25K

5K

1.5 x FS

ELHS-B4

ELHS-T420 mV/V

50K

10K

1.5 x FS

ELHS-B6

ELHS-T620 mV/V

N

lbf

1K

200

2.5K

1. Sensors are calibrated with 15Vdc power supply as standard.
2. For B1, B4 and B6 version, signal goes positive in compression with standard wiring configuration
For T1, T4 and T6 version, signal goes positive in tension with standard wiring configuration

PERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)PARAMETERSVALUES

Supply voltage

1 to 15Vdc

Input resistance

1500Ω nom.

Output resistance500Ω nom.

Repeatability± 0 . 2 5 % FS

Combined non-linearity & hysteresis±0.5%FS

Thermal Zero Shift "TZS" in CTR±0.02%FS /°C

Thermal Sensitivity Shift "TSS" in CTR±0.05% /°C

Operating temperature Range (OTR)-20°C to 80°C

Compensated Temperature Range (CTR)

20°C to 80°C

Zero offset± 10mVNOTES

See option table for other temperatures

Note:
1. Electrical Termination: 3m shielded cable output as standard
2. Type B units have positive output in compression. Type T units have positive output in tension.
3. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// ELHS_en06/2017

Page 2ELHS
Load CellDIMENSIONS
ELHS-B1, -B4 & -B6

ELHS-T1, -T4 & T6

ITEM

ELHS-B1
1kN to 5kN
200 to 1,000Lb

ELHS-B4
10kN to 25kN
2kLb to 5kLb

ELHS-B6
50kN
10kLb

A

3m (10 ft)

3m (10 ft)

3m (10 ft)

Bø 10 (0.39”)

ø 20 (0.78”)

ø 28 (1.1”)C3 (0.12”)

3 (0.12”)6 (0.24”)B

3m (10 ft)

3m (10 ft)

3m (10 ft)

Dø 3 (0.12”)

ø 3 (0.12”)

ø 5 (0.2”)Cø 3 (0.12”)

ø 3 (0.12”)

ø 5 (0.2”)

Eø 15 (0.59”)

ø 25 (0.98”)

ø 35 (1.38”)

D

3 (0.12”)3 (0.12”)6 (0.24”)F

15 (0.59”)

25 (0.98”)

35 (1.38”)

Eø 15 (0.59”)

ø 25 (0.98”)

ø 35 (1.38”)

G

1 (0.04“)

1 (0.04“)

1 (0.04“)F

15 (0.59”)

25 (0.98”)

30 (1.18”)G

15 (0.59”)

25 (0.98”)

30 (1.18”)

H

15 (0.59”)

25 (0.98”)

30 (1.18”)

INSTALLATIONSENSOR SOLUTIONS /// ELHS_en

ITEMA

ELHS-T1
1kN to 5kN
200 to 1,000Lb

ELHS-T4
10kN to 25kN
2kLb to 5kLb

ELHS-T6
50kN
10kLb-T1E : 3/8-24 UNF -T4E : 5/8-18 UNF -T6E : 3/4-16 UNF-T1M : M10x1.5-T4M : M16x2-T6M : M20x2.5

CONNECTIONS

06/2017

Page 3ELHS
Load CellOPTIONS AND ACCESSORIESOPTIONSCODES

DESCRIPTIONS

Compensated temperature ranges

Z0Z1Z2
Z*-40°C to 20°C-20°C to 40°C
0°C to 60°C
Non-standard, contact factory

Supply voltage

V00

Non-standard power supply calibration, replace “00” with value inVolt (standard 15Vdc)

Special cable length

L00F
L00M

Replace “00” with total length in feet with Lb range.
Replace “00” with total length in meters with N range.

Connector wired to cable

CRS

Microtech type male or equivalent (w/o mate)
RJ telephone type male (w/o mate)

ORDERING INFORMATION
Model-

BodyThread Type-

Range & Unit (1)-/Options

ELHS-

B1/T1
B4/T4B6/T6

M : Metric
E : SAE-1KN
2.5KN
5KN
10KN
25KN
50KN

-/Z0, Z1, Z2, or Z*/V1 thru V14/L00F or L00M
/C or RS

(1)Note :200L
500L
1KL
2KL
5KL
10KL

L=lbf ; N=Newton.
Metric threaded units must have range in Newton and cable length in meters specified.
SAE threaded units must have range in lbf and cable length in inches specified.
Options can be multiple if compatible (e.g. ELHS-B1-5KN-/Z1 /L3M/C)

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// ELHS_en

All Rights Reserved.

06/2017

Page 4