INSTRUCTION MANUALPRECISION
MEASURING TOOLSSCRIBING CALIPERS

TVC-15S

Model No.

Thank you for purchasing the SCRIBING CALIPERS.
This product is a precision measurement tool with a main scale and a vernier scale to indicate the measured value.
Carbide tipped measuring surfaces for scribing steel and hardened surfaces.
●For safe and proper use of this product, please read this instruction manual before use and follow the procedures described. Pleasekeep manual where it is accessible to user for future reference.
●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact distributor or place of purchase.

■PART IDENTIFICATION AND FUNCTION⑤Measuring surfaces for internal measurements⑥Lock Screw④Inside Jaws
③Measuring surfaces forstep measurements

⾿⿀

⿁㇎

⿂㇏

㇐㇑

⿄⿃㇒

㇓㇔

㇕

㇖

②Outside Jaws
①Measuring surfaces foroutside measurements

⿅㇗⿆

⑦Main Scale

⿇⿈

⿀
⾿

⑧Depth Bar⿀⿀㇎

⑩Thumb Knob

⑨Measuring surfaces for
depth measurements⑪Vernier Slide (Vernier Scale)①Measuring Surfaces for outside measurements … Object to be measured is positioned between these surfaces.
②Outside Jaws ………………………………………… Measures extremal dimensions. Move Slide to open and close.
③Measuring surfaces for step measurements …… Measures the height of steps.
④Inside Jaws…………………………………………… Measures internal dimensions. Move Slide to open and close.
⑤Measuring surfaces for internal measurements … Position surfaces inside object to be measured.
⑥Lock Screw …………………………………………… Locks the Vernier Slide position.
⑦Main Scale …………………………………………… Indicates measured value in 1mm graduations.
⑧Depth Bar …………………………………………… Bar for measuring depth. Bar moves in and out with Vernier Slide.
⑨Measuring surfaces for depth measurements …… Depth is measured between these two surfaces.
⑩Thumb Knob ………………………………………… Position thumb on knob for precise control of vernier slide.
⑪Vernier Slide (Vernier Scale) ……………………… Scale for measuring with a resolution of 0.05mm.
Refer to following page for usage instructions.

SAFETY PRECAUTIONS

Please Observe

Always follow the proper operating procedures in order to prevent
harm to yourself or others, and to prevent damage to property.

PREPARATION FOR USE・Use of product other than as described in the manual
may cause accident.

Use only for measuring.

●Do not apply excessive force to Vernier Slide.
Pushing against slide, or attempting to move without pressing
the Thumb Knob may cause damage or inaccuracy.

Denotes a prohibition-You MUST NOT do

Denotes a requirement-You MUST do・Please do not use instrument in rain or locations withexcessive humidity. Do not use at extreme high orlow temperatures. Keep out of direct sunlight.

Handle with Care.

・Excessive force applied to Slide may cause damage
or affect accuracy.

CALIBRATION

Wear of measuring surfaces from repeated use may affect accuracy andperiodic accuracy checks should be performed.

●EXAMPLE１

Contamination of surfaces may cause measurement error.

●Allow the caliper and object to be measured time toreach the same temperature.

㇎⿀㇏㇐

㇑

⿁㇒

㇓

⿂㇔

㇕

㇖

⿃㇗

・Careless handling may cause injury.

Do not move the Vernier slide without pressing
Thumb Knob.
Do not disassemble or modify.

・Do not remove the vernier slide, Please do not
attempt to disassemble or modify as it may cause
damage or poor accuracy.

We provide calibration services.
Please contact agent in country ofpurchase to make arrangements.

Reading73＋ 0.0 0
7 3. 0 0

Vernier Scale(0.05mm)●Performing an accuracy check.
①Jaw Alignment
In closed position, make sure the measuring surfaces on theoutside jaws align properly and there is no gap (no lightshould leak past.)②Confirm “0” points on scale (see diagram at below)
With jaws closed as in step ① above, make sure the readingis zero and the vernier scale graduations align at the “0” and“39” positions as shown.
③Depth Confirmation
Position caliper on a flat surface for depth measurement.
Reading should be “0” as in step ② above, with the vernier
graduations aligned at the “0” and “39” positions.

⾿Measured Value: 73.00㎜

Main Scale(1mm)

A temperature difference between the object to be measured
and the caliper may cause measurement error. Allow enoughtime for the temperatures to equilibrate.

⿄㇎Proper alignment

HOW TO USE①Loosen the lock screw, and open the Jaw the required amount.
②Lock the position by tightening the Lock Screw.
③Hold the Jaws as shown in the diagram below and scribe theworkpiece.

●EXAMPLE２ Measured Value: 73.50㎜
Main Scale(1mm)

Reading
73＋ 0.5 07 3. 5 0

Vernier Scale(0.05mm)●EXAMPLE３ Measured Value: 73.55㎜
Main Scale(1mm)

Reading
73＋ 0.5 5
7 3. 5 5

Vernier Scale(0.05mm)PREVENTING ERRORS
⿆to prevent⿇⿈⿀⾿⿀⿀ please note the following.
⿅In order
measurement
error,POSITIONING CALIPER ON OBJECT＜Flat surfaces＞
When measuring featureswith flat surfaces, make surethe measuring surface of thejaws are parallel to thesurfaces to be measured.
＜Cylindrical surfaces＞
When measuring cylindricalfeatures, make sure thecaliper jaws areperpendicular to the cylinder
axis as shown.

Jaw

Object being measured

VIEWING THE SCALE⾿

Scale should be read fromdirectly above the calipers.
Due to step in height ofvernier scale relative to main
scale, if viewing direction is
not directly above the reading
may have parallax error.

Use in proper environment.

Jaw Tips are sharp - Handle with care.

●In order to maintain instrument accuracy, it is recommended thataccuracy is confirmed through calibration on a periodic basis.

Measured value = Main Scale Reading + Vernier Scale Reading●Wipe off any corrosion, dirt or oil from measuring
surfaces and slide.

●Scribing・Use for any purpose other than measuring may
damage or wear the instrument. Improper use may
also cause accident.
・Do not drop or subject instrument to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage or poor accuracy.
・Do not scratch instrument, for example by writing IDnumber.

The measured value is determined by reading the main scale,and then adding the Vernier Scale reading as determined by
the position where the vernier and main scale graduations align.

Before using the caliper, follow the steps below:

WARNING Indicates risk of personal injury or property damage if not followed.
Read the manual and follow the instructions.

READING THE SCALE

beriScWorkpiece

AFTER USE CARE, STORAGE NOTES●As required, wipe slide and measuring surfaces with a dry cloth
treated with anti-corrosive oil. When not in use, apply
anti-corrosive treatment or keep in anti-corrosive bag.
●When not in use, keep jaws slightly open to prevent thermal
expansion from creating excessive forces on jaws and affecting
accuracy.
●Store in supplied case when not in use in a cool, dry location.

⿀⿁㇎

⿂㇏

㇐㇑

⿄⿃㇒

㇓㇔

㇕

㇖⿅㇗

⿆

⿇㇎

SPECIFICATIONS●Material : Stainless Steel●Measuring Range : 150mm●Resolution : 0.05mm●Maximum permissible error : ±0.05mm●Weight : 150g

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

精密測定工具

取扱説明書 品番 TVC-15S

ケガキノギス

ご使用前に、下記の準備を行ってください。
指かけを押さずにバーニヤを動かすと、無理な力がかかり、精度不良や破損の原因となります。

●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。
⑤内側用測定面⑥止めねじ④内側用ジョウ

③段差測定面⑦本尺（本尺目盛）

⾿⿀

⿁㇎⿂㇏㇐

②外側用ジョウ㇑⿄⿃㇒㇓㇔㇕㇖⿅㇗⿆⿈⿀⾿

⑧デプスバー⿀⿀

⑨深さ測定面

㇎⑩指かけ

●測定例１

●本体と被測定物の温度慣らしを十分に行う。

本尺目盛（1㎜）
副尺目盛（0.05㎜）

そのまま測定を行うと誤差の原因となります。

測定器本体と被測定物の間に温度差があると、誤差の原因となります。同じ温度になるよう、温度慣らしは十分に行ってください。

●使用前精度確認を行う。

①ジョウの確認
外側用ジョウを閉じた状態で外側用ジョウの部分を光に透かして、隙間ががないか（光が漏れていないか）確認してください。
②副尺の０点が合致しているか（下図）①の状態で、副尺目盛の２つの０が、本尺目盛の０と39に合致していることを確認してください。
③デプスの確認
深さ測定面を平らな面に垂直に当て、②と同様に副尺目盛の２つの０が、本尺目盛の０と39に合致していることを確認してください。

⾿①外側用測定面 …… 被測定物をこの面で挟んで測定します。
②外側用ジョウ …… 被測定物の外側を測ります。バーニヤをスライドさせると開閉します。
③段差測定面 ……… 段差面に当てて長さを測定します。

④内側用ジョウ …… 被測定物の内側を測ります。バーニヤをスライドさせると開閉します。
⑤内側用測定面 …… 被測定物の内側にこの面を当てて測定します。

⑥止めねじ …………

⿀

⿁⑦本尺（本尺目盛）…… １ｍｍ単位を測定する目盛です。

㇎㇏

㇐

㇑

㇒

㇓

㇔

⑨深さ測定面 ………… 測定面に当てて深さを測定します。

⑪バーニヤ …………… 0.05ｍｍ単位を測定する目盛です。目盛の読み方は、右記を参照してください。
（副尺目盛）

してはいけない内容『禁止事項』です。

校正について●測定精度を保つため、一定周期で校正による精度確認をおすすめします。

使用を繰り返す間に測定面の磨耗などにより、精度を保てなくなる可能性があります。定期的な精度確認の実施をおすすめします。

㇕㇖㇗㇎

合致していること

使用方法●ケガキ作業

必ず実行していただく
内容『強制事項』です。

注 意 お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

測定作業にのみ使用する・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。
大切に取り扱う・ぶつける、落とすなどの衝撃を与えたり、重量物を上に乗せたりしないでください。精度不良や破損の原因となります。
・番号の書き込みなど、製品に傷をつけないでください。

⿄①止めねじをゆるめ、けがきたい幅にジョウを開きます。
②止めねじを締めて幅を固定してください。
③ジョウを下図のように対象物に当て、けがいてください。

必ずお守りください

取扱説明書をよく読み、指示に従う
・取扱説明書に記載された内容以外での使用は、事故の原因となります。

⿃

使用環境に注意する
・雨の中や湿気の多い場所、高温・低温環境下、直射日光の当たる場所での使用はおやめください。
ジョウ先端は尖っているので、扱いに注意する
・けがの原因となります。
指かけを押さずに、バーニヤをスライドさせない。
・無理な力がかかり、精度不良や破損の原因となります。
分解・改造を行わない
・バーニヤを外したり、分解・改造を行わないでください。精度不良や破損の原因となります。

弊社にて校正を受け付けております。全国にある弊社代理店、取扱店を経由してお申し込みください。

お問い合わせ：( 02 56 ) 3 3 - 5502

測定値 73.00㎜
読み
730.0 07 3. 0 0

測定値 73.50㎜●測定例２本尺目盛（1㎜）
副尺目盛（0.05㎜）

読み730.5 07 3. 5 0

測定値 73.55㎜●測定例３本尺目盛（1㎜）
副尺目盛（0.05㎜）
⿅

読み
730.5 57 3. 5 5

誤差の防止
⿆⿇⿈⿀⾿⿀⿀

測定誤差を防ぐため、下記の事項に注意してください。

⑧デプスバー ………… 深さを測定するバーです。バーニヤをスライドさせると一緒に動きます。

お使いになる人や他の人への危害、財産への損害を未然に防止するた
め、必ずお守りいただくことを右記のように説明しています。

⿂

バーニヤのスライドを固定します。

⑩指かけ ……………… ここに指をかけてバーニアをスライドさせます。

測定値＝本尺目盛＋副尺（バーニヤ）目盛●測定面、摺動面、外観などの汚れや防錆油などを拭き取る。

⑪バーニヤ（副尺目盛）①外側用測定面

安全上のご注意

⿇本尺目盛に、本尺目盛と副尺（バーニヤ）目盛が合致したところの副尺目盛をプラスして、測定値を読みます。

●バーニヤに無理な力をかけない。

この度は「ケガキノギス」をお買上げいただきありがとうございます。
この商品は、本尺目盛と副尺（バーニヤ）目盛を使い、測定値を読み取る精密測定工具です。
測定面に超硬チップ付で、一般鋼材・焼入品などへのケガキ作業が行えます。

■各部の名称と機能

目盛の読み方使用前の準備被測定物への当て方

ジョウ＜直方体を測定する場合＞
平行する面に対して垂直にジョウ部分を当てます。
＜円筒体を測定する場合＞
円筒体の軸方向に対して、ノギスが直角になるようにジョウ部分を当てます。

被測定物

目盛の見方
ノギスの目盛のある方向に垂直に読み取ります。垂直でないと本尺と副尺（バーニヤ）の面の高さの段差により、視差ができ誤差が発生します。

⨎⨐⨋

⾿

⿀

⿁㇎

⿂㇏

㇐㇑

⿄

⿃㇒

㇓

㇔

㇕㇖⿅㇗

⿆⿇㇎

対象物

使用後のお手入れ・保管方法●測定面・摺動面・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う。
防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。

●止めねじをゆるめ、ジョウを少し開いた状態で保管する。

ジョウを閉じた状態で止めねじを締めて固定すると、温度変化による膨張などで無理な力がかかり、精度不良の原因となります。

●付属のケースに入れ、乾燥した冷暗所に保管する。

直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

製品仕様●材質：ステンレス●最大測定長：150mm●最小読取値：0.05 mm●最大許容誤差：±0.05mm●本 体 質 量：150g
発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I202-K12204⿈⿀⾿