AC DIGITAL CLAMP METER

KEW 2007R/2117R/2127R

CAT Ⅳ 1000A True-RMS
NEW AC CLAMP METER SERIES

Measures both low and high power circuits
KEW 2117R

 Peak Hold for inrush

current *2127R Only

 Backlight LCD provides

easy readingin dark locations *2127R Only Fully Safety jaw
 Ergonomic over-molded

body gives convenient
one-hand operation

 Large easy-to-read display

with 0.01 A resolution*2117R, 2127R

 Safety Standard

IEC61010-1 CAT Ⅳ 300V / CAT Ⅲ 600V

KEW 2127R

KEW 2007R

● KEW 2007R/2117R/2127R Specifications
KEW 2007R
600.0/1000A (Auto-ranging)±1.5%rdg±4dgt [45 - 65Hz]±2.0%rdg±4dgt [40 - 400Hz]
600.0V
±1.2%rdg±3dgt [45 - 65Hz]±1.5%rdg±4dgt [40 - 400Hz]
600.0V
±1.2%rdg±3dgt
600.0Ω/6.000kΩ (Auto-ranging)±1.3%rdg±5dgt (600Ω)±2.0%rdg±3dgt (6kΩ)
600Ω (Buzzer sounds below 90Ω)

AC A
AC V
DC VΩ

Continuity buzzer
Hz
Capacitance test
Diode test
Conductor size
Applicable Standards

KEW 2117R
KEW 2127R
60.00/600.0/1000A (Auto-ranging)±1.5%rdg±4dgt [45 - 65Hz]±2.0%rdg±5dgt [40 - 1kHz]
60.00/600.0V (Auto-ranging)±1.0%rdg±2dgt [45 - 65Hz] (600V)±1.5%rdg±4dgt [40 - 1kHz] (60/600V)
60.00/600.0V (Auto-ranging)±1.0%rdg±3dgt (60V), ±1.2%rdg±3dgt (600V)
600.0Ω/6.000/60.00/600.0kΩ (Auto-ranging)
600.0Ω/6.000/60.00/600.0kΩ/6.000/40.00MΩ(Auto-ranging)±1.0%rdg±5dgt (600Ω)±1.0%rdg±5dgt (600Ω), ±2.0%rdg±3dgt (6/60/600kΩ)±2.0%rdg±3dgt (6/60/600kΩ)±3.0%rdg±3dgt (6MΩ), ±5.0%rdg±3dgt (40MΩ)

̶̶̶̶

̶̶

999.9Hz/9.999kHz (Auto-ranging) ±0.1%rdg±3dgt(Input sensitivity Current:more than 4A Voltage:more than 2V)
1.000/10.00/100.0μF±3.0%rdg±15dgt (1μF)±3.0%rdg±10dgt (10/100μF)
2.000V ±4%rdg±5dgt Open-loop Voltage:<3.5V∅33mm max.

IEC 61010-1 CAT Ⅳ 300V, CAT Ⅲ 600V Pollution degree 2
IEC 61010-031, IEC 61010-2-032, IEC 61010-2-033, IEC 61326-2-2(EMC), IEC 60529 IP40, EN 50581(RoHS)
Power source
R03/LR03(AAA)(1.5V)x2
Continuous measuring time Approx. 170 hours (when R03 is used) (Auto power save : approx. 10 minutes) (NCV_LED*1, Backlight*2:off)
Dimensions
204(L) × 81(W) × 36(D)mm
Weight220g approx. (including batteries)
230g approx. (including batteries)
Accessories
7066A (Test leads), 9079 (Carrying case), R03 (AAA)×2, Instruction manual*1 2117R, 2127R *2 2127R only

● Included Accessories

NCV Function (2117R, 2127R)

Easy to read backlight LCD display*

NCV Sensor can detect electrical field
only from the direction indicated in thebelow figure.
Put the fixed element (left side) closer tothe conductor under test.
Detection against in-wall outlet is impossible.
Red LED lights up in electric field exceeding AC70V.

7066A*2127R only

9079

True RMS function for accurate readings also inpresence of modern loads such as LED lamps,motor inverters, PCs, TVs, Air conditioners, etc.

● Selection Guide

Peak hold Function (2127R only)

2007R

2117R

2127R

Continuity buzzer

〇

〇

〇

NCV×

〇

〇

Capacitance test

××〇

Diode test

××〇

Back light××〇

Peak hold××〇

Press the SELECT key on ACA, ACV or DCV
function to start PEAK measurement. TheLCD shows updates the max measured
value repeatedly during a measurement.

ONPEAKOFF

Peak Hold for inrush current.

STEADYSTATE

Waveform

True RMS type

Averagingvalue typecorrect		 reading
correct		 reading
correct		 reading
correct		 reading

correct		 reading
higherreadinglower
readinglower
reading








Please read the "Safety Warnings" in the instruction manual supplied with the instrument thoroughly and completely

correct use. Failure to follow the safety rules can cause fire, trouble, electrical shock, etc. Therefore, make sure
Safety Warnings : forto operate the instrument on a correct power supply and voltage rating marked on each instrument.

For inquires or orders :2-5-20, Nakane, Meguro-ku, Tokyo, 152-0031 Japan
Phone:+81-3-3723-0131
Fax:+81-3-3723-0152
E-mail:info-eng@kew-ltd.co.jp

The contents of this leaflet are subject to change without notice.

KEW 2007R/2117R/2127R-1E Feb. 17 AD