Touch Screen Programmable Controller
PCT-200• Easy to view & operate via Touch Screen• Choose from 4 languages (Japanese, English, Chinese, Korean)• Fast sampling period: 25 ms• Large capacity memory: 100 patterns, 100 steps• Logging data can be saved on USB memory stick (Recorders unnecessary)• Data can be loaded from USB memory stick.

Model-PCT-20
Temperature control

Ordering Example-PCT-20 9 – R – 1

Number of connected WCL-13A units

1 to 9module

RRelay contact output

Control output

SNon-contact voltage output(WCL-13A-SA/MM PCT)A

Direct current output(WCL-13A-AA/MM PCT)

(WCL-13A-RA/MM PCT)

Temperature control module

1

For Y-type terminals only; Finger-safe (ASK-001-1)

Socket2

For Y-type and ring terminals (ASK-002-1)

9 units of WCL-13A are connected.
Output : Non-contact voltage output
Socket : For Y-type terminals only;
Finger-safe

Specifications
Temperature control module
Input

Power Supply
Mounting

For details, refer to the catalog of the WCL-13A.
Thermocouple, RTD, Direct current, DC voltage
Number of Patterns:
100 patterns
Number of Steps:
100 steps
Number of Repetition:
0 to 99 times
Program Time Range:
0 to 99 min 59 sec/Step or 0 to 99 hr 59 min/stepTime Setting Accuracy:
Timer accuracy ( 65 sec/month at 10 to 40 )Wait Block Parameters:
Setting range: 0 to 1000 (Decimal point depends on the input range.)
Block Setting:
PID block (10 types), Time signal block (16 types), Wait block (10 types), Alarm block (10 types)
Time Signal:
20 points (Open collector output)
Alarm Output:
4 points (Open collector output)
Pattern End Output:
1 point (Open collector output)
External Operation Input: Pattern selection (Selects 1 to 100 pattern numbers via BCD code, BCD 8-point input)
Start, Stop, Advance, Fast (speeds up program progression.)
100 to 240 V AC
Touch Screen: Flush
Temperature control module: DIN rail

External Dimensions
WeightTouch Screen: W167.5 x H135 x D78 mm
Touch Screen: 1.0 kg or less

Program Performance

Temperature control module: W30 x H88 x D108 mm (including socket)
Temperature control module: Approx. 200 g (including socket)

Touch Screen Monitoring Display
Opening display
Menu button
Window Name(1)Setting Contents(1)

OperationOperated by pressing buttons, and control status is displayed.

(2)

Operation monitoring

Program pattern graph is displayed.

(3)

Monitoring all WCL units

Indicates PV, SV and MV of the connected WCL-13A units.

(4)

(4)

Program setting

Sets program pattern and steps.

(5)

(5)

Block setting

Sets Time signal block, PID block, Wait block and Alarm block.

(6)

(6)

Engineering setting

Sets alarm type, alarm hysteresis, proportional cycle and sensor(7)

Setting lock selection,Unlocks, or locks all setting contents so that they cannot be changed.

(8)

Selection after power

Selects RUN/STOP after power restoration.

(9)

restoration

(2)
(3)

correction value.

(7)

By pressing the Menu button in the Opening display, windows
from (1) to (9) are displayed.
For the setting method from (1) to (9), see the Instruction Manual.

(8)

SV bias setting

Sets SV bias from the second WCL-13A and all following units.

(9)

Pattern selection &Selects the pattern number.

USB management

Reads/Writes data from/on the USB memory stick.

Contents of Package
When delivered, check that the following contents to confirm that all items you have ordered are contained.
Consult us regarding the wiring-completed (customized) products.
ModelQuantity
Touch Screen
LT-3300-S1-D24-K
1
Temperature control module (*1)
WCL-13A- A/MM PCT
1 to 9 (*2)
Connection cable (Between Touch Screen - WCL-13A) CWD-C5
1
CDD
(*3)
Connection cable (Between WCL-13As)
Switching power supply 24 V DC
Cable for front mounted USB port
EX module
MIL socket connector harness
USB memory stick
Socket (*1)

PS3L-D24-AFC
CA5-USBEXT-01
EXM-DDO16UK
HIFS-SY-SB-20-5
MF-SU202GWH
ASK-00 -1

11111(*3)(*1) Specify one when ordering. When WCL units are connected, their specifications should be the same.
(*2) For Instrument Number setting for the WCL-13A, start from 1 (one) and progress chronologically.
(*3) When multiple WCL units are ordered, corresponding amount of cables and sockets are included.

Configuration Example
DIO connector

Touch Screen

WCL-13A

When connecting multiple WCL-13A units, use included CDD cable.
For details, refer to the Instruction Manual for the WCL-13A.
Instruction Manual for the WCL-13A can be downloaded from Shinko Website.
http://www.shinko-technos.co.jp/e/
Support & Downloads
Downloads
Manuals
Connection cable CWD-C5(Touch Screen - WCL)Power connector(included)
Cable for front mounted USB port
(CA5-USBEXT-01)
CH1 input

Socket for WCL

Powersupply

24 V AC

CH1 output

100 to 240 V AC
Switching power supply
PS3L-D24-AFC■ Touch screen Terminal Arrangement (For details, refer to LT3300 Series Hardware Manual.)
DIO Connector
Pin Connection

Pin No.
A1Signal Name
IN1
BCD 100-2

Pin No.
B1A2

IN3

BCD 100-8

B2IN2

BCD 100-4

A3IN5

BCD 10 -2

B3IN4

BCD 10 -1A4IN7BCD 101-8

B4IN6BCD 101-4A5IN9STOPB5IN8RUN/HOLD(*)

A6A7IN11IN13FASTB6B7IN10IN12ADV

A8IN15B8IN14A9NC

B9COM

1EX Module Connector

Signal Name
IN0BCD 100-1
1A10Sink output type: NC

B10Sink output type: +24 V

A11Sink output type: 0 V

B11Sink output type: 0 V

A12A13OUT1OUT3AL1AL3B12B13OUT0OUT2A14OUT5TS1B14OUT4P.END
AL2AL4A15OUT7TS3B15OUT6TS2A16OUT9TS5B16OUT8TS4A17OUT11TS7B17OUT10TS6A18A19OUT13OUT15TS9TS11B18B19

OUT12OUT14TS8TS10(*) Contact CLOSED: RUN, Contact OPEN during RUN: HOLD

■ Touch Screen External Dimensions (Scale: mm)■ Panel Cutout (Scale: mm)
Panel thickness: 1.6 to 5.0 mm

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, Japan
http://www.shinko-technos.co.jp
URL:E-mail:overseas@shinko-technos.co.jp

Tel: +81-72-727-6100
Fax: +81-72-727-7006
2013.09