取扱説明書

電源品質アナライザ

KEW

6315

共立電気計器株式会社

目次

KEW6315

目次 ········································································································································································ 1
梱包内容の確認 ················································································································································ 5
安全に関するご使用上の注意 ···················································································································· 8
1章

2章

3章

4章

製品の概略·········································································································································· 11
1.1

機能概略 ········································································································································ 11

1.2

特長 ·················································································································································· 13

1.3システム構成図 ··························································································································· 14

1.4測定までの手順··························································································································· 15

各部の名称·········································································································································· 16
2.1

表示部（ＬＣＤ）／キー操作部 ································································································· 162.2コネクタ部 ······································································································································· 17

2.3

側面部 ············································································································································· 182.4電圧用測定コードとクランプセンサ······················································································ 19

基本操作··············································································································································· 20
3.1

操作キーの説明 ·························································································································· 20

3.2

ＬＣＤの上部に表示するマーク ······························································································ 21

3.3

画面の表示記号·························································································································· 22

3.4

バックライトとコントラストの調整 ··························································································· 22

3.5

画面表示と画面構成 ················································································································· 23•

瞬時値／積算値／デマンド ··························································································· 23•

ベクトル ·································································································································· 24•

波形 ········································································································································· 24•

高調波 ···································································································································· 25•

電源品質································································································································ 26•

設定 ········································································································································· 26

測定前の準備 ····································································································································· 27
4.1

購入後はじめにすること··········································································································· 27•

入力端子プレートを貼る ·································································································· 27•

識別マーカーを取り付ける ····························································································· 28

4.2電源について ································································································································ 29•

電池の使用··························································································································· 29•

画面の表示／電池の残量 ····························································································· 30-1-KEW6315

KEW6315

目次•

乾電池のセット方法··········································································································· 31•

電源コードの接続··············································································································· 31•

電源の定格··························································································································· 32

4.3

ＳＤカードの挿入／取り出し方法 ·························································································· 33•

挿入方法································································································································ 34•

取り出し方法 ························································································································ 34

4.4電圧用測定コードとクランプセンサの接続 ······································································· 35

4.5電源の投入 ··································································································································· 36•

初期画面表示 ······················································································································ 36•

注意表示································································································································ 36

4.6

5章

記録の手順 ··································································································································· 37•

記録の開始 ··························································································································· 37•

記録の終了 ··························································································································· 38•「かんたんスタートナビ」で記録を開始する ······························································ 39

設定 ························································································································································ 47
5.1

設定項目 ········································································································································ 47

5.2

基本設定 ········································································································································ 48•

結線に関する設定 ············································································································· 49•

結線について ······················································································································· 51•

電圧測定に関する設定 ··································································································· 53•

ＶＴ／ＣＴについて ··············································································································· 54•

電流測定に関する設定 ··································································································· 56•

外部入力端子／基準周波数に関する設定 ···························································· 58

5.3測定設定 ········································································································································ 59•

デマンドに関する設定 ······································································································ 59•

デマンドについて ················································································································ 62•

高調波解析に関する設定······························································································· 63•

電源品質（イベント）しきい値設定················································································ 65•

フリッカのフィルタ係数に関する設定 ········································································· 69•

進相コンデンサの目標力率値に関する設定 ·························································· 70

5.4

記録設定 ········································································································································ 71
記録項目に関する設定 ··············································································································· 72

KEW6315-2-

目次

KEW6315•

記録項目について ············································································································· 73•

記録方法に関する設定 ··································································································· 74•

保存可能な時間について ······························································································· 76

5.5

その他 ············································································································································· 77•

システム環境に関する設定 ··························································································· 77•

本体システムに関する設定 ··························································································· 79

5.6

6章

保存データ ····································································································································· 82•

記録データに関する操作 ································································································ 82•

保存データの種類について ··························································································· 87•

本体設定の保存と読み込みに関する操作······························································ 89

画面ごとの表示項目 ························································································································ 92
6.1

瞬時値「Ｗ」 ··································································································································· 92•

測定値を一覧で表示する ······························································································· 92•

大きな文字で測定値を表示する ·················································································· 96•

トレンドグラフを表示する ································································································· 97•

表示測定項目と表示位置を変更する ········································································ 99

6.2

積算値「Ｗｈ」 ····························································································································· 100

6.3

デマンド ········································································································································ 102•

測定値を表示する··········································································································· 102•

時間内推移図を表示する ···························································································· 103•

デマンド推移図を表示する ·························································································· 104

6.4ベクトル ········································································································································ 105

6.5

波形 ··············································································································································· 107

6.6

高調波 ·········································································································································· 108•

高調波をバーグラフで表示する ················································································ 108•

高調波をリストで表示する ··························································································· 112

6.7

電源品質 ····································································································································· 114•

電源品質を低下させる事象とその現象 ································································· 114•

イベントの発生状況を表示する ················································································· 116•

フリッカ測定値を一覧表示する ·················································································· 120•

Ｐｓｔ，１ｍｉｎのトレンドグラフを表示する·································································· 121•

Ｐｌｔの推移図を表示する ······························································································· 122-3-

KEW6315

KEW6315

目次

7章

その他の機能 ·································································································································· 1238章

周辺機器との接続 ························································································································· 125
8.1

ＰＣ：コンピュータへのデータ転送 ······················································································ 125

8.2

Bluetooth®を使用する ············································································································ 126

8.3

外部機器との信号制御 ········································································································· 126
•

8.4
9章

入出力端子を接続する ································································································· 126
測定ラインから電源を供給する ························································································· 128

設定・解析用ＰＣソフトウェア······································································································ 129

10章 仕様 ····················································································································································· 130
10.1 安全要求仕様 ··························································································································· 130
10.2 一般仕様 ····································································································································· 130
10.3 測定仕様 ····································································································································· 133•

測定項目別、解析データ数 ························································································· 133•

瞬時測定項目 ··················································································································· 134•

演算項目····························································································································· 137•

積算測定項目 ··················································································································· 140•

ＤＥＭＡＮＤ測定項目 ······································································································ 143•

高調波測定項目 ·············································································································· 144•

電源品質測定項目 ········································································································· 149

10.4 クランプセンサの仕様 ············································································································ 152
11章 故障かなと思ったら ······················································································································· 157
11.1 トラブルシューティング ··········································································································· 157
11.2 エラーメッセージの内容とその対処方法 ········································································ 158
アフターサービス ·········································································································································· 161

KEW6315-4-

梱包内容の確認

KEW6315

梱包内容の確認
このたびは弊社電源品質アナライザ KEW6315 をご購入していただきありがとうございます。
まずお手元に届きました本製品の梱包内容を確認してください。

●梱包内容
1

本体

KEW6315：1 台

2電圧用測定コード3

電源コード

MODEL7169 ：1 本

4

USB ケーブル

MODEL7219 ：1 本

5

クイックマニュアル

1冊

6

CD-ROM

1枚

7

電池

単３形アルカリ乾電池（LR6）：6 個

8

SD カードM-8326-02

9

キャリングバッグ

MODEL9125 ：1 個

10

入力端子プレート

1枚

11

識別マーカー

各 4 本×8 色（赤/青/黄/緑/茶/灰/黒/白)MODEL7255：1 セット
赤、白、青、黒、各 1 本 （ワニグチ付き）：1 枚（2GB）

オプション
12

クランプセンサ

ご購入台数

13

クランプセンサの取扱説明書

1冊

14

マグネット付携帯ケース

MODEL9132

15

電源供給アダプタ

MODEL8312（CAT.Ⅲ150V、CAT.Ⅱ240V）-5-

KEW6315

KEW6315

梱包内容の確認

1.本体

5.ｸｲｯｸﾏﾆｭｱﾙ

2.電圧用測定コード

6.CD-ROM

3.電源コード

7.電池

8.ＳＤカード

2GB

10.入力端子プレート

KEW6315

11.識別マーカー-6-

M-8326-02

4.USB ケーブル

9.キャリングバッグ

梱包内容の確認

KEW6315

12.クランプセンサ（ご購入台数）
50Aﾀｲﾌﾟ(φ24mm/φ75mm)

M-8128/KEW8135

100A ﾀｲﾌﾟ(φ24mm)

M-8127200A ﾀｲﾌﾟ(φ40mm)

M-8126500A ﾀｲﾌﾟ(φ40mm)

M-8125

1000A ﾀｲﾌﾟ(φ68/110mm) M-8124/KEW8130
13．クランプセンサの取扱説明書

14.マグネット付携帯ケース

3000A ﾀｲﾌﾟ(φ150/170mm) KEW8129/8133

15.電源供給アダプタ

10A ﾀｲﾌﾟ(φ24mm)

M-8146

10A ﾀｲﾌﾟ(φ40mm)

M-8147

10A ﾀｲﾌﾟ(φ68mm)

M-8148

1A ﾀｲﾌﾟ(φ24mm)

M-8141

1A ﾀｲﾌﾟ(φ40mm)

M-8142

1A ﾀｲﾌﾟ(φ68mm)

M-8143

販売終了品：
KEW8129/M-8141/M-8142/M-8143●収納方法
ご使用後は下記のように収納してください。

●製品の間違い、品不足、破損、印刷不良等がございましたらお買上店（販売店）
までご連絡をください。
●クイックマニュアルには保証書が付いています。大切に保管してください。

-7-

KEW6315

KEW6315

安全に関する使用上の注意

安全に関するご使用上の注意
本製品は IEC 61010-1：電子測定装置に関する安全規格に準拠して、設計・製造の上、検査合格した最良の状態にて出荷されています。
この取扱説明書には、使用される方の危険を避けるための事項および本製品を損傷させずに長期間良好な状態で使用して頂くための事柄が書かれています。ご使用前に必ずこの取扱説明書をお読みください。

警告
取扱説明書について●本製品を使用する前に必ずこの取扱説明書をよく読んでご理解ください。
●この取扱説明書は手近な所に保管し、必要な時にいつでも取り出せるようにしてください。
●取扱説明書で指定した製品本来の使用方法を守ってください。
●取扱説明書の安全に関する指示に対しては、指示内容を理解の上、必ず守ってください。
●付属のクイックマニュアルはこの取扱説明書をよくお読みになってから使用してください。
●使用するクランプセンサの取扱いについては、クランプセンサの取扱説明書も必ずよくお読みになって、ご理解ください。
以上の指示を必ず厳守してください。指示に従わないと、怪我や事故の恐れがあります。危険および警告、注意に反した使用により生じた事故や損傷については、弊社として責任と保証を負いかねます。
本製品に表示のマークは、安全に使用するため取扱説明書を読む必要性を表しています。尚、このマークには次の 3 種類がありますので、それぞれの内容に注意してお読みください。

危険
警告
注意

：

この表示を無視して誤った取扱いをすると、人が死亡または重傷を負う危険性が高い内容を示しています。

： この表示を無視して誤った取扱いをすると、人が死亡または重傷を負う可能性が想定される内容を表示しています。
： この表示を無視して誤った取扱いをすると、人が傷害を負う可能性が想定される内容および物的損害の発生が想定される内容を示しています。

○測定カテゴリについて
安全規格 IEC61010 では測定器の使用場所についての安全レベルを測定カテゴリという言葉で規定し、以下のように O～CAT.Ⅳの分類をしています。この数値が大きいほど過渡的なインパルスが大きい電気環境であることを意味します。
CAT.Ⅲで設計された測定器は CAT.Ⅱで設計されたものより高いインパルスに耐えることができます。
O(None, Other) ： 主電源に直接接続しないその他の回路です
CAT II ： コンセントに接続する電源コード付機器の１次側の電気回路
CAT III ： 直接配電盤から電気を取り込む機器の１次側および分岐部からのコンセントまでの電路
CAT IV ： 引込み線から電力量計および１次過電流保護装置（配電盤）までの電路

KEW6315-8-安全に関する使用上の注意

KEW6315

危険●指定した操作方法および条件以外で使用した場合、本体の保護機能が正常に動作せず本器を破損したり感電等の重大な事故を引き起こす可能性があります。本器の使用前あるいは指示結果に対する対策を取る前に、既知の電源で正常な動作を確認してください。
●測定カテゴリに準じてCAT ⅣではAC300V、CAT ⅢではAC600V、CAT ⅡではAC1000Vより高い電圧のある回路では絶対に使用しないでください。
●引火性ガスや爆発性のガスおよび、蒸気のある場所で使用すると大変危険ですので、使用しないでください。
●本製品や手が濡れている状態や、湿気などの水滴が付着した状態では、絶対に使用しないでください。
測定について●測定の際には、測定範囲を超える入力を加えないでください。
●測定中は絶対に電池蓋を開けないでください。
電池について●測定中は絶対に電池交換を行わないでください。
●銘柄や種類の違う電池を混ぜて使用しないでください。
電源コードについて●電源コードは必ずコンセントに接続してください。
●使用する電源コードは、付属の専用コードをご使用ください。
電源コネクタについて●電池駆動時の電源コネクタは絶縁されていますが、絶対に触らないでください。
電圧用測定コードについて●付属のものをご使用ください。
●測定コードのキャップの装着は、測定カテゴリに適した使用をしてください。
●測定コードと本体の測定カテゴリが違っている場合は低い方の測定カテゴリが優先されます。測定電圧と定格が合っているか必ず確認してください。
●測定に必要のない電圧用測定コードは絶対に接続しないでください。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定の際は指先等が、保護用フィンンガガードを超えることのないよう充分注意してください。
保護用フィンンガガード：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●先端の金属部で測定ラインの 2 線間を接触させないでください。
●先端の金属部には絶対に触れないでください。
●測定コードのコード内部から金属部分または外装被覆と異なる色が露出したときは、直ちに使用を中止してください。
クランプセンサについて●本製品専用のものをご使用ください。
●測定電流と定格が合っているか必ず確認し、対地間最大定格電圧以下の電路で使用してください。
●測定に必要のないものは絶対に接続しないでください。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定の際は指先等が、バリアを超えることのないよう充分注意してください。
バリア：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●必ずブレーカーの二次側に接続してください。1 次側は電流容量が大きく危険です。
●コアを開いたとき、金属部で測定ラインの 2 線間を接触させないでください。

-9-

KEW6315

KEW6315

安全に関する使用上の注意

注意●被測定導線が高温の場合がありますので注意してください。
●各レンジの測定範囲を超える電流や電圧を長時間入力しないでください。
●電源が OFF の状態で、電圧用測定コードやクランプセンサに電圧や電流を入力しないでください。
●埃の多い場所や、水のかかる環境で使用しないでください。
●強力な電磁波が発生したり、帯電したりしているものの近くで使用しないでください。
●振動や衝撃を与えたり、落下させたりしないでください。
●ＳＤカードの向きを必ず確認して正しい方向で本体へ挿入してくだい。無理に挿入しようとするとＳＤカー
ドまたは、本体を破損する恐れがります。
●ＳＤカードを挿入／取り出すときは必ず、ＳＤカードへアクセス中でないことを確認してください。

（アクセス中はが点滅します。）ＳＤカードへアクセス中に取り出しを行うと、保存されたデータや本体を破損する恐れがあります。
クランプセンサについて●クランプセンサのケーブルを折ったり引っ張ったりしないでください。
●測定に使用する電流クランプセンサは、同じ種類を使用してください。
使用後について●必ず電源を OFF にし、電源コード、電圧用測定コードおよびクランプセンサを外してください。
●長期間ご使用にならない場合は、電池を取り外した状態で保管してください。
●持ち運ぶときは、ＳＤカードを本体から抜いてください。
●運搬の際には振動や衝撃を与えたり、落下させたりしないでください。
●高温多湿、結露するような場所および直射日光の当たる場所に放置しないでください。
●クリーニングには研磨剤や溶剤を使用しないで、中性洗剤か水に浸した布を使ってください。
●濡れているときは、乾燥後保管してください。

また、各章の危険、警告、注意、注記（）の内容も必ず守ってください。

本製品に使用している安全記号
取扱説明書を参照する必要があることを示します。
二重絶縁または強化絶縁で保護されている機器を示します。

～ 交流(AC)を示します。
(機能)接地端子を示します。

KEW6315- 10 -

1.1 機能概略
機

KEW6315

1章
章 製品の概略
1.1

機能概略
略
記録の開始
記
始と終了
通
通常の記録開始と、記録に必要なな設定をナビゲーートしてくれる「「かんたんスタートナビ」との２種類類で記録を開始すすることができきます。

詳
詳細は、「4.6
記録
録の手順」３７頁をを参照してください
い。
瞬時値／積
瞬
積算／デマン
ンドの表示
電
電流／電圧／有効電力／皮相電
電力／無効電力のの瞬時値から平均
均値／最大値／
最小値を表示します。画面を切りり替えれば積算
値
値を表示することともできます。またたデマンド目標値値を設定して、測定
定開始から終了ままでのデマンド値を表示します。

詳細は、「6.1 瞬時値「Ｗ」９２頁／
詳／6.2 積算値「Ｗｈ」１００頁／
6 デマンド」１０
6.3
０２頁を参照してください。
ベクトルの表示
示と結線確
確認
測定
定ＣＨの電圧と電電流に対応したベク
クトル図の表示とと、結線
線確認を行いますす。
詳細
細は、「6.4 ベクト
トル」１０５頁を参照
照してください。

- 11 -

KEW6315

1.1 機能概略

KEW6315
波形の
の表示
測定ＣＨの
の電圧と電流に対対応した波形を表示示します。

詳細は、「6.5 波形」１０７頁
頁を参照してくださ
さい。

高調波
高
測
測定ＣＨの電圧と電電流に重畳した高高調波成分を表示示します。

詳
詳細は、「6.6
高調
調波」１０８頁を参照してください。

設定
定(SET UP)機器のの設定や、測定のの設定をします。

詳細はは、「５章 設定」４７頁を参照してください。

KEW
W6315

電源品質(QUALITY))イベントの表
表示
電圧のスウェル
ル／ディップ／瞬停
停／トランジェント
ト、インラッシュカレント、フフリッカを表示しまます。

詳細は、「6.7 電源品質」１１４頁
頁を参照してくだささい。

- 12 -

1.2 特長

1.2

KEW6315特長

本製品は多彩な結線方式に対応したクランプ式電源品質アナライザです。従来の瞬時値、積算値、電力管理のためのデマンド値、高調波解析、電源品質に関するイベント、力率改善のための進相コンデンサ値のシミュレーショ
ンを全て同時に行います。電圧と電流を、それぞれ波形やベクトルで表示することも可能です。
測定データは SD カードまたは内部メモリへ保存し、そのファイルは USB 通信によりパソコンへ転送できます。また、Bluetooth®によって市販の Android 端末にてリアルタイムに測定データを確認することもできます。
安全設計
安全規格 IEC 61010-1 CAT.Ⅳ 300V／CAT.Ⅲ 600V／CAT.Ⅱ 1000V に準拠した安全設計です。
電源品質測定電源品質測定国際規格 IEC61000-4-30 Class S に対応しています。高精度の周波数／電圧実効値測定／高調波解析に加え、電源異常の捕捉・監視に必要なスウェル／ディップ／瞬停／トランジェント／インラッシュカレント／フ
リッカをギャップなしで全て同時に測定することができます。
電力測定
有効／無効／皮相電力および電力量、力率、電流実効値、位相角、中性線の電流を全て同時に測定することが
できます。
結線方式
単相 2 線(4 系統)、単相 3 線（2 系統）、三相 3 線（2 系統）、三相 4 線の各種測定ラインに対応できます。
デマンド測定設定した目標値(契約電力)を超えないように使用状況を簡易的に監視することができます。
波形／ベクトル表示
電圧と電流を波形／ベクトルで表示することができます。
測定データの保存
記録間隔が設定可能なロギング機能を搭載しています。測定データは手動または日時指定で保存できます。また、プリントスクリーン機能でＳＤカードへ画面データの保存ができます。
２つの電源方式
AC 電源と電池のどちらでも駆動できる 2 電源方式です。電池は、単３形アルカリ乾電池（ＬＲ６）と市販の充電式単３形ニッケル水素電池(Ni-MH)の使用が可能です。充電式単３形ニッケル水素電池(Ni-MH)への充電は、ご使用になる電池メーカの充電器をご使用ください。本体での充電は行えません。ＡＣ電源で駆動中に停電が発生した場合に、電源の供給を自動的に電池へ切り換えます。
大画面表示
カラーTFT 液晶の採用により、見易い大画面表示です。
簡単結線で小型軽量設計
クランプ式で簡単に結線ができ小型軽量設計のため、設置や持ち運びに非常に便利です。
アプリケーション
SD カードや内部メモリに保存したファイルを USB 経由でパソコンへダウンロードできます。ダウンロードしたファイルは、付属のＰＣソフト（KEW Windows for KEW6315）で簡単に解析することが可能です。またパソコンから本製品の設定を簡単に変更することもできます。Bluetooth®通信により、Android 端末からリアルタイムに測定値の確認ができます。
外部信号入出力機能
2ｃｈのアナログ入力(DC 電圧)により、温度計や照度計などのアナログ信号を同時に測定することができます。
電源品質にイベントが発生した場合に 1ｃｈのデジタル出力から、接点信号を警報機等に送ることができます。

- 13 -

KEW6315

KEW6315

1.3

1.3 システム構成図

システム構成図

電流入力

交流電圧入力電源コード

デジタル出力(１ch)

記録計・警報機等

アナログ入力(２ch)

温度計・照度計

単３形アルカリ乾電池(ＬＲ６)
単３形ニッケル水素電池（Ｎi-ＭＨ）

ＵＳＢ

SD カードKEW6315- 14 -

ＰＣ

1.4 測定までの手順

KEW63151.4 測定までの手順
ご使用になる前に必ず「安全に関するご使用上の注意」8 頁をお読みください。
測定準備「４章 測定前の準備」２７頁

本体へコードとセンサを接続「4.4 電圧用測定コードとクランプセンサの接続」３５頁

電源を入れる「4.5 電源の投入」３６頁

各測定共通の項目を設定する「5.2 基本設定」４８頁

本体設定の読み込み「5.6 保存データ」８２頁

測定ラインへ結線「5.2 基本設定-結線について」５１頁

結線を確認する「6.4 ベクトル」１０５頁

測定ごとの設定と保存方法を設定する「5.3 測定設定」５９頁／「5.4 記録設定」７１頁

測定値を確認する「6.1 瞬時値「W」」９２頁／「6.4 ベクトル」１０５頁

記録を開始する／記録を終了する「4.6 記録の手順」３７頁

測定データを確認する「６章 画面ごとの表示項目」９２頁

測定ラインからコードとセンサを外し、電源を切る

PC で記録データを解析する「8.1 PC：コンピュータへのデータ転送」１２５頁「９章 設定・解析用 PC ソフトウェア」１２９頁- 15 -

KEW6315

KEW6315

2.1 表示部（LCD）／キー操作部

2章 各部の名称
2.1

表示部（LCD）／キー操作部

表示部（LCD）

キー操作部

KEW6315- 16 -

2.2 コネクタ部

2.2

KEW6315コネクタ部
交流電圧入力端子（VN、V1、V2、V3）

電流入力端子（A1、A2、A3、A4）

電流入力端子カバー
電源コネクタ

結線方式

電流入力端子※交流電圧入力端子

単相 2 線（1 系統）

1P2W×1

VN、V1

A1

単相 2 線（2 系統）

1P2W×2

VN、V1

A1、A2

単相 2 線（3 系統）

1P2W×3

VN、V1

A1、A2、A3

単相 2 線（4 系統）

1P2W×4

VN、V1

A1、A2、A3、A4

単相 3 線（1 系統）

1P3W×1

VN、V1、V2

A1、A2

単相 3 線（2 系統）

1P3W×2

VN、V1、V2

A1、A2、A3、A4

三相 3 線（1 系統）

3P3W×1

VN、V1、V2

A1、A2

三相 3 線（2 系統）

3P3W×2

VN、V1、V2

A1、A2、A3、A4

三相 3 線 3A

3P3W3A

V1、V2、V3

A1、A2、A3

三相 4 線

3P4W×1

VN、V1、V2、V3

A1、A2、A3※結線に使用していない電流端子は、実効値と高調波のみ測定できます。
※測定に使用する電流クランプセンサは、同じ種類を使用してください。

- 17 -

KEW6315

2.3 側面部

KEW6315

2.3 側面部＜コネクタカバーを閉じた状態＞

SD カード部

USB 部
アナログ入力／デジタル出力部＜コネクタカバーを開いた状態＞

SD カードコネクタ

USB コネクタ

アナログ入力／デジタル出力端子

KEW6315- 18 -

2.4 電圧用測定コードとクランプセンサ

KEW63152.4 電圧用測定コードとクランプセンサ＜ワニグチクリップ＞※電圧用測定コード先端部保護用フィンガガード＜クランプセンサ＞

バリア

保護用フィンガガード及びバリアは操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
測定の際は指先等が、保護用フィンガガードまたはバリアを超えることのないよう充分に注意してください。

- 19 -

KEW6315

3.1 操作キーの説明

KEW6315

3章 基本操作3.1

操作キーの説明PRINT SCREEN キー

ファンクションキー

LCD に表示中の画面を、BMP(ビットマップ)ファイルとし

画面ごとの機能を実行します。

て保存します。
DATA HOLD キー／KEY LOCK キー
表示値をホールドします。
※
ホールド時も測定は継続しています。

KEY
LOCK

キーの誤操作を防止したい場合には 2
秒以上の長押しで、全てのキー操作を無効にできます。再度 2 秒以上長押しすると解除します。
ＬＣＤキー
バックライトの ON/OFF します。
2 秒以上の長押しで、明るさ／コントラス
トを変更できます。
カーソルキー
選択項目の移動、表示の切り換えに使用します。
ENTER キー
選択を決定します。

START／STOP キー

電源キー

測定を開始／

電源を ON／

終了します。

OFF します。

ESC キー／RESET キー
カーソルキーで設定選択したものを決定せずに前回の設定に戻します。

ステータス LED
緑
赤

設定キー

点灯：記録測定中
点滅：スタンバイ中

基本設定／測定設定／記録設定／保存データ編

点滅：バックライト OFF

集／その他の設定を変更します。
表示切替キー

W/Wh
ベクトル
波形

KEW6315

瞬時値／積算値／デマン

高調波

高調波電圧／電流／電力を表示しまドを表示します。

解析

す。

位相を表示します。

QUALITY

電圧／電流波形を表示し(電源品質)ます。
- 20 -

スウェル／ディップ／瞬停／トランジ
ェント／インラッシュカレント／フリッカの発生情報を表示します。

3.2 ＬＣＤの上部に表示するマーク

3.2

KEW6315

LCD の上部に表示するマーク
マーク

表示時の状態
電池で駆動しています。残量によって 4 種類に変化します。
AC 電源で駆動しています。

画面の表示更新をホールドしています。

キーがロックされています。

ブザーをオフしています。

SD カードが使用可能です。

SD カードに記録中です。

SD カードに記録できるだけの容量がありません。

SD カードへのアクセスに失敗しています。
内部メモリへ記録可能です。SD カードを挿入していない状態で記録を開始した場合に表示します。

内部メモリへ記録中です。

内部メモリに記録できるだけの容量がありません。
記録待機状態です。
測定値を記録中です。
記録対象の媒体が、いっぱいになっています。
USB が使用可能です。

Bluetooth®が使用可能です。

- 21 -

KEW6315

3.3 画面の表示記号

KEW6315

3.3

画面の表示記号
画面表示記号
相電圧

V※１

P

有効電力

PF

力率

VL※１

＋

消費

－

回生

＋

遅れ

－

進み

アナログ入力DC1

1ch の電圧

線間電圧
無効電力

Q

f

＋

遅れ

－

進み

A

電流

S

皮相電力

C※３

進相コンデンサ容量

周波数

DC2

アナログ入力2ch の電圧
＋

遅れ

－

進み

An※２

中性線の電流PA※３

WP+

有効電力量(消費)WS+

皮相電力量(消費)WQi+

無効電力量(遅れ)WP-

有効電力量(回生)WS-

皮相電力量(回生)WQc+

無効電力量(進み)THD

電圧／電流歪み率
Pst

短期電圧フリッカ

Plt

長期電圧フリッカ

Pst

1 分電圧フリッカ(1min)

位相角※１

Ｗ画面：結線３Ｐ４Ｗを選択した場合に、V と VL 表示を[カスタマイズ]できます。

※２

Ｗ画面：An は結線３Ｐ４Ｗを選択した場合のみ表示します。

※３

Ｗ画面：PA と C は[カスタマイズ]で表示できます。

3.4

バックライトとコントラストの調整
バックライトが点灯している状態で、LCD キーを 2 秒以上長押しするとバックライトの輝度と、表示のコントラストを調整する、スライドバーを画面上に表示します。調整するにはカーソルキーを操作し、スライドバーを移動して調整してください。調整後に ENTER キーを押すと決定します。スライドバーに関係なく調整を終了する場合には ESC キーか LCD キーを再度押して終了してください。
輝度調整
バックライトの明るさを 11 段階に調整できます。

コントラスト調整
表示のコントラストを 11 段階に調整できます。

KEW6315- 22 -

瞬時値／／積算値／デマンド

KEW6315

3.5 面表示と画
画面構成
瞬
瞬時値／積算
算値／デマ
マンド
画面の切り
り替え
を押すごとに表示画面がが切り替わりま
ます。
W（瞬時
時値）

Wh（積
積算値）

デマンド

カスタマイズ
表示していいる測定項目を変更
更します。

トレンド
測定値の変動をトレンドグラフにに表示します。

拡大
大表示
拡大したい測定項目をを選択して表示し
します。
分
4 分割

8 分割

- 23 -

KEW6315

KEW63115

ベクトル

ベ
ベクトル
画面の切り替え

結線確
確認
確認項目
目に対して判定した
た結果を表示しま
ます。

結線図
図
設定
定している結線図をを表示します。

波
波形
の切り替え
表示項目の

KEW
W6315

- 24 -

高調波

KEW6315高調波
高表示項目の切り替え
電 圧・リニア・全体表示
電 流

リスト・含有率

電 力

対 数

拡大表示

位相角

実効値

電 流

電 力

- 25 -

KEW6315

KEW63115

電源品質

電
電源品質
表示項目の切り替え
フリッカ

イベン
ント

設
設定
表示項目
目の切り替え左右右のカーソルキ
キーで移動しま
ます。

KEW
W6315

- 26 -

入力端子プレートを貼る

KEW6315

4章 測定前の準備
4.1 購入後はじめにすること
入力端子プレートを貼る
付属の入力端子プレート６タイプの内から、配線色にあった入力端子プレートを１枚はがします。
プレートを貼る前に入力端子部分のゴミやホコリをふき取り、入力端子プレートの向きを確認して、下図の位置へ貼り付けてください。
TYPE1

TYPE4

TYPE2

TYPE5

TYPE3

TYPE6

入力端子プレートを貼り付けます
入力端子プレート

VN

V1/A1

V2/A2

V3/A3

A4

TYPE 1

青

赤

緑

黒

黄

TYPE 2

青

茶

黒

灰

黄

TYPE 3

黒

黄

緑

赤

白

TYPE 4

青

黒

赤

白

黄

TYPE 5

白

黒

赤

青

黄

TYPE 6

黒

赤

黄

青

白

- 27 -

KEW6315

KEW6315

識別マーカーを取り付ける

識別マーカーを取り付ける
電圧用測定コードとクランプセンサの両端に、入力端子と同じ色の識別マーカーを取り付けます。
識別マーカーは 8 色（赤・青・黄・緑・茶・灰・黒・白）各 4 本の計 32 本あります。

クランプセンサの両端に取り付けます

識別マーカー（計 32 本）

0電圧用測定コードの両端に取り付けます

KEW6315- 28 -

電池の使用4.2

KEW6315

電源について電池の使用
本製品は、AC 電源／電池駆動の 2 電源方式です。
停電などが原因で AC 電源の供給が止まった場合でも、電源の供給を電池に切り換えて測定を行います。
電池駆動では、単３形アルカリ乾電池（LR6）と単３形ニッケル水素電池(Ni-MH)の使用が可能です。
充電式電池への充電は、ご使用になる電池メーカの充電器をご使用ください。本体での充電は行えません。
※

単３形アルカリ乾電池（ＬＲ６）は付属品となっています。

危険●測定中は絶対に電池交換を行わないでください。
●銘柄や種類の違う電池を混ぜて使用しないでください。
●電池駆動時の電源コネクタは絶縁されていますが、絶対にさわらないでください。

警告●電池の交換の際には電源コード、電圧用測定コードおよびクランプセンサを本体からはずし、電源を OFF にしてください。

注意●電池は古いものと混ぜて使用しないでください。
●電池の極性をまちがえないよう、ケース内の彫刻の向きに合わせて入れてください。

本製品のご購入時は電池が内蔵されておりません。必ず付属の電池をセットしてください。
電源 OFF の状態でも電池を消費しますので、長時間使用されない場合は電池を抜き取って保管してください。
AC 電源から供給がある場合は、電池から電源の供給はされません。
本体に電池が内蔵されていない状態で AC 電源の供給が止まった場合、本体の電源が切れ、記録中のデータが失われる可能性があります。充分注意してください。

- 29 -

KEW6315

KEW6315

画面の表示／電
電池の残量

画
画面の表示
示／電池の残残量
画
画面右上に表示しした電源アイコンがが、電源の状態にによって下記のよううに変化します。
ＡＣ電源駆動

4 分割

電池駆
駆動

電池の残量

画面の表示

単３形アルカ
カリ乾電池の場合
合は約３時間測定定可能です。
単３形ニッケ
ケル水素電池(19000ｍA/h)の場合は約４．５時間測定可能能です。※LCD を OFFF にした状態での参参考値です。
通常測定※

ニッケル水素電
電池は満充電時のの電圧が、ア

ルカリ乾電池
池に比べて低いたため、左の表示以以上にならな
い場合があります。
測定は続行行しますが、保存はは終了します。
(終了までのの測定データは保存存されます。)

KEW
W6315- 30 -

乾電池のセット方法

KEW6315乾電池のセット方法
以下の手順で乾電池をセットします。
ネジ

電池蓋

単３形アルカリ乾電池：ＬＲ６

１

電源コード、電圧用測定コードおよびクランプセンサを本体からはずし、電源を切ります。

２

本体裏側のネジ２個を緩めて電池蓋をはずします。

３

電池を全て取りはずします。

４

正しい極性で単３形アルカリ乾電池：ＬＲ６を必ず６本セットします。

５

電池蓋を取り付けて、ネジ２個を締めます。

電源コードの接続
必ず確認してください。

危険●電源コードは付属の専用コードを使用してください。
●電源コードは必ずコンセントに接続してください。また AC240V より高い電位のある場所には絶対に接続しないでください。（付属の電源コード MODEL7169 の最大規格電圧は AC125V です。）

警告
●本体の電源が OFF になっていることを確認してから接続してください。
●接続は必ず先に本体側から行い、根元まで確実に差し込んでください。
●使用しているうちに亀裂が生じたり、金属部分が露出したときは、直ちに使用を中止してください。
●本製品を使用しない場合は、電源コードをコンセントから抜いてください。
●電源コードのプラグをコンセントから抜くときは必ず差し込みプラグを持って抜いてください。

- 31 -

KEW6315

KEW6315

電
電源の定格

以
以下の手順で電電源コードを接続
続します。

※

１

本体に電
電源が入っていなないことを確認しま
ます。

２

本体の電
電源コネクタに、付付属の電源コードドを接続します。

３

※

コンセン
ントに接続します。
。

コンセントに接続
続してから約 2 秒間
間は本体機能確認
認のためを押押しても電源が入入りません。

その後は通常通り使用できます。

電
電源の定格
電
電源の定格は下下表のとおりです
す。

KEW
W6315

定格電
電源電圧

1000～240V AC（±
±10％）

定格電源
源周波数

45～
～65Hz

最大消
消費電力

7V
VA max- 32 -

4.3 ＳＤカードの挿入／取り出し方法

4.3

KEW6315

ＳＤカードの挿入／取り出し方法
必ず確認してください。

注意●「挿入方法」に従って正しい向きでＳＤカードを本体に挿入してください。間違った方向で挿入するとＳＤカードまたは、本体が破損する恐れがあります。
●ＳＤカードを挿入／取り出すときは必ず、ＳＤカードへアクセス中でないことを確認してください。

（アクセス中はが点滅します。）ＳＤカードへアクセス中に取り出しを行うと、保存されたデータや本体を破損する恐れがあります。
●記録中（が点滅します。）にはＳＤカードを取り出さないでください。保存されたデータや本体を破損する恐れがあります。取り出す場合には、必ず記録を終了して「記録を停止しました」のメッセージが消えた後に取り出してください。

注記●新しいＳＤカードを使用する場合には、必ず本体にてフォーマットを行ってください。ＰＣでフォーマットを行うと正しく記録できない場合があります。詳細は「フォーマットする」８６頁を参照してください。
●頻繁に使用したＳＤカードや長期間使用したＳＤカードは、フラッシュメモリの寿命からデータを記録できなくなる場合があります。記録できないＳＤカードは新しいものと取り換えてください。
●SD カードの記録データは、事故や故障によって、消失または変化してしまうことがあります。記録したデータは定期的にバックアップを取って保存してください。なお、データが消失または変化した場合の損害につきましては、原因および、内容にかかわらず保証できません。あらかじめご了承ください。

- 33 -

KEW6315

KEW6315

挿入方法

挿入方法
１

コネクタカバーを開きます。

２

ＳＤカードの向きを確認し表面を上にした状態で図の矢印の方向へコネクタの奥まで押し込みます。

３

コネクタカバーを閉じます。特に必要のない場合には必ずコネクタカバーを閉めて使用してください。

取り出し方法１

コネクタカバーを開きます。

２

ＳＤカードを奥に押すとカードを取り出せる状態になります。

３

ＳＤカードをつまんで引き抜いてください。

４

コネクタカバーを閉じます。特に必要のない場合には必ずコネクタカバーを閉めて使用してください。

SD カードコネクタカバー

KEW6315- 34 -

4.4 電圧用測定コードとクランプセンサの接続

4.4

KEW6315

電圧用測定コードとクランプセンサの接続必ず確認してください。

危険●電圧用測定コードは付属の専用コードを使用してください。
●クランプセンサは本製品専用のものを使用してください。また、測定電流と定格が合っているか必ず確認してください。
●測定に必要のない電圧用測定コードおよびクランプセンサは絶対に接続しないでください。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●測定の際は指先等が、保護用フィンガガードを越えることのないよう充分注意してください。

警告●本体の電源が OFF になっていることを確認してから接続してください。
●接続は必ず先に本体側から行い、根元まで確実に差し込んでください。
●測定コードのコード内部から金属部分または外装被覆と異なる色が露出したときは、直ちに使用を中止してください。

以下の手順で電圧用測定コードおよびクランプセンサを接続します。

１

本体に電源が入っていないことを確認します。

２

本体の交流電圧入力端子へ測定に必要な電圧用測定コードを接続します。

３

本体の電流入力端子へ測定に必要なクランプセンサを接続します。このときクランプセンサの出力端子の矢印と本体の電流入力端子の矢印が向き合うように接続してください。

矢印の向きに合わせる

電圧用測定コードおよびクランプセンサの使用数、接続場所は結線方式によって異なります。
「結線図」５０頁
を参照してください。

- 35 -

KEW6315

KEW6315

4.5

初期
期画面表示

電源の投
投入
初
初期画面表
示
画
画面が表示されれるまで電源キ
キーを押し続けると電源が入ります。電源をを切る場合は、2 秒
以以上電源キーをを押し続けてください。
本
本体の電源を入入れると、下記記のように画面がが表示されますす。

１

本体の電源をを入れると、モデル名／バージョン
ン画面を表示します。
正しく起動ししない場合には、直直ちに使用を中止
止し「11 章 故障かかなと思ったら」１
１５７頁を参照しててください。

２

前回、電源をＯＦＦした時の画面を表示し
します。

注
注意表示
前
前回と異なるクラ
ランプセンサを接接続している場合
合に、今回接続したクランプセン
ンサが 5 秒間表表示されます
す。但し、表示されれたクランプセン
ンサは、自動でで設定されませんん。クランプセン
ンサの接続設定を変更した
い
い場合には、

から「センンサ識別」を行う
うか、直接クラン
ンプセンサの種類
類を変更してくだ
ださい。クラン
ンプセンサを接続
続していない場合
合には、前回の設定をそのまま
ま引き継ぎます。
。

KEW
W6315

- 36 -

記録の開始4.6

KEW6315

記録の手順
記録の開始
キーを押す

記録の開始方法を「かんたんスタートナビ」と「すぐに記録開始」から選択できます。「かんたんスタートナビ」では画面の指示に従って項目を順次設定することで、記録に必要な設定を完了し、簡単に記録を開始できます。ただし「かんたんスタートナビ」にて設定する項目は、結線と記録設定に関するもののみです。その他の設定を行う場合には、から設定してください。記録に必要な設定が既に完了してるか、設定を変更する必要がない場合には「すぐに記録開始」を選択することで、現在の設定に準じて直ぐに記録を開始します。測定前には必ず『安全の確認』、『測定準備』を行ってください。

「かんたんスタートナビ／すぐに記録開始」の項目へ移動- 37 -

決定キャンセル

KEW6315

KEW6315

記録の終了

記録の終了
キーを押す

記録データナンバー
記録開始方法

記録項目

記録に関する情報の表示と、記録を終了できます。

画面表示項目
記録データナンバー
経過時間

記録データの識別番号を表示します。測定データを保存するフォルダ名としても使用します。
記録開始からの経過時間を表示します。
手動記録開始方法「記録開始日時」を表示します。

連続記録「記録開始日時」／「記録終了日時」を表示します。

時間帯指定「記録開始日時」／「記録期間」／「記録時間帯」を表示します。

記録先

測定データを記録している保存対象を表示します。

記録項目

記録している測定項目を表示します。

「キャンセル／ストップする」の項目へ移動

KEW6315

決定- 38 -

キャンセル

「かんんたんスタートナ
ナビ」で記録を開開始する

KEW6315「かかんたんス
スタートナビ」で記録を開開始する
記
記録項目の設
設定

結結線の設定

結線接続確認

測定環境テススト①記録したたい項目を選択
択してください。
※項目が多多いほどファイルサイズが大きくな
なるため、
記録で
できる時間が短くな
なります。

３７頁
頁参照

②測定
定対象の結線をを選択してくだださい。
※設定を誤りますと正正しい測定ができまませんのでご注
注意ください。

４４１頁参照③③測定対象の回路に接続してください。
※付属取扱説明明書に記載されてております注意事項を確認いいただき、安全に作業してくださ
い。

２７
７頁参照

④⑤測定
定環境のテストトを実施してくだださい。
※本器のの自己診断，結線
線状態の確認，接接続しているセン サの識別を行います。
※正しい状態で記録するたためにテストの実
実施をお勧めいた
たします。テ
テスト時間は約 100 秒です。

４２頁参照

- 39 -

KEW6315

KEW6315「か
かんたんスタートトナビ」で記録を
を開始する

記録インタ
ターバル

記録開始方法
法

記録開
開始の確認

記録開
開始

⑥記録間隔を選択しててください。
※間隔
隔が短いとファイル
ルサイズが大きくくなるため長期の記録ができません
ん。

７６頁参照

⑦
⑦⑧⑨記録方法
法を選択してく
ください。
４５頁参照

⑩
⑩準備完了。記記録を開始しますす。
記
記録を開始するとと画面上に REC を表示し、Logging
L が緑色に点灯
LED
灯します。
強
強制的に記録を止止めたいときには
は

を押して
て、

L
LCD
画面の案内にしたがって操作作してください。

KEW
W6315

- 40 -

「かんたんスタートナビ」で記録を開始する

KEW6315②結線の設定
以下の中から結線を選択できます。
L

L

負荷
①
N
1P2W
L
負荷
N
②
1P2W
L
負荷
N
③
1P2W
L
負荷
N
④
1P2W

電源 N

V1VN

A1 A2

1P2W×1

電源

L1

L1

N

負荷
①
1P3W
L2
N

L2
1P2W×2

負荷
②
1P3W
L2

1P2W×3N

1P2W×4
V1 V2
VN

L1(R)
負荷
L2(S) ①
L3(T) 3P3W

電源 L2(S)L3(T)

1P3W×2

L1

A3 A4

L1(R)

1P3W×1

3P3W×1

A1 A2

A3 A4

L1(R)

L1(R)

電源 L2(S)

L2(S)

L3(T)

L3(T)

負荷
3P3W
3P3W3A

3P3W×2
L1(R)負荷②
L3(T) 3P3W

L2(S)

V1 V2
VN

A1 A2

V1 V2

V3

A1 A2

A3

電流クランプセンサの向き

A3 A4

負 荷

電源

L1(R)

L1(R)

L2(S)

L2(S)

L3(T)

L3(T)

N

N

V1 V2

A1 A2
V3
VN

3P4W
負荷
3P4W

矢印：負荷
電 源

A3

※ 逆にクランプすると有効電力(Ｐ)の値の符号が逆転します。
※測定に使用する電流クランプセンサは、同じ種類を使用してください。

- 41 -

KEW6315

KEW6315「か
かんたんスタートトナビ」で記録をを開始する④／／⑤測定環境境テスト
画面
面の切り替え
測定環
環境テスト
テスト
トを開始する を選選択して決定した
た場合、

結線確
確認

測定環
環境をテストして、 その判定した結
結果を表
示します
す。

確認項目に
に対して判定した結結果を表示します
す。
※

力率が著し
しく悪い測定現場場では、正しい結線
線を行っていても、NG と判定することがあります。

自己診診断
OK ／ NG の判定結
結果を選択し決決定する

本体システ
テムの動作状態をを確認し判定したた結果をすることができます。
と、詳細な情報を表示す

表示します
す。

センサ
サ識別
接続された
た電流センサを自自動的に識別し最
最大レン
ジで設定し
します。

KEW
W6315

- 42 -

「かんんたんスタートナ
ナビ」で記録を開開始する KEW6315

KEW6315

NG
G 判定
結線
線確認
判定結果を
を閉じると、NG の値とベクトル線がが点滅します。全てて OK ならば参考としてて結線設定に準じじた理想的なベク
クトル図のみを画面面左下へ表示します。

結線
線確認の合格
格判定基準と原
原因
確
確認事項

合格判定基
基準

周
周波数

・V
V1 の周波数が 40～70Hz
4
で
あ
あること。

交流
流電圧入力

・交
交流電圧入力が(公称電圧×VVT)の10％以上でであること。

原因・電電圧クリップが被被測定物に確実に接続されています
すか？
・高高調波の成分が大大きくないですかか？
・電電圧クリップが被被測定物に確実に接続されています
すか？
・電電圧用測定コードが本製品の交流
流電圧入力端子子に正常に挿挿入されています
すか？

電圧
圧バランス

・交流電圧入力が基準電圧((V1)の ± 20% 以 内 で あ る こ

・電電圧クリップが被被測定物に確実に接続されています
すか？

と
と。

・電
電圧用測定コード
ドが交流電圧入力
力端子に正常に挿挿入されて単単相結線では判定
定しない。

い
いますか？

・交
交流電圧入力の位
位相が基準

・電電圧コードの接続
続先が間違っていいませんか？

値
値(正しいベクトルル)の±10°(接続するチャンネ
ネルを間違っていいませんか？)

※

電
電圧位相

・測
測定ラインの結線線方式と設定が合合っていますか？。
。

以以内であること。
電
電流入力

・電流入力が(電流レンジ×・ク
クランプセンサがが本製品の交流電
電力入力端子に確
確実に挿入

C
CT)の
5％以上、110%以下でさされていますか？
？

あ
あること。

・電
電流レンジの設定
定が入力レベルにに対して大きすぎ
ぎたり、小さす
すぎたりしていませ
せんか？

電
電流位相

・各
各 CH の力率(PFF､絶対値)が0 以上あること。
0.5。
・各各 CH の有効電力力（P）が正のクランプセンサのの電流方向マークは『電源→負荷』』の方向を・ク
向
向いていますか？
？
・ク
クランプセンサの接続先は間違っていませんか？

数
数であること。

- 43 -

KEW6315

KEW6315「か
かんたんスタートトナビ」で記録を
を開始する

自己診断
頻繁
繁に NG が表示さされる場合には、本体が故障していいる可能性がありります。直ちに使用用を中止し「１１章章 故障かなと思思ったら」１５７頁
頁を参照してください。

セン
ンサ識別
NG と判断された場合
合には、電流クラ
ランプセンサの種別別を赤字で表示しします。

識別
別が NG となる
る原因
確認事項

原因
因

電流
流クランプセンサのの・異なる種類
類の電流クランプ
プセンサを各ｃｈにに接続していませせんか？測定に使使用する電種類？？？（識別不能））流クランプ
プセンサは、同じ種種類を使用してくだださい。
・電流クランンプセンサが本体体に確実に接続されれていますか？
・故障かなとと思ったら？
NG と識識別された電流クランプセンサの接接続ｃｈを正しく識別別されているｃｈと
と変更して
再テスト
トを行ってみてくだ
ださい。この時、前前回と同じｃｈが NG と識別された場
場合には、
本体が故
故障している可能能性があります。前前回 NG と識別さされた電流クランプ
プセンサを
接続して
ているｃｈが、NG と識別された場合
合には、電流クランプセンサが故障
障している
可能性が
があります。故障障が疑われる場合
合は、直ちに使用をを中止し「１１章 故障かなと思った
たら」１５７頁を参照
照してください。

KEW
W6315

- 44 -

「かんんたんスタートナ
ナビ」で記録を開開始する

KEW6315⑧／／⑨ 開始方法
法ごとの設定日付
付と日時を指定してて、記録を開始すすることができますす。

⑧ 日時を指定しして予約する設定
定した開始日時かから終了日時までの間、インターバ
バルごとに記録を行行います。

『例例』 表示のようにに設定した場合にには、次の期間記録
録を行います。
２０１３年８月２２日の８時～２０１３年８月７日の１８時⑨ 繰り返し記録
録を予約する

定した時間帯の間
間のみインターバル
ルごとに記録し、ここれを設定した期
期間繰り返します。
。
設定『例例』 表示のように設設定した場合にはは、次の（ⅰ）～（ⅷⅷ）の時間帯に記記録を行います。18 時から翌日の 8 時までの間は記録しませせん。
（ⅰ）２０１３年８月１日の８時～１８時（ⅱ）２０１３年８月２日の８時～１８時（ⅲ）２０１３年８月３日の８時～１８時（ⅳ）２０１３年８月４日の８時～１８時（ⅴ）２０１３年８月５日の８時～１８時（ⅵ）２０１３年８月６日の８時～１８時（ⅶ）２０１３年８月７日の８時～１８時（ⅷ）２０１３年８月８日の８時～１８時- 45 -

KEW6315

KEW6315「か
かんたんスタートトナビ」で記録を
を開始する

項
項目選択
本製
製品は、
カ
カーソルキーで選
選択し、ENT
TER キーで決定、
ESC キー
ーで決定せずに元元の設定に戻る操作を基本とししています。スタートナビの入力
力操作を例に、詳詳細を説明します
す。他の表示画面
面での入力
作も、ほぼ同様のの操作方法です。
操作

青い字で表示している項目((未選択)、または青い背景に白抜抜き文字の項目(選択中中)がカーソルキー
ーで移動可能な項
項目になります。記
記録スター
トの左の画
画面では上下のカ
カーソルキーで記
記録開始方法をを選択し、ENTER キーーで決定します。選
選択に関係なくス
スタートナビを終了
了する場合には ESC キーを押してくださ
キ
さい。

選択項目が
が縦横に表示してあある画面では上下
下左右のカーソル
ルキーで項
目の移動が
が可能です。結線をを選択する左の画
画面では、上下左左右のカー
ソルキーで測測定対象の結線をを選択し、ENTER
R キーで決定しまます。選択に関係なく前
前画面に戻るにはは ESC キーを押ししてください。

日付／時刻
刻のように数値を入入力するには左右右のカーソルキーーで変更したい桁を移動動し、上下のカーーソルキーで数値
値のカウントアップ
プ／ダウンを行います。
。記録日時を選択
択する左の画面ででは、左右のカーーソルキー
で日付の十の桁を選択していいます。この状態
態で上下のカーソ
ソルキーを操作すると、、十の桁を１ずつカ
カウントアップ、ままたはカウントダウ
ウンできます。変更を決
決定するには ENTER キーを押しててください。数値にに関係なく前画面に戻るには ESC キー
ーを押してください。

設
設定の注意
電流
流レンジを「AUTOO」に設定しているる場合には①記録
録項目の設定で「電力＋高調波」「電力のみ」しか選選択できません。電電源品質を記録ししたい場合には、電流レンジを固定定レンジに変更してから行ってくだださい。「かんたんスタートナビ」では、結線と記記録設定に関するるもののみ設定しします。基本設定定に含まれる公称
称電圧、公称周波数、測定定に含まれる電源
源品質イベントののしきい値、フリリッカ測定のフィィルタ係数（ランンプ）などは、事事前に設定設定する必要があります
す。これらの設定
定は、かから行ってください。また、「＋ククランプ」オプショョンクランプ

センサ
サの設定は強制的に「なし」に設定
定します。

KEW
W6315

- 46 -

5.1

設定項目

KEW6315

5章 設定
5.1

設定項目
測定を始める前にあらかじめ測定条件やデータの保存について設定をする必要があります。設定を行う
場合はメニューキーのを押して、SET UP モードにしてください。

は、下記の５つの項目に分かれています。それぞれの項目へは変更した設定は画面から別の画面に切り替えると、画面左上にで移動します。

を表示した後に有効になります。

設定を変更しても、そのまま電源をきると変更した設定が有効になりません。注意してください。

基本設定 ············ 各測定の共通項目を設定します。
測定設定 ············ 各測定独自の項目を設定します。
記録設定 ············ 保存方法の項目を設定します。
保存データ ········· 記録済みのデータや本体設定データを編集します。
その他 ················· 環境設定をします。
各

設

定

基本設定

測定設定

記録設定

保存データ

その他

（結線）

（デマンド）（記録項目）（記録データ）（環境設定）

（電圧値）

（高調波）

（記録方法）（本体設定）（本体設定）（電流値）（電源品質）（手動／連続記録／時間帯指定）（外部入力ＤＣ）（フリッカ）（周波数）（進相コンデンサ）- 47 -

KEW6315

5.2 基本設定

KEW6315

5.2

基本設定
キーを押す

KEW6315「基本設定」タブへ移動- 48 -

結線に関する設定

KEW6315

結線に関する設定「基本結線」
測定対象の結線に合わせて１０種類の結線方式の中から 1 つを選択します。
設

定

内

容

①1P2W×1⑤1P3W×1⑦3P3W×1②1P2W×2⑥1P3W×2⑧3P3W×2③1P2W×3⑨3P3W3A④1P2W×4⑩3P4W※「+クランプ」結線に使用していない電流端子は、実効値と高調波のみ測定できます。
※測定に使用する電流クランプセンサは、同じ種類を使用してください。
※「基本結線」の項目へ移動

決定は初期値です。

プルダウンメニュー表示

結線方式を選択

キャンセル「+クランプ」 オプションクランプセンサ「+クランプ」の項目へ移動

決定

プルダウンメニュー表示

オプションクランプを選択

キャンセル- 49 -

KEW6315

KEW6315

結線に関する設定

結線図「基本結線」の項目へ移動すると結線図を表示後、

KEW6315「前の結線」「結線図」にて、結線方式に準じた「結線図」を表示できます。
「次の結線」で結線方式を変更- 50 -

決定キャンセル

結線についてKEW6315

結線について必ず確認してください
危険●本製品は測定カテゴリに準じてCAT ⅣではAC300V、CAT ⅢではAC600V、CATⅡではAC1000Vより高い電圧のある回路では絶対に使用しないでください。
●電圧用測定コードとクランプセンサは本製品専用のものをご使用ください。
●クランプセンサ、電圧用測定コード、電源コードは必ず測定物や電源よりも先に本体に接続してください。
●測定コードと本体の測定カテゴリが違っている場合は低い方の測定カテゴリが優先されます。測定電圧と定格が合っているか必ず確認してください。
●測定に必要のない電圧用測定コードおよびクランプセンサは絶対に接続しないでください。
●クランプセンサは必ずブレーカーの２次側に接続してください。1 次側は電流容量が大きく危険です。
●通電中は CT の２次側が開放しないよう充分注意してください。万一開放状態になりますと、２次側に高電圧が発生して大変危険です。
●結線時に電圧用測定コードの先端の金属部で電源ラインを短絡しないように注意してください。また、先端の金属部には絶対に触れないでください。
●クランプセンサのコア先端部は被測定物を短絡しないような構造になっていますが、絶縁されていない導線を測定する場合コアで被測定物を短絡しないように注意してください。
●測定の際は指先等が、保護用フィンガガードまたはバリアを超えることのないよう充分注意してください。
保護用フィンガガードまたはバリア：操作中の感電事故を防ぐため、最低限必要な沿面および空間距離を確保するための目印です。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●コアを開いたとき、金属部で測定ラインの２線間を接触させないでください。

警告●感電、短絡事故をさけるため、接続をする場合は測定ラインの電源を切ってください。
●電圧用測定コードの先端の金属部には絶対にさわらないでください。
●測定コードのコード内部から金属部分または外装被覆と異なる色が露出したときは、直ちに使用を中止してください。

- 51 -

KEW6315

KEW6315

結線について正確に測定するために●測定ラインと本製品の結線方式の設定は正しく行ってください。
●クランプセンサは下記のように矢印を負荷側に向けてクランプしてください。
負荷側

矢印：負荷側
電源側※

逆にクランプすると有効電力（Ｐ）の値の符号が逆転します。

KEW6315- 52 -

電圧測定に関する設定

KEW6315

電圧測定に関する設定「電圧レンジ」
使用する電圧レンジを選択します。
※

電力品質国際規格 IEC61000-4-30 Class S に対応した測定を行う場合には「600V」レンジを選択してください。
設

定

内

容

600V／1000V※

プルダウンメニュー表示「電圧レンジ」の項目へ移動

決定は初期値です。

電圧レンジを選択

キャンセル

「ＶＴ比」
外部にＶＴ（変圧器）が設置されている場合に設定します。設定したＶＴ比は電圧に関する全ての測定値に対して掛け合わせます。
設

定

内

容

0.01～9999.99（1.00）※「ＶＴ比」の項目へ移動

決定

キャンセル

は初期値です。

数値入力ウインドウを表示※

ＶＴ比を入力※入力できる範囲を同時にポップアップに表示します。

- 53 -

KEW6315

KEW6315

ＶＴ／ＣＴについて※

VT／CT について※「電流測定に関する設定」中の設定項目

危険●測定カテゴリに準じてCAT ⅣではAC300V、CAT ⅢではAC600V、CATⅡではAC1000Vより高い電圧のある回路では絶対に使用しないでください。
●電源コードは必ずコンセントに接続してください。またＡＣ２４０Ｖより高い電位のある場所には絶対に接続しないでください。
●本製品は必ずＶＴ（変圧器）、ＣＴ（変流器）の２次側で使用してください。
●通電中はＣＴの２次側が開放しないよう充分注意してください。万一開放状態になりますと、２次側に高電圧が発生して大変危険です。

注意●本製品はＶＴ、ＣＴを使用した場合の確度は保証していません。ＶＴ、ＣＴを使用する場合、本製品の確度にＶＴ、ＣＴの確度、位相特性等を考慮してください。

測定ラインの電圧値や電流値が本製品の最大測定レンジを超える場合、下記のように特定ラインの電圧値、電流値に適した仕様のＶＴ、ＣＴを使用して２次側を測定することによって、１次側の値を表示させることができます。

単相２線（１系統）“1P2W-1 の例
L
電源

負荷

N

CT

VT

i

VN

V1

A1

CT の２次側が 5A 定格の場合、クランプセンサは 8128/8135（50A タイプ）を使用し 5A レンジで使用することをおすすめします。
この場合、使用するＶＴ、ＣＴの比を設定してください。

KEW6315- 54 -

電圧測定に関する設定

KEW6315「公称電圧」
測定対象から入力される公称電圧を設定します。
設

定

内

容

50V～600V（100V）※

数値入力ウインドウを表示※「公称電圧」の項目へ移動

決定は初期値です。

公称電圧を入力※

キャンセル

入力できる範囲を同時にポップアップに表示します。

既定値「公称電圧」の項目へ移動すると「既定値」にて、一般的な公称電圧をリストから選択できます。

設

定

内

容

100Ｖ／101Ｖ／110Ｖ／120Ｖ／200Ｖ／202Ｖ／208Ｖ／220Ｖ／230Ｖ／
240Ｖ／277Ｖ／346Ｖ／380Ｖ／400Ｖ／415Ｖ／480Ｖ／600Ｖ

公称電圧を選択決定キャンセル- 55 -

KEW6315

KEW6315

電流測定に関する設定

電流測定に関する設定「クランプ」 電流測定用クランプセンサ
使用するクランプセンサを選択します。測定に使用する電流クランプセンサは、同じ種類を使用してください。
「+クランプ」 にてオプションクランプセンサを選択していた場合には4ｃｈに限り、測定結線対象に接続するクランプセンサと異なる種類のクランプセンサを選択できます。プルダウンメニューを表示している状態で選択する
クランプセンサへ移動すると、各クランプセンサの定格電流と被測定導体径をポップアップに表示します。
設

定

内

容

8128/8135：5／50A／AUTO
8127：10／100A／AUTO
8126：20／200A／AUTO
電力測定用クランプセンサ

8125：50／500A／AUTO
8124/8130：100／1000A／AUTO
8129：300／1000／3000A
8133：300／3000A／AUTO
8141：
8142：

500mA／AUTO

8143：

リーク電流測定用クランプセンサ

8146：
8147：

1／10A／AUTO

8148：
※

「クランプ」の項目へ移動

決定KEW6315は初期値です。

プルダウンメニュー表示

キャンセル- 56 -

クランプセンサを選択

電流測定に関する設定

KEW6315「電流レンジ」
使用する電流レンジを選択します。電源品質に関するイベントを「記録する」と設定している場合には「AUTO※」を選択できません。電流レンジを自動で切り替えた場合は、電源品質に関するイベントを「記録しない」設定にしてください。
電源品質に関するイベント設定の詳細は「ＶＴ/ＣＴについて」５４頁にて説明しております。
※「AUTO」レンジでは電力品質国際規格 IEC61000-4-30 Class S に対応した測定を行えません。

「電流レンジ」の項目へ移動

決定

プルダウンメニュー表示

レンジを選択

キャンセル

「ＣＴ比」
外部にＣＴ（変流器）が設置されている場合に設定します。設定したＣＴ比は電流に関する全ての測定値に対して掛け合わせます。ＣＴに関する詳細は「ＶＴ/ＣＴについて」５４頁を参照してください。
設

定

内

容

0.01～9999.99（1.00）※「ＣＴ比」の項目へ移動

決定

キャンセル

は初期値です。

数値入力ウインドウを表示※

ＣＴ比を入力※入力できる範囲を同時にポップアップに表示します。

センサ識別「センサ識別」にて、自動で接続している電流クランプセンサを設定することができます。ただし結線方式にて測定対象となる各クランプセンサ接続ｃｈに種類の異なるクランプセンサを接続していた場合または、接続している
クランプセンサの種類を識別できなかった場合には、エラーメッセージをポップアップに表示後、「クランプ」「電流レンジ」「ＣＴ比」をクリアします。「センサ識別」に関する詳細は「センサ識別」４４頁を参照してください。

- 57 -

KEW6315

KEW6315

外部入力端子／基準周波数に関する設定

外部入力端子／基準周波数に関する設定「ＤＣレンジ」
入力する直流電圧信号に準じてＤＣレンジを選択します。
設

定

内

容

100mV／1000mV／10V※「ＤＣレンジ」の項目へ移動

決定は初期値です。

プルダウンメニュー表示

レンジを選択

キャンセル

「周波数」
測定対象の公称周波数を設定します。停電時のように電圧の周波数が特定できない状態では、設定した公称周波数を基準にして測定を行います。
設

定

内

容

50Hz／60Hz※「公称周波数」の項目へ移動

決定KEW6315は初期値です。

プルダウンメニュー表示

キャンセル- 58 -

周波数を選択5.3 測定設定

5.3

KEW6315

測定設定
キーを押す

「測定設定」タブへ移動

デマンドに関する設定「測定周期」
デマンド測定なし、または記録測定時間内の 1 回のデマンド測定時間を選択します。デマンド測定を開始すると「測定周期」ごとにデマンド値を決定し記録していきます。また、デマンド測定を行う場合、インターバルは下記の時間のみしか設定できません。この他のインターバルを設定している状態で「測定周期」を設定すると自動的にインターバルが「測定周期」と同じ設定に変わります。注意してください。
設定可能なインターバル：1 秒､2 秒､5 秒､10 秒､15 秒､20 秒､30 秒､1 分､2 分､5 分､10 分､※15 分､※30 分※「測定周期」より長いインターバルは設定できません。
設

定

内

容

測定なし／10 分／15 分／30 分※「測定周期」の項目へ移動

決定は初期値です。

プルダウンメニュー表示

時間を選択

キャンセル

- 59 -

KEW6315

KEW6315

デマンドに関する設定「目標値」
デマンド測定の目標値を設定します。
設

定

内

容

0.001mW～999.9TW(100.0kW)※「目標値」の項目へ移動

決定

キャンセル

は初期値です。

数値入力ウインドウを表示※

目標値を入力※入力できる範囲を同時にポップアップに表示します。

目標値の数値入力ウインドウを表示している状態で、次の操作が有効になります。デマンドの目標値には
有効電力または、皮相電力を設定することが可能です。有効電力と皮相電力の切り替えは

「VA」「
/ W」

にて電力を示す単位※を切り替えることで変更できます。数値を示す単位※を変更するには

にて単位へ

移動し、

で変更してください。

では、小数点位置を左右へ移動することができます。

※

皮相電力：mVA､_VA､kVA､MVA､GVA､TVA ／ 有効電力：mW､_W､kW､MW､GW､TW

KEW6315- 60 -

デマンドに関する設定

KEW6315「判定周期」
デマンド測定中に予測値が目標値を上回った場合にブザーで警告する周期（時間）を選択します。判定周期には測定周期よりも長い時間は設定できません。測定周期の設定によって選択できる判定周期は下記のようになります。
設定「測定周期」「判定周期」選択できる周期（時間）10 分／15 分

1 分／2 分／5 分

30 分

1 分／2 分／5 分／10 分／15 分※「判定周期」の項目へ移動

決定

キャンセル

は初期値です。

数値入力ウインドウを表示※

時間を選択

※

入力できる範囲を同時にポップアップに表示します。

- 61 -

KEW6315

KEW6315

デマンドについてデマンドについて
通常 30 分間（「測定周期」）の平均電力をデマンドといいます。工場などの契約電力は、このデマンドによって決定されるため、デマンドの値が大きければ電気料金が上がる可能性があります。デマンドを抑える方法を下記を例に説明します。仮に最大デマンドを 500kW（「目標値」）に抑えたいとした場合、測定周期 1 のデマンドは 500kW のため問題ありませんが、測定周期 2 では前半の 15 分間で 600kW の電力を消費しています。この場合、後半の 15 分間を 400kW に抑えることで、測定周期 2 も測定周期 1 と同じ 500kW にデマンドを抑えることができます。なお、測定周期 2 の電力消費が前半の 15 分間で 1000kW、後半の 15 分間で無負荷（0kW）でも同じ 500kW となります。また、「判定周期」を 15 分と設定していれば、測定周期 2 の前半 15 分の位置でブザー音が鳴ります。
(kW)
600kW

600
500

500kW

500kW
400kW

0

15

15
（ブザー）

30

測定周期 1
判定周期

KEW6315

測定周期 2

判定周期

判定周期- 62 -

判定周期

30(分)

高調波解析に関する設定

KEW6315

高調波解析に関する設定「ＴＨＤ算出方法」
ＴＨＤ：総合高調波歪み率の算出方法を選択します。基本波を基準とした総合高調波歪み率が「ＴＨＤ－Ｆ」、全実
効値を基準とした総合高調波歪み率が「ＴＨＤ－Ｒ」です。
設

定

内

容

THD-F(基本波を基準)／THD-R(全実効値を基準)※

プルダウンメニュー表示「ＴＨＤ算出方法」の項目へ移動

決定は初期値です。

算出方法を選択キャンセル「ＭＡＸホールド」
ＭＡＸホールド「ON」を選択すると測定開始からの最大含有率を高調波のグラフ上にマークで表示します。
設

定

内

容

ON／OFF※「ＭＡＸホールド」の項目へ移動

決定は初期値です。

プルダウンメニュー表示

ＯＮ／ＯＦＦを選択

キャンセル- 63 -

KEW6315

KEW6315

高調波解析に関する設定「許容値範囲を編集する」
高調波に関する電磁両立性（ＥＭＣ）レベルの許容値範囲（含有率）を次数ごとに設定します。設定した許容値範囲は高調波のグラフ上にバーグラフで表示します。
設

定

内

容

規定値／カスタマイズ（電圧／電流）※「許容値範囲を編集する」へ移動

設定したい次数へ移動

決定

キャンセル

は初期値です。

許容値範囲入力画面を表示

数値入力ウインドウを表示※

許容値を入力※入力できる範囲を同時にポップアップに表示します。

許容値範囲入力画面を表示している状態で、次の操作が有効になります。各次数の許容値には初期値として電磁両立性（EMC）国際規格 IEC 61000-2-4産業環境 Class 3 の値が設定してあります。許容値を変更していた場合であっても「既定値」で初期値に戻すことができます。

「電流／電圧[%]」で高調波電流の許容値入力画面と高調波電圧の許容値入力画面とを切り替えます。入力を終了して「測定設定」の画面に戻るには

KEW6315「戻る」を操作してください。

- 64 -

電源品質（イベント）しきい値設定

KEW6315

電源品質（イベント）しきい値設定「ＯＦＦ／ＯＮ」で、それぞれのイベント「しきい値」を入力できる状態にします。「しきい値」を設定していてもＯＦＦの状態ではイベントを判定しません。再度ＯＮにすると前回設定した「しきい値」を表示します。

しきい値設定の注意「スウェル」「ディップ」「瞬停」のしきい値は、公称電圧に対してパーセンテージで設定するため、公称電圧の設定を変更すると、しきい値の電圧が変わります。「トランジェント」は公称電圧を変更した時点で変更後の公称電圧の３倍のピーク電圧（３００％）を初期値として設定します。また「インラッシュカレント」では電流レンジに対してパーセン
テージで設定するため、電流レンジの設定を変更すると、しきい値の電流が変わります。注意してください。

「ヒステリシス」
イベント判定をしない測定領域を、しきい値に対する％で設定します。適切なヒステリシスを設定することで、しきい値付近の電圧変動／電流変動によって起こる必要以上のイベント判定を防ぐことができます。
設

定

内

容

しきい値に対して 1～10%（5%）※

「ヒステリシス」の項目へ移動

決定

キャンセル

は初期値です。

数値入力ウインドウを表示※

ヒステリシス[%]を入力※入力できる範囲を同時にポップアップに表示します。

- 65 -

KEW6315

KEW6315

電源品質（イベント）しきい値設定「トランジェント」 オーバー電圧（インパルス）
トランジェントのしきい値を瞬間の電圧値で設定します。設定できる範囲はＶＴ比の設定によって設定範囲×VT 比に変わります。
設

定内容±50～±2200Vpeak（公称電圧の 300%）※「トランジェント」の項目へ移動

決定

キャンセル

は初期値です。

数値入力ウインドウを表示※

電圧値を入力※入力できる範囲を同時にポップアップに表示します。

トランジェント検出例
詳細は「イベントの発生状況を表示する」１１６頁を参照してください。
しきい値（電圧値）
電圧波形

KEW6315- 66 -

電源品質（イベント）しきい値設定

KEW6315「スウェル」 瞬間の電圧上昇
スウェルのしきい値（1 周期の電圧実効値）を公称電圧に対する％で設定します。設定できる範囲はＶＴ比の設定によって設定範囲×VT 比に変わります。このしきい値にはヒステリシスが有効になります。
設

定

内

容

公称電圧に対して 100～200%（110%）※は初期値です。

数値入力ウインドウを表示※「スウェル」の項目へ移動

公称電圧の％を入力※入力できる範囲を同時にポップアップに表示します。

「インラッシュカレント」 瞬間の電流上昇
インラッシュカレントのしきい値（1 周期の電流実効値）を電流レンジの最大値に対する％で設定します。設定できる範囲はＣＴ比の設定によって設定範囲×ＣT 比に変わります。このしきい値にはヒステリシスが有効になります。
設

定

内

容

電流レンジに対して 0～110%（100%）※は初期値です。

数値入力ウインドウを表示※「インラッシュカレント」の項目へ移動

電流レンジ最大値に対するの％を入力※入力できる範囲を同時にポップアップに表示します。

スウェル／インラッシュカレントの検出例
詳細は「イベントの発生状況を表示する」１１６頁を参照してください。
実効値②

実効値④

実効値⑥

実効値⑧

電圧／電流波形

実効値①

実効値③

実効値⑤

実効値⑦

実効値⑨

⑤

しきい値
実効値波形

①

②

④

③

⑥

ヒステリシス⑦- 67 -⑧⑨

KEW6315

KEW6315

電源品質（イベント）しきい値設定「ディップ」 瞬間の電圧降下
ディップのしきい値（1 周期の電圧実効値）を公称電圧に対する％で設定します。設定できる範囲はＶＴ比の設定によって設定範囲×VT 比に変わります。このしきい値にはヒステリシスが有効になります。
設

定

内

容

公称電圧に対して 0～100%（90%）※

は初期値です。

数値入力ウインドウを表示※「ディップ」の項目へ移動

公称電圧の％を入力※入力できる範囲を同時にポップアップに表示します。

「瞬停」 瞬間の電力供給停止状態
瞬停のしきい値（1 周期の電圧実効値）を公称電圧に対する％で設定します。設定できる範囲はＶＴ比の設定によって設定範囲×VT 比に変わります。このしきい値にはヒステリシスが有効になります。10V 以下の電圧実効値にてイベントを検出する場合には、必ず瞬停イベントを有効にしてください。同様のしきい値をディップに対して設定しても正しく検出できない可能性があります。
設

定

内

容

公称電圧に対して 0～100%（10%）※

は初期値です。

数値入力ウインドウを表示※「瞬停」の項目へ移動

公称電圧の％を入力※入力できる範囲を同時にポップアップに表示します。

ディップ／瞬停の検出例
詳細は「イベントの発生状況を表示する」１１６頁を参照してください。
実効値②

実効値④

実効値⑥

実効値⑧

電圧波形

実効値①

実効値③

①

③

②

実効値波形

④

しきい値

KEW6315実効値⑤⑥⑤- 68 -

実効値⑦

実効値⑨

⑦

⑨

⑧

ヒステリシスフリッカのフィルタ係数に関する設定

KEW6315

フリッカのフィルタ係数に関する設定「フィルタ係数」
公称電圧に準じてフィルタ係数を選択します。フリッカを正しく測定すためには、実際の測定対象に合わせて公称
電圧値、公称周波数、フィルタ係数を正しく設定する必要があります。可能ならば公称電圧値とフィルタ係数は同じ
電圧を設定してください。
設

定

内

容

230V／220V／120V／100V※「フィルタ係数」の項目へ移動

決定は初期値です。

プルダウンメニュー表示

フィルタ係数を選択

キャンセル- 69 -

KEW6315

KEW6315

進相コンデンサの目標力率値に関する設定

進相コンデンサの目標力率値に関する設定「目標力率値」
進相コンデンサ設置後の力率値を設定します。電源に接続されている設備がモーター等の誘導性負荷の場合は電圧よりも電流の位相が遅れるために力率が悪くなります。高圧の受電設備では、これを改善するために通常進相コンデンサが設置されています。低圧電力、高圧電力、業務用電力等の契約では力率を改善すると電気料金を節約でできる場合があります。
設

定

内

容

0.5～1（1.000）※

は初期値です。

数値入力ウインドウを表示※「目標力率値」の項目へ移動※入力できる範囲を同時にポップアップに表示します。

KEW6315- 70 -

改善後の力率を入力

5.4 記録設定

5.4

KEW6315

記録設定
キーを押す

「記録設定」タブへ移動- 71 -

KEW6315

KEW6315

記録項目に関する設定

記録項目に関する設定

測定データをＳＤカードまたは内部メモリに記録できる時間は記録する項目数とインターバルにて決まります。必要のない記録項目は「記録しない」に設定することで記録時間を長くできます。詳細は「保存可能な時間について」
７６頁を参照してください。

「電力関連」
電力に関する測定項目は常に記録します。記録していることを認識できるように「電力関連：記録する」を表示しています。そのため、この項目を選択することはできません。

「高調波」
電圧、電流、電力に関する高調波測定データを記録するまたは、記録しないを選択します。
設

定

内

容

記録する／記録しない※は初期値です。

「イベント」
電源品質にてイベントが発生した場合に詳細データを記録するまたは、記録しないを選択します。電流レンジを「ＡＵＴＯ※」と設定している場合には「記録する」を選択できません。電流レンジを固定レンジに変更してから「記録する」を選択してください。
※「AUTO」レンジでは電力品質国際規格 IEC61000-4-30 Class S に対応した測定を行えません。
設

定

内

容

記録する／記録しない※「高調波／イベント」の項目へ移動

決定KEW6315は初期値です。

プルダウンメニュー表示

キャンセル- 72 -

記録する／しないを選択

記録項目について

KEW6315

記録項目について記録方法に応じて全測定 CH の以下のデータが記録されます。
記録項目は記録方法、結線方式によって異なります。
記録設定
記録ファイル

記録項目
電力

＋高調波

＋イベント

●

●

●

電圧実効値（線間／相）
電流実効値
有効電力
無効電力
皮相電力
力率
周波数
中性線電流（３P４W）
電圧／電流位相角（１次）
アナログ入力電圧１、２CH
電圧／電流不平衡率
電力測定データ

１分電圧フリッカ
短期電圧フリッカ（Pst）
長期電圧フリッカ(Plt)
進相コンデンサ容量
有効電力量(消費／回生)
無効電力(消費)遅れ／進み
皮相電力量(消費／回生)
無効電力(回生)遅れ／進み
デマンド値（W／VA）
デマンド目標値（W／VA）
総合高調波電圧歪み率(F/R)
総合高調波電流歪み率(F/R)
高調波電圧／電流（1～50 次）

高調波測定

電圧／電流位相角（1～50 次）

データ

電圧電流位相差（1～50 次）●

高調波電力（1～50 次）
電圧／電流変動

半周期ごとの電圧実効値

データ

半周期ごとの電流実効値

イベント種別
データ
波形データ

●

イベント検出日時
イベント種別●イベント検出時の測定値
電圧／電流波形- 73 -●

KEW6315

KEW6315

記録方法に関する設定

記録方法に関する設定「インターバル」
測定データを SD カードまたは内部メモリに記録する時間の間隔を選択します。設定可能なインターバルは 17 種類ありますが、デマンド測定の「測定周期」を既に設定している場合には、それよりも長いインターバルが設定できません。
デマンド測定の測定周期を「測定なし」に変更してから選択してください。先にインターバルの設定を行っていた場合であっても、自動的に「測定周期」と同じ設定に変わるため注意が必要です。詳細は「デマンドに関する設定」５９頁を参照してください。

設

定

内

容

1 秒／2 秒／5 秒／10 秒／15 秒／20 秒／30 秒／
1 分／2 分／5 分／10 分／15 分／20 分／30 分／
1 時間／2 時間／150,180 周期（約 3 秒）※

は初期値です。

※「150,180 周期（約 3 秒）」は電力品質国際規格 IEC61000-4-30 にて指示されたインターバルです。50Hz の公称周波数では 150 周期、60Hz の公称周波数では 180 周期で集計します。

「インターバル」の項目へ移動

決定KEW6315

選択ウインドウを表示

キャンセル- 74 -

記録間隔を入力

記録方法に関する設定

KEW6315「開始方法」
記録開始の方法を選択します。
設

定

内

容

手動／連続記録／時間帯指定※

プルダウンメニュー表示「開始方法」の項目へ移動

決定は初期値です。

記録開始方法を選択

キャンセル

「手動」
キーを操作して記録開始／記録終了を行った間のみ記録します。

「連続記録」
記録を開始する日時と記録を終了する日時とを設定します。設定した日時の間連続して、インターバルの間隔で記録します。詳細は「⑧／⑨ 開始方法ごとの設定」４５頁を参照してください。
設

定

項

目

定

内

容

開始日時

年／月／日

時：分 （0000／00／00

00：00）

終了日時

年／月／日

時：分 （0000／00／00

00：00）「開始日時／終了日時」の項目へ移動

決定

設

日時入力ウインドウを表示

日時を入力

キャンセル

- 75 -

KEW6315

KEW6315

保存可能な時間について「時間帯指定」記録する期間を開始する日と終了する日とで設定し、この記録期間のそれぞれの日に対して記録する共通の時間帯を設定します。記録期間の日ごとに記録時間帯の間、インターバルの間隔で記録します。詳細は「⑧／⑨ 開始方法ごとの設定」４５頁を参照してください。
設

定

項

目

設

定

内

容

記録期間

開始－終了

年／月／日（YYYY／MM／DD）－年／月／日（YYYY／MM／DD）

記録時間帯

開始－終了

時：分（hh：mm）－時：分（hh：mm）「記録期間」の項目へ移動

日付入力ウインドウを表示

日付を入力

キャンセル

「記録時間帯」の項目へ移動

時間入力ウインドウを表示

時間を入力

決定

決定

キャンセル

保存可能な時間について
2GB の SD カードを使用した場合に記録可能な時間の目安
記録項目
インターバル

電力関連

記録項目

＋高調波

インターバル

電力関連

＋高調波

１秒

１３日

３日

１分

１年以上

３ヶ月

２秒

1５日

３日

２分

２年以上

６ヶ月

５秒

３８日

７日

５分

６年以上

１年以上

１０秒

２．５ヶ月

１５日

１０分

２年以上

１５秒

３．５ヶ月

２３日

１５分

３年以上

２０秒

５ヶ月

１ヶ月

２０分

３０秒

７．５ヶ月

１．５ヶ月

３０分

１０年以上

１時間

７年以上
１０年以上

２時間
150／180 周期

５年以上

２３日

４日

※

上記には電源品質のイベントデータを含んでおりません。イベントの記録を設定していた場合には、その発生量によって記録可能な残り時間が減少します。１回の記録開始／終了で保存できるイベントデータは最大 1GB です。

※

本製品に使用可能な SD カードは、弊社が提供する付属およびオプションの SD カードに限ります。

KEW6315- 76 -

5.5 その他

5.5

KEW6315

その他
キーを押す「その他」タブへ移動

システム環境に関する設定「言語」
表示言語を選択します。
設

定

内

容

日本語／英語※は初期値です。ただしシステムリセットを行っても変更した設定は初期化されません。

「言語」の項目へ移動

プルダウンメニュー表示- 77 -

言語を選択

決定

キャンセル

KEW6315

KEW6315

システム環境に関する設定「日付形式」
日付の表示形式を選択します。画面の右上に表示している現在の日付、記録開始終了に関する日付表示と設定入力
形式などが全て変更できます。
設

定

内

容

YYYY/MM/DD ／ MM/DD/YYYY ／ DD/MM/YYYY※

は初期値です。ただしシステムリセットを行っても変更した設定は初期化されません。

プルダウンメニュー表示「日付形式」の項目へ移動

決定

形式を選択

キャンセル

「CH 配色」
電圧と電流の各 CH に対応した色を選択します。項目ラベルの文字色、各グラフの色、結線図の CH 色などが変更で
きます。
設

定

内

容

白色／黄色／ｵﾚﾝｼﾞ／赤色／灰色／青色／緑色
VN は結線図にのみ対応しています。
VN：黄色／1CH：赤色／2CH：白色／3CH：青色／4CH：緑色が初期値です。ただしシステムリセットを行っても変更した設定は初期化されません。

※「CH 配色」の項目へ移動

決定KEW6315

選択ウインドウを表示

キャンセル- 78 -

色を選択

本体システムに関する設定

KEW6315

本体システムに関する設定「現在日時」
システムクロックの時間を設定します。現在の日時に合わせてください。
設

定内yyyy/mm/dd

容

hh：mm

※

「日付形式」に同期して入力形式も変わります。

「現在日時」の項目へ移動

決定

日時入力ウインドウ表示

日時を入力

キャンセル

「ＩＤ番号」
本体の認識番号を設定します。複数台使用する場合や１台で複数の場所を定期的に測定する場合に番号を使い分けると、記録データを解析する時に便利です。
設

定

内

容

00-001～99-999(00-001)※「ＩＤ番号」の項目へ移動

決定は初期値です。

数値入力ウインドウ表示

認識番号を入力キャンセル- 79 -

KEW6315

KEW6315

本体システムに関する設定「ブザー音」
キーを操作した時のブザー音のあり、なしを選択します。デマンド判定や乾電池の警告ブザーは、この設定には関係
しません。
設

定

内

容

あり／なし
※

「ブザー音」の項目へ移動

決定は初期値です。

プルダウンメニュー表示

ブザー音あり／なしを選択

キャンセル「Bluetooth®」
内蔵している Bluetooth®の有効、無効を選択します。Bluetooth®通信を行わない場合には無効を選択してください。
設

定

内

容

有効／無効
※

「Bluetooth®」の項目へ移動

決定は初期値です。

Bluetooth®有効／無効を選択

プルダウンメニュー表示

キャンセル

「電源」
記録をしていない状態で無操作状態が続いた場合に、自動で電源をＯＦＦするか、ＯＦＦしないか選択します。乾電池で駆動している場合には乾電池の容量を節約するために「自動でＯＦＦしない」を選択できません。
設

定

対

象

設

AC 電源

定

5 分後に OFF※

決定

KEW6315容

5 分後に OFF ／自動で OFF しない

乾電池「電源」の項目へ移動

内は初期値です。

プルダウンメニュー表示

キャンセル- 80 -

自動で電源をＯＦＦする／しないを選択

本体システムに関する設定

KEW6315「バックライト」
無操作状態が続いた場合に、自動でバックライトをＯＦＦするか、ＯＦＦしないか選択します。乾電池で駆動している場
合には乾電池の容量を節約するために「自動でＯＦＦしない」を選択できません。
設

定

対

象

設

AC 電源

定

容

5 分後に OFF ／自動で OFF しない

乾電池

2 分後に OFF※

「バックライト」の項目へ移動

内は初期値です。

プルダウンメニュー表示

自動でバックライトをＯＦＦする／しないを選択

決定

キャンセル

「システムリセット」
環境設定「言語」､「日付形式」､「ＣＨ配色」および､「現在日時」を除いた設定を出荷時の状態に初期化します。

「システムリセット」の項目へ移動

確認メッセージ表示

はい／いいえを選択決定- 81 -

KEW6315

5.6 保存データ

KEW6315

5.6

保存データ
キーを押す

「

：測定データ」、「「保存データ」タブへ移動：プリントスクリーン」、「：設定データ」を：SD カードまたは：内部メモリに保存します。本体に SD カードを挿入していた場合には、自動的に SD カードへデータを記録します。内部メモリへ記録したい場
合には SD カードを挿入しないでください。記録先を本体の設定などで選択することはできません。
データの記録先は SD カードを推奨します。内部メモリに保存可能なファイル数は測定データは３ファイル、その他のフ
ァイルは８ファイルまでです。

記録データに関する操作

操作項目へ移動KEW6315- 82 -

決定記録データに関する操作

KEW6315「データを削除する」：SD カードと：内部メモリに記録している「：測定データ」、「：プリントスクリーン」、「：設定データ」を、一覧から選択して削除します。データの表示は記録日時の順番に並んでいません。データを記録した順序を知りたい
場合にはファイル名の右側に表示している記録日時を確認してください。ただし内部メモリからＳＤカードへ転送したデータは転送時の日時に更新されています。スクロールバーは１画面に全てのデータが表示できない場合に限り表示します。
操作対象
スクロールバー

チェックボックス

削除したいデータへ移動

はい／いいえを選択

選択「削除する」確認メッセージ表示

決定

削除したいデータを選択するとチェックボックスがに変わり、選択されていることを表示します。一度に複数のデータを削除したい場合には続けて選択してください。

削除する「削除する」で確認メッセージを表示し「はい」で決定するとデータを削除します。

内部メモリ／ＳＤカード「内部メモリ／ＳＤカード」では操作対象を変更します。現在の操作対象は画面左上にマークで表示します。
画面を切り替えると、それまでのデータ選択を解除します。

- 83 -

KEW6315

KEW6315

記録データに関する操作

空き容量「空き容量」で操作対象の記録情報をポップアップで表示します。

「閉じる」でデータを削除する画面

へ戻ります。

項目
容量

表示内容
合計サイズ

使用領域+空き領域の容量

空き容量

空き領域のみの容量

電力のみ
記録可能時間電力+高調波

保存可能数※

内部メモリのみ

測定データ
本体設定／
プリントスクリーン「記録項目」が電力関連のみの場合に記録できる
目安の時間「記録項目」が電力関連と高調波の場合に記録できる
目安の時間
記録測定済みの回数※

内部メモリへの記録は最大３件です

本体設定とプリントスクリーンを合わせた記録回数※

内部メモリへの記録は最大８回です。

戻る「戻る」を操作すると入力を終了して「保存データ」の画面に戻ります。

KEW6315- 84 -

記録データに関する操作

KEW6315「データを転送する」：内部メモリに記録している「して：測定データ」、「：プリントスクリーン」、「：設定データ」を、一覧から選択：SD カードへ転送します。データの表示は記録日時の順番に並んでいません。データを記録した順序を知りたい場合にはファイル名の右側に表示している記録日時を確認してください。ただし内部メモリからＳＤカードへ転送したデータは転送時の日時に更新されています。スクロールバーは１画面に全てのデータが表示できない場合に限り表示します。

操作対象スクロールバー

チェックボックス

転送したいデータへ移動

はい／いいえを選択

選択「転送する」確認メッセージ表示

決定

転送したいデータを選択するとチェックボックスがに変わり、選択されていることを表示します。一度に複数のデータを削除したい場合には続けて選択してください。

転送する「転送する」で確認メッセージを表示し「はい」で決定するとデータを転送します。

ＳＤカード「ＳＤカード」では転送先のＳＤカードに記録してあるデータを確認できます。転送データ選択画面へ戻るには再度「内部メモリ」を操作してください。画面を切り替えると、それまでのデータ選択を解除します。

空き容量「空き容量」で操作対象の記録情報をポップアップで表示します。

「閉じる」でデータを削除する画面

へ戻ります。詳細は「空き容量」８４頁を参照してください。

- 85 -

KEW6315

KEW6315

記録データに関する操作

戻る「戻る」を操作すると入力を終了して「保存データ」の画面に戻ります。

「フォーマットする」：SD カードまたは：内部メモリをフォーマットします。データの表示は記録日時の順番に並んでいません。デー

タを記録した順序を知りたい場合にはファイル名の右側に表示している記録日時を確認してください。ただし内部メ
モリからＳＤカードへ転送したデータは転送時の日時に更新されています。スクロールバーは１画面に全てのデータが表示できない場合に限り表示します。
操作対象
スクロールバー「フォーマット」確認メッセージ表示

はい／いいえを選択

決定

フォーマット「フォーマット」で確認メッセージを表示し「はい」で決定するとフォーマットを開始します。

内部メモリ／ＳＤカード「内部メモリ／ＳＤカード」では操作対象を変更します。現在の操作対象は画面左上にマークで表示します。

空き容量「空き容量」で操作対象の記録情報をポップアップで表示します。

「閉じる」でデータを削除する画面

へ戻ります。詳細は「空き容量」８４頁を参照してください。

戻る「戻る」を操作すると入力を終了して「保存データ」の画面に戻ります。

KEW6315- 86 -

保存データの種類について

KEW6315

保存データの種類について
データの取り扱いに関してファイル名は自動的にファイル№をカウントアップしてつけられます。カウントアップしたファイル№は電源をオフしても本体で記憶しています。そのため記録先を変更してもシステムリセットを行うか、最大のカウント数を越えるまでカウントアップし続けます。
保存するファイル№のファイル名と同じ名前のファイルを記録先に保存していると、データフォルダにあるファイル
は自動的に次のファイル№にカウントアップしてファイル名を作成し、データを保存しますが、「プリントスクリーン」と「本体設定」のファイルは同じファイル名で上書きします。システムリセットを行ってファイル№がゼロから始まる
場合と複数の本体で１枚のＳＤカードを使用する場合にはファイル名が重複しないように注意してください。ただし全てのファイル No.のファイル名、9999 種類を記録先に保存しているとデータフォルダにあるファイルも上書きします。注意してください。
ＰＣ：コンピュータ等でファイルを削除したり保存したフォルダまはたファイルの名前を変更すると、本体でのデータ
操作と解析用ＰＣソフトウェアでの解析ができなくなります。フォルダ名および、ファイル名を変更しないでください。

「プリントスクリーン」のＢＭＰデータを記録します。
ファイル名： ＰＳ－

ＳＤ

０００

．ＢＭＰ

｜

｜

｜

保存先ｺｰﾄﾞ
SD：SD ｶｰﾄﾞ
ME：内部ﾒﾓﾘ

ファイル№

拡張子

（000～999）（ＢＭＰファイル）「本体設定」の「保存データ」-本体設定から本体設定データを記録します。
ファイル名： SUP

Ｓ

００００

｜

｜

保存先コード
S：SD ｶｰﾄﾞ
M：内部ﾒﾓﾘ- 87 -.ＰＲＥ

ファイル№（0000～9999）

KEW6315

KEW6315

保存データの種類について「データフォルダ」
１回の測定ごとにインターバルデータ・電源品質データを記録するためのフォルダを、順次作成します。
フォルダ名：

／ ＫＥＷ

／

Ｓ

００００

｜

｜

保存先ｺｰﾄﾞ
S：SD ｶｰﾄﾞ
M：内部ﾒﾓﾘ

ファイル№（0000～9999）「インターバルデータ」
本体設定データ

ファイル名： SUP

Ｓ

００００

.ＫＥＷ

測定設定データ

ＩＮＩ

Ｓ

００００

.ＫＥＷ

電力測定データ

ＩＮＰ

Ｓ

００００

.ＫＥＷ

高調波測定データ

ＩＮＨ

Ｓ

００００

.ＫＥＷ

｜

｜

保存先コード
S：SD ｶｰﾄﾞ
M：内部ﾒﾓﾘ

ファイル№（0000～9999）「電源品質データ」
イベント種別データ

ファイル名： ＥＶＴ

Ｓ

００００

.ＫＥＷ

波形データ

ＷＡＶ

Ｓ

００００

.ＫＥＷ

電圧／電流変動データ

ＶＡＬ

Ｓ

００００

.ＫＥＷ

｜

｜

保存先コード
S：SD ｶｰﾄﾞ
M：内部ﾒﾓﾘ

KEW6315- 88 -

ファイル№（0000～9999）

本体設定の保存と読み込みに関する操作

KEW6315

本体設定の保存と読み込みに関する操作

操作項目へ移動決定「設定を保存する」：SD カードまたは：内部メモリに「：設定データ」を保存します。データの表示は記録日時の順番に並んでいません。データを記録した順序を知りたい場合にはファイル名の右側に表示している記録日時を確認してください。ただし内部メモリからＳＤカードへ転送したデータは転送時の日時に更新されています。スクロールバーは１画面に全てのデータが表示できない場合に限り表示します。

操作対象
スクロールバー「保存する」確認メッセージ表示

はい／いいえを選択決定

保存する「保存する」で確認メッセージを表示し「はい」で決定すると本体設定をＳＤカードまたは、内部メモリに保存します。

内部メモリ／ＳＤカード「内部メモリ／ＳＤカード」では保存先を変更します。現在の保存先は画面左上にマークで表示します。

- 89 -

KEW6315

KEW6315

本体設定の保存と読み込みに関する操作

空き容量「空き容量」で操作対象の記録情報をポップアップで表示します。

「閉じる」でデータを削除する画面

へ戻ります。詳細は「空き容量」８４頁を参照してください。

戻る「戻る」を操作すると入力を終了して「保存データ」の画面に戻ります。

以下の本体設定を保存します
基 本 設 定設

測 定 設 定
定

項

設

目

定

項
測定周期

結線
デマンド

電圧レンジ

判定周期

VT 比

目標値

公称電圧

THD(総合高調波歪曲)算出方法

クランプ／電流レンジ

高調波

許容値範囲の設定

CT 比

MAX ホールド

DC レンジヒステリシスしきい値

周波数

トランジェントしきい値
スウェルしきい値

電源品質

ディップしきい値

そ の 他 設 定
設
環境設定
本体設定

目

定

瞬停しきい値
項目

インラッシュカレントしきい値

日付形式
ID 番号

フリッカ

フィルタ係数（ランプ）

進相コンデンサ

目標力率値

ブザー音

記 録 設 定
設
記録項目

記録方法

定

項

高調波
電源品質（発生イベント）
インターバル
開始方法
開始日時

連続記録
時間帯指定

KEW6315- 90 -

目

終了日時
記録期間

開始－終了

記録時間帯

開始－終了

本体設定の保存と読み込みに関する操作

KEW6315「設定を読み込む」：SD カードまたは：内部メモリから「：設定データ」を読み込んで本体の設定を変更します。データの表示

は記録日時の順番に並んでいません。データを記録した順序を知りたい場合にはファイル名の右側に表示している
記録日時を確認してください。ただし内部メモリからＳＤカードへ転送したデータは転送時の日時に更新されています。
操作対象
スクロールバー
チェックボックス

転送したいデータへ移動

はい／いいえを選択

選択「読み込む」確認メッセージ表示

決定スクロールバーは１画面に全てのデータが表示できない場合に限り表示します。読み込みたいデータを選択するとチェックボックスがに変わり、選択されていることを表示します。

読み込む「読み込む」で確認メッセージを表示し「はい」で決定するとデータを転送します。

内部メモリ／ＳＤカード「内部メモリ／ＳＤカード」では読み込み元を変更します。現在の読み込み元は画面左上にマークで表示します。

空き容量「空き容量」で操作対象の記録情報をポップアップで表示します。

「閉じる」でデータを削除する画面へ戻ります。詳細は「空き容量」８４頁を参照してください。

戻る「戻る」を操作すると入力を終了して「保存データ」の画面に戻ります。

- 91 -

KEW6315

KEW6315

6.1 瞬時値「Ｗ」

6章 画面ごとの表示項目
6.1

瞬時値「W」
キーを押す「Ｗ」瞬時値画面を表示

測定値を一覧で表示する「一覧表示」（／「拡大表示」）（例）３Ｐ３Ｗ３Ａ+１Ａ（三相３線式＋オプション電流）を表示しています。
１ｃｈの測定値
２ｃｈの測定値
３ｃｈの測定値

オプションクランプセンサ
の測定値

合計値
（1ｃｈ+２ｃｈ+３ｃｈ）

インターバル内の経過時間／インターバル設定値

複数の測定値を１画面に表示します。表示項目とその表示位置はキーを操作することで変更できます。

KEW6315- 92 -

測定値を一覧で表示する

KEW6315

画面表示記号
V※１

P

有効電力

PF

力率

DC1
An※２

VL※１

相電圧
＋

消費

－

回生

＋

遅れ

－

進み

アナログ入力1ch の電圧
中性線の電流

Q

線間電圧

無効電力

f
DC2
PA※３

＋

遅れ

－

進み

A

電流

S

皮相電力

C※３

進相コンデンサ容量

周波数
アナログ入力2ch の電圧
電圧電流
位相差

＋

遅れ

－

進み

※１

Ｗ画面：結線３Ｐ４Ｗを選択した場合に、V と VL 表示を[カスタマイズ]できます。

※２

Ｗ画面：An は結線３Ｐ４Ｗを選択した場合のみ表示します。

※３

Ｗ画面：PA と C は[カスタマイズ]で表示できます。３P３W３ＡのＰＡは線間電圧を相電圧へ変換して電流と位相角を求めています。

（例）１Ｐ３Ｗ－２（２系統）を表示しています。

- 93 -

KEW6315

KEW6315

測定値を一覧で表示する「表示する系統を変更する」
を操作することで表示する系統を変更できます。結線と系統数によって 1 画面に表示できる項目は以下のようになります。点線で区切られた範囲が 1 画面に表示できる範囲です。
１Ｐ２Ｗ－１～４（単相２線式、１系統～４系統）
１系統

２系統

３系統

４系統

１ｃｈ
の測定値
(V1／A1)１ｃｈの測定値(V1／A2)１ｃｈの測定値(V1／A3)１ｃｈの測定値(V1／A4)

全系統の合計値

１Ｐ３Ｗ－１～２（単相３線式、１系統～２系統）
１系統

１ｃｈ
の測定値
(V1／A1)

２系統

１ｃｈ
の測定値
(V1／A3)２ｃｈの測定値(V2／A2)

Σ
１系統
１ｃｈ＋
２ｃｈ
の合計値

２ｃｈ
の測定値
(V2／A4)２系統
１ｃｈ＋２ｃｈの合計値

１系統
１ｃｈ＋２ｃｈ
の合計値

２系統
１ｃｈ＋
２ｃｈ
の合計値

全系統の合計値

３Ｐ３Ｗ－１～２（三相３線式２電力計法、１系統～２系統）
１系統

１ｃｈ
の測定値
(V12／A1)２ｃｈの測定値(V23／A2)

３ｃｈ
の演算値

１系統
１ｃｈ＋２ｃｈ＋３ｃｈの合計値

KEW6315

Σ

２系統

１ｃｈ
の測定値
(V12／A3)２ｃｈの測定値(V23／A4)

３ｃｈ
の演算値

２系統
１ｃｈ＋２ｃｈ＋３ｃｈの合計値- 94 -

１系統
１ｃｈ＋
２ｃｈ＋
３ｃｈ
の合計値

２系統
１ｃｈ＋
２ｃｈ＋
３ｃｈ
の合計値

全系統の合計値

測定値を一覧で表示する

KEW6315
３Ｐ４Ｗ（三相４線式）

３Ｐ３Ｗ３Ａ（三相３線式）

１ｃｈ
の測定値
(V12／A1)２ｃｈの測定値(V23／A2)３ｃｈの測定値(V31／A3)

１ｃｈの測定値(V1／A1)

１ｃｈ＋２ｃｈ＋３ｃｈ
の合計値

２ｃｈ
の測定値
(V2／A2)３ｃｈの測定値(V3／A3)

１ｃｈ＋２ｃｈ＋３ｃｈの合計値「表示種別を変更する」
を操作することで表示種別を瞬時値、インターバル範囲内での平均値／最大値／最小値に切り替えます。
インターバルを 1 秒に設定している場合には、表示更新とインターバルが同じ時間であるため瞬時値、平均値、最大値、最小値は全て同じ値になります。

「Ｗｈ」積算値「Ｗｈ」で積算値を表示する画面に切り替えます。詳細は「6.2 積算値「Ｗｈ」」１００頁を参照してください。

「拡大表示」「拡大表示」で４つの測定値を選択して拡大表示する「４分割」または、８つの測定値を選択して拡大表示する「８分割」画面に切り替えます。詳細は「大きな文字で測定値を表示する」９６頁を参照してください。

「トレンドグラフ」「トレンド」で表示項目のトレンドグラフを表示する画面に切り替えます。表示範囲は現在から過去 60 分
間分を表示します。詳細は「トレンドグラフを表示する」９７頁を参照してください。

「カスタマイズ」「カスタマイズ」で表示する測定項目と表示位置を変更します。詳細は「表示測定項目と表示位置を変更する」９９頁を参照してください。

- 95 -

KEW6315

KEW6315

大きな文字で測定値を表示する

大きな文字で測定値を表示する（例）「８分割」画面を表示しています。
「表示項目」「表示種別」（瞬時値／平均値・・・）

４つの測定値または８つの測定値を選択して、１つの画面に表示します。一覧表示画面よりも大きく文字を表示するため、測定値の確認がし易くなります。

「表示項目」

分割したそれぞれの位置に表示する測定項目を選択します。左側メニューの表示項目へ移動すると、選択でき
るｃｈと合計の測定項目を自動的に右側へ表示します。右側のメニューから拡大表示したい測定項目を選択してください。

表示したい位置の「表示項目」へ移動「表示項目」を選択

KEW6315

決定

選択メニュー表示

キャンセル- 96 -

トレンドグラフを表示する

KEW6315「表示種別」
分割したそれぞれの位置に表示する測定項目の「表示種別」を瞬時値：INST、インターバル範囲内での平均値：
AVG／最大値：MAX／最小値：MIN から選択します。インターバルを 1 秒に設定している場合には、表示更新とイ
ンターバルが同じ時間であるため瞬時値、平均値、最大値、最小値は全て同じ値になります。

表示したい位置の「表示種別」へ移動「表示種別」を選択

決定

プルダウンメニュー表示

キャンセル「一覧表示」「一覧表示」で一覧表示画面に切り替えます。

「４分割／８分割」「４分割／８分割」で１画面に拡大表示する項目数を、４項目と８項目にて切り替えます。

トレンドグラフを表示する（例）１Ｐ３Ｗ―２（単相３線式２系統）の有効電力をｃｈごとに表示しています。
系統 1 の１ｃｈ瞬時値
系統 1 の２ｃｈ瞬時値
系統２の１ｃｈ瞬時値
系統２の２ｃｈ瞬時値

ｃｈごとに色分けした
トレンドグラフ

経過時間

測定値を選択して、それぞれの経時変化をグラフで確認できます。

- 97 -

KEW6315

KEW6315

トレンドグラフを表示する（例）１Ｐ３Ｗ－２（単相３線式２系統）を表示しています。

「トレンドグラフ表示対象を変更する」
を操作することでトレンドグラフを表示する測定対象を変更します。

「Σ／ＣＨ」「Σ／ＣＨ」で系統ごとの合計値と総合計値のトレンドグラフまたは、ｃｈごとのトレンドグラフとを切り替えて表示します。「Σ／ＣＨ」の操作は表示している測定値のトレンドグラフに限定されません。全てのトレンドグラフに対して有効になります。「Σ」を操作すると系統ごとの合計値と総合計値のトレンドグラフを表示します。「ＣＨ」を操作するとｃｈごとのトレンドグラフを表示します。なお、３Ｐ４ＷのＡ：電流実効値を選択じた状態で「Σ」を操作す
るとＡｎ：中性線電流の測定値をトレンドグラフに表示します。

「一覧表示」「一覧表示」で一覧表示画面に切り替えます。

KEW6315- 98 -

表示測定項目と表示位置を変更する

KEW6315

表示測定項目と表示位置を変更する
現在の表示項目

変更後の表示項目

現在の表示項目を選択して別の測定値を表示できます。

変更したい位置の「表示項目」へ移動

決定

キャンセル

プルダウンメニュー表示

OK／キャンセルを選択表示する測定値を選択

決定

キャンセル

表示項目変更ウインドウを表示すると「現在の表示項目」と同じ項目を「変更後の表示項目」にも表示します。「変更後の表示項目」を変更すると「現在の表示項目」の測定値を表示している位置に「変更後の表示項目」を表示します。ただし同じ種類の表示項目の間でしか移動できません。表示位置は電圧／電流に関する測定項目と電力／進相コンデンサに関する測定項目の大きく２つに分かれています。画面表示記号の詳細に関しては「測定値を一覧で表示する」９３頁を参照してください。

- 99 -

KEW6315

KEW6315

6.2

6.2 積算値「Ｗｈ」

積算値「Wｈ」
キーを押す「Ｗｈ」積算値画面を表示（例）１Ｐ３Ｗ－２（単相３線式２系統）を表示しています。

記録開始からの
経過時間

Σは総合計です。
Σは系統ごとの合計です。

ある期間内に流れた電力を積算電力量として表示します。積算電力量は電気料金の支払いやエネルギー管理などに利用されています。

画面表示記号
WP

KEW6315

有効電力量

＋

消費

－

回生

WQ

無効電力量

- 100 -

＋

遅れ

－

進み

WS

皮相電力量

＋

消費

－

回生

6.2 積算値「Ｗｈ」

KEW6315（例）１Ｐ３Ｗ－２（単相３線式２系統）を表示しています。

「表示する系統を変更する」
を操作することで表示する系統を変更します。結線と系統数との関係は「結線に関する設定」
４９頁を参照してください。

「表示するｃｈを変更する」
を操作することで表示するｃｈを変更できます。結線とｃｈとの関係は「結線に関する設定」４９頁を参照してください。

「デマンド」「デマンド」でデマンド値を表示する画面に切り替えます。詳細は「6.3

デマンド」１０２頁を参照してください。

- 101 -

KEW6315

KEW63115

6.3

6.3 デマンド

「デマンド
ド」
キーを押す
す

「デマンド」画面をを表示「測定値
値／時間内推移図
図／デマンド推移図」画面を表示

測
測定値を表示示する「測定値」画面面を表示

る期間の平均電力
力をデマンドといいます。デマンド測定中に予測値が目標値を上回った場合には判定
定周期ごと
ある
にブ
ブザーで警告しまます。

画面表示項目残り時間

デデマンド「測定周期期」で設定した時間
間をカウントダウン
ンします。

目標値

デ
デマンド「目標値」でで設定した値を表表示します。

予測値

現負荷の測定周期
現期後のデマンド値を予測します。
現在値×測
測定周期
を
を時間の経過と共共に算出して表示します。
測定開始からのの経過時間

現在値

測定周期時間内の測のデマンド値(平均均電力)です。
(測定開始からのの WP＋の積算値
値)×1 時間
測定周期を
を時間の経過と共に算出して表示しします。

最大デマンド／ 測定開始から終了
測
了までの最大デマンドを表示します
す。測定値が最大デデマンドを超える度に表
記
記録年月日

示
示を更新します。

瞬
瞬時値「Ｗ」「瞬時値「Ｗ」」で瞬時値値を表示する画面
面に切り替えますす。詳細は「6.1 瞬時値「Ｗ」」９２頁」
を参参照してください。

KEW
W6315

- 102 -

時間内推推移図を表示する

KEW6315時
時間内推移図
図を表示す
する

画面表示項目残り時間

デデマンド「測定周期期」で設定した時間をカウントダウンします。

負荷率

目標値と現在値の目の割合です。
現在値
目標値 を表示します。

予測

目標値に対する予測
目
測値の割合です。
予測値を表示します。
目標値

残り時間
目標値

「判
判定周期」分の時
時間が経

予測値

過
過した時点で「予測測値」が「目目標値」を上回ってているとブザ
ザーにてお知らせせします。

デマンド値（現在
在値）

測定周期

判定周期

判定周期

判
判定周期

保存ポイント- 103 -

KEW6315

KEW6315

デマンド推移図を表示する

デマンド推移図を表示する

カーソル

スクロールバーを操作することでカーソルの移動と、デマンド推移図を左右にスクロールさせます。スクロールバーには測定期間全体の範囲を白色で表示し、現在の表示範囲を濃いオレンジ色で表示します。
画面表示項目
デマンド測定値／
記録年月日

カーソル位置のデマンド測定値と記録年月日を表示します。

最大デマンド測定値(測定画面に表示)

目標値

最新のデマンド測定値

デマンド測定値

デマンド開始／記録開始時間デマンド終了／
最新の記録時間

デマンド開始／記録開始時間はデマンド推移図を 1 画面に表示できない場合に表示します。

KEW6315- 104 -

6.4 ベクトル

6.4

KEW6315「ベクトル」
キーを押す
ベクトル表示部（例）３Ｐ４Ｗ（三相４線式）を表示しています。

各項目の測定値
Ｖ：電圧実効値※１ ／位相角※２
Ａ：電流実効値／位相角※２※１

３Ｐ３Ｗ３Ａでは線間電圧実効値を表示します。

※２

Ｖ１の位相を基準（０°）とした角度を表示します。

円の実線：電圧レンジ／電流レンジの最大値
位相角（進み）－０°～－１８０°

ベクトル表示部
電圧実効値（実線）
電流実効値（点線）
±１８０°

±０°

位相角（遅れ）＋０°～＋１８０°
電圧レンジと電流レンジの最大値を円の実線で表示して電圧実効値と電流実効値を線の長さ、それぞれの位相関係をＶ１を基準（０°）にした線の角度にて表示します。３Ｐ３Ｗ３Ａ／３Ｐ４Ｗの結線では同時に不平衡率を表示します。測定電圧と測定電流との関係が平衡な状態では下記のようなベクトル図を表示します。
1P2W
A

±0°

±0°

V

3P3W

1P3W

V

3P3W3A

A

1CH:±0° 1CH:＋30°
2CH:＋120° 2CH:＋150°
3CH:－120° 3CH:－90°

VA

1CH:±0° 1CH:±0°
2CH:±180° 2CH:±180°
V

3P4W

V

A

1CH:±0° 1CH:＋30°
2CH:－60° 2CH:－90°

A

1CH:±0°
1CH:±0°
2CH:＋120° 2CH:＋120°
3CH:－120° 3CH:－120°- 105 -

KEW6315

KEW6315

6.4 ベクトル（例）３Ｐ４Ｗを表示しています。

「Ｖ×倍率」「Ｖ×倍率」で電圧ベクトル図の線の長さを倍率で順番に切り替えます。
1倍

2倍

5倍

10 倍

「Ａ×倍率」「Ａ×倍率」で電流ベクトル図の線の長さを倍率で順番に切り替えます。
1倍

2倍

5倍

10 倍

「結線図」
「結線図」で結線方式に準じた「結線図」を表示できます。詳細は「結線図」５０頁を参照してください。

「結線確認」「結線確認」で確認項目に対して判定した結果を表示します。※力率が著しく悪い測定現場では、正しい結線を行っていても、NG と判定することがあります。詳細は「結線確認」４３頁を参照してください。

KEW6315- 106 -

6.5

波形

6.5

KEW6315

「波形」
キーを押す

（例）１Ｐ３Ｗ－２（単相３線式２系統）を表示しています。

各項目の測定値
Ｖ：電圧実効値※
Ａ：電流実効値
ｃｈごとに色分けした
測定波形※

３Ｐ３Ｗ３Ａでは線間電圧実効値を表示します。

電圧の波形と電流の波形を５０Hz ならば最長１０周期分、６０Hz ならば最長１２周期分表示します。「波形」画面に切り替えると波形振幅と期間を最大で表示できる倍率を自動で選択して表示します。

「波形表示対象を変更する」
を操作することで波形を表示する測定対象を変更します。

「V×倍率」「V×倍率」で電圧波形（縦方向）の倍率を下記の順番で切り替えます。
0.1 倍

0.5 倍

1倍

2倍

5倍

- 107 -

10 倍

KEW6315

6.6
6 高調波

KEW63115

「A
A×倍率」
「A×倍率
率」で電流波形（縦縦方向）の倍率を下
下記の順番で切りり替えます。
0.1 倍

0.5 倍

1倍

2倍

5倍

10 倍

「ｔｔ×倍率」「ｔ×倍率」で時間軸（横方向）の倍率を下記記の順番で切り替
替えます。
1倍

2倍

5倍

10 倍

「ｆｆｕｌｌ ｓｃａｌｅ」」「ｆｕｌｌ ｓｃｃａｌｅ」で電圧／電流波形を画面に最最大で表示できるる倍率に設定しま
ます。

6.6

「高調波
波」
キーを押す
す

高
高調波をバー
ーグラフで表表示する「グラフ」画画面を表示（例例）３Ｐ４Ｗ（三相４４線式）の「リニア」」「全体表示」を表表示しています。

画面表示記号
V
ＴＨＤ
P

KEW
W6315

電圧
※

３Ｐ３Ｗ
Ｗ３Ａでは線間電電圧を表示します。

A

電流

圧を表示している場合には電圧総総合高調波歪み率
率、「Ａ」電流を表示示している場合には電流
「Ｖ」電圧
総合高調
調波歪み率を「ＴＨＤ算出方法」に準じて表示します
す。
各ｃ
ｃｈ有効電力

＋

流入

－

流出

- 108 -

ΣＰ

系統合計／総合
合計

＋

流入

有効電力

－

流出

高調波をバーグラフで表示する

KEW6315

ーグラフ表示部バー（例例）「リニア」「全体表示」を表表示しています。
含有率

高調波解
解析次数 最大５００次の例では「リニア
ア」「全体表示」を選択しているるため含有率の上
上限は１００％ににて１次～５０次
次までの全
この
ての
の高調波を１画面に表示します
す。
画面表示項目
含有率

1 次の基本波にに対する各高調波
波の含有率です。

例）３Ｐ４Ｗ（三相４線式）の「対数」「拡大表表示」を表示して
ています。
（例

含有率

スクロールババー
高調波解析
析次数 表示範囲１
１５次の例では「対数」」「拡大表示」を選択しているたため含有率の上限
限は１０％になりり１画面の表示は
は１５次の
この
範囲
囲となります。表示範囲はで画面をス
スクロールするこ
ことで移動します
す。ただし、基準準となる

１次
次の基本波はスクロールしませせん。スクロールバ
バーには１次～５０次までの全体体の範囲を白色で表示し、現在在の表示範囲を濃いオレンジ色
色で表示します。

- 109 -

KEW6315

KEW63115

高調波
波をバーグラフでで表示する（例例）３Ｐ４Ｗ（三相４線式）の「対数」「拡大表表示」を表示して
ています。

軸値超過
最大値

しきい値超過

グラフ色
許容範囲
囲

画面表示項目
軸値超過

各次数の高調波
波含有率が１０％％を越えている場合に表示します。基本波１次は常常に含有率が１００％にななるため、「対数」」の表示では常にに「軸値超過」にななります。
測定開始からのの最大値を表示しします。表示のリセ
セットは設定変更
更、記録開始、最大値の２秒以上の長押
押しの、いずれかかの操作でリセット
トします。ただし記記録中にはリセット
トできま
せん。

グラフ色
しきい値超過
許容範囲

KEW
W6315

測定ｃｈ数が多い場合には各ｃｈｈに対応した色でグ
グラフを分割します。
設定した許容範
範囲を測定値が越越えています。
IEC61000-2-4 Class3 に準じてて設定してあります
す。変更したい場場合には[SETUP]の高調
る「許容範囲を編編集する」から変更
更してください。
波項目中にある- 110 -

高調波をバーグラフで表示する

KEW6315「表
表示するｃｈｈを変更するる」
を操作することとで表示するｃｈをを変更します。結線
線とｃｈとの関係は
は「結線に関するる設定」４９頁をを参照
照してください。

「リ
リスト／グララフ」「リスト／ググラフ」で１次～５０次までの電圧／／電流／電力高調
調波を、それぞれ
れの項目ごとにリス
ストで表示
しまます。バーグラフのの画面では含有率率のみの表示です
すが、リストでは実
実効値／含有率／位相角※をそれれぞれ選択してて表示します。
※「Ｐ」電力を表示ししている場合は、電電圧と電流の位相相差を表示しますす。±０°～±９００°の範囲が流入
入、±９０°～～１８０°の範囲がが流出です。

「対
対数／リニア」「対数／リニニア」で含有率（ババーグラフの縦軸軸）の上限を１０％に変更してバーググラフを表示します
す。レベルの低低い高調波成分をを解析するのに有
有効です。

「拡
拡大表示／
／全体表示」」「拡大表示
示／全体表示」で１次～５０次の高高調波から１５次のの範囲を拡大してて電圧／電流／電
電力高調波
を、、それぞれの項目
目ごとにバーグラフ
フで表示します。１５次の表示範囲
囲はで画
画面をスクロール
ルして切り

替え
えます。

「Ｖ
Ｖ/Ａ/Ｐ／Σ
ΣＰ」「Ｖ/Ａ/Ｐ／／ΣＰ」で高調波の解析対象を、Ｖ
Ｖ：電圧／Ａ：電流
流／Ｐ：電力（ΣＰ：系統合計、総合計
計）から
選択
択します。

- 111 -

KEW6315

KEW6315

高調波をリストで表示する

高調波をリストで表示する「リスト」画面を表示（例）１Ｐ３Ｗ－２（単相３線式２系統）の「Ｐ：高調波電力」「電力」を表示しています。

１次～５０次までの電圧／電流／電力高調波の実効値／含有率／位相角を、それぞれの項目ごとにリストで表示します。
画面表示記号VP※２

電圧
各ｃｈ有効電力

※１

A
＋

流入

－

流出

ΣＰ※２

電流
系統合計／総合計

＋

流入

有効電力

－

流出

※１

３Ｐ３Ｗ３Ａでは線間電圧を表示します。

※２

記号と数字との関係はＰ[ｃｈ番号]＿[系統番号]のように表示します。Ｐと番号との間に間隔があいている
場合には系統番号のみを表示している状態です。この時の表示測定値は系統ごとの合計を表示しています。
Ｐのみ表示している場合は総合計を表示しています。

KEW6315- 112 -

高調波をリストで表示する

KEW6315「表示する次数を変更する」
を操作することで画面を縦にスクロールして表示する次数を変更します。

「グラフ／リスト」「グラフ／リスト」で１次～５０次までの電圧／電流／電力高調波を、それぞれの項目ごとにバーグラフで表示します。バーグラフの画面では含有率のみを表示します。

「含有率／位相角／実効値（電力）」「含有率／位相角／実効値（電力）」でリストに表示する測定項目を変更します。表示する測定対象に「Ｖ：電圧」「Ａ：電流」を選択している場合には含有率／位相角（Ｖ１を基準（０°）とした位相角）／実効値から選択します。「Ｐ（ΣＰ）：電力」を選択している場合には含有率／位相角（ｃｈごとの電圧電流位相差）／電力から選択します。

「Ｖ/Ａ/Ｐ／ΣＰ」「Ｖ/Ａ/Ｐ／ΣＰ」で高調波の解析対象を、Ｖ：電圧／Ａ：電流／Ｐ：電力（ΣＰ：系統合計、総合計）から選択します- 113 -

KEW6315

KEW6315

6.7

6.7 電源品質「電源品質」
キーを押す

電源品質を低下させる事象とその現象
事象

波形表示

主な現象
機器の制御回路は、インバータ
回路(コンデンサインプット型整流回路)およびサイリスタ制御回路

高調波(位相制御回路)を使用しています。これらの回路は電流に歪みを生じさせ、その歪みが高調波を発生させます。

主な弊害
高調波電流が流れると進相コンデンサおよびリアクトルの焼損、トランスのうなり、ブレーカの誤作動、テレビ映像のちらつき、ステレオ等へ雑音の影響があります。

RMS

電力ラインの開閉器の電源投入
電圧スウェル

時に突入電流が発生し、瞬時的に電圧が上昇します。

RMS

モータ負荷等の起動時に突入電
電圧ディップ流が発生し、電流降下を発生さ
せます。

RMS

電圧瞬停

KEW6315

落雷等により電力供給が一瞬停止状態になります。

- 114 -

機器／溶接ロボット等の動作
停止やコンピュータ等の OA
機器のリセットを引き起こします。

電源品質を低下させる事象とその現象
事象

KEW6315

波形表示

トランジェント・

主な現象
ブレーカ、マグネット、リレーの接オーバー電圧点不良等により発生します。

(インパルス)

モーター、白熱灯、大容量の平滑インラッシュ

コンデンサを持つ機器等の起動

カレント

時等に、一時的に流れる大電流(サージ電流)です。

主な弊害
急峻な電圧変化(スパイク)のため、機器の電源を破壊、リセット動作を引き起こします。
電源スイッチ接点の溶着、ヒューズの溶断、ブレーカのト
リップ、整流回路などへの悪影響、電源電圧の不安定化を引き起こします。

動力ライン負荷の増減、また偏った設備機器増設等により、特定の相が重負荷になる。そのため、不平衡

電圧・電流波形の歪、また電圧降
下および逆相電圧が発生しま

電圧・電流のアンバランス・モ
ータの回転ムラ・逆相電圧・高調波等が発生します。

す。

動力ラインなど各相ごとに接続された負荷の増減や、偏った設備
フリッカ測定RMS
RMS

機器の稼動により、特定相だけの負荷が重くなり、電圧降下が発生します。

- 115 -

電圧のアンバランス・逆相電圧・高調波の発生などにより、モータの回転ムラやブレーカのトリップ、トランスの過負荷発熱などの事故につながることがあります。

KEW6315

KEW6315

イベントの発生状況を表示する

イベントの発生状況を表示する「イベント」画面を表示
イベント測定値

イベント記号

画面表示記号
開始

終了

スウェル
ディップ

イベント記号

瞬停
トランジェント
インラッシュカレント
イベントの開始と終了を判定した時点の瞬時値を表示します。イベントの開始と終了時
間が短い場合には終了時の測定値のみ表示しない場合があります。判定前後の実効値
イベント測定値を確認するには実効値変動データを参照してください。長期間継続しているイベント中の測定値はインターバル測定データから確認できます。電源品質を記録する場合には、できるだけ短いインターバルの設定が解析には有効です。

発生日時

イベントの開始と終了を判定した時間を表示します。

多相システムでのイベント判定「瞬停」結線方式にて測定対象となる全ｃｈの電圧が瞬停になった時点をイベント開始と判定します。その後、いずれか１ｃｈの瞬停が終わった時点でイベント終了と判定します。

「スウェル／ディップ／インラッシュカレント／トランジェント」
結線方式にて測定対象となる、いずれか１ｃｈの電圧／電流が、イベント状態になった時点をイベント開始と判定します。その後、全ｃｈのイベントが終わった時点でイベント終了と判定します。

KEW6315- 116 -

イベントの発生状況を表示する

KEW6315

スウェル／ディップ／瞬停／インラッシュカレントの測定方法
ギャップなしで半周期毎にオーバーラップさせた１波形の実効値から各イベントを検出します。１波形の実効値で初めてイベントを検出した場合、その 1 波形の先頭をイベント開始時刻とし、その後の１波形の実効値でイベントが検出されなかった場合に、その 1 波形の先頭をイベント終了時刻とします。なお、開始から終了までの間はイベントが継続していると見なします。
ディップ検出例※

瞬停も同じ方法で検出します。
実効値②

実効値④

実効値⑥

実効値⑧

測定波形

実効値波形

実効値①

実効値③

①

③

②

実効値⑤

④

しきい値

実効値⑦

実効値⑨

⑦

⑨

⑧

⑥

ヒステリシス⑤

ディップ開始ディップ終了発生期間
スウェル検出例※

インラッシュカレントも同じ方法で検出します。
実効値②

実効値④

実効値⑥

実効値⑧

測定波形

実効値①

実効値③

実効値⑤

実効値⑦

実効値⑨

⑤

しきい値
実効値波形

①

②

④

⑥

③

ヒステリシス⑦

スウェル開始⑧⑨スウェル終了発生期間- 117 -

KEW6315

KEW6315

イベントの発生状況を表示する

トランジェントの判定方法
電圧波形をギャップなしで約４０ｋｓｐｓでモニタし、約２００ｍｓ区間ごとにトランジェントを集計しながら判定します。
そのため、初めてトランジェントを検出した約２００ｍｓ区間の最初がイベント開始時刻となり、その後の２００ｍｓ区間で、トランジェントが検出されなかった場合に、その区間の最初をイベント終了時刻とします。なお、開始から終了までの間はイベントが継続していると見なします。
トランジェント検出例

しきい値

測定波形

トランジェント開始

トランジェント終了

約２００ｍｓ間
発生期間

保存データ

イベントが発生した場合には、そのイベントの種別、開始時間、終了時間、測定値を記録しながら、イベント波形と実効値変動を同時に記録します。ただしイベント波形はデータ更新タイミング１秒間の内、約２００ｍｓ区間のみ記録します。
イベント波形
イベントデータを含んだ約２００ｍｓ（５０Ｈｚ：１０周期／６０Ｈｚ：１２周期）区間の全ｃｈ電圧／電流波形データを８１
９２ポイントで記録します。データ更新タイミング１秒以内に複数の異なるイベントが同時に発生した場合には、最も優先順位の高いイベントを含んでいる、約２００ｍｓ区間の波形データのみを記録します。ただし同じ項目のイベントが同時に発生した場合には、より最高値（最深値）を示すイベントを選択します。これも同じ場合には、発生期間の長いイベントを選択して記録します。なお接続チャンネル間に優先順位はありません。
【最優先】→電圧トランジェント→瞬停→ディップ→スウェル→インラッシュカレント
実効値変動
イベントデータを含んだデータ更新タイミング約１秒間の全ｃｈの電圧／電流実効値（分解能：半周期）変動データを記録します。
約８００ｍｓ間のディップ検出例（保存データ）
イベント開始波形（約２００ｍｓ）

イベント終了波形（約２００ｍｓ）

測定波形

実効値波形

イベント開始実効値変動（約１秒）

KEW6315- 118 -

イベント終了実効値変動（約 1 秒）

イベントの発生状況を表示する

KEW6315「表示範囲を変更する」
を操作することで画面を縦にスクロールして表示する範囲を変更します。

「フリッカ」「フリッカ」でフリッカを表示する画面に切り替えます。詳細は「フリッカ測定値を一覧表示する」１２０頁
を参照してください。

「イベント抽出」「イベント抽出」で表示するイベント種別を下記の順番で切り替えます。

「全イベント」
スウェル

ディップ- 119 -

瞬停

トランジェント

インラッシュカレント

KEW6315

KEW6315

フリッカ測定値を一覧表示する

フリッカ測定値を一覧表示する「フリッカ」画面を表示「Ｖ：一覧表示／Ｐｓｔ(１ｍｉｎ)：トレンドグラフ／Ｐｌｔ：推移図」画面を表示

残り時間

アーク炉負荷のような変動負荷がある場合に、電圧が変動して照明などがちらつく場合があります。このような状態を
電圧フリッカといい、Ｐｓｔ、Ｐｌｔでその厳しさを示します。

画面表示記号
残り時間

Ｐｓｔ算出までの残り時間をカウントダウンします。Ｐｓｔの評価には１０分間の測定時間が
必要です。

V

相電圧※３Ｐ３Ｗ、３Ｐ３Ｗ３Ａでは線間電圧を表示します。

f

周波数

Ｐｓｔ，１ｍｉｎ
Ｐｓｔ
Ｐｓｔ，ＭＡＸ
Ｐｌｔ
Ｐｌｔ，ＭＡＸ

１分間で測定した短期間フリッカの厳しさです。判断基準として電力調査や研究時の評価に有効です。
１０分間で測定した短期間フリッカの厳しさです。
記録を開始してからのＰｓｔの最大値です。測定値が現在の最大値を超えるごとに表示を更新します。
２時間で測定した長期間フリッカの厳しさです。
記録を開始してからのＰｌｔの最大値です。測定値が現在の最大値を超えるごとに表示を更新します。

「イベント」「イベント」でイベントの発生状況を表示する画面に切り替えます。詳細は「イベントの発生状況を表示
する」１１６頁を参照してください。

KEW6315- 120 -

Ｐｓｔ，１ｍｉｎのトレンドグラフを表示する

KEW6315

Ｐｓｔ，１ｍｉｎのトレンドグラフを表示する

最新のＰｓｔ，1ｍｉｎＰｓｔ，1ｍｉｎ最大値

経過時間

最新１２０分間に測定したＰｓｔ，1ｍｉｎのトレンドグラフを表示します。

画面表示記号
Ｐｓｔ，１ｍｉｎ最大値
経過時間

１分間で測定した最新の短期間フリッカの厳しさです。
記録を開始してからのＰｓｔ，1ｍｉｎの最大値です。Ｐｓｔ，1ｍｉｎの測定値が現在の最大値を超えるごとに表示を更新します。
最新の測定値を右端（経過時間 0 分）に表示し、時間が経過するごとに表示位置を左へ移動します。表示する範囲は現在の時間から、最大で１２０分間です。

- 121 -

KEW6315

KEW6315

Ｐｌｔの推移図を表示する

Ｐｌｔの推移図を表示する

カーソル

スクロールバーを操作することでカーソルの移動と、Ｐｌｔの推移図を左右にスクロールさせます。スクロールバーには測定期間全体の範囲を黒色で表示し、現在の表示範囲を濃いオレンジ色で表示します。

画面表示項目
Ｐｌｔ測定値／
記録年月日

カーソル位置の各ｃｈのＰｌｔと記録年月日を表示します。

最新のＰｌｔ

最大値

Ｐｌｔの測定値

最新の記録時間

記録開始時間

記録開始時間はＰｌｔ推移図を 1 画面に表示できない場合に表示します。
画面表示記号
最大値

KEW6315記録を開始してからのＰｌｔの最大値です。測定値が現在の最大値を超えるごとに表示を更新します。

- 122 -

7章

その他の機能KEW6315

7章 その他の機能「データホールド機能」
DATA HOLD キーを押すと測定状態に関わらず表示更新をストップして、画面上にび DATA HOLD キーを押すとのマークを表示します。再のﾏｰｸは消え、表示更新をスタートします。データホールド中でも表示を切り替

えて、各画面の測定値を確認できます。記録中に表示更新をストップしても、測定値とイベントの情報を常に記録しています。

「キーロック機能」
DATA HOLD ｷｰを 2 秒以上押すと、画面上にを表示し、その時点から LCD キーを除く全てのキー入力をキャンセルします。再び DATA HOLD キーを２秒以上押し続けるとが消え、キーロックを解除します。

「バックライトの消灯」
LCD キーでバックライトを消灯します。再点灯する場合には、電源キーを除くキーを操作してください。

「自動バックライトオフ」
AC 電源接続時何も操作しない状態が 5 分間経過すると自動的にバックライトを消灯します。再点灯する場合には、電源キーを除くキーを操作することで再度 5 分間点灯します。SETUP から「自動で OFF しない」を選択することで、バックライトを常時点灯することも可能です。

電池駆動時
電池駆動に切り替わった時点で消費電流を節約するために、バックライトの明るさが AC 電源で駆動している状態の約 1／2 になります。何も操作しない状態が 2 分間経過すると､自動的にバックライトを消灯します。再点灯する
場合には、電源キーを除くキーを操作することで再度 2 分間点灯します。電池駆動では常時点灯できません。

「自動電源オフ」
AC 電源接続時何も操作しない状態が 5 分間経過すると自動的に電源をオフします。ただし記録中にはオフしません。再度電源をオンしたい場合には電源キーを操作してください。SETUP から「自動で OFF しない」を選択することで、常に電源オンすることも可能です。

電池駆動時何も操作しない状態が 5 分間経過すると自動的に電源をオフします。ただし記録中にはオフしません。再度電源をオンしたい場合には電源キーを操作してください。電池駆動の場合は常時電源オンし続けることはできません。

- 123 -

KEW6315

7章

KEW6315

その他の機能「電流オートレンジ」
測定した電流実効値から各センサの電流レンジを自動で切り替えます。電源品質のイベントを記録する場合には選択できません。切り替えは下側レンジの 300%peak を上回るとレンジアップ、下側レンジの 100%peak を下回るとレンジダウンです。ただし表示は上側レンジに固定します。

「センサ識別機能」
SETUP からセンサ識別を操作することにより、現在本体に接続しているクランプセンサを自動で識別します。
電源起動時には、その時点で接続しているクランプセンサと、前回測定時のクランプ設定との相関を、自動で確認します。

「停電復帰機能」
記録中に電源の供給が無くなり電源がオフされた場合には、その後、電源が再供給された時点で、自動的に記録を再開します。

「プリントスクリーン機能」
PRINT SCREEN キーを操作することで、現在表示している画面を BMP ファイルで保存します。1 ファイルのサイズ
は約 77KB です。

「設定記録機能」
電源をオフしても前回測定時の全ての設定を記憶してるため、次に電源をオンした時には前回測定時と同じ状態
で起動します。※出荷後初回の起動時は初期設定値です。

「かんたんスタートナビ」
START/STOP キーを操作することによりスタートナビを開始します。画面の表示に従って項目を順次設定すること
で簡単に記録を開始できます。

「ステータス LED」
バックライトが消えている状態では赤色で点滅し、記録中はバックライトの点灯に関係なく、常に緑色で点灯します。記録スタンバイ中は緑色で点滅しています。

KEW6315- 124 -

8.1 ＰＣ：コンピュータへのデータ転送

KEW6315

8章 周辺機器との接続
8.1

ＰＣ：コンピュータへのデータ転送

SD カードおよび内部メモリに保存したデータは、USB 接続又は SD カードリーダーを使用することにより PC に転送することが可能です。
PC 転送方法
USB

※1

※1

カードリーダーSD カードデータ（ファイル）△○

内部メモリデータ（ファイル）○---------

ＰＣへのデータ移動にはＳＤカードリーダーの使用をおすすめします。（本製品の転送時間 約 320MB／時）保

存容量の大きいデータを直接 USB 接続で PC に転送すると、SD カードリーダーを使用た場合に比べて長い時間が必要になります。なお、使用する SD カードの取り扱いについては、カードに付属されている取扱説明書を確認してください。確実にデータを保存するために、本製品のファイル以外は SD カードに記録しないでください。必要のないファイルは事前に全て削除してください。

USB 接続

内部メモリデータ／
SD カードデータ

SD カードSD カードスロットもしくはSD カードリーダー- 125 -

KEW6315

8.2 Bluetooth®を使用する

KEW6315

Bluetooth®を使用する

8.2

本体内蔵の Bluetooth® によってＡｎｄｒｏｉｄ ＯＳ対応の機器を用いたリアルタイムでのデータ確認が可能です。
Bluetooth®を使用する際は、「設定２６：Bluetooth®の電源」をＯＮにする必要があります。
Bluetooth®

Ａｎｄｒｏｉｄ端末※

Ａｎｄｒｏｉｄ端末と通信を行うには、専用のアプリケーションソフト「ＫＥＷ Ｓｍａｒｔ」が必要です。「ＫＥＷ Ｓｍａｒｔ」はＧｏ

ｏｇｌｅ Ｐｌａｙ ストア(旧Ａｎｄｒｏｉｄ マーケット)で無料配信しています。注意：インターネット接続にかかる費用はお客様のご負担になります。
「商標について」Bluetooth®は Bluetooth SIG の商標です。

8.3 外部機器との信号制御
入出力端子を接続する注意●入力端子には±１１Ｖの範囲内の電圧、出力端子には０～３０Ｖ(５０ｍＡ、２００ｍＷ)の範囲内の電圧を入力してください。これを超えると本体を破損する恐れがあります。
●各チャンネルの L 端子は、つながっています。L 端子側にグランドレベルの異なる入力を同時に接続しないでく
ださい。

出力端子

入力端子

接続の際には、入力端子と出力端子の接続を間違えないように確認してください。
接続可能な信号線の寸法は下記のとおりです。
適合電線：単線φ１．２ｍｍ(ＡＷＧ１６)
撚線１．２５ｍｍ２(ＡＷＧ１６)素線径φ０．１８ｍｍ以上使用可能電線：単線φ０．４～１．２(ＡＷＧ２６～１６)
撚線０．２～１．２５ｍｍ２(ＡＷＧ２４～１６)素線径φ０．１８ｍｍ以上
標準むき線長：１１ｍｍ

KEW6315- 126 -

入出力端子を接続する
１

KEW6315コネクタカバーを開きます。

２

端子の上にある四角の部分をマイナスドライバーなどで押しながら、信号線を挿入します。

３ドライバーを離すと信号線が固定されます。

マイナスドライバーなど

信号線「入力端子」
温度センサーなどの電圧出力信号をモニターします。他の機器から出力される信号と、その電源に起こる異常を同時に測定する場合に有効です。
チャンネル数：２ｃｈ
入力抵抗：約２２５．６ｋΩ

「出力端子」
電源品質のイベントが発生している間、出力端子をＬｏｗにします。通常はＨｉｇｈです。イベントの継続している時間が１秒未満であった場合には、1 秒出力端子をＬｏｗにします。ただし、出力対象イベントはイベント設定している内で最も優先順位が高いイベントに対してのみです。優先順の低いイベントに同期して出力端子をＬｏｗにしたい場合には、そのイベントよりも優先順位の高いイベントを「ＯＦＦ」に設定してください。詳細は「電源品質（イベント）しきい値設定」６５
頁を参照してください。【最優先】→トランジェント→瞬停→ディップ→スウェル→インラッシュカレント

+5V

出力形式：オープンコレクタ出力
最大入力：３０Ｖ、５０ｍＡ、２００ｍＷ
出力電圧：Ｈｉ －４～５Ｖ
Ｌｏ－０～１VH

OUTPUTL- 127 -

KEW6315

8.4 測定ラインから電源を供給する

KEW63158.4

測定ラインから電源を供給する

コンセントからの AC 電源の供給ができない場合、ＭＯＤＥＬ８３１２（電源供給アダプタ)を用いることで、電圧用測
定コードから電源供給を行うことができます。
危険●測定コードと本体の測定カテゴリが違っている場合は低い方の測定カテゴリが優先されます。測定電圧と定格が合っているか必ず確認してください。
●測定に必要のない電圧用測定コードは絶対に接続しないでください。
●本体に接続していない状態で測定ラインに接続しないでください。
●測定中（測定ラインからの通電中）は絶対に本体のコネクタから取りはずさないでください。
●必ずブレーカーの二次側に接続してください。1 次側は電流容量が大きく危険です。

警告●本体の電源が OFF になっていることを確認してから接続してください。
●接続は必ず先に本体側から行い、根元まで確実に差し込んでください。
●測定コードのコード内部から金属部分または外装被覆と異なる色が露出したときは、直ちに使用を中止してください。
以下の手順で本製品を接続します。

２４０Ｖ以上注意
５●安全に使用するため、以下の手順通り接続してください。
●手順通り接続しないとヒューズが溶断することがあります。
１

ＭＯＤＥＬ８３１２の電源スイッチがＯＦＦになっていることを確認します。

２

ＭＯＤＥＬ８３１２のプラグをＫＥＷ６３１５のＶＮとＶ１端子に差し込みます。

３

ＭＯＤＥＬ８３１２の電源プラグをＫＥＷ６３１５の電源コネクタに差し込みます。

４

ＭＯＤＥＬ８３１２の VN とＶ１端子へそれぞれの電圧用測定コードを接続します。

５

電圧用測定コードのワニグチを被測定回路に接続します。

６

ＭＯＤＥＬ８３１２の電源スイッチをＯＮにします。

７

ＫＥＷ６３１５の電源スイッチをＯＮにします。
４

本製品をはずす場合は、逆に ７ から １ へ向かっての手順になります。

１ / ６

使用方法の詳細はＭＯＤＥＬ８３１２の取扱説明書を参照ください。
２

ＭＯＤＥＬ８３１２

３

測定 ＣＡＴ Ⅲ１５０Ｖ ＣＡＴ Ⅱ２４０Ｖ
ヒューズ定格：ＡＣ５００ｍＡ／６００Ｖ
速断タイプ、φ６．３×３２ｍｍ

７KEW6315- 128 -

9 章 設定・解析用ＰＣソフトウェア

9章 設定・解析用ＰＣソフトウェア

KEW6315

ＰＣから「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ６３１５」を使用することで、本体で記録したデータの解析 ※と、本体の設定を行うことができます。※記録データからグラフとリストを 1 クリック自動生成、記録データのＣＳＶ形式変換、複数台の設定データ、記録データを一元管理、省エネ法に準じた原油・ＣＯ２換算値をレポート形式で出力など。

「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ６３１５」を使用するには、別ＰＤＦファイルの「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ６３１５」用のインストールマニュアルを参考にして、アプリケーションとＵＳＢドライバをＰＣ：コンピュータへインストールしてください。
●インターフェース
本製品は USB、Bluetooth®インターフェースを装備しています。
通信方式：USB Ver2.0 準拠

Bluetooth®仕様：Bluetooth® Ver.5.0準拠

対応プロファイル：GATT®

USB、Bluetooth 通信で以下のことが行えます。
・本体の内部メモリ内のファイルをパソコンへダウンロード
・パソコンから本体の設定
・本体からリアルタイムで測定値を取得し、パソコン上で測定値とグラフを表示●パソコンの推奨動作環境
・OS（オペレーションシステム）
Windows 8/7/Vista/XP
WindowsのOSはCDケースのバージョンラベルを参照してください。
Ⓡ

・画面表示
解像度 1024×768 ドット、65536 色以上
・HDD（ハードディスク）空き容量 1Gbyte 以上(Framework を含む)
・.NET Framework(4.6.1以上)●商標について・Windows®は米国マイクロソフト社の商標です。
・Bluetooth®は Bluetooth SIG の商標です。

最新のソフトのダウンロードは、弊社ホームページから行うことができます。
www.kew-ltd.co.jp- 129 -

KEW6315

KEW6315

10.1 安全要求仕様

10章 仕様10.1 安全要求仕様
使用環境： 屋内使用高度 2000m 以下

確度保証温湿度範囲 ： 23℃±5℃

相対湿度 85%以下（結露しないこと）

使用温湿度範囲： 0℃～45℃

相対湿度 85%以下（結露しないこと）

保存温湿度範囲：-20℃～60℃

相対湿度 85%以下（結露しないこと）

耐電圧：

AC5160V／5 秒間(交流電圧入力端子)－(外装)間

AC3310V／5 秒間(交流電圧入力端子)－(電流入力端子、電源コネクタ、通信(USB)コネクタ)間AC2210V／5 秒間(電源コネクタ)－ (電流入力端子、通信(USB)コネクタ、外装)間

絶縁抵抗：50MΩ 以上／1000V

適合規格：IEC 61010-1

防塵/放水性：IEC 60529 IP40

環境規制規格：EN 50581(電圧／電流入力端子、電源コネクタ)－(外装)間

測定 CAT Ⅳ300V CAT Ⅲ600V CATⅡ1000V 汚染度 2

IEC 61010-031、IEC61326 Class A

10.2 一般仕様
測定ラインと入力ｃｈ：結線に関係しない電流 ch(A2～A4)は、独立して測定できる
入力ｃｈ

結線

電 圧

単相２線式－1 系統（1P2W-1）

VN-V1

A1

単相２線式－２系統（1P2W-2）

VN-V1

A1,A2

単相２線式－３系統（1P2W-3）

VN-V1

A1,A2,A3

単相２線式－４系統（1P2W-4）

VN-V1

A1,A2,A3,A4

単相３線式－1 系統（1P3W-1）

VN-V1,V2

A1,A2

単相３線式－２系統（1P3W-2）

VN-V1,V2

A1,A2,A3,A4

三相３線式－1 系統（3P3W-1）

VN-V1,V2

A1,A2

三相３線式－２系統（3P3W-2）

VN-V1,V2

A1,A2,A3,A4

三相３線式（3P3W3A）

V1-V2,V2-V3,V3-V1

A1,A2,A3

三相４線式（3P4W）

VN-V1,V2,V3

A1,A2,A3

使用表示：3.5 型 TFT 液晶 QVGA(320×RGB×240)

表示更新周期

：1 秒※

※

電 流

測定演算処理の関係から、画面上に実際の測定値が反映されるまでに、最大で２秒の遅延があります。ただし

記録データとタイムスタンプに遅延はありません。
バックライト

消灯：点灯状態から LCD ｷｰを押す
点灯：消灯状態から電源ｷｰを除くｷｰ操作を行う

電源品質測定：IEC 61000-4-30 Ed.2 Class S

外形寸法：175（Ｌ）×120（Ｗ）×68（Ｄ）mm

質量：約 900g（電池含む）

KEW6315- 130 -

10.2 一般仕様
付属品

KEW6315：電圧用測定ｺｰﾄﾞ MODEL7255(赤､白､青､黒､各１本（ワニグチ付き）） 1 ｾｯﾄ
電源ｺｰﾄﾞ MODEL7169 ····························································································· 1 本
USB ケーブル MODEL7219 ··················································································· 1 本
クイックマニュアル ······································································································· 1 冊
CD-ROM ·························································································································· 1 枚
設定・解析用 PC ｿﾌﾄｳｴｱ（KEW Windows for KEW6315）
取扱説明書ﾃﾞｰﾀ（PDF ﾌｧｲﾙ）
単３形ｱﾙｶﾘ乾電池（LR6）························································································ 6 個
SD カード M-8326-02 ······························································································· 1 枚
キャリングバッグ MODEL9125·············································································· 1 個
入力端子プレート ········································································································ 1 枚

オプション

識別マーカー ································· 各 4 本×8 色（赤/青/黄/緑/茶/灰/黒/白)：クランプセンサ
MODEL8128(
ｸﾗﾝﾌﾟｾﾝｻ
50A ﾀｲﾌﾟ φ24mm)
KEW8135 （ﾌﾚｷｼﾌﾞﾙｸﾗﾝﾌﾟｾﾝｻ 50A ﾀｲﾌﾟ φ75mm）
MODEL8127(
ｸﾗﾝﾌﾟｾﾝｻ
100A ﾀｲﾌﾟ φ24mm)
MODEL8126(
ｸﾗﾝﾌﾟｾﾝｻ
200A ﾀｲﾌﾟ φ40mm)
MODEL8125(
ｸﾗﾝﾌﾟｾﾝｻ
500A ﾀｲﾌﾟ φ40mm)
MODEL8124(
ｸﾗﾝﾌﾟｾﾝｻ 1000A ﾀｲﾌﾟ φ68mm)
KEW8129 ( ﾌﾚｷｼﾌﾞﾙｾﾝｻ 3000A ﾀｲﾌﾟ φ150mm) ※販売終了品
KEW8130 ( ﾌﾚｷｼﾌﾞﾙｾﾝｻ 1000A ﾀｲﾌﾟ φ110mm)
KEW8133 ( ﾌﾚｷｼﾌﾞﾙｾﾝｻ 3000A ﾀｲﾌﾟ φ170mm）
MODEL8146(
ﾘｰｸｾﾝｻ
10A ﾀｲﾌﾟ φ24mm)
MODEL8147(
ﾘｰｸｾﾝｻ
10A ﾀｲﾌﾟ φ40mm)
MODEL8148(
ﾘｰｸｾﾝｻ
10A ﾀｲﾌﾟ φ68mm)
MODEL8141(
ﾘｰｸｾﾝｻ
1A ﾀｲﾌﾟ φ24mm) ※販売終了品
MODEL8142(
ﾘｰｸｾﾝｻ
1A ﾀｲﾌﾟ φ40mm) ※販売終了品
MODEL8143(
ﾘｰｸｾﾝｻ
1A ﾀｲﾌﾟ φ68mm) ※販売終了品
クランプセンサの取扱説明書
マグネット付携帯ケース MODEL9132
電源供給アダプタ MODEL8312（CAT.Ⅲ150V、CAT.Ⅱ240V）

実時間確度：±5 秒／日以内

電源：AC 電源

電圧範囲

AC100V(AC90V)～AC240V(AC264V)

周波数50Hz(47Hz)～60Hz(63Hz)

消費電力

7VAmax：DC 電源

電圧

乾電池

充電式電池

DC3.0V (＝1.5V×2 直列×3 並列)

DC2.4V (＝1.2V×2 直列×3 並列)

使用電池

単３形ｱﾙｶﾘ電池(LR6)単3形ﾆｯｹﾙ水素電池(1900mA/h)

消費電流1.0A typ.(@3.0V)

1.1A typ.(@2.4V)

連続使用時間 ※23℃参考値

ﾊﾞｯｸﾗｲﾄ OFF：3 時間

ﾊﾞｯｸﾗｲﾄ OFF：4.5 時間 ※ﾌﾙ充電時

リアルタイム OS：

本製品は、T-Engine フォーラム(www.t-engine.org)の T-License に基づき T-Kernel ソースコードを利用しています。
本製品のソフトウェアの著作権の一部は、(c) 2010 The FreeType Project (www.freetype.org)のものです。すべての権利はその所有者に帰属します。

- 131 -

KEW6315

KEW6315外部通信

10.2 一般仕様：USB※

接続 USB ｹｰﾌﾞﾙ長は、2m 以下であること

ｺﾈｸﾀ形状mini-B

通信方式

USB Ver2.0 準拠

USB 認識番号

ﾍﾞﾝﾀﾞｰ ID:12EC(Hex)
ﾌﾟﾛﾀﾞｸﾄ ID:6315(Hex)
ｼﾘｱﾙ番号:0+7 桁機体番号

通信速度

12Mbps (ﾌﾙｽﾋﾟｰﾄﾞ)：Bluetooth

®

通信方式

Bluetooth® Ver.5.0準拠

ﾌﾟﾛﾌｧｲﾙ

GATT

周波数2402 ～ 2480MHz

変調方式

GFSK(1Mbps), π/4-DQPSK(2Mbps), 8DPSK(3Mbps)

伝送方式

周波数ﾎｯﾋﾟﾝｸﾞ方式

デジタル出力端子：

通常は High です。測定値が電源品質イベントしきい値を越えている間、出力端子を Low にします。イベント継続区
間が 1 秒未満であった場合には 1 秒間 Low になります。複数のイベント判定をオンにしている場合にはイベント設定している内の、最も優先順位が高いイベントのみが出力対象です。優先順位が低いイベントに対して出力する場合には、そのイベントよりも優先順位の高いイベント判定をオフしてください。
【最優先】→トランジェント→瞬停→ディップ→スウェル→インラッシュカレント
ｺﾈｸﾀ形状貫通型スクリューレス端子台 6 極(黒/赤/灰 ML800-S1H-6P)

出力形式

ｵｰﾌﾟﾝｺﾚｸﾀ出力、Low ｱｸﾃｨﾌﾞ

入力電圧0～30V、50mAmax、200mW

出力電圧

High：4.0V～5.0V Low：0.0～1.0V

KEW6315- 132 -

10.2 一般仕様
記録データ

KEW6315：内蔵 FLASH メモリ

記録容量4MB（ﾃﾞｰﾀ記録領域 3,437,500byte）

ﾃﾞｰﾀ容量14,623byte/data (最大記録件数：234 件)※

3P3W-2/1P3W-2(電力+高調波)設定時

最大 3 測定ﾃﾞｰﾀ ※記録ｽﾀｰﾄできる回数

保存可能ﾌｧｲﾙ数
表示記号

内蔵ﾒﾓﾘが記録先として有効な場合は､記録中のみのﾏｰｸが点灯する

FULL 表示
保存ﾃﾞｰﾀが､記録容量を超えた場合は、のﾏｰｸが点滅する

点灯している状態では記録できない
積算/ﾃﾞﾏﾝﾄﾞ測定は継続されるが記録しない：ＳＤカード対応容量

2GB(ﾃﾞｰﾀ記録領域 1.86Gbyte)

ﾃﾞｰﾀ容量(2GB)

14,623byte/data (最大記録件数：1,271,964 件)※

3P3W-2/1P3W-2(電力+高調波)設定時

保存可能ﾌｧｲﾙ数(2GB)
表示記号

最大 65536 測定ﾃﾞｰﾀ ※記録ｽﾀｰﾄできる回数

ﾌｫｰﾏｯﾄ形式（2GB）
FULL 表示

FAT16

SD ｶｰﾄﾞが記録先として有効な場合は､ﾏｰｸが点灯

保存ﾃﾞｰﾀが記録容量を超えた場合、保存可能ﾌｧｲﾙ数を越えた場合に

ﾏｰｸ

が点滅する
点灯している状態では記録できない
積算/ﾃﾞﾏﾝﾄﾞ測定は継続されるが記録しない

10.3 測定仕様
測定項目別、解析データ数
200ms(50Hz：10 周期、60Hz：12 周期)を 1 測定範囲として、8192 ﾎﾟｲﾝﾄのﾃﾞｰﾀで演算する項目
周波数、電圧実効値、電流実効値、有効電力、皮相電力、無効電力、力率、進相ｺﾝﾃﾞﾝｻ

200ms(50Hz：10 周期、60Hz：12 周期)を 1 測定範囲として、2048 ﾎﾟｲﾝﾄのﾃﾞｰﾀで演算する項目
電圧不平衡率、電流不平衡率、高調波電圧実効値(含有率)、高調波電流実効値(含有率)、高調波無効電力、
総合高調波電圧歪み率(THDV-F/R)、総合高調波電流歪み率(THDA-F/R)、高調波電圧位相角、高調波電流
位相角、高調波電圧電流位相差

半波ごとにｵｰﾊﾞｰﾗｯﾌﾟした 1 波形を 1 測定範囲として、50Hz ならば約 819 ﾎﾟｲﾝﾄ、60Hz ならば約 682 ﾎﾟｲﾝﾄで演算する項目
電圧ﾃﾞｨｯﾌﾟ、電圧ｽｳｪﾙ、瞬停、ｲﾝﾗｯｼｭｶﾚﾝﾄ

40.96ksps の瞬時測定値にて表示する項目
電圧波形、電流波形、外部入力電圧- 133 -

KEW6315

KEW6315

瞬時測定項目瞬時測定項目
周波数 f〔Hz〕
表示桁数

4桁

確度

±2dgt (40.00Hz～70.00Hz，V1 ﾚﾝｼﾞ 10%～110％，正弦波)

表示範囲10.00～99.99Hz

信号ｿｰｽ

V1 固定

10 秒平均周波数 f10〔Hz〕
表示桁数
測定方式
確度
表示範囲
信号ｿｰｽ

4 桁、ｲﾝﾀｰﾊﾞﾙ 10 秒時の周波数平均値がこれに該当する
IEC61000-4-30 に準じる
±2dgt (40.00Hz～70.00Hz，V1 ﾚﾝｼﾞ 10%～110％，正弦波)
10.00～99.99Hz
V1 固定

電圧実効値 V〔Vrms〕
ﾚﾝｼﾞ

600.0／1000V

表示桁数

4桁

有効入力範囲表示範囲

ﾚﾝｼﾞの 1%～120% (rms)および、ﾚﾝｼﾞの 200% (peak)
ﾚﾝｼﾞの 0.15%～130%(0.15%未満は 0 表示)

ｸﾚｽﾄﾌｧｸﾀ
測定方式
確度

3 以下
IEC61000-4-30 に準じる
測定波形 正弦波 40～70Hz において、600V ﾚﾝｼﾞにて
公称電圧 100V 以上に対し 10%～150% ：公称電圧±0.5%
上記範囲外と 1000V ﾚﾝｼﾞ：±0.2%rdg±0.2%f.s.
約 1.67MΩ

入力ｲﾝﾋﾟｰﾀﾞﾝｽ演算式⎛ 1 ⎛ n −1⎞⎞⎜⎜ ⎜ ∑ (V ci )2 ⎟ ⎟⎟⎠⎠⎝ n ⎝ i=0

Vc =i ：ｻﾝﾌﾟﾘﾝｸﾞﾎﾟｲﾝﾄ№※
n：10 周期,12 周期間でのｻﾝﾌﾟﾙ数
c ：測定ﾁｬﾝﾈﾙ

50Hz は、10 波形 8192 ポイント、60Hz は 12 波形 8192 ポイントで演算する※

1P2W-1～4

V1

1P3W-1～2

V1 、V2

3P3W-1～2

線間電圧 V12 、V23 、V31 = √(V23^2 ＋ V12^2＋2×V23×V12×cosθV)θV=V12 , V23 の相対角※

3P3W3A

線間電圧 V12 、V23 、V31

3P4W

相電圧

V1 、V2 、V3線間電圧 V12 = √( V1^2 + V2^2－2×V1×V2×cosθV1 )V23 = √( V2^2 + V3^2－2×V2×V3×cosθV2 )V31 = √( V3^2 + V1^2－2×V3×V1×cosθV3 )θV1 = V1 ,V2 の相対角、θV2 = V2 ,V3 の相対角、θV1 = V3 ,V1 の相対角※

KEW6315- 134 -

瞬時測定項目

KEW6315

電流実効値 A〔Arms〕
ﾚﾝｼﾞ

表示桁数
有効入力範囲表示範囲
ｸﾚｽﾄﾌｧｸﾀ
測定方式
確度
入力ｲﾝﾋﾟｰﾀﾞﾝｽ
演算式

MODEL8128/KEW8135 (50A ﾀｲﾌﾟ)：5000m／50.00A／AUTO
MODEL8127(100A ﾀｲﾌﾟ)：10.00／100.0A／AUTO
MODEL8126(200A ﾀｲﾌﾟ)：20.00／200.0A／AUTO
MODEL8125(500A ﾀｲﾌﾟ)：50.00／500.0A／AUTO
MODEL8124/KEW8130 (1000A ﾀｲﾌﾟ) ：100.0／1000A／AUTO
MODEL8141/8142/8143 (1A ﾀｲﾌﾟ)：500.0mA
MODEL8146/8147/8148 (10A ﾀｲﾌﾟ)：1000m／10.00A／AUTO
KEW8129(3000A ﾀｲﾌﾟ) ：300.0／1000／3000A
KEW8133(3000A ﾀｲﾌﾟ) ：300.0／3000A／AUTO
4桁
各ﾚﾝｼﾞの 1%～110% (rms)およびﾚﾝｼﾞの 200% (peak)
各ﾚﾝｼﾞの 0.15%～130%(0.15%未満は 0 表示)
3 以下
IEC61000-4-30 に準じる
測定波形 正弦波 40～70Hz において±0.2%rdg±0.2%f.s.＋ｸﾗﾝﾌﾟｾﾝｻ精度
約 100kΩ⎛1⎛n −1⎞
2⎞
Ac = ｜ ｜∑ ( Aci ) ｜｜⎠⎝n⎝i = 0⎠c ：測定ﾁｬﾝﾈﾙ A1 、A2 、A3 、A4i ：ｻﾝﾌﾟﾘﾝｸﾞﾎﾟｲﾝﾄ№※
n：10 周期,12 周期間でのｻﾝﾌﾟﾙ数※

50Hz は 10 波形 8192 ポイント、60Hz は 12 波形 8192 ポイントで演算する※

3P3W-1～2 結線時の A3 は電流実効値より算出する
A3 = √(A1^2 ＋ A2^2＋2×A1×A2×cosθA)- 135 -

θA = A1 , A2 の相対角

KEW6315

KEW6315

瞬時測定項目

有効電力 P〔W〕
ﾚﾝｼﾞ
8128/8135

電流

8127

8126

50.00A

5000mA

100.0A

10.00A

200.0A

20.00A

1000V

50.00k

5000

100.0k

10.00k

200.0k

20.00k

600.0V

30.00k

3000

60.00k6000

120.0k12.00k

電圧

8125

電流

8124/30

8146/47/48

500.0A

50.00A

1000A

100.0A

10.00A

1000mA

1000V

500.0k

50.00k

1000k

100.0k

10.00k1000

600.0V

300.0k30.00k

600.0k

60.00k

6000

600.0

電圧

電流

8141/42/43

8129

8133

500.0mA

3000A

1000A

300.0A

3000A

300.0A

1000V

500.0

3000k

1000k

300.0k

3000k

300.0k

600.0V

300.0

1800k

600.0k

180.0k1800k

180.0k

電圧

表示桁数

4桁

確度

±0.3%rdg±0.2%f.s.＋ｸﾗﾝﾌﾟｾﾝｻ精度 (力率 1，正弦波，40～70Hz)
Sum 値は使用ﾁｬﾝﾈﾙの総合値

※

力率の影響

±1.0%rdg (40Hz～70Hz、力率 0.5)

極性表示
演算式

消費(流入)：＋(符号無し)，

回生(流出)：－

1⎛⎞
Pc = ｜∑ (Vci × Aci)｜
n⎝i=0
⎠
n−1

c:測定ﾁｬﾝﾈﾙ
i:ｻﾝﾌﾟﾘﾝｸﾞﾎﾟｲﾝﾄ№ ※n:ｻﾝﾌﾟﾙ数

50Hz は 10 波形 8192 ポイント、60Hz は 12 波形 8192 ポイントで演算する※

1P2W-1～4

P1 、P2 、P3 、P4 、Psum＝P1+P2+P3+P4

1P3W(3P3W)-1～2

P1 、P2 、Psum1＝P1+P2
P3 、P4 、Psum2＝P3+P4
Psum＝Psum1+Psum2

3P3W3A

P1 、P2 、P3 、Psum＝P1+P2+P3

3P4W

P1 、P2 、P3 、Psum＝P1+P2+P3※

相電圧を使用する

外部入力電圧 DCi〔V〕
ﾚﾝｼﾞ

100.0mV／1000mV／10.00V

表示桁数

4桁

有効入力範囲各ﾚﾝｼﾞの 1%～±100% (DC)

表示範囲

ﾚﾝｼﾞの 0.3%～±110% (0.3%未満は 0 表示)

確度±0.5%f.s (DC)

入力ｲﾝﾋﾟｰﾀﾞﾝｽ

約 225.6kΩ

保存項目

外部入力電圧

KEW6315- 136 -

演算項目

KEW6315

演算項目
皮相電力 S〔VA〕
ﾚﾝｼﾞ

有効電力に同じ

表示桁数

有効電力に同じ

確度

各測定値からの演算に対して±1dgt(合計値は±3dgt)

符号

極性なし

演算式

Sc＝Vc×Ac

1P2W-1～4

S1 、S2 、S3 、S4 、Ssum＝S1+S2+S3+S4

1P3W-1～2

S1 、S2 、Ssum1＝S1+S2

Pc＞Sc 時は Pc＝Sc とする

c：測定ﾁｬﾝﾈﾙ

S3 、S4 、Ssum2＝S3+S4
Ssum＝Ssum1+Ssum2
3P3W-2

S1 、S2 、Ssum1＝√3／2( S1+S2 )S3 、S4 、Ssum2＝√3／2( S3+S4 )Ssum＝Ssum1+Ssum2

3P3W3A

S1 、S2 、S3 、Ssum＝S1+S2+S3

3P4W

S1 、S2 、S3 、Ssum＝S1+S2+S3※

相電圧を使用する

無効電力 Q〔Var〕
ﾚﾝｼﾞ
表示桁数
確度
符号

有効電力に同じ
有効電力に同じ
各測定値からの演算に対して±1dgt(合計値は±3dgt)－符号：進位相(電圧に対する電流位相)＋符号(符号無) ：遅位相(〃)
極性符号は、測定ﾁｬﾝﾈﾙ毎に高調波無効電力を求め、基本波の逆の符号を付加する

演算式

Qc＝ sign

1P2W-1～4

Q1 、Q2 、Q3 、Q4 、Qsum＝Q1+Q2+Q3+Q4

1P3W(3P3W)-1～2

Q1 、Q2 、Qsum1＝Q1+Q2

Sc 2 − Pc 2

sign：極性符号、c：測定ﾁｬﾝﾈﾙ

Q3 、Q4 、Qsum2＝Q3+Q4
Qsum＝Qsum1+Qsum2
3P3W3A(3P4W)

Q1 、Q2 、Q3 、Qsum＝Q1+Q2+Q3- 137 -

KEW6315

KEW6315

演算項目

力率 PF
表示範囲

-1.000～0.000～1.000

確度

各測定値からの演算に対して±1dgt(合計値は±3dgt)

符号

－符号

：進位相

＋符号(符号無)：遅位相※極性符号は、測定ﾁｬﾝﾈﾙ毎に高調波無効電力を求め、基本波の逆の符号を付加する
演算式

PF c = sign P cSc

sign：極性符号、c：測定ﾁｬﾝﾈﾙ、

1P2W-1～4

PF1 、PF2 、PF3 、PF4 、PFsum

1P3W(3P3W)-1～2

PF1 、PF2 、PFsum1
PF3 、PF4 、PFsum2
PFsum

3P3W3A(3P4W)

PF1 、PF2 、PF3 、PFsum

中性電流 An〔A〕※

3P4W の結線時のみ

ﾚﾝｼﾞ

電流実効値に同じ

表示桁数

電流実効値に同じ

表示範囲

電流実効値に同じ

演算式
An＝
※

{A1 + A2 cos(θ 2 − θ 1) + A3 cos(θ 3 − θ 1)}2 + {A2 sin (θ 2 − θ 1) + A3 sin(θ 3 − θ 1)}2

θ1,2,3 は、それぞれ V1 と A1,2,3 の位相差を示す

電圧不平衡率 Uunb [%]
表示桁数

5桁

表示範囲

0.00%～100.00%

結線

3P3W、3P4W

測定方式

IEC61000-4-30 に準じる

確度

測定波形

正弦波 50/60Hz おいて ±0.3%(IEC61000-4-30 の試験にて 0%～5%の範囲)
演算式⎛1−
Vumb = ⎜⎜1+⎝(3 − 6β ) ⎞⎟× 100
(3 − 6β ) ⎟⎠β=

V124 + V234 + V314

(V + V + V )212高調波電圧の 1 次成分を使用する※

3P4W では、相電圧を線間電圧に変換し演算する※

V12 = V1－V2 、V23 = V2－V3 、V31 = V3－V1

KEW6315- 138 -

2
23

2 2
31

演算項目

KEW6315電流不平衡率 Aunb [%]
表示桁数

5桁

表示範囲

0.00%～100.00%

結線
演算式

3P3W、3P4W⎛ 1−
Iumb = ⎜⎜1+⎝(3 − 6β ) ⎞⎟×100
(3 − 6β ) ⎟⎠β=

A124 + A234 + A314

(A + A + A )
2
12

2
23

2 2
31

高調波電流の 1 次成分を使用する※

3P4W では相電流を線間電流に変換し演算する※

進相コンデンサ

A12 = A1－A2 、A23 = A2－A3 、A31 = A3－A1

表示桁数

4 桁、単位：nF、μF、mF、kvar

表示範囲
演算式

0.000nF～9999F、0.000kvar～9999kvar⎛
1
C C = PC × ⎜−1 −⎜ PF 2C⎝=⎞⎟ [k var ]1−
2⎟
PFC _ T arg et
⎠
1

⎛
⎞
1
1
⎜
⎟ [μF ]
×
−
−
−
1
1
2
2
2
⎜
⎟
2πf × VCPFC _ T arg et⎝ PFC⎠
PC × 10 9

Cc：改善に必要なコンデンサ容量

Pc ：負荷電力(有効電力)〔kW〕
f ：周波数Vc ：電圧実効値
PFc ：測定した力率
PFc_Target ：改善後の力率(目標)c ：測定ﾁｬﾝﾈﾙ
1P2W-1～4

C1、C2、C3、C4、Csum＝C1+C2+C3+C4

1P3W(3P3W)-1～2

C1、C2、Csum1＝C1+C2
C1、C2、Csum2＝C3+C4
Csum＝Csum1+ Csum2

3P3W3A(3P4W)

C1、C2、C3、Csum＝C1+C2+C3- 139 -

KEW6315

KEW6315

積算測定項目

積算測定項目消費電力（P≧0 の場合）
有効電力量+WP〔Wh〕
表示桁数
表示範囲

6 桁、単位：m、k、M、G、T ( + WS にあわせる)

0.00000mWh ～ 9999.99TWh ( + WS にあわせる)※表示範囲を越えた場合は OL を表示する

演算式

⎞
1⎛
+ WPc = ⎜⎜ ∑ (+ Pci )⎟⎟h⎝ i⎠h：積算時間(3600 秒)、c：測定ﾁｬﾝﾈﾙ、i：ﾃﾞ－ﾀﾎﾟｲﾝﾄ№

1P2W-1～4+WP1 、+WP2 、+WP3 、+WP4 、+WPsum

1P3W(3P3W)-1～2+WP1 、+WP2 、+WPsum1+WP3 、+WP4 、+WPsum2+WPsum

3P3W3A(3P4W)+WP1 、+WP2 、+WP3 、+WPsum

皮相電力量＋WS〔VAh〕
表示桁数
表示範囲

6 桁、単位：m、k、M、G、T ( + WS にあわせる)

0.00000mVAh ～ 9999.99TVAh ( + WS にあわせる)※表示範囲を越えた場合は OL を表示する

演算式+ WSc =⎞
1⎛⎜⎜ ∑ (Sci )⎟⎟h⎝ i⎠h：積算時間(3600 秒)、c：測定ﾁｬﾝﾈﾙ、i：ﾃﾞ－ﾀﾎﾟｲﾝﾄ№、
1P2W-1～4+WS1 、+WS2 、+WS3 、+WS4 、+WSsum

1P3W(3P3W)-1～2+WS1 、+WS2 、+WSsum1+WS3 、+WS4 、+WSsum2+WSsum

3P3W3A(3P4W)+WS1 、+WS2 、+WS3 、+WSsum

保存項目

皮相電力量値

KEW6315- 140 -

積算測定項目

KEW6315

無効電力量＋WQ〔Varh〕
表示桁数
表示範囲

6 桁、単位：m、k、M、G、T ( + WS にあわせる)

0.00000mvarh ～ 9999.99Tvarh ( + WS にあわせる)
表示範囲を越えた場合はを OL 表示する

※

演算式

進位相

遅位相

⎞
1⎛
+ WQc _ c = ⎜⎜ ∑ (+ Qci )⎟⎟ 、
h⎝ i
⎠
⎞
1⎛
+ WQi _ c = ⎜⎜ ∑ (− Qci )⎟⎟ 、h⎝ i⎠h：積算時間(3600 秒)、n：:系統№、c：測定ﾁｬﾝﾈﾙ、i：ﾃﾞ－ﾀﾎﾟｲﾝﾄ№、※

遅位相：Q ≧0 の時，進位相：Q ＜0 の時

1P2W-1～4+WQ1 、+WQ2 、+WQ3 、+WQ4 、+WQsum

1P3W(3P3W)-1～2+WQ1 、+WQ2 、+WQsum1+WQ3 、+WQ4 、+WQsum2+WQsum

3P3W3A(3P4W)+WQ1 、+WQ2 、+WQ3 、+WQsum

回生電力（P＜0 の場合）
有効電力量－WP〔Wh〕
表示桁数
表示範囲

6 桁、単位：m、k、M、G、T ( + WS にあわせる)

0.00000mWh ～ 9999.99TWh ( + WS にあわせる)※

表示範囲を越えた場合は OL を表示する

演算式− WPc =⎞
1⎛⎜⎜ ∑ (− Pci )⎟⎟h⎝ i⎠h：積算時間(3600 秒)、c：測定ﾁｬﾝﾈﾙ、i：ﾃﾞ－ﾀﾎﾟｲﾝﾄ№
1P2W-1～4
1P3W(3P3W)-1～2-WP1 、-WP2 、-WP3 、-WP4 、-WPsum-WP1 、-WP2 、-WPsum1-WP3 、-WP4 、-WPsum2-WPsum

3P3W3A(3P4W)-WP1 、-WP2 、-WP3 、-WPsum- 141 -

KEW6315

KEW6315

積算測定項目

皮相電力量－WS〔VAh〕
表示桁数
表示範囲

6 桁、単位：m、k、M、G、T ( + WS にあわせる)

0.00000mVAh ～ 9999.99TVAh ( + WS にあわせる)
表示範囲を越えた場合は OL を表示する※

演算式− WSc =⎞
1⎛⎜⎜ ∑ (Sci )⎟⎟h⎝ i⎠h：積算時間(3600 秒)、c：測定ﾁｬﾝﾈﾙ、i：ﾃﾞ－ﾀﾎﾟｲﾝﾄ№、
1P2W-1～4-WS1 、-WS2 、-WS3 、-WS4 、-WSsum

1P3W(3P3W)-1 ～-WS1 、-WS2 、-WSsum1

2-WS3 、-WS4 、-WSsum2-WSsum

3P3W3A(3P4W)-WS1 、-WS2 、-WS3 、-WSsum

無効電力量－WQ〔Varh〕
表示桁数
表示範囲

6 桁、単位：m、k、M、G、T ( + WS にあわせる)

0.00000mvarh ～ 9999.99Tvarh ( + WS にあわせる)
表示範囲を越えた場合は OL を表示する

※

演算式

進位相

遅位相

⎞
1⎛
− WQc _ c = ⎜⎜ ∑ (+ Qci )⎟⎟ 、
h⎝ i
⎠
⎞
1⎛
− WQi _ c = ⎜⎜ ∑ (− Qci )⎟⎟h⎝ i⎠h：積算時間(3600 秒)、n：:系統№、c：測定ﾁｬﾝﾈﾙ、i：ﾃﾞ－ﾀﾎﾟｲﾝﾄ№、※

遅位相：Q ≧0 の時，進位相：Q ＜0 の時

1P2W-1～4－WQ1 、－WQ2 、－WQ3 、－WQ4 、－WQsum

1P3W(3P3W)-1～2－WQ1 、－WQ2 、－WQsum1－WQ3 、－WQ4 、－WQsum2－WQsum

3P3W3A(3P4W)－WQ1 、－WQ2 、－WQ3 、－WQsum

積算時間
表示範囲

KEW631500:00:00 (0 秒)～ 99:59:59 (99 時間 59 分 59 秒) ,0100:00～ 9999:59 (9999 時間 59 分) ,010000～ 999999 (999999 時間)- 142 -※

表示は順次遷移する

ＤＥＭＡＮＤ測定項目

KEW6315

ＤＥＭＡＮＤ測定項目
目標値(DEMTarget)
表示桁数

4桁

表示単位

m、k、M、G、T

表示範囲

0.000mW(VA)～ 999.9TW(VA) ※設定値に固定する予測値(DEMGuess)
表示桁数

6桁

表示単位

m、k、M、G、T (DEMTarget にあわせる)

表示範囲

0.00000mW(VA)～99999.9TW(VA)
小数点位置を DEMTarget にあわせる※

表示範囲を越えた場合は OL を表示する※

演算式

DEM Guess = ΣDEM ×

測定周期時間測定開始から現在までの経過時間

現在値・ﾃﾞﾏﾝﾄﾞ測定値(ΣDEM)
表示桁数

6 桁、単位：m、k、M、G、T ※DEMTarget にあわせる

表示単位

m、k、M、G、T ※DEMTarget にあわせる

表示範囲

0.00000mW(VA)～99999.9TW(VA) ※小数点位置を DEMTarget にあわせる
表示範囲を越えた場合は OL を表示する

※

演算式

ΣDEM＝
１時間
（測定開始から現在までの"+WPsum (+WSsum)"の積算値）×

測定周期時間

負荷率
表示桁数

6桁

表示範囲

0.00～9999.99% ※表示範囲を越えた場合は OL を表示する

演算式

Σ DEM

DEM Terget

予測
表示桁数

6桁

表示範囲

0.00～9999.99% ※表示範囲を越えた場合は OL を表示する

演算式

DEM Guess

DEM Terget- 143 -

KEW6315

KEW6315

高調波測定項目

高調波測定項目
測定方式：ﾃﾞｼﾞﾀﾙ PLL 同期方式

測定方法

：高調波解析後の、整数次に隣接する中間高調波成分を加算して表示する

有効周波数範囲：40～70Hz

解析次数：1～50 次ｳｲﾝﾄﾞｳ幅：50Hz は 10 周期、60Hz は 12 周期

ｳｲﾝﾄﾞｳの種類：ﾚｸﾀﾝｷﾞｭﾗ

解析ﾃﾞｰﾀ数：2048 ﾎﾟｲﾝﾄ

解析ﾚｰﾄ：50Hz/60Hz にて、1 回/200m 秒

高調波電圧実効値 Vk〔Vrms〕
ﾚﾝｼﾞ

電圧実効値に同じ

表示桁数

電圧実効値に同じ

表示範囲

電圧実効値に同じ ※含有率 0.0%～100.0% 基本波に対する割合

測定方式

IEC61000-4-30、IEC61000-4-7、IEC61000-2-4 に準じる
解析ｳｲﾝﾄﾞｳ幅は 50/60Hz に対して 10/12 ｻｲｸﾙである
測定値は解析次数に隣接する中間高調波成分を含んでいる

確度

600V ﾚﾝｼﾞの 10%～100%の入力範囲で IEC61000-2-4Class3 の確度を規定する
公称電圧 100V 以上に対し 3%以上：±10%rdg

公称電圧 100V 以上に対し 3%未満

：公称電圧±0.3%

1000V ﾚﾝｼﾞ：±0.2%rdg±0.2%f.s.

演算式

V ck =

1∑ (Vc (10 k + n )r ) + (Vc (10 k + n )i )2

n = −1

2

含有率 =V ck × 100
V c1

c: 測定ﾁｬﾝﾈﾙ、k: 高調波の次数、
Vr: 電圧 FFT 変換後の実数成分、Vi: 電圧 FFT 変換後の虚数成分
演算式の測定周期は 10 周期であり、12 周期ならば式中の"10k+n"を"12k+n"に置き換える
1P2W-1～4

V1k

1P3W-1～2

V1k，V2k

3P3W-1～2

線間電圧 V12k 、V32k

3P3W3A

線間電圧 V12k 、V23k 、V31k

3P4W

V1k，V2k，V3k

KEW6315- 144 -

高調波測定項目

KEW6315高調波電流実効値 Ak〔Arms〕
ﾚﾝｼﾞ

電流実効値に同じ

表示桁数

電流実効値に同じ

表示範囲

電流実効値に同じ

測定方式

IEC61000-4-7、IEC61000-2-4 に準じる※

含有率

0.0%～100.0% 基本波に対する割合

解析ｳｲﾝﾄﾞｳ幅は 50/60Hz に対して 10/12 ｻｲｸﾙ、測定値は解析次数に隣接する中間
高調波成分を含んでいる
確度

測定ﾚﾝｼﾞの 10%～100%の入力範囲で IEC61000-2-4Class3 の確度を規定する
ﾚﾝｼﾞ最大に対し 10%以上：±10%rdg + ｸﾗﾝﾌﾟｾﾝｻ精度

ﾚﾝｼﾞ最大に対し 10%未満：ﾚﾝｼﾞ最大±1.0% + ｸﾗﾝﾌﾟｾﾝｻ精度

演算式

A ck =

1∑ ( Ac (10 k + n )r ) + ( Ac (10 k + n )i )
2

2

含有率 =

n = −1

A ck × 100
A c1

c: 測定ﾁｬﾝﾈﾙ A1k 、A2k 、A3k 、A4k 、k: 高調波の次数r: FFT 変換後の実数成分、i: FFT 変換後の虚数成分
演算式の測定周期は 10 周期であり、12 周期ならば式中の"10k+n"を"12k+n"に置き換える

高調波電力 Pk〔W〕
ﾚﾝｼﾞ

有効電力に同じ

表示桁数

有効電力に同じ

表示範囲

有効電力に同じ

測定方式

IEC61000-4-7 に準じる

確度

±0.3%rdg±0.2%f.s.＋ｸﾗﾝﾌﾟｾﾝｻ精度 (力率 1，正弦波，50／60Hz)※

含有率

0.0%～100.0%

基本波の絶対値に対する割合(Sum 値は使用ﾁｬﾝﾈﾙの総合値)
演算式

Pck＝Vc(10k)r×Ac(10k)r－Vc(10k)i×Ac(10k)i

含有率 =P ck × 100
P c1

c: 測定ﾁｬﾝﾈﾙ、k: 高調波の次数、r: FFT 変換後の実数成分、
i: FFT 変換後の虚数成分
上記演算式の測定周期は 10 周期であり、12 周期ならば式中の"(10k)"を"(12k)"に置き換える
1P2W-1～4

P1k 、P2k 、P3k 、P4k 、Psumk＝P1k+P2k+P3k+P4k

1P3W-1～2

P1k 、P2k 、Psum1k＝P1k+P2k
P3k 、P4k 、Psum2k＝P3k+P4k
Psumk＝Psum1k+Psum2k

3P3W-1～2

P1k 、P2k 、Psum1k＝P1k+P2k
P3k 、P4k 、Psum2k＝P3k+P4k
Psumk＝Psum1k+Psum2k

3P3W3A

相電圧

P1k：V1 = ( V12－V31 )／3 、P2k：V2 = ( V23－V12 )／3 、
P3k：V3 = ( V31－V23 )／3 、Psumk＝P1k+P2k+P3k

3P4W

P1k 、P2k 、P3k 、Psumk＝P1k+P2k+P3k- 145 -

KEW6315

KEW6315

高調波測定項目

高調波無効電力 Qk〔var〕(内部演算のみに使用)
演算式

Pck＝Vc(10k)r×Ac(10k)i－Vc(10k)i×Ac(10k)rc: 測定ﾁｬﾝﾈﾙ、k: 高調波の次数、r: FFT 変換後の実数成分、
i: FFT 変換後の虚数成分
上記演算式の測定周期は 10 周期であり、12 周期ならば式中の"(10k)"を"(12k)"に置き換える

1P2W-1～4

Q1k 、Q2k 、Q3k 、Q4k 、Qsumk＝Q1k+Q2k+Q3k+Q4k

1P3W-1～2

Q1k 、Q2k 、Qsum1k＝Q1k+Q2k
Q3k 、Q4k 、Qsum2k＝Q3k+Q4k
Qsumk＝Qsum1k+Qsum2k

3P3W-1～2

Q1k 、Q2k 、Qsum1k＝Q1k+Q2k
Q3k 、Q4k 、Qsum2k＝Q3k+Q4k
Qsumk＝Qsum1k+Qsum2k

3P3W3A

相電圧

Q1k：V1 = ( V12－V31 )／3 、Q2k：V2 = ( V23－V12 )／3 、
Q3k：V3 = ( V31－V23 )／3 、Qsumk＝Q1k+Q2k+Q3k

3P4W

Q1k 、Q2k 、Q3k 、Qsumk＝Q1k+Q2k+Q3k

総合高調波電圧歪み率 THDVF [%]
表示桁数

4桁

表示範囲
演算式

0.0%～100.0%
50

THDVF c =∑ (V ck ) × 100
2

k =2

V c1

c: 測定ﾁｬﾝﾈﾙ
V: 高調波電圧k: 高調波の次数

1P2W-1～4

THDVF1

1P3W-1～2

THDVF1 、THDVF2

3P3W-1～2

線間電圧 THDVF12 、THDVF32

3P3W3A

線間電圧 THDVF12 、THDVF23 、THDVF31

3P4W

THDVF1 、THDVF2 、THDVF3

総合高調波電流歪み率 THDAF [%]
表示桁数

4桁

表示範囲
演算式

0.0%～100.0%
50

THDAF c =KEW63152∑ ( Ack ) × 100
k =2

A c1

- 146 -

c: 測定ﾁｬﾝﾈﾙ THDAF1 、THDAF2 、THDAF3 、THDAF4
A: 高調波電流k: 高調波の次数高調波測定項目

KEW6315

総合高調波電圧歪み率 THDVR [%]
表示桁数

4桁

表示範囲
演算式

0.0%～100.0%
50

2∑ (Vck ) × 100
k =2

THDVR c =50

c: 測定ﾁｬﾝﾈﾙ
V: 高調波電圧k: 高調波の次数∑ (Vck )2

k =1

1P2W-1～4

THDVR1

1P3W-1～2

THDVR1 、THDVR2

3P3W-1～2

線間電圧 THDVR12 、THDVR32

3P3W3A

線間電圧 THDVR12 、THDVR23 、THDVR31

3P4W

THDVR1 、THDVR2 、THDVR3

総合高調波電流歪み率 THDAR [%]
表示桁数

4桁

表示範囲
演算式

0.0%～100.0%50∑ ( A ck ) × 100
2k =2

THDAR c =50

∑ ( A ck )2

c: 測定ﾁｬﾝﾈﾙ THDAR1 、THDAR2 、
THDAR3 、THDAR4
A: 高調波電流k: 高調波の次数k =1

高調波電圧位相角 θVk [deg]
表示桁数

4桁

表示範囲
演算式

0.0°～±180.0°⎧ V ckr ⎫⎬⎩ − V cki ⎭θ V ck = tan −1 ⎨c： 測定ﾁｬﾝﾈﾙ
V：高調波電圧k：高調波の次数r：FFT 変換後の実数成分
i：FFT 変換後の虚数成分

1P2W-1～4θV1 k

1P3W-1～2

θV1k 、θV2k

3P3W-1～2

θV12k 、θV32k ※線間電圧を使用する

3P3W3AθV12k 、θV23k 、θV31k

3P4W

θV1k 、θV2k 、θV3k※線間電圧を使用する- 147 -

KEW6315

KEW6315

高調波測定項目

総合高調波電流位相角 θAk [deg]
表示桁数

4桁

表示範囲
演算式

0.0°～±180.0°⎧ A ckr ⎫⎬⎩ − A cki ⎭θ A ck = tan −1 ⎨c: 測定ﾁｬﾝﾈﾙ θA1k 、θA2k 、θA3k 、θA4k
A: 高調波電流k: 高調波の次数r: FFT 変換後の実数成分
i: FFT 変換後の虚数成分

高調波電圧電流位相差 θk [deg]
表示桁数

4桁

表示範囲

0.0°～±180.0°

演算式
1P2W-1～4

1P3W(3P3W)-1～2

3P3W3A(3P4W)-1

KEW6315

θck = θAck − θVckc: 測定ﾁｬﾝﾈﾙ、k: 高調波の次数⎧ Q sumk ⎫⎨⎬⎩ P sumk ⎭−1 ⎧ Q sum 1k ⎫
θ1k 、θ2k 、 θ sum 1k = tan ⎨⎬⎩ P sum 1k ⎭−1 ⎧ Q sum 2 k ⎫θ3k 、θ4k 、 θ sum 2 k = tan ⎨⎬⎩ P sum 2 k ⎭⎧ Q sumk ⎫θ sumk = tan −1 ⎨⎬⎩ P sumk ⎭−1 ⎧ Q sumk ⎫θ1k 、θ2k 、θ3k 、 θ sumk = tan ⎨⎬⎩ P sumk ⎭θ1k 、θ2k 、θ3k 、θ4k 、 θ sumk = tan- 148 -

−1

電源品質測定項目

KEW6315

電源品質測定項目
電圧トランジェント
測定方式
表示桁数
有効入力範囲
表示範囲
確度
入力ｲﾝﾋﾟｰﾀﾞﾝｽ
しきい値
検出ﾁｬﾝﾈﾙ（ｃｈ）
1P2W-1～4
1P3W-1～2
3P3W-1～2
3P3W3A
3P4W

約 40.96ksps (24μs 間隔)でｷﾞｬｯﾌﾟ無しに、ｲﾍﾞﾝﾄの有無を判定する(50Hz/60Hz)
4桁
50V～2200V (DC)50V～2200V (DC)
0.5%rdg ※1000V (DC)にて規程
約 1.67MΩ
絶対値ﾋﾟｰｸ電圧を指定する
V1
V1 、V2線間電圧 V12 、V32
線間電圧 V12 、V23 、V31
V1 、V2 、V3

電圧スウェル・ディップ・瞬停
ﾚﾝｼﾞ

電圧実効値に同じ

表示桁数

電圧実効値に同じ

有効入力範囲
表示範囲

電圧実効値に同じ
電圧実効値に同じ

ｸﾚｽﾄﾌｧｸﾀ
入力ｲﾝﾋﾟｰﾀﾞﾝｽ

電圧実効値に同じ
電圧実効値に同じ

しきい値
測定方式

公称電圧に対する%にて指定する
IEC61000-4-30 に準じる
半波ごとにｵｰﾊﾞｰﾗｯﾌﾟした 1 波形で実効値を算出する
多相ｼｽﾃﾑ ｽｳｪﾙ、ﾃﾞｨｯﾌﾟ 判定条件：
何れかの一つの ch がｲﾍﾞﾝﾄを開始した時にｲﾍﾞﾝﾄ開始、全ての ch がｲﾍﾞﾝﾄを終了した
時にｲﾍﾞﾝﾄ終了とする
多相ｼｽﾃﾑ 瞬停 判定条件：
全ての ch がｲﾍﾞﾝﾄを開始した時にｲﾍﾞﾝﾄ開始、何れかの一つの ch がｲﾍﾞﾝﾄを終了した
時にｲﾍﾞﾝﾄ終了とする
公称電圧 100V 以上に対し 10%～150%：公称電圧±1.0%
上記範囲外：±0.4%rdg±0.4%f.s.
周波数 40～70Hz においてｲﾍﾞﾝﾄ継続時間の測定誤差 ：1 周期以内

確度

検出ﾁｬﾝﾈﾙ（ｃｈ）
1P2W-1～4

V1

1P3W-1～2

V1 、V2

3P3W-1～2

線間電圧 V12 、V32

3P3W3A

線間電圧 V12 、V23 、V31

3P4W

V1 、V2 、V3- 149 -

KEW6315

KEW6315

電源品質測定項目

インラッシュカレント
ﾚﾝｼﾞ

電流実効値に同じ

表示桁数

電流実効値に同じ

有効入力範囲

電流実効値に同じ

表示範囲

電流実効値に同じ

ｸﾚｽﾄﾌｧｸﾀ

電流実効値に同じ

入力ｲﾝﾋﾟｰﾀﾞﾝｽ

電流実効値に同じ

しきい値

ﾚﾝｼﾞに対する%にて指定する

測定方式

半波ごとにｵｰﾊﾞｰﾗｯﾌﾟした 1 波形で実効値を算出する

確度±0.4%rdg±0.4%f.s.＋ｸﾗﾝﾌﾟｾﾝｻ精度

検出ﾁｬﾝﾈﾙ（ｃｈ）

A1 、A2 、A3 、A4

KEW6315- 150 -

電源品質測定項目

KEW6315

フリッカ
表示項目

Pst 残算出時間： 次の Pst 算出までの残り時間
V： 半波毎の電圧実効値、1 秒間の平均値
Pst(1min)： 1 分間のﾌﾘｯｶ値(Pst 参考値)
Pst： 短期間(10 分間)ﾌﾘｯｶの厳しさ
Plt： 長期間(2 時間)ﾌﾘｯｶの厳しさ
最大 Pst： Pst 最大値および、更新時間
最大 Plt： Plt 最大値および、更新時間
Pst(1min)最新 120 分間のﾄﾚﾝﾄﾞｸﾞﾗﾌ
Plt 最新 600 時間のﾄﾚﾝﾄﾞｸﾞﾗﾌ

表示桁数

4桁

分解能：対数 0.001～6400 P.U.を 1024 分割

ﾗﾝﾌﾟﾓﾃﾞﾙ

230V ﾗﾝﾌﾟ／220V ﾗﾝﾌﾟ／120V ﾗﾝﾌﾟ／100V ﾗﾝﾌﾟ

測定方式

IEC61000-4-30 および IEC61000-4-15 Ed.2 に準じる

確度

IEC61000-4-15 Ed.2 Class F3 の試験方法において Pst(最大 20)：±10%rdg

演算式
Pst(1min)c 、Pstc＝0 . 0314 × P 0 .1 + 0 . 0 525 × P 1 S + 0 . 0 657 × P 3 S + 0 . 28 × P 10 S + 0 . 08 × P 50 S
V1S＝(P0.7+P1+P1.5)／3 、V3S＝(P2.2+P3+P4)／3 、V10S＝(P6+P8+P10+P13+P17)／5 、
V50S＝(P30+P50+P80)／3
c：測定ﾁｬﾝﾈﾙ
10 分間※の測定から非線形分類で 1024 ｸﾗｽ(0～6400P.U.)に分けた累積確率関数(CPF)を求める
これを非線形補間法にて補正した後、平滑化した値を用いて演算する ※Pst(1min)は 1 分間
N

Plt C = 3 ×

∑ Pst
i =1

N

3
i

c：測定ﾁｬﾝﾈﾙ、N：12 回(2 時間の測定)

1P2W-1～4

Pst(1min)1 、Pst1 、Plt1

1P3W-1～2

Pst(1min)1、Pst1 、Plt1 、Pst(1min)2、Pst2 、Plt2

3P3W-1～2

線間電圧 Pst(1min)12、Pst12 、Plt12 、Pst(1min)32、Pst32 、Plt32

3P3W3A

線間電圧 Pst(1min)12、Pst12 、Plt12 、Pst(1min)23、Pst23 、Plt23 、Pst(1min)31、Pst31 、Plt31

3P4W

Pst(1min)1、Pst1 、Plt1 、Pst(1min)2、Pst2 、Plt2 、Pst(1min)3、Pst3 、Plt3- 151 -

KEW6315

KEW631510.4 クランプセンサの仕様

10.4 クランプセンサの仕様＜MODEL8128 ＞＜MODEL8127 ＞＜MODEL8126 ＞

定格電流

AC 5Arms[最大定格 AC50Arms(70.7Apeak)]AC 100Arms(141Apeak)AC 200Arms(283Apeak)

出力電圧0～50mV (AC 50mV/AC 5A)[Max.AC 500mV/AC50A]：10mV/A

AC0～500mV(AC500mV/AC100A)：5mV/A

AC0～500mV(AC 500mV/AC200A)：2.5mV/A

測定範囲

AC0～50Arms

AC0～100Arms

AC0～200Arms

確度(正弦波入力)
位相特性±0.5％rdg±0.1mV (50/60Hz)±1.0％rdg±0.2mV (40Hz～1kHz)±2.0°以内(0.5～50A／45～65Hz)

確度保証温湿度範囲
使用温湿度範囲
保存温湿度範囲
AC50Arms(50／60Hz)

出力インピーダンス
使用環境

約 20Ω

耐電圧

絶縁抵抗
被測定導体径
外形寸法

AC100Arms(50／60Hz)

AC200Arms(50／60Hz)

約 10Ω
約 5Ω
屋内仕様、高度 2000m 以下
IEC 61010-1，IEC 61010-2-032
IEC 61010-1，IEC 61010-2-032
測定 CAT.Ⅲ(300V) 汚染度 2
測定 CAT.Ⅲ(600V) 汚染度 2
IEC61326
IEC61326
AC3540V／5 秒間
AC5350V／5 秒間
コア嵌合部と外箱間
コア嵌合部と外箱間
外箱と出力端子間
外箱と出力端子間
コア嵌合部と出力端子間
コア嵌合部と出力端子間50MΩ以上／1000V
コア嵌合部と外箱間、外箱と出力端子間、コア嵌合部と出力端子間
最大約φ24mm
最大約φ40mm
100(L)×60(W)×26(D)mm

128(L)×81(W)×36(D)mm

ケーブル長

約 3m

出力端子

MINI DIN 6PIN

重量

±1.0°以内(2～200A／45～65Hz)

23±5℃、相対湿度 85％以下(結露しないこと)
0～50℃、相対湿度 85％以下(結露しないこと)-20～60℃、相対湿度 85％以下(結露しないこと)

最大許容入力適応規格±2.0°以内(1～100A／45～65Hz)

約 160g

約 260g付属品

取扱説明書
ケーブルマーカー

オプション

7146（バナナФ4 変換プラグ）・7185（延長コード）

KEW6315- 152 -

10.4 クランプセンサの仕様

KEW6315＜MODEL8125 ＞＜MODEL8124 ＞

定格電流

AC 500Arms (707Apeak)AC 1000Arms (1414Apeak)

出力電圧AC0～500mV(AC500mV/500A)：AC 1mV/A

AC0～500mV(AC500mV/1000A)：0.5mV/A

測定範囲

AC0～500Arms

AC0～1000Arms

確度(正弦波入力)

被測定導体径
外形寸法±0.5％rdg±0.1mV (50/60Hz)±0.5％rdg±0.2mV (50/60Hz)±1.0％rdg±0.2mV (40Hz～1kHz)±1.5％rdg±0.4mV (40Hz～1kHz)±1.0°以内±1.0°以内(5～500A／45～65Hz)(10～1000A／45～65Hz)
23±5℃、相対湿度 85％以下(結露しないこと)
0～50℃、相対湿度 85％以下(結露しないこと)-20～60℃、相対湿度 85％以下(結露しないこと)
AC500Arms(50／60Hz)
AC1000Arms(50／60Hz)
約 2Ω
約 1Ω
屋内仕様、高度 2000m 以下
IEC 61010-1，IEC 61010-2-032
測定 CAT.Ⅲ(600V) 汚染度 2
IEC61326
AC5350V／5 秒間
コア嵌合部と外箱間／外箱と出力端子間／コア嵌合部と出力端子間50MΩ以上／1000V
コア嵌合部と外箱間、外箱と出力端子間、コア嵌合部と出力端子間
最大約φ40mm
最大約φ68mm
128(L)×81(W)×36(D)mm
186(L)×129(W)×53(D)mm

ケーブル長

約 3m

位相特性
確度保証温湿度範囲
使用温湿度範囲
保存温湿度範囲
最大許容入力出力インピーダンス
使用環境
適応規格
耐電圧
絶縁抵抗

出力端子
重量
付属品
オプション

MINI DIN 6PIN
約 260g

約 510g
取扱説明書、ケーブルマーカー
7146（バナナФ4 変換プラグ）・7185（延長コード）- 153 -

KEW6315

KEW631510.4 クランプセンサの仕様＜KEW8129＞＜KEW8129＞

レンジ : AC 300 Arms（424Apeak）
300A レンジ：300A
AC30～300Arms(424Apeak)
定1000A格 電
流 1000A
レンジ : AC 1000 Arms（1414Apeak）
レンジ：
AC100～1000Arms(1414Apeak)
レンジ : AC 3000 Arms（4243Apeak）
3000A レンジ：3000A
AC300～3000Arms(4243Apeak)＜ KEW8130＞＜KEW8130＞＜ KEW8133 ＞＜ KEW8135 ＞＜KEW8133＞

AC 1000（1850Apeak）
AC 3000 Arms（5515A Peak）
AC 50 Arms（92A Peak）
AC Arms
1000
Arms(1850Apeak)AC 3000 Arms(5515Apeak)

300A レンジ：
300A レンジ :(AC500mV/AC300A)
AC0～500mVAC0300A）:1.67mV/A～ 500mV（AC500mV/AC 1.67mV/A
1000A レンジ：
AC0～500mV
1000A レンジ :AC0 - AC0～500mV
500mV
AC0 - 500mV
AC0 - 500mV
出 力
圧 (AC500mV/AC1000A) 0.5mV/A
AC0電～500mV(AC500mV/AC1000A)：0.5mV/A(AC500mV/AC3000A)：0.167mV/A
AC0 ～ 500mV（AC500mV/AC1000A）:0.5mV/A （AC500mV/AC1000A）:0.5mV/A（AC500mV/AC3000A）:0.167mV/A（AC500mV/AC50A）:10mV/A
3000A レンジ：
3000A レンジ :(AC500mV/AC3000A)
0.167mV/A
AC0～500mVAC0500mV（AC500mV/AC3000A）:0.167mV/A～

300A レンジ： AC30～300Arms
300A AC100～1000Arms
レンジ : 30 - 300Arms
1000A レンジ：
測 定 3000A
範 囲レンジ：
1000A AC300～3000Arms
レンジ : 100 - 1000Arms

3000A レンジ : 300 - 3000Arms±1.0％rdg (45～65Hz)(センサ中央で測定において)

AC0～1000Arms
AC0 - 1000Arms

AC0 - 3000Arms±0.8％rdg±0.2mV (45～65Hz)±1.5％rdg±0.4mV (40Hz～1kHz)

AC0～3000Arms
AC0 - 50Arms±1.0％rdg±0.5mV (45～65Hz)±1.5％rdg±0.5mV(40Hz～1kHz)± 1.0%rdg± 0.5mV（45Hz - 65Hz）（0-50A）

確度± 1.0%rdg（45 - 65Hz）± 0.8%rdg ± 0.2mV（45 - 65Hz） ± 1.0%rdg ± 0.5mV（45 - 65Hz）± 1.5%rdg ±0.5mV（40Hz - 300Hz）（0-20A）±1.0°以内（正弦波入力） （センサ中央で測定において） ±1.5%rdg±0.4mV（40Hz -±2.0°以内(45～65Hz)
1kHz）±1.5%rdg±0.5mV（40Hz - 1kHz）± 1.5%rdg ± 0.5mV（300Hz - 1kHz）（0-5A）(各レンジの測定範囲±3.0°以内(40Hz～1kHz)
45～65Hzにおいて)± 1.0°
以内± 2.0°以内（45 - 65Hz）± 3.0°以内（45 - 65Hz）
位 相 特 性85％以下
（各レンジの測定範囲 45 - 65Hz において） 23±5℃、相対湿度± 3.0°以内（40 - 1kHz）± 4.0°以内（40 - 1kHz）(結露しないこと)
確度保証温湿度範囲
23 ± 5℃、相対湿度 85％以下（結露しないこと）-10～50℃、相対湿度 85％以下
0～50℃、相対湿度 85％以下
使用温湿度範囲− 10 ～ 50℃、相対湿度 85％以下（結露しないこと）(結露しないこと)(結露しないこと)-20～60℃、相対湿度
85％以下
保存温湿度範囲− 20 ～
60℃、相対湿度 85％以下（結露しないこと）(結露しないこと)最 大許容入力
AC3600Arms（50/60Hz）
AC1300Arms（50/60Hz）AC3900Arms（50/60Hz） AC65Arms（50/60Hz）
AC3600Arms 連続(50／60Hz)
AC1300Arms(50／60Hz)
AC3900Arms(50／60Hz)
出力インピーダンス
100 Ω以下
約 100Ω約以下
屋内仕様、高度2000m
以下以下
使 用 環 境屋内仕様、高度2000m
IEC 61010-1，IEC 61010-2-032
IEC 61010-1，IEC 61010-2-032
IEC 61010-1,IEC 61010-2-032
IEC 61010-1,IEC 61010-2-032
測定 CATⅢ600V 汚染度 2
測定 CATⅢ600V／CATⅣ300V 汚染度 2
適 応 規 格CAT. III（600V）汚染度 2
CAT. III（600V）/CAT.IV（300V）汚染度 2
IEC61326
IEC61326
IEC61326
IEC61326
耐電圧

AC5350V／5
秒間 秒間
AC5350V/5
回路―センサ間回路―センサ間

絶 縁 抵 抗被測定導体径 最大約φ150mm
最大約 ø150mm
111(L)×61(W)×43(D)mm
111（L）× 61（ W）× 43（D）mm （突外 形 寸 法 (突起物を含まない)
起物を含まない）
センサ部：約2m
センサ部：約2m
ケーブル 長出力ケーブル：約 1m
8129-1:約 410g
8129-2:約 680g
8129-1：約 410g ／ 8129-2：約 680g ／重量 8129-3:約 950g
8129-3：約 950g
取扱説明書
取扱説明書、出力ケーブル
出力ケーブル (M-7199) （M-7199）、
付
属
品
携帯ケース
携帯ケース
オプション

出 力 端 子KEW6315AC5160V／5
AC5160V ／ 5 秒間
回路―センサ間回路―センサ間

50M Ω 以上 / 1000V
50MΩ以上／1000V
回路―センサ間回路―センサ間
最大約φ110mm
最大約ø110mm
最大約 ø170mm

最大約φ170mm
最大約 ø75mm

65(L)×24(W)×22(D)mm
65（L）× 25（W）× 22（D）mm
センサ部：約 2.7m
センサ部：約 2.7m
出力ケーブル：約
0.2m
出力ケーブル：約 0.2m
MINI DIN 6PINMINI DIN 6PIN
約 180g
約 180g－

約 200g
約 200g
約 170g
取扱説明書
ケーブルマーカー
取扱説明書、ケーブルマーカー、携帯ケース
携帯ケース―- 154 -

10.4 クランプセンサの仕様＜MODEL8141 ＞

KEW6315＜MODEL8142 ＞

定格電流AC1000mArms

出力電圧AC0～100mV(AC100mV/AC1000mA)

測定範囲

AC0～1000mArms

確度(正弦波入力)＜MODEL8143 ＞±1.0％rdg±0.1mV (50/60Hz)±2.0％rdg±0.1mV (40Hz～1kHz)

位相特性-------------

確度保証温湿度範囲
使用温湿度範囲
保存温湿度範囲
最大許容入力出力インピーダンス
使用環境

23±5℃、相対湿度 85％以下(結露しないこと)
0～50℃、相対湿度 85％以下(結露しないこと)-20～60℃、相対湿度 85％以下(結露しないこと)
AC100Arms(50／60Hz)
AC200Arms(50／60Hz)
AC500Arms(50／60Hz)
約 180Ω
約 200Ω
約 120Ω
屋内仕様、高度 2000m 以下
IEC 61010-1，IEC 61010-2-032
測定 CAT.Ⅲ(300V) 汚染度 2
IEC61326 (EMC 規格)
AC3540V／5 秒間
コア嵌合部と本体外装の間
コア嵌合部と出力端子の間
本体外装と出力端子の間
50MΩ以上／1000V
コア嵌合部と本体外装の間
コア嵌合部と出力端子の間
本体外装と出力端子の間
最大約φ24mm
最大約φ40mm
最大約φ68mm
100(L)×60(W)×26(D)mm
128(L)×81(W)×36(D)mm
186(L)×129(W)×53(D)mm(突起部除く)(突起部除く)(突起部除く)
約 2m
MINI DIN 6PIN
約 150g
約 240g
約 490g
取扱説明書
携帯ケース7146（バナナФ4 変換プラグ）
7185（延長コード）

適応規格

耐電圧

絶縁抵抗
被測定導体径
外形寸法
ケーブル長
出力端子
重量
付属品
オプション

- 155 -

KEW6315

KEW631510.4 クランプセンサの仕様＜KEW8146 ＞＜KEW8147 ＞＜KEW8148 ＞

AC 30Arms (42.4Apeak)AC 70Arms (99.0Apeak)AC 100Arms (141.4Apeak)AC0～1500mV(AC50mV/A)

AC0～3500mV(AC50mV/A)

AC0～5000mV(AC50mV/A)

AC0～30Arms
0～15A±1.0％rdg±0.1mV (50/60Hz)±2.0％rdg±0.2mV (40Hz～1kHz)
15～30A
±5.0％rdg (50/60Hz)±10.0％rdg (45～1kHz)

AC0～70Arms
0～40A±1.0％rdg±0.1mV (50/60Hz)±2.0％rdg±0.2mV (40Hz～1kHz)
40～70A
±5.0％rdg (50/60Hz)±10.0％rdg (45～1kHz)

AC0～100Arms
0～80A±1.0％rdg±0.1mV (50/60Hz)±2.0％rdg±0.2mV (40Hz～1kHz)
80～100A±5.0％rdg (50/60Hz)±10.0％rdg (45～1kHz)------------23±5℃、相対湿度 85％以下(結露しないこと)
0～50℃、相対湿度 85％以下(結露しないこと)-20～60℃、相対湿度 85％以下(結露しないこと)
AC70Arms(50／60Hz)
約 100Ω
屋内仕様、高度 2000m 以下
IEC 61010-1，IEC 61010-2-032
測定 CAT.Ⅲ(300V) 汚染度 2
IEC61326
AC3540V／5 秒間
コア嵌合部と外箱間
外箱と出力端子間
コア嵌合部と出力端子間

AC30Arms(50／60Hz)
約 90Ω

AC100Arms(50／60Hz)
約 60Ω

50MΩ以上／1000V
コア嵌合部と外箱間、外箱と出力端子間、コア嵌合部と出力端子間
最大約φ24mm

最大約φ40mm

最大約φ68mm

100(L)×60(W)×26(D)mm

128(L)×81(W)×36(D)mm

186(L)×129(W)×53(D)mm

約 150g

KEW6315

約 2m
MINI DIN 6PIN
約 240g
取扱説明書
ケーブルマーカー
7146（バナナФ4 変換プラグ）
7185（延長コード）- 156 -

約 510g11.1 トラブルシューティング

KEW6315

11章 故障かなと思ったら
11.1 トラブルシューティング
本製品を使用しているときに故障かなと思われる内容が発生した場合、下記の事項を確認してください。下記以外の不具合が認められる場合は、弊社または販売店までご連絡をください。
症状
電源キーを操作しても電源が入らない。(何も表示しない。)

確認事項
AC 電源の場合
・電源コードがコンセントに正しく接続されているか確認してくだ
さい。
・電源コードが断線していないか確認してください。
・電源電圧が許容範囲内か確認してください。
電池駆動の場合
・電池の極性が正しくセットされているか確認してください。
・単３形ニッケル水素電池（Ni－MH）の場合、充分充電されているか確認してください。
・単３形アルカリ乾電池の場合、電池が消耗していないか確認してください。
上記が問題ない場合
・ＡＣの電源コードを抜いた状態で、一旦全ての電池を取り外し、再度全ての電池をセットし直してＡＣ電源を接続し電源を入れてください。この状態でも電源が入らない場合には本体が故障している可能性があります。

キー操作ができない。

・キーロック機能が動作していないか確認してください。
・本書にて各測定レンジの有効キーを確認してください。

測定値を表示しない。

・電圧の 1ｃｈに入力している周波数が確度保証の範囲内か確認し測定表示値が不安定、またはおかしてください。４０～７０Ｈｚが測定できる範囲です。

い。

・電圧用測定コード、クランプセンサが正しく接続されているか確認
してください。
・測定ラインに対して本製品の設定および結線が正しいか確認してください。
・使用しているクランプセンサとクランプの設定が正しいか確認してください。
・電圧用測定コードが断線していないか確認してください。
・入力信号にノイズがのっている可能性がないか確認してください。
・近くに強い電磁波がないか確認してください。
・使用環境が本製品の仕様内かどうか確認してください。
・結線方式で測定するｃｈに異なる種類の電流センサを接続していないか確認してください。

内部メモリに保存ができない。

・保存しているファイル数を確認してください。
・ＳＤカードを挿入していないか確認してください。ＳＤカードを挿入している場合には内部メモリへ保存できません。
- 157 -

KEW6315

KEW631511.2 エラーメッセージの内容とその対処方法
症状

ＳＤカードに保存ができない

確認事項
・ＳＤカードが正しく挿入されているか確認してください。
・ＳＤカードがフォーマットされているか確認してください。
・ＳＤカードの容量がオーバーしていないか確認してください。
・使用するＳＤカードの保存ファイル数又は容量を確認してください。
・使用するＳＤカードが本製品の動作確認済みのカードであるか確認してください。
･既知のハードウエアで正常に動作することを確認してください。

USB 通信でダウンロードおよび設定が・本体をパソコンに USB コードで正しく接続されているか確認してく

できない

ださい。
・通信アプリケーションソフト(KEW Ｗｉｎｄｏｗｓ for KEW6315)において接続デバイスが表示されていることを確認ください。表示されていない場合、USB ドライバが正常にインストールできていない可能性があります。別ＰＤＦファイルの「ＫＥＷ Ｗｉｎｄｏｗｓ ｆｏｒ ＫＥＷ６
３１５」用のインストールマニュアルを参考にしてＵＳＢドライバをＰ
Ｃへ再インストールしてください。

自己診断で頻繁に NG を表示する

・「ＳＤ Ｃａｒｄ」の場合は別項目「ＳＤカードに保存ができない」を参照してください。その他の診断結果でＮＧを表示する場合には、Ａ
Ｃの電源コードを抜いた状態で、一旦全ての電池を取り外し、再度全ての電池をセットし直してＡＣ電源を接続し、自己診断をやり直してください。この状態でもＮＧを表示する場合には本体が故障している可能性があります。

11.2 エラーメッセージの内容とその対処方法
本製品使用中に、画面にメッセージが表示されることがあります。
ここでは、その内容と対処方法を説明します。
メッセージ

内容／対処方法

SD カードが挿入されていません
転送を行うためには使用可能な SD カードを挿入してください
設定を読み込むには使用可能な SD カードを挿入してください

・ＳＤカードが正しく挿入されているか確認してください。詳しくは「4.3 Ｓ
Ｄカードの挿入／取り出し方法」３３頁を参照してください。

設定を保存するには使用可能な SD カードを挿入してください
空き容量の十分な SD カードが挿入されているか確認してください
SD カードに空き領域がありません
フォーマットまたはデータ削除を行ってください

KEW6315・ＳＤカードの空き容量を確認してください。容量が少ない場合には保存しているファイルを削除、またはフォーマットするか、本体でフォーマッ
トした別のＳＤカードを使用してください。詳しくは「記録データに関する操作」８２頁を参照してください。

- 158 -

11.2 エラーメッセージの内容とその対処方法

KEW6315

メッセージ

内容／対処方法

正しく識別できませんでした。

・電流センサが本体に確実に接続されているか確認してください。

接続されているセンサをもう一度確認

・故障が疑われる場合には下記の手順で確認してください。

してください。

NG と識別された電流センサの接続ｃｈを正しく識別されているｃｈと変更
して再テストを行ってみてください。この時、前回と同じｃｈが NG と識別された場合には、本体が故障している可能性があります。前回 NG と識別された電流センサを接続しているｃｈが、NG と識別された場合には、
電流センサが故障している可能性があります。故障していた場合には直ちに使用を中止してください。

電池残量が少ないため電源を OFF します。

・ＡＣ電源を使用するか、新しい単３形アルカリ乾電池（LR6）または充電済みの単３形ニッケル水素電池(Ni-MH)、６本と入れ替えてください。詳しくは「乾電池のセット方法」３１頁を参照してください。
・内部メモリの空き容量と保存ファイル数を確認してください。内部メモリ

内部メモリに空き領域がありませんに保存可能なファイル数は測定データは３ファイル、その他は８ファイ

フォーマットまたはデータ削除を行ってルまでです。容量が少ない場合には保存しているファイルを削除、またください

はフォーマットしてください。詳しくは「記録データに関する操作」８２頁を参照してください。
・再度設定ファイルの読み込みを行ってください。それでも読み込めない設定ﾌｧｲﾙを読み込むことができせん
ﾌｧｲﾙが壊れている可能性があります

場合、ＳＤカードに記録した設定ファイルであればＳＤカードまたは本体、内部メモリにある設定ファイルであれば本体が故障している可能性があります。本体が故障していた場合には直ちに使用を中止してください。

十分な空き領域がありません

・ＳＤカードの空き容量と、内部メモリの空き容量と保存ファイル数を確認

SD カードおよび内部メモリの空き容量

してください。内部メモリに保存可能なファイル数は測定データは３ファを確認してくださいイル、その他は８ファイルまでです。容量が少ない場合には保存しているファイルを削除、またはフォーマットするか、ＳＤカードの場合には本記録可能な領域がありません

体でフォーマットした別のＳＤカードを使用してください。詳しくは「記録データに関する操作」８２頁を参照してください。
・記録「開始方法」に連続記録／時間帯指定を選択し、記録終了の日時

過去の日時が指定されていますに過去の日時を設定していた場合に表示します。それぞれの記録方法

記録開始方法を確認してくださいにて設定した日付を確認してください。詳しくは「⑧／⑨ 開始方法ごとの設定」４５頁を参照してください。
・SET UP の「記録設定」に矛盾がないか確認してください。詳しくは「5.4記録設定」７１頁を参照してください。

記録を開始できませんでした

・再度記録を開始してください。それでも開始できない場合、記録対象がＳＤカードであればＳＤカードまたは本体、内部メモリであれば本体が故障している可能性があります。本体が故障していた場合には直ちに使用を中止してください。

- 159 -

KEW6315

KEW631511.2 エラーメッセージの内容とその対処方法
メッセージ

記録中および記録待機中は本体設定を変更できません

前回と異なるｾﾝｻを識別しました。
測定前に SET UP の基本設定を再確認してください。

内容／対処方法
記録中には設定の確認しかできません。設定を変更する場合には、必ず記録をストップし「記録を停止しました」のメッセージが消えた後に変更してください。
・前回の測定時と異なるクランプセンサを接続してる場合に表示します。
SET UP の「基本設定」にて現在接続しているクランプセンサを設定するか「センサ識別」で接続しているクランプセンサを自動認識してください。

センサの接続が正しくありません。

・結線方式で測定するｃｈに異なる種類の電流センサを接続していないか

接続されているセンサをもう一度確認

確認してください。測定に使用する電流センサは同じ種類しか使用でき

してください。

ません。
・必ず記録をストップし「記録を停止しました」のメッセージが消えた後に

SD カードの空き領域が足りません。
記録を中止しました。

保存しているファイルをＰＣ等にバックアップして、ファイルを削除、またはフォーマットするか、本体でフォーマットした別のＳＤカードを使用して記録を再開してください。詳しくは「記録データに関する操作」８２頁を参照してください。
・必ず記録をストップし「記録を停止しました」のメッセージが消えた後に内部メモリの空き領域が足りません。

保存しているファイルをＰＣ、ＳＤカード等にバックアップして、ファイルを記録を中止しました。

削除、またはフォーマットして記録を再開してください。詳しくは「記録データに関する操作」８２頁を参照してください。

KEW6315- 160 -

この取扱説明書に記載されている事項を断りなく変更することがありますのでご了承ください。

■ホームページのご案内
www.kew-ltd.co.jp●新製品情報 ●取扱説明書／ソフトウェア／単品カタログのダウンロード●販売終了製品情報

03-18
92-2161B
8-2192-1261E