AVAILABLE
with Signature
Calibration

SEE PG. IN-5

General Purpose
Gage/Absolute
Pressure Transducers
Model A-5

AMPLIFIED OUTPUT AVAILABLE

1-888-282-9891

STAINLESS STEEL
0.5 to 30,000 psi, 35 mbar to 2000 bar

www.honeywell.com/sensing

Sensotec Sensors

The Model A-5 is designed as a general industrial pressure transducer with a wide variety ofavailable options to meet specific application requirements. Pressure ranges span from 0.5 to30,000 psi. All models are constructed of stainless steel and utilize complete four arm 350 ohm
strain gage bridges. Model A-5 transducers use a standard gage design. The absolute models
have an internal sealed 0 psia reference.

Dimensions
Model A-5 Gage

Model A-5 Absolute(Order Code AP141)(Order Code AP142)

Available Ranges
D”L”.5; 1; 2; 5 psig
2.25 (5.72 cm) 2.42 (6.15 cm)
10; 15; 25; 50; 75; 100 psig 1.5 (3.81 cm) 2.35 (5.97 cm)
200; 300; 500; 750;
1000; 1500 psig
1.5 (3.81 cm) 2.35 (5.97 cm)
2000; 3000; 5000; 7500;
10,000 psig
1.5 (3.81 cm) 2.35 (5.97 cm)
15,000; 20,000; 30,000 psig 1.5 (3.81 cm) 2.21 (5.61 cm)

Available Ranges
D”
1; 2; 5 psia
2.25 (5.72 cm)
10; 15; 25; 50; 75; 100;
150; 200; 300; 500; 750;
1000; 1500 psia
1.5 (3.81 cm
2000; 3000; 5000;
7500; 10,000 psia
1.5 (3.81 cm)
15,000; 20,000; 30,000 psia 1.5 (3.81 cm)

0.5 to 1500 psi
35 mbar to 100 bar

Options (See Appendix)

PRESSURE PORT
1/4–18 NPT MALE

2000 to 30,000 psi
140 bar to 2000 bar

L”
2.09 (5.31 cm)
2.35 (5.97 cm)
2.39 (6.07 cm)
2.21 (5.61 cm)

2,000 TO 10,000 PSI
PRESSURE PORT
1/4–18 FEMALE NPT
15,000 TO 30,000 PSI
PRESSURE PORTAUTOCLAVE F-250C

Temperature compensation: 1b, 1c, 1d, 1e, 1f; Internal amplifiers: 2a, 2n or 2N intrinsically safe
amp - see page AP-6, 2j, 2k, 2t;
Amp enhancements 3d; Pressure ports: 5a, 5b, 5c, 5d (<15,000 psi); Electrical terminations 6e,6f, 6g, 6h, 6l; Int. shunt cal 8a; Special calibration 9a (>5 psi gage only); Signature calibration
53e
Premium Options: 1g (>15 psi only); 1i (>15 psi only); 2c, 2q, 3a, 3c, 6b, 6c, 6j; 9b (Gage only,>5 psi); 10a, 12a, 12b

Accessories: Mating connectors and connector/cable assemblies; Pressure port adapters.
PR-10

1-888-282-9891
Model A-5
Gage:
Absolute:
Pressure Ranges..........................
Accuracy (min) .............................
Non-Linearity ................................
Hysteresis.....................................
Non-Repeatability .........................
Output (standard) .........................
Resolution ....................................

ENVIRONMENTAL

Temperature, Operating
1 to 1,000psi ...........................
1,500 to 30000psi ...................
Temperature, Compensated ........
Temperature Effect– Zero (max).............................
– Span (max .............................

ELECTRICALMECHANICALINTERNALLY AMPLIFIEDUNITS (Optional)

Mating Connector (not incl.) .........
Media............................................
Overload-Safe
Overload-Burst
0.5 to 5000 psi .........................
7500 to 10,000 psi ...................
15,000 to 30,000 psi ................
Pressure Port
1 to 1500 psi ............................
2000 to 10,000 psi ...................
15,000 to 10,000 psi ................
Dead Volume................................
0.5 to 5 psi ...............................
10 to 15 psi ..............................
25 to 1500 psi ..........................
2000 to 30,000 psi ...................
Wetted Parts Material...................
Type .............................................
Weight ..........................................
Case Material ...............................

-100° to 325° F (-73° to 162° C)-100° to 250° F (-73° to 95°C)
60° to 160° F (16° to 71° C)
0.0075% F.S./ ° F (0.0135% FS./ ° C)
0.01% Rdg./ ° F (0.018% Rdg./ ° C)
Bonded foil
10VDC
Up to 12VDC or AC
5000 megohm @ 50VDC
350 ohms
Included#2 (See Pg. AP-8)
PTIH-10-6P or equiv.
(Hermetic stainless)
PT06A-10-6S or equiv.
Gas Liquid
50% over capacity
300% over capacity
200% over capacity
70% over capacity
1/4-18 NPT male
1/4-18 NPT female
Autoclave AE F250-C
0.32 cu. in.
0.17 cu. in.
0.17 cu. in.
0.12 cu. in.
17-4 PH Stainless /15-5 Stainless
Absolute or Gage * * *10 oz. (50 psi)
Stainless Steel

Outputs Available .........................

0-5VDC, 0-10VDC, 4-20mA

Additional Length..........................

1.12" (2.84 cm)

NOTES

GENERAL PURPOSE

Strain Gage Type .........................
Excitation (calibration) ..................
Excitation (acceptable) .................
Insulation Resistance ...................
Bridge Resistance ........................
Shunt Calibration Data .................
Wiring Code (std) .........................
Electrical Termination (std) ..........

0.5 to 30,000 psi (35 mbar to 2000 bar)*+ /-0.5% F.S.
+ /-0.25% F.S.
+ /-0.13% F.S.
+ /-0.07% F.S.
3mV/V* *Infinite

PRESSUREPERFORMANCE

Order Code AP141
Order Code AP142

4-20mA Intrinsically safe
See page AP-62.0" (5.08 cm)* 0.5 psi (35 mbar) is available in gage only.
** Output for 0.5, 1, 2 psi units is 1-2mV/V*** Gage pressure units greater than 200 psi (7 bar) are sealed.

General Information
How to order (See Pg. AP-19)
Gage/Absolute pressure selection flow chart (See Pg. PR-1)●お問い合わせは下記へ ̶PR-11