Handpiece Kit
Accessories
Handpiece GripTip

Chip andchip coveris optional．

Spiral Protection TubeTipcover

HP IRR Cover
HP Angle Coverby Advanced Technologies

Sterilization torque wrench

Foot Switch

Sterilization spanner
Sterilization cap

SC Cart

Filter Tube set

Stelirization Case

Cleaning Wire

Disposable Tubing Set

Tip Lineup

A New Approach to Veterinary Medicine Enabled

Disposable Canister

For Soft tissue
Soft Tissue Tip

P/N: CS1

Large Soft Tissue Tip

P/N: CS2

Spatula Tip

P/N: CS3

Dia.

2.9mm

Max.
width5mm

Length 114.5mm

Length112mm

Length101mm

Vibrationlongitudinal
type
SuctionYES
port

Vibration
longitudinal
typeSuction
YES
port

Vibration
longitudinal
type
SuctionYES
port

Cholecystectomy, Synechiotomy, Tumor removal...

Cholecystectomy, Synechiotomy, Tumor removal...

Cholecystectomy, Synechiotomy, Tumor removal...

Dia.

1.9mm

For Hard tissue
Bone Tip

P/N: CH1
Dia.

2mm

Length119mm

Vibration
type
Suctionport

Large Bone Tip

Dia.

3.5mm

Length116mm

Vibration
type
Suction
port

LTYES

Hernia laminectomy, Nasal bone removal...

P/N: CH2Saw Tip

Dia.

0.8mm

Length 114.5mm
Vibration
typeSuctionportLTYES

Hernia laminectomy, Nasal bone removal...

P/N: CH3LTNO

Craniotomy, bone plasty...

■Specifications- Ultrasonic output power: Method for oscillation/ electrostrictive vibration, rated oscillatory frequency/ 25 kHz, rated output/ 100 W- Irrigation (water supply): Rated flow rate/ 40 ml/min- Suction (aspiration): Rated pressure/ 65 kPa- Electrical rating: Power supply voltage/ 100 V AC, power supply frequency/ 50/60 Hz, power supply input/ 300 VA- Dimension and weight: Main unit/ W 280 x D 440 x H 195 mm, 13 kg

Tokyo Iken Co., Ltd.

Head Oﬃce：1131-1 Higashi Naganuma, Inagi, Tokyo 206-0802 Japan Phone ＋81-42-378-6630 FAX ＋81-42-378-6614
Sales Department：
2-27-2 Yushima Bunkyou-Ku, Tokyo 113-0034 Japan Phone ＋81-3-5807-3011 FAX ＋81-3-5807-3012

veterinary ultrasonic
surgical device

Combination of the Most Advanced
Technologies Makes Tough Surgery Saferand Less Invasive

Feature1

Sono Cure has a vibrating metal tip in the ultrasonic range and isused in surgeries to crush, emulsify, and aspirate the target site, aswell as to cut bone. During surgeries, elastic tissues such as bloodvessels or nerves can be conserved selectively, and consequently thedamage to affected animals can be minimized.

With the increased aging of pets and changes in living conditions, animal diseases
have become more varied. Surgery for animals requires expert technologies evenmore.

Minimally invasive surgeries can be performed
for crushing, emulsification, and aspiration oftissues, and bone cutting Debut as a device
exclusively for use on animals in Japan

Sono Cure , the first ultrasonic surgical device exclusively for

veterinary use in Japan, is a new approach in the field of animal surgery.
Since not only crushing, emulsification, and aspiration of tissues but
also highly developed technologies including bone cutting can be

Featureperformed safely and in a minimally invasive way, this substantially

A compact and light handpiece with high crushingstrength has been adopted. Variation of six different
tips for use in soft tissue, bone and hard tissue
The selection of tips suitable for symptoms and/or affected sites
enables efficient crushing, emulsification, and aspiration.

mitigates the intraoperative stress of veterinarians and thephysical damage of affected animals.

For Soft tissue●SoftTissue Tip

Feature●Large SoftTissue Tip●Spatula Tip

For Hard tissue●Bone Tip

●LargeBone Tip●Saw Tip

The tips for bone, which have no rotational moment,
do not damage adjacent tissues.
The tips for bone are useful for bone cutting in the vicinity of nerves,
blood vessels or dura mater. Addition of torsional movement tolongitudinal vibration increases the crushing power and achieves
better operability during cutting.
Longitudinal
vibration

LT vibration

Torsionalvibration■LT（Longitudinal-Torsional)Vibration chip
Image view

Excellent operability as well as sophisticated design
Featureveterinary ultrasonic
surgical device
FeatureAll of the necessar y set tings are
implemented on the front panel.
In t r o d u c i n g t h e f o o t s w i t c h, t h e
h a n d p i e c e portion operates quickly
without a time lag. The main body has
a compact design, and the SC wagon
accommodates the main body andaccessories while saving space.

Water supply, ultrasonic crushing and aspiration areincorporated into one system.
Sono Cure is equipped with all the driving functions forsaline supply, ultrasonic power and aspiration in the main body.
It is an all-in-one system that requires no auxiliary devices.