AVAILABLE
with Signature
Calibration

SEE PG. IN-5

Ultra Precision Gage/Absolute
Pressure Transducers
Model Super TJE

0.05% ACCURACY
TRUE GAGE OR ABSOLUTE

1-888-282-9891

SUPERIOR TEMP. SPECSAMPLIFIED OUTPUT AVAILABLE

www.honeywell.com/sensing

Sensotec Sensors

Model Super TJE is one of the most accurate industrial pressure transducers available today.
The Super TJE features 0.05% FS Accuracy and a zero temperature error of less than 0.0015%
FS/°F. These specifications are maintained by welding a double jacket shell of stainless steel
along with our proprietary “True Gage” second diaphragm that is olates the strain gage circuitry
from atmospheric contamination. High output options of 5 or 10 Vdc and 4-20 mA eliminate theneed for an amplifier card in your data system. An optional internal Signature Calibration chipprovides calibration information for automatic set up with the Model SC four or twelve-channel
digital indicators.

PR-4Dimensions
Absolute (Order Code AP112)
True Gage (Order Code AP111)

1 to 1500 psi

Available Ranges
15; 25; 50 75; 100; 150; 200; 300; 500; 750; 1000;
1500; 2000; 3000; 5000; 7500 psi
10; 15; 25; 50; 75; 100; 150; 200; 300; 500; 750;1000; 1500; 2000; 3000; 5000; 7500 psi

2000 to 7500 psi
Ø1.63 *
[Ø4.14 cm]

Ø1.63[Ø4.14 cm]

2.62 [6.65 cm]
2.62 [6.65 cm*]

PRESSURE PORT
1/4–18 NPT MALE
WITH 3/4" HEX

PRESSURE PORT
1/4–18 NPT FEMALE
WITH 1.00" HEX

Options (See Appendix)
Internal amplifiers: 2a (for replacement only, see page AP-20); 2c (0-5VDC); 2j.
Amplifier enhancements: 3d; Int. shunt cal 8a; Signature calibration 53e
Accessories: Mating connectors and connector/cable assemblies; Pressures port adapters*For 10 and 15 psi (700 mbar and 1 bar) true gage models, the diameter is 2.00” (5.08 cm).

1-888-282-9891

Model Super TJE
True Gage: Order Code AP111
Absolute: Order Code AP112PERFORMANCE

Pressure Ranges..........................
Accuracy (min.) ............................
Output (standard) .........................
Resolution ....................................

10 to 7,500 psi (700 mbar to 500 bar)±0.05% F.S. (BFSL)
2mV/VInfinite

ENVIRONMENTAL

Temperature, Operating ...............
Temperature, Compensated ........
Temperature Effect– Zero (max.)............................
– Span (max.)...........................

-65° F to 250° F (-50° to 120° C)
60° F to 160° F (15° to 70° C)

Strain Gage Type .........................
Excitation (calibration) ..................
Excitation (acceptable) .................
Insulation Resistance ...................
Bridge Resistance ........................
Shunt Calibration Data .................
Wiring Code (std.) ........................
Electrical Termination (std.) .........

MECHANICALINTERNALLY AMPLIFIEDUNITS (Optional)

Media............................................
Overload-Safe ..............................
Overload-Burst .............................
Pressure Port
10 to 1500 psi ..........................
2000 to 7500 psi ......................
Dead Volume................................
Wetted Parts Material...................
Type .............................................
Weight ..........................................
Case Material ...............................

1/4-18 NPT male
1/4-18 NPT female
0.17 cu. in.
17-4 PH Stainless
True Gage or Absolute
12 oz. (Nominal)
17-4 PH StainlessOutputs Available .........................
Additional Length..........................

0-5VDC, 4-20mA
1.12" (28.5 mm)

Gas Liquid
50% over capacity
300% over capacity

ULTRA PRECISION

Mating Connector (not incl.) .........

Bonded foil
10VDC
Up to 12VDC or AC
5000 megohm @ 50VDC
350 ohms
Included#2 (See Pg. AP-8)
PTIH-10-6P or equiv.
(Hermetic stainless)
PT06A-10-6S or equiv.

PRESSURE

ELECTRICAL

0.0015% F.S./ ° F (0.0027% F.S./ ° C)
0.0015% Rdg./ ° F (0.0027% Rdg./ ° C)

NOTES *Gage pressure units greater than 500 psi (35 bar) are sealed at atmospheric pressure.

General Information
How to order (See Pg. AP-19)
Gage/Absolute pressure selection flow chart (See Pg. PR-1)●お問い合わせは下記へ ̶PR-5