Product information

FEATURES:· Easy to use

LabTEC-X
Corona Lab System· Multi-layer counterelectrode· Usable for a numberof different
applications· Generator· Touch panel· Comes as apackage

Head OfficeTantec A/SIndustrivej 6
DK-6640 Lunderskov(+45) 7558 5822
USA:(+1) 630-351-1320
Mail:sales@tantec.com
Web:
www.tantec.com

LabTEC-X is a corona system that has been adapted for use in a laboratory
setting. The generator provided with this system has been adapted to provide
tighter control as the system is used as a test device. The system is strictly
manual to ensure that this will be a safe addition to any laboratory. You can easily
treat a variety of parts by holding the insulated handle of the electrode.
Adhesive bonding and finishing of moulded and extruded parts made of polyethylene, polypropylene,
TPO and EPDM are constant manufacturing problems. Medical laboratory products made of polystyrene have poor wettability and poor cell attatchment. Surface treatment technology offers innovative
solutions to adhesion and wetting problems in many industries. Corona surface treatment is a
modification of polymer surfaces with electrical discharge at atmospheric pressure.
The corona process creates the necessary polar functional groups on the polymer surface, which
promote adhesion and wettability. Tantec’s bench top LabTEC-X is specially designed for treatment ofsmall polymer parts.
This includes both three-dimensional parts and films. The LabTEC-X is specially configured for R&D
lab and proof of concept work. This system consists of a high-frequency power generator and atreating station with high-voltage transformer integrated into the station. The high-voltage transformer
steps up the output signal from the generator to produce the high frequency. The output of thetransformer is connected to a stainless steel or ceramic roller electrode by way of an insulated
high-voltage cable.

Technical Data

Easy to use
This system is built for LAB use and is very
flexible. It can be operated easily with safety
shielding enclosure.
Multi-layer counter electrodeTests a variety of polymers.
Usable for a number of different applications
Both conductive and non-conductive material
can be treated as the system is supplied witha hand-ball and a ceramic electrode.

Touch panelProface color touch screen 5.7” with built in PLC
unit allows complete control of all available
generator parameters. Anodized aluminium
housing has foot stand and holes for wall
mounting.
Comes as a package
The unit is a complete package: Generator,
remote control, cables, LAB station, ball and wire
electrode. Transformer type is
HT-X01-28: 28 kW.

Generator
Generator type is HV-X02: 0-200 Watt output.

Technical Specifications

HV-X02 Generator

LAB-station

Mains Voltage and Frequency

100-240VAC 50/60Hz

N/A

Output voltage/plasma power

Max. 400 Vp/0-200 Watt

N/A

Power consumption

300 VA

N/A

Ramp up time

5-30 ms, depending on power load

N/A

Shut down time

< 1 msN/A

Treatment area

N/A

300 x 350 mm

Dimensions in mm (LxWxH)

430 x 470 x 200

635 x 500 x 410

Weight in kg

12.5

40

Operator interface control(Optional)

5.7” STN color touch display
Ports: (1) USB - (1) Ethernet- (1) Serial RS485

N/A

Bus communication system

Standard: RTU 4-Wire Modbus,
57.6Kbaud, 8bit. Optional:
CANopen/Profibus and others

N/A

www.tantec.com

150822-PLA

CE - ROHS - WEEE compliant