MMM4011C
Coriolis Mass Flowmeter
Mountable on Fillers
OUTLINE
MMM4011C, a lightweight, compact Coriolis mass flowmeter,
is ideal for mounting on various filling machines for liquids suchas beverages and cooking oil, and for controlling their mass ﬂowrate.
By combining our expertise and customers demands, we havedeveloped a lightweight, compact, highly accurate, and durable
flowmeter. This product has a simple design; it comes without
an indicator, operates on 24 V DC, and the output information ispulse output (open collector) only.
MMM4011C can directly measure the mass flow rate of liquids
regardless of changes in their physical properties (temperature,density, viscosity, conductivity, and others).
With two meter sizes of 10 and 15 mm this ﬂowmeter is ideal forthe process of filling low-conductivity liquids including oil, pure・Materials:

water, and molasses.

FEATURES● Lightweight, compact design; mountable on ﬁllers
189 × 333 × 59.5 mm / 2.8 to 2.9 kg● Measuring mass ﬂow rate
Directly measures the mass ﬂow rate of liquids.
● Measuring low-conductivity liquids
Measures the ﬂow rate of oil, pure water, molasses, and otherParts
Material
Measuring tube (wetSS316Lted part)
Fitting (wetted part)
SS316L
Housing
SS316
・Output
Pulse output (open collector)
Load rating: up to 30 V DC, 20 mA

Pulse rate

: up to 10,000 Hz/F.S.

Pulse width: duty ratio of 1:1

low-conductivity liquids.
Suitable for ﬁlling processes that are diﬃcult for electromagnetic ﬂowmeters to measure.

・Power supply
24 V DC ±20%Power consumption : 3 W● Excellent resistance
The fully-sealed structure withstands cleaning and high hu-(Inrush current: 5 A/100 μs)

midity. The perfect weld without any openings keeps out wa・Accuracy

ter and moisture.

±0.15% of reading (see the next page for details)● High resolution and accuracy with high-speed pulse outputs
A setting of 10,000 Hz allows 0.1 g per pulse to be mea-Conditions: water, 20 ℃ , and 0.1 MPa (accredited byUKAS EN17025)

Repeatability : depending on the ﬁlling time

sured.

STANDARD SPECIFICATIONS・ Measuring Principle: Coriolis force

・ Meter Size

: 10, 15 mm

・ Fluid: Liquid・ Enclosure: P67 (IEC 60529/NEMA6)・ Fluid Temperature: 0 to +100 ℃

1.5 to 3 seconds
3 to 5 seconds
5 seconds or more

Within 0.13%
Within 0.07%
Within 0.04%

・ Ambient Temperature : ‒40 to +55 ℃
・ Pressure: 0 Pa (abs) to +4 MPa (for the・ Mounting: Sanitary ﬁttings

measuring tube only)
1" ISO2852 ferrule (equiv. to IDF ferrule)1/2" Tri Clover clamp

TG-F1068-3E
RevisedMay 2016 K
1st edition Jun 2012

MMM4011C Coliolis Mass Flowmeter Mountable on Fillers

RECOMMENDED FILLING FLOW RATELENGTH OF STRAIGHT TUBE

Recommendedﬁlling ﬂow rate
(g/sec)Minimum ﬁllingvolume(g)Minimum ﬁllingtime(sec)

10 mm

15 to 370

231.515 mm

50 to 1200

75

1.5

Meter size

T here is no need to connect straight tubes upstream anddownstream.

The measuring range can be up to 130% of the recommended
ﬁlling ﬂow rate.

DETAILS OF ACCURACY
Accuracy

Condition±0.15% of reading

At least 10% of the maximum recommended ﬁlling ﬂow rate

0.1% of reading + zerostability

Up to 10% of the maximum recommended ﬁlling ﬂow rate

Zero stability: 0.005% of the maximum recommended ﬁlling ﬂow rate

（％）
0.5

0.45
0.4Accuracy

0.35
0.3
0.25
0.2
0.15
0.1
0.05

0102030405060708090

100 110 120 130

Maximum recommended ﬁlling ﬂow rate (%)
Meter size 10 mm: 370 g/sec
Meter size 15 mm: 1200 g/sec

EXAMPLEA typical conﬁguration for a ﬁlling system

Tank
Filling startcommand
MMM
4011C
Flow rate pulse
signal

PLC, batch controllers,
etc.

Valve
Valve control signal

Containers*1 NotePreliminary tests are recommended; mount the product on your equipment and check its performance.

2TOKYO KEISO CO., LTD.

TG-F1068-3E

MMM4011C Coliolis Mass Flowmeter Mountable on Fillers

EXTERNAL DIMENSIONS
DCA

B
Meter size(mm)Dimensions (mm)A

B

CDMass (kg)

10189

59.5

35333

2.8

15189

59.5

35333

2.9

ELECTRICAL CONNECTION
Wiring terminal (M12 male)

Terminal
numberColor*

Description

1

White

24 V DC power supply (+)

2Brown

RS485A (for service tools only)

3

Green

RS485B (for service tools only)

4

Yellow

Power supply (‒)

5GrayPulse output (+)

6PinkPulse output (‒)

7BlueNC

8

Translucent NC*These colors indicate those of the optional cable set including a connector plug for wiring.

TG-F1068-3E

TOKYO KEISO CO., LTD.

3

MMM4011C Coliolis Mass Flowmeter Mountable on Fillers

MODEL AND SPECIFICATION CODES
Model：MMM4011C-S □□* □□ : size code (10 or 15)
Flowmeter
spec. code
Sensor code

VE

4B

0K

0

H

000

Description

3VE

Meter size

MMM4011C

Meter size 10
Meter size 159192(Fixed code)

4○：

4Standard

Measuring tube materialNoneStandard (Inner surface ofwetted parts is polished at Raof 0.5 μm. Surface of outerhousing is polished.)
1/2" Tri Clover clamp(only for meter size 10)
1" ISO2852 ferrule(only for meter size 15)

0Measuring tube surface ﬁnish−：

B (stainless steel SS316L)

B

2KR

Process connection
MT(Fixed code)
Outer housing

0Not
available
Meter size
1015

○−−

○0

K (stainless steel SS316)
None
2-m cable + connector and plug (Lumberg)
10-m cable + connector and plug (Lumberg)
0 (not certiﬁed)
None3A (American Dairy Approval)*Specify code 2 also for the measuring
tube surface ﬁnish.
H (integral type)3-point calibration (standard)
5-point calibration (UKAS certiﬁed)

K0

Cable and connector for power supply and output
1wiringAExplosion proof certiﬁcation

00

Hygiene/sanitary approvals(Fixed code)

2H0Calibration

D(Fixed code)(Fixed code)(Fixed code)(Fixed code)

00
0

000
33(Blank) Not provided/Z Provided

Additional function

SPECIFICATIONS WHEN ORDERING
1. Model: MMM4011C
2. Meter size: 10 or 15 mm
3. Pulse output: totalized pulse of mass ﬂow rate
Specify the volume of mass ﬂow rate per pulse. Example: 0.1 g, 1 g, 10 g, etc. per pulse

STANDARD ACCESSORIES● Instruction manual: 1 copy

OPTIONS● Lumberg 8-pole female connector (M12 × 1) for power supply and output wiring* Specification is subject to change without notice.

Head Office : Shiba Toho Building, 1 – 7 – 24 Shibakoen, Minato-ku, Tokyo 105 – 8558
Tel : +81-3 – 3431 – 1625 (KEY) ; Fax : +81-3 – 3433 – 4922
e-mail : overseas.sales@tokyokeiso.co.jp ; URL : http://www.tokyokeiso.co.jp
4TG-F1068-3E