High Power
3 µm DPSSL Modules. Compact monolithic laser systems. Highly efficient diode pumping. Fiber-coupled versions available. No high-voltage required. Reduced waste heat
. Maintenance free. Process variability

Specifications
DPM-2 (Er:YAG) free / fiberDPM-25 (Er:YAG) free / fiber

DPM-50 (Er:YAG) free / fiber

Wavelength2940 nm

2940 nm

2940 nmAverage Output Power (max)

2 / 1.2 W

25 / 16 W

50 / 33 W

Pulse Energy (max)

20* / 13* mJ

300* / 200* mJ

600* / 400* mJ

Pulse Repetition Rateup to 2 kHzup to 2 kHzup to 2 kHz

Pulse Duration

40 to 1000** µs

40 to 1000** µs

40 to 1000** µs

Average Current (max)

30 A

25 A

25 A

Mode of Operation

PulsedPulsedPulsedEfficiency (optical-optical)› 10 %› 10 %› 10 %

Beam Shape (focus)

top hat like

top hat like

top hat like

Free Beam Quality

M2 < 5M2 ‹ 25M2 ‹ 50

Free Beam Diameter

0.6 mm

1.6 mm

1.6 mmFree Divergence (half angle)‹ 25 mrad‹ 25 mrad‹ 50 mrad

Fiber Diameter GeO2

~ 230 µm (NA < 0.2)

~ 230 µm (NA < 0.2)

~ 420 µm (NA < 0.2)

Distilled water with AlgaecideDistilled water with AlgaecideDistilled water with Algaecideand Corrosion Inhibitor

and Corrosion Inhibitor

and Corrosion Inhibitor

Coolant Temperature

20 to 35 °C

20 to 25 °C

20 to 25 °C

Coolant Flow Rate

≥ 1 lpm› 5 lpm≥ 6 lpm

Coolant Pressure(1 - 3) bar(2 - 5) bar(3 - 5) bar

Required Cooling Power~ 150 W @ 25 °C Environment≥ 540 W @ 25 °C Environment≥ 780 W @ 25 °C Environment

Temperature

Temperature

Temperature

Diode Forward Voltage

2V~ 20 V~ 30 V

Diode Forward Current

350 A Pulsed

300 A Pulsed

300 A Pulsed

Average Power Consumption (max)‹ 120 W incl. 2 TECs

‹ 450 W

‹ 650 W

WxDxH

30 x 32 x 25 mm

120 x 96 x 75 mm

120 x 120 x 75 mm

Weight60 g

1.5 kg

1.7 kgEmission Height-47.5 mm

47.5 mm

Optical Parameters

Cooling Requirements
Coolant

Electrical Parameters

Mechanical Dimensions*with pulse durations > 600 µs
600 µs standard, 1000 µs on request**

www.pantec-medicallaser.com

Laser Diode Drivers
The LDD series are economic QCW laser diode driver modules designed to provide high current pulses to drive 3m.i.k.r.o.n.™
modules in various applications. It delivers output currents up to 300 A and pulse widths variable from 50 up to 1000* μs operation. Up to 1000 W average output power is available with the supplied heatsink and forced air flow. Several safety features areintegrated to protect both laser diode and driver.
DPM-2 (Er:YAG) / DPM-25 (Er:YAG)

DPM-50 (Er:YAG)

Laser Diode Driver

LDD-20300

LDD-30300

Output Currentup to 300 Aup to 300 A

Rise Time (10 - 90%)

‹ 20 µs‹ 20 µs

Mechanical Dimensions (W x D x H)

200 x 150 x 85 mm

200 x 150 x 85 mm

Additional Features

Safety circuit andSafety circuit andcommunication interface

communication interface

Test and Evaluate
The 3m.i.k.r.o.n.™ evalution kits are ready-to-use and straightforward laboratory systems for first feasibility studies in research environment. The evaluation kits are available with three different kind of laser
sources (see front page), shortens the development time, enables flexibility and a fast demonstration offeasibility. The test systems are delivered with your requested laser source, a laser control system and acooling system for laboratory use only.
Please contact us for more information on rental or purchase conditions: 3um@pantec.com

3m.i.k.r.o.n.™ Applications
MedicalIndustrial

Aesthetics / Dermatology

Material Processing (Drilling, Cutting, Melting, Welding, Evaporation)

Dentistry

Analytics

ENT

Security

Lithotripsy

DefenseMinimally-Invasive Surgery
Orthopedics
etc.

More Services

株式会社 ルクスレイManagemityealnt

Qu

Customized laser sources
Optical and mechanical design
Contract development and manufacturing
Medical device consulting (IP research, Medical CE, ...)

URL: http://www.LxRay.jp E-mail: info@LxRay.jp

関西本社 〒663-8113 兵庫県西宮市甲子園口3-28-22
東京事務所 〒113-0033 東京都文京区本郷2-11-6 第1谷口ビル1F

TEL: 0798-31-0500, FAX: 0798-31-0505
TEL: 03-3868-0200, FAX: 03-6912-6394

IS O13485

Pantec Biosolutions AG . Medical Laser
Industriering 21 . 9491 Ruggell . Liechtenstin
Tel: +423 377 13 33 . Fax: +423 377 13 34
3um@pantec.com . www.pantec-medicallaser.com

Illustrations, descriptions and technical data are not binding and may be changed without notice.
Copyright Pantec Biosolutions AG, Ruggell, Liechtenstein, 2019 – All rights reserved.