FC23Compression Load Cell

SPECIFICATIONSFEATURES

Interchangeable
Low Noise
Robust: High Over-Range Capability
High Reliability
Low Deflection
Low Off Center Errors
Fast Response TimeLong Cycle Life Expectancy

APPLICATIONS

Batch Weighing
Robotics End-Effectors
Variable Force Control
Load and Compression Sensing
Assembly Line Force Measurement
Pumps
Hoist and Winch Loads
Weighing

50 – 2000 lbf Ranges
High Level or mVInterchangeable
Compact Load Button DesignIndustry Standard Packaging
CE Compliance

The FC23 is a high compression force sensor thatcreates new markets previously unrealizable due to costand performance constraints. The FC23 offers normalized
zero and span for interchangeability and is thermallycompensated for changes in zero and span with respect totemperature.
The FC23 incorporates MEAS’ proprietary
Microfused™ technology which employs micromachined
silicon piezoresistive strain gages fused with hightemperature glass to a high performance stainless steel
substrate. Microfused™ technology eliminates agesensitive organic epoxies used in traditional load cell
designs providing excellent long-term span and zerostability. The FC23 measures direct force and is therefore
not subject to lead-die fatigue failure common withcompetitive designs which use a pressure capsuleembedded within a silicone gel-filled cavity. Operating atvery low strains, Microfused™ technology provides anessentially unlimited cycle life expectancy, superior
resolution, and high over-range capabilities.
The FC23 brings your OEM product to life whether
you need thousands or millions of load cells annually.
Although the standard model is ideal for a wide range ofapplications, our dedicated design team at our Load Cell
Engineering Center is ready to provide you with custom
designs for your OEM applications.
Please refer to models FS20 and FC22 for lower force
applications.

SENSOR SOLUTIONS /// FC2301/2020

Page 1

FC23

Compression Load Cell

STANDARD RANGESRange
0 to 0050
0 to 0100
0 to 0250
0 to 0500
0 to 1000
0 to 2000

lbf






PERFORMANCE SPECIFICATIONS
Unless otherwise specified: Supply Voltage: 5.0V, Ambient Temperature: 25°C
PARAMETERS
MIN
TYP
Span (mV)

MAXUNITS

NOTES

95

100105mV

1Span (Amplified)

3.8

4.0

4.2

V1Zero Force Output (mV)

-20020mV

1Zero Force Output (Amplified)

0.3

0.50.7V

1Accuracy (non-linearity, hysteresis, and repeatability)-1

1%Span2

Input Resistance (mV)

kΩ

3.0

Output Resistance (mV)

kΩ

2.2

Temperature Error – Span (Amplified & mV)

-2.5

±1

2.5%Span3

Temperature Error – Zero (Amplified & mV)

-2.5

±1

2.5%Span3

Supply Voltage (mV)

25510V

15.5

V

1

Supply Voltage (Amplified)

4.5

Response Time (10% to 90%)

1.0msLong Term Stability (1 year)±1%SpanMaximum Overload

2.5X

Rated

Compensated Temperature

050°C

Operating Temperature-40+85°C

Storage Temperature-40+85Isolation Resistance (250Vdc)

50Deflection at Rated Load
Humidity

0°CMΩ0.05mm

90%RH

For custom configurations, consult factory.
Notes
1. Ratiometric to supply.
2. Best fit straight line.
3. Maximum temperature error over compensated range with respect to 25°C.

CE Compliance
IEC61000-4-2: [4kV/ 4 kV (Air/Contact)]
IEC61000-4-3: (3 V/m)SENSOR SOLUTIONS /// FC2305/2019

Page 2

FC23

Compression Load CellDIMENSIONSSENSOR SOLUTIONS /// FC2305/2019

Page 3

FC23

Compression Load Cell

CONNECTIONSSENSOR SOLUTIONS /// FC2305/2019

Page 4

FC23

Compression Load Cell

ORDERING INFORMATIONFC23 3 1 – 0000 – 0100 - N
Force Units
Outputs

LN

20 mV/V
0.5 – 4.5 V

13Lbf
Newtons

Multiplier
K

None
X1000

Connection
2ft Cable

1Type0000
SXXX
XXXX

Standard Production
SpecialEWR Customize

Force Range (L)Force Range (N)

0050
0100
0250
0500
1000
2000

0250
0500
1000
2500
50000010 K (Multiplier)TE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred toherein might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard
Terms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or
misuse of the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

SENSOR SOLUTIONS /// FC23

All Rights Reserved.

05/2019

Page 5