A world-wide
success story
ISO9001

ISO14001（HQ only）

91
106.2

9648

91
106.2

3 types of controller selectablebased on installation space110□961102-point SV externally selectable
CE, UL conformity

Indicating Controller

Digital Indicating Controllers
JC series■ Model nameＪＣ□ -33Ａ-□ /□ □，□□□Ｓ
Ｒ
Ｄ
Ｒ
Control output
Ｓ(OUT1)
ＡInputＭSeriesnameSupply voltageOption

W48 x H48 x D95mm
W48 x H96 x D98.5mm
W96 x H96 x D98.5mm
Relay contact
Non-contact voltage (for SSR drive)
DC current
Multi-input
100 to 240V AC
24V AC/DC (*1)
１Ａ２Alarm 2 (A2)Ｗ
（ 5A） Rated current: 5A
HeaterＷ（10A） Rated current: 10A
burnoutＷ（20A） Rated current: 20A
alarm (*2)
Ｗ（50A） Rated current: 50AＤＲ(*3)
ＤＳ(*3) Control output (OUT2)
ＤＡ(*3) (Heating/Cooling control output)
ＤＴ(*4)
Serial communication
Ｃ５(Based on EIA RS-485) (*5)
ＳＭ
SV1/SV2 external selection (*4)
ＬＡ
Loop break alarm
Ｐ２４ Insulated power supply output (*3)
ＴＣ
Terminal cover
ＢＫ
Color Black■ Rated scale
Input type-200 to 1370 ℃-199.9 to 400.0 ℃-200 to 1000 ℃
ＪＲ
0 to 1760 ℃
Ｓ
0 to 1760 ℃
Thermocouple Ｂ
0 to 1820 ℃-200 to 800 ℃
Ｅ-199.9 to 400.0 ℃
Ｔ-200 to 1300 ℃
Ｎ
PL-Ⅱ
0 to 1390 ℃
C（W/Re5-26）
0 to 2315 ℃Ｋ(*1): Standard voltage is 100 to 240V AC.
Only when ordering 24V AC/DC,enter [1] after the input code.
(*2): For DC current output type, [W] option cannot be added.
(*3): Can be added only to the JCR-33A and JCD-33A.
(*4): Can be added only to the JCS-33A.
(*5): When [C5] option is added to the JCR-33A or JCD-33A,
SV1/SV2 external selection cannot be used.

[Option combination for the JCS-33A]
Combination 1
Combination 2
Combination 3
Combination 4Ｗ
○×○
×ＤＴ
×○
×
○

Ｃ５○
○
××

ＳＭ
××
○
○

ＬＡ
○
×
○
×

ＴＣ
○
○
○
○

ＢＫ
○
○
○
○

ＴＣ
○
○
○
○

ＢＫ
○
○
○
○

[Option combination for the JCR-33A and JCD-33A]
Combination 1
Combination 2
Combination 3
Combination 4

Scale-320 to 2500 °F-199.9 to 750.0 °F-320 to 1800 °F
0 to 3200 °F
0 to 3200 °F
0 to 3300 °F-320 to 1500 °F-199.9 to 750.0 °F-320 to 2300 °F
0 to 2500 °F
0 to 4200 °F

Ａ２○×○
×Ａ２○○×○

Input type
RTD

Pt100
JPt100

Ｗ
○×○
×Ｄ□
×○
○
×Ｃ５○
○
○
○

ＬＡ
○
○
×
○

P24×××○Scale-200 to 850 ℃-300 to 1500 °F-199.9 to 850.0 ℃-199.9 to 999.9 °F-200 to 500 ℃-300 to 900 °F-199.9 to 500.0 ℃-199.9 to 900.0 °F

4 to 20mA DC
0 to 20mA DC-1999 to 9999，-199.9 to 999.9
0 to 1V DC-19.99 to 99.99，-1.999 to 9.999
0 to 10V DC
DC voltage
1 to 5V DC(*2)
0 to 5V DC(*1): For DC current input, connect 50Ωshunt resistor (sold separately)
externally.
(*2): For DC current input and DC voltage input, scaling and decimal point
place are changeable.
DC current(*1)■ Standard specifications
Display

Input

Accuracy(Setting, Indication)
Input sampling period
Control output(OUT 1)

Control action

Alarm 1 (A1)

JCS-33A: PV [Red 4 digits, Character size: 10.2 x 4.9mm (H x W)], SV [Green 4 digits, Character size: 8.8 x 4.9mm (H x W)]
JCR-33A: PV [Red 4 digits, Character size: 11.2 x 5.4mm (H x W)], SV [Green 4 digits, Character size: 11.2 x 5.4mm (H x W)]
JCD-33A: PV [Red 4 digits, Character size: 18 x 8mm (H x W)], SV [Green 4 digits, Character size: 12.6 x 6mm (H x W)
Thermocouple : K, J, R, S, B, E, T, N, PL-Ⅱ, C (W/Re5-26) External resistance: 100Ω or less (However, for B input: 40Ω or less)
RTD: Pt100, JPt100 3-wire system (Allowable input lead wire resistance per wire: 10Ω or less)
DC current: 0 to 20mA DC, 4 to 20mA DC Input impedance: 50Ω (Connect shunt resistor 50Ω between input terminals.)
Allowable input current: 50mA or less (When shunt resistor 50Ω is used)
DC voltage: 0 to 1V DCInput impedance: 1MΩ or more
Allowable input voltage: 5V or less
Allowable signal source resistance: 2kΩ or less0 to 5V DC, 1 to 5V DC, 0 to 10V DC Input impedance: 100kΩ or more
Allowable input voltage: 15V or less
Allowable signal source resistance: 100Ω or less
Thermocouple: Within ±0.2% of each input span ±1 digit or ±2℃(4°°F) whichever is greater
However, R or S input, 0 to 200℃(0 to 400°F) : Within ±6℃(12°°F)
B input, 0 to 300℃(0 to 600°°F) : Accuracy is not guaranteed.
K, J, E, T, and N input , less than 0℃(32°F) : Within ±0.4% of each input span ±1 digit
RTD: Within ±0.1% of each input span ±1 digit or ±1℃(2°°F) whichever is greater
DC current, DC voltage : Within ±0.2% of each input span ±1 digit
250ms
Relay contact: 1a1b (JCS-33A: 1a), 3A 250V AC (Resistive load), 1A 250V AC (Inductive load cosφ=0.4), Electric life: 100,000 cycles
Non-contact voltage : 12+20 V DC Max. 40mA (Short circuit protected)
DC current: 4 to 20mA DC Load resistance: Max. 550ΩActions mentioned below can be selected by keypad. [Default: PID]
PID (with auto-tuning function), PI, PD (with auto reset function), P (with auto reset function), ON/OFFOUT1 proportional band (P) : Thermocouple: 0 to 1000℃(0 to 2000°°F) (ON/OFF action when set to 0)
RTD: 0.0 to 999.9℃(0.0 to 999.9°°F) (ON/OFF action when set to 0.0)
DC current and DC voltage: 0.0 to 100.0% (ON/OFF action when set to 0.0)
Integral time (I): 0 to 1000 seconds (OFF when set to 0)
Derivative time (D): 0 to 300 seconds (OFF when set to 0)
OUT1 proportional cycle: 1 to 120 seconds (Not available for DC current output type)
OUT1 ARW: 0 to 100%Hysteresis: Thermocouple and RTD: 0.1 to 100.0℃(°°F)
DC current and DC voltage: 1 to 1000 (The placement of the decimal point follows the selection.)
OUT1 output limit: -5 to 105%
Alarm action and Energized/De-energized can be selected by keypad. [Default: No alarm]
・No alarm
・High limit alarm(Deviation setting) Setting range: -(Input span) to Input span・Low limit alarm(Deviation setting) Setting range: -(Input span) to Input span・High/Low limits alarm(Deviation setting) Setting range: 0 to Input span・High/Low limit range alarm(Deviation setting) Setting range: 0 to Input span・Process high alarm
Setting range: Input range low limit value to Input range high limit value
・Process low alarm
Setting range: Input range low limit value to Input range high limit value
・High limit alarm w/standby(Deviation setting) Setting range: -(Input span) to Input span・Low limit alarm w/standby(Deviation setting) Setting range: -(Input span) to Input span・High/Low limits alarm w/standby (Deviation setting) Setting range: 0 to Input span

When input uses a decimal point, the negative low limit value is -199.9 and the positive high limit value is 999.9.
When input is DC current or DC voltage, input span is scaling span.
When input is DC current or DC voltage, input range low (or high) limit value is scaling low (or high) limit value.
Setting accuracy : The same as the indicating accuracy
Alarm 1 (A1)
Action: ON/OFF action
Hysteresis: Thermocouple and RTD: 0.1 to 100.0℃(°F)
DC current and DC voltage: 1 to 1000 (The placement of the decimal point follows the selection.)
Output: Relay contact 1a, 3A 250V AC (Resistive load), Electric life: 100,000 cycles
Selects SV1 or SV2 from the external contact. (For JCS-33A, [SM] option needs to be added.)
SV1/SV2
SV1: Contact open between terminals 14 and 17 SV2: Contact closed between terminals 14 and 17
external selection
Supply voltage
100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz Allowable voltage fluctuation range: 85 to 264V AC, 20 to 28V AC/DCPower consumption
Approx. 8VA
When control output (OUT1) is Non-contact voltage or DC current output with [DS or DA] option is added, insulation test between
Control output (OUT1) terminal and Heater burnout alarm output terminal, between Control output (OUT1) terminal and Controloutput (OUT2) terminal, between Control output (OUT1) terminal and Insulated power output terminal must not be carried out.
Insulation resistance When control output (OUT1) is Non-contact voltage or DC current output, insulation test between Control output (OUT1) andSV1/SV2 external switch terminal, between control output (OUT1) and communication terminal must not be carried out.
When control output (OUT2) is Non-contact voltage or DC current output, insulation test between Control output 2 (OUT2) andSV1/SV2 external switch terminal, between control output (OUT2) and communication terminal must not be carried out.
Other combinations: 10MΩ or more, at 500V DC
1.5kV AC for 1min between input terminal and ground terminal, between input terminal and power terminal
1.5kV AC for 1min between power terminal and ground terminal
Dielectric strength
1.5kV AC for 1min between output terminal and ground terminal, between output terminal and power terminal
Environment
Ambient temperature: 0 to 50℃ Ambient humidity: 35 to 85%RH (Non-condensing) Drip-proof/Dust-proof (IP66 for the front panel)
Safety standard
UL: Power input rating 100 to 240V AC, 24V AC/DC File No. E159038
Case (Material, Color) Material: Flame-resistant resin Color: Light gray
Mounting
Screw type mounting brackets. (Mountable panel thickness:1 to 8mm)
Setting
Sheet key input
External dimensions JCS-33A: W48 x H48 x D95mm, JCR-33A: W48 x H96 x D98.5mm, JCD-33A: W96 x H96 x D98.5mm
JCS-33A (Approx.200g), JCR-33A (Approx. 250g), JCD-33A (Approx. 370g)
WeightSensor correction, Set value LOCK, Power failure countermeasure, Self-diagnosis, Automatic cold junction temperature
Attached functions
compensation (Only for thermocouple), Sensor burnout alarm, Input burnout, Warm-up indication, Auto/Manual control selection■ Options[Select options according to your needs. When ordering, designate the Option code to be added.]
JCS-33A has Event output which involves Alarm 2 (A2), Heater burnout alarm (W), Control output (OUT 2) and Loop break alarm (LA) output.
Alarm 2 (A2) [A2]When this option is added, 1 alarm point is added. Specifications are the same as those of Alarm1 (A1).
Watches the heater current with CT (current transformer), and detects the burnout.
Heater rated current must be selected from 5A, 10A, 20A and 50A.
Heater burnout alarm Setting accuracy : Within ±5% of heater rated current[W]
Output: Relay contact 1a 3A 250V AC (Resistive load), Electric life: 100,000 cycles
Accessories: CT (for single phase: 1 piece)
Upon returning to set limits, the alarm will stop.
If this option is applied, control output 2 is added and enables Heating/Cooling control.
Heating control action (Heating side): The same as control output (OUT1)
Cooling control action (Cooling side):
Proportional band (P): 0.0 to 10.0 times the control output (OUT1) proportional band (ON/OFF action when set to 0.0)
Integral time (I): The same as that of the control output (OUT1).
Derivative time (D): The same as that of the control output (OUT1).
Proportional cycle: 1 to 120 seconds (Not available for DC current output type)
Overlap band/Dead band : Thermocouple and RTD: -100.0 to 100.0℃(°°F)
DC current and DC voltage: -1000 to 1000 (The placement of the decimal point follows the selection.)Control output (OUT2)Hysteresis: 0.1 to 100.0℃(°°F)(Heating/Cooling control) Control output[DR, DS, DA, DT]・Relay contact (DR)
: 1a, 3A 250V AC (Resistive load), 1A 250V AC(Inductive load cosφ=0.4), Electric life:100,000 cycles
・Non-contact voltage (DS) : 12+20 V DC Max. 40mA (Short circuit protected)
・DC current (DA): 4 to 20mA DC Load resistance: Max. 550Ω・Non-contact relay (DT) : 0.3A 250V AC (Resistive load)
Cooling action mode (This must be selected by key operation from below.)・Air cooling (Linear characteristic)・Oil cooling (1.5th Power of the linear characteristic)・Water cooling (2nd Power of the linear characteristic)
Various setting status changing, reading and setting of the JC□-33A can be performed from an external computer.
By combining Shinko programmable controller (SVTC option added) with JC□-33A (C5 option added), it is possible to transmit theSV (set value) of the programmable controller digitally to the JC□-33A
Communication interface: EIA, RS-485
Communication method: Half-duplex communicationSynchronization method: Start-stop synchronization
Communication speed: (2400/4800/9600/19200bps) Selectable by keypad: (Even/ Odd/ No parity) Selectable by keypad
Serial communication Parity
Stop bit: (1 or 2) Selectable by keypad[C5]
Communication protocol: Based on Shinko standard protocol or Modbus (Selectable by keypad)(When Modbus is selected, RTU mode or ASCII mode can be selected by keypad.)Number of connectable units : A maximum of 31 units per host computer
Communication error detection : Parity check and Checksum
SV1 and SV2 can be changed by external contact. (Can be added only to the JCS-33A.)
SV1/SV2
external selection [SM] SV1: Contact open between terminals 13 and 14 SV2: Contact closed between terminals 13 and 14
This option enables Heater burnout, Sensor burnout and actuator trouble to be detected.
Loop break alarm time
: 0 to 200 minutes
Loop break alarm
Loop break alarm action span : Thermocouple and RTD: 0 to 150℃(°°F), 0.0 to 150.0℃(°°F)[LA]
DC current and DC voltage: 0 to 1500
Output: Relay contact 1a, 3A 250V AC (Resistive load), Electric life:100,000 cycles
When this option is added, 24V DC is outputted from the terminal 9 to 10 of JCD-33A and JCR-33A and can be the power source of
2-wire transmitter.
Insulated power output
Output voltage: 24±3V DC (When load current is 30mA.)[P24]
Ripple voltage: 200mV (When load current is 30mA.)
Maximum load current: 30mA
Terminal cover
Electrical shock protection cover[TC]
Be sure to use this terminal cover by adding this option if operator may touch the back of the controller while running the controller.
Case and base: Black.
Color Black [BK]

■ External dimensions ( Scale : mm )■ Panel cutout ( Scale : mm )・JCS-33A

11.5□4845＋0.5
059.7

44.5

7545＋0.50・JCS-33A

Lateral close mounting
n: Number of units mounted

45＋0.5095

・JCR-33A

n×48−3＋0.5
0130

92＋0.80

・JCR-33A

9691
106.2

92＋0.80

n×48−3＋0.50

Lateral close mounting
n: Number of units mounted

45＋0.50

・JCD-33A

98.5

130

・JCD-33A□92＋0.8011.5

48n×96−3＋0.5091
106.2

Lateral close mounting
n: Number of units mounted□92＋0.80CautionCT２250VAC
0.3A

３OUT A123A
250V
AC

NO

６OUT1 NO

１３＋ ＋A

２24V AC/DC

１２ − ７YA

POWER
SUPPLY ＋−１４NOYB

B

１２２３A2/LA（HB）

YB（＋）
NC

１５CTNO

６ 3A

１６７１７21A1NO

８ 3A

SG＋ A ＋

１８30

250V AC＋＋OUT2/HB− −24V DC
30mA● Other productsin the same series！

SAFETY
PRECAUTIONS

９NO

１０ 3A

250V AC

1040

250V ACP24

2.80.5250V AC

SV2５OUT1−3015１４４＋

φ5.8NO

１３ 3A

３８１５ − − B１０SG
RS-485 DC TC RTD

５CTL-12-S36-10L1U（for 50A）

RS-485100to240V AC

９CTL-6S（for 5A, 10A, 20A）

１１１POWERSUPPLYSV2４EVT＋100２１１YA（−）

2−φ 3.540

100 to240V AC

24V AC/DC−１GND

10.5

POWER
SUPPLY１■ CT dimentions ( Scale : mm )・JCR-33A, JCD-33A
3A
250V
AC

7.5・JCS-33A

25■ Terminal arrangement
POWER
SUPPLY＋

If lateral close mounting is used for the controller, Drip-proof/Dust-proof
IP66 may be Compromised, and all warranties will be invalidated.

98.5

311.5

□9640

φ12302−M315− B −１９DCTCB２０RTD● JCL-33A(W48 H24mm)● JCM-33A(W72 H72mm)● To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
● This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify

correct usage after consulting purpose of use with our agency or main oﬃce.
(Never use this instrument for medical purposes with which human lives are involved.)● External protection devices such as protection equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Also proper
periodic maintenance is required.
● This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in this manual.

Caution with respect toExport Trade Control Ordinance
To avoid this instrument from being used as acomponent in, or as being utilized in themanufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe ﬁnal use of this instrument.
In the case of resale, ensure that this
instrument is not illegally exported.

・This catalog is as of March 2012 and its contents are subject to change without notice.
・If you have any inquiries, please consult us or our agency.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION

Head Oﬃce : 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel: 81-72-727-6100

Fax: 81-72-727-7006

URL: http://www.shinko-technos.co.jp

E-mail: overseas@shinko-technos.co.jp
32 113 JC (E)