LINEARIZER SAAP (DC CURRENT INPUT)
LINEARIZER SAVP (DC VOLTAGE INPUT)
INSTRUCTION MANUALPreface
Thank you for purchasing the Linearizer SAAP (DC current input) and SAVP (DC voltage input).
This manual contains instructions for the mounting, functions, operations and notes when
operating the SAAP or SAVP. To prevent accidents arising from the misuse of this instrument,
please ensure the operator receives this manual.

Notes• This instrument should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed, serious injury
or malfunction may occur.
• Specifications of the SAAP and SAVP and the contents of this instruction manual are subject tochange without notice.
• Care has been taken to ensure that the contents of this instruction manual are correct, but if there
are any doubts, mistakes or questions, please inform our sales department.
• This instrument is designed to be installed on a DIN rail. If it is not, measures must be taken to
ensure that the operator does not touch power terminals or other high voltage sections.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos CO., LTD. is not liable for any damage or secondary damage(s) incurred as aresult of using this product, including any indirect damage.
SAFETY PRECAUTIONS (Be sure to read these precautions before using our products.)
The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on circumstances, procedures indicated by
Caution may cause serious results,
so be sure to follow the directions for usage.
which may lead to dangerous conditions andWarning Procedurescause death or serious injury, if not carried out properly.

CautionProcedures which may lead to dangerous conditions and cause
superficial to medium injury or physical damage or may
degrade or damage the product, if not carried out properly.

Warning• To prevent an electric shock or fire, only Shinko or other qualified service personnel may
handle the inner assembly.
• To prevent an electric shock, fire or damage to the instrument, parts replacement may only
be undertaken by Shinko or other qualified service personnel.

Safety precautions• To ensure safe and correct use, thoroughly read and understand this manual before using this
instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring
equipment. Verify correct usage after consulting purpose of use with our agency or main office.
(Never use this instrument for medical purposes with which human lives are involved.)• External protection devices such as protection equipment against excessive temperature rise, etc.
must be installed, as malfunction of this product could result in serious damage to the system or
injury to personnel. Also proper periodic maintenance is required.
• This instrument must be used under the conditions and environment described in this manual.
Shinko Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring
due to the instrument being used under conditions not otherwise stated in this manual.
Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the manufactureof weapons of mass destruction (i.e. military applications, military equipment, etc.), please
investigate the end users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.
2

1. Installation precautions

CautionThis instrument is intended to be used under the following environmental conditions(IEC61010-1): Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of -5 to 55 (23 to 131 )that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85%RH• No large capacity electromagnetic switches or cables through which large current is flowing• No water, oil or chemicals or where the vapors of these substances can come into
direct contact with the unit
• When installing this unit within a control panel, take note that ambient temperature ofthis unit must not exceed 55 (131 ). Otherwise the life of electronic components(especially electrolytic capacitor) may be shortened.
Note: Avoid setting this instrument directly on or near flammable material even though thecase of this instrument is made of flame-resistant resin.

2. Wiring precautions

Caution• Do not leave bits of wire in the instrument, because they could cause fire and malfunction.
• When wiring terminals, use ferrules with an insulation sleeve and crimping pliers madeby Phoenix Contact GMBH & CO. applicable to terminals.
• Tighten the terminal screw using the specified torque.
If excessive force is applied to the screw when tightening, the screw or case may bedamaged.
• This instrument has no built-in power switch, circuit breaker or fuse. It is necessary to
install them near the instrument.
(Recommended fuse: Time-lag fuse, rated voltage 250V AC, rated current 2A)• For wiring of AC power source, be sure to use exclusive terminals as described in thismanual. If AC power source is connected to incorrect terminals, the unit will burn out.
• For a 24V DC power source, do not confuse polarity when wiring.
• For DC voltage and current inputs, do not confuse polarity when wiring.

3. Operation and maintenance precautions

Caution• Do not touch live terminals. This may cause electric shock or problems in operation.
• Turn the power supply to the instrument OFF when retightening the terminal andcleaning. Working or touching the terminal with the power switched ON may result insevere injury or death due to Electric Shock.
• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)• As the display section is vulnerable, do not strike or scratch it with a hard object orpress hard on it.
3Characters used in this manual
Indication
Number,

/

-1

0

1

2

3

4

5

6

7

8

9

A

B

C

D

E

F

G

H

I

J

K

L

M

N

O

P

Q

R

S

T

U

V

W

X

Y

Z

Indication
Alphabet
Indication
Alphabetmeans that no character is indicated (unlit) on the display.

4

---

CONTENTS ---1. ModelPage
1.1 Model ---------------------------------------------------------------------------- 6
1.2 How to read the model label ----------------------------------------------- 6
2. Name and functions of sections ----------------------------------- 6

3. Mounting

3.1 External dimensions (Scale: mm) ---------------------------------------- 7
3.2 Mounting and removal to/from the DIN rail ----------------------------- 7

4. Wiring

4.1 Recommended ferrules ----------------------------------------------------- 8
4.2 Terminal arrangement and circuit configuration ---------------------- 9
4.3 Wiring of terminals ----------------------------------------------------------- 9
4.3.1 Power source wiring ---------------------------------------------------- 9
4.3.2 Output wiring -------------------------------------------------------------- 9
4.3.3 Input wiring ---------------------------------------------------------------- 9
5. Operation flowchart ------------------------------------------------------ 10

6. Setup

6.1 Indication after power ON ------------------------------------------------- 12
6.2 Basic operation of setup --------------------------------------------------- 13
6.3 Setup of the unit ------------------------------------------------------------- 13
6.4 Linearization function ------------------------------------------------------- 15
6.5 Basic operation of Linearization ----------------------------------------- 16
6.6 Linearization setting mode ------------------------------------------------ 17

7. Adjustment

7.1 Basic operation of adjustment ------------------------------------------- 18
7.2 Adjustment -------------------------------------------------------------------- 18

8. Operation

8.1 Indication after power-on -------------------------------------------------- 19
8.2 Operation --------------------------------------------------------------------- 20
8.2.1 When using this unit as a signal conditioner -------------------- 20
8.2.2 When using the Reverse function ---------------------------------- 20
8.2.3 When using the 1st order lag filter function ---------------------- 20
9. Specifications -------------------------------------------------------------- 21

10. Troubleshooting

10.1 Indication -------------------------------------------------------------------- 22
10.2 Key operation -------------------------------------------------------------- 22
10.3 Operation ------------------------------------------------------------------- 22
11. Character table ------------------------------------------------------------- 23

5

1. Model
1.1 Model

SA
Signalconditioner

P
A
V

Power supply

0
1

Series name: SA
DC current input linearizer
DC voltage input linearizer
100 to 240V AC
24V AC/DC(e.g.) SAAP-0 (DC current input linearizer), Supply voltage: 100 to 240V AC.
Default: Input: 4 to 20mA DC, Output: 4 to 20mA DC
SAVP-0 (DC voltage input linearizer), Supply voltage: 100 to 240V AC.
Default: Input: 1 to 5V DC, Output: 4 to 20mA DC
1.2 How to read the model label
The model label is attached to the left side of the case.
Model (e.g. SAAP-0)
Input (Enter a type selected during Input type selection.)
Output (Enter a type selected during Output type selection.)
Serial number(Fig. 1.2-1)

2. Name and functions of sections
1 Power indicator (Green)

Lights when the power to the instrument is turned on.

1

2 Input display (Red)

2
3

4

6

5

7

(Fig.2.1)

Indicates the input value during Run mode.
Indicates setting (or adjustment) characters in the
Setup, Adjustment and Linearization setting mode.
3 Output display (Green)
Indicates the output value (%) during Run mode.
Indicates the set (or adjusted) value in the Setup,
Adjustment and Linearization setting mode.
4 Up key ()
Increases the numeric value, or switches theselection items.
5 Down key ()
Decreases the numeric value, or switches theselection items.
6 Mode key ()
Switches the setting mode and registers theset (or selected) value.
By holding down this key for approx. 3 seconds,
the unit proceeds to the Adjustment mode.
7 Sub-mode key (Unmarked)
If the Mode key is pressed while holding down thiskey, the unit proceeds to the Setup mode.
By holding down this key for approx. 5 seconds,
the unit proceeds to the Linearization setting mode.
6

3. Mounting
3.1 External dimensions (Scale: mm)

75

DIN rail

97
22.5

100

4

(Fig. 3.1-1)
3.2 Mounting and removal to/from the DIN rail

Caution• Mount the DIN rail horizontally.
• To remove this instrument, a flat blade screwdriver is required for pulling down
the lever.
Never turn the screwdriver when inserting it into the release lever.
If excessive power is applied to the lever, it may break.
• Be sure to use commercially available fastening plates at both ends of the unit if it
is in a position susceptible to vibration or shock.
Recommended fastening plate
Manufacturer
Omron Corporation
IDEC Corporation
Panasonic Electric Works, Co., Ltd.

Model
End plate
PFP-M
Fastening plate BNL6
Fastening plate ATA4806

Mounting to the DIN rail (Fig. 3.2-1)• Hook 1 of the instrument on theupper side of the DIN rail.
• Making 1 part of the instrument
as a support, fit the lower part 2of the instrument to the DIN rail.
The unit will be completely fixed to
the DIN rail when a “Click” sound is heard.

12(Fig. 3.2-1)
7

Removal from the DIN rail (Fig.3.2-2)• Insert a flat blade screwdriver
into the release lever ( 1 ).
• Remove the instrument from the
DIN rail by pulling down the lever ( 2 ).

Release lever

2

1(Fig. 3.2-2)

4. Wiring
Warning
Turn the power supply to the instrument off before wiring.
Working or touching the terminal with the power switched on may result in severe
injury or death due to Electric Shock.

Caution• For 100 to 240V AC, if AC power source is connected to incorrect terminals,
this instrument will burn out.
• For a 24V DC power source, do not confuse polarity when wiring.

4.1 Recommended ferrules
When using ferrules, use the following recommended ferrules and crimping pliers made by
Phoenix Contact GMBH &CO. See (Table 4.1-1).
Take note that screw size and tightening torque differ depending on the terminal number.
(Table 4.1-1)
Terminal Terminal Ferrules withnumberscrew insulation sleeve
1 to 4
M2.6
AI 0.25-8 YE
AI 0.34-8 TQ
AI 0.5-8 WH
AI 0.75-8 GY
AI 1.0-8 RD
AI 1.5-8 BK
5 to 9M2.0
AI 0.25-8 YE
AI 0.34-8 TQ
AI 0.5-8 WH

Conductor
Tightening
Crimping pliers
cross sections
torque0.2 to 0.25mm2 0.5 to 0.6N•m
CRIMPFOX2
ZA 3
0.25 to 0.34mm
20.34 to 0.5mm
CRIMPFOX
0.5 to 0.75mm2UD 6
20.75 to 1.0mm
1.0 to 1.5mm2
0.2 to 0.25mm2 0.22 to 0.25N•m
0.25 to 0.34mm2
0.34 to 0.5mm2

8

4.2 Terminal arrangement and circuit configuration
SAAP+

Insulation circuit
5

Input

-

6

Input
circuitOutputcircuitCPU

3 +Output4-

7
8

Display/Setting key
circuit9Power
circuit1 +Power supply
2-(Fig. 4.2-1)
SAVP
Input
+
mV DC
0 to 1V DC-

0 to 5V DC
1 to 5V DC
0 to 10V DC+-

Insulation circuit
5

OutputcircuitCPU
63

Output4

Input
circuit+-7
8

Display/Setting key
circuit9Power
circuit1+
Power supply

2-(Fig. 4.2-2)
4.3 Wiring of terminals
4.3.1 Power source wiring
Use terminals 1 (+) and4.3.2 Output wiring
Use terminals 3 (+) and2 (-) for the power supply to the instrument.

4 (-) for the output wiring.

4.3.3 Input wiring
SAAP: Use terminals 5 (+), 6 (-) for input wiring and shunt resistor (sold separately)connection (Table 4.3.3-1).
SAVP: Terminals for wiring differs depending on the input specifications.
Refer to (Fig. 4.2-2).
(Table 4.3.3-1)

Shunt resistor
Model
Specification
4 to 20mA DC, 0 to 20mA DC, 0 to 16mA DC RES-S02-050
50
0.1%
2 to 10mA DC, 0 to 10mA DC
RES-S02-100
100
0.1%
1 to 5mA DC
RES-S02-200
200
0.1%
0 to 1mA DC
RES-S02-01K
1k
0.1%
Input

9

5. Operation flowchart
Power ONWarm-up indication (Approx. 3sec.)

Run mode

Input, Output display
The Input display indicates setting (or adjustment)characters, and the Output display indicates thedefault value.

Hold down the
key for approx. 3sec.
Adjustment modeIN
Output zeroadjustment
OUT

IN
OUT

Press thekey.
Output span
adjustment
Press the

IN
OUT

IN
OUT

Press thekey while holding
down the Sub-mode key.
Setup modeIN
Set value lockSensorcorrection
OUT
Press thekey.
Input type
(SAAP)

IN
OUT

key.
IN

(SAVP)

IN

OUT

IN
OUT

IN
OUT

IN
OUT

IN
OUT

OUT

Press thekey.
Decimal point
place

IN

Press thekey.
Output 0%
value

IN

Press thekey.
Output 100%value
Press thekey.
Filter time
constant
Press the

10

key.

Press thekey.
Output type

OUT

Press thekey.
OutputNormal/Reverse
Press thekey.
Display
selection
Press thekey.
Indication time

OUT

Press thekey.

Key operation: This means that if the
key is pressed, the set value is saved, andthe instrument proceeds to the next setting item.
• Set the value with theorkey, and register it with thekey.
• If thekey is pressed for approx. 3sec, the instrument reverts to the Run
mode from any mode.
• To return from Adjustment to Run mode, press the
key for approx. 3sec.
From any setting item in Adjustment mode, it is possible to return to Run mode.
•IN

Press the Sub-mode key for 5sec.
Linearization setting mode
Linearization Enabled/Disabled

Press theOUT

Press theIN(when Enabled is selected)

Input X1

OUT

Press theIN

key.

Output Y1

OUT

Press theIN

key.

Input X2

OUT

Press theIN

key.

Output Y2

OUT

Press theIN

key.

Input X25

OUT

Press theIN

key.

Output Y25

OUT

Press the

key.

11

(when Disabled
is selected)

6. Setup
Setup should occur before using this unit, to set the Input type, Output 0% value, Output
100% value, Output type, Linearization Enabled/Disabled, etc. according to the users’
conditions.
If the users’ specifications are the same as the default value of the instrument, or if setup
has already been completed, it is not necessary to set up the instrument. Proceed to
Section “7. Adjustment”.
(Table 6-1)
Setting item
Default value
Set value lock
UnlockInput type
SAAP 4 to 20mA DC (-1999 to 9999)
SAVP 1 to 5V DC (-1999 to 9999)
Decimal point place
No decimal point
Output 0% value-1999
Output 100% value
9999
Filter time constant
0.0 seconds
Sensor correction0
Output type
4 to 20mA DC
Output Normal/Reverse
NormalDisplay selection
Input/Output indication
Indication time
00.00 (Continuous)
Linearization Enabled/Disabled Linearization Disabled
6.1 Indication after power-on
After power-on, warm-up status below (Fig. 6.1-1) is indicated for approx. 3sec.
Indicates the input type.
SAAP: Indicateswhen default value is
4 to 20mA DC (-1999 to 9999).
SAVP: indicateswhen default value is
1 to 5V DC (-1999 to 9999).

Lights after
power-on.

Indicates the output type.
Indicateswhen default value is 4 to 20mA DC.
(Fig. 6.1-1)
After that, the unit switches to the Run mode as shown below.

Indicates the input value.

Indicates the output value in a percentage
form (%) corresponding to the input value.
(Fig. 6.1-2)
12

6.2 Basic operation of setup
Setup is conducted in the Setup mode.
To enter the Setup mode, press the
key while holding down the Sub-mode key in the
Run mode. (Fig. 6.2-1)
To set (select) each item, use theorkey, and register the value with the
key.
(Fig. 6.2-2)(1) Run mode

To enter the Setup mode, press theMode ( ) key while holding down the
Sub-mode key in the Run mode.
Mode key
Sub-mode key (Unmarked)(Fig. 6.2-1)(2) Setup mode

To proceed to each setting item andto register the set (or selected) value,please use this key.

To set (or select)each item, use
these keys.
(Fig. 6.2-2)

6.3 Setup of the unit
The following shows all setup items. Set up the unit referring to the explanation of each item.
DisplayName, Function, Setting range
Default value
Set value lock
UnlockIN
Locks the set values to prevent setting errors.
Selection item:
OUT: Unlock: Lock (None of the set values and adjusted values can be changed.)IN
4 to 20mA DC, -1999 to 9999
Input type (SAAP)
Selects the input type.
OUT
Selection item:: 4 to 20mA DC-1999 to 9999
: 0 to 20mA DC-1999 to 9999
: 0 to 16mA DC-1999 to 9999
: 2 to 10mA DC-1999 to 9999
: 0 to 10mA DC-1999 to 9999
: 1 to 5mA DC-1999 to 9999
: 0 to 1mA DC-1999 to 9999

13

IN

OUT

IN

OUT

IN

OUT

IN

OUT

IN

OUT

IN

OUT

IN

OUT

IN

OUT

1 to 5V DC, -1999 to 9999
Input type (SAVP)
Selects the input type.
Selection item:: 0 to 10mV DC-1999 to 9999
: -10 to 10mV DC-1999 to 9999
: 0 to 50mV DC-1999 to 9999
: 0 to 60mV DC-1999 to 9999
: 0 to 100mV DC-1999 to 9999
: 0 to 1V DC-1999 to 9999
: 0 to 5V DC-1999 to 9999
: 1 to 5V DC-1999 to 9999
: 0 to 10V DC-1999 to 9999
No decimal point
Decimal point place
Selects the decimal point place.
Selection item:: No decimal point: 1 digit after decimal point: 2 digits after decimal point: 3 digits after decimal point-1999
Output 0% value
Sets the value (indicated on the Input display) at 0% output.
Setting range: -1999 to Output 100% value (The placement of the decimal
point follows the selection)
9999
Output 100% value
Sets the value (indicated on the Input display) at 100% output.
Setting range: Output 0% value to 9999 (The placement of the decimal
point follows the selection)
0.0 seconds
Filter time constantSets the filter time constant.
Reduces input fluctuation caused by noise.
Setting range: 0.0 to 10.0 seconds
0
Sensor correction
Sets the sensor correction value.
Input value = Current input value + Sensor correction value
Setting range: -1000 to 1000 (The placement of the decimal point follows
the selection)
4 to 20mA DC
Output type
Selects the output type.
Selection item:: 4 to 20mA DC: 0 to 20mA DC: 0 to 12mA DC: 0 to 10mA DC: 1 to 5mA DC: 0 to 1V DC: 0 to 5V DC: 1 to 5V DC: 0 to 10V DC
NormalOutput Normal/Reverse
Selects either Normal mode (0.0 to 100.0%) or Reverse mode (100.0 to0.0%) for output status.
Selection item:: Normal: Reverse
14

IN

OUT

IN

OUT

Input/Output indication
Display selection
Selects an indication type on the display.
Selection item:: Input/Output indication: Input indication: Output indication: No indication (Only power indicator is lit.)
00.00 (Continuous)
Indication time
Sets the indication time of the display after the final key operation.
Not available if No indication (Only the power indicator is lit) is selectedduring Display selection
After the indication time has elapsed, the displays go off (Only the power
indicator is lit.).
If power is turned on again, or if any of the keys,,
and theSub-mode key is pressed while displays are unlit, the displays will light again.
Setting range:
00.00: Continuous indication
00.01 (1 second) to 60.00 (60 minutes) [Minute.Second]

Output Yn

6.4 Linearization function
If Linearization Enabled is selected during Linearization Enabled/Disabled selection mode,a maximum of 25 points of curved line characteristic can be programmed.
One point consists of one pair of input Xn (%) and output Yn (%). (n: Numbers from 1 to 25).
Set necessary points from the smallest input value (X1, Y1), (X2, Y2) … (X25, Y25) innumeric order.
(Setting example)
110.0%
100.0%nXn (%)Yn (%)
Curved line
1
10.0
20.0
characteristic
2
40.0
30.0
Y3
Linear3
60.0
70.0
characteristic
Action explanation
Y2
1 If input is lower than X1(10.0%), outputs
Y1
linearly between -10.0% and Y1(20.0%).
2 If input is between X1(10.0%) and
0.0%
1
2
3
4
X2(40.0%), outputs linearly between-10.0%
0.0% X1
X2
X3
100.0%
Y1(20.0%) and Y2(30.0%).
110.0%-10.0%
3 If input is between X2(40.0%) andInput Xn
X3(60.0%), outputs linearly between(Fig. 6.4-1)
Y2(30.0%) and Y3(70.0%).
4 For other inputs, outputs linearly as steps 2 and 3 depending on points.
For other inputs after X3(60.0%), outputs linearly between Y3(70.0%) and 110.0%.
In the case of Xn=Xn+1 or Xn>Xn+1: Settings after Xn+1 will be invalidated, and will output
linearly between Yn and 110%.
(e.g.) In the above example, if X3 is set to 40.0%, then X2=X3. Therefore, settings after X3
will be invalidated, and will output linearly between Y2(30.0%) and 110.0%.
If X1 and X2 are set to 0.0%, this is considered as no setting, and will output linearly
between -10.0% and 110.0% [the same as the linear characteristic in (Fig.6.4-1)].
(e.g.) In the case of X1(0.0%), Y1(20.0%), X2(0.0%) Y2(30.0%), this is considered as no
setting, and will output linearly between -10.0% and 110.0%.
15

6.5 Basic operation of Linearization
Set the Linearization function in the Linearization setting mode.
To enter the Linearization setting mode, hold down the Sub-mode key for approx. 5sec in
the Run mode. (Fig. 6.5-1)
To set (or select) values, use the(Fig. 6.5-2)(1) Run modeorkey, and register the value with thekey.

To enter the Linearization setting mode,
hold down the Sub-mode key for approx
5sec in the Run mode.

Sub-mode key (Unmarked)(Fig. 6.5-1)(2) Linearization setting mode

Please use
these keys
for setting (orselection).

To proceed to each item in the
Linearization setting mode and toregister the value, please use this key.

(Fig. 6.5-2)
6.6 Linearization setting mode
The following shows all items for Linearization function setting. Refer to the explanation ofeach item.
DisplayName, function, Setting range
Default valueIN
Linearization Enabled/Disabled Linearization Disabled
OUT
Selects Linearization function Enabled or Disabled.
Selection item:: Linearization Disabled: Linearization Enabled
If Linearization Disabled is selected during the Linearization Enabled/Disabled
selection, the following items will not appear. The unit reverts to the Run mode.
IN
Input X1
0.0%
OUT
Sets Input X1.
Setting range: -10.0 to 110.0%
IN
Output Y1
0.0%
OUT
Sets Output Y1 corresponding to Input X1.
Setting range:-10.0 to 110.0%
16

IN

Input X2
0.0%
Sets Input X2.
Setting range:-10.0 to 110.0%
Output Y2
0.0%
Sets Output Y2 corresponding to Input X2.
Setting range:-10.0 to 110.0%

OUT

IN
OUT

IN

Input X25
0.0%
Sets Input X25.
Setting range:-10.0 to 110.0%
Output Y25
0.0%
Sets Output Y25 corresponding to Input X25.
Setting range:-10.0 to 110.0%

OUT

IN
OUT

7. Adjustment

Performs the output zero and span adjustments.
Connect an mV generator to the input terminals of this instrument.
Connect a digital multimeter to output terminals.
7.1 Basic operation of adjustment
Adjustment can be conducted in the Adjustment mode.
To enter Adjustment mode, hold down the
key for approx. 3 seconds in the Run mode.
(Fig. 7.1-1)
For output adjustment, use theorkey, and register the value with the
key.
(Fig. 7.1-2)
To revert to the Run mode, press the

key again for approximately 3 seconds.

(1) Run mode

To enter Adjustment mode, pleasehold down the
key for approx.
3 seconds.

(Fig. 7.1-1)(2) Adjustment mode

To proceed to each item in the
Adjustment mode and to register
the adjusted value, please use
this key.

Please use thesekeys for adjustment.

(Fig. 7.1-2)
17

7.2 Adjustment
The following shows all adjustment items. Adjust values referring to explanation of each item below.
DisplayName, Function, Setting range
Default valueIN
0.00%
Output zero adjustment
Adjusts output zero.
OUT
Input the value corresponding to 0% output, then adjust the value with theor
key while viewing the output value (on the digital multimeter).
When the output range lower limit is zero, (even if zero adjustment results
in a negative value), the output value will not be negative.
Setting range: -5.00 to 5.00%
Effective range of adjustment differs depending on the output types.
4 to 20mA DC: -5 to 5%
0 to 1V DC : 0 to 5%
0 to 20mA DC: 0 to 5%
0 to 5V DC : 0 to 5%
0 to 12mA DC: 0 to 5%
1 to 5V DC : -5 to 5%
0 to 10mA DC: 0 to 5%
0 to 10V DC : 0 to 5%
1 to 5mA DC : -5 to 5%
IN
Output span adjustment
0.00%
Adjusts output span.
OUT
Input the value corresponding to 100% output, then adjust the value with theor
key while viewing the output value (on the digital multimeter).
Setting range: -5.00 to 5.00%
Effective range of adjustment is 95 to 105%.

8. Operation

8.1 Indication after power-on
After power-on, the following warm-up status is indicated for 3 seconds (Fig. 8.1-1).
Indicates the input type. (Table 8.1-1)
SAAP: Indicateswhen default value is
4 to 20mA DC (-1999 to 9999).
SAVP: indicateswhen default value is
1 to 5V DC (-1999 to 9999).

Lights after power-on.

Indicates the output type (Table 8.1-2).
Indicateswhen default value is 4 to 20mA DC.
(Table 8.1-1)
Input
4 to 20mA DC
0 to 20mA DC
0 to 16mA DC
2 to 10mA DC
0 to 10mA DC
1 to 5mA DC
0 to 1mA DC
0 to 10mV DC-10 to 10mV DC
0 to 50mV DC
0 to 60mV DC
0 to 100mV DC
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC(Fig. 8.1-1)
Input display: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
: -1999 to 9999
18

(Table 8.1-2)
Output4 to 20mA DC
0 to 20mA DC
0 to 12mA DC
0 to 10mA DC
1 to 5mA DC
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

Output display

8.2 Operation
The unit enters the Run mode as shown in (Fig. 8.2-1). The input signal selected during
Input type selection is converted to the output selected during Output type selection.
Indicates the input value.

Indicates the output value in a percentage form (%).
Output value corresponding to the input value is indicated in apercentage form (%).
(Fig. 8.2-1)• Indication when input value is -2000 or less
For the indication of -2000 or less (up to –10% output), the input value and the minus (-)
sign are indicated alternately.
(e.g.) Indication of -2000
IN
IN• Indication when input value is 10000 or more
For the indication of 10000 or more (up to 110% output), the lower 4 digits of input value
are flashing.
(e.g.) Indication of 10020
IN
Flashes.
• Underrange, Overrange and Sensor burnout alarm indication
Even if any selection is made during the Display selection, the following indications appear.
Underrange : "" flashes on the Input display.
Overrange : "" flashes on the Input display.
• Indication time setting
If indication time is set, the displays will go off after the indication time has elapsed.
(Only the power indicator is lit.)
If power is turned on again, or if any of the keys,,
and the Sub-mode key
is pressed while displays are unlit, the displays will light again.
8.2.1 When using this unit as a standard linearizer
Set the filter time constant to 0.0 seconds, andset the Output Normal/Reverse selection to “Normal”.
8.2.2 When using the Reverse function
This function reverses the output (100 to 0%)
Input
that corresponds to the input (0 to 100%).
Set the Output Normal/Reverse selection to “Reverse”.
8.2.3 When using the first order lag filter function
The value is outputted by performing the firstorder lag computation using the filter time
constant “T”. (Fig. 8.2.3-1)
Set the filter time constant to a random value(0.0 to 10.0 seconds).
19

Time
OutputT(Fig. 8.2.3-1)

Time9. Specifications

Input specifications
SAAP
Input
Shunt resistor
4 to 20mA DC
50
0 to 20mA DC
0 to 16mA DC
2 to 10mA DC
100
0 to 10mA DC
1 to 5mA DC
200
0 to 1mA DC
1k
Connect a shunt resistor (sold separately) between input terminals.
See (Table 4.3.3-1) on page 9.
SAVP
Allowable signal
Input
Input resistance source
resistance
0 to 10mV DC
20 or less
-10 to 10mV DC
40 or less0 to 50mV DC
200 or less0 to 60mV DC
1M
0 to 100mV DC
0 to 1V DC
2k or less0 to 5V DC
1k or less
1 to 5V DC
0 to 10V DC
Output specifications
DC current
Output4 to 20mA DC
0 to 20mA DC
0 to 12mA DC
0 to 10mA DC
1 to 5mA DC

Allowable load Zero adjustment
resistancerange
700 or less
-5 to 5%700 or less0 to 5%1.2k or less0 to 5%1.2k or less0 to 5%2.4k or less
-5 to 5%

Span adjustment
range
95 to 105%
95 to 105%
95 to 105%
95 to 105%
95 to 105%

DC voltage
Allowable load Zero adjustment
Span adjustment
resistance
range
range
0 to 1V DC
100 or more
0 to 5%95 to 105%
0 to 5V DC
500 or more
0 to 5%95 to 105%
1 to 5V DC
500 or more
-5 to 5%95 to 105%
0 to 10V DC
1k or more
0 to 5%95 to 105%
When the output range lower limit is zero, (even if zero adjustment results
in a negative value), the output value will not be negative.
Performance
Accuracy
Input : Within 0.1%
Output : Within 0.1%
Display accuracy
Within Input accuracy 1 digit
Response time
0.5 seconds (typical) (090%)
Temperature coefficient 0.015%/
Insulation resistance
Input – Output – Power: 10M or more, at 500V DC
Dielectric strength
Input – Output – Power: 2.0kV AC for 1 minute20
OutputGeneral structure
Case
Front panel
Setting
Display, indicator

Flame-resistant resin, Color: Light gray
Membrane sheet
Setting by the front keypad
Input display : 7 segments Red LED display 4 digits
Character size: 7.4 x 4.0mm (H x W)
Output display : 7 segments Green LED display 4 digits
Character size: 7.4 x 4.0mm (H x W)
Power indicator: Green LED

Installation specifications
Power supply
100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz
Allowable voltage range 85 to 264V AC, 20 to 28V AC/DCPower consumption
Approx. 6VA
Ambient temperature -5 to 55 (23 to 131 )Ambient humidity
35 to 85%RH (Non-condensing)
WeightApprox. 120g
Mounting
DIN rail mounting
External dimensions
W22.5 x H75 x D100mm
Attached function• Power failure countermeasure: The setting data is backed up in the non-volatile IC memory.
• Self-diagnosis: The CPU is monitored by a watchdog timer, and if an abnormal status isfound on the CPU, the unit is switched to warm-up status with all outputs off.

10. Troubleshooting
10.1 Indication
Problem
Input display is flashing"" or "".

Presumed cause and solution• The sensor may be burnt out.
Change each sensor.
• Check whether the sensor is securely connected to the input
terminals of the instrument.
Ensure that the sensor terminals are securely connectedto the input terminals of the instrument.
• Check the input signal source.
The indication of the Input • Check whether the sensor correcting value is suitable.
Set it to a suitable value.
display is irregular or• There may be equipment that interferes with or makes
unstable.
noise near the unit.
Keep equipment that interferes with or makes noiseaway from the unit.

10.2 Key operation
Problem
Setting or adjustment is
not possible.
10.3 Operation
Problem
Input value does notchange.

No output

Presumed cause and solution• “Lock” has been selected during Set value lock selection.
Select “Unlock”.
Presumed cause and solution• The sensor may be out of order. Change the sensor.
• Check whether input and output wires are securely
connected to the Input/Output terminals of the instrument.
Ensure that input and output wires are securely connectedto the Input/Output terminals.
• Check whether the wiring of input and output is correct.
• Check whether Output 100% and Output 0% value havebeen set to suitable values.
• Check whether Output type and Output Normal/Reverse havebeen selected correctly during Output type and Output
Normal/Reverse selection.
21

11. Character table

All setting items are indicated in the following tables, however, some items will not be
indicated depending on the specifications.

Setup mode
Display
Setting item
Set value lock
Input type (SAAP)
Input type (SAVP)
Decimal point place
Output 0% value
Output 100% value
Filter time constantSensor correction
Output typeOutput Normal/Reverse
Display selection
Indication time

Default value
Unlock4 to 20mA DC (–1999 to 9999)
1 to 5V DC (–1999 to 9999)
No decimal point

Data

-1999
9999

0.0 seconds
0
4 to 20mA DC
NormalInput/Output indication
00.00 (Continuous)

Adjustment mode
Display
Setting item
Output zero
Output span

Default value

Data

0.00%
0.00%

Linearization setting mode
Display
Setting item
Linearization Enabled/Disabled
Input X1
Output Y1
Input X2
Output Y2
Input X3
Output Y3
Input X4
Output Y4
Input X5
Output Y5
Input X6
Output Y6
Input X7
Output Y7
Input X8
Output Y8
Input X9
Output Y9
Input X10
Output Y10
Input X11
Output Y11
Input X12
Output Y12

Default value
Linearization Disabled
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%

Data

22

Input X13
Output Y13
Input X14
Output Y14
Input X15
Output Y15
Input X16
Output Y16
Input X17
Output Y17
Input X18
Output Y18
Input X19
Output Y19
Input X20
Output Y20
Input X21
Output Y21
Input X22
Output Y22
Input X23
Output Y23
Input X24
Output Y24
Input X25
Output Y25

0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%
0.0%

23

Linearization table (photocopiable)
Xn(%)

Yn(%)
110.0%
100.0%

Output Yn

n
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

50.0%

0.0%
-10.0%
0.0%-10.0%

50.0%

100.0%

Input Xn

110.0%******* Inquiry *******For any inquiry about this unit, please contact the vendor where youpurchased the unit or our agency after checking the following.
(e.g.)• Model ........................………. SA P• Serial number ….................... No. xxxxxx
In addition to the above, please let us know the details of malfunction,
if any, and the operating conditions.

SHINKO TECHNOS CO.,LTD.
OVERSEAS DIVISION

Reg. Office : 2-5-1, Senbahigashi, Minoo, Osaka, JapanURL: http://www.shinko-technos.co.jp
Tel : 81-72-727-6100
E-mail: overseas@shinko-technos.co.jp
Fax: 81-72-727-7006
No.SA3E2 2009.04
24