SE series

SPEC. SHEET

Thermocouple Transmitter

Model:SE2E(with indication function)■ Model

SE2E - □ - □Socket
1: Finger-safe(For Y terminal)
2: For Ring terminal
Power supply
0: 100 to 240V AC
1: 24V AC/DC■ How to orderSpecify the model (e.g.) SE2E-1-0
Default value
CH1 input
K: -200 to 1370
CH2 input
K: -200 to 1370
CH1 output
4 to 20mA DC
CH2 output
4 to 20mA DC■ Accessories (sold separately)
Communication cable for the console software: CMB-001

DC voltage
Output range

Allowable loadresistance

0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

100 or more
500 or more
500 or more
1k or more

Zeroadjustment
range
0 to 5%0 to 5%-5 to 5%0 to 5%Span
adjustment
range
95 to 105%
95 to 105%
95 to 105%
95 to 105%

■ Input specification
Thermocouple
Input resistance: 1M or more
External resister: 100 or less, however, B input, 40 or less
Burnout: Upscale, Downscale (Selectable by keypad）
Input
Thermocouple
Input range
K-200 to 1370
-328 to 2498
J-200 to 1000
-328 to 1832
R-50 to 1760
-58 to 3200
S-50 to 1760
-58 to 3200
B0 to 1820
32 to 3308
E-200 to 800-328 to 1472
T-200 to 400-328 to 752
N-200 to 1300
-328 to 2372
PL0 to 1390
32 to 2534
W5Re/W26Re
0 to 2315
32 to 4199
W3Re/W25Re
0 to 2315
32 to 4199
Minimum span: 50(100 )■ Output specification
When the output range lower limit is zero, (even if zeroadjustment results in a negative value), the output value
will not be negative.
DC current
ZeroSpan
Allowable load
Output range
adjustment
adjustment
resistance
range
range
4 to 20mA DC
700 or less
-5 to 5%95 to 105%
0 to 20mA DC
700 or less0 to 5%95 to 105%
0 to 12mA DC
1.2k or less0 to 5%95 to 105%
0 to10mA DC
1.2k or less0 to 5%95 to 105%
1 to 5mA DC
2.4k or less
-5 to 5%95 to 105%■ Performance
Accuracy (When ambient temperature is 23 ):Input:
Within 0.1% of each input span, however,R, S input, -50 to 200 (-58 to 392 ): Within 6 (12 )B input, 0 to 300 (32 to 572 ): Accuracy is not guaranteed.
K, J, E, T, N input, less than 0 (32 ):Within 0.4% of each input span
Output: Within 0.1%
Cold junction compensation accuracy: Within 1 at -5 to 55
Indication accuracy: Within input accuracy 1 digit
Input sampling period: 25ms, 125ms, 250ms (Selectable by keypad)
Response time: 65ms (typ.) (0→90%) (Input sampling period 25ms)
225ms (typ.) (0→90%) (Input sampling period 125ms)
425ms (typ.) (0→90%) (Input sampling period 250ms)(Selectable by keypadTemperature coefficient: 0.015%/ or lessInsulation resistance: 10M or more, at 500V DC(Input – Output – Power supply)
Dielectric strength: 2.0kV AC for 1 minute(Input – Output – Power supply)

SE2E■ General structure■ Displays and indicators

Case: Flame-resistant resin, Color: Light gray
Front panel: Membrane sheet
Setting: By the front keypad
Connector for console software: Only for CMB-001
Indication: Input display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Output display: 7-segment, Red LED display 4-digit
Character size 10×4.6mm (H×W)
Power indicator: Green LED
CH1 indicator: Yellow LED
CH2 Indicator: Yellow LED

Input display: Indicates the input value
Indication of -200.0 or less (for the range withdecimal point):
The minus (-) sign and input value light alternately.
Under range: “” flashes on the input display.
Over range: “” flashes on the input display
Warm-up indication: For approx. 3sec. after the power to theinstrument is turned on, the input type of CH1
is indicated on the input display, the input
type of CH2 is indicated on the output display.
Output display: Indicates output volume in percentage (%) form.
Power indicator: The green LED lights when the power to theinstrument is turned on.
CH1 indicator: The yellow LED lights when CH1 is selected during
Display selection mode.
CH2 indicator: The yellow LED lights when CH2 is selected during
Display selection mode.

■ Solderless terminal
Y typeRing type

Console connector■ Installation specifications

TCTC

Power supply: 100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz■ Circuit configuration and terminal arrangement
Allowable voltage range: 85 to 264V AC, 20 to 28V AC/DCPower consumption: Approx. 8VA
Ambient temperature: -5 to 55+ 1+13PowerPowerAmbient humidity: 35 to 85%RH (Non-condensing)
CH1
Display/Setting
2
CH1
circuitsupplykeycircuit
Mounting: DIN rail mounting
14
input
External dimensions: W30×H88×D108mm (including the socket)
3circuit
Insulation circuit
Weight: Approx. 190g (including the socket)
4■ Attached functions

CH2

Auto-light function: Display brightness is controlled in accordancewith the surrounding area. Unnecessary brightness is reduced,saving energy.
Power failure countermeasure: The data is backed up in nonvolatile IC memory.
Self diagnosis: The CPU is monitored by a watchdog timer, andwhen an abnormal status is found on the CPU, the unit isswitched to warm-up status with tuning all outputs off.
Cold junction compensation: Available- 67

CH1output
circuit

CPU+ 59+

CH1 output
10-

Insulation circuit

CH2input
circuit

CH2output
circuit811+
CH2 output

12-

Insulation circuit■ External dimensions (Scale: mm)■ Environmental specification
RoHS directive compliance■ Settings
Function keys
(1) UP Key: Increases the numeric value.
(2) DOWN Key: Decrease the numeric value.
(3) MODE Key: Selects the setting mode.
(4) SUB-MODE Key: Turns the displays ON again when theyare in OFF status.
(The UP, DOWN or MODE Key also turns the displays ON again
when they are in OFF status.)

SHINKO TECHNOS CO., LTD.

Tel: 81-72-727-6100

URL: http://www.shinko-technos.co.jp/e/ E-mail: overseas@shinko-technos.co.jp

20120106