Lot production type purgemeter for measuring gases
and liquids

P-010
Purgemeter■ OUTLINEThe P-010 is a lot production version of the popular P-100. Althoughmeasuring ﬂuids are limited to water, air, and nitrogen, the P-010 is ideal
if more than 10 units with the same specifications are required. Laser
inscription of graduations on the convex surface makes reading easier.

■ MAJOR APPLICATIONSGeneral purpose (for small ﬂow rates, water lines and air/nitrogen lines)■ STANDARD SPECIFICATIONS
Gases(air/nitrogen)
Water
Min. 0.1 to 1 L/min（nor）Air
Max. 5 to 50 L/min（nor）
Min. 5 to 50 mL/min
Water
Max. 0.4 to 2 L/min
Measuring range
・When selecting a ﬂow range, refer to "Index &Quick Reference for P Series Purgemeters"(TG-S0001).
・When selecting a ﬂow range,refer to thestandard ﬂow rate table.
Range ability
10:1(10:2 for some models)
Indication accuracy±5％ F.S.
Fluid pressure
Max. 0.8MPa
Fluid temperature
Max. 120ºC（Depends on the gasket material）
Wetted part SCS14Tapered tube Heat-resistant glass
Nitrile rubber（NBR）（Max. 80℃）
Fluoro rubber (Max. 120ºC)
Gasket・The maximum ﬂuid temperature for each
Material
gasket material may vary depending on usage
and ambient conditions.
Side plate
A6063-T5
Front panel
ABSScale plate
Polycarbonate
Rc : 1/4"（Standard）
Process Connection
NPT : 1/4"
Mounting
Lock nut mount on the panel front (Standard)
Massapprox. 0.35ｋ
ｇMeasuring ﬂuid

■ BASIC MODEL CODE
P-010

Model Code‒ □ 0 ‒ 4 □ ‒ □ □‒ LValve‒ U
Alarm output
Wetted parts
material

Descriptions
Inlet (gas foratmospheric pressurescale)
Outlet (gas forpressure scale or fornegative pressure onthe secondary side)
Not provided

0‒ 4Notes
Refer to "Index& QuickReference
for P Series
Purgemeters"(TG-S0001).

SCS14/SUS304

Nitrile rubber（NBR）(Standard)
F
Fluoro rubber
ZSpecial‒ RRc（Standard）
Lock nut
Process connection type
mounting on a
‒ N
NPT female
panel front
2 1/4"（Standard）
Process connection size
Z SpecialN

Gasket material■ ALARM CONTACT AND OUTPUT
TypeGeneral
Reed switch type alarm unit
UL-approved
PAU optical alarm unit

Availability
×××

For CE approval, refer to "Index & Quick Reference for P Series
Purgemeters" (TG-S0001).

■ STANDARD FLOW RATE TABLE
When the ﬂuid pressure is other than 0 MPa, refer to "Index & Quick Reference for P
Series Purgemeters" (TG-S0001).

Air (0ºC, 0 MPa)
0.1 to 1L/min（nor）
0.3 to 3 L/min（nor）
0.5 to 5 L/min（nor）
1 to 10 L/min（nor）
2 to 20 L/min（nor）
3 to 30 L/min（nor）
4 to 40 L/min（nor）
5 to 50 L/min
（nor）

Water
5 to 50mL/min
10 to 100 mL/min
20 to 200 mL/min
30 to 300 mL/min
50 to 500 mL/min
0.1 to 1L/min
0.3 to 1.5
L/min
0.4 to 2
L/min

Although other flow rate specifications are available, some flow ranges
need to be changed. Contact us for details.

TG-F1106-2ERevisedMay 2016 K
1st edition Jun 2014 K

P-010Purgemeter■ ORDERING INFORMATION
Basic model code

Items to be speciﬁed①②③④⑤
Fluid name − Flow range − Press. − Temp. − Mounting option

Ｐ-010- □ 0- □□ - □□（Select in the model code table.）−⑥
Other options（Refer to "Index & Quick Reference for P Series Purgemeters" (TG-S0001).）■ DIMENSIONS● Standard typeP − 010 − U0 − 4N −□□■ Lock nut mounting on the panel front■ Valve at the outlet(30)(MAX.27)

(31)

17
2-φH

2-D
OUTO C▼ ▼115

115

(140)

2-dIN

Panel cut● Panel cut size
Connection size
Dd
Rc1/4
M18
1/4NPT
M18

Diameter (mm)H2020

● Standard material
Parts name
Wetted part
Tapered tube
Float ＊Gasket
Side plate
Front panel
Scale plate◯◯◯◯

Standard material
Available material
SCS14/SUS304
̶Heat-resistant glass
̶SUS304, Glass, PTFE, Ruby
̶NBR
Fluoro rubber
A6063-T5̶
ABS̶Polycarbonate
̶Parts marked ○ come in contact with the measuring ﬂuid.
＊ We will select a proper material for your application.

· SUS316 may be used instead of SUS304 due to production reasons.
· ASTM or AISI materials equivalent to JIS materials may be used due
to production reasons.

* Specification is subject to change without notice.

Head Office : Shiba Toho Building, 1 – 7 – 24 Shibakoen, Minato-ku, Tokyo 105 – 8558
Tel : +81-3 – 3431 – 1625 (KEY) ; Fax : +81-3 – 3433 – 4922
e-mail : overseas.sales@tokyokeiso.co.jp ; URL : http://www.tokyokeiso.co.jp
2

TOKYO KEISO CO., LTD.

TG-F1106-2E