Buffered USB2.0 CCD Line Camera with External Trigger (16-bit)(Part Number: TCN-1304-U)

FEATURES• USB2.0 compatible• Board-level camera, idealfor OEM applications• No external power supply
required• Optical integration time

adjustable from 100µs to6.5s• 3648 pixel silicon linear
CCD array• 8µm x 200µm pixel size

• 16-Bit A/D converter forhigh intensity resolution• High scan rate (up to 138scans/second)• External trigger capability• 4 GPIOs• Full-featured SDK• Linux driver available• USB command set protocol
for non-Windows basedapplications• LabVIEW support is provided• Demo graphical user interface• Compatible to Windows
2000XP or higher

APPLICATIONS• Industrial Process Control• Optical Spectroscopy

PRODUCT DESCRIPTION
Mightex’s TCN-1304-U line camera is a cost-effective high-performance B/W board-level line camera, based on a single-line, 3648-pixel CCD chip with USB2.0 (480 Mb/s) interface. CCD line cameras have several advantages over their area-array counterparts, including high optical linear resolution that allows systems developers to use the cameras to capture two-dimensional (2-D) imagesby moving the object or the CCD perpendicularly to the scan line. The TCN-1304-U is a compact,
board-level line-scan camera ideal for a variety of OEM applications in industry process control,
optical spectroscopy and bio-medical imaging etc. Setting up the TCN-1304-U line camera is very
easy, the user simply installs the latest version of the operating software onto any desktop or notebook PC and then connects the USB cable from the line camera to the PC. There is no need for
installing a DAC card, or using an external power supply. A user-friendly GUI based application
software, LabVIEW support and an SDK are provided. In addition a USB command set protocol anda Linux driver are available for non-Windows based applications.

PERFORMANCE SPECIFICATIONSParameters

TCN-1304-U

Board-level/enclosed

Board-level

CCD chipUnit

Toshiba TCD1304DG

Number of pixels

3648

pixelsPixel size

8 x 200µm

Spectral range

200 to 1000 (see spectral response below)

nm

Pixel output clock

0.5

MHz

Data storage on camera

4frames

ADC resolution

16bitsExternal trigger

Yes

Exposure time range

0.1 ~ 6,500Number of GPIOs

4 programmable I/O’s

Frame rate

138

Host interface

USB2.0ms

scans/second** Frame Rate is dependent on exposure time. When exposure time is set to 0.1 ms, the camera can achieve
138 scans / second.

SDK Features
Operation Systems

Windows 2000, XP, Vista and Windows 7

Minimum Requirement

RAM > 64M, hard disk space used > 10M

USB Port

2.0

Multiple Cameras

Supported

Device Driver

Yes

Demo Application

Yes

Library Files

Yes (DLL files and static library file)

Example Codes

Yes (VC++ and Delphi)

Frame Attributes*

Exposure Time, Time Stamp, Trigger Event Count,Over-Exposure Detection.

* SDK will provide call back, which will send user frame data and the related attributes of the frame. The attributes
include: Exposure Time, Time Stamp, Trigger Event Count and Over-Exposure Flag.
Mightex Systems, 2343 Brimley Road, Suite 868, Toronto, Ontario M1S 3L6, Canada
Tel: 1-416-840-4991 or 1-925-218-1885, Fax: 1-416-840-6541, Email: sales@mightex.com
www.mightex.com or www.mightexsystems.com

Buffered USB2.0 CCD Line Camera with External Trigger (16-bit)(Part Number: TCN-1304-U)

SPECTRAL RESPONSE

OPERATION CONDITION
Operating Temperature Range:
Storage Temperature Range:
Relative Humidity, Non-condensing:-10°C ~ +50°C-25°C ~ 85°C
5% ~ 95%

EXAMPLE OF GRAPHICAL USER INTERFACEMECHANICAL DIMENSIONS
Controller Board:
CCD Daughter Board:
Daughter Board Cable Length:69.0mm x 49.5mm
20.0mm x 49.5mm
120mm【お問い合せ先】〒���-���� 東京都千代田区富士見�-�-�ステージビルディング��階TEL ��-����-���� https://www.ads-img.co.jp

With a world-class OEM design team, Mightex offers a broad range of customized solutions in order to meet individual customer’sunique requirements. Please call 1-416-840 4991 or email sales@mightex.com for details.
Mightex Systems, 2343 Brimley Road, Suite 868, Toronto, Ontario M1S 3L6, Canada
Tel: 1-416-840-4991 or 1-925-218-1885, Fax: 1-416-840-6541, Email: sales@mightex.com
www.mightex.com or www.mightexsystems.com