TC80-CCL
TC80-D3V
TC80-EIPDIN RAIL MOUNT TORQUE CONVERTERFOR UTMϪ/UTMϩ/UTMVSpecifications
Analogsection- Torque sensor input (voltage input)
Signal input range-10 to +10 V (UTM⋝) Input resistance 1 MΩ or more
-5 to +5 V (UTM⋜/UTMV) Input resistance 1 MΩ or more
AccuracyNon-linearity: Within 0.02% FS±1 digit
Zero drift: Within 0.2 mV/Υ RTI
Gain drift: Within 0.01%/Υ
A/D converter
Speed: 16000 times/sec.
Resolution: 24 bit (binary) 1/30000 with respect to 10 V
Digital low pass filterPASS, 3 Hz to 1 kHz (TC80-CCL)/ PASS, 1 Hz to 1 kHz (TC80-D3V, TC80-EIP)
Digital high pass filter PASS, 1 Hz to 1 kHz (TC80-D3V, TC80-EIP)- Rotation speed input (Pulse input, open collector) (UTM⋝/UTM⋜/UTMV)
Maximum input frequency
Conforming to the maximum rotation speed of UTM⋝/UTM⋜/UTMV
Minimum input frequency
Selectable from 15, 10, 5, 3, or 2 rpm (when pulse rate is 4 ppr)
Selectable from 60, 40, 20, 12, or 8 rpm (when pulse rate is 1 ppr)
Minimum detection pulse width50 ȣs
Circuit configuration
No-voltage contact input (minus common)
Open collector connectable (Ic = Approx. 10 mA)- Rotation speed input (Pulse input, open collector) * For using rotary encoder option
Maximum input frequency
Conforming to the maximum rotation speed of UTMϪ/UTMϩ
Minimum input frequency
0.1 rpm * The resolution depends on the settings of the rotation speed
and the number of the output pulse.
Minimum detection pulse width
5 ȣs
Circuit configuration
No-voltage contact input (minus common)
Open collector connectable (Ic = Approx. 10 mA)- Encoder input (Pulse input)
Compatible with encoder option of UTM⋝/UTM⋜

Display
section

Display unit

Character height 8 mm Numerical display by 7-segment green LED

Display content

Switchable numeric display: torque, rotation speed, angle (TC80-CCL)

Display frequency

Selectable from 1, 3 , 6, 13, and 25 times/sec.

Status display

RUN / SD / RD / ERR (TC80-CCL), MS / NS (TC80-EIP)

Output signal(2)

Selectable/ configurable
Transistor’s open collector output

Input signal(1)

Selectable/ configurable
Contact (relay, switch, etc.) or non contact (transistor, photo coupler, etc.)
Input is ON when shorted to COM terminal

torque, rotation speed, power/ angle (TC80-D3V, TC80-EIP)

External
I/O section

Interface

6XLWDEOHWRUTXHPHWHU

870⋝

870⋜

8709

87)⋜

GeneraPower supply voltage
performance Power consumption
Operating conditions‫ ە‬High-speed sampling at 16000 times/sec.
‫ ە‬Equipped with display and comparison function forconnection data*1 of UTM⋝/UTF⋜‫ ە‬Torque, rotation speed and torque-angle curve
can be monitored.
‫ ە‬Equipped with hold data memory function(Torque, rotation speed & angle. Store 100 latest data)‫ ە‬The unit supplies electric power to UTMϪUTMϩ/UTMV.
(Also to an optional rotary encoder*2)*1 RS-485 interface (torque, rotation speed)*2 UTMϪ is compatible with rotary encoder option(R)(H).
Rotary encoder option is not available for UTMV.

CCL: CC-Link interface TC80-CCL
D3V: D/A converter voltage output (3 ch) TC80-D3V
EIP: EtherNet/IP interface TC80-EIP

485: RS-485 interface
USB: USB interface

DC 24 V±15%
6 W typ. (TC80-CCL, TC80-D3V), 7 W typ. (TC80-EIP)
Operation temperature: -10 to +50ΥStorage temperature: -20 to +85Υ
Humidity: 85% RH or less (non-condensing)34(W) × 88(H) × 91(D) mm (Not including projections)
Approx. 230 gDimension
WeightAttachments Quick manualҩҩҩҩҩҩҩҩҩҩҩҩҩҩҩҩҩҩҩ 2
CC-Link connector (TC80-CCL)ҩҩҩ 1
Operating tool (TC80-D3V)ҩҩҩҩҩҩҩҩ 1

Mini driverҩҩҩҩҩҩҩҩҩҩҩҩҩҩҩ 1
Ferrite core (TC80-CCL)ҩҩҩ 1
Various I/O connectorҩҩҩҩҩ2(TC80-CCL, TC80-EIP), 3(TC80-D3V)

Optional
CATM321: Cable for UTM⋝ connection 2 m
accessories CATM351: Cable for UTM⋝ connection 5 m
CATM312: Cable for UTM⋝ connection 10 m
CATM21: Cable for UTM⋜ connection 2 m
CATM51: Cable for UTM⋜ connection 5 m
CATM12: Cable for UTM⋜ connection 10 m
CATM(R)321: Cable for connection UTM ⋝ rotary encoder 2 m
CATM(R)351: Cable for connection UTM ⋝ rotary encoder 5 m
CATM(R)21: Cable for connection UTM ⋜ rotary encoder 2 m
CATM(R)51: Cable for connection UTM ⋜ rotary encoder 5 m
CATM(RC)321: Cable for UTM⋝(RC) connection 2 m
CATM(RC)351: Cable for UTM⋝(RC) connection 5 m
CE marking EMC directives EN61326-1
certification

CA81-USB: miniUSB-computer USB cable 1.8 m
CN7A: Connector for D/A converter (3 ch)(Same as the attachment)
CN74: CC-Link connector (Same as the attachment)
CN75: CC-Link connector (Y type branch connector)
CN76: CC-Link connector (Terminator connector)
CN85: 13p connector for power source/sensor/RS-485‫( ؙؙؙ‬Same as the attachment)
CN87: 10p connector for external I/O(Same as the attachment)
E04SR211132: Ferrite core
TSU03: Lightning surge protector (DC)* Please note that there are possibilities of individual differences in a color tone on display devices such as LEDs,fluorescent display tubes and LCDs due to manufacturing process or production lots.

Application software for USB interface
With communication through USB interface, logging,graph display, setting parameters, and calibration
can be done.

‫ۋ‬ceo = 30 V, Ic = 50 mA

External dimension
Unit: mm‫ ڦ‬TC80-CCL‫ ڦ‬TC80-D3V

Compact size

34

88
101

88
101

It can be fit into a limited space!
Light weight & compact body for 35 mm DIN rail mount.

34‫ ڦ‬TC80-EIP

91
DIN rail mountadapter

Structure of product codeձ Standard unit
SignInterface

TC80-CCL CC-Link
RS-485, USB

88
101

TC80‫ٸ‬ձ

TC80-D3V D/A converter voltage output (3 ch)
RS-485, USB
TC80-EIP EtherNet/IP
RS-485, USB

34

unlPULXE

www.unipulse.tokyo/en/