page 1 of 2

CS65 SHUTTERSPECIFICATIONSFEATURES•••
•
••
•
••••

Small form factor, a 65 mm aperture fits into a 5-inch
diameter housing!
Multi-bladed design in combination with the UNIBLITZ
actuator provide increased speed over other designsof this type.
State of the art damping system for increased reliabilityand speed.
Minimum exposure time exceeds that of the VS65!
Reflective blades available as with Vincent LS, VS andCS45 series.
Can be driven with our existing VMM drive units.
Special driver not required.
Available housed or un-housed for OEM applications.
#102 Mounting Ring available for universal mounting
applications.
Electronic Synchronization System option available.
Exposure repetition rates from DC to 2 Hz.
Design, accuracy, and reliability that you have come toknow as with all other UNIBLITZ products!

The CS65 is the newest patent-pending product andis the next release in the new UNIBLITZ® CS series.
The small form factor allows a 65 mm aperture to beinstalled into applications not presently accessible
with existing UNIBLITZ VS series shutters. As with theCS45, the CS65 has been designed to provide accurate,
repeatable exposures for a wide variety of applications.
The small form factor allows a 65 mm aperture to beinstalled into a five-inch diameter housing – about the same
overall size as the existing VS35 shutter! To increase theunit’s flexibility, the shutter can be supplied in an unhoused version for OEM applications or in situations
inaccessible to most shutters due to spatial limitations.
The CS65, along with its state-of-the-art damping system,
provides increased reliability over other designs of its type.
In addition, the system provides the reliable actuatorsystem found in all other UNIBLITZ shutter designs
allowing it to be driven with our existing VMM drive units.
As an option, the shutter can be equipped the electronicsynchronization system.
When gating high intensity light sources, the CS65 can beequipped with polished stainless steel reflective blades.
This option protects the shutter blades from the lightsource’s damaging effects by reflecting the energy awayfrom the blade surface. Look for future availability of “Z”(AlSiO) and “ZM” (AlMgF2) coated blades in this device.
To further enhance the flexibility of the CS65; an optional#102 mounting ring is available to allow the shutter to beeasily mounted in many non-specific applications.
Additional information regarding the #102 mounting ringcan be found in the specific data sheet entitled“MICROSCOPE & VIDEO MOUNTING SYSTEMS” or online under “products”, “Mounting Systems”.

ELECTRICAL
Coil Res.:

12 ohms

Pulse Voltage to Open

+70VDC

Hold Volt.:+7VDC/+5VDC*(*VMM dual hold level driver system included in UNIBLITZ VMM shutter controller)

MECHANICAL
Wgt. Un-Cased

4.0 oz (0.112 kg)

Wgt. Cased

13.2 oz (0.374 kg)

Operating Temp.

0°C to +80°C

Max. Frequency of Operat.

2Hz

Max. Opening Bounce

15%

Max. Closing Bounce

5%

Number of Blades

6TIMING

Typical timing values (msec.) using UNIBLITZ drive equipment andmeasured with UNIBLITZ shutters equipped with standard TEFLON®
coated shutter blades.

TIMING OF PULSE INPUT AND SYNCHRONIZATIONOUTPUT RELATIVE TO SHUTTER STATETYPICAL PULSEINPUT TOCONTROLLERSHUTTERSTATE
TYPICAL
ELECTRONICSYNCHRONIZATIONOUTPUT FROMCONTROLLER(Timing in msec.)
6.0
29.0
35.0
78.0
18.0
55.0
107.0

O-A Delay time on opening after current is applied
A-C Transfer time on opening
O-C Total opening time
B-F Min. equivalent exp. time
C-E Min. dwell time with min. input pulse
E-G Transfer time on closing
A-G Total window time
MET: Min. exposure time
TEP: Typical exposure pulse

40.0
>40.0

The question regarding enhancement of shutter speed with the application of user
supplied lubricants has been repeatedly asked. It is our experience that lubricating
the shutter blades will actually slow the shutter down and eventually render theshutter inoperable. UNDER NO CIRCUMSTANCES SHOULD ANY TYPE OF
LUBRICANT BE APPLIED TO THE SHUTTER BLADE AREA.

HOW TO ORDER

CS65S
APERTURE SIZE

CS65S - 65mm

HOUSING3TBLADE FINISH0-102

SOLID STATESYNCHRONIZATION

1 - UNCASEDT - TEFLON COATED 0 - OMIT SYNC.

3- #3 CASE

S - POLISHEDSTAINLESSZM - AlMgF2
COATED BeCu
BLADESZ - AlSiO
COATED BeCu
BLADES

1 - SOLID STATESYNC.

MOUNTING
OPTIONS- 102 MOUNTING
RING

page 2 of 2

CS65 SHUTTERSPECIFICATIONSUN-CASED STYLEFigure 1Ø[123]
Ø4.86[15.7].62

4.5°
TERNINATES TO
7-PIN MALE IN
LINE CONNECTOR

C'SK FOR 0-80
FLAT HEAD SCREWON Ø4.725 B. C.
120° APART (3)[65.0]
Ø2.56

CLEARANCE HOLES
FOR 2-56 SCREWON Ø4.725 B. C.
120° APART (3)

[1.8]
.07

4.5°

[7.6]
.30

ALL DIMENSIONS MAX.
[MM]IN

CASED STYLE

[1.0]
.04

Figure 2[27.3]
1.08
SHUTTER
PLATE[Ø127.0]
Ø5.00

SEE NOTES
1 AND 2[17.0]
.67

[7.4]
.29
[Ø76.2]
Ø3.00
x40 THD[Ø65.0]Ø2.56#3 CASE
CS65

[6.4]
.25

SHUTTER REFLECTIVE
SURFACE THIS SIDE
[6.4]
x 40 THD.
Ø.25

[3.2]
.13

NOTES:
1. (3) 2-56 MOUNTING
HOLES ON A Ø2.938
BOLT CIRCLE 120°APART (UNDER NAME
PLATE)
2. (3) 4-40 MOUNTING
HOLES ON A Ø3.500
BOLT CIRCLE 120°APART (UNDER NAME
PLATE)
3. ALL DIMENSIONS MAX.
[MM]
INCH

[0.8]
.03

[Ø9.5]
Ø.38
[8.7]
.34

HOUSING/CONNECTOR LAYOUT

COVER

CS65 CASE
OUTLINE(3) 4-40 THD x.150
DEEP ON A Ø3.500
B.C. 120° APART(UNDER NAME PLATE)
45°

SEAM

The CS65 #3 housing style allows a number ofmounting configurations. A ¼-20 threaded holeis provided for post mounting. The 3.00inch x
40TPI external thread located on the rear side,and the specific mounting holes located on thefront side (see Figure #2 and Figure #3) can beinterfaced directly into your application or fittedwith a variety of user specific mounting options.
For the CS65, the #102 Mounting Ring is presently the only mounting option available. Additional information regarding the #102 mounting
ring can be found in the specific data sheet entitled “MICROSCOPE & VIDEO MOUNTING SYSTEMS” or on-line under “products”, “MountingSystems”. The unit terminates with a 7-pin maleconnector as illustrated.

Figure 3127
5.00

CASE

The CS65 un-housed style is the basic configuration of this device and is best suited for OEM applications. Mounting can be accomplishedthrough three 0-80 (flat head required) clearance
holes located around the unit’s perimeter on a4.725 inch diameter bolt circle. These holes, asindicated, are 120 degrees apart. Unless otherwise specified, this standard unit is terminated toa 7-pin male connector through a 7-wire six-inch
cable assembly. (The three 2-56 holes are notrecommended for mounting due to the potential
for interference with the shutter’s blades.)This drawing illustrates 7-pin connector and ¼-20
threaded hole layout for the CS65 series #3
housed style.

CLEARANCEAPERTURE[65.0]
2.56

[8.3]
.33

(3) 2-56 THD x.150
DEEP ON A Ø2.938
B.C. 120° APART(UNDER NAME PLATE)

VIEW A-A

COVER
CASE
SEAM

10.8.42
VIEW B

A
1/4-20 THD
POST MOUNTING
L OCATIONNOTES:
1. DRAWING SHOWINGCONNECTOR AND
1/4-20 POSTMOUNTING LOCATION

24°A
32.61.28

B

2. FINISH BLACK ANODIZE
3. TERMINATES TO710P OR 710C
CABLE ASSEMBLY
4. ALL DIMENSION MAXIMUM

WHEN FEMALE CONNECTORIS ATTACHED, MAXINUMDISTANCE FROM CASE WILL BE
1.375 (34.9) AS SHOWN[MM]INCH
19.4.76

Due to our ongoing product development program, Vincent Associates reserves the right to discontinue or change specifications or designs at any time, without incurring anyobligations. Teflon is a registered trade mark of E.I. Dupont U. S. Pat. No. 3.427.576; 3.595.553; 3.967.293. Drawing shown for illustrative purposes only.
Updated 05/2003.