Next-Generation
Optical Parametric Amplifier
From UV to MIR

Continuous power monitoringand diagnostics
Pumped by PHAROS-UP
for ultrashort pulses
Up to 80 W, 800 µJ pumpat up to 2 MHz

Fully integrated
wavelength extensions

Second repetition rate
pump option

Exceptional output stability

ORPHEUS-NEO-ONE typical tuning curves in ONE configuration.
Pump: 80 W, 800 µJ, 100 kHz
200

10000

Output power, mW

5000
3000
2000

100

1000

10500

5300
200

32100

150

0.530200300500 700 1000

2000 3000

6000

SignalIdler
DFG
VIS–NIR extension

10000

50302020020000Output power, mWSignal
Idler
SHSSHITHSDFG1DFG2Pulse energy, µJ

20000

5000
4000
3000

5040302000

20100010500
4003005432002100500 700

0.310000 16000

1001000 1400

2000 3000 4000

6000

10000

116000

Wavelength, nm

Wavelength, nm

ORPHEUS-NEO typical long-term power stability at 800 nm

Output power, a.u.

1.10
1.05
1.00
0.95
NRMSD = 0.4%
0.90051015

REV. 240522 | SALES@LIGHTCON.COM

202530354045Time, h50556065

70758085

Pulse energy, µJ

ORPHEUS-NEO typical tuning curves in HP configuration.
Pump: 80 W, 800 µJ, 100 kHz

ORPHEUS-NEO specifications
ModelConfiguration

ORPHEUS-NEO

ORPHEUS-NEO-ONE

ORPHEUSORPHEUS-ONE

Applications | Microscopy Sources | Spectroscopy Systems | OPCPA Systems | Wavelength-Tunable Sources | Femtosecond Lasers

Pump power

Up to 80 W

Pump pulse energy

20 – 800 µJ

Repetition rate 1)

Up to 2 MHz
640 – 1000 nm (signal)1050 – 2600 nm (idler)

1400 – 2000 nm (signal)
2100 – 4200 nm (idler)> 7% @ 700 nm(40 – 800 μJ pump; up to 1 MHz)

> 9% @ 1550 nm(40 – 800 μJ pump; up to 1 MHz)> 3.5% @ 700 nm(20 – 40 μJ pump; up to 2 MHz)

> 6% @ 1550 nm(20 – 40 μJ pump; up to 2 MHz)

60 – 220 cm-1 @ 700 – 960 nm

50 – 150 cm-1 @ 1450 – 2000 nm

Tuning range

Conversion efficiency

Spectral bandwidth

120 – 400 fs

100 – 400 fs

2

Beam quality (M )< 1.3 @ 800 nm< 1.3 @ 1550 nm

Beam diameter3)

2.1 ± 0.6 mm @ 800 nm

2.1 ± 0.6 mm @ 1550 nm< 2 mrad @ 800 nm< 4 mrad @ 1550 nm< 1% @ 800 nm< 1% @ 1550 nm< 1% @ 800 nm< 1% @ 1550 nm

Pulse duration 2)

Beam divergence (full-angle)
Long-term power stability, 8 h4)

Pulse-to-pulse energy stability, 1 min 4)

210 – 320 nm (THS); > 0.4% @ 250 nm
Wavelength extension options;conversion efficiency

640 – 1000 nm and 1050 – 1350 nm (VIS–NIR);> 1% @ 700 nm

320 – 500 nm (SHS) and 525 – 640 nm (SHI);> 1.2% @ 350 nm
2500 – 4200 nm (DFG1); > 3% @ 3000 nm

4000 – 16000 nm (DFG);> 0.3% @ 10000 nm (for > 40 µJ pump)4000 – 16000 nm (DFG2); > 0.2% @ 10000 nmPUMP LASER REQUIREMENTS
Configuration

PHAROS or CARBIDE

Center wavelength1030 ± 10 nm

Maximum pump power

80 W

Maximum repetition rate

2 MHz

Pump pulse energy

20 – 800 µJ

Pump pulse duration

180 – 500 fs

ENVIRONMENTAL & UTILITY REQUIREMENTS
Operating temperature 5)
Relative humidity

19 – 25 ºC (air conditioning recommended)
20 – 70% (non-condensing)5)

Electrical requirements

100 – 240 V AC, 4.5 A; 50 – 60 Hz

Rated power

280 WStandby: 20 W
Max during wavelength tuning: 200 W

Power consumption1)

Second repetition rate pump option (up to 20 µJ)
is available for signal and extension range in HP
configuration; contact sales@lightcon.com.

2)

Output pulse duration depends on selectedwavelength and pump laser pulse duration.

3)

FW 1/e2, measured at laser output, using maximum
pulse energy.

4)

Expressed as NRMSD (normalized root mean squareddeviation).

5)

Specifications are guaranteed for a maximum
temperature variation of ± 1 ºC and humidity variation of± 10%.

DANGER: VISIBLE AND/OR INVISIBLE
LASER RADIATION AVOID EYE OR SKINEXPOSURE TO DIRECT, REFLECTED OR
SCATTERED RADIATION

CLASS 4 LASER PRODUCTWWW.LIGHTCON.COM

ORPHEUS-NEO-UP specifications
Model

ORPHEUS-NEO-UP

ORPHEUS-NEO-ONE-UP

ORPHEUSORPHEUS-ONE

Configuration

Up to 20 W

Pump pulse energy

20 – 400 µJ

Repetition rate 1)

Up to 1 MHz
640 – 1000 nm (signal)1050 – 2600 nm (idler)

Tuning range

1450 – 2000 nm (signal)
2100 – 4500 nm (idler)

Conversion efficiency> 7% @ 700 nm> 9% @ 1550 nm

Spectral bandwidth

120 – 300 cm-1 @ 700 – 2600 nm

150 – 250 cm-1 @ 1500 – 1900 nm & 2200 – 3500 nm 2)< 100 fs @ 700 – 1000 nm< 120 fs @ 1060 – 2000 nm< 100 fs @ 1500 – 1900 nm

Pulse duration 3)
Beam quality (M2)

< 1.3 @ 800 nm< 1.3 @ 1550 nm

Beam diameter 4)

2.1 ± 0.6 mm @ 800 nm

2.1 ± 0.6 mm @ 1550 nm< 2 mrad @ 800 nm< 4 mrad @ 1550 nm< 1% @ 800 nm< 1% @ 1550 nm< 1% @ 800 nm< 1% @ 1550 nm

Beam divergence (full-angle)
Long-term power stability, 8 h5)

Pulse-to-pulse energy stability, 1 min 5)

210 – 320 nm (THS); > 0.2% @ 250 nm
Wavelength extension options;conversion efficiency

640 – 1000 nm and 1050 – 1450 nm (VIS–NIR);> 1% @ 700 nm

320 – 500 nm (SHS) and 525 – 640 nm (SHI);> 1.2% @ 350 nm
2500 – 4500 nm (DFG1); > 3% @ 3000 nm

4500 – 14000 nm (DFG);> 0.2% @ 10000 nm

4500 – 14000 nm (DFG2); > 0.1% @ 10000 nm

PUMP LASER REQUIREMENTS
Configuration

PHAROS-UP

Center wavelength1030 ± 10 nm

Maximum pump power

20 W

Maximum repetition rate

1 MHz

Pump pulse energy

20 – 400 µJ

Pump pulse duration

80 – 100 fs

ENVIRONMENTAL & UTILITY REQUIREMENTSOperating temperature 6)
Relative humidity

19 – 25 ºC (air conditioning recommended)
20 – 70% (non-condensing)

6)

Electrical requirements

100 – 240 V AC, 4.5 A; 50 – 60 Hz

Rated power

280 WStandby: 20 W
Max during wavelength tuning: 200 W

Power consumption

1)

Second repetition rate pump option (up to 20 µJ)
is available for signal and extension range in HP
configuration; contact sales@lightcon.com.

2)

Spectral bandwidth is equal to 150 – 250 cm-1 @5000 – 12000 nm.

3)Output pulse duration depends on selected
wavelength and pump laser pulse duration.

4)

FW 1/e2, measured at laser output, using maximum
pulse energy.

5)

Expressed as NRMSD (normalized root meansquared deviation).

6)

Specifications are guaranteed for a maximum
temperature variation of ± 1 ºC and humidity
variation of ± 10%.

DANGER: VISIBLE AND/OR INVISIBLE
LASER RADIATION AVOID EYE OR SKINEXPOSURE TO DIRECT, REFLECTED OR
SCATTERED RADIATION

CLASS 4 LASER PRODUCTDrawings
ORPHEUS-NEO / ORPHEUS-NEO-UP drawings
25 (1”)

525 (21”)

For M6 (1/4-20 UNC) screw
6 mounting holes

76646.5
26260

30125624125

425 (17”)39067714453

www.phototechnica.co.jp
REV. 240522 | SALES@LIGHTCON.COM〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067 FAX:048-871-0068

e-mail:voc@phototechnica.co.jp

Femtosecond Lasers | Wavelength-Tunable Sources | OPCPA Systems | Spectroscopy Systems | Microscopy Sources | Applications

Pump power