Compact, low-cost, quick-delivery purgemeter with
POM body for measuring gases and liquids

XPPurgemeter■ OUTLINE
The body of the standard compact purgemeter is made ofPolyacetal(POM).
This purgemeter can reduce your stock with a quick delivery and lowcost.

■ MAJOR APPLICATIONSGeneral purpose, quick delivery■ STANDARD SPECIFICATIONS
Measuring fluidGasesLiquids

Min. 0.1 to 1 L/min（nor）
Max. 2 to 20 L/min（nor）
・0 MPa (1atm), 20 ℃
Measuring range(Standard)
Min. 0.02 to 0.1 L/min
Water
Max. 0.2 to 1 L/min
・1.0 g/cm3, 1.0 mPa・s
Min. 0.1 to 1 L/min（nor）
Air
Max. 2 to 20 L/min（nor）
Min. 0.02 to 0.1 L/min
Measuring range
Water
Max. 0.2 to 1 L/min(Optional)
・When selecting a flow range, refer to "Index& Quick Reference for P Series Purgemeters"(TG-S0001).
Range ability
10:1(10:2 for some models)
Indication accuracy±5％ F.S.
Fluid pressure
Max. 0.4 MPa
Fluid temperature
Max. 50℃
Wetted part Polyacetal（POM）
Tapered tube Heat-resistant glass
MaterialGasket
Fluoro rubber
Cover
Polycarbonate
Process Connection
Rc: 1/4"（Standard）
Mounting
Screw mount on the panel front (Standard)
Massapprox. 0.1 kg
Air

The maximum fl uid temperature for each material may vary dependingon usage and ambient conditions.

■ BASIC MODEL CODE
Model CodeDescriptions□ □ — □ □L 1
Water 0.02 to 0.1 L/min
L 2
Water 0.04 to 0.2 L/min
L 3
Water 0.06 to 0.3 L/min
L 4
Water 0.1 to 0.5 L/min
L 5
Water 0.2 to 1 L/min
Flow
G 1
Air 0.1 to 1 L/min (nor)
range
G 2
Air 0.3 to 3 L/min (nor)G 3
Air 0.6 to 6 L/min (nor)
G 4
Air 1 to 10 L/min (nor)G 5
Air 2 to 20 L/min (nor)0 0Special flow range— 0
Not provided
Inlet (gas for atmospheric— LValvepressure scale)
Outlet (gas for pressure— Uscale)
0 Not provided
Alarm output
E PAU alarm unit provided
XP-Notes

Refer to "Index& Quick
Reference forP Series
Purgemeters"(TG-S0001).

■ ALARM CONTACT AND OUTPUT
TypeGeneral
Reed switch type alarm unit
UL-approved
PAU optical alarm unit

Availability
××◯

For CE approval, refer to "Index & Quick Reference for P Series
Purgemeters" (TG-S0001).

TG-F1105-1ERevised
Jan 2016 K
1st edition Jul 2014 K

XPPurgemeter■ ORDERING INFORMATION
Basic model code

Items to be specified①②③④⑤
Fluid name − Flow range − Press. − Temp. − Mounting optionXP- □□ - □□（Select in the model code table.）−⑥
Other options（Refer to "Index & Quick Reference for P Series Purgemeters" (TG-S0001).）■ DIMENSIONS● Standard typeXP －□□－ L0● Standard typeXP －□□－ U0■ Screw mounting on the panel front■ Valve at the inlet

2-M4
Depth 52-M4
Depth 5（MAX.23）

32258030

4.5φ2×25

Rc1/4IN2824

φ

2×80

30
2580112

Rc1/4
ＯＵＴ112

Rc1/4
ＯＵＴ30

■ Screw mounting on the panel front■ Valve at the outlet

Rc1/4IN

528（MAX.23）

32

5Panel cut● Standard material
Parts nameStandard material
Body○ Polyacetal (POM)
Tapered○ Heat-resistant glasstube
Float※○ SUS316, Glass
Packing○ Fluoro rubber
Valve○ Polyacetal (POM)
Cover
Polycarbonate

Available material
−−
Ruby
−
−
−

Parts marked“○”come in contact with the measuring fluid.
※We will select a proper material for your application.

* Specification is subject to change without notice.

Head Office : Shiba Toho Building, 1 – 7 – 24 Shibakoen, Minato-ku, Tokyo 105 – 8558
Tel : +81-3 – 3431 – 1625 (KEY) ; Fax : +81-3 – 3433 – 4922
e-mail : overseas.sales@tokyokeiso.co.jp ; URL : http://www.tokyokeiso.co.jp
2TOKYO KEISO CO., LTD.

TG-F1105-1E