Remote RadEye HR Product Datasheet

Remote RadEye™ HR
X-Ray Detector
Overview

Key Features


Active area 33.0 x 24.9 mm

20µm resolution



0.7 fps max. frame rate

Sensor weighs 0.2kg

USB interface with 12-bit
digital video output

Supports x-ray energies as
low as 5keV and up to 90kVp



Ready-to-run software anddrivers

Applications
Industrial inspection,
biomedical and scientific

The Remote RadEye HR x-ray detector is a slim, lightweight,
rugged solution for high-resolution radiation imaging. The detector
is suitable for industrial inspection applications where images aretaken in tight or difficult-to-reach spaces. This revolutionary x-ray
camera is a cost-effective imaging solution for NDT/industrial
inspection, scientific research such as x-ray crystallography, andgeneral radiography applications.
Each detector features a rugged aluminum enclosure with astainless steel cover and a carbon-fiber or Beryllium window thatshields the sensor against ambient light and protects the sensitiveelectronics from accidental damage.
A Gd2O2S scintillator screen, placed in direct contact with thephotodiode array, converts incident x-ray photons to light, which inturn is detected by the photodiodes. A model featuring a Beryllium
entrance window can be used in low-energy applications down to~5keV.
The detector is compatible with our ShadoCam image acquisition
software, and is available with programming examples and SDKs
for custom application software development.

Remote RadEye HR Product Datasheet

Specifications
Sensor Specifications
Resolution

RadEye HRUnits

20µm

Number of Pixels

1650 x 1246

Active area

33.0 x 24.9mm

Avg. dark current (23°C) (1)

6ADU/s(2)Read noise (rms)

2ADU

Dynamic range

2000:1Digitization

12bitsConversion gain155

elec/ADUSensor data rate

5000

kHz

Readout period(3)

680

ms

Image transfer to PC

0.7

sec

Weight of sensor head(4)0.2kg

Operating temperature

0 to 50°C

Storage temperature-10 to +65°C

Humidity (non-condensing)

10 to 80% R.H.

Camera Module – Direct USB, 2m

Generaldark current doubles approx. every 8°C
ADU = Analog-Digital Unit = 1 LSB (Least Significant Bit)
(3)
time required to transfer image from sensor to camera memory(4)
not including sensor cable
(1)

(2)

www.teledynedalsa.com/ndt

2

Remote RadEye HR Product Datasheet

Resolution
The actual Modulation Transfer Function (MTF) for various scintillator options is shown in the followingcharts. A thicker phosphor screen will produce more signal, but at the expense of high-frequency contrast.
Please refer to our application note AN07 for more information on scintillator performance and tradeoffs.
RadEye HR MTF

1.0

0.8
CsI
Min-R

0.6

0.4

0.2

0.0
048

121620lp/mm

Ordering Information
The RadEye HR can be ordered in several configurations (see table below). The default scintillator option isCarestream Min-R® 2190. Additional scintillator options may be available on request.
P/NDescription

Notes

RM1426-02

RadEye HR, Min-R 2190

std. model, 10-90 kV energy range

RM1426-06

RadEye HR, CsI scintillator

higher spatial resolution

RM1426-08

RadEye HR, Be window

for low-energy applications

RM1426-09

RadEye HR, CsI + Be window

low energy + high resolution

www.teledynedalsa.com/ndt

3

Remote RadEye HR Product Datasheet

Mechanical Drawings
50.0 mm
5.3 mm36.4 mm

3.0 mm27.4 mm40.0 mm(1246,1650)

pixel (1,1)

12.7 mm

3.0 mmM2.5 mounting hole (4x)Remote RadEye HR Sensor Head

Contact Information
Teledyne Rad-icon Imaging Corp.
765 Sycamore Drive, Milpitas, CA 95035(408) 736-6000
sales.rad-icon@teledynedalsa.com
www.teledynedalsa.com/ndt

Teledyne Rad-icon Imaging Corp. is a wholly owned subsidiary of Teledyne DALSA.
Teledyne DALSA has its corporate offices in Waterloo, Canada.
Teledyne DALSA reserves the right to make changes at any time without notice. © 2018 | v1.0

www.teledynedalsa.com/ndt

4