XA01403C/07/A3/04.18
71391367

Products

Solutions

Services

Safety Instructions
OUSAF12, OUSAF21, OUSAF22,
OUSAF44, OUSAF46, OUSTF10
Ergänzung zu / Supplement to / Complément à:
BA00497C, BA00471C, BA00472C, BA00416C,
BA00500C, BA01349C
Sicherheitshinweise für elektrische Betriebsmittel im
explosionsgefährdeten Bereich
Safety instructions for electrical apparatus for explosionhazardous areas

0

Conseils de sécurité pour matériels électriques destinés
aux zones explosibles

XA01403C

2OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

Endress+Hauser

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

Endress+Hauser

XA01403C

3

Flexible conduit

ATEX
approved
EXP-1 Lamp

II 2G

Explosion Proof
Junction Box(supplied bycustomer)

Ex cable gland(supplied bycustomer)

OUSAF12OUA 260or
OUA 261
CUA 261

Attach to Earth Ground

Cable set
CUK80

Intrinsic safe
optical detector

II 3G

Zone 2

9- The equipment must be connected to ground and the resistance shall not exceed 10 Ω

- Do not remove cover while circuit is alive.

- Wiring between sensor and CM44P to conform with applicable safety standards.

Attach to Earth Ground

Conduit or armored cable

Title

Material

Project

CM44P

Zulassungszeichnung CM44P
ATEXControl Drawing CM44P ATEX

Designed

SAP material no.

WeightTameggerR

1 of 3

Page

Doc. partTameggerR

TameggerR

A3

FormatVersion03
Volume2017-11-22

1001186

Pressure contr.
Released

Released

2016-02-05
2017-11-22

No. of document

StatusApproval contr.

Welding contr.

Drawing chng.
Number of changes

107908Tolerance

Liquiline CM44P

Notif. no.

Scale

Conduit or armored cable

Non hazardous location

Refer to protection notice ISO 16016
Classification
Confidential

Conduit or armored cableI n t r i n s i c S a f e t y B a r r i e r (supplie d)
MTL7760AC- After de-energizing delay 5 minutes before opening the enclosure of the explosion proof lamp.
assembly EXP-1 allowing the lamp to cool down.

- Sealing fittings and conduit supplied by customer.

Notes:- For installation details refer to CM44P and sensor operating / installation instruction.

II 1G

Zone 0

Hazardous location
Gas Group IIC, IIB, IIA
Temperature class T5
Zone 1

Control Drawing CM44P
with OUSAF12

Cable set CUK80

4Endress+Hauser Conducta GmbH & Co. KG
Gerlingen, Germany

XA01403C
OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

Endress+Hauser

Endress+Hauser
Flexible conduit

ATEX
approved
EXP-1 Lamp

ATEX
approved
EXP-1 Lamp

II 2G

Explosion Proof
Junction Box(supplied bycustomer)

Ex cable gland(supplied bycustomer)

OUSAF22
OUSTF10OUA 260or
OUA 261
CUA 261

OUSAF21

Flow assembly

Attach to Earth Ground

Cable set
CUK80

Intrinsic safe
optical detector

Intrinsic safe
optical detector

II 3G

Zone 2

9- The equipment must be connected to ground and the resistance shall not exceed 10 Ω

- Do not remove cover while circuit is alive.

- Wiring between sensor and CM44P to conform with applicable safety standards.

Attach to Earth Ground

Conduit or armored cable

Material

Project

CM44P

Zulassungszeichnung CM44P
ATEXControl Drawing CM44P ATEX

Title

SAP material no.

VolumeA3

Format03

1001186
WeightVersionPressure contr.
Released

Released
No. of document

2 of 3

Page

Doc. partTameggerR

Welding contr.

Status

2017-11-22

2017-11-22
Approval contr.

TameggerR
TameggerR2016-02-05
Designed

Drawing chng.

107908Tolerance

Liquiline CM44P

Number of changes

Notif. no.

Scale

Conduit or armored cable

Non hazardous location

Refer to protection notice ISO 16016
Classification
Confidential

Conduit or armored cableI n t r i n s i c S a f e t y B a r r i e r ( s u pp l i ed )MTL7760AC
I n t r i n s i c S a f e t y B a r r i e r ( s u pp l i ed )MTL7760AC- After de-energizing delay 5 minutes before opening the enclosure of the explosion proof lamp.
assembly EXP-1 allowing the lamp to cool down.

- Sealing fittings and conduit supplied by customer.

Notes:- For installation details refer to CM44P and sensor operating / installation instruction.

II 1G

Zone 0Endress+Hauser Conducta GmbH & Co. KG
Gerlingen, Germany

Hazardous location
Gas Group IIC, IIB, IIA
Temperature class T5
Zone 1

Cable set CUK80

Control Drawing CM44P
with OUSAF21, OUSAF22 or OUSTF10

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10
XA01403C

5

Flexible conduit

ATEX
approved
EXP-1 Lamp

II 2G

Explosion Proof
Junction Box(supplied bycustomer)

Ex cable gland(supplied bycustomer)

OUSAF44
OUSAF46

OUA 260or
OUA 261
CUA 261

Attach to Earth Ground

Cable set
CUK80

Intrinsic safe
optical detector+ reference
detector

II 3G

Zone 2

9- The equipment must be connected to ground and the resistance shall not exceed 10 Ω

- Do not remove cover while circuit is alive.

- Wiring between sensor and CM44P to conform with applicable safety standards.

Attach to Earth Ground

Conduit or armored cable

Title

Material

Project

CM44P

Zulassungszeichnung CM44P
ATEXControl Drawing CM44P ATEX

Designed

SAP material no.

TameggerR

3 of 3

Page

Doc. partTameggerR

TameggerR

A3

Format03
Weight1001186
Volume2017-11-22
VersionReleased

2017-11-22

2016-02-05

No. of document

Pressure contr.
Released

StatusApproval contr.

Welding contr.

Drawing chng.
Number of changes

107908Tolerance

Liquiline CM44P

Notif. no.

Scale

Conduit or armored cable

Non hazardous location

Refer to protection notice ISO 16016
Classification
Confidential

Conduit or armored cableI n t r i n s i c S a f e t y B a r r i e r ( supp li ed)
MTL7760AC- After de-energizing delay 5 minutes before opening the enclosure of the explosion proof lamp.
assembly EXP-1 allowing the lamp to cool down.

- Sealing fittings and conduit supplied by customer.

Notes:- For installation details refer to CM44P and sensor operating / installation instruction.

II 1G

Zone 0

Hazardous location
Gas Group IIC, IIB, IIA
Temperature class T5
Zone 1

Control Drawing CM44P
with OUSAF44 or OUSAF46

Cable set CUK80

6Endress+Hauser Conducta GmbH & Co. KG
Gerlingen, Germany

XA01403C
OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

Endress+Hauser

Explosion Proof
Junction Box(supplied)

OUSAF12OUA 260or
OUA 261
CUA 261

Attach to Earth Ground

Cable set
CUK80

Intrinsic safe
optical detector

DIV2

Conduit9- The equipment must be connected to ground and the resistance shall not exceed 10 Ω

- Do not remove cover while circuit is alive.

- Wiring between sensor and CM44P to conform with applicable safety standards.

- After de-energizing delay 5 minutes before opening the enclosure of the explosion proof lamp.
assembly EXP-1 allowing the lamp to cool down.

- Sealing fittings and conduit supplied by customer.

Attach to Earth Ground

ConduitConduitNon hazardous location

Material

Project

CM44P

Zulassungszeichnung CM44P FM

Control Drawing CM44P FM

Title

Refer to protection notice ISO 16016
Classification
Confidential

I n t r i n s i c S a f e t y B a r r i e r (supplied)
MTL7760AC

Notes:- For installation details refer to CM44P and sensor operating / installation instruction.

Flexible conduit

FM approved
EXP-1 Lamp

Hazardous location Class I
Gas Group B, C, D
Temperature class T5
DIV1

Endress+Hauser Conducta GmbH & Co. KG
Gerlingen, Germany

Endress+Hauser
Cable set CUK80

Control Drawing CM44P
with OUSAF12

Designed

SAP material no.

Weight1 of 3

Page

Doc. partTameggerR

TameggerR

TameggerR

A3

FormatVersion03
Volume2017-11-22

1001187

Pressure contr.
Released

Released

2017-11-22

2016-02-05

No. of document

StatusApproval contr.

Welding contr.

Drawing chng.
Number of changes

107908Tolerance

Notif. no.

Scale

Liquiline CM44P

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10
XA01403C

7

Explosion Proof
Junction Box(supplied)

OUSAF22
OUSTF10OUA 260or
OUA 261
CUA 261

OUSAF21

Flow assembly

Attach to Earth Ground

Cable set
CUK80

Intrinsic safe
optical detector

Intrinsic safe
optical detector

DIV2

Conduit9- The equipment must be connected to ground and the resistance shall not exceed 10 Ω

- Do not remove cover while circuit is alive.

- Wiring between sensor and CM44P to conform with applicable safety standards.

- After de-energizing delay 5 minutes before opening the enclosure of the explosion proof lamp.
assembly EXP-1 allowing the lamp to cool down.

- Sealing fittings and conduit supplied by customer.

Attach to Earth Ground

ConduitConduitNon hazardous location

Material

Project

CM44P

Zulassungszeichnung CM44P FM

Control Drawing CM44P FM

Title

Refer to protection notice ISO 16016
Classification
Confidential

I n t r i n s i c S a f e t y B a r r i e r (s u pp l i ed )MTL7760AC
I n t r i n s i c S a f e t y B a r r i e r (s u pp l i ed )MTL7760AC

Notes:- For installation details refer to CM44P and sensor operating / installation instruction.

Flexible conduit

FM approved
EXP-1 Lamp

FM approved
EXP-1 Lamp

Hazardous location Class I
Gas Group B, C, D
Temperature class T5
DIV1

Endress+Hauser Conducta GmbH & Co. KG
Gerlingen, Germany

8
Cable set CUK80

Control Drawing CM44P
with OUSAF21, OUSAF22 or OUSTF10

Designed

SAP material no.

VolumeTameggerR

2 of 3

Page

Doc. partTameggerR

TameggerR

A3

Format03
WeightVersion2017-11-22

1001187

Pressure contr.
Released

Released

2017-11-22

2016-02-05

No. of document

StatusApproval contr.

Welding contr.

Drawing chng.
Number of changes

107908Tolerance

Notif. no.

Scale

Liquiline CM44P

XA01403C
OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

Endress+Hauser

Explosion Proof
Junction Box(supplied)

OUSAF44
OUSAF46

OUA 260or
OUA 261
CUA 261

Attach to Earth Ground

Cable set
CUK80

Intrinsic safe
optical detector+ reference
detector

DIV2

Conduit9- The equipment must be connected to ground and the resistance shall not exceed 10 Ω

- Do not remove cover while circuit is alive.

- Wiring between sensor and CM44P to conform with applicable safety standards.

- After de-energizing delay 5 minutes before opening the enclosure of the explosion proof lamp.
assembly EXP-1 allowing the lamp to cool down.

- Sealing fittings and conduit supplied by customer.

Attach to Earth Ground

ConduitConduitNon hazardous location

Material

Project

CM44P

Zulassungszeichnung CM44P FM

Control Drawing CM44P FM

Title

Refer to protection notice ISO 16016
Classification
Confidential

I n t r i n s i c S a f e t y B a r r i e r (supplied)
MTL7760AC

Notes:- For installation details refer to CM44P and sensor operating / installation instruction.

Flexible conduit

FM approved
EXP-1 Lamp

Hazardous location Class I
Gas Group B, C, D
Temperature class T5
DIV1

Endress+Hauser Conducta GmbH & Co. KG
Gerlingen, Germany

Endress+Hauser
Cable set CUK80

Control Drawing CM44P
with OUSAF44 or OUSAF46

Scale

VolumeSAP material no.

A3

Format03

1001187
WeightVersionPressure contr.
Released

Released
No. of document

3 of 3

Page

Doc. partTameggerR

Welding contr.

Status

2017-11-22

2016-11-22
Approval contr.

TameggerR
TameggerR2016-02-05
Designed

Drawing chng.

107908Tolerance

Number of changes

Notif. no.

Liquiline CM44P

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10
XA01403C

9

XA01403C

10OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

Endress+Hauser

XA01403C

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

OUSAF12, OUSAF21, OUSAF22, OUSAF44,
OUSAF46, OUSTF10
Ergänzung zu / Supplement to / Complément à:
BA00497C, BA00471C, BA00472C, BA00416C, BA00500C, BA01349C
Inhaltsverzeichnis
EU-Konformitätserklärung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

13

Zugehörige Dokumentation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Ergänzende Dokumentation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Temperaturklasse . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Identifizierung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

12

14Endress+Hauser

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

EU-Konformitätserklärung

→ 3

Zugehörige Dokumentation

Dieses Dokument ist fester Bestandteil folgender Betriebsanleitungen:• Betriebsanleitung OUSAF12, BA00497C• Betriebsanleitung OUSAF21, BA00471C• Betriebsanleitung OUSAF22, BA00472C• Betriebsanleitung OUSAF44, BA00416C• Betriebsanleitung OUSAF46, BA01349C• Betriebsanleitung OUSTF10, BA00500C

Ergänzende Dokumentation

Sicherheitshinweise

Endress+Hauser

XA01403C

Kompetenzbroschüre CP00021Z• Explosionsschutz: Richtlinien und Grundlagen• www.endress.com

Die Prozessphotometer-Messstelle, bestehend aus den bescheinigten Prozessphotometern
OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46 oder OUSTF10 und Messkabel CUK80 darf
nur über die bescheinigte Sicherheitsbarriere MTL7760 ac an den Messumformer CM44P angeschlossen werden.
• Der elektrische Anschluss muss entsprechend den zugehörigen Control Drawings in diesem Dokument und den Installationshinweisen in den Prozessphotometer-Betriebsanleitungen ausgeführt
werden.
• Die Prozessphotometer dürfen in Umgebungen eingesetzt werden, die entflammbare Gase undDämpfe enthalten und den Gerätekategorien IIA, IIB, IIC (ATEX) und B, C, D (FM) der Temperaturklasse T5 zugeordnet sind. Gerätekategorie IIC (ATEX) ist gleich der nicht zugelassenen Gerätekategorie A (FM).
• Die Montage muss durch geschultes Personal in Überstimmung mit anwendbaren Richtlinien, z.B.
IEC/EN 60079-14 oder ANSI/NFPA-70 und entsprechend der Betriebsanleitung des jeweiligen
Prozessphotometers und des anzuschließenden Messumformers erfolgen.
• Die Reparatur muss durch geschultes Personal in Übereinstimmung mit anwendbaren Richtlinien,
z.B. IEC/EN 60079-19 und entsprechend seiner und der Betriebsanleitung des anzuschließenden
Messumformers erfolgen.
• Bauteile, die in diese Prozessphotometer als Zubehör oder Ersatzteile eingebaut werden, müssen
durch geschultes Personal unter Beachtung der Dokumentation des Herstellers montiert werden.
• Wenn das Prozessphotometer mit aggressiven Substanzen in Kontakt kommen kann, muss der
Benutzer geeignete Schutzmaßnahmen treffen, damit der zugesagte Schutzgrad der Messeinrichtung nicht beeinträchtigt wird. Aggressive Substanzen sind z.B. saure Flüssigkeiten oder Gase, die
Metall angreifen oder Lösungsmittel, die polymere Materialien beschädigen können. Geeignete
Schutzmaßnahmen sind regelmäßige Prüfungen als Teil von Routinekontrollen oder die Prüfung
der Widerstandsfähigkeit gegenüber speziellen Chemikalien anhand des Werkstoffdatenblatts.
• Der optische Empfänger mit Photodioden wird im Strommodus betrieben. Die Photodioden erzeugen Energie unterhalb der Spannungs-, Strom- und Leistungsgrenzen, die in IEC/EN 60079-11,
Kapitel 5.7 genannt werden. Daher ist die Einrichtung als einfaches Betriebsmittel einzustufen.
• Die Prozessphotometer sind in Temperaturklasse T5 eingestuft, die eine maximale Oberflächentemperatur von 100 °C (212 °F) erlaubt.
• Die Trennung des sicheren vom gefährlich eingestuften Bereich wird erreicht durch die Verwendung von einer oder zwei zugelassenen Sicherheitsbarrieren MTL7760 ac. Der Anschluss derSicherheitsbarriere muss gemäß den zugehörigen Control Drawings in diesem Dokument und den
Installationshinweisen in den Prozessphotometer-Betriebsanleitungen erfolgen.
• Die zugelassene Sicherheitsbarrieren und der Messumformer CM44P dürfen nicht in der gefährlichen Umgebung installiert werden.
• Vor dem Öffnen des Gehäuses muss die explosionsgeschützte Lampe EXP-1 abkühlen. Öffnen Sie
das Photometergehäuse erst 5 Minuten nach dem Abschalten der Energiezufuhr.
• Die explosionsgeschützte Lampe EXP-1 muss so eingebaut werden, dass das Frontfenster vor
mechanischer Beschädigung geschützt ist.
• Die explosionsgeschützte Lampe EXP-1 ist mit einem dauerhaft verbundenem Kabelende versehen, das an einen externen Kreis entsprechend den Anweisungen dieser Anleitung und den Installationshinweisen in den Prozessphotometer-Betriebsanleitungen angeschlossen werden muss.

13

XA01403C

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10• Das Messkabel CUK80 muss vor elektrostatischer Aufladung geschützt werden, wenn es durch die
ATEX Zone 0 geführt wird.
• Kabel, die durch explosionsgefährdete Umgebungen (Class I, Division 1/2) geführt werden, erfordern ein Kabelschutzrohr.
• Die Durchflussarmaturen OUA260/CUA261 dürfen im explosionsgefährdeten Bereich nur mit
Edelstahladaptern verwendet werden.

Temperaturklasse

Photometer

Prozesstemperatur

Umgebungstemperatur

Temperaturklasse

OUSAF12

0 … 90 °C (32 °F … 190 °F) (Dauerbetrieb)
max. 130 °C (266 °F), für max. 2 h-20 … 40 °C (0 … 100 °F)

T5 ≤ 100 °C (≤ 212 °F)

OUSAF21
OUSAF22
OUSAF44
OUSAF46
OUSTF10Bei Einhaltung der angegebenen Prozesstemperaturen treten an den Betriebsmitteln keine für die
jeweilige Temperaturklasse unzulässigen Temperaturen auf.

Identifizierung

Typenschlüssel
Typ

Ausführung

OUSAF12OUSAF21
OUSAF22OUSTF10-

*
*
*
*

OUSAF12OUSAF21
OUSAF22OUSTF10-

*
*
*
*

Armatur, nicht exrelevant

*
*
*
*

A/B/C/D
C/D
C/D
B

*
*
*
*

*
*
*
*

Ausführung

OUSAF44-*

OUSAF46-*22
2

für den Einsatz in explosionsgefährdeten Bereichen, ATEX II 2G Ex db IIC T5 Gb

Armatur, nicht exrelevant

B*

Wellenlänge, Kalibrierung,
nicht ex-relevant

für den Einsatz in explosionsgefährdeten
Bereichen, FM Cl.1, Div. 1, Groups B, C, D

Armatur, nicht exrelevant

*

C

*

Wellenlänge, Kalibrierung,
nicht ex-relevant

für den Einsatz in explosionsgefährdeten
Bereichen, ATEX II 2G Ex db IIC T5 Gb

Armatur, nicht exrelevant

*

B

*

Wellenlänge, Kalibrierung,
nicht ex-relevant

für den Einsatz in explosionsgefährdeten
Bereichen, FM Cl.1, Div. 1, Groups B, C, D

Armatur, nicht exrelevant

*

C

*

für den Einsatz in explosionsgefährdeten
Bereichen, ATEX II 2G Ex db IIC T5 Gb

Armatur, nicht exrelevant

*

*

*

Wellenlänge, Kalibrierung,
nicht ex-relevant

14

*
*
*
*

für den Einsatz in explosionsgefährdeten Bereichen, FM Cl.1, Div. 1, Groups B, C, DTypOUSAF46-

1
1
1

Wellenlänge, Kalibrierung, nicht ex-relevant

Wellenlänge, Kalibrierung, nicht ex-relevant

OUSAF44-

A/B/C/D
C/D
C/D
B

Endress+Hauser

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

XA01403C

Typenschild
Folgende Informationen zu Ihrem Gerät können Sie dem Typenschild entnehmen:• Herstelleridentifikation• Bestellcode• Erweiterter Bestellcode• Seriennummer• Sicherheits- und Warnhinweise• Ex-Kennzeichnung bei Ex-Ausführungen‣ Angaben auf dem Typenschild mit Bestellung vergleichen.
Zertifikate und Zulassungen-Zeichen
Das Produkt erfüllt die Anforderungen der harmonisierten europäischen Normen. Damit erfüllt es
die gesetzlichen Vorgaben der EU-Richtlinien. Der Hersteller bestätigt die erfolgreiche Prüfung desProdukts durch die Anbringung des -Zeichens.
Ex-Zulassungen• ATEX II 2G Ex db IIC T5 Gb• FM Cl.1, Div. 1, Groups B, C, D
Benannte Stelle
Sira Certification Service
Rake Lane, Ecclestone, Chester, CH4 9JN, England

Endress+Hauser

15

XA01403C

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

OUSAF12, OUSAF21, OUSAF22, OUSAF44,
OUSAF46, OUSTF10
Ergänzung zu / Supplement to / Complément à:
BA00497C, BA00471C, BA00472C, BA00416C, BA00500C, BA01349C
Table of contents
EU Declaration of conformity . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17
Associated documentation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17
Supplementary documentation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

17

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17
Temperature class . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18
Identification . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

16

18

Endress+Hauser

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

EU Declaration of conformity→ 3

Associated documentation

This document is an integral part of the following Operating Instructions:• Operating Instructions OUSAF12, BA00497C• Operating Instructions OUSAF21, BA00471C• Operating Instructions OUSAF22, BA00472C• Operating Instructions OUSAF44, BA00416C• Operating Instructions OUSAF46, BA01349C• Operating Instructions OUSTF10, BA00500C

Supplementary
documentation

Safety instructions

Endress+Hauser

XA01403C

Competence Brochure CP00021Z• Explosion Protection: Guidelines and General Principles• www.endress.com

The process photometer measuring point, consisting of the certified process photometers OUSAF12,
OUSAF21, OUSAF22, OUSAF44, OUSAF46 or OUSTF10 and measuring cable CUK80 may only beconnected to the CM44P transmitter via the certified safety barrier MTL7760 ac.
• The electrical connection must be established as illustrated in the relevant control drawings in thisdocument and the installation instructions in the Operating Instructions for the process
photometer.
• The process photometers may be used in environments that contain flammable gases and vapors
and are assigned to apparatus groups IIA, IIB, IIC (ATEX) and B, C, D (FM), temperature class T5.
Apparatus group IIC (ATEX) is equivalent to not approved apparatus group A (FM).
• Installation must be performed by trained personnel in compliance with applicable standards andguidelines, e.g. IEC/EN 60079-14 or ANSI/NFPA-70 and as specified in the Operating Instructions
for the particular process photometer and the transmitter to be connected.
• Sensor repairs must be performed by trained personnel in compliance with applicable standards
and guidelines, e.g. IEC/EN 60079-19 and as specified in the sensor's Operating Instructions andthe Operating Instructions for the transmitter to be connected.
• Component parts which are installed as accessories or replacement parts in these process
photometers must be mounted by trained personnel in compliance with the manufacturer's
documentation.
• If the process photometer can come into contact with aggressive substances, the user must takesuitable protective measures to ensure that the confirmed level of protection of the measuringsystem is not compromised. Examples of aggressive substances include acidic liquids or gases that
corrode metal, or solvents that can damage polymeric materials. Suitable protective measuresinvolve regular checks as part of routine inspections, or checks to verify the resistance of thematerials to special chemicals using the material data sheet.
• The optical receiver with photodiodes is operated in the current mode. The photodiodes generateenergy below the voltage, current and power limits that are specified in IEC/EN 60079-11, Section
5.7. Therefore the equipment is categorized as a simple apparatus.
• The process photometers are assigned to temperature class T5, which permits a maximum surface
temperature of 100 °C (212 °F).
• One or two approved safety barriers MTL7760 ac are used to separate the safe area from the areaclassified as hazardous. The safety barrier must be connected as illustrated in the relevant control
drawings in this document and the installation instructions in the Operating Instructions for theprocess photometer.
• The approved safety barriers and the CM44P transmitter may not be installed in the hazardous
environment.
• The explosion-protected lamp EXP-1 must cool down before the housing is opened. Afterswitching off the power supply, first wait for 5 minutes before you open the photometer housing.
• The explosion-protected lamp EXP-1 must be installed in such a way that the front window isprotected against mechanical damage.
• The explosion-protected lamp EXP-1 has a permanently connected cable end which must beconnected to an external circuit according to the instructions in these Operating Instructions andthe installation instructions in the Operating Instructions for the process photometer.

17

XA01403C

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10• The CUK80 measuring cable must be protected against electrostatic charge if it is routed through
ATEX Zone 0.
• Cables routed through hazardous environments (Class I, Division 1/2) require a cable conduit.
• The OUA260/CUA261 flow assemblies may only be used in the hazardous area with stainlesssteel adapters.

Temperature class

Photometer Process temperature

Ambient temperature

OUSAF12-20 to 40 °C (0 to 100 °F) T5 ≤ 100 °C (≤ 212 °F)

OUSAF21

0 to 90 °C (32 °F to 190 °F) (continuous
operation)
max. 130 °C (266 °F), for max. 2 hTemperature class

OUSAF22
OUSAF44
OUSAF46
OUSTF10

If the specified process temperatures are complied with, temperatures that are not permitted for therespective temperature class will not occur on the equipment.

Identification

Type code
Type

VersionOUSAF12OUSAF21
OUSAF22OUSTF10-

*
*
*
*

OUSAF12OUSAF21
OUSAF22OUSTF10-

*
*
*
*

Assembly, no Exrelevance

*
*
*
*

A/B/C/D
C/D
C/D
B

*
*
*
*

VersionOUSAF44-*

OUSAF46-

*
*
*
*

*

2
2
2

For use in hazardous areas,
ATEX II 2G Ex db IIC T5 Gb

Assembly, no Exrelevance

B*

Wavelength, calibration, no
Ex-relevanceFor use in hazardous areas, FM Cl.1, Div. 1,
Groups B, C, D

Assembly, no Exrelevance

*

C

*

Wavelength, calibration, no
Ex-relevanceFor use in hazardous areas,
ATEX II 2G Ex db IIC T5 Gb

Assembly, no Exrelevance

*

B

*

Wavelength, calibration, no
Ex-relevanceFor use in hazardous areas, FM Cl.1, Div. 1,
Groups B, C, D

Assembly, no Exrelevance

*

C

*

For use in hazardous areas,
ATEX II 2G Ex db IIC T5 Gb

Assembly, no Exrelevance

*

*

*

Wavelength, calibration, no
Ex-relevance

18

*
*
*
*

For use in hazardous areas,
FM Cl.1, Div. 1, Groups B, C, D

Type

OUSAF46-

1
1
1

Wavelength, calibration,
no Ex-relevance

Wavelength, calibration,
no Ex-relevance

OUSAF44-

A/B/C/D
C/D
C/D
B

Endress+Hauser

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

XA01403C

Nameplate
The nameplate provides you with the following information on your device:• Manufacturer identification• Order code
• Extended order code
• Serial number• Safety information and warnings• Ex labeling on hazardous area versions‣ Compare the information on the nameplate with the order.
Certificates and approvals
 markThe product meets the requirements of the harmonized European standards. As such, it complies
with the legal specifications of the EU directives. The manufacturer confirms successful testing of the
product by affixing to it the  mark.
Ex approvals• ATEX II 2G Ex db IIC T5 Gb• FM Cl.1, Div. 1, Groups B, C, D
Designated authority
Sira Certification Service
Rake Lane, Ecclestone, Chester, CH4 9JN, England

Endress+Hauser

19

XA01403C

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

OUSAF12, OUSAF21, OUSAF22, OUSAF44,
OUSAF46, OUSTF10
Ergänzung zu / Supplement to / Complément à:
BA00497C, BA00471C, BA00472C, BA00416C, BA00500C, BA01349C
Sommaire
Déclaration de conformité UE . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

21

Documentation associée . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

21

Documentation complémentaire . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

21

Consignes de sécurité . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

21

Classe de température . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22
Identification . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

20

22Endress+Hauser

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

Déclaration de conformité UE→ 3

Documentation associée

Le présent document fait partie intégrante des manuels de mise en service suivants :• Manuel de mise en service OUSAF12, BA00497C• Manuel de mise en service OUSAF21, BA00471C• Manuel de mise en service OUSAF22, BA00472C• Manuel de mise en service OUSAF44, BA00416C• Manuel de mise en service OUSAF46, BA01349C• Manuel de mise en service OUSTF10, BA00500C

Documentation
complémentaire

Consignes de sécurité

Endress+Hauser

XA01403CBrochure Compétence CP00021Z• Protection contre les explosions : Directives et principes généraux
• www.fr.endress.com

Le point de mesure avec photomètre de process, comprenant les photomètres de process certifiés
OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46 ou OUSTF10 et le câble de mesure CUK80, ne
peut être raccordé au transmetteur CM44P que via la barrière de sécurité certifiée MTL7760 ac.
• Le raccordement électrique doit être réalisé conformément aux schémas de contrôlecorrespondants dans ce document et aux instructions de montage du manuel de mise en service duphotomètre de process.
• Les photomètres de process peuvent être utilisés dans des environnements contenant des gaz etdes vapeurs inflammables et sont affectés aux groupes d'appareils IIA, IIB, IIC (ATEX) et B, C, D(FM), classe de température T5. Le groupe d'appareils IIC (ATEX) est équivalent au grouped'appareils non agréé A (FM).
• L'installation doit être réalisée par du personnel formé conformément aux normes et directives envigueur, par ex. IEC/EN 60079-14 ou ANSI/NFPA-70 et conformément aux spécifications dumanuel de mise en service du photomètre de process concerné et du transmetteur à raccorder.
• Les réparations du capteur doivent être réalisées par du personnel formé conformément aux
normes et directives en vigueur, par ex. IEC/EN 60079-19 et conformément aux spécifications dumanuel de mise en service du capteur et du transmetteur à raccorder.
• Les composants installés comme accessoires ou pièces de rechange dans ces photomètres deprocess doivent être montés par du personnel formé conformément à la documentation dufabricant.
• Si le photomètre de process peut entrer en contact avec des substances agressives, l'utilisateur doit
prendre des mesures de protection appropriées afin de ne pas compromettre le niveau deprotection confirmé de l'ensemble de mesure. Parmi les substances agressives, on trouve lesliquides ou gaz acides qui corrodent les métaux, ou les solvants qui peuvent endommager lesmatières polymérisées. Des mesures de protection appropriées exigent des contrôles réguliers dansle cadre d'inspections de routine, ou des contrôles de vérification de la résistance des matériaux
aux produits chimiques spéciaux à l'aide de fiches techniques de matériaux.
• Le récepteur optique avec photodiodes fonctionne en mode courant. Les photodiodes produisentde l'énergie sous les limites de tension, courant et puissance spécifiées dans IEC/EN 60079-11,
Section 5.7. L'appareil est par conséquent classé comme simple appareil.
• Les photomètres de process sont affectés à la classe de température T5 qui permet une
température de surface maximale de 100 °C (212 °F).
• Une ou deux barrières de sécurité agréées MTL7760 ac sont utilisées pour séparer la zone sûre dela zone classée comme dangereuse. La barrière de sécurité doit être raccordée conformément auxschémas de contrôle correspondants dans ce document et aux instructions de montage du manuelde mise en service du photomètre de process.
• Les barrières de sécurité agréées et le transmetteur CM44P ne doivent pas être installés dans unenvironnement dangereux.
• La lampe antidéflagrante EXP-1 doit refroidir avant que le boîtier ne puisse être ouvert. Après
coupure de l'alimentation électrique, attendre 5 minutes avant d'ouvrir le boîtier du photomètre.
• La lampe antidéflagrante EXP-1 doit être installée de telle sorte que la fenêtre avant est protégéecontre les dommages mécaniques.
• La lampe antidéflagrante EXP-1 a une extrémité de câble qui doit être raccordée en permanence à
un circuit externe conformément au manuel de mise en service et aux instructions de montage dumanuel de mise en service du photomètre de process.

21

XA01403C

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10• Le câble de mesure CUK80 doit être protégé contre le chargement électrostatique s'il traverse lazone ATEX 0.
• Les câbles qui traversent des environnements explosibles (Class I, Division 1/2) ont besoin d'un
conduit de câble.
• Les chambres de passage OUA260/CUA261 ne peuvent être utilisées en zone explosible qu'avec
des adaptateurs en inox.

Classe de température

Photomètre Température de process

Température ambiante Classe de température

OUSAF12-20 à 40 °C (0 à 100 °F)

OUSAF210 à 90 °C (32 °F à 190 °F) (fonctionnement
continu)Max. 130 °C (266 °F), pendant max. 2 h

T5 ≤ 100 °C (≤ 212 °F)

OUSAF22
OUSAF44
OUSAF46
OUSTF10Si les températures de process spécifiées sont respectées, les températures qui ne sont pas autoriséespour la classe de température correspondante ne se produiront pas sur l'appareil.

Identification

Code de typeType

VersionOUSAF12OUSAF21
OUSAF22OUSTF10-

*
*
*
*

OUSAF12OUSAF21
OUSAF22OUSTF10-

*
*
*
*

Sonde, pas relatif à
Ex

*
*
*
*

A/B/C/D
C/D
C/D
B

*
*
*
*

VersionOUSAF44-*

OUSAF46-

*
*
*
*

2
2
2

Pour utilisation en zone explosible,
ATEX II 2G Ex db IIC T5 Gb*

B

Sonde, pas relatif àEx*Longueur d'onde, étalonnage,pas relatif à Ex

Pour utilisation en zone explosible, FM Cl.1, Sonde, pas relatif àDiv. 1, Groups B, C, D
Ex

*

C

*

Longueur d'onde, étalonnage,pas relatif à Ex

Pour utilisation en zone explosible,
ATEX II 2G Ex db IIC T5 Gb

Sonde, pas relatif à
Ex

*

B

*

*

*

Longueur d'onde, étalonnage,pas relatif à Ex

Pour utilisation en zone explosible, FM Cl.1, Sonde, pas relatif àDiv. 1, Groups B, C, D
Ex

*

C

*

Pour utilisation en zone explosible,
ATEX II 2G Ex db IIC T5 Gb

Sonde, pas relatif àEx*Longueur d'onde, étalonnage,pas relatif à Ex

22

*
*
*
*

Pour utilisation en zone explosible,
FM Cl.1, Div. 1, Groups B, C, D

Type

OUSAF46-1
1
1Longueur d'onde,étalonnage, pas relatif àExLongueur d'onde,étalonnage, pas relatif à
Ex

OUSAF44-

A/B/C/D
C/D
C/D
B

Endress+Hauser

OUSAF12, OUSAF21, OUSAF22, OUSAF44, OUSAF46, OUSTF10

XA01403C

Plaque signalétique
Sur la plaque signalétique, vous trouverez les informations suivantes relatives à l'appareil :• Identification du fabricant• Référence de commande• Référence de commande étendue• Numéro de série• Consignes de sécurité et avertissements• Marquage Ex sur les versions pour zones explosibles‣ Comparez les indications de la plaque signalétique à votre commande.
Certificats et agréments
Marquage Le système satisfait aux exigences des normes européennes harmonisées. Il est ainsi conforme aux
prescriptions légales des directives UE. Par l'apposition du marquage , le fabricant certifie que leproduit a passé les tests avec succès les différents contrôles.
Agréments Ex• ATEX II 2G Ex db IIC T5 Gb• FM Cl.1, Div. 1, Groups B, C, D
Autorité désignée
Sira Certification Service
Rake Lane, Ecclestone, Chester, CH4 9JN, Angleterre

Endress+Hauser

23

*71391367*
71391367

www.addresses.endress.com