M210 DISPLAY

SPECIFICATIONSFeatures••••

Analog Output: ±10V
Red LED Display: 2,000countHigh Bandwidth on analogueoutput (1 000 Hz)
Low Noise LevelApplications••••

Instrumentation Laboratory
Test stand
Process monitoring
Rack Mounts

TE CONNECTIVITY SENSORS /// M210_EN•

Signal Conditioning•

Digital display meter•

Analogue output•Casing mounted ABS in option

The M210 electronic is an all-in-one signal conditioning
and display meter that adapts to most strain gauge sensors. The
M210 device powers the sensor, amplifies the signals receivedfrom the sensor, and displays the value in real time.
The analog output is compatible with control systems anddata acquisition cards. A power supply filter helps protect against
incorrect voltage supply. The M210 features high bandwidth andlow noise level.
The large high visibility ±2,000 count LED display allows
quick reading of the measurements.
Optionally, the M210 offers many options including RTC
switches and can be housed in an ABS case.

03/2018

Page 1

M210 DISPLAYPERFORMANCE SPECIFICATIONS

All values are typical at temperature 23°C

General Characteristics
Dimensions (H x L x W)
Panel Cut-out

48 x 96 x 155 mm [1.9 x 3.8 x 6.1 in]

Operating Temperature

0° C to 50° C [32 to 122° F]

Storage Temperature-40° C to 85° C [-40 to 185° F]

Relative Humidity<95% at 40° C [104° F]

92 x 45 mm [3.6 x 1.8 in]Rear panel connectors
Weight600g [1.3lb]

Electrical Characteristics
115/230 Vac ± 10%, 50 to 60 Hz

Standard Operating Power Voltage
Power Supply Filter with Overvoltage Protection
Optional Operating Power Voltage

12, 24, Vdc

Consumption

6VA

Sensor Excitation

10Vdc, 50mA maximum (±5V) (12Vdc with option CD)

Sensor Sensitivity

10 to 1000mV F.S.

Amplification
Span

10 to 1000

Analog Output

10Vdc ± 0.05% Full Scale (FS)

Analogue output accuracy± 0.01% FS

Bandwidth at -3dB

1kHz

Noise Level<10mV peak to peak

Display (1)TypeDisplay accuracy

Red LED Display ± 2000 count display, 14mm
0.2% F.S.

Conversion Rate

2.5 readings/second

Notes
1. The display works as an analogue output copy. If a calibration test is needed, it should be done thru the analogue
output which is more accurate, and not with a display check only.

TE CONNECTIVITY SENSORS /// M210_EN

03/2018

Page 2

+TE CONNECTIVITY SENSORS /// M210_EN

03/2018

Mains or + Supply

Mains or - Supply

Earth

Shield Output- Output (±10 V)+ Output (±10 V)

NC (Optional mA-)

NC (Optional mA+)

+ Excitation- Excitation+ Signal- Signal

ShieldM210 DISPLAY

DIMENSIONS & WIRING SCHEMATIC

Front+Back

Page 3

M210 DISPLAYOPTIONS
12: Excitation 12 Vdc

24: Excitation 24 Vdc

4-20 : Analog Output Current 4 to 20mA
AL2 : Two adjustable threshold levels by internal relay
ABS : House in ABS Case

Type of sensor
Wheatstone bridge sensors
D: Position sensor LVDT

CD: Torquemeter type CD1140; CD9510

G: Pressure sensor

N: Potentiometric position sensor

ORDERING INFORMATION

M210- 12- /ABS
Options (ABS, AL2,..)

Power supply (none, 12, 24 or 48)

Modelte.com/sensorsolutions
Measurement Specialties, MEAS, American Sensor Technologies, AST, TE Connectivity, TE, and the TE Connectivity (logo) are trademarks of the TE Connectivity Ltd. family of companies.
Other logos, products and/or company names referred to herein may be trademarks of their respective owners.
The information given herein, including drawings, illustrations and schematics which are intended for illustration purposes only, is believed to be reliable. However, TE Connectivity makes nowarranties as to its accuracy or completeness and disclaims any liability in connection with its use. TE Connectivity‘s obligations shall only be as set forth in TE Connectivity‘s Standard Terms andConditions of Sale for this product and in no case will TE Connectivity be liable for any incidental, indirect or consequential damages arising out of the sale, resale, use or misuse of the product.
Users of TE Connectivity products should make their own evaluation to determine the suitability of each such product for the specific application.
© 2018TE Connectivity Ltd. family of companies

All Rights Reserved.

TE CONNECTIVITY SENSORS /// M210_EN

03/2018

Page 4