EA01285D/06/A2/01.20
71478140
2020-07-31

Products

Solutions

Installation Instructions
Replacing the Sensor
t-mass I, F 500-digital

EN Installation Instructions: EA01285D →  3
DE Einbauanleitung: EA01285D →  13

Services

EA01285D

Replacement of the sensor
t-mass I, F 500-digital
Table of contents
1

Overview of replacement sensors . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

3Personnel authorized to carry out repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

5

Safety symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

6

Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

7Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

8t-mass I, F 500-digital . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

9Cust-potted sensor connection housing, stainless steel cast housing IP68 . . . . . . . . . . . 11

10

Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11

Endress+Hauser

3

EA01285D

1

Overview of replacement sensors

The Installation Instructions apply to the following replacement sensors:
Order structure

Device component

6F5B**-**XXXXXXXAXA******A2

1 × sensor, aluminum coatedfor t-mass F 500-digital complete, including sensor
nameplate, 1 × Installation Instructions

6F5B**-**XXXXXXXAXL******A2

1 × sensor, cast, stainless steel
for t-mass F 500-digital complete, including sensor
nameplate, 1 × Installation Instructions

6F5B**-**XXXXXXXAXL******A2+*CC

1 × sensor, cast, stainless steel, IP68, cust-potted for tmass F 500-digital complete,
including sensor nameplate,
1 × potting compound, 2 × Installation Instructions

6I5B**-**XXXXXXXAXA******A2

1 × sensor, aluminum coatedfor t-mass I 500-digital complete, including sensor
nameplate,
1 × Installation Instructions

6I5B**-**XXXXXXXAXL******A2

1 × sensor, cast, stainless steel
for t-mass I 500-digital complete, including sensor
nameplate,
1 × Installation Instructions

6I5B**-**XXXXXXXAXL******A2+*CC

1 × sensor, cast, stainless steel, IP68, cust-potted for tmass I 500-digital complete, including sensor
nameplate,
1 × potting compound, 2 × Installation Instructions

4

Endress+Hauser

EA01285D

2

Designated use

The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type. Only original parts from Endress+Hauser may be used. Only
spare parts kits designed by Endress+Hauser for the measuring device can be used at any
time.
Inspection is performed using the W@M Device Viewer. The procedure involved is describedas follows.
A spare parts overview is located in the interior of some measuring devices. If the spareparts kit is specified on this list, no inspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

Order code

4.

=

Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for productdata and click Spare Parts.

3.

All spare parts for the measuring device aredisplayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order Thenumber of the spare parts kit corresponds to anorder number on the spare parts list.

3

Personnel authorized to carry out repairs

Authorization to carry out repairs depends on the measuring device's approval type. The tablebelow shows the authorized group of people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out
safely and to the required quality standard. He/she must also guarantee the safety of thedevice following repair.

1)

Measuring device approval

Personnel authorized to perform repairs 1)

Without approval

2, 3With approval (e.g. IECEx)

2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/alteration must be performed under supervision.

Endress+Hauser

5

EA01285D

4

Safety instructions• Check whether the spare part matches the labeling on the measuring device as described on
the cover page.
• The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type.
Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.
• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:• Specialized technical staff must be trained in instrument safety.
• They must be familiar with the individual operating conditions of the devices.
• In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is de-energized.
• For measuring devices intended for use in hazardous locations, please observe theguidelines in the Ex documentation (XA).
• In the case of measuring devices in safety-related applications in accordance withIEC 61508 or IEC 61511: Commission in accordance with Operating Instructions after
repair. Document the repair procedure.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Hot surfaces! Risk of injury! Before commencing work: allow the system and measuring
device to cool down to a touchable temperature.
• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removing the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn the measuring device off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open the housing for a brief period. Avoid the penetration of foreign bodies, moisture
or contaminants.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.

6

Endress+Hauser

EA01285D• If spacing is reduced or the dielectric strength of the measuring device cannot be
guaranteed during repair work, perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service plug:• Do not connect in explosive atmospheres.
• Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your Endress+Hauser service organization.

5

Safety symbols
DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

6

Symbols used

6.1

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

Series of steps

Endress+Hauser

7

EA01285D

7

Tools list

3 mm

0.5 x3.5 mm

T10, T20

8t-mass I, F 500-digital

8.1

Replacement of the sensor

8.1.1

Removing the sensor connection housing

LCAUTION
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is deenergized.
1→

2→

2.

1.

3.
3 mm

T20
4.
24 mm

6.

5.

0.4 × 2.5

‣ Sensor connection housing: aluminum coated8

Endress+Hauser

EA01285D

32.
circuits

are

1.

alive

3.
3 mm

T20
4.
24 mm

6.

5.

0.4 × 2.5

‣ Sensor connection housing: stainless steel cast housing

8.1.2

Mounting the replacement sensor

Reassembly is carried out in reverse order. Note the following:
Determining the insertion depth for t-mass I 300, 500-digital

9
9

230

230

9

220

220

210

210

200
190

8

180

8

200
190

200

7

180
7

190
180
7

A

A

B

B

C1 C2

230
220

210
8

C1 C2

8.1.3

A = Internal pipe diameter for circular pipes
Channel height for rectangular ducts if the sensor is to be installed vertically,or the channel width if the sensor is to be installed horizontally.
(a = min. 80 mm or 3 in)
B = Thickness of pipe wall or channel wall
C1 = Height of the weld-in nipple on the pipe or channel
C2 = Sensor pipe union and valve unit

Determining the insertion depth before mounting for the first time:
Insertion depth = (0.3 × a) + b + c

Endress+Hauser

9EA01285D

8.1.4

Insert the measuring device to the calculated insertion depth and align it

NOTICE
Risk of damage to the sensor tip:
‣ Do not insert the sensor too far.
1→

9

230
220
210

8

200
190
180

7

‣ Set the scale to the calculated insertion depth.
2→

3

120
1¼
36 mm
90° (±3°)‣ Check and ensure that the sensor on the pipe is aligned at a 90° angle tothe direction of flow.
‣ The arrow engraved on the sensor shaft must match the flow direction.
‣ Align the scale to the pipe axis.

‣ Tighten the coupling nut by hand.
‣ Mounting for the first time:
Tighten coupling nut with 1 ¼
turns.
‣ Repeat mounting: Tighten
coupling nut with 1 turn.
‣ NOTE!  If strong vibrations canbe expected, tighten the coupling
nut with 1½ turns when
mounting for the first time.

For detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

10

Endress+Hauser

EA01285D

9

Cust-potted sensor connection housing, stainless steel
cast housing IP68

For detailed information on the potting for the sensor connection housing, stainless steel casthousing IP68, see EA01025D

10

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize the disposalof WEEE as unsorted municipal waste. Do not dispose of products bearing this marking as
unsorted municipal waste. Instead, return them to Endress+Hauser for disposal under theapplicable conditions.

Endress+Hauser

11

EA01285D

Austausch Messaufnehmer
t-mass I, F 500-digital
Inhaltsverzeichnis
1Übersicht Ersatzmessaufnehmer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

2

Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

3

Reparaturberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

5Warnhinweissymbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17

6

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17

7

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

8t-mass I, F 500-digital . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

9

Feldverguss Sensor Anschlussgehäuse Guss rostfrei IP68 . . . . . . . . . . . . . . . . . . . . . . . . . . 21

10

Entsorgung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21

Endress+Hauser

13

EA01285D

1Übersicht Ersatzmessaufnehmer

Die Einbauanleitung ist für folgende Ersatzmessaufnehmer gültig:
Bestellstruktur

Gerätekomponente

6F5B**-**XXXXXXXAXA******A2

1 × Messaufnehmer Alu beschichtet
für t-mass F 500-digital komplett, inklusive
Messaufnehmer Typenschild, 1 × Einbauanleitung

6F5B**-**XXXXXXXAXL******A2

1 × Messaufnehmer Guss rostfrei
für t-mass F 500-digital komplett, inklusive
Messaufnehmer Typenschild, 1 × Einbauanleitung

6F5B**-**XXXXXXXAXL******A2+*CC

1 × Messaufnehmer Guss rostfrei, IP68, Feldverguss
für t-mass F 500-digital komplett,inklusive Messaufnehmer Typenschild,
1 × Vergussmasse, 2 × Einbauanleitung

6I5B**-**XXXXXXXAXA******A2

1 × Messaufnehmer Alu beschichtet
für t-mass I 500-digital komplett, inklusive
Messaufnehmer Typenschild,
1 × Einbauanleitung

6I5B**-**XXXXXXXAXL******A2

1 × Messaufnehmer Guss rostfrei
für t-mass I 500-digital komplett, inklusive
Messaufnehmer Typenschild,
1 × Einbauanleitung

6I5B**-**XXXXXXXAXL******A2+*CC

1 × Messaufnehmer Guss rostfrei, IP68, Feldverguss
für t-mass I 500-digital komplett, inklusive
Messaufnehmer Typenschild,
1 × Vergussmasse, 2 × Einbauanleitung

14

Endress+Hauser

EA01285D

2

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

3

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

Reparaturberechtigte Personen

Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die
Sicherheit während der Arbeiten, die Qualität der Ausführung und die Sicherheit desGeräts nach der Reparatur.

1)

Zulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Ohne Zulassung

2, 3

Mit Zulassung (z.B. IECEx)

2, 3

Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Endress+Hauser

15

EA01285D

4

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:• In Gerätesicherheit ausgebildet.
• Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
• Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Messgeräten für den explosionsgefährdeten Bereich: Hinweise in der ExDokumentation (XA) beachten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.

16

Endress+Hauser

EA01285D• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:• Nicht in explosionsfähiger Atmosphäre anschließen.
• Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

5

Warnhinweissymbole
GEFAHR

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen wird.
WARNUNG

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen kann.
VORSICHT

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu leichter oder mittelschwerer Körperverletzung führen kann.
HINWEIS

Dieser Hinweis enthält Informationen zu Vorgehensweisen und weiterführenden
Sachverhalten, die keine Körperverletzung nach sich ziehen.

6Verwendete Symbole

6.1

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

Handlungsschritte

Endress+Hauser

17

EA01285D

7

Werkzeugliste

3 mm

0,5 x 3,5 mm

T10, T20

8t-mass I, F 500-digital

8.1

Austausch Messaufnehmer

8.1.1

Ausbau Sensor Anschlussgehäuse

LVORSICHT
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
1→

2→

2.

1.

3.
3 mm

T20
4.
24 mm

6.

5.

0.4 × 2.5

‣ Sensor Anschlussgehäuse: Alu beschichtet

18

Endress+Hauser

EA01285D

32.
circuits

are

alive

1.

3.
3 mm

T20
4.
24 mm

6.

5.

0.4 × 2.5

‣ Sensor Anschlussgehäuse: Guss rostfrei

8.1.2

Ersatzmessaufnehmer montieren

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
Bestimmung der Einstecktiefe für t-mass I 300, 500-digital

9
9

230

230

9

220

220

210

210

200
190

8

180

8

200
190

200

7

180
7

190
180
7

A

A

B

B

C1 C2

230
220

210
8

C1 C2

8.1.3

A = Rohrinnendurchmesser bei einem runden Rohr
Kanalhöhe bei einem rechteckigen Kanal, wenn der Messaufnehmer senkrecht eingebaut werden solloder die Kanalbreite, wenn er waagerecht eingebaut werden soll.
(a = min. 80 mm oder 3 in)
B = Rohrwandstärke oder Kanalwandstärke
C1 = Höhe des Einschweißstutzens am Rohr oder Kanal
C2 = Messaufnehmer-Rohrverschraubung und Ventileinheit

Bestimmung der Einstecktiefe vor Erstmontage:
Einstecktiefe = (0,3 × a) + b + c

Endress+Hauser

19

EA01285D

8.1.4

Messgerät auf berechnete Einstecktiefe einführen und ausrichten

HINWEIS
Gefahr von Schäden an der Sensorspitze:‣ Messaufnehmer nicht zu tief einführen.
1→

9

230
220
210

8

200
190
180

7

‣ Messskala auf die berechnete Einstecktiefe einstellen.
2→

3

120
1¼
36 mm
90° (±3°)‣ Prüfen und sicherstellen, dass der Messaufnehmer am Rohr 90° zur

Durchflussrichtung ausgerichtet ist.
‣ Der eingravierte Pfeil auf dem Messaufnehmerschaft muss mit derDurchflussrichtung übereinstimmen.
‣ Skala zur Rohrachse ausrichten.

‣ Überwurfmutter von Hand
anziehen.

‣ Erstmontage: Überwurfmutter

mit 1 ¼ Umdrehungen anziehen.

‣ Wiederholmontage:

Überwurfmutter mit
1 Umdrehung anziehen.
‣ HINWEIS!  Wenn mit starken
Vibrationen zu rechnen ist, dannbei der Erstmontage dieÜberwurfmutter mit
1½ Umdrehungen anziehen.

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.
20

Endress+Hauser

EA01285D

9

Feldverguss Sensor Anschlussgehäuse Guss rostfrei
IP68

Detaillierte Angaben zum Verguss Sensor Anschlussgehäuse Guss rostfrei IP68: EA01025D

10

Entsorgung

Gemäß der Richtlinie 2012/19/EG über Elektro- und Elektronik-Altgeräte (WEEE) ist das
Produkt mit dem abgebildeten Symbol gekennzeichnet, um die Entsorgung von WEEE als
unsortierten Hausmüll zu minimieren. Gekennzeichnete Produkte nicht als unsortierter
Hausmüll entsorgen, sondern zu den gültigen Bedingungen an Endress+Hauser
zurückgeben.

Endress+Hauser21

*71478140*
71478140

www.addresses.endress.com