SE series
IsolatorSPEC. SHEET(with indication function)■ ModelModel:SE1A

SE1A - □ - □Socket
1: Finger-safe(For Y terminal)
2: For Ring terminal
Power supply
0: 100 to 240V AC
1: 24V AC/DC■ How to orderSpecify the model (e.g.) SE1A-1-0
Default value
Input
4 to 20mA DC
Output4 to 20mA DC■ Output specification■ Accessories (sold separately)
Name

Model
Spec.
RES-S01-050
500.1%Ring
RES-S01-100
1000.1%terminal RES-S01-200
2000.1%Shunt
RES-S01-01K
1k0.1%
resistor
RES-S06-050
500.1%RES-S06-100
Y
1000.1%terminal RES-S06-200
2000.1%RES-S06-01K
1k0.1%
Communication cable for the console software: CMB-001

When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value
will not be negative.
DC current
Output range

Allowable loadresistance

4 to 20mA DC
0 to 20mA DC
0 to 12mA DC
0 to10mA DC
1 to 5mA DC

700
700
1.2k
1.2k
2.4k

or lessor lessor lessor lessor less

Zeroadjustment
range-5 to 5%
0 to 5%0 to 5%0 to 5%-5 to 5%Span
adjustment
range
95 to 105%
95 to 105%
95 to 105%
95 to 105%
95 to 105%

Zeroadjustment
range
0 to 5%0 to 5%-5 to 5%0 to 5%Span
adjustment
range
95 to 105%
95 to 105%
95 to 105%
95 to 105%

DC voltage■ Input specification

Output range

Allowable load
resistance

0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

100 or more
500 or more
500 or more
1k or more

DC current
Connect a shunt resistor (sold separately) between input terminals

Input
4 to 20mA DC
0 to 20mA DC0 to 16mA DC
2 to 10mA DC
0 to 10mA DC
1 to 5mA DC
0 to 1mA DC

Shunt resistor
50
1002001k

SE1A■ Performance■ Settings

Accuracy (When ambient temperature is 23 ):Input: Within 0.1%
Output: Within 0.1%
Indication accuracy: Within input accuracy 1 digit
Input sampling period: 25ms, 125ms, 250ms (Selectable by keypad)
Response time: 65ms (typ.) (0→90%) (Input sampling period 25ms)
225ms (typ.) (0→90%) (Input sampling period 125ms)
425ms (typ.) (0→90%) (Input sampling period 250ms)(Selectable by keypad)
Temperature coefficient: 0.015%/ or lessInsulation resistance: 10M or more, at 500V DC(Input – Output – Power supply)
Dielectric strength: 2.0kV AC for 1 minute(Input – Output – Power supply)

Function keys
(1) UP Key: Increases the numeric value,(2) DOWN Key: Decrease the numeric value,(3) MODE Key: Selects the setting mode,(4) SUB-MODE Key: Turns the displays ON again when they
are in OFF status.
(The UP, DOWN or MODE Key also turns the displays ON again
when they are in OFF status,)■ Displays and indicators
Input display: Indicates the input value
Indication of -2000 or less (for DC input):
The minus (-) sign and input value light alternately.
Indication of 10000 or more:The lower 4 digits flash.
Under range: “” flashes on the input display.
Over range: “” flashes on the input display
Warm-up indication: For approx. 3sec. after the power to theinstrument is turned on, the input type isindicated on the input display, and Output
type is indicated on the Output display.
Output display: Indicates output volume in percentage (%) form.
Power indicator: The green LED lights when the power to theinstrument is turned on.

■ General structure
Case: Flame-resistant resin, Color: Light gray
Front panel: Membrane sheet
Setting: By the front keypad
Connector for console software: Only for CMB-001
Indication: Input display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Output display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Power indicator: Green LED■ Solderless terminal
Y type■ Circuit configuration and terminal arrangementmA DC+Console connector-123■ Installation specifications
Power supply: 100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz
Allowable voltage range: 85 to 264V AC, 20 to 28V AC/DCPower consumption: Approx. 6VA
Ambient temperature: -5 to 55
Ambient humidity: 35 to 85%RH (Non-condensing)
Mounting: DIN rail mounting
External dimensions: W30×H88×D108mm (including the socket)
Weight: Approx. 190g (including the socket)

Ring type

Input
circuit

Display/Setting
key circuit+Powercircuit

1314-

Outputcircuit9+

10-Powersupply

Insulation circuit

4CPU
5Output

6117128■ External dimensions (Scale: mm)■ Attached functions
Auto-light function: Display brightness is controlled in accordancewith the surrounding area. Unnecessary brightness is reduced,saving energy.
Power failure countermeasure: The data is backed up in nonvolatile IC memory.
Self diagnosis: The CPU is monitored by a watchdog timer, andwhen an abnormal status is found on the CPU, the unit isswitched to warm-up status with tuning all outputs off.

■ Environmental specification
RoHS directive compliance

SHINKO TECHNOS CO., LTD.

Tel: 81-72-727-6100

URL: http://www.shinko-technos.co.jp/e/ E-mail: overseas@shinko-technos.co.jp

20120106