XFTC302
Miniature Load Cell High Temperature
SPECIFICATIONSRange from 0-5kN to 0-10kN[0-1.124klbf to 2.148klbf]
Tension and Compression
High Temperature Range : -55 to +175° C[-67 to +347° F]

The XFTC302 series has been specifically developed tomeasure tension and/or compression in static and dynamic
applications.
Fitted with metallic strain gages in a Wheatstonebridge circuit, the XFTC302 has the capacity of measuring
heavy loads of up to 10,000 N, 2000 lbf while providing
excellent temperature stability. With two threaded M10studs, the XFTC302 is easily installed in industrial or OEMapplications. A strain relief spring strengthens the cableoutput.

FEATURES

Wide temperature compensation range
Optional IP rating improvement
Heavy duty
Integrated Amplifier optional
Easy to customized signal and design

For lower force and temperature ranges, TECONNECTIVITY developed another product XFTC301.
This other models covers the ranges from 2N to 2000N.

APPLICATIONS

On request, Instruction documents can be provided
to ease the selection and use of our sensors and provide
helpful tips.

Dynamic strain cylinder regulation
Miniature force & temperature tests
Motor sport clutch monitoring
Robotics regulation
Heat room environment

SENSOR SOLUTIONS /// XFTC302_en

04/2019

Page 1XFTC302

Miniature Load Cell High Temperature

PERFORMANCE SPECIFICATIONS (typical values at temperature 23º C)
Ranges (FS) (N)

5k

10 k

Ranges (lbf)1124.02248.1

Stiffness (N/m)

3.4E+08

6.8E+08

Stiffness (lbf/ft)

2.3E+07

4.7E+07

StandardA1

Power supply

10Vdc

10Vdc to 30Vdc

Sensitivity (FSO)

10 mV±2Vdc ±0.2VOffset<±1 mV

2.5Vdc ±0.2V

Input Impedance / Consumption

700 ohms<30mA

Output Impedance

350 ohms

1 kohms max

Operating Temperature Range (OTR)-55°C to +175°C (-67 to 347°F)-40°C to +85°C (-40 to 185°F)

Compensated Temperature Range (CTR)

0°C to +150°C (32 to 302°F)

0°C to +85°C (32 to 185°F)

VersionOverrange Without Damage

2x FS

Overrange Without Destruction

3x FS

Linearity< ±0.5% FS

Hysteresis< ±0.5% FS

Thermal Zero Shift in CTR< 1% FS/50°C

Thermal Sensitivity Shift in CTR< 1%/50°C

Insulation> 100 Mohms

Protection Index

IP50

Material

Stainkess steel

Temperature restriction notes
Sensors integrating an A1 amplifier have their OTR limited from -40°C to 85°C [-40°F to 185°F] max. due to internal components.
Notes
1. Signal goes positive in tension with standard wiring configuration. Custom outputs available on request
2. Shielded cable with 4 wires (AWG28), standard length 2 m [6.5 ft] with strain relief spring
3. Output impedance standard, available <100Ω on request.
4. CE compliance according to EN 61010-1, EN 50081-1, EN 50082-1

SENSOR SOLUTIONS /// XFTC302_en

04/2019

Page 2XFTC302

Miniature Load Cell High Temperature

DIMENSIONS & WIRING SCHEMATIC (IN METRIC AND IMPERIAL)
Wiring Schematic

Separate data sheets are available for load cells with two female threads
reference XFTC322.

SENSOR SOLUTIONS /// XFTC302_en

04/2019

Page 3

XFTC302

Miniature Load Cell High Temperature

OPTIONS
Z0
Z04
Z1
Z3
Z35
Z36
HA
L5M , L10M , L15M

CTR -40°C to +20°C (-40°F to 68°F)
CTR -40°C to +90°C (-40°F to 194°F) Not available on A1 version
CTR -20°C to +40°C (-4°F to 104°F)
CTR +20°C to +80°C (68°F to 176°F)
CTR +20°C to +120°C (68°F to 248°F) Not available on A1 version
CTR +20°C à +150°C (68°F to 302°F) Not available on A1 version
Accuracy (CNL&H) ≤ ±0.5% F.S.
Special cable length (ex : L5M = 5m length) (Standard cable length 2m )ORDERING INFOXFTC302 - A1 - 100N -/L5M
Options (Z1, HA, L00M, ...)
Range in Newton
Amplifier version (none, A1)
ModelTE.com/sensorsolutions
Measurement Specialties, Inc., a TE Connectivity company.
Measurement Specialties, TE Connectivity, TE Connectivity (logo) and EVERY CONNECTION COUNTS are trademarks. All other logos, products and/or company names referred to herein
might be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes
no warranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s StandardTerms and Conditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuseof the product. Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2015

TE Connectivity Ltd. family of companies

All Rights Reserved.

SENSOR SOLUTIONS /// XFTC302_en04/2019

Page 4