Tension/ Compression Load Cell

TU - MXR2(T) n n N - G3

Features
Body
Material

Compact &Small Capacity
AluminumTension/ Compression Load Cell

DurableRobot Cable
standardized

Plug & Playwith
built-in TEDS

Enhanced durability against bendingthat occurs in moving parts withfrequent repetitive motion, such asindustrial robots and machine tools.
High stability and reliability are realized.

With the TD series indicators,
equivalent input calibration,
likely to forget in manual setting,can be performed automaticallyand help prevetion.

Mounting
MethodApplications
Load measurements for Robotand Test equipment

M3/M4 screws to mountTension/Compression loadcertified※Product image for illustration purposes only. Actual product may varySpecifications
TypeModelTension/Compression Load Cell

TU-MXR2(T) n n N - G3

(Embeded in the tip of the cable)

Rated Capacity (R.C.)

10N

20N

50N

100N

200N

500N

Natural Frequency

2.2kHz3.0kHz5.2kHz8.0kHz6.6kHz(T.B.A.)

Weight (Approx.)

9g

69g

10g

10g

21g

24 g

Safe overload rating

120％ R.C.

Rated Output (R.O.)

Approx. 1.5mV/V ±30%

Linearity0.1％ R.O.

Hysterisis

0.1％ R.O.

Repeatability

0.1％ R.O.

Safe Excitation Voltage

8V DC / ACInput Terminal Resistance350Ω ±2%Output Terminal Resistance
Insulation Resistance350Ω ±2%1000ＭΩ or more (50V DC)

Compensated Temperature Range–10 to 45℃

Permissible Temperature Range–20 to 60℃

Temperature Effect on Zero Balance

0.5% R.O. /10℃

Temperature Effect on Output

0.5％ R.C. / 10℃

Cable
Mounting Method
Body Material(See the reverse page for detail on TEDS)

Main unit to built-in TEDS section: Φ2, 4-core shielded cable,
Built-in TEDS section to the end: Φ3, 6-core shielded cable, approx. 15cm,Total 3m, direct connection robot cable with bare lead wires
M3 Screw holeAluminum

M4 Screwhole

Dimensional drawings (Units: mm)(10 substances)

Advantages of the TEAC Load Cells

TEAC Load Cells

Robot Cable standardized

TEDS-compatible

Since the 1980s, when TEAC started manufacturing
and selling load cells, we have cultivated technologies
to achieve higher precision and smaller size with ourunique structures. With these technologies, a numberof load cells that achieve high response, high accuracy,
and high stability, as well as products that takeenvironmental conservation into consideration havebeen developed to match customers’ applications.

Robot cables provide enhanced durability andstable performance against bending that occurs inmoving parts with frequent repetitive motion, suchas industrial robots and machine tools.

The TEDS (Transducer Electronic Data Sheet)system is a generic term for a description format
standardized by IEEE that electronically reads andwrites sensor’s specific characteristic, which isrecorded in an EEPROM built into the sensor and canbe read and written electronically.

Every TEAC’s ultra-compact load cells employ
robot cables, together with the TEDS function,
contribute to factory automation and labor savings.
* Customized proposals that match your applicationand environment are available. Please contact oursales representatives for detail.

We also offer customization for specific conditions(usage environment, space) that are difficult to meetwith standard ones. From one-off prototypes to mass
production, we support engineers involved in researchand development on manufacturing technology.
Examples of appplication
Pressuer of mounters

Tension of wire saw

Dicing wafers

Polishing wafers

Model name, serial number, sensitivity (output valueagainst physical quantity) and other calibration
factors are digitized and recorded in the memory
built into the load cell body. Sensor’s specific values
can be set electronically, automating the reading ofrecorded information and equivalent input calibration,
eliminating human error in setting and reducing theburden of load cell replacement.
Load Cell Embedded
Equipment

TEDS-compatible
digital indicators

Sending individual specific values of each loadcell indicated in the unit’s Data SheetAs shown above, fix the core wire so that it does not move,
bend it 90 degrees to the left or right, and confirm that nowire breakage occurs.

TEAC has been strongly promoting TEDS (IEEE 1451.4
Transducer Electronic Data Sheet) compliance for loadcells and load cell indicators.We are the first Japanesemanufacturer that obtained a “Manufacturer ID”, making
our load cells and indicators TEDS-compatible.

Related Products (Indicators and Signal Conditioners)

NewNew

EtherNet/IP™
model

CC-Linkmodel

92 x 92mm
Panel opening size

Color Graphics Digital Indicator

Digital Indicator

RS-485 model
EtherNet/IP™ model
CC-Link modelStandard model
CC-Link model
RS-485 model

TD-9000T

Attaches tocommon DIN rails

92 x 45 mm
Panel opening size

TD-700T

Weighs only 320g
(incl. batteries)

Portable Digital Indicator

Signal Conditioner

TD- 01 Portable

TD-SC1
D/A model
RS-485 modelHigh performance modelwith large LCD

Excellent model with compact
and high functionality

Slim and light-weightsignal conditioner

On-site checking toolwith versatility

Supporting two inputs, force sensor
and displacement sensor, various
comparison judgments function,

Supporting five key functions in one unit,
numeric display, graph display, TEDS function,
static strain display, and signal conditioner.
This small and cost-effective TD-700T
achieves equal or even higher performance toupper-class models, with high-visibility color
LCD and various hold functions.

Supporting high-speed samplingof 20,000 times/second, PC-based
configuration via USB connection,
selectable network, and TEDS
calibration function.

Supporting various functions
that equal to embeded systems,
in hand-held size, allowing youto take measurements anytime
anywhere, according to yourpurpose.

and direct saving of waveform
data onto large capacity internalmemory.

***Under planning

EtherNet/IP is a trademark of ODVA, Inc. Other company names, product names and logos in this document are the trademarks or registered trademarks of their respective holders.

TEAC CORPORATIONTEAC America, Inc.,

1-47 Ochiai, Tama-shi, Tokyo
206-8530, JapanTEAC EUROPE GmbH.

E-mail: cs_ipd@teac.jp
Web: https://loadcell.jp/en/

E-mail: datarecorder@teac.com
E-mail: info@teac.eu

TEAC SALES & TRADING (ShenZhen) CO., LTD.
E-mail: teacservice3@teac.com.cn© Copyright TEAC CORPORATION 2022 2206TCJ-PDF / ISD-202