PT2-2472

定置式無線ガス検知部
ＳＤＷＬ-1 シリーズ
ＳＤＷＬ-１ＲＩ(ＴＹＰＥ Ｒ)
ＳＤＷＬ-１ＯＸ(ＴＹＰＥ Ｒ)
ＳＤＷＬ-１ＥＣ (ＴＹＰＥ Ｒ)

取扱説明書〒174-8744 東京都板橋区小豆沢 2-7-6
ホームページ http://www.rikenkeiki.co.jp/

目次
1. 製品のアウトライン .......................................................................................................................................... 4
はじめに ........................................................................................................................................................... 4
SDWL-1RI(TYPE R)の使用目的と使用上の留意事項 ..................................................................................... 5
SDWL-1OX(TYPE R)の使用目的と使用上の留意事項.................................................................................... 5
SDWL-1EC (TYPE R)の使用目的と使用上の留意事項 ................................................................................... 5
危険、警告、注意、注記の定義 ......................................................................................................................... 6
2. 安全上、大切なお知らせ.................................................................................................................................. 72-1. 危険事項 .................................................................................................................................................. 72-2. 警告事項 .................................................................................................................................................. 82-3. 注意事項 ................................................................................................................................................ 103. 製品の構成 ................................................................................................................................................... 113-1. 本体および標準付属品........................................................................................................................... 113-2. 各部の名称と働き .................................................................................................................................. 133-3. ブロックダイアグラム ............................................................................................................................... 16
4. 設置方法....................................................................................................................................................... 174-1. 設置に関する留意事項 ........................................................................................................................... 174-2. システム設計上の留意事項 .................................................................................................................... 184-3. 接点ケーブルの配線方法 ....................................................................................................................... 204-4. 取付方法 ................................................................................................................................................ 25
4-5. アンテナモジュールの取り付け ............................................................................................................... 30
4-6. 電波輻射の調整治具の取り付け ............................................................................................................ 30
4-7. 無線通信 ................................................................................................................................................ 31
5. 使用方法....................................................................................................................................................... 35
5-1. ご使用にあたって ................................................................................................................................... 35
5-2. 始動準備 ................................................................................................................................................ 35
5-3. 始動方法 ................................................................................................................................................ 35
5-4. 操作方法 ................................................................................................................................................ 37
5-5. 警報値設定の確認 ................................................................................................................................. 38
5-6. 基本動作フロー ...................................................................................................................................... 39
5-7. ユーザーモード ....................................................................................................................................... 425-8. ユーザーモードへの入り方 ..................................................................................................................... 436. 警報動作....................................................................................................................................................... 526-1. ガス警報動作、酸素警報動作 ................................................................................................................. 52
6-2. 故障警報動作 ........................................................................................................................................ 54
6-3. 正確な警報を行うための機能 ................................................................................................................. 55
7. 保守点検....................................................................................................................................................... 57
7-1. 点検の頻度と点検項目........................................................................................................................... 57
7-2. メンテナンス（定期点検）モード ................................................................................................................ 59
7-3. メンテナンスモードへの入り方 ................................................................................................................. 60
7-4. ガス校正の準備 ..................................................................................................................................... 62
7-5. テストモード ............................................................................................................................................ 63
7-6. ガス校正方法 ......................................................................................................................................... 69
7-7. 環境設定 1 ............................................................................................................................................. 74
7-8. 環境設定 2 ............................................................................................................................................. 78
7-9. 部品の交換 ............................................................................................................................................ 83
8. 保管・移設および廃棄について ...................................................................................................................... 90
8-1. 保管または長期間使用しない場合の処置 ............................................................................................... 90
8-2. 移設、または再度使用する場合の処置 ................................................................................................... 91
8-3. 製品の廃棄 ............................................................................................................................................ 91
9. トラブルシューティング ................................................................................................................................... 92
9-1. 機器の異常 ............................................................................................................................................ 92
9-2. 指示値の異常 ........................................................................................................................................ 93
10. 製品仕様 .................................................................................................................................................... 94-2-10-1. 製品仕様一覧 ...................................................................................................................................... 9410-2. 標準付属品 .......................................................................................................................................... 9910-3. 特別付属品（オプション品） ................................................................................................................... 99
11. 付録 .......................................................................................................................................................... 10011-1. 検知原理 ............................................................................................................................................ 10011-2. 用語の定義 ........................................................................................................................................ 102-3-

製品のアウトライン

はじめに

1

製品のアウトライン
はじめに
この度は、定置式無線ガス検知部 SDWL-1RI(TYPE R)／SDWL-1OX(TYPE R)／SDWL-1EC (TYPE R)（以降「本器」）をお買い上げいただきありがとうございます。本器は、非防爆構造の機器です。お買い求めの製品型番と本取扱説明書に記載の仕様を照合し、ご確認をお願いします。
本書は、SDWL-1RI(TYPE R)、SDWL-1OX(TYPE R)、SDWL-1EC (TYPE R)、3 機種の取扱方法と仕様を説明したものです。本器を正しく使用していただくために必要な事項が記載されています。
各機種の記載について説明が異なる内容については、それぞれ下記のアイコンで示しています。

SDWL-1RI(TYPE R)

RI

SDWL-1OX(TYPE R)

OX

SDWL-1EC(TYPE R)

EC

アイコンのない操作方法や仕様は、3 機種が共通する内容となっています。初めて使用される方はもちろん、既に使用されたことのある方も、知識や経験を再確認する上で、よくお読みいただき、内容を理解した上で使用してください。
弊社では、本器あるいはその付属品に対して、以下の行為を禁止しています。
（１） 改変および修正（２） 分解や解析（３） 逆アセンブルおよび逆コンパイルなど、すべての解析（４） 第三者への譲渡、販売および担保提供（５） 貸与、再使用許諾その他名目の如何を問わず、第三者に使用させること
なお、製品改良のために、この説明書の内容を将来予告なしに変更することがあります。また、この説明書の全部または一部を無断で複写または転載することを禁じます。
保証期間の内外を問わず本器を使用することによって生じたいかなる事故および損害の補償はいたしません。
保証書に記載されている保証規定を必ずご確認ください。

-4-

製品のアウトライン

SDWL-1RI(TYPE R)の使用目的と使用上の留意事項

SDWL-1RI(TYPE R)の使用目的と使用上の留意事項
RI
SDWL-1RI(TYPE R)は、大気中に漏洩した可燃性ガスを検知する定置式のガス警報器です。
検知したガス濃度値などのパラメータは、ISA100.11a に対応した無線通信機能により、上位システム（アクセスポイントやゲートウェイなど）へ伝送することができます。また、本器は警報接点を 3 つ内蔵しており、ガス警報(ALM1、ALM2)と故障警報(FAULT)に使用することができます。ご使用時には、以下の点に留意してください。
1. 本器は、検知対象ガス以外のガス、蒸気による干渉を受けますので、干渉による警報作動に注意してください。また、設置場所の環境変化（温度・湿度など）により変動することがあります。
2. 警報の設定は機器の性能に見合う範囲で使用してください。高圧ガス保安法対応設備では、弊社標準警報
設定値未満の警報設定の場合には、誤警報の原因となることがあります。
3. 本器は保安機器であり、制御機器ではありません。本器の警報接点出力は、外部警報ランプ・ブザー用に使用してください。それ以外の制御用に使用した場合、誤作動などに対して当社は一切を補償しません。
4. 本器の保守に際しては、取扱説明書に記載されている定期交換部品の交換調整を含めた定期点検が必要です。また本器は保安機器ですので、法令に準拠して 6 ヶ月ごとの定期点検とガス校正を推奨しています。
本器は、検知結果により、生命・安全の保障をするものではありません。

SDWL-1OX(TYPE R)の使用目的と使用上の留意事項
OX
SDWL-1OX(TYPE R)は、大気中の酸素を検知する酸欠警報器です。
検知した酸素濃度値などのパラメータは、ISA100.11a に対応した無線通信機能により、上位システム（アクセスポイントやゲートウェイなど）へ伝送することができます。また、本器は警報接点を 3 つ内蔵しており、ガス警報(ALM1、ALM2)と故障警報(FAULT)に使用することができます。ご使用時には、以下の点に留意してください。
1. 本器は、大気圧の変動に対し指示が若干変動します。特に台風などによる低気圧時には警報作動に注意してください。また、設置場所の環境の変化（温度・湿度など）によっても変動することがあります。
2. 警報の設定は機器の性能に見合う範囲で注意してください。標準警報設定値は 18.0%です。
3. 本器は保安機器であり、制御機器ではありません。本器の警報接点出力は、外部警報ランプ・ブザー用に使用してください。それ以外の制御用に使用した場合、誤作動などに対して当社は一切を補償しません。
4. 本器の保守に際しては、取扱説明書に記載されている定期交換部品の交換調整を含めた定期点検が必要です。また保安機器ですので、6 ヶ月ごとの定期点検とガス校正を推奨しています。
本器は、検知結果により、生命・安全の保障をするものではありません。

SDWL-1EC(TYPE R)の使用目的と使用上の留意事項
EC
SDWL-1EC (TYPE R)は、大気中の毒性ガスを検知するガス警報器です。
検知したガス濃度値などのパラメータは、ISA100.11a に対応した無線通信機能により、上位システム（アクセスポイントやゲートウェイなど）へ伝送することができます。また、本器は警報接点を 3 つ内蔵しており、ガス警報(ALM1、ALM2)と故障警報(FAULT)に使用することができます。ご使用時には、以下の点に留意してください。
1.
2.

3.

本器は、検知対象ガス以外のガス、溶剤蒸気などにより干渉を受けます。干渉による警報作動に注意してください。また、設置場所の環境の変化（温度・湿度など）によって変動することがあります。
本器は保安機器であり、制御機器ではありません。本器の警報接点出力は、外部警報ランプ・ブザー用に使用してください。それ以外の制御用に使用した場合、誤作動などに対して当社は一切を補償しません。
警報の設定は機器の性能に見合う範囲で使用してください。高圧ガス保安法対応設備では、弊社標準警報設定値（じょ限量）未満の警報設定の場合には、誤警報の原因となることがあります。
-5-

製品のアウトライン
4.

5.

危険、警告、注意、注記の定義

本器のガス検知センサのガス接触部は多孔質高分子膜のため、溶剤の影響により膜の撥水性が低下し、内部電解液が浸み出す恐れがあります。
検知部の近傍では溶剤を使用しないでください。止むを得ず、溶剤を使用される場合は、使用時間中と使用終了後の約１時間の間は、弊社推薦のフィルタをガス警報器の吸入口などに装着してください。
本器の保守に際しては、取扱説明書に記載されている定期交換部品の交換調整を含めた定期点検が必要です。また保安機器ですので、法令に準拠して 6 ヶ月ごとの定期点検とガス校正を推奨しています。

本器は、検知結果により、生命・安全の保障をするものではありません。

危険、警告、注意、注記の定義
本取扱説明書では、安全かつ効果的な作業が行えるように、次の見出しを使用しています。

危険取り扱いを誤った場合、「人命、人体または物に重大な被害を及ぼすことが想定される」ということを意味します。

警告

取り扱いを誤った場合、「身体または物に重大な被害を及ぼすことが想定される」ということを意味します。

注意

取り扱いを誤った場合、「身体または物に軽微な被害を及ぼすことが想定される」ということを意味します。

注記

取り扱い上のアドバイスを意味します。

-6-製品の構成

本体および標準付属品

3製品の構成3-1. 本体および標準付属品
梱包箱を開けて、本器と付属品を確認します。
万一、足りないものがありましたら、販売店または最寄りの弊社営業所までお問い合わせください。

3-1-1. 本体
本器の各部の名称と働きおよび LCD 表示は、「3-2 各部の名称と働き」（P.13）を参照してください。

＜本体＞(ケーブルグランド含む)

3-1-2. 標準付属品
RI

OXEC

取扱説明書・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・1 冊
ケーブル結線用操作レバー・・・・・・・・・・・・・・・・・・・・・・・・・・1 個
U ボルトユニット ・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・1 個取付アングルセット・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・1 個
六角棒スパナ（対辺 4）・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・(※1)
六角棒スパナ（対辺 2）・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・(※1)
コントロールキー・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・(※2)
バッテリーケース【リチウム電池 単 1 形（2 本）】・・・・・・・・・1 個アンテナコネクタ用キャップ ・・・・・・・・・・・・・・・・・・・・・・・・・1 個- 11 -

設置方法

無線通信

4-7-2. 赤外線通信の準備（IrDA 通信モード移行）
プロビジョニングを実施するための赤外線（IrDA）通信モードに移行します。

RI

OX
共通画面

1 検知待機状態で、MENU/ESC キーを長押しする（3 秒間）

A1

○

A2○

F

○

2 ユーザーモードで、▼/ALM2 キーを押して、M MODE（メンテンナスモード）を選択する

A1

○

A2○

F

○

3

A1

○

A2○

F

○

A1

○

A2

○

F

○

A1

○

A2○

F

○

SET/ALM1 キーを押す

4 続けて、もう一度SET/ALM1 キーを長押しする（3 秒間）
メンテナンスモードに入ります。

5▲ または▼/ALM2 キーを押して、SET 2（環境設定 2）を選択する- 32 -EC

設置方法

6

SET/ALM1キーを押す

無線通信

A1○A2○F○

7▲ または▼/ALM2 キーを押して、IRDA（IrDA 通信モード移行）を選択する

8SET/ALM1キーを押す
IrDA 通信モードに移行します。

A1

○A2○F

○

注記
IrDA 通信モードを解除するには
 IrDA 通信モードを解除するときは、MENU/ESC キーを押してください。

- 33 -

使用方法

警報値設定の確認

5-5. 警報値設定の確認
本器に設定されている警報設定値を確認します。

5-5-1. ALM1 の警報設定値の確認
下記手順で、A1、A2、F は警報ランプ ALM1、ALM2、故障ランプ FAULT、●は点灯、○は消灯を示します。
ここでは、ALM1 の警報設定値を確認します。

1 検知待機状態で、SET/ALM1 キーを押す

ALM1 の警報値が表示されます。
SET/ALM1 キーからコントロールキーを放すと、検知待機状態に戻ります。

A1

○

A2○F

○

A1○

A2○

F

○RIOX

EC

RIOX

EC

5-5-2. ALM2 の警報設定値の確認
ここでは、ALM2 の警報設定値を確認します。

1 検知待機状態で、▼/ALM2 キーを押す

ALM2 の警報値が表示されます。
SET/ALM2 キーからコントロールキーを放すと、検知待機状態に戻ります。

A1○

A2○

F

○

A1○

A2○

F

○- 38 -

使用方法

ユーザーモードへの入り方

5-8. ユーザーモードへの入り方
ここでは、ユーザーモードへの入り方と、表示される画面について説明しています。

RI
1 検知待機状態で、MENU/ESC キーを長押しする（3 秒間）

A1○

A2○

F

○

A1○

A2○

F○EC

A1○

ゼロ調整を実施できます。

A2○F○OX

EC

ユーザーモード（ROM/SUM 表示）に切り替わり、プログラムのバージョンが表示されます。

2 ユーザーモードで、▼/ALM2 または ▲キーを押す表示項目が切り替わります。

RIゼロ調整⇒P.46

ゼロ調整⇒P.46

OX

A1

○

スパン調整を実施できます。

A2

○F

○
スパン調整⇒P.48

設定値表示
各種設定値を表示します。
⇒P.50

A1○

A2○F

○

メンテナンスモード
メンテナンスモードに切り替えます。
⇒P.60

A1

○

A2

○

F

○- 43 -

使用方法

ユーザーモードへの入り方

5-8-2. ゼロ調整
ゼロ調整を行うことにより、ガス濃度を正確に測定します。

RI
1 検知待機状態で、MENU/ESC キーを長押しする（3 秒間）

A1

○

A2

○

F

○

2 ゼロ調整用ガスを供給する
以降の操作は、指示が安定してから行ってください。

3 ユーザーモードで、▼/ALM2 キーを押して、ZERO（ゼロ調整）を選択する

A1

○A2○F

○

4A1○

A2○

F

○

A1○

A2○

F

○

A1○

A2○

F

○

SET/ALM1 キーを押す
現在のガス濃度値が表示されます。

5SET/ALM1 キーを押す
ゼロ調整が実行されます。

ゼロ調整が実行されている間には、「CAL」が表示されます。

- 46 -EC

使用方法

ユーザーモードへの入り方

5-8-4. 設定値表示
各種設定値を確認することができます。

RI
1 検知待機状態で、MENU/ESC キーを長押しする（3 秒間）

A1

○

A2○

F

○

2 ユーザーモードで、▼/ALM2 キーを押して、CONF（設定値表示）を選択する

A1

○

A2○

F

○

3

A1○

A2○F

○

A1○

A2○

F

○

A1○

A2○

F○

A1○

A2○

F

○

SET/ALM1 キーを押す設定値表示に切り替わります。

4▲ または▼/ALM2 キーを押す
ALM1 の警報点を表示します。

ALM2 の警報点を表示します。

警報遅延時間（秒）を表示します。

- 50 -OX

EC

警報動作

ガス警報動作、酸素警報動作

6-1-2. 接点動作
警報設定値を超えたガス濃度もしくは酸素濃度が検知された場合、接点は以下の図のように動作します。また、警報設定値未満になると、接点動作は自動で復帰します。

濃度

警報設定値（ALM1）

0時間

警報ランプ(ALM1)
警報接点(端子台 1、2)

警報遅延時間(2 秒)

上記の図は、ガス濃度が ALM1 の警報設定値を超えて、時間の経過と共にガス濃度が低くなった場合の遷移を示しています。

6-1-3. 警報時の対応
警報が発報されたときの対応は、お客様の管理ルールに従い、速やかに対応してください。
一般的には、以下のような対応を行っています。
本器の指示値の確認をする

警報管理濃度に基づき、監視区域から人を遠ざけて安全を確保する

注記
 瞬間的なガス漏れの場合、確認した時点では既に指示が低下している場合があります。ノイズや偶発な条件など、ガス警報以外でも、一時的な警報状態になったときに指示が低下している場合があります。

濃度表示が継続している場合は、原因となるガス（酸素）の元栓を閉じて、濃度指示が低下したことを確認する

万一、ガスが残っていることを想定し、危険を回避できる装備をした上でガス（酸素）漏洩現場に行き、ポータブルのガス検知器などによりガスの残存状況を確認する

危険のないことを確認し、ガス漏洩に対する処置を施す- 53 -

保守点検

メンテナンスモードへの入り方

7-3. メンテナンスモードへの入り方
ここでは、メンテナンスモードへの入り方と、表示される画面について説明しています。

RI

OX

EC

共通画面
1 検知待機状態で、MENU/ESC キーを長押しする（3 秒間）

A1

○

A2○F

○

2 ユーザーモードで、▼/ALM2 キーを押して、M MODE（メンテナンスモード）を選択する

A1○

A2○

F

○

3A1○

A2○

F

○

4 続けて、もう一度SET/ALM1 キーを長押しする（3 秒間）

A1○

A2○

メンテナンスモードに入ります。

F○▲ または▼/ALM2 キーを押して、設定項目を選択する

A1○

A2○

F

○

5SET/ALM1 キーを押す

テストモード
各種テストを行います。
ガステスト⇒P.63
警報テスト⇒P.65
故障テスト⇒P.67
LED・LCD 表示テスト⇒P.68- 60 -保守点検

メンテナンスモードへの入り方
A1○

A2○

F

○

A1○

A2○

F

○

A1○

A2○

F

○

A1○

A2○

F

○

A1○

A2○

F

○

A1○

A2○

F

○

A1○

A2○

F

○

ゼロ調整
ゼロ調整を行います。
⇒P.69

スパン調整
スパン調整を行います。
⇒P.71

ゼロ・スパン値初期化
センサ交換後に、ゼロ調整、スパン調整の初期化を行います。
⇒P.73

環境設定 1
各種環境設定を行います。
⇒P.74

環境設定 2
各種環境設定を行います。
⇒P.78

各種電気設定表示
お客様においては使用しません。

各種電気設定表示
お客様においては使用しません。

- 61 -保守点検

テストモード

7-5. テストモード
ここでは、ガステスト、警報テスト、故障テスト、LCD テストについて、説明しています。

7-5-1. ガステスト
ガスによるテストを行います。

RI
1 メンテナンスモードで、▲ または ▼/ALM2 キーを押して、TEST（テストモード）を選択する

2 ▲ または ▼/ALM2
キーを押して、GAS T（ガステスト）を選択す
る

3 SET/ALM1 キーを押す

4 テストガスを導入する

RI

EC

OX

A1

○

A2○

F

○

A1○

A2○

F○A1○

A2○

F○

A1●

A2○

F

○

A1●

A2●F

○

5 テストガスを止める- 63 -OX

EC保守点検

テストモード

7-5-2. 警報テスト
指示値を警報点まで上げて、警報が発報されるかどうかのテストを行います。

RI
1 メンテナンスモードで、▲ または ▼/ALM2 キーを押して、TEST（テストモード）を選択する

2 ▲ または ▼/ALM2キーを押して、AL T（警報テスト）を選択す
る

3 SET/ALM1 キーを押す

A1

○

A2○

F

○

A1

○

A2○

F

○

4 ▲ または ▼/ALM2 キーを押して、接点出力の ON/OFF を選択する
ON を選択すると、警報テスト時に接点を出力します。

5 ▲ または ▼/ALM2キーを押して、指示値を上下させる

A1○A2○

警報が発報されることを確認してください。

F○RIA1●

A2○F○

A1●

A2●F○

EC

OX

6 警報テストが終わったら、- 65 -OX

EC保守点検

ガス校正方法7-6-2. スパン調整
ゼロ調整を行うことにより、ガス濃度を正確に測定します。

RIOX

1 スパン調整用ガスを供給する
以降の操作は、指示が安定してから行ってください。

2 メンテナンスモードで、▲または ▼/ALM2 キーを押して、SPAN（スパン調整）を選択する

3

SET/ALM1キーを押す

A1

○

A2○

F

○

A1○

A2○F

○

A1○

A2

○

▲ または
▼/ALM2 キーを押して、合わせてください。

F○

スパン調整が実行されている間には、「CAL」が表示されます。

A1○A2○F○

現在のガス濃度値が表示されます。

4

SET/ALM1キーを押す
10 ％ F.S. 未 満 では 、 SET/ALM1キーは押せません。

5 指示値を導入したガス濃度に合わせる- 71 -EC

保守点検

環境設定 1

7 操作終了後はMENU/ESC キーを長押しする検知待機状態に戻ります。

注記 手順 6 で検知待機状態の画面に戻った場合、INHIBIT が設定されていると画面下部に「INHI」と表示されています。

RI
A1

○

A2○

F

○OX

- 75 -EC

保守点検

環境設定 1

7-7-2. 警報設定値設定ガス警報 ALM1 と ALM2 の警報値を設定します。警報設定値の確認操作について詳しくは、「5-5 警報値設定の確認」（P.38）を参照してください。

RI
1 メンテナンスモードで、▲ または ▼/ALM2 キーを押して、SET 1（環境設定 1）を選択する

2 ▲ または ▼/ALM2

A1○A2○F○

3 SET/ALM1 キーを押す
AL 1（ALM1）の警報値設定が表示されます。

A1

○A2○

F

○4 ▲ または ▼/ALM2A1○キーを押して、ALM1または ALM2 から設定する警報を選択する

A2○F○キーを押して、AL P（警報設定値表示）を選択する

5 SET/ALM1 キーを押す
ここでは、AL 1（ALM1）の警報値設定例を示しています。

6 ▲ または ▼/ALM2キーを押して、設定値にする

A1

○A2○

F

○- 76 -OXEC