SDR-Kits, 14 Ragleth Grove, Trowbridge Wilts, BA14-7LE, United Kingdom sdrkits@gmail.com

DG8SAQ VNWA 2.x Self Installer Manual

1.

Introduction:

This document describes the DG8SAQ Installation process using the VNWA Self Installer. The VNWA
Self installer is an executable file which contains the VNWA Driver, the VNWA Application and the
VNWA helpfile and assists VNWA 2.x users with easy installation option.
The Self Installer installs VNWA Application 35.x upwards.

2.

Important Information

Copyright notice
The design of the is copyright by Thomas Baier DG8SAQ & SDR-Kits - all right reserved 2.1

Caution - Safety Information:
1. VNWA 2.x is powered and controlled through the USB-cable supplied. The USB-cableshould be connected to the USB port of a Personal Computer capable of supplying +5V DC
at 500mA maximum.
2. The maximum safe input voltage into the TX-out and Rx-in ports is an RF Voltage of 0dBm
(225mV RMS with a frequency between 1 kHz up to 1300 MHz). Exceeding this value may
cause damage to the VNWA and may invalidate product warranty.
3. Always check the SDR-Kits Website for current Product information including Safety
Information and latest Product Updates.
http://www.sdr-kits.net/VNWA/VNWA_Documentation.html

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 1 of 32

2.2
Product Documentation:
The documentation of this product is supplied as a VNWA Application Helpfile, through the VNWA
Installer or may be downloaded from the Internet fromhttp://sdr-kits.net/DG8SAQ/VNWA/VNWA_HELP.pdf
2.3

Requirements:

The Self Installer is available from VNWA Application 35.0 upwards. After VNWA Application 35.x is
installed it is important that VNWA 2.x is firmware is manually upgraded to the latest firmware. The
latest firmware can be found in the VNWA Application folder used by the Self Installer.
Note: Firmware version 4.14 comes after version 4.9 and 4.10!!
2.4

VNWA License code:

Make sure you have the VNWA License code which was supplied with the VNWA available prior to
the installation. It will be required for the upgrade.

3.

VNWA Self Installer: Driver - Software and Helpfile Installation

Caution: Although the installation package is provided for easy installation, it is recommended that theappropriate installation procedure for your Operating System is reviewed before starting the installer.
The procedure below assumes that VNWA package has NOT been installed previously on theComputer in question.
3.1
Windows 7 & Vista 64 bit and 32 bit Installation
This section shows installation procedure for Windows 7 (64 or 32 bits)

Note: For Installation on Windows XP refer to section 3.4
IMPORTANT: Information for Windows 7 and Vista users:
If you are installing the VNWA application for the first time, the VNWA Setup program will guide you
to install an application Winhlp32.exe from the Microsoft website. The Winhlp32.exe program isrequired to display the VNWA help file. However Microsoft license conditions does not allow
theWinhelp32.exe to be distributed as part of the VNWA setup program.
VNWA installation is easier and friendlier if Winhlp32.exe application is already installed beforehand.
It is therefore recommend to install Winhlp32.exe from the link
http://support.microsoft.com/kb/917607 prior to starting the VNWA installation program in step
3.1.1. You can always check if Winhlp32.exe is already installed by inspecting the file size of this filein C:\windows. If file size of Winhlp32.exe is 290 KBytes then the application is already installed. If
this file is present but it is only 9Kbytes in size then Winhlp32.exe should be installed first:
3.1.1 Download and save the Installation Package from the following location to your desktop
http://www.sdr-kits.net/DG8SAQ/VNWA-installer.exe

3.1.2 Make sure the VNWA is NOT connected to the PC
Double Click on the VNWA-installer Icon to start the VNWA installation process.
SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 2 of 32

Confirm the VNWA-installer can make changes to your Computer, then screen below is shown.
Press "Next"

3.1.3 If the file Winhlp32.exe is NOT installed on the Computer than the screen below is shown.
Winhlp32.exe file should be installed otherwise VNWA helpfile will not be displayed. If you
click the box the Web browser will be launched. Follow the instructions on the Microsoft
Website to download winhlp32.exe. Note: For 64 bit Operating Systems the filename ending in-x64.msu should be downloaded and -x86.msu for 32 bit Operating Systems.
Note: The link to the microsoft winhlp32.exe support is:

SDR-Kits

http://support.microsoft.com/kb/917607– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 3 of 32

3.1.4 Press "Next" – The default location where VNWA will be installed will be shown:
Note: Select a different Destination Location (ie C:\VNWA_2) if you s to install a second VNWA
application on the same PC to prevent overwriting the setup files of your first VNWA application.

3.1.5 Press "Next" – Default where program shortcuts will be installed

3.1.6 Press "Next" – Screen to enter VNWA license key is now displayed as shown below:
There are two options to enter the VNWA license key for your VNWA:

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 4 of 32

Directly enter the VNWA license code exactly as shown in the Manual supplied with the VNWA
2.x hardware. The VNWA license code is also specified in the VNWA shipping advice email
Note: the last colon is the last character and must also be entered.

Note: A Screen (not shown in this manual) is now displayed with option to create a Desktop Icon for:A) All users or B) For Current User only. Make your selection and press "Next"
3.1.7 Press "Next" – Information on Audio Configuration is displayed

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 5 of 32

3.1.8 Press "Next" and Ready to Install screen is displayed

3.1.9 Press "Install" – VNWA Application and helpfile are now installed and Driver Installation
window is displayed. Note: Do NOT plug-in VNWA at this time

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 6 of 32

3.1.10 Press "Next" – Following screen may be displayed – Select “Install this driver software
anyway”

3.1.11 Press "Next" – Driver installation will take from 10-30 seconds

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 7 of 32

3.1.12 Press "Finish" 3.1.13 If you have selected to view the “Show Vista/Windows 7 Audio setup help” in point 3.1.5 readthe information and close the window prior to next step
3.1.14 Press “Finish” to complete installation.

END OF VNWA Installation

3.2

Windows 7 & Vista 64 bit and 32 bit Application Configuration:

3.2.1 Plug in the VNWA USB cable into USB Port of your Computer. Note: It is recommended toconnect the VNWA same Computer USB Port next time you use the VNWA.
When connecting the VNWA for the first time after Driver installation, the VNWA will be recognizedand device driver installation should complete automatically.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 8 of 32

3.2.2 Click “Start” “All Programs” “VNWA” and VNWA Icon to start the application or use the
Shortcut created on the Desktop.

3.2.3 Overclocking Warning may be displayed when you run VNWA Application software forthe first time. Select "ok" to allow overclocking of the DDS chips beyond 400 MHz. Note:
VNWA performance over 500 MHz will be worse affected if overclocking of the DDS is not
permitted.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 9 of 32

3.2.4 VNWA Application starts: Click “OK” to enter VNWA setup information

3.2.5 The tab "USB Setting" should now displayed. If not click on the USB-Setting Tab
Check whether VNWA license code is shown, In event this information is missing then enter
the VNWA license code again. Now press “Rescan USB Bus” and “Test USB Interface”
Check bottom line for message “Test passed without errors”

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 10 of 32

3.2.6 Select Tab “Audio Settings” and set “Audio Capture Device” to “USB Audio Codec”
Note: Variants of the USB Audio Codec” may be shown instead
For Factory VNWA2.x check Reference = Right Channel has been selected. (Note: some
VNWA 2.x built from a Kit may use "Left Channel" depending on wiring of the 3.5mm audio
connectors) Now press “Test Audio”

3.2.7

Click on Tab “Audio Level” and check whether two sine waves are present as show
below in “Reflect” mode. (100% Reflection with TX Port unterminated)
If sine waves are distorted then USB Audio Codec needs to be configured as shown insection 3.3.
When the Audio level adjust is performed according to point 3.4 click on Tab "Calibrate Sample
Rate"
Note: when the THRU Mode is active the Audio levels min and max are higher than when in
REFL Mode

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 11 of 32

3.2.7 Connect the TX port and RX port with SMA cable and check that the Reflect (Left Channel)
sine wave has now disappeared. (TX port now sees 50 Ohms and no reflection)
Next select “Thru” and check that the Left Channel "sine wave is displayed.

3.2.8 Click “Stop“ cease the Audio test. Next click on Instrument Settings” and check whether clock
is set to “auto" as shown below. Note: for VNWA 2, the clock is generally a little below 36
MHz, somewhere between 35.855 MHz and 36.000 MHz depending on Crystal fitted.
When performing Audio test press ++ / + or -- / - buttons until the sinewaves are almost
stationary on the screen. If you have a frequency counter perform Calibrate DDS Clockcalibration as described in the VNWA helpfile.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 12 of 32

3.2.9 Press the top right button to close the VNWA Setup screen. Next we test whether the VNWA
helpfile is available. Press “Help” and select “Help” again. The screen below may be shown.
Press “Yes” and “Browse” to select the location where the Helpfile is located. If standard
VNWA installation defaults were selected then the default path will be:
My Computer C:\VNWA\VNWA.hlp

3.2.10 IMPORTANT: To save the VNWA configuration (after any changes are made) exit the Setup
VNWA application. This saves the VNWA configuration files for the next time the VNWA applicationis started.
End of VNWA Configuration

3.3

Windows 7 - 64 bit and 32 bit Audio Codec Configuration

3.3.1 Windows 7 configures the USB Audio Codec as a mono application with levels set too high asshown below. This prevents VNWA application from functioning.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 13 of 32

3.3.2 Start the VNWA Application and select Audio Test as shown in 3.2.7
Now to configure USB Codec properties select “Start”, “Control Panel”

3.3.3 Select “Hardware and Sound” and click on "Manage Audio Devices"

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 14 of 32

3.3.4 and click on tab “Recording”

3.3.5 Click on “Microphone USB Audio Codec” and select “Advanced”
Ensure that Default Format is set to “2 channel, 16 bit, 48000 Hz”

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 15 of 32

3.3.6 Next select “Levels” and adjust Microphone level to value of between 3 and 5, (typically 4)
and press OK
Note: If 4 is not selectable e.g. the displayed number jumps from 3 to 5 then use the left and rightarrow keys and count you way. You will find whilst only 5 is shown both the setting for 4 and 5
is available.

3.3.7 Check on VNWA Setup Screen that sinewave covers approx 50% to 60% as shown in step 3.3.6
and 3.3.7. If necessary adjust Microphone level until this is achieved.

End of Audio Codec Configuration

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 16 of 32

3.4

Windows XP Installation

Note: For Installation on Windows 7 and Vista refer to section 3.1
This section shows installation procedure for Windows XP Operating System.
3.4.1 Download and save the Installation Package from the following location to your desktop
http://www.sdr-kits.net/DG8SAQ/VNWA-installer.exe
3.4.2 Make sure the VNWA is NOT connected to the PC
Double Click on the VNWA-installer Icon to start the VNWA installation process.
Confirm that the VNWA-installer can make changes to your Computer

3.4.3 Press Next – The default location where VNWA will be installed will be shown:
Note: Select a different Destination Location (ie C:\VNWA_2) if you want to install a second VNWA
application on the same PC. This prevents overwriting the setup files of your first VNWA application

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 17 of 32

3.4.4 Press Next – Default Location where VNWA program shortcuts will be installed

3.4.5

Press Next – Screen to enter VNWA license key is now displayed as shown below:
There are two options to enter the VNWA license key for your VNWA:
Directly enter the VNWA license code exactly as shown in the Manual supplied with the VNWA
2.x hardware
The VNWA license code is also specified in the VNWA shipping advice email
Note: the last colon is the last character and must also be entered.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 18 of 32

Note: A Screen (not shown in this manual) is now displayed with option to create a Desktop Icon for:A) All users or B) For Current User only. Make your selection and press "Next"
3.4.6

Press "Next" - Read and execute the instructions displayed on the screen below:
Note: The VNWA hardware must be connected in this step otherwise driver installation may
fail.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 19 of 32

3.4.7 If the screen below is displayed after plugging in VNWA then press "Cancel button on thisscreen3.4.8 Press "Install" on Screen below. The Installation of VNWA Application and Driver should now
start and screen below is shown.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 20 of 32

3.4.9 The Installation of VNWA Application and Driver now start and screen below is displayed.
press "Next"

3.4.10 Driver installation now completes - Press "Finish"

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 21 of 32

3.4.11 Press "Finish" - to exit the VNWA installation routine.

END of Windows XP Installation

3.5 Windows XP VNWA Application Configuration
3.5.1 Ensure the VNWA USB cable into USB Port of your Computer.
3.5.2. Click “Start” “All Programs” “VNWA” and "VNWA" to start the VNWA application or use the
Shortcut created on the Desktop.

3.5.3 VNWA Application is started. The following Notice is displayed when VNWA application isstarted for the first time:
Note: To allow for VNWA operation above 500 MHz auto clock multipliers must be selected.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 22 of 32

3.5.4 VNWA Application starts: Click “OK” to acknowledge that VNWA setup information needs tobe entered.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 23 of 32

3.5.5 Now click on Tab "USB Setting"
Check whether VNWA license code is shown, In event this information is missing then enter
the VNWA license code shown on page 2, 2.4 again. Now press “Rescan USB Bus” and “Test
USB Interface” Check bottom line for message “Test passed without errors”

3.5.6 Select “Audio Settings” and set “Audio Capture Device” to “USB Audio Codec”
Set Recording Mode to "Capture" and Recoding Device to "Microphone"
For Factory assembled VNWA 2.x check Reference = Right Channel has been selected.
(Note: Older VNWA 2.x built from a Kit may use "Left Channel" depending on wiring of
the 3.5mm audio connectors) Now press “Test Audio”

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 24 of 32

3.5.7 Select “Audio Level” and ensure whether Reflect Mode is selected. The sinewave levels should
be as displayed below (100% Reflection as TX port is not terminated)
Note: If screen display is different from shown below (for instance when Volume Sliders areshown) then return to previous step 2.6.5 and change Recording Mode and Recording Device.

3.5.8 Connect the TX port and RX port with an SMA cable and check that Left Channel sinewave
has disappeared (No signal reflected) . Now select “Thru” (by pressing on softbutton "Reflect"and check that the Left Channel sinewave is displayed as shown in screenshot below.
Note: when the THRU Mode is active the Audio levels min and max are a little higher than when in
REFL Mode

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 25 of 32

3.5.9 Click “Stop“ cease the Audio test. Next click on "Instrument Settings” and check whether clock
is set to “auto”. Note that for VNWA 2, the clock is generally a little below 36 MHz,
somewhere between 35.855 MHz and 36.000 MHz depending on Crystal fitted.
When performing Audio test press ++ / + or -- / - buttons until the sinewaves are almost
stationary on the screen. If you have a frequency counter perform Calibrate DDS Clockcalibration as described in the VNWA helpfile

3.5.10 Press the top right button to close the VNWA Setup screen. Next we test whether the VNWA
helpfile is available. Press “Help” and select “Help” again. The screen below may be shown.
Press “Yes” and “Browse” to select the location where the Helpfile is located. If standard
VNWA installation defaults were selected then the default path will be:
My Computer C:\VNWA\VNWA.hlp

3.5.11 IMPORTANT: To save the VNWA configuration (after any changes are made) exit the VNWA
Application. This saves the VNWA setup for the next time the VNWA application is started.
End of VNWA Configuration for Windows XP

4.

VNWA Master Calibration

Prior to taking VNWA measurements the VNWA needs to be calibrated - Calibration uses SOLT Short - Open - Load and Through Method. Ideal Calibration standards are used by default. For High
precision measurements in VHF and UHF range you need to specify more precise model of thecalibration standards used - please consult the VNWA helpfile
To calibrate the TX port (One Port) you need an SMA Short connector and and a SMA 50 Ohm
Termination. For RX port (Two Port) calibration a short SMA to SMA cable is also required.
4.1

Master Calibration procedure

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 26 of 32

1. Set frequency range for Master Calibration (for instance 0 MHz - 500 MHz). Click on"Center=" and "Span=" and specify Start and Stop frequency of the Master Calibration
2. From VNWA top menu select "Setting" and select "Sweep"
Specify: 8192 points
time per sample: 3.3 mS

3. From VNWA top menu select "Measure" and "Calibrate" and select Short - Open - Load for
TX calibration and follow the instructions shown on screen. For 2 Port Thru calibration"Crosstalk Cal - Thru Cal - Thru Match Cal" steps also need to be done using a short cable.
4. Once Calibration has been done, save the results in a Master Cal file: for instance withfilename: Master Cal 0-500Mhz. Note: you may save any number of cal files.

4.2

Reloading saved Master Calibration File
To load a previously save Master Calibration File:

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 27 of 32

1.
2.

From Top Menu select "Options" and "Setup" and "Misc. Settings"
Now select the required Calibration file for loading

End of VNWA Master Calibration

5.

Some Practical VNWA Measurements

5.1

Antenna VSWR and Impedance (S11 - 1 Port Device)

Antenna VSWR or Impedance measurements are typical examples of VNWA "One Port
Measurements" The example below illustrates how VSWR measurement is obtained of an (144 MHz)
Antenna from 400MHz to 450MHz (2 Meter Antenna used on 70 cm Band)

Method:(assume VNWA has been calibrated before)
Connect Device Under Test (DUT) ie Antenna to the VNWA TX Socket
1. Set Frequency
Start and Stop frequency
2. Set Measurements - select measurements to be displayed(S11, dB), (S11, Real Z), (S11, Imag Z)
Press"Single Sweep"
results are now displayed
3. Adjust the measurement scale if required to improve the graph appearance
Add Marker (Right Mouse Click and select "Normal Marker") and place on point of interest
In the example the Marker displays: Frequency = 434 MHz, VSWR = 1.61 Real Z = 78.75 Ohm andImag Z = 10.43 Ohm (Inductive)
5.2

VNWA Transmission Line Measurements (2 Port Measurement)

Filter Attenuation measurement is an example of VNWA "Two Port Measurements"
SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 28 of 32

The measurements of attenuation (S21 dB) and input impedance (S11 Z) of a 400 MHz Surface
Acoustic Wave (SAW filter)

Method:(assume VNWA has been calibrated before)
Connect Device Under Test (DUT) input to VNWA TX port and Device output to VNWA RX port1.
Set Start and Stop frequency
2
Select measurements to be displayed (S21, dB), (S11, Real Z), (S11, ImagZ)
Press "Single Sweep" Results should be displayed3
You can change the measurement scale if required to improve graph appearance
4
Add Marker (Right Mouse Click and select "Normal Marker") and place on point ofinterest
5
Graph position may be adjusted by placing curser here and dragging graph up or down.
In the example Marker 1 displays: Frequency = 400.09 MHz Attenuation = -12.05dB Input RealZ =22.28 Ohm and Input imagZ = -5.76 Ohm (capacitive). Note: SAW filter is not matched to 50 Ohm
impedance of VNWA TX and RX port.
5.3

Time Domain Reflectometry Measurement (T D R)

Below is an example of using the TDR capability of the VNWA to determine the characteristic impedance (andlength) of a coaxial cable.

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 29 of 32

Method:
1.
2.
3.
4.
5.

Calibrate VNWA between 0 kHz and 1300 MHz with sweep of 2000 points and 4mSConnect the Coax cable to VNWA TX port. Other end of Coax is left unconnected.
Perform S11 dB measurement as (shown as Trace 1)
Add Trace 2 as |Z| and Time - (Use right hand mouse click and select "Trace Options")
Move curser over Trace 2 (shown as S11 |Z|), Next do right hand mouse click and select"Other" and "Time Domain. Time domain Setting window now appears as in picture.
Change the settings as shown in the example, - Select "Stop Time" to be a little longer
than the length of your coax cable. (10ns shown - for measuring cable length up to 1m)

6.

In the example Marker 1 displays: the characteristic impedance as 49.78 Ohm (each division is 1
Ohm!). The total length of the coax cable is actually 0.63m. Accuracy of this measurements depends
on correct Velocity Factor of the cable type to be entered

6.

VNWA Help file

Check you can open and read the VNWA HELPFILE. Open the VNWA Helpfile and take out time tofamiliarize yourself with the VNWA application: particularly with sections such as:- Installation & Configuration- Calibration- Known Issues- Verification of Proper Operation (Troubleshooting section)
SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 30 of 32

Note: a PDF version of the latest helpfile may be downloaded from:http://sdr-kits.net/DG8SAQ/VNWA/VNWA_HELP.pdf

7.

VNWA User Support

7.1

VNWA User group

It is recommended you become a member of the DG8SAQ VNWA Forum on Yahoo where VNWA
announcements and VNWA user experiences are shared. You will have the benefit of useful information
including advice when new software and helpfile updates are available for download.
http://groups.yahoo.com/group/VNWA/
Please use the DG8SAQ VNWA Forum to check for important announcements, documentation updates and toshare your experiences building and using the DG8SAQ Vector Network Analyser.
7.2

Acknowledgements

Acknowledgements and thanks to all the VNWA 2.x owners who tested 34.x and 35.x releases and patiently
reported issues via the VNWA reflector. A special thanks for Kurt Poulsen OZ7OU for his diligent review of thethis manual and for his sterling work on the VNWA Calibration Standards supplied by SDR-Kits.
Tom Baier and I acknowledge the contribution of Alan Rowe M0PUB for his expertise and efforts in providing theInstaller package.
15th June 2011
Jan Verduyn
Proprietor SDR-Kits

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 31 of 32

9.8

Contact adress

SDR-Kits(Internet & mail order only)
14 Ragleth Grove
Trowbridge, Wilts BA14-7LE
United Kingdom

email: sdrkits@gmail.com
www.SDR-Kits.net

SDR-Kits– VNWA 2.x Self Installer Manual 1.2 - copyright 2011 by SDR-Kits

Page 32 of 32