XFL225D
Compression Miniature Load Cell
SPECIFICATIONS

0-10 N to 0-5 kN [0-2.25 lbf to 0-1.12 klbf]
Diameter 25mm [0.98”]
For Static and Dynamic Applications
Strain Relief Spring

The XFL225D is a very flat miniature load cell with a temperaturecompensation module integrated into the output cable. This
design allows TE CONNECTIVITY to manufacture extremely
small sensors without sacrificing thermal zero and sensitivity
performance.

FEATURES

Load Washer ModelFlat sensor
Inside hole  8mm or flat surface
Temperature compensatedAPPLICATIONS

Strain table measurement
Micro component assembly tools
Screw and insert test benches
Press fit monitoring
Laboratory

SENSOR SOLUTIONS /// XFL225D_en

The sensing element is fitted with a fully temperaturecompensated Wheatstone bridge equipped with high stability
micro-machined silicon strain gages.
The XFL225D can withstand considerable overloads
and measure strain during compression in static and dynamic
applications. A strain relief spring strengthens the cable output.
This sensor is available in a load washer model (the diameter ofthe hole is 8.2 mm) to measure bolt tightening forces.
With a long standing experience as a designer andmanufacturer of sensors, TE CONNECTIIVTY often works withcustomers to design or customize sensors for specific uses andtesting environments.
On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpfultips.

04/2019

Page 1

XFL225D

Miniature Load CellPERFORMANCE SPECIFICATIONS (typical values at temperature 23°C)

Ranges (FS)

10 N

20 N

50 N

100 N

200 N

500 N

1 kN

2 kN

5 kN

Ranges (lbf)

2.248

4.50

11.24

22.48

45.0

112.4

225

4501124

Aluminium

Material

Stainless Steel

Stiffness (N/m)

1.2E+06 2.5E+06 6.8E+06 1.3E+07 2.8E+07 5.7E+07 8.3E+07 1.8E+08 3.4E+08

Stiffness (lbf/ft)

8.2E+04 1.7E+05 4.7E+05 8.9E+05 1.9E+06 3.9E+06 5.7E+06 1.2E+07 2.3E+07

Specifications
Power supply

10Vdc

Overrange Without Damage

2 x FS

Overrange Without Destruction

3 x FS

Sensitivity (FSO)

150 mV

Offset<±10 mV

Input Impedance

1500 ohms

Output Impedance

500 ohms

Linearity< ±1% de l'EMHysteresis< ±1% de l'EM

Operating Temperature Range (OTR)-40°C to +120°C (-40 to 250°F)

Compensated Temperature Range (CTR)

0°C to +60°C (32 to 140°F)

Thermal Zero Shift in CTR< 2% de l'EM/50°C

Thermal Sensitivity Shift in CTR< 2%/50°C

Insulation> 100 Mohms

Protection Index

IP50Notes
1. Signal goes negative in compression with standard wiring configuration
2. Electrical Termination: Cable: Shielded cable with 4 wires (AWG36), standard length 2 m [6.5 ft] with strain relief spring ;Compensation module at 1m [3.25 ft] from transducer
3. Material: Body in stainless steel or aluminum alloy depending on F.S.
4. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// XFL225D_en

04/2019

Page 2

XFL225D

Miniature Load CellDIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)
Wiring Schematic

Identical dimensions for model XFL225 without the through hole of 8.2 mm.

Dimensions in mm [inch]
F.S. Ranges in N

10 - 20 - 50 - 100

A

SENSOR SOLUTIONS /// XFL225D_en

200 - 500 - 1000
3 [0.12]04/2019

2000 - 5000
5 [0.2]

Page 3

XFL225D

Miniature Load CellOPTIONSZ0

CTR -40°C to +20°C (-40°F to 68°F)

Z04

CTR -40°C to +90°C (-40°F to 194°F)

Z1

CTR -20°C to +40°C (-4°F to 104°F)

Z3

CTR +20°C to +80°C (68°F to 176°F)

Z35

CTR +20°C to +120°C (68°F to 248°F)

L5M,L10M,L15M

Special total cable length (ex: "L05M= 5m length).(Standard total cable length = 2m)

M0.25M , M0.5M

Special distance between sensor and module(ex:M0.5M=0.5m) .(Standard distance =1m)

ORDERING INFO

XFL225D - 10KN - /Z0
Options (Z0, L3M, ...)
Range in Newton
ModelTE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// XFL225D_en

All Rights Reserved.

04/2019

Page 4