BEAMAGE

CMOS beam profiling cameras

INTUITIVE SOFTWARE INTERFACE
Easy to navigate interface, with many displays and control features:
2D, 3D and XY Displays
Background Subtraction Function
Unique “Animate” Function
Gaussian Fit
Semi-Log Graph

KEY FEATURESACCESSORIES

USB 3.0 FOR THE FASTEST TRANSFER RATESUp to 10X faster than regular USB 2.0
connections
HIGH RESOLUTION
4.2 Mpixels resolution gives accurate profile
measurements of very small beams
LARGE APERTURES

Stand with delrin post

BA series
optical attenuators

Stackable ND filters(0.5, 1.0, 2.0, 3.0, 4.0 & 5.0)

UV and IR filters

Pelican carrying case

Fiber adaptors & connectors(FC, ST and SMA)

• 11.3 x 11.3 mm for the Beamage-4M• 20.5 x 20.5 mm for the Beamage-4M-FOCUS
AVAILABLE WITH IR COATING
Beamage 4M-IR cameras have a special phosphor
coating for IR wavelengths (1495-1595 nm)
ISO COMPLIANT
D4Ơdefinition of diameter, centroid, ellipticity
and orientation are ISO 11146:2004 and 11146:2005
compliant
EXTERNAL TRIGGER
To synchronize the camera with a pulsed laser

UV converters &IR adaptors

107

gentec-eo.com/laser-beam-diagnostics

BEAMAGE
Specifications

BEAMAGE-4M-IR

BEAMAGE-4M-FOCUS

CMOS

CMOS

(with phosphor coating)(with fiber optic taper)

SENSOR TECHNOLOGYCMOS

EFFECTIVE APERTURE

11.3 x11.3 mm

11.3 x 11.3 mm

20.5 X 20.5 mm a

Camera only

350 - 1150 nm

1495 - 1595 nm

350 - 1150 nm

With UG11-UV filter

250 - 370 nm------

With B3-IR-filter

POWER DETECTORSBEAMAGE-4M

MEASUREMENT CAPABILITY
Wavelength range------

4.2 MPixels

4.2 MPixels

4.2 MPixels

HxV

2048 x 2048

2048 x 2048

2048 x 2048

Minimum measurable beam

55 µm70 µm

120 µmFrame rate

ENERGY DETECTORS

1250 - 1350 nm

Pixel count

6.2 fps at 4.2 MPixels (Full Frame)
11.4 fps at 2.1 MPixels (2048 x 2048)
18.6 fps at 1.1 MPixels (2048 x 544)

RMS noise
DAMAGE THRESHOLDS
Maximum average power
Maximum density (1064 nm)

1000:1 (60 dB)

1000:1 (60 dB)
1 W with ND filter

1 W with ND filter

10 W/cm2
0.1 J/cm2

10 W/cm2
0.1 J/cm2

10 W/cm2
0.1 J/cm2

2D, 3D, XY and Beam Tracking

D4σ (ISO compliant),
1/e2 along crosshairs (13.5%)
FWHM along crosshairs (50%)
Custom (%)

Beam Diameter Definitions

Buffer Controls

Open File, Save Current Data, Save All Data, Previous/Next Image, Clear Buffer, Animate

11.3 x 11.3 mm

11.3 x 11.3 mm

1.28 cm2

1.28 cm2

1.28 cm2

Effective aperture

Same as sensor

Same as sensor

20.5 x 20.5 mm

Dimensions (not including filter)

61H x 81.1W x 19.7D mm

61H x 81.1W x 19.7D mm

61H x 81.1W x 46.5D mm

Weight (head only)

138 g

138 g

235 g

STAND-D-233

STAND-D-233

STAND-D-233

Product page

Specifications are subject to change without noticea. With a typical pixel multiplication factor (PMF) of 1.8.

T 418.651.8003 | info@gentec-eo.com〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

PRODUCT GUIDE 2023

TEL:048-871-0067

FAX:048-871-0068,

https://www.phototechnica.co.jp
e-mail:voc@phototechnica.co.jp.com

CUSTOM / OEM PRODUCTS

11.3 x 11.3 mm

Sensor area

DISPLAYS & PC INTERFACES

Full Report in Print Ready Format (2D, 3D, XY, Measures, Parameters)
Print Screen in BMP format (2D and 3D)

Printing and Reports

ORDERING INFORMATION
Compatible standTERAHERTZ DETECTORS

2D: Print Screen, Reset View, Show/Hide Beam Diameter
3D: Print Screen, Reset View, Top View
XY: Save Data, Zoom, Gaussian Fit, Semi-Log, Show/Hide Cursor, Show/Hide FWHM, Show/Hide 1/e2
Beam Tracking: Save Data, Print Screen, Reset View, Zoom

Display Features

PHYSICAL CHARACTERISTICS
Sensor size

1000:1 (60 dB)

1 W with ND filter

SOFTWARE
Displays

BEAM PROFILING

32 fps at 0.066 MPixels (256 x 256)108