FN2317
HandBrake Load Cell

SPECIFICATIONSHigh accuracy ±0.5 % FS

With many years of experience as a designer andmanufacturer, TE Connectivity (TE) offers solutions toautomotive industry and can supply standard or custom
sensors for specific uses and testing environments.

On-board equipment testing
Production quality control
Laboratory and Research

SENSOR SOLUTIONS /// FN2317_en“Easy to mount” through clamping collars

Through careful placement of metallic strain gages
inside, the sensors provides accurate measurements
regardless of the point of application of force.

Compact and ergonomic design
Tension measurements
High accuracy regardless force
application point

APPLICATIONS

Compatible with most handbrakes

The FN2317 is designed with the purpose of allowing quick
and easy installation when measuring forces applied to thehandle of the hand brake in automobiles. Clamping collars
facilitate the prompt and uncomplicated installation of theload cell under the hand brake. As such, the FN2317 iscompatible with most models of hand brakes.

FEATURES

Ranges 500 N and 1000N (100 and 200 lbf)

On request, Instruction documents can be provided
to ease the selection and use of our sensors and provide
helpful tips.

07/2017

Page 1

FN2317
HandBreak Load Cell

STANDARD RANGES
Ranges in N (FS)

500

1000

Ranges in lbf (FS)

100200PERFORMANCE SPECIFICATIONS (typical values at temperature 23±3°C)
Parameters
Operating Temperature Range (OTR)-20 to 80° C [-4 to 176° F]

Compensated Temperature Range (CTR)

0 to 60º C [32 to 140° F]

Zero Shift in CTR<0.5% F.S. / 50º C [/100° F]

Sensitivity Shift in CTR<1% of reading / 50º C [/100° F]

Over-Range
Without Damage

1.5 x F.S.

Without Destruction

3 x F.S.

Accuracy
Combined non-linearity & hysteresis

±0.5% F.S.

Electrical Characteristics
ModelFN2317

FN2317-A1

FN2317-A2

Supply Outage

10Vdc

10 – 30Vdc±15Vdc (±12 to ±18Vdc)

Sensitivity “FSO” 4±1.5mV/V

4V ±0.2V

5V ±0.2V

Zero Offset 4±5% F.S.

0.5V ±0.2V

0V ±0.2V

Input Impedance/Consumption

350 to 700Ω

<50mA

50mA

Output Impedance

350 to 700Ω

1 kΩ 51 kΩ 5Insulation under 50Vdc≥100MΩ≥100MΩ≥100MΩ

Notes
1. standard electrical termination: cable gland with 3mm shielded cable, 2meters length
2. Material: stainless steel.
3. Protection Index: IP50
4. Other signal output on request
5. Output impedance < 100Ω on request
6. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN2317_en

07/2017

Page 2

FN2317
HandBreak Load CellDIMENSIONS & WIRING SCHEMATIC (IN METRIC)
Wiring Schematic

SENSOR SOLUTIONS /// FN2317_en

07/2017

Page 3

FN2317
HandBreak Load CellOPTIONS
A1 : Amplified Tension output with unipolar power supply
A2 : Amplified Tension output with bipolar power supply
SC : LEMO Integrated connector output instead of standard cable gland, Mating supplied
L00M : Special cable length, replace “00” with total length in meters

ORDERING INFORMATIONFN2317 - A1 - 1KN - /L5M
Options (L05M, SC, ...)
Range in Newton
Amplified version (none, A1 or A2)
Model

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s StandardTerms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN2317_en

All Rights Reserved.

07/2017

Page 4