FN3030
Load Cell Tension and Compression
SPECIFICATIONS

S-Beam Load Cell

Stainless Steel or Aluminum

FEATURES

Optional Rod Ends
Accuracy: 0.1% F.S.
Integrated Amplifier optional
Low Cost

APPLICATIONS

Process Control Equipment
Weighing Calibration Tool
Robotics and Effectors
Laboratory and Research

SENSOR SOLUTIONS /// FN3030_en

50 N to 100,000 N (11.2 lbf to 22,480 lbf)

Cable Gland or Connector Output
Built In Amplifier per Request

The FN3030 measures tension and compression in standard
ranges from 0-50 N to 0-100 000 N. The FN3030 is well suitedfor customization in industrial applications and protection levels
can be upgraded for use in hostile environments. Additionally,for high-level output a model with integrated amplifier isavailable.
With a long-standing experience as a designer and amanufacturer of sensors, TE CONNECTIVITY often works withcustomers to design or customize sensors for specific uses andtesting environments.
To meet your needs, we also offer extensive turnkey
systems. The matched components (sensor, power, amplifierand digital display) are formatted, calibrated and ready for
immediate use.
On request, Instruction documents can be provided toease the selection and use of our sensors and provide helpful
tips.

03/2019

Page 1FN3030

Load Cell Tension and Compression

CHARACTERISTICS (typical values at temperature 23°C)
Ranges (FS) (N)

50100

200

500

1k

2k

5k

10k

20k

50k

100k

Ranges (lbf)

11.2

22.5

45.0

112.4

224.8

449.6

1124

2.25k

4.50k

11.2k

22.5k

Material

Aluminium

Stainless Steel

Stiffness (N/m)

8.0E+05 1.4E+06 2.2E+06 1.4E+07 2.1E+07 3.0E+07 8.1E+07 9.6E+07 1.3E+08 2.0E+08 3.0E+08

Stiffness (lbf/ft)

5.5E+04 9.3E+04 1.5E+05 9.9E+05 1.4E+06 2.1E+06 5.5E+06 6.6E+06 9.1E+06 1.4E+07 2.1E+07
VersionStandard

A1A2

Power supply

10Vdc

10Vdc to 30Vdc±12Vdc to ±18Vdc

Sensitivity (FSO)±20mV±2Vdc ±0.2V±5Vdc ±0.25V

Offset<±1 mV

2.5Vcc ±0.2V

0V ±0.25V

Input Impedance / Consumption

350 ohms<30mA

Output Impedance

350 ohms

1 kohms max

Overrange Without Damage

1.5x FS

Overrange Without Destruction

3x FS

Combined Non-Linearity & Hysteresis< ±0.1% FS

Operating Temperature Range (OTR)-20°C à +80°C (-4 to 176°F)

Compensated Temperature Range (CTR)

0°C à +60°C (32 to 140°F)

Thermal Zero Shift in CTR< 0.5% FS/50°C

Thermal Sensitivity Shift in CTR< 1%/50°C

Insulation> 1000 Mohms

Protection Index

IP50Notes
1. Signal goes positive in tension with standard wiring configuration. Other signal output on request
2. Electrical Termination: Cable gland termination; 2 m cable length standard
3. Material: Body in stainless steel or aluminum alloy depending on F.S., with stainless steel cover
4. Output impedance < 100Ω on request
5. CE conformance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// FN3030_en

03/2019

Page 2

FN3030

Load Cell Tension and Compression

DIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)
Wiring Schematic
Wiring Schematic

Dimensions in mm [inch]
F.S. Ranges in N

50 - 100 - 200

500 – 1k – 2k

5k – 10k

A60[2.36]

60[2.36]

60[2.36]

75[2.95]

110[4.33]

130[5.12]

B

25[0.98]

25[0.98]

25[0.98]

30[1.18]

40[1.57]

55[2.17]

C29[1.14]

29[1.14]

29[1.14]

34[1.34]

40[1.57]

55[1.77]

D60[2.36]

60[2.36]

60[2.36]

60[2.36]

80[3.15]

110[4.33]F (Thread)M6 depth 6

SENSOR SOLUTIONS /// FN3030_en

M12 depth 1203/2019

M12 depth 1220k

M16 x 2depth 1650k

M20 x 1.5
depth20100k

M32 x 2depth 32

Page 3

FN3030

Load Cell Tension and Compression

OPTIONSZ04

CTR -40°C to +90°C (OTR -40°C to +90°C)

Z35

CTR +20°C to +120°C (OTR -20°C to +120°C)

Z36

CTR +20 to +150°C (OTR -20°C to +150°C) - Not available on A1 / A2 versions

HAAccuracy (CNL&H) ≤ ±0.05% F.S. (only on FS 50N to 10kN)

SC

Connector output instead of standard cable gland

L00M

Special cable length, replace “00” with total length in meters

ORDERING INFOFN3030 - A1 - 5KN - /Z04
Options (Z04, L05M,...)
Range in Newton
Amplified version (none, A1 or A2)
Model

SUPPLIED ACCESSOIRES (ONLY WITH SC OPTION)
EFMX-4M: mating plug Jaeger 530-801-006 with clamp 530-841-006 for option /SC; /Z04/SC
EFMX-4H: mating plug Jaeger 530-804-006 with clamp 530-844-006 for option /SC; /Z35/SC; /Z36/SC

RECOMMENDED ACCESSORIESOPTERxxx: Rod Ends
OPTEHxxx: Hemispherical Load Button

TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FN3030_en

All Rights Reserved.

03/2019

Page 4