Torque Gauge

Digital Torque Gauge

BTGE
PAT. PEND

Digital torque gauge series for very small torque range
2cN·m - 200cN·m

Handheld battery-operated torque gauge combined with flip-up
digital display
Ideal for measurement, inspection and tightening for very small
torqueCan be used for checking accuracy of torque screwdrivers
Read the peak torque both in clockwise and counterclockwise
directions
999 data memory storage with USB output
7 models cover torque capacities from 2cN·m to 200cN·m

Digital Torque Gauge

BTGE

External View

Digital display details

Body (with display opened)
Three-jaw chuck Chuck holder Fixing groove

Counter

Battery remaining

Unit

Battery cover

Liquid crystal display

External output terminal Reset switchMeasuring
Direction

TorqueAuto memory
ONFeatures

Compact & Versatile design

Photo 1

Handheld body with built in flip-up digital display that can be adjusted for
optimal reading and accommodates various testing positions.
Turn the work.

Three-jaw chuck equipped to clamp the object firmly.
Physical stopper prevents accidental over-loading.
Wide torque range
7 models available covering a torque range of 2cN·m to 200cN·m.
Capture the peak torque in both directions. (clockwise and counterclockwise)

Fix the torque gauge.

Variety of uses
Ideal for measuring very small torque of precision machinery, electronic
devices and various assembly components.

Photo 2Use either by fixing the body (see Photo 1), or by turning the body (seePhoto 2) for optimal measurement method.

Turn the torque gauge.

Torque screwdriver checker
Can be used for checking accuracy of torque screwdrivers. (See Photo 3)
Screwdriver bit is secured in three fingered chuck, with display flipped out foreasy viewing.

Fix the work.

Memory & Data transfer
Measured data storage up to 999 readings, with statistical mode displaying:sample quantity, maximum value, minimum value, and mean value.
USB cable can be used for data transfer to PC. (sold separately)

Photo 3Torque screwdriver

Battery-operated
Coin-type lithium battery (CR2450) allows 10-hour continuous operation.
Expanded 50-hours of continuous operation using optional battery pack.
(sold separately)International use
Applicable for international use including the EU region. CE mark applied.

While holding thetorque gauge
fixed, measurethe torque.

Torque Gauge

ModelsCAPACITYMETRIC[cN·m]

S.I. MODELCAPACITY[kgf·cm]

MODEL
MIN. to MAX. GRAD.
cN·m

cN·m

BTGE10CN

2 to 100.01

BTGE20CN

4 to 20

BTGE50CN

DIMENSION [mm]

CAPACITYAMERICANWEIGHT[lbf·in]

TOTAL

OUTER

MIN. to MAX. GRAD.

LENGTH

DIAMETERCHUCKDIAMETER
RANGE

13075

ø1 to ø8.5

MODEL
MIN. to MAX. GRAD.
kgf·cmkgf·cmlbf·inlbf·in

BTGE1KM

0.2 to 1

0.001

BTGE09I

0.2 to 0.9

0.001

0.02

BTGE2KM

0.4 to 20.002

BTGE1.8I

0.4 to 1.80.002

10 to 500.05

BTGE5KM

1 to 50.005

BTGE4.4I

1 to 4.40.005

BTGE100CN

20 to 100

0.1

BTGE10KM

2 to 100.01

BTGE9I

2 to 90.01

BTGE200CN

40 to 200

0.2

BTGE20KM

4 to 20

0.02

BTGE18I

4 to 18

0.02

OUTER
MAIN BODY DIAMETER OFHOLDER

72

ø66

Standard Accessories: Coin-type lithium battery (built-in the body), Carrying caseCommon Specifications
Measurement direction

Bi-directional

Display

7-segment LCD display

Measurement mode

PEAK/RUN

MemoryMax. 999 readings

Statistical processing

Sample q'ty, max. value, min. value, mean value

Data output

USB output (USB mini B connector)

Power source

Coin-type lithium battery (CR2450)

Continuous use

10 hours with coin-type battery, 50 hours with battery pack
(BP-C1)Other functions

Auto power off (3 min). Auto memory reset (0.5-5sec), Autozero adjustment, residual battery indicator (4 steps)

Operative temperature range

0 to 40°C (no condensation)

Display adjustable range

0 to 163 degree

Optional Accessories

USB Connecting Cable

Battery Pack

(Part No. 584)

(Part No. BP-C1)
Appearance(Cord length: 1.8m)(Cord length: 1m)

Inside[kg]0.65Dimensions

Fixing groove

Approx. 130
ø66
Chuck operating diameterø1 to ø9

Maximum turning
position of the display

ø78

Display73°
Three-jaw chuck90°

Chuck holder
3Dodecagon width
across flats 75

Body 72
Overall length Approx. 130

Alternative Models

BTG (-S) Analog version of torque gauge
S.I. MODELCAPACITY[cN·m]WITH
MEMORYPOINTERMETRIC

MIN. to MAX. GRAD.

BTG15CN

BTG15CN-S

2 to 15

BTG24CN

BTG24CN-S

3 to 24

BTG36CN

BTG36CN-S

4 to 36

BTG60CN

BTG60CN-S

6 to 60

BTG90CN

BTG90CN-S

10 to 90

BTG150CN BTG150CN-S20 to 150

0.20.05

12MODELCAPACITY[gf·cm/kgf·cm]

AMERICAN

MIN. to MAX. GRAD.

1.5BTG-S

kgf·cm
0.2 to 1.5

2.4BTG-S

0.3 to 2.4

3.6BTG-S

0.4 to 3.66BTG-S

0.6 to 6

9BTG-S

1 to 915BTG-S

2 to 15MODEL

CAPACITY[ozf·in/lbf·in]
MIN. to MAX.

kgf·cm
0.02 1.5BTG-A-S

lbf·in0.1 to 1.5

2.4BTG-A-S

0.3 to 2.4

3.6BTG-A-S

0.4 to 3.6

6BTG-A-S

0.6 to 6

0.05

0.10.29BTG-A-S

1 to 915BTG-A-S

2 to 15

WEIGHT[kg]

0.52

Complimentary Products

TGTCL Calibration Kit for BTG/BTGE models
MODEL

COMPONENTAPPLICABLE MODELBTGTCL150CN

Main unit, Calibration Pulley x 2
Calibration Wire x 3
Scale Pan (5g and 100g)

BTG15CN, BTG24CN, BTG36CN,BTG60CN, BTG90CN, BTG150CN, BTGE

08.06.D