Multifunction Tester
KEW 6016

The most User Friendly Multifunction
Tester complete with Anti-Trip Technology

10 in 1CONTINUITYINSULATIONLOOPPSC

PFC

RCD

EARTH

PHASE ROTATIONVOLTSFREQUENCY

+%7 

The most User

KEW 6016 Multifunction tester makes ‘Easeof Use’ it’s top priority with such features as:
Advanced Anti-Trip Technology(ATT) to avoid RCDs to trip
during loop testing.

A single rotary dial to make your selection
and a fine Dot Matrix display gives
multiple information
on one screen.
Slim remote probe with test button as well asa lockdown option on the instrument for themost convenient hands
free testing.

L-PE

Auto test (RCD Measurement)

UL50V
OK

Auto test enables complete testing of RCD (6 tests) while you simply stand by andreset the RCD. All the results are displayed on one screen. No need to scroll.

AUTO
30 mA

L-N!

X1/2 0°>2000ms
180°>2000ms
X1 0°
27ms
180°
17ms
X5 0°
5ms
180°
15ms

Null facility (Continuity Measurement)
When you press Null (F1) button, the test leads resistancewill be eliminated and the display will read zero ohm.
The nulled value will be stored, even if the instrument isswitched off, unless the nulling is cancelled.

MeasuringNULLON

MEMORY and PC SOFTWARE
KEW6016 has a memory of up to 1000 results and it issupplied with a comprehensive PC software that allows
preparing complete reports with 3 levels of description forevery stored result such as Circuit number, Board numberand Site number.

SAVE MODEDATA No.001
UPDOWNSELECT
OKCIRCUIT:
01
BOARD:
01
SITE:01

ΩFriendly Multifunction Tester complete with Anti-Trip Technology
Continuity Measurement
Continuous testing can be carried out by use of the test button lockdownfeature. A selectable buzzer gives instantaneous indication of continuity. Null facility eliminates the test lead resistance from the results. Livecircuit warnings are given by a flashing LED, buzzer and indication on thedisplay. 200mA test current complies with IEC61557.

Insulation Measurement
3 selectable test voltages : 250V, 500V, 1000V. Measurement up to 2000MΩ
(auto ranging) with quick testing. Auto discharge function ensures that circuits are not hazardous after testing. A red LED gives warning of high voltage
output during testing and discharging of the circuit. Live circuit warnings are given
by a flashing LED, buzzer and indication on the display. High test current complies with IEC61557.

Loop Impedance Measurement
A patented (ATT) low current loop impedance test enables high accuracy loop measurements (resolution 0.01Ω) andquick testing without tripping RCDs rated at 30mA or more.
A high current alternative is selectable for even higher accuracy and instantaneous results. The KEW6016 allows also forphase to phase loop tests from 100 to 500V and 2-wire or 3-wire measurements.

PSC / PFC Measurement
Prospective Short Circuit Current (PSC) and Prospective Fault Current (PFC) are automatically calculated and shownon the display in auto ranging. As loop testing, the function has low and high test current options, phase to phase
tests from 100 to 500V and 2-wire or 3-wire measurements.

RCD Measurement
KEW6016 has a comprehensive RCD test feature for RCD type AC (Alternative Current), RCD type A (Pulsating DirectCurrents), General and Selective (delayed).
Measures at 1/2 x, 1x, 5x of nominal RCD current. It also has Ramp test and Auto test functions where all results areshown on one screen. Touch voltage limit can be selected for 25V or 50V depending on application.

Earth Measurement
Using the classical Volt-Amper method with two auxiliary earth spikes and without external power source. All test leads andspikes are supplied as standard accessories.

Phase Rotation
KEW6016 can check the phase rotation of three phase lines with clear indication of the sequence on the display.

Voltage Measurement
In addition to the voltage measurement from 25 to 500V AC and DC, this function gives also the Frequency of the voltage under test.

WE OFFER MORE.....
Demo Panel KEW 8205AThis panel demonstrates all the functions of KEW6016. Also, other Kyoritsu single function
testers like Insulation, Continuity, Loop, RCD or Earth testers can be demonstrated withthis panel. Useful for trainings, seminars, exhibitions and schools of electrical engineers.

A PRACTICAL GUIDETO THE MEASUREMENTS ON ELECTRICAL INSTALLATIONSIn accordance with IEC 60364-6 standard
With KEW6016 and the practical guide, you can test in accordance with IEC 60364-6.
You may download the guide from Kyoritsu web-site: http://www.kew-ltd.co.jp

Multi LanguagesPRACTICAL
GUIDE

▲

6 languages (English, French, Italian, Spanish, Russian, Polish) are available for LCD display, Instruction manual and PC software.

● KEW 6016 SPECIFICATIONSContinuity
Range
Open circuit voltage (DC)Short circuit current

RCD
20/200/2000W (Auto-ranging)
5V±20%(*1)> 200mA

Insulation resistance
Range
Open circuit voltage (DC)Rated current

20/200MW
20/200/2000MW
20/200MW
20/200/2000MW
20/200MW

Accuracy
20/200/2000MW
Loop impedance
Function
L-PE, L-PE (ATT):
L-N:
L-L:
20W:
Nominal test current at 0W 200W:
external loop:
2000W:Magnitude/Duration(*2)
L-N:
N-PE:
Range
L-PE, L-N / L-L:
Accuracy
L-PE (ATT):PSC (L-N/L-L) / PFC (L-PE)
Function
PSC:
Rated voltage
PFC, PFC (ATT):
Nominal test current at 0W PSC:
PFC:
external loop:
Magnitude/Duration(*5)
PFC (ATT):
Range
Rated voltage

Accuracy
RCD
Function
Trip current setting

Trip current Duration±0.1W (0~0.19W )±2%rdg+8dgt (0.2~2000W)

Accuracy

X1/2, X1,Uc:
X5:
Ramp:

20/200/2000MW (Auto-ranging)
250V+25% -0%
500V+25% -0%, 1000V+20% -0%1mA or > @ 250kW
1mA or > @ 500kW, @ 1MW±2%rdg+6dgt (0~19.99MW)±5%rdg+6dgt (20~200MW)±2%rdg+6dgt (0~199.9MW)±5%rdg+6dgt (200~2000MW)
L-PE, L-PE (ATT), L-N / L-L
100~260V (50/60Hz)
100~300V (50/60Hz)
300~500V (50/60Hz)
6A/20ms
2A/20ms
15mA/500ms
6A/60ms
10mA/approx. 5s
20/200/2000W Auto-Ranging (L-N < 20W)
±3%rdg+4dgt*3 ±3%rdg+8dgt*4±3%rdg+6dgt*3 ±3%rdg+8dgt*42000ms
G:550ms / S: 1000ms
410ms

Ramp:

Goes up by 10% from 20% to 110%
G:300ms/S:500msX10 times

X1/2, X1, X5,Ramp, Uc:

230V+10%-15% 50/60Hz

Rated voltage
Auto:

AC TypeAccuracy

Trip current

A TypeEarthRange
20W:
200/2000W:

Accuracy
Phase Rotation
Rated Voltage
Remarks
Volts
Function
Rated voltage
Measuring range
Accuracy
GeneralApplicable standards

PSC, PFC, PFC (ATT)
100~500V 50/60Hz
100~260V 50/60Hz
6A/20ms
6A/20ms, 2A/20ms, 15mA/500ms
L-N: 6A/60ms, N-PE: 10mA/approx. 5s
2000A/20kA Auto-Ranging
PSC/PFC accuracy is derived from measured loop impedance
specification and measured voltage specification
X1/2, X1, X5, Ramp, Auto,Uc
10/30/100/300/500/1000mA
10/30/100mA
10/30/100/300/500mA

X1/2:
X1:
X5:

Power source
Dimensions
WeightAccessories

Optional

*1:
*2:
*3:
*6:

Depending on the accuracy at each function. Measurement
sequence:
X1/2 0°→X1/2 180°→X1 0°→X1 180°→X5 0°→X5 180°
Measurements with x5 are not carried out for RCDs withnominal current of 100mA or more.
X1/2: -8%~-2%, X1, X5: +2%~+8%, Ramp: ±4%
X1/2: -10%~0%, X1, X5: 0%~+10%, Ramp: ±10%
Uc: +5%~+15%rdg±8dgt
20/200/2000W Auto-Ranging±3%rdg+0.1dgt±3%rdg+3dgt (Auxiliary earth resistance 100±5%)50-500V 50/60Hz
Correct phase sequence: are displayed “1.2.3” and mark
Reversed phase sequence: are displayed “3.2.1” and markVolts
25~500V, 45~65Hz
25~500V±2%rdg+4dgt

Frequency
45~65Hz±0.5%rdg+2dgt

IEC 61010-1 CAT.Ⅲ 300V(500V L to L) Pollution degree 2
IEC 61010-031, IEC 61557-1,2,3,4,5,6,7,10
IEC 60529(IP40), IEC 61326(EMC)
LR6 × 8136(L) × 235(W) × 114(D)mm
1350g (including batteries.)
7218(Mains cord)(*6) 7196(Remote test lead)
7188(Distribution board fused test leads)
7228(Earth test leads) 8032(Auxiliary earth spikes)
8212-USB(USB adaptor with KEW Report(Software))Cord case, 9142(Carrying Case), 9121(Shoulder strap), Buckle,Battery, Instruction manual
8212-RS232C(RS232C adaptor with KEW Report(Software))Voltages are output when measurement resistance is under 2100 ohm.
at 230V*4: Other voltages except for *3
230V+10%-15%*5: at 230V
7187(Main cord [UK]), 7221(Main cord [SA]), 7222(Main cord [AU])EVERYTHING YOU NEEDﾉ....
KEW6016 comes complete with everything you need for testing an electricalinstallation according to the IEC 60364-6.
A full set of new fused test leads is included. The practical slim leads offer hands
free operation via a test button located within the probe itself. The safety test
leads have an overmould for maximum grip and are extremely robust.
Two auxiliary spikes and test leads for earth measurements, PC software fordownloading and interpreting of data and interface cable are included.
The instrument also comes in a soft carry case, a quick reference guide locatesinside the integrated case lid, and it is supplied with a calibration certificate.

*i>ÃiÊÀi>`ÊÌ iÊ¸->viÌÞÊ7>À}Ã¸ÊÊÌ iÊÃÌÀÕVÌÊ>Õ>ÊÃÕ««i`ÊÜÌ ÊÌ iÊÃÌÀÕiÌÊÌ ÀÕ} ÞÊ>`ÊV«iÌiÞÊ

iÊÃ>viÌÞÊÀÕiÃÊV>ÊV>ÕÃiÊvÀi]ÊÌÀÕLi]ÊiiVÌÀV>ÊÃ V]ÊiÌV°ÊÊ/ iÀivÀi]Ê>iÊÃÕÀiÊ->viÌÞÊ7>À}ÃÊ\ vÀÊVÀÀiVÌÊÕÃi°ÊÊ>ÕÀiÊÌÊvÜÊÌÌÊ«iÀ>ÌiÊÌ iÊÃÌÀÕiÌÊÊ>ÊVÀÀiVÌÊ«ÜiÀÊÃÕ««ÞÊ>`ÊÛÌ>}iÊÀ>Ì}Ê>Ài`ÊÊi>V ÊÃÌÀÕiÌ°ÀÊµÕÀiÃÊÀÊÀ`iÀÃÊ\)N CONSIDERATION OF THE ENVIRONMENT SOY INK AND RECYCLED PAPER WERE USED IN THIS PUBLICATION/ iÊVÌiÌÃÊvÊÌ ÃÊi>viÌÊ>ÀiÊÃÕLiVÌÊÌÊV >}iÊÜÌ ÕÌÊÌVi° 7Èä£È£ Ê-i«°ÊänÊ