F701-C

BASIC TYPE ACCUMULATION VALUE DISPLAY
WEIGHING INDICATORAnalog‫ە‬The basic type of weighing indicator.
‫ە‬High speed A/D conversion and powerful digital processing
capabilities of 200 times/sec., for rapid response to input signal.
‫ە‬To serve its purpose, the operation has been made simple.ࠉ
‫ە‬High resolution of 1/10000 in all input range.
‫ە‬DIN-size 144 × 72 for easy installation

Display

Excitation voltage DC 10 V±5% Output current: Within 120 mA
Remote sense type (Up to 4 350 Ω load cells can be connected in parallel.)Zero adjustment
range

0 to 2 mV/V
Rough adjustment: Digitally controlled by rough adjustment circuit
Fine adjustment: Automatic adjustment by digital processing

Gain adjustment
range

0.3 to 3.2 mV/V
Rough adjustment: Digitally controlled by rough adjustment circuit
Fine adjustment: Automatic adjustment by digital processing

Minimum input
sensitivity

0.3 ȣV/countAccuracyNon-linearity: Within 0.01% FS
Zero drift: Within 0.2ȣV/°C RTI
Gain drift: Within 15 ppm/°C

Analog filter

Bessel type low-pass filter (-12 dB/oct.)
Selectable from 2, 4, 6, 8 Hz

A/D converter

Speed: 200 times/sec.
Resolution: 24 bit (binary)

Display unit

Numerical display of 10.4 mm character height via vacuum fluorescent display (7 digits)
Sub display: 4.0 mm Character height

Measured value 5 digits; Signs: Minus sign on most significant digit

External
signalUnit

Selectable from t, kg, g, N, Ib, none

Status display

SP1, SP2, SP3, near zero, Hold, zero alarm, stable, tare, net, zero,battery low indicator

External output
signal (12)Near zero/ SP1/ SP2/ SP3/ under/ over/ HI/ LO/ stable/ weight alarm or error/
GO or complete/ RUN or accumulation error
Transistor open collector output (Emitter = COM terminal)
Output is set to LO when transistor is ON
Vceo = 30 V (max), Ic = 120 mA (max)External input
signal (8)Gross or Net/ digital zero/ tare/ tare reset/ hold*/ judge*/ feed or discharge*/
start*/ stop*/ accumulation command*/ accumulation clear** Signals are selectable. Up to 4 points can be selected.
Set to ON when shorted to COM terminal through contact point (relay, switchetc.) or non-contact point (transistor, open collector output such as TTL etc.)
Ic = 10 mA or lower

18 digits sub display
Displays weighing data such as accumulation value aswell as various setting values.
Display mode can be selected from 6 types of modes.
InterfaceЋMain display
ЋSub display(Example) Accumulation times & accumulation value

SIF:
2-wire type serial interface
232:
RS-232C communication interface
CCL: CC-Link interface (Option)
BCO: BCD parallel data output interface (Option)
DAC: D/A converter interface (Option)
485:
RS-485 communication interface (Option)

Power supply voltage
Generalspecifications Power consumption‫ە‬Standard built-in of RS-232C.
‫ە‬Allows one option to be selected from either CC-Link,
BCD output, D/A converter or RS-485

Operating conditions*Only one option is avalable.

AC 100 to 240 V (+10%-15%) (Free power source at 50/60 Hz)
11 VA typ.
Operation temperature: -10 to +40°C
Storage temperature: -20 to +85°C
Humidity: 85% RH or less (non-condensing)

Exclusive PC application for setup

External dimension

144(W) × 72(H) × 160(D) mm (Not including projections)

WeightApprox. 1.0 kg

Attachments AC input cord (3 m) (nominal rating 125 V)… 1
Mini driver………………………………………... 1
FCN series I/O connector (with cover)……..... 1
Jumper wire………………………………........... 2
Operation manual………………………………. 1
Load cell input connector terminal block……. 1(Already mounted on the main unit)

With RS-232C interface thatallow you to read, write,
save and retrieve parametersettings with a PC.

CC-Link connector..................................... 1(When CC-Link option is selected)
BCD output connector............................... 1(when BCD output option is selected)
D/A converter connector........................... 1
㸦When D/A converter option is selected)
RS - 485 connector.................................... 1(when RS-485 option is selected)

Optional
AC input cord 3 m (Same as the attachment)
CA325AC3P-B3:
accessories CA325AC3P-CEE7/7B2: AC input cord (voltage resistance: 250 V) 2 m
CN34:
CN50:
CN55:
CN70:
CN71:
CN72:
CN80:
F720 fixing panel:
LB01-XH:

External dimension(Front)(Side)

D-sub 9 p connector for RS-232C
FCN series I/O connector (with cover)(Same as the attachment)
FCN series I/O connector (with diagonal cover)D/A converter connector
RS-485࣭CC-Link connector
Double row connector for CC-Link
Load cell input connector terminal block (Same as the attachment)
F252/F280 Fixing panel for conversion of F701-C
Lithium battery (with XH connector)

72* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,fluorescent display tubes and LCDs due to manufacturing process or production lots.

144

10150

Panel cutout size
Panel thickness 1.6 to 3.2 mm

137168-067(Rear)

2138 -0

Unit: mmunlPULXE

www.unipulse.tokyo/en/