Instructions for Use

Color LCD monitor

ImportantPlease read this “Instructions for Use”, and “Installation Manual”(separate volume) carefully to familiarize yourself with safe andeffective usage.
Please retain this manual for future reference.
• For monitor adjustment and settings, refer to the “Installation
Manual”.
• For the latest product information including the “Instructions forUse”, refer to our web site :http://www.eizoglobal.com

SAFETY SYMBOLSThis manual and this product use the safety symbols below. They denote critical information. Please readthem carefully.

WARNINGCAUTION

Failure to abide by the information in a
WARNING may result in serious injury
and can be life threatening.

Failure to abide by the information in a
CAUTION may result in moderate injury and /or property or product damage.

Indicates a warning or caution. For example,

indicates an “electrical shock” hazard.

Indicates a prohibited action. For example,

means “Do not disassemble”.

Indicates a mandatory action. For example,

means “Ground the unit”.

This product has been adjusted specifically for use in the region to which it was originally shipped. If
operated outside this region, the product may not perform as stated in the specifications.
No part of this manual may be reproduced, stored in a retrieval system, or transmitted, in any form or by
any means, electronic, mechanical, or otherwise, without the prior written permission of EIZO Corporation.
EIZO Corporation is under no obligation to hold any submitted material or information confidential unlessprior arrangements are made pursuant to EIZO Corporation’s receipt of said information. Although every
effort has been made to ensure that this manual provides up-to-date information, please note that EIZO
monitor specifications are subject to change without notice.

2

﻿PRECAUTIONS● IMPORTANT• This product has been adjusted specifically for use in the region to which it was originally shipped.
If the product is used outside the region, it may not operate as specified in the specifications.
• To personal safety and proper maintenance, please read carefully this section and the caution
statements on the monitor.

Location of the Caution Statements
WARNINGAVERTISSEMENTWARNUNG

Symbols on the unit
SymbolThis symbol indicates
Main Power Switch:

Press to turn the monitor’s main power off.

Main Power Switch:

Press to turn the monitor’s main power on.

Power button:

Press to turn the monitor’s power on or off.

Alternating current
Alerting electrical hazard
CAUTION:

Refer to “SAFETY SYMBOLS” (page 2).

WEEE marking:Product must be disposed of separately; materials may
be recycled.

CE marking:EU conformity mark in accordance with the provisions ofCouncil Directive and/or Regulation (EU).

Manufacturer
Date of manufacture
Medical device in EU
Importer in EU
Mark signifying compliance with UK regulations
UK Responsible Person
Authorised representative in Switzerland

PRECAUTIONS

3

WARNING
If the unit begins to emit smoke, smells like something is burning, or makes strange noises,
disconnect all power connections immediately and contact your EIZO representative for advice.
Attempting to use a malfunctioning unit may result in fire, electric shock, or equipment damage.
Do not open the cabinet or modify the unit.
Opening the cabinet or modifying the unit may result in fire, electric shock, or burn.
Refer all servicing to qualified service personnel.
Do not attempt to service this product yourself as opening or removing covers may result in fire, electric
shock, or equipment damage.
Keep small objects or liquids away from the unit.
Small objects accidentally falling through the ventilation slots into the cabinet or spillage
into the cabinet may result in fire, electric shock, or equipment damage. If an object or
liquid falls / spills into the cabinet, unplug the unit immediately. Have the unit checked bya qualified service engineer before using it again.
Place the unit at a sturdy and stable place.
A unit placed on an inadequate surface may fall and result in injury or equipment damage. If the unit falls,
disconnect the power immediately and ask your local EIZO representative for advice. Do not continue using
a damaged unit. Using a damaged unit may result in fire or electric shock.
Use the unit in an appropriate location.
Otherwise, fire, electric shock, or equipment damage may result.
• Do not place outdoors.
• Do not place in any form of transportation (ships, aircraft, trains, automobiles, etc.).
• Do not place in dusty or humid environments.
• Do not place in locations where water may be splashed on the screen (bathrooms, kitchens,etc.)• Do not place in locations where steam comes in direct contact with the screen.
• Do not place near heat generating devices or humidifiers.
• Do not place in locations where the product is subject to direct sunlight.
• Do not place in environments with inflammable gas.
• Do not place in environments with corrosive gases (such as sulfur dioxide, hydrogen sulfide,
nitrogen dioxide, chlorine, ammonia, and ozone)• Do not place in environments with dust, components that accelerate corrosion in theatmosphere (such as sodium chloride and sulfur), conductive metals, and so on

To avoid danger of suffocation, keep the plastic packing bags away from babies and children.
Use the enclosed power cord and connect to the standard power outlet in your country.
Be sure to use within the rated voltage of the power cord. Otherwise, fire or electric shock may result.
Power supply: 100 - 240 Vac, 50 / 60 Hz
To disconnect the power cord, grasp the plug firmly and pull.
Tugging on the cord may damage and result in fire or electric shock.

The equipment must be connected to a grounded main outlet.
Failure to do so may result in fire or electric shock.
Use the correct voltage.
• The unit is designed for use with a specific voltage only. Connection to another voltage than specified in this“Instructions for Use” may cause fire, electric shock, or equipment damage.
Power supply: 100 - 240 Vac, 50 / 60 Hz• Do not overload your power circuit, as this may result in fire or electric shock.

4

PRECAUTIONSWARNING
Handle the power cord with care.
• Do not place the cord underneath the unit or other heavy objects.
• Do not pull on or tie the cord.

If the power cord becomes damaged, stop using it. Using of a damaged cord may resultin fire or electric shock.
For electrical safety, do not connect or disconnect the power cord in the presence of patients.
Never touch the plug and power cord if it begins to thunder.
Touching them may result in electric shock.
When attaching an arm stand, please refer to the user’s manual of the arm stand and install the unit
securely.
Otherwise, the unit may become detached, resulting in injury and / or equipment damage. Before installation,
make sure that desks, walls, or any other installation surface has adequate mechanical strength. If the unit
falls, please ask your local EIZO representative for advice. Do not continue using a damaged unit. Usinga damaged unit may result in fire or electric shock. When reattaching the tilt stand, please use the samescrews and tighten them securely.
Do not touch a damaged LCD panel directly with bare hands.
Liquid crystal is poisonous. If any part of your skin comes in direct contact with thepanel, wash thoroughly. If liquid crystal enters your eyes or mouth, immediately flushwith large amounts of water and seek medical attention.
Fluorescent backlight lamps contain mercury (the products that have LED
backlight lamps contain no mercury), dispose according to local, state or federal
laws.
Exposure to elemental mercury can result in effects on the nervous system, including
tremors, memory loss, and headaches.

PRECAUTIONS

5CAUTION
Handle with care when carrying the unit.
Disconnect the power cord and cables when moving the unit. Moving the unit with the power cord or cables
attached is dangerous and may result in injury.
Carry or place the unit according to the correct specified methods.
• When carrying, grasp and firmly hold the unit as shown in the illustration below.
• Monitors of size 30 inches and above are heavy. When unpacking and / or carrying the monitor, ensure at least two
people are utilized.

Dropping the unit may result in injury or equipment damage.

Do not block the ventilation slots on the cabinet.
• Do not place any objects on the ventilation slots.
• Do not install the unit in a place with poor ventilation or inadequate space.
• Do not use the unit laid down or upside down.

Blocking the ventilation slots prevents proper airflow and may result in fire, electric
shock, or equipment damage.
Do not touch the plug with wet hands.
Doing so may result in electrical shock.

Use an easily accessible power outlet.
This is to facilitate disconnecting the power in case of a problem.
Periodically clean the area around the power plug and the ventilation slot of the monitor.
Dust, water, or oil on the plug may result in fire.
Unplug the unit before cleaning it.
Cleaning the unit while it is plugged into a power outlet may result in electric shock.
If you plan to leave the unit unused for an extended period of time, disconnect the power cord fromthe wall socket after turning off the power switch for the safety and the power conservation.
This product is suitable for placing near patients, but not for direct contact with patients.
For users in the territory of the EEA and Switzerland:
Any serious incident that has occurred in relation to the device should be reported to the
Manufacturer and the Competent Authority of the Member State in which the user and/or patient isestablished.

6

PRECAUTIONS

Notice for this monitor
Intended use
This product is intended to be used in displaying and viewing digital images for review, analysis anddiagnosis by trained medical practitioners. It does not support the display of mammography images fordiagnosis.
Caution: Federal law (USA) restricts this device to sale by or on the order of a physician.
Attention• This product should be set to horizontal view mode when used for the purposes mentioned above.
• This product may not be covered by warranty for uses other than those described in this manual.
• The specifications noted in this manual are only applicable when the following are used:- Power cords provided with the product- Signal cables specified by us• Only use optional products manufactured or specified by us with this product.

Precautions for Use• Parts (such as the LCD panel) may deteriorate over the long term. Periodically check that they areoperating normally.
• When the screen image is changed after displaying the same image for extended periods of time, anafterimage may appear. Use the screen saver or power save function to avoid displaying the same image
for extended periods of time.
• If the monitor displays continuously over a long period of time, dark smudges or burn-in may appear. To
maximize the life of the monitor, we recommend the monitor be turned off periodically.
• An afterimage may appear even after a short time period has elapsed depending on the displayed
image. If this occurs, changing the image or leaving the power off for a few hours may solve the problem.
• The backlight of the LCD panel has a fixed lifetime. When the screen becomes dark or begins to flicker,
please contact your local EIZO representative.
• The screen may have defective pixels or a small number of light dots on the screen. This is due to thecharacteristics of the panel itself, and is not a malfunction of the product.
• Do not press on the panel or edge of the frame strongly, as this may result in display malfunctions, suchas interference patterns, etc. If pressure is continually applied to the panel, it may deteriorate or damage
your panel. (If the pressure marks remain on the panel, leave the monitor with a black or white screen.
The symptom may disappear.)• Do not scratch or press on the panel with any sharp objects, as this may result in damage to the panel.
Do not attempt to brush with tissues as this may scratch the panel.
• Do not touch the built-in calibration sensor (Integrated Front Sensor). Doing so may reduce themeasurement accuracy or result in equipment damage.
• Depending on the environment, the value measured by the built-in illuminance sensor may differ fromthe value shown on a stand-alone illuminometer.
• When the monitor is cold and brought into a room or the room temperature goes up quickly, dew
condensation may occur on the interior and exterior surfaces of the monitor. In that case, do not turn themonitor on. Instead wait until the dew condensation disappears, otherwise it may cause some damage
to the monitor.

Notice for this monitor

7

To Use the Monitor for a Long Time● Maintenance• The display quality of monitors is affected by the quality level of input signals and the degradation of
the product. Perform visual checks and periodic constancy tests to comply with medical standards /guidelines according to your application, and carry out calibration as necessary. Use of the RadiCS
monitor quality control software enables you to perform high-level quality control that meets medical
standards / guidelines.
• It takes about 15 minutes for the performance of electrical parts to stabilize. Please wait 15 minutes
or more after the monitor power has been turned on or the monitor has recovered from the powersaving mode, and then adjust the monitor.
• We recommend that monitors be set to the recommended level or lower to reduce changes inluminosity caused by long-term use and maintain stable brightness.
• Perform correlation using RadiCS / RadiCS LE to adjust the measurement result of the Integrated
Front Sensor to the measurement result of the UX1 sensor for reference. Periodical correlation
enables the measurement accuracy of the Integrated Front Sensor equivalent to that of the UX1
sensor. For details, refer to RadiCS / RadiCS LE User’s Manual.

● Cleaning

Periodic cleaning is recommended to keep the monitor looking new and to prolong its operation
lifetime.
Gently wipe off any dirt on the cabinet or panel surface with a soft cloth soaked in a small amount ofwater or one of the chemicals listed below.

Chemicals that may be used for cleaning
Material name

Ethanol
Isopropyl alcohol
Chlorhexidine
Benzalkonium chloride
Alkyldiaminoethylglycine
GlutaralGlutaralEthanol
Isopropyl alcohol
Hibitane
WelpasTego 51
Sterihyde
Cidex Plus28

Product name

Attention• Do not use chemicals on a frequent basis. Chemicals such as alcohol and antiseptic solution may cause
gloss variation, tarnishing, and fading of the cabinet or panel, and also quality deterioration of the image.
• Never use any thinner, benzene, wax, and abrasive cleaner, which may damage the cabinet or panel.
• Do not let chemicals come into direct contact with the monitor.
Note• The optional ScreenCleaner is recommended for cleaning the cabinet and panel surface.

To use the monitor comfortably• Staring at the monitor for a long time tires your eyes. Take a 10-minute rest every hour.
• Look at the screen from a proper distance and from a proper angle.

8

Notice for this monitor

CONTENTS
PRECAUTIONS....................................................... 3● IMPORTANT.................................................... 3

Notice for this monitor.......................................... 7
Intended use.............................................................. 7
Precautions for Use.................................................. 7
To Use the Monitor for a Long Time........................ 8● Maintenance..................................................... 8● Cleaning........................................................... 8

To use the monitor comfortably.............................. 8

CONTENTS.............................................................. 9
Chapter 1

Introduction...................................... 10

1-1.

Features.........................................................10

1-2.

Package Contents.........................................12

1-3.

Controls and Functions...............................14● EIZO LCD Utility Disk......................................12

Chapter 2
2-1.

●

Installation / Connection................. 15

Before Installing the Product......................15
Installation Requirements...............................15

2-2.

Connecting Cables.......................................16

2-3.

Turning On the Power...................................192-4.

Adjusting the Screen Height and Angle.....19

Chapter 3

No-Picture Problem......................... 20

Chapter 4

Specifications.................................. 21

4-1.

Specifications List........................................21

4-2.

Compatible Resolutions............................. 22

4-3.

Optional Accessories.................................. 22

Appendix............................................................... 23
Medical Standard.................................................... 23
FCC Declaration of Conformity............................. 24
Declaration of compliance with India RoHS........ 25
EMC Information..................................................... 25

CONTENTS

9

Chapter 1 Introduction
Thank you very much for choosing an EIZO color LCD monitor.

1-1. Features● High degree of freedom of layout
This product has the PbyP (Picture by Picture) and PinP (Picture in Picture) functions that can display
up to three signals at the same time.

SinglePbyP

PinP

PbyP + PinP

● Simple wiring
In addition to a DisplayPort input terminal, an output terminal is also provided.
• From the output terminal (), a signal can be output to a different monitor.

• In order to display images in PbyP, two signal cables usually need to be connected between the
PC and the monitor. On this model, you can connect it to the PC using one cable. You only needto connect the supplied DisplayPort cable (PP028) between the output terminal (terminal ().

10Chapter 1 Introduction) and the input

● Space-saving design
The monitor has two USB upstream ports. You can operate two PCs using one set of USB devices(mouse, keyboard, etc.) by switching between the PCs.

● Monitor operation from the mouse and keyboard
Using the RadiCS / RadiCS LE monitor quality control software, you can perform the following
monitor operations using the mouse and keyboard:• Switching CAL Switch modes• Switching input signals• Displaying or hiding the PinP sub window (Hide-and-Seek)• Switching PCs used to operate USB devices (Switch-and-Go)
Note• The RadiCS / RadiCS LE software allows you to display or hide the PinP sub window and switch the PC
used to operate USB devices at the same time. For more information about the setup procedure, refer to
RadiCS / RadiCS LE User's Manual.

● Quality control• This monitor has a built-in calibration sensor (Integrated Front Sensor). This sensor enables themonitor to perform calibration (SelfCalibration) and Grayscale Check independently.

• Using RadiCS LE that is attached to the monitor, you can manage history related to the monitor,
and the SelfCalibration target and execution schedule.
• The RadiCS monitor quality control software enables you to perform quality control that meets
medical standards / guidelines.

Chapter 1 Introduction111-2. Package Contents
Check that all of the following items are contained in the package. If any of these are missing or
damaged, contact your dealer or local EIZO representative listed on the attached sheet.
Note• It is recommended that the box and packing materials be stored so that they can be used to move or transport
this product.
• Monitor• Power cord

• Digital signal cable: DD300DL x 1
DVI - DVI (dual link)• Digital signal cable: PP300 x 2
DisplayPort - DisplayPort• USB cable: UU300 x 2• Digital signal cable: PP028 x 1
DisplayPort - DisplayPort• EIZO LCD Utility Disk (CD-ROM)• Instructions for Use● EIZO LCD Utility Disk

The CD-ROM contains the following items. Refer to "Readme.txt" on the disk for software startup
procedures or file reference procedures.
• Readme.txt file
• RadiCS LE monitor quality control software (for Windows)• User’s Manual
Monitor Installation Manual
RadiCS LE User's Manual• Outline dimensions

RadiCS LE
RadiCS LE enables you to perform the following quality control and monitor operations. For more
information about the software or setup procedures, refer to RadiCS LE User's Manual.

Quality control• Executing calibration• Displaying test results in a list and creating a test report• Setting the SelfCalibration target and execution schedule

Monitor operations• Switching CAL Switch modes• Switching input signals• Displaying or hiding the PinP sub window (Hide-and-Seek)• Switching PCs used to operate USB devices (Switch-and-Go)• Entering power saving mode (Backlight Saver)
Attention• The specifications of RadiCS LE are subject to change without notice. The latest version of RadiCS LE isavailable for download from our web site: http://www.eizoglobal.com

12

Chapter 1 Introduction

To use RadiCS LE
For information on how to install and use RadiCS LE, refer to RadiCS LE User's Manual (on the
CD-ROM).
When using RadiCS LE, connect the monitor to your PC using the supplied USB cable. For more
information about how to connect the monitor, see “2-2. Connecting Cables” (page 16).
Note

• In an environment where it is difficult to connect the USB cable, enabling DDC communication allows you to
RadiCS LE without having to use the USB cable. For information on how to configure DDC communication,refer to the Installation Manual (on the CD-ROM). In view of the response time and operation stability, USB
connection is recommended.

Chapter 1 Introduction13

1-3. Controls and Functions
1

12

2

13

Approx. 30°

14
15
16
*1

Setting menu

3

5

4

6

7

8

9

10

11

1. Integrated Front Sensor This sensor is used to perform calibration and Grayscale Check.
2. Ambient Light Sensor
This sensor measures environmental illumination. Environmental illuminancemeasurement is performed using the RadiCS / RadiCS LE quality control software.
3. Presence Sensor
This sensor detects movements made by a person in front of the monitor.
4. Operation switches
Displays the operation guide. Set menus according to the operation guide.
5.

switch

6. Main power switch
7. Power connector
8. DVI-D connector
9. DisplayPort input
connector
10. DisplayPort output
connector

Turns the power on or off.
The switch indicator is lit when you turn the power on. The indicator color differs
depending on the monitor's operation status.
Green: Monitor in operation, Orange: Power saving mode,
Off: Main power / power off
Turns the main power on or off.
: Off, : On
Connects the power cord.
Connect it to the PC. The connector to be used differs depending on the screen
layout. For more information, see “2-2. Connecting Cables” (page 16).
To set up a daisy-chain connection, connect the cable to the input port of anothermonitor.
When this connector is connected to, the signal input tocan be displayed in
PbyP mode.

11. USB upstream port

12. Handle

For more information, see “2-2. Connecting Cables” (page 16).
Connect this port to the PC when you use software that needs a USB connection orconnect a USB device (peripheral device that supports USB) to the USB downstream
port.
This handle is used for transportation.
Attention• When carrying the monitor, hold the lower part of the monitor and this handle tightly
so that you do not drop it.
13. Stand
14. USB downstream port

The height and angle can be adjusted.
Connect it to a USB device. To set up a daisy-chain connection, connect the cable to
the USB upstream port of another monitor.
15. Security lock slot
Complies with Kensington’s MicroSaver security system.
16. Cable holder
Holds the monitor cables.
*1 For information on how to use, refer to the Installation Manual (on the CD-ROM).

14

Chapter 1 Introduction

Chapter 2 Installation / Connection
2-1. Before Installing the Product
Carefully read “PRECAUTIONS” (page 3) and always follow the instructions.
If you place this product on a lacquer-coated desk, the color may adhere to the bottom of the stand dueto the composition of the rubber. Check the desk surface before use.

● Installation Requirements

When installing the monitor in a rack, ensure that there is adequate space around the sides, back andtop of the monitor.

Attention• Position the monitor so that there is no light to interfere with the screen.

Chapter 2 Installation / Connection15

2-2. Connecting Cables
Attention• Check that the monitor and the PC are powered off.
• When replacing the current monitor with this monitor, refer to “4-2. Compatible Resolutions” (page 22) tochange the PC settings for resolution and vertical scan frequency to those that are available for this monitor,
before connecting the PC.
• A protective cap may be attached to the connector on the monitor. Remove the cap before connecting the cable.

1. Connect signal cables.

Check the shapes of the connectors, and connect the cables. After connecting the DVI cable, tighten
the fasteners to secure the connector.
When enabling PinP Display, refer to the Installation Manual (on the CD-ROM).
Single screen display

Signal cable:
DD300DL
Signal cable: PP300

or

Attention• When using the monitor for single screen display via DisplayPort, connect the monitor to the input connector.
• When the monitor is used for single screen display via DVI, the refresh rate (the frequency at which theimage on the screen is refreshed) will be reduced. Using DisplayPort 1 is therefore recommended for single
screen display. When using the monitor for single screen display via DVI, you need to select “DVI” for “Input”
in “Preferences” in the Setting menu. For details, refer to the Installation Manual (on the CD-ROM).

16

Chapter 2 Installation / Connection

PbyP (dual-screen) display
Attention• You need to configure “Input” in “Preferences” in the Setting menu to select a combination of signals to be
displayed. For details, refer to the Installation Manual (on the CD-ROM).
• When the monitor is used in PbyP mode to display images from two PCs, use of some quality control
features such as calibration may be restricted.

For DisplayPort 1 / DisplayPort 2

For DisplayPort 1 / DVI

Signal cable:
DD300DL

Signal cable:
PP300

Signal cable: PP300

Signal cable: PP300

Note• When the supplied DisplayPort cable (PP028) is connected toand , the signal input tocan bedisplayed in PbyP mode. In this case, you need to set “Signal Format” - “DisplayPort 1” - “Version” in the
Administrator Setting menu to “1.2”, and “Daisy Chain” to “On”.

Signal cable: PP300

Signal cable: PP028

Chapter 2 Installation / Connection

17

Setting up a daisy-chain connection to another monitor
The signal input tois output to another monitor.

Attention• Visit the EIZO website for information about monitors and graphic boards that can be used for the daisychain connection: http://www.eizoglobal.com• You need to set “Signal Format” - “DisplayPort 1” - “Version” in the Administrator Setting menu to “1.2”, and“Daisy Chain” to “On”.

Connecting to a Computer
Signal cable: PP300

Signal cable: PP300

2. Plug the power cord into a power outlet and the power connector on themonitor.

Insert the power cord fully into the monitor.

3. When you use RadiCS / RadiCS LE or connect a USB device (peripheral device
that supports USB) to the monitor, connect the USB cable to the monitor's
USB upstream port and the PC.
Connecting to the Monitor

Connecting to a Computer

Attention• The PC on which RadiCS / RadiCS LE has been installed for monitor quality control purposes must beconnected to .

18

Chapter 2 Installation / Connection

2-3. Turning On the Power
1. Touch

to turn on the power to the monitor.

The power switch indicator of the monitor lights up green.
If the indicator does not light up, see “Chapter 3 No-Picture Problem” (page 20).
Note

• When the power of the monitor is not on, touching any button except

causesto flash.

2. Turn on the PC.

The screen image appears.
If an image does not appear, refer to “Chapter 3 No-Picture Problem” (page 20) for additional
advice.
Attention• For the maximum power saving, it is recommended that the Power button be turned off. When not using
the monitor, you can turn off the main power supply or disconnect the power plug so that the power is cut
completely.
Note• In order to maximize the monitor's lifespan by impeding brightness degradation and to reduce power
consumption, carry out the following:- Use the power saving function of the PC or monitor.
- Turn off the monitor after using it.

2-4. Adjusting the Screen Height and Angle
Hold left and right edge of the monitor with both hands, and adjust the screen height, tilt and swivel of thescreen to the best condition for working.

Attention• After the adjustment is finished, make sure that the cables are correctly connected.

Chapter 2 Installation / Connection

19Chapter 3 No-Picture Problem
Problem
1. No picture• Power switch indicator does not light up• Power switch indicator lights up: Green• Power switch indicator lights up: Orange

Possible cause and remedy• Check whether the power cord is connected properly.
• Turn the main power switch on.
• Touch .
• Turn off the main power, and then turn it on again.
• Increase "Brightness", "Contrast", or "Gain" in the Setting
menu. For details, refer to the Installation Manual (on the
CD-ROM).
• Turn off the main power, and then turn it on again.
• Switch the input signal. For details, refer to the Installation
Manual (on the CD-ROM).
• Move the mouse or press any key on the keyboard.
• Check whether the PC is turned on.
• If the Presence Sensor is set to "On", the monitor may be inpower saving mode. Try moving closer to the monitor.
• Check whether the signal cable is connected properly. If"DisplayPort 1" is selected for the input signal, connect it to.
is used for output when a daisy-chain connection is
set up.
• Turn off the main power, and then turn it on again.
• Power switch indicator blinks: Orange,• Connect via the signal cable specified by EIZO. Turn off theGreenmain power, and then turn it on again.
• If the signal cable is connected to DisplayPort 1 (), tryswitching the DisplayPort version. For details, refer to the
Installation Manual (on the CD-ROM).
2. The message below appears.
This message appears when the signal is not input correctly
even though the monitor is functioning properly.
• The message shown left may appear, because some PCs do• This message appears when no signal is
not output the signal immediately after power-on.
input.
• Check whether the PC is turned on.
Example:• Check whether the signal cable is connected properly.
• Switch the input signal. For details, refer to the Installation
Manual (on the CD-ROM).
• If the signal cable is connected to DisplayPort 1 (), tryswitching the DisplayPort version. For details, refer to the
Installation Manual (on the CD-ROM).
• Is the signal cable connected toto display a DisplayPort
signal on a single screen? The signal cannot be displayed ifor .
the cable is connected to• Turn off the main power, and then turn it on again.
• Check whether the PC is configured to meet the resolution• The message indicates that the input signal
and vertical scan frequency requirements of the monitor (seeis out of the specified frequency range.
“4-2. Compatible Resolutions” (page 22)).
Example:• Reboot the PC.
• Select an appropriate setting using the graphics board’s
utility. For more information, refer to the User’s Manual of thegraphics board.

20

Chapter 3 No-Picture Problem

Chapter 4 Specifications
4-1. Specifications List
LCD Panel

Type
Backlight
Size

Color (IPS)LED
79 cm (31.1 inch)

Resolution (H x V)

4096 × 2160

Display Size (H x V)

698.0 mm × 368.1 mm

Pixel Pitch

0.1704 mm

Display Colors

10-bit (DisplayPort*1): 1073.74 million colors (Max.)
8-bit (DisplayPort*1 / DVI): 16.77 million colors

Video Signals

Viewing Angles(H / V, typical)
Recommended
Brightness
Response Time (typical)

178° / 178°

Input Terminals

DisplayPort × 2, DVI-D (dual link) × 1

Output Terminal

DisplayPort × 1

Horizontal scanningfrequency
Vertical scan frequency

31 kHz - 134 kHz

Frame Synchronizationmode
Dot clock

270 cd/m2
20 ms (black -> white -> black)

DisplayPort: 29 Hz - 61 Hz (720 × 400 : 69 Hz - 71 Hz)
DVI:
14 Hz - 61 Hz (720 × 400 : 69 Hz - 71 Hz)
29.5 Hz - 30.5 Hz, 59 Hz - 61 Hz

USB

Port

DisplayPort : 25 MHz - 570 MHz (Single screen display)
25 MHz - 300 MHz (PbyP / PinP)DVI :25 MHz - 300 MHz
Upstream port × 2, downstream port × 3

Standard

USB Specification Revision 2.0

Power

Input
Maximum Power
Consumption
Power Save Mode
Standby Mode
Dimensions ( W × H × D )100 - 240 VAC ±10 %, 50 / 60 Hz 1.30 - 0.60 A
125 W or lessPhysicalSpecifications

Operating
Environmental
Requirements

1.6 W or less*2
1.6 W or less*3
733.0 mm × 474.0 mm - 574.0 mm × 225.0 mm (Tilt: 0˚)
733.0 mm × 508.0 mm - 608.0 mm × 267.0 mm (Tilt: 30˚)Dimensions ( W × H × D ) 733.0 mm × 408.0 mm × 64.5 mm(Without Stand)
Net Weight
Approx. 11.7 kg
Net Weight
Approx. 7.5 kg(Without Stand)
Height Adjustment Range 100 mm (Tilt: 0˚)
Tilt
Up 30˚, down 5˚Swivel70˚Temperature
0 ˚C - 35 ˚C (32 ˚F - 95 ˚F)
Humidity
20 % - 80 % R.H. (no condensation)
Air Pressure
540 hPa - 1060 hPa

Chapter 4 Specifications

21

Temperature-20 ˚C - 60 ˚C (-4 ˚F - 140 ˚F)
Transportation/ Storage
Humidity
10 % - 90 % R.H. (no condensation)
Environmental
Air Pressure
200 hPa - 1060 hPa
Requirements*1 The colors of DisplayPort signals displayed on the monitor may differ depending on the monitor settings. For
details of display colors and settings, see “4-2. Compatible Resolutions” (page 22).
*2 When DisplayPort 1 input is used, the USB upstream port is not connected, “Input”: “Single DisplayPort 1”,“DDC”: “Off”, “DP Power Save”: “On”, “DisplayPort 1” - “Version”: “1.1”,
Other settings: Default settings, no external load is connected*3 When the USB upstream port is not connected, “DP Power Save”: “On”, “DisplayPort 1” - “Version”: “1.1,“DDC”: “Off”, Other settings: Default settings, no external load is connected

4-2. Compatible Resolutions
The monitor supports the following resolutions. For information about the compatible resolutions of the
PinP sub window, refer to the Installation Manual.
√: Supported
DisplayPort
Single screen
Dual screen
display
display(PbyP)*3
Ver. 1.2*1 Ver. 1.1*2

Resolution(H x V)

Verticalfrequency(Hz)

720 × 400

70

√

√

640 × 48060√

800 × 600

60

1024 × 768

DVI
Single screen
display

Dual screen
display (PbyP)

√

√

√

√

√

√

√

√

√

√

√

√

60

√

√

√

√

√

1280 × 1024

60

√

√

√

√

√

1600 × 1200

60

√

√

√

√

√

2048 × 2160

30

-

-

√

-

√

2048 × 2160

60

-

-

√*4

-

√*4

3840 × 2160

15

-

-

-

√

-

3840 × 2160

30

-

√

-

√

-

3840 × 2160

60

√

-

-

-

-

4096 × 2160

15

-

-

-

√

-

30

-

√

*4

-

√

*4

-

60

*4

-

-

-

-

4096 × 2160
4096 × 2160√

*1 When "Daisy Chain" is set to "Off": 10-bit input, when "Daisy Chain" is set to "On": 8-bit input*2 10 bit input*3 Dual screen display using "DisplayPort 1 / DisplayPort 2": 10-bit input, dual screen display using "DisplayPort
1 / DVI": 8-bit input*4 Recommended resolution

4-3. Optional Accessories
The following accessories are available separately.
For the latest information about the optional accessories and information about the latest compatible
graphics board, refer to our web site. http://www.eizoglobal.com
Calibration Kit
Network QC Management Software
Cleaning Kit
Comfort Light for Reading Rooms

22

Chapter 4 Specifications

EIZO "RadiCS UX1" Ver. 4.5.5 or later
EIZO "RadiCS Version Up Kit" Ver. 4.5.5 or later
EIZO "RadiNET Pro" Ver. 4.5.5 or later
EIZO "RadiNET Pro Lite" Ver. 4.5.5 or later
EIZO "ScreenCleaner"
EIZO "RadiLight"

Appendix
Medical Standard• It shall be assured that the final system is in compliance to IEC60601-1-1 requirement.
• Power supplied equipment can emit electromagnetic waves, that could influence, limit or result in
malfunction of the monitor. Install the equipment in a controlled environment, where such effects
are avoided.

Classification of Equipment-

Type of protection against electric shock : Class I
EMC class: EN60601-1-2:2015 Group 1 Class B
Classification of medical device (EU): Class I
Mode of operation : Continuous
IP Class : IPX0

Appendix

23

FCC Declaration of Conformity
For U.S.A., Canada Only
FCC Declaration of Conformity
We, the Responsible Party

EIZO Inc.
5710 Warland Drive, Cypress, CA 90630
Phone: (562) 431-5011

declare that the product

Trade name: EIZO
Model: RadiForce MX315W

is in conformity with Part 15 of the FCC Rules. Operation of this product is subject to thefollowing two conditions: (1) this device may not cause harmful interference, and (2) thisdevice must accept any interference received, including interference that may cause undesired
operation.
This equipment has been tested and found to comply with the limits for a Class B digital
device, pursuant to Part 15 of the FCC Rules. These limits are designed to provide reasonable
protection against harmful interference in a residential installation. This equipment generates,
uses, and can radiate radio frequency energy and, if not installed and used in accordance with
the instructions, may cause harmful interference to radio communications. However, there is
no guarantee that interference will not occur in a particular installation. If this equipment does
cause harmful interference to radio or television reception, which can be determined by turning
the equipment off and on, the user is encouraged to try to correct the interference by one ormore of the following measures.
* Reorient or relocate the receiving antenna.
* Increase the separation between the equipment and receiver.
* Connect the equipment into an outlet on a circuit different from that to which the receiver isconnected.
* Consult the dealer or an experienced radio/TV technician for help.
Changes or modifications not expressly approved by the party responsible for compliance could
void the user’s authority to operate the equipment.
Note
Use the attached specified cable below or EIZO signal cable with this monitor so as to keepinterference within the limits of a Class B digital device.
- AC Cord- Shielded Signal Cable (enclosed)

Canadian NoticeThis Class B digital apparatus complies with Canadian ICES-003.
Cet appareil numérique de le classe B est comforme à la norme NMB-003 du Canada.

24

AppendixDeclaration of compliance with India RoHS
We, EIZO Corporation, hereby declare and guarantee that this product has been designed andmanufactured in compliance with the E-Waste management rule which prohibit the inclusion of thefollowing substances except for the exemptions listed in schedule II.
- Lead, Mercury, Hexavalent Chromium, Polybrominated Biphenyls or Polybrominated Diphenyl Ethers
exceeding a concentration of 0.1% by weight in homogeneous materials- Cadmium exceeding a concentration of 0.01% by weight in homogeneous materials
For information on proper disposal and recycling of the product, please refer to the following website.
www.eizoglobal.com/i/ewaste-in/

EMC Information

The RadiForce series has a performance that appropriately displays images.

Environments of Intended Use
The RadiForce series is intended to be used in Professional healthcare facility environments such as
clinics and hospitals.
The following environments are not suitable for the RadiForce series to be used:• Home healthcare environments• In the vicinity of high-frequency surgical equipments such as electrosurgical knives• In the vicinity of short-wave therapy equipments• RF shielded room of the medical equipment systems for MRI• In shielded location Special environments• Installed in vehicles including ambulances.
• Other special environments

WARNINGThe RadiForce series requires special precautions regarding EMC and need to be installed. You
need to carefully read EMC Information and the "PRECAUTIONS" section in this document, andobserve the following instructions when installing and operating the product.
Be sure to use the cables attached to the product, or cables specified by EIZO.
Use of cables other than those specified or provided by EIZO of this equipment could result inincreased electromagnetic emissions or decreased electromagnetic immunity of this equipment andimproper operation.
Cable length: 3 m max.
The RadiForce series should not be used adjacent to or stacked with other equipment. If adjacentor stacked use is necessary, the equipment or system should be observed to verify normal
operation in the configuration in which it will be used.
When using a portable RF communication equipment, keep it 30 cm (12 inches) or more away from any part,
includingcables, of the RadiForce series. Otherwise, degradation of the performance of this equipment could result.
Anyone who connects additional equipment to the signal input part or signal output parts,
configuring a medical system, is responsible that the system complies with the requirements of
IEC / EN60601-1-2.

Appendix25

Technical Descriptions
Electromagnetic emissions

The RadiForce series is intended for use in the electromagnetic environment specified below.
The customer or the user of the RadiForce series should assure that it is used in such an environment.

Emission test

Compliance

RF emissions
CISPR11 / EN55011

Group 1

RF emissions
CISPR11 / EN55011
Harmonic emissions
IEC / EN61000-3-2
Voltage fluctuations /flicker emissions
IEC / EN61000-3-3Class BClass DElectromagnetic environment - Guidance

The RadiForce series uses RF energy only for its internal function.
Therefore, its RF emission are very low and are not likely to cause any interference innearby electronic equipment.
The RadiForce series is suitable for use in all establishments, including domestic
establishments and those directly connected to the public low-voltage power supply
network that supplies buildings used for domestic purposes.

Complies

Electromagnetic immunity
The RadiForce series has been tested at the following compliance levels according to the testing requirements for professional healthcare facility environments defined in IEC / EN60601-1-2.
Customers and users of the RadiForce series must ensure that the RadiForce series is used in the following environments:

Immunity test
Electrostatic discharge (ESD)
IEC / EN61000-4-2
Electrical fast transients / bursts
IEC / EN61000-4-4
Surges
IEC / EN61000-4-5
Voltage dips, shortinterruptions andvoltage variations on
power supply input
lines
IEC / EN61000-4-11
Power frequency
magnetic fields
IEC / EN61000-4-8

26

Appendix

Test level for professional
healthcare facility
environments

Compliance level

Electromagnetic environment - Guidance±8 kV contact discharge±15 kV air discharge±8 kV contact discharge±15 kV air discharge±2 kV power lines±1 kV input / output lines±2 kV power lines±1 kV input / output lines

Floors should be wood, concrete or ceramic tile.
If floors are covered with synthetic material, therelative humidity should be at least 30%.
Mains power quality should be that of a typicalcommercial or hospital environment.

±1 kV line to line
±2 kV line to ground
0 % UT (100 % dip in UT)
0.5 cycles and 1 cycle
70 % UT (30 % dip in UT)
25 cycles
0 % UT (100 % dip in UT)
5 sec±1 kV line to line
±2 kV line to ground
0 % UT (100 % dip in UT)
0.5 cycles and 1 cycle
70 % UT (30 % dip in UT)
25 cycles
0 % UT (100 % dip in UT)
5 sec

30 A/m(50 / 60 Hz)

30 A/mMains power quality should be that of a typicalcommercial or hospital environment.
Mains power quality should be that of a typical commercial or hospital environment. If theuser of the RadiForce series requires continued
operation during power mains interruptions, it
is recommended that the RadiForce series be
powered from an uninterruptible power supply ora battery.
Power frequency magnetic fields should be at levels
characteristic of a typical location in a typical commercial
or hospital environment.
The product should be kept at least 15 cm away fromthe source of power frequency magnetic fields duringuse.

Electromagnetic immunity
The RadiForce series has been tested at the following compliance levels according to the testing requirements for professional healthcare facility environments defined in IEC / EN60601-1-2.
Customers and users of the RadiForce series must ensure that the RadiForce series is used in the following environments:

Immunity test

Test level for professional
healthcare facility
environments

Compliance level

Electromagnetic environment - GuidancePortable and mobile RFcommunications equipment
should be used no closer toany part of the RadiForceseries, including cables,
than the recommended
separation distance calculated from the equation
applicable to the frequencyof the transmitter.

Conducted disturbances induced by
RF fields
IEC / EN61000-4-6

Radiated RF fields
IEC / EN61000-4-3

Recommended separation
distance
3 Vrms3 Vrms150 kHz - 80 MHz
6 Vrms
ISM bands between
150 kHz and 80 MHz

6 Vrms3 V/m
80 MHz - 2.7 GHz

3 V/m

d = 1.2√P

d = 1.2√P, 80 MHz - 800 MHz
d = 2.3√P, 800 MHz - 2.7 GHz
Where “P” is the maximum output power rating of
the transmitter in watts (W) according to the transmitter manufacturer and “d” is the recommended
separation distance in meters (m).
Field strengths from fixed RF transmitters, as
determined by an electromagnetic site surveya),should be less than the compliance level in eachfrequency rangeb).
Interference may occur in the vicinity of equipment marked with the following symbol.

Note 1
Note 2
Note 3

UT is the a.c. mains voltage prior to application of the test level.
At 80 MHz and 800 MHz, the higher frequency range applies.
Guidelines regarding conducted disturbances induced by RF fields or radiated RF fields may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people.
Note 4The ISM bands between 150 kHz and 80 MHz are 6.765 MHz to 6.795 MHz, 13.553 MHz to 13.567 MHz, 26.957
MHz to 27.283 MHz, and 40.66 MHz to 40.70 MHz.
a)
Field strengths from fixed transmitters, such as base stations for radio (cellular / cordless) telephones and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered.
If the measured field strength in the location in which the RadiForce series is used exceeds the applicable RF compliancelevel above, the RadiForce series should be observed to verify normal operation. If abnormal performance is observed,
additional measures may be necessary, such as reorienting or relocating the RadiForce series.
b)Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m.

Appendix27

Recommended separation distances between portable or mobile RF communication equipment
and the RadiForce Series

The RadiForce series is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled.
The customer or the user of the RadiForce series can help prevent electromagnetic interference by maintaining a minimum
distance between portable and mobile RF communications equipment (transmitters) and the RadiForce series.
Immunity to proximity fields from following RF wireless communication equipments has been confirmed:
Minimum
Compliance
Test freMaximum
separation IEC / EN60601
Bandwidth a)
a)
b)
level
quencypower
ServiceModulation
distance test level (V/m)
(MHz)
(V/m)
(MHz)
(W)
(m)
385

380 - 390TETRA 400450

430 - 470

GMRS 460,
FRS 460

710
745
780
810
870
930

704 - 787

LTE Band 13, 17

800 - 960

GSM 800 / 900,TETRA 800,
iDEN 820CDMA 850,
LTE Band 5

1720
1845
1970

1700 - 1990 GSM 1800;
CDMA 1900;
GSM 1900;
DECT;
LTE Band 1, 3, 4,
25;
UMTS2400 - 2570 Bluetooth,
WLAN,
802.11 b/g/n,
RFID 2450,
LTE Band 75100 - 5800 WLAN 802.11 a/n

2450

Pulse modulation b) 1.8
18 Hz
FM
2
±5 kHz deviation
1 kHz sine
Pulse modulation b) 0.2
217 Hz

0.3

27

27

0.3

28

28

0.3

9

9

Pulse modulation b) 2
18 Hz

0.3

28

28

Pulse modulation b) 2
217 Hz

0.3

28

28

Pulse modulation b) 2
217 Hz

0.3

28

28

0.3

9

9

5240
Pulse modulation b) 0.2
217 Hz55005785a) For some services, only the uplink frequencies are included.
b) carrier are modulated using a 50 % duty cycle square wave signal.

The RadiForce series is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. For other portable and mobile RF communication equipments (transmitters), minimum distance between portable and mobile RF communications equipment(transmitters) and the RadiForce series as recommended below, according to the maximum output power of the communications equipment.

Rated maximum output
power of transmitter(W)

Separation distance according to frequency of transmitter(m)
150 kHz to 80 MHz
80 MHz to 800 MHz
800 MHz to 2.7 GHz
d = 1.2√P
d = 1.2√P
d = 2.3√P

0.01

0.12

0.12

0.23

0.1

0.38

0.38

0.73

1

1.2

1.2

2.3

10

3.8

3.8

7.3

100

12

12

23

For transmitters rated at a maximum output power not listed above, the recommended separation distance “d” in meters (m)can be estimated using the equation applicable to the frequency of the transmitter, where “P” is the maximum output power
rating of the transmitter in watts (W) according to the transmitter manufacturer.

Note 1
Note 2

At 80 MHz and 800 MHz, the separation distance for a higher frequency range applies.
These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption andreflection from structures, objects and people.

Cable

LengthSignal cable:

PP300

3m

Signal cable:

PP028

0.28 mSignal cable:

DD300DL3m

USB cable:

UU300

3m

Power cord(with earth)3m

28

Appendix153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan
Carl-Benz-Straße 3, 76761 Rülzheim, Germany
中国苏州市苏州工业园区展业路 8 号中新科技工业坊 5B
1 Queens Square, Ascot Business Park, Lyndhurst Road,
Ascot, Berkshire, SL5 9FE, UK
Moosacherstrasse 6, Au, CH-8820 Wädenswil, Switzerland

www.eizoglobal.com
Copyright © 2023 EIZO Corporation. All rights reserved.

00N0N386A1
IFU-MX315W
1st Edition - February 20th, 2023 Printed in Japan.