MODEL 4000A MEMS DC
ACCELEROMETERSPECIFICATIONS.

FEATURES

±2g to ±100g Dynamic Range
High Over-Range Protection
Gas Damped MEMS Sensors
Low Power Consumption
Screw Mounting
8 to 36Vdc Excitation Voltage

APPLICATIONS

Low Frequency Monitoring
Transportation
Vibration Sensing
Test & Instrumentation
Machine Control
Motion Analysis

SENSOR SOLUTIONS /// Model 4000A Rev B

MEMS DC Accelerometer
DC to 1300Hz Frequency Response
Amplified Output, Signal ConditionedTemperature CompensatedEconomical, Light Weight

The Model 4000A is an economical signal conditioned
MEMS DC accelerometer packaged in an anodized
Aluminum housing. The accelerometer is available inranges from ±2 to ±100g with a wide bandwidth from DCto 1300Hz. The model 4000A accelerometers incorporate
gas damped MEMS sensing elements that provide
outstanding performance over an operating temperature
range of -20°C to +85°C.
The accelerometer is designed for 8 to 36Vdc excitation
voltage and the signal conditioned output incorporates a2.5V reference that offers the user an optional differential
or single-ended output.
For a triaxial version, TE Connectivity also offers themodel 4030 and 4630 accelerometers.

11/2019

Page 1

MODEL 4000A ACCELEROMETERPERFORMANCE SPECIFICATIONS
All values are typical at +24°C, 80Hz and 12Vdc excitation unless otherwise stated. TE Connectivity reserves the right to update and changethese specifications without notice.
Parameters
DYNAMIC
Range (g)Sensitivity, Differential (mV/g)
Frequency Response (Hz)Natural Frequency (Hz)Non-Linearity (%FSO)
Transverse Sensitivity (%)
Damping Ratio
Shock Limit (g)Residual Noise (µV RMS)
Spectral Noise (µg/Hz)±210000-200700±1<30.7
500050035ELECTRICAL
Zero Acceleration Output (mV)
Excitation Voltage (Vdc)
Excitation Current (mA)
Bias Voltage (Vdc)
Full Scale Output (differential)
Full Scale Output (single-ended)
Output Resistance (Ω)
Insulation Resistance (MΩ)
Turn On Time (msec)
Ground Isolation±100
8 to 36<52.5±2 Vpk+0.5 to 4.5 Vpk<100>100<100Isolated from Mounting Surface

Differential

ENVIRONMENTAL
Thermal Zero Shift (%FSO/°C)
Thermal Sensitivity Shift (%/°C)
Operating Temperature (°C)
Storage Temperature (°C)
Humidity±0.014±0.028-20 to 85-20 to 85
Epoxy Sealed

TypicalTypicalPHYSICAL
Case Material
Cable
Weight (grams)
Mounting
Mounting TorqueAnodized AluminumPFA Insulated Leads, Braided Shield, PU Jacket
7
2x #4 or M3 Screws
3 lb-in (0.3 N-m)

Notes
±5400
0-300
800±1
<30.7
5000
300
38±102000-350
1000±1
<30.7
5000
300
75±20
100
0-600
1500±1
<30.7
5000
350
132±50400-800
4000±1<30.65000
400
316±100
200-1300
6000
±1
<30.5
5000
350
516

±10%
±5%
<1 Typical
PassbandPassband@100Vdc

Calibration supplied:

CS-FREQ-0100

NIST Traceable Amplitude Calibration from 20Hz to ±5% Frequency Response Limit

Optional accessories:

AC-D02652
121

Triaxial Mounting Block
3-Channel Precision Low Noise DC Amplifier

The information in this sheet has been carefully reviewed and is believed to be accurate; however, no responsibility is assumed for inaccuracies. Furthermore, thisinformation does not convey to the purchaser of such devices any license under the patent rights to the manufacturer. TE Connectivity reserves the right to makechanges without further notice to any product herein. TE Connectivity makes no warranty, representation or guarantee regarding the suitability of its product for any
particular purpose, nor does TE Connectivity assume any liability arising out of the application or use of any product or circuit and specifically disclaims any and allliability, including without limitation consequential or incidental damages. Typical parameters can and do vary in different applications. All operating parameters must
be validated for each customer application by customer’s technical experts. TE Connectivity does not convey any license under its patent rights nor the rights ofothers.

SENSOR SOLUTIONS /// Model 4000A Rev B

11/2019

Page 2

MODEL 4000A ACCELEROMETERDIMENSIONSSCHEMATICSENSOR SOLUTIONS /// Model 4000A Rev B

11/2019

Page 3

MODEL 4000A ACCELEROMETERTYPICAL FREQUENCY AND PHASE RESPONSE

2g Range
5g Range
10g Range
20g Range
50g Range
100g-200g Range

Typical Frequency Response
40353025Sensitivity Deviation (%)

2015
1050-5-10-15-20-25-30-35-40
101001000

10000Frequency (Hz)2g Range
5g Range
10g Range20g Range
50g Range
100g Range
200g Range

Typical Phase Response
4035302520Phase (degrees)

151050-5-10-15-20-25-30-35-40
10100

1000

10000

Frequency (Hz)SENSOR SOLUTIONS /// Model 4000A Rev B

11/2019

Page 4

MODEL 4000A ACCELEROMETER

ORDERING INFORMATION
4000A

GGG

ZZZ

Range
002=2g005=5g010=10g020=20g050=50g
100=100g
Cable length
060 = 60 inches, 5 feet
120 = 120 inches, 10 feet
240 = 240 inches, 20 feet
197 = 197 inches, 5 meters
394 = 394 inches, 10 meters
Example;4000A-002-060
Model 4000A, 2g range, 60inch (5ft) cable length

MEAS France SAS and Measurement Specialties (China) , Inc., are TE Connectivity companies.
TE Connectivity, TE, TE connectivity (logo) are trademarks. All other logos, products and/or company names referred to herein might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes nowarranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard Termsand Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuse of theproduct. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2017

TE Connectivity Ltd. All Rights Reserved.

SENSOR SOLUTIONS /// Model 4000A Rev B

11/2019

Page 5