IEC 61010-1 CAT.Ⅲ 300V
■Accessories
Discontinued
Product■Model 7098

■Model 8210

Discontinued
Product■Model 7099
■Model 8032
■Model 7025

10 in1

Discontinued
Product■Model 7109
■Model KSLP5(Model 7132)

Discontinued
ProductAccessories

7098 Distribution board test leads(1 set)for Loop/PSC/RCD testing
7099 Moulded plug test leads(1 set)for Loop and RCD testing at sockets
7025 Lead for insulation and continuity testing7109 Earth resistance test leads(1 set)
KSLP5 external earth probe
8032 auxiliary earth spikes
8210 lrDA Adaptor
Durable Holdall
Accessory pouch
Battery 8×1.5V R6 or LR6
Instruction Manual■Durable Holdall &Accessory pouch

6050

Earth Resistance Tester
Phase Rotation Tester
Mains Frequency Measurement
Mains Voltage Measurement

6010A

6011A

6015

Discontinued
ProductContinuity Testing

Loop Impedance TestingContinuity TestingContinuity Testing

Rated Voltage 230V + 10%‐15% (20/200/2000 Ω)

Measuring Ranges

Measuring Ranges

Measuring Ranges

400V + 10%‐15% (200/2000 Ω)

20/200Ω Autoranging

20/200/2000 Ω Autoranging

20/200/2000 Ω Autoranging

Voltage Measuring Range 100-260V

Open Circuit Voltage ＞ 4V

Open Circuit Voltage ＞ 6V

Open Circuit Voltage ＞ 6V

100-440V (200/2000 Ω ,200A)

Short Circuit Current ＞ 200mA DC

Short Circuit Current ＞ 200mA DC

Short Circuit Current ＞ 200mA DC

Impedance Range 20/200/2000 ΩAccracy ± (3%rdg+3dgt)

Accuracy ± (1.5%rdg+3dgt)

Accuracy ±(1.5%rdg+3dgt)

Nominal Test Current

Insulation Testing

Insulation Testing

Insulation Testing

15mA (200/2000 Ω ) 3A (20 Ω)

Measuring Ranges

Measuring Ranges

Measuring Ranges

20/200MΩ Autoranging

20/200M Ω Autoranging

20/200/2000MΩ Autoranging

20Ωrange±(3%rdg+8dgt)200Ωrange±(3%rdg+8dgt)

Test Voltage 500V DC

Test Voltages 250V,500V and 1000V DC

Test Voltages 250V,500Vand 1000V DC

2000Ωrange±(3%rdg+8dgt)

Output Voltage on Open Circuit 570V DC ± 6%

Output Voltage on Open Circuit

Output Voltage on Open Circuit

PSC Testing

Rated Current ＞ 1mA

250V+40%-0% 500V+30%-0% 1000V+20%-0%

250V+20%-0% 500V+20%-0% 1000V+20%-0%

Rated Voltage 230V + 10%‐15% (20/200/2000 A)

Accuracy ± (3%rdg+3dgt)

Rated Current ＞ 1mA

Rated Current ＞ 1mA

400V + 10%‐15% (200 A)

Loop Impedance Testing

Accuracy ± (1.5%rdg+3dgt)

Accuracy 0~100M Ω ±(2%rdg + 3dgt)
100M Ω ~ 2000M Ω ±(2%rdg + 4dgt)

Base Accuracy

PSC Ranges 200/2000A/20kA

Rated Voltage 230V AC ± 10%@ 50Hz

Loop Impedance Testing

Accuracy

Voltage Measuring Range 100-250V @ 50Hz

Rated Voltage 230V AC + 10%‐15%@ 50Hz

Loop Impedance Testing

PSC accuracy derived from measured loop impedance

Impedance Ranges 20/2000 Ω

Voltage Measuring Range 100-250V AC @ 50Hz

Rated Voltage:230VAC ± 10%@ 50Hz

Nominal Test Current

Impedance Ranges 20/200/2000 Ω

Voltage Measuring Range 100-250V AC @ 50Hz

Nominal Test Current

Impedance Range 20/200/2000 Ωspecification and measured voltage specification

24A (20 Ω range) 15mA (2000 Ω range)

RCD Testing
Rated Voltage 230V + 10%‐15%

Accuracy

3A(20Ωrange),15mA(200Ωrange),15mA(2000Ωrange)

Nominal Test Current

Trip Current Settings

20Ωrange±(3%rdg+8dgt)2000Ωrange±(3%rdg+8dgt)

Accuracy

25A(20 Ω range) 1.2A(200 Ω range) 1.2A(2000 Ω range)
Test Period 1/2 cycle(10ms)

RCD × 1/2

: 10,30,100,300,500,1000mA

RCD Testing

20 Ω range ±(3%rdg+4dgt) 200 Ω range ±(3%rdg+8dgt)

RCD ×1: 10,30,100,300,500,1000mA

Rated Voltage 230V AC ± 10%@ 50Hz

2000 Ω range ±(3%rdg+4dgt)

Base Accuracy

RCD ×5: 10,30,100mA

Trip Current Settings

PSC Testing

20 Ω range ±(3%rdg+5dgt)

RCD ×DC: 10,30,100,300,500mA

RCD × 1/2 : 5,15,50,150,250mA

Rated Voltage 230V AC + 10%‐15%@50Hz

RCD ×Ramp : 10,30,100,300,500mA

RCD × 1

: 10,30,100,300,500mA

RCD fast

PSC Testing
Rated Voltage 230VAC ± 10%@ 50Hz

PSC Ranges 200A,2000A and 20kA

PSC Range 2000A and 20kA: 150mA

Accuracy

RCD × 1/2,× 1,: ×DC: 1000ms

Trip Current Duration RCD × 1/2 × 1 : 2000ms

PSC accuracy derived from measured loop impedance

Accuracy

RCD × 5,DC (500 and × 1 1000mA):200ms

RCD fast :40ms

specification and measured voltage specification

PSC accuracy derived from measured loop impedance
specification and measured voltage specification

Trip Current Duration

Contact Voltage Reading

Accuracy

RCD Testing

1.0-100.0V

Trip Current ± 3% of test current at 230V

Rated Voltage 230V AC + 10%‐15%@ 50HzRCD Testing

General Specifications

Trip time ± (3%rdg+3dgt)

Trip Current Settings

Rated Voltage 230VAC ± 10%@ 50Hz

Safety Standard IEC 61010-1 CAT.Ⅲ 300V

General Specifications

RCD ×1/2 : 10,30,100,300,500,1000mA

Pollution degree 2

Safety Standard IEC 61010-1 CAT.Ⅲ 300V

RCD ×1: 10,30,100,300,500,1000mA

RCD × 1/2 : 10,30,100,300,500mA

Withstand Voltage 3700V AC for one minutePollution degree 2

RCD ×5: 10,30,100,300mA

RCD × 1

: 10,30,100,300,500mA

Power Source 8×1.5V R6P

Withstand Voltage 3700V AC for one minute(on × 5 range max current 1A)

RCD × 5: 10,30,100,300,500mA

Dimensions 186(L)×167(W)×89(D)mm

Power Source 8×1.5V R6P

Trip Current Duration RCD×1/2×1 : 2000ms,

Trip Current Duration RCD × 1/2 × 1 : 2000ms

Weight 980g approx including batteries

Dimensions 175(L)×115(W)×86(D)mm

RCD fast : 50ms

RCD × 5 : 200ms fast trip 50ms

Weight 780g approx including batteries

Accuracy

Accuracy

Trip current + 10%-0% of test current at 230V

Trip current + 10%-0% of test current at 230V

Trip time ± (1%rdg+3dgt)

Trip time ± (2%rdg+3dgt)

CE Marking:signifies conformance toEMC directive(89/336/EEC, 92/31/EEC, 93/68/EEC)
and Low Voltage directive(73/23/EEC, 93/68/EEC)

Trip Current Settings

General Specifications

Fault Current

Safety Standard IEC 61010-1 CAT.Ⅲ 300V

Rated Voltage 230V AC ± 10% @ 50Hz

Pollution degree 2

Fault Current Ranges 2000A and 20kA

Withstand Voltage 3700V AC for one minuteNominal Test Current 1.2A (2000A) 25A (20kA)

Power Source 8×1.5V R6P

Fault Current Accuracy

Dimensions 130(L)×183(W)×100(D)mm

Fault Current accuracy derived from measured loop impedance

Weight 1000g approx including batteries

specification and measured voltage specification
Earth ResistanceRanges 20Ω 200Ω 2000Ω

8205 KEW Demonstration Kit
Discontinued
ProductContinuity Tester
Insulation Tester
Loop Impedance Tester
RCD Tester
PSC Tester
Fault Current Tester

Discontinued
ProductKEW Demonstration Kit is alsoavailable from Kyoritsu to help
customers with sales promotion anduser training. This single unit can beused to give a demonstration of how
to use the Insulation/Continuity, Loopand RCD(ELCB) testers. For moredatails contact the overseas sales
department of KYORITSU.

Output Voltage 70V peak to peak square wave
Frequency 720Hz ± 5%Accuracy
20 Ω (±2%rdg + 5dgt) 200/2000Ω (± 2%rdg + 3dgt)
General Specifications
Safety Standard IEC 61010-1 CAT.Ⅲ 300V
Pollution degree 2
Withstand Voltage 3700V AC for one minute
Power Source 8×1.5V R6 or LR6Dimensions 105(L)×210(W)×240(D)mm

KEW Multifunction Tester

Discontinued
ProductThe top of the range KEW6015 canperform TEN separate test functions
and provides all the test functions
generally required to verify the safetyof electrical installations in fullcompliance with IEC61557.
The KEW6015 can store test results in
its on-board memory for later
downloading to a computer or printer.
Data can be downloaded via the Infra
Red Communication Port(IRDA Port).
The KEW6015 is software driven andguides the user through the test
functions required making
the instrument very easy to use.
It's numerous features include theability to conduct 3 or 4 wire earth
resistance tests, D-Lok technology toprevent tripping of most passive type
RCD's when loop testing andmains voltage and frequency
measurement.
■Features●Live circuit and wiring
connection check LED's●D-Lok technology on loop test range●On-board memory to store
test results●Downloading of data via IRDA port
●Auto power off●Back lit display●Polarity switch●Continuity Auto-null●Auto ramp and auto test on
RCD ranges●Designed to IP54 Rating●Complies with IEC61557

Discontinued
ProductKEW
Multifunction
Tester★Provide Basic, Medium & Sophisticated MFT series.
★Fully meet with the needs of convenient and fast
installation works.
★World top-ranking MFT's with the latest
standards and regulations.

Weight 1924g approx including batteries

Safety Warnings

Digital Multifunction Tester

Please read the Safety Warnings in the instruction manual supplied with the instrument thoroughly andcompletely for correct use.
Failure to follow the safety rules can cause fire, trouble, electrical shock, etc. Therefore,make sure to operate
the instrument on a correct power supply and voltage rating marked on each instrument.
The contents of this catalogue may be subject to change without notice.

DISTRIBUTOR
No.5-20 Nakane 2-chome, Meguro-ku, Tokyo,152-0031 Japan
Phone : 81- 3 - 3723 - 0131 Fax:81- 3 - 3723 - 0152
URL : http://www.kew-ltd.co. jp
E-Mail : info@kew-ltd.co. jp
Factories : Uwajima & Ehime
03.01F PRINTED IN JAPAN

6050

6010A

6011A

6015

3 in1

4 in1

5in1

10in1

The Model 6010A is an advanced
multifunction tester that can perform
all the testing required by IEC
364-6-61 with a single instrument.
It can be used to test insulation
resistance, continuity,loop impedanceand RCDs(ELCBs), four major test
requirement in the electrical testing
industry.This four-in-one instrument
eliminates the need to carryindividual single function testers.
Model 6010A is designed to IEC61010-1
and available with CE marking for
EMC compliance.

KEW COMBITESTER MODEL 6050 isNew Low Cost 3 in 1 MFT withMemory and Download Results to PC!

Discontinued
ProductIEC 61010-1 CAT.Ⅲ 300V

■Accessories
Moulded plug test leads

Model 7123(AU)Australian plug
Model 7124(UK)British plug(13A)
Model 7125(EU)European SHUKO plug
Model 7126(SA)South African plug

■Optional Accessory■Features●Kew CombiTester provides theultimate in Microprocessor technology
offering Loop and RCD tests with acomprehensive memory anddownloading test results.
●LOOP Impedance 200/2000 Ohm
range measurement is carried out
using a test current of 15mA.
This current will not trip
RCD's rated at 30mA or greater.
●High resolution for low Loop
Impedance (resolution of 0.01 Ohm)on 20 ohm range.
●Direct reading of Prospective
Short Circuit Current (PSC) andearth fault current.
●RCD tests at 0 and 180 degree
phase angle switch permits quick
tests and consistent readings.
●Test of a large kind of RCD's.
Standard, Selective, AC and A(DC sensitive breakers).
●Constant current source circuitryensures that a fluctuating mains
voltage does not affect the accuracyof readings.
●RCD Ramp test in order to indicate
the actual tripping current of RCD.
●Contact Voltage Reading (UC).
with two selectable contact voltage
limits (50 and 25v).
●Visual indication of correct wiring
status and automatic lock-out if
test resistors overheats.
●Designed to IEC 61010-1, IEC61557
and high protection degree IP54.

■Features

Discontinued
Product

IEC 61010-1 CAT.Ⅲ 300V
Accessories
KAMP10 Moulded plug test leads(1 set)

Continuity and Insulation Testing●Live circuit warning
LED:Immediate illumination of awarning lamp if the circuit under
test is live
Allows fast and safe discharge●Auto Discharge:of electrical energy stored incapacitive circuits following
testing●Integral overload protection on internal continuity andinsulation test circuitry●Autoranging●500V DC test voltage for insulation tests●200 mA short circuit current on continuity ranges
Loop and RCD Testing●Wiring connection
check LED's:●Overheat Protection:

for Loop and RCD testing at sockets
7025 Lead for insulation and continuity testing
Pouch for Test Leads
Shoulder Strap, Shoulder PadBattery 8×1.5V R6P
Instruction Manual
Optional Accessories
OMA DIEC Distribution board or lighting circuit
test leads(1 set) for LOOP and RCD testing●Supply Voltage
Indication:●Low Current Loop
Impedance Test:

Accessories

Model 7121(Distribution board test leads)

Moulded plug test leads(1 set)for Loop and RCD testing at socket
Pouch for Test Leads

Discontinued
Product■Accessories
Continuity and Insulation Testing●Auto Data Hold:
LOOP and RCD Testing●Phase Angle
Selection:

Shoulder Strap
Battery 8×1.5V R6P
Instruction Manual
Optional Accessories
7121 Distribution board test leads(1 set)

Immediate indication if wiring of
circuit under test is correct
Rapid detection of overheatingof internal circuitry, display of
warning signal to user andautomatic halting of
measurement
Can display supply voltage value
in loop impedance mode
Short duration low test current
will prevent tripping of many
passive RCDs
Retains displayed reading for set
period following test
Selection of test current from+ve or -ve half cycle ofwaveform can prevent tripping ofpolarised type RCDsKew Report)

Model 8212 Data Communication Adapter"KEW Report" Software for report"KEW Report" transfer
measurement data from the COMBITESTER to a PC.
Print or import the data into your report.
COMBITESTER are connected
to a PC via an optically-isolated
RS232C interface cable(Model 8212 Data Communication Adapter)3 in 1Loop Impedance Tester
RCD Tester
PSC Tester

Accessories■Accessories■KSLP5 (Model 7132)■Model 7025

■KAMP10(Test Leads)(Test Leads with IEC Connector)■Optional Accessory
Discontinued
Product4 in 1

Continuity Tester
Insulation Tester
Loop Impedance Tester
RCD Tester■OMA DIEC (Model 7133)

KAMP10 Moulded plug test leads(1 set)for Loop and RCD testing at sockets
7122 Lead for insulation and continuity testing
KSLP5 external earth probe
Pouch for Test Leads
Shoulder Strap
Battery 8×1.5V LR6Instruction Manual
Optional Accessories
OMA DIEC Distribution board or lighting circuit
test leads(1 set) for LOOP/PSC/RCD testing■KAMP10■Model 7122 Test Leads

8212 Data Communication Adapter( including Software

IEC 61010-1 CAT.Ⅲ 300V

The Model 6011A can perform FIVE
separate test functions; insulation,
continuity, earth loop impedance,
prospective short circuit current
and RCD trip testing in full compliancewith IEC61557.
It features a triple voltage insulation
test at 250V,500V and 1000V andcan measure mains voltage when
operating in the loop impedanceand RCD testing functions.
Additional features include a phaseangle selector switch for RCD testing,a polarity switch for switchingpolarity during continuity andinsulation tests and a continuity
null to facilitate automatic
subtraction of test lead resistancefrom continuity measurements.
■Features●Live circuit and wiring
connection check LED's●Triple range insulation tests
at 250,500 and 1000V●Triple range loop and PSC tests 15mAtests current at 200Ω/2000Ω range(No RCD Trip)●Phase angle selection on
RCD range●Polarity switch for continuity tests●Auto-null facility for continuity tests●DC test for DC sensitive RCDs
●Earth probe for loop impedance
testing on extraneous metalwork

Discontinued
Product■Optional Accessory
Discontinued
Product■OMA DIEC (Model 7133)

5 in 1

Continuity Tester
Insulation Tester
Loop Impedance Tester
RCD Tester
PSC Tester