AVAILABLE
with Signature
Calibration

SEE PG. IN-5General Purpose
Gage/Absolute
Pressure Transducers
Model ZAMPLIFIED OUTPUT AVAILABLE

1-888-282-9891

STAINLESS STEEL
0.5 to 60,000 psi, 35 mbar to 4000 bar

www.honeywell.com/sensing

Sensotec Sensors

The Model Z is designed as a general industrial pressure transducer with a wide variety ofavailable options to meet specific application requirements. Pressure ranges span from 0.5 to30,000 psi. All models are constructed of stainless steel and utilize complete four arm 350
ohm strain gage bridges. Model Z transducers use a standard gage design. The absolutemodels have an internal sealed 0 psia reference.

Dimensions
Model Z Gage

Model Z Absolute(Order Code AP131)
Available Ranges*
0.5; 1; 2; 5 psig
10; 15 psig
25; 50; 75; 100; 150; 200;
300; 500 psig
750; 1000; 1500; psig
2000; 3000; 5000; 7500;
10,000 psig
15,000; 20,000; 30,000;
50,000; 60,000 psig

(Order Code AP132)
D"
2.25 (5.72 cm)
1.50 (3.81 cm)

L"
2.42 (6.15 cm)
2.35 (5.97 cm)

1.5 (3.81 cm)
1.5 (3.81 cm)

2.35 (5.97 cm)
2.35 (5.97 cm)

1.5 (3.81 cm)

2.39 (6.07 cm)

1.5 (3.81 cm)

2.21 (5.61 cm)

Available Ranges*
0.5; 5 psia
10; 15; 25; 50; 75; 100; 150;
200; 300; 500; 750;
1000; 1500 psia
2000: 3000; 5000; 7500;
10,000 psia
15,000; 20,000; 30,000;
50,000; 60,000 psia

D"
2.25 (5.72 cm)

L"
2.09 (5.31 cm)

1.5 (3.81 cm)

2.35 (5.97 cm)

1.5 (3.81 cm)

2.39 (6.07 cm)

1.5 (3.81 cm)

2.21 (5.61 cm)*Stocked ranges are in bold face print.
**Consult Sensotec

2000 to 60,000 psi
140 bar to 4000 bar2,000 to 10,000 PSI
PRESSURE PORT
1/4–18 FEMALE NPT
15,000 TO 60,000 PSI
PRESSURE PORTAUTOCLAVE F-250C

0.5 to 1500 psi
35 mbar to 100 bar

PRESSURE PORT
1/4–18 NPT MALE

Options (See Appendix)
Temperature compensated 1b, 1c, 1d, 1e, 1f; Internal amps 2a; 2c; 2k; 2j; 2t; 2n or 2N Intrinsically Safe
Amp - see page AP-6.
Amp enhancements 3d; Pressure ports 5a, 5b, 5c, 5d (</=15,000 psi); Electrical terminations 6e, 6f, 6g,6h, 6i; Int. shunt cal 8a; Signature calibration 53e
Premium Options: 1g (>/=15 psi); 1i (>/=15 psi only); 2q; 3a, 3c; 6b, 6c, 6j; Special calibration 9a (>/=5psi gage only); 9b (Gage only, >/=5 psi); 10a; 12a, 12b
Accessories: Mating connectors and connector/cable assemblies; Pressure port adapters.

PR-8

1-888-282-9891
Model Z
Gage:
Absolute:

PERFORMANCE

Pressure Ranges..........................
Accuracy (min) .............................
Non-Linearity ................................
Hysteresis.....................................
Non-Repeatability .........................
Output (standard) .........................
Resolution ....................................

ENVIRONMENTAL

Temperature, Operating
1 to 1,000 psi ...........................
1,500 to 60000 psi ...................
Temperature, Compensated ........
Temperature Effect– Zero (max).............................
– Span (max .............................

0.5 to 60,000 psi* (35 mbar to 4000 bar)+ /-0.25% F.S.
+ /-0.15% F.S.
+ /-0.10% F.S.
+ /-0.05% F.S.
3mV/V* *Infinite-100° to 325° F (-73° to 162° C)-100° to 250° F (-73° to 95°C)
60° F to 160° F (15° C to 71° C)
0.005% F.S./ ° F (0.009% FS./ ° C)
0.005% Rdg./ ° F (0.009% Rdg./ ° C)
Bonded foil
10VDC
Up to 12VDC or AC
5000 megohm @ 50VDC
350 ohms
Included#2 (See Pg. AP-8)
PTIH-10-6P or equiv.
(Hermetic stainless)
PT06A-10-6S or equiv.

Mating Connector (not incl.) .........

MECHANICALINTERNALLY AMPLIFIEDUNITS (Optional)

Media............................................
Overload-Safe
Overload-Burst
0.5 to 5000 psi .........................
7500 to 10,000 psi ...................
15,000 to 60,000 psi ................
Pressure Port
0.5 to 1500 psi .........................
2000 to 10,000 psi ...................
15,000 to 60,000 psi ................
Dead Volume................................
0.5 to 5 psi ...............................
10 to 15 psi ..............................
25 to 1500 psi ..........................
2000 to 10,000 psi ...................
Wetted Parts Material...................
Type .............................................
Weight ..........................................
Case Material ...............................

Gas Liquid
50% over capacity
300% over capacity
200% over capacity
70% over capacity
1/4-18 NPT male
1/4-18 NPT female
Autoclave AE F250-C
0.32 cu. in.
0.17 cu. in.
0.17 cu. in.
0.12 cu. in.
17-4 PH Stainless / 15-5 Stainless
Absolute or Gage * * *10 oz. (50 psi)
Stainless Steel

Outputs Available .........................

0-5VDC, 4-20mA

Additional Length..........................

1.12" (2.84 cm)

NOTES

GENERAL PURPOSE

Strain Gage Type .........................
Excitation (calibration) ..................
Excitation (acceptable) .................
Insulation Resistance ...................
Bridge Resistance ........................
Shunt Calibration Data .................
Wiring Code (std) .........................
Electrical Termination (std) ..........

PRESSURE

ELECTRICALOrder Code AP131
Order Code AP132

4-20mA Intrinsically safe
See page AP-6
2.0" (5.08 cm)* 0.5 psi (35 mbar) is not available in absolute pressure** Output for 1 and 2 psi units is 1mV/V/psi minimum.
*** Gage pressure units greater than 500 psi (35 bar) are sealed at atmospheric pressure.

General Information
How to order (See Pg. AP-19)
Gage/Absolute pressure selection flow chart (See Pg. PR-1)●お問い合わせは下記へ ̶PR-9