Digimatic Indicator for Bore Gage ID-C112GB
Digimatic Bore Gage CG-D100/160
Key operation

The name and function of a key
Press shortly(less then 2 sec.)
Press and hold(2 sec. or more)

Master setting mode/
Initial state change
Return to the state at
RESET
the time of battery
insertion
ON/OFFPower ON/OFFin/mm
Switches inch/mm display
DATA/HOLD Data hold / Data output
PRESET
Operation of PRESET
STARTStarting the measurement
TOL.
Setting the toralance
M/W

Setup
①Install the batteries
It is unnecessary toshut off a power②User's Manual
No.99MAH015B
No.543, 511Operation Manual

PRESET

ID-C112GB

CG-D100/160

Master setting

Master setting

When a master value is registered newly

When the master value is registering
beforehand① To the master settingmode⑨ Detect the minimum valueof the master ring

① Recall the master value
M/W

Measuring the inside diameter① Starting of minimum value
measurement
START

M/W

The registered master value② Selection of a memory② Setting the master ring

M/W

START

Secretion of M1, M2, M3.

Please push a key after setting abore gage to master ring.

③ Presetting the master value
PRESET

Shake at forward and backward or
right and left, and explore theminimum value.

④ Moving a digit
PRESET

Please push a key after setting abore gage to master ring.

② Detect the minimum valueof the work

③ Detect the minimum valueof the master ring

⑩ Holding the displayed value
DATA/HOLD

Hold until a desired digit
starts blinking.

⑤ Change of a value
PRESET

Only the pushed number oftimes changes a value.

⑥ PRESET
Please continue pushing until"P" blinks.

⑦ Comoletion of presetting
the master value
PRESET

The master value preseted byoperation to ⑥ is blinks.
Although the value which
preseted by the operation so
far is memorized by thememory, it is not registered asa master value. Please
perform subsequent⑧ Setting the master ring
START

Repeat operation ⑧ to ⑨ several
times and check a display isstabilized in zero.
If there is a possibility of an
accidental change of the minimum
value detected, press the[DATA/HOLD] key shortly to holdthe display value.("H" is lighting.)

Shake at forward and backward or
right and left, and explore theminimum value.

⑪ Recall the master value④ Holding the displayed value

PRESET

DATA/HOLDPlease push a key after drawing
out a bore gage from master ring.
The value preseted by operation
to ⑥ is called and it blinks.

⑤ Recall the master value⑫ Registering the master value⑥ Registering the master value

PRESETIf a key is pushed, blinking willchange to lighting, and a mastervalue will be registered.

⑬ Completion of registrationof the master value
M/W

Please push a key after setting
a bore gage to master ring.

Shake at forward and backward or
right and left, and explore theminimum value.

③ The minimum value isdisplayed

If the minimum value is detected,
a digital display will hold the value.

PRESETCAUTIONPRESET⑦ Completion of registrationof the master value

Even when measuring the same
work measured on the previous
day, in case you startmeasurement, please operate"Master setting" again, and be
sure to check a zero point.

M/W

Returns to the state where
measurement is stood by.

Returns to the state where
measurement is stood by.

The value displayed here is reference. It differs from the value actually displayed.