LASE
neo

brilliance in customized laser solutions

Laser Amplifier

Boost Your Energy
The neoVAN series is an OEM type solidstate laser amplifier system to boost thepulse energy or average output power fora couple of applications. The flexible
system design allows a selection ofdifferent power and energy levels basedon high reliable and long lifetime gain
modules. The ultra-compact and nearlymonolithic modules allow easy integrationand cost effective upgrading of laser
application machines, scientific lasers orlow power oscillators.

No Matter What!
Whether high peak power short pulse
picosecond lasers for micromachining
applications or single frequency radiation
for gravitational wave detection theneoVAN amplifier modules will boost
your application. While the fiber coupled,high gain module allows direct
amplification of mode-locked oscillators,
gain switched or narrow linewidth diodes,
the standard free-space modules scales
microchip lasers either into high average
power or high energy levels.

neoVAN Laser Amplifier
Key features
Output power
Pulse energy
Mode of operation
Beam quality
Amplification factor

5 to 100 Wup to 5mJ
cw to ps-pulses
TEM0,0 / M²<1.3up to 40 dB

Advantages● Easy and compact energy or power boosting● Higly flexible and scalable amplifier units● Proven long term stability and industrial reliability
neoLASE GmbH | Hollerithallee 17 | 30419 Hannover | Germany
Phone: +49 511 515160-0 | E-Mail: info@neolase.com | www.neolase.com

System Specifications
Amplifier module
Typical input power
Typical output power
Wavelength
Beam quality
Power noise
Polarization ratio
Operating regime
Operating voltage
Power consumtion
Ampient temp.
Laser controllerCoolingOptions

||
|
|
|
|
|
|
|
|||
|
|

2P
1 - 100 mW
1 - 10 W

|
|
|

2S

|
1 - 10 W|

10 - 25 W
1064 nm
TEM0,0 / M²<1.3<1 % RMS>100:1
Continuous wave to ps pulses
90 - 240 VAC
max. 350 W / gain module
15 - 35 °C
19'' Rackmount 4 U height
Water cooled
Fiber coupled input, air cooling|

4S25 - 50 W

The shown parameters are examples of standard system combinations, other parameters can be offered onrequest. Visit the neoLASE laser matrix on www.neolase.com for tested amplifier combinations.

Dimensions Amplifier Heads
5,39in137mm
0,16in4mm

Typical Output versus Input Power
1,89in48mm
1,18in30mm

3,15in80mm

0,59in15mm

4-stage

4,66in118,40mm

4,33in110mm

2-stage

1,57in40mm

Top View

1,57in40mm

0,98in25mm

1,18in30mm
0,00in0mm

0,94in24mm

2,20in56mm

3,50in89mm

4,76in121mm

Laser Beam Output/Input

Input configuration:
2P: free-space /single-mode fiber FC connector
2 / 4S: free-space
0,79in20mm

0,79in20mm

Notes: 1. Due to neoLASE continuous product improvement, all specifications are subject to change without
notice.
2. Laser light emitted from this system is invisible and will be harmful to the human eye. Proper laser
safety eyewear must be worn during operation.

neoLASE GmbH | Hollerithallee 17 | 30419 Hannover | Germany
Phone: +49 511 515160-0 | E-Mail: info@neolase.com | www.neolase.com
株式会社 ルクスレイ

URL: http://www.LxRay.jp E-mail: info@LxRay.jp

関西本社 〒663-8113 兵庫県西宮市甲子園口3-28-22
東京事務所 〒113-0033 東京都文京区本郷2-11-6 第1谷口ビル1F

TEL: 0798-31-0500, FAX: 0798-31-0505
TEL: 03-3868-0200, FAX: 03-6912-6394