Super Cell

HIGH STIFFNESS / HIGH RESPONSE TYPE LOAD CELL

Flexible
exible cable"Hardly breakable!"e!" - Load cell with an innovative concept!
A strain gauge sensor with phenomenal stiffness and responsiveness.
RobustHigh responsivenessAmazing stiffness is achieved with a strain gauge!
Suitable for applications requiring extra-low deformation duringforce measurements, such as measuring of grinding or
machining forces.

High resonant frequency is possible with strain gauge typesensor!
It can be used to monitor a high-speed phenomenon.

Safe overload of 500%Overload up to 500% of the rated capacity does not affect theaccuracy.
Stable measurement can be performed for a long term even insever loading conditions (e.g. press fitting machines).

Example of use
PLC
Press machine

STARTetc.

OK/NOK
judgment etc.

Displacement sensor

High stiffness &high response load cell

SD cardLoad cell

Indicator

unlPULXE

USBPC

www.unipulse.tokyo/en/Specifications
ModelSCB-‫ڧ‬N

Rated capacity

100 N 200 N 500 N

Rated output

0.4 mV/V±0.2%

1 kN

2 kN

‫ە‬Wiring diagram‫ ڧ‬: Rated capacity

5 kN10 kN 20 kN 50 kN 100 kN 200 kN
0.4 mV/V±0.1%

Safe overload

ٔ EXC (Red)500% R.C.

Zero balance±1% R.O.*1±0.3% R.O.

Non-linearity

0.2 or less R.O.

Hysteresis

0.2 or less R.O.

Repeatability

0.1 or less R.O.

Compensated temperature range

ٕ10 to +60Υ

Safe temperature range

ٕ20 to +70ΥTemperature effect on zero

0.015% R.O./Υ or below

Temperature effect on span(Yellow)

ٕS(Orange)

ٕ EXC (Black)ٔ SIG (Green)ٕ SIG (White)
SHIELD (Blue)

0.01% R.O./Υ or belowInput resistanceApprox. 700 Ω

Approx. 1000 Ω

Output resistanceApprox. 700 Ω

Approx. 1000 Ω

Recommended excitation voltage

5V

10 VMaximum excitation voltage

7.5 V

15 V

Insulation resistance (DC 50 V)

1000 MΩ or more
ȭ2.5 6-conductor flexible cable ȭ5 6-conductor flexible cable
3 m, Cable end: 7 wires
3 m, Cable end: 7 wires

Cable

ٔS

ȭ7 6-conductor flexible cable 3 m
Cable end: 7 wires

Load cell material

Stainless steel

RoHS directives

2011/65/EU (EU)2015/863*1 Specification changed.

External dimension‫ ڦ‬SCB-100N/200N/500N‫ ڦ‬SCB-1KN/2KN/5KN

Load direction
(+)

SR50

ȭ12SR75

252221188.5(3)

ȭ10(3)

(4.5)

SR30

910(1)

ȭ4‫ ڦ‬SCB-10KN/20KN

Load direction(+)

Load direction
(+)

8

ȭ20
ȭ60

ȭ40

P.

C
P.

P.C.D.17

C.

.3.D

2-M2 Depth 3.5

90°

3D.

48
22.5°

4 5°

ȭ2.5 6-conductor
flexible cable 3 m
Cable end: 7 wires *67. 5°

2-M2.5 Depth 4

ȭ5 6-conductor
flexible cable 3 m
Cable end: 7 wires *ȭ7 6-conductor
flexible cable 3 m
Cable end: 7 wires *4-M4 Depth 6‫ ڦ‬SCB-100KN

SR150

ȭ5035

ȭ100C
P.
.8.D°

P.

0C.

D.
80

90

ȭ154°

ȭ7 6-conductor
flexible cable 3 m
Cable end: 7 wires *9 0°

P.C.D.110

22.5°

ȭ7 6-conductor
flexible cable 3 m
Cable end: 7 wires *Model
Deflection at rated capacity Natural frequency Weight
20 g
0.015 mm
40 kHzSCB-100N
20 g
0.015 mm
40 kHzSCB-200N
20 g
0.015 mm
40 kHzSCB-500N
130 g
0.015 mm
26 kHzSCB-1KN
130 g
0.020 mm
26 kHzSCB-2KN
130 g
0.025 mm
26 kHzSCB-5KN
400 g
0.025 mm
26 kHzSCB-10KN
400 g
0.035 mm
26 kHzSCB-20KN
2.0 kg0.065 mm
15 kHzSCB-50KN
2.3 kg0.100 mm
15 kHzSCB-100KN
8.2 kg0.100 mm
8 kHz
SCB-200KN

unlPULXE

2-M6, Effective
depth 12
Not used to fix loadcell main body

22.5°

2 2 .5°

4-M5 Depth 8SR20020

ȭ1009064

1750(5)

ȭ24

Load direction
(+)

45

1340

SR15035(5)ȭ24‫ ڦ‬SCB-200KN

Load direction(+)‫ق‬6 ‫ك‬

Load direction(+)

70‫ ڦ‬SCB-50KN

ȭ7 6-conductor
flexible cable 3 mCable end: 7 wires *4-M5 Depth 10
4-M8 Depth10,
pilot hole depth14
Equal circumference

Unit: mm

* Connector can be attached to the cable according to your requests.
Refer to 84 pages for details.

www.unipulse.tokyo/en/