Flush DiaphragmPressure Transducers
Model A-105 and A-205
UNITIZED FLUSH DIAPHRAGM
15 TO 15,000 PSI

1-888-282-9891

STAINLESS STEEL
Model A-105
Model A-205

www.honeywell.com/sensing

Sensotec Sensors

Models A-105, A-205 Subminiature Pressure Transducers are manufactured with a unitizedstainless steel diaphragm. The advantage of this type of design is that a thin diaphragm andheavy sidewalls are made from one piece of stainless steel. This unitized diaphragm is rugged,
but at the same time can be made thin enough to measure low pressures. Available pressureranges span from 15 to 15,000 psi. These models can be used in corrosive fluid environments.
Models A-105 and A-205 have welded stainless steel electrical connectors as an integral part of
the transducer body. A-105’s and A-205’s are recommended for applications involving roughhandling or where a completely hermetically sealed transducer is required.

PR-22

Dimensions
Model A-105 (Order Code AP311)
Model A-205 (Order Code BP312)

Available Ranges* (All Models)
15; 25; 50; 75; 100; 150; 200; 300; 500; 750; 1000;
1500; 2000; 3000; 5000; 7500; 10,000; 15,000 psi*Stocked ranges (in bold) for Models A-105 only.

Options (See Appendix)
Temperature compensated 1b, 1c, 1f
Premium Options: 1d, 1e, 1g, 1h, 1i
Accessories: Mating connectors and connector/cable assemblies

1-888-282-9891

ModelsA-105
A-205

PERFORMANCEENVIRONMENTAL

High Pressure Ranges

Pressure Ranges..........................
Non-Linearity andHysteresis (max)* .....................
Non-repeatability (max)*...............
Output (standard)*........................
Resolution ....................................

<100 psi (1 bar to 7 bar)

100 to 15,000 psi (10 bar to 1000 bar)

1% F.S.
1% F.S.
.1mV/psi (@ 5VDC)
Infinite+ /-0.5% F.S.
+ /-0.1% F.S.
2mV/V
Infinite

Temperature, Operating ...............
Temperature, Compensated ........
Temperature Effect– Zero (max)* ...........................
– Span (max)* ..........................

-65° to 300° F (-54° to 150° C)
60° to 160° F (15° to 71° C)-65° to 300° F (-54° to 150° C)
60° to 160° F (15° to 71° C)

0.001mV/ ° F (0.0018mV/ ° C)
0.002mV/ ° F (0.0036mV/ ° C)

0.01% F.S. / ° F (0.018% F.S. / ° C)
0.02% Rdg. / ° F (0.036% Rdg. / ° C)

Strain Gage Type .........................
Excitation (calibration) ..................
Excitation (acceptable) .................
Insulation Resistance ...................
Bridge Resistance ........................
Shunt Calibration Data .................
Wiring Code (std)A-105/A-205 .............................
Electrical Termination (std)A-105/A-205 .............................

Bonded foil
5VDC
Up to 5VDC or AC
5000 megohm @ 50VDC
350 ohms
Included

Bonded foil
5VDC
Up to 5VDC or AC
5000 megohm @ 50VDC
350 ohms
Included#2 (See Pg. AP-8)#2 (See Pg. AP-8)

PTIH-10-6P or equiv.
(Hermetic stainless)

PTIH-10-6P or equiv.
(Hermetic stainless)

PT06A-10-6S or equiv.

PT06A-10-6S or equiv.

Media............................................
Overload-Safe ..............................
Overload-Burst .............................
Pressure Port
A-105 ........................................
A-205 ........................................
Dead Volume................................
Wetted Parts Material...................
Type .............................................
Weight ..........................................
Case Material ...............................

Gas, Liquid
100% over capacity400% over capacity

Gas, Liquid
50% over capacity
400% over capacity

7/16-20 UNF male
1/4-18 NPT male
Flush diaphragm17-4 PH StainlessGage
1-2 oz.
17-4 PH Stainless

7/16-20 UNF male
1/4-18 NPT male
Flush diaphragm17-4 PH StainlessGage
1-2 oz.
17-4 PH Stainless

IN-LINE AMPLIFIERS

Outputs Available .........................

0-5VDC, 4-20mA (>50 psi)
0-1VDC (<50 psi)

0-5VDC, 4-20mA

FLUSH DIAPHRAGM

Low Pressure Ranges

Mating Connector (not incl.)
A-105/A-205 .............................

MECHANICAL

7/16-20 UNF
1/4-18 NPT

PRESSURE

ELECTRICAL

Pressure Threads
Order Code AP311
Order Code BP312

NOTES *Unit of measure in specifications is different for low ranges than high ranges.

General Information
How to order (See Pg. AP-19)
Gage/Absolute pressure selection flow chart (See Pg. PR-1)●お問い合わせは下記へ ̶PR-23