Acryl-mold type purgemeter for measuring gases andliquids
Compact, low-cost, and quick in delivery

P-060
Purgemeter■ OUTLINEThe P-060 is an acryl-mold, standard type purgemeter.
This ultra-compact, low-price purgemeter is ideal for lot production.
The P-060 is available in two sizes; 105 mm long, 22 mm wide and 114mm long and 30 mm wide.

■ MAJOR APPLICATIONS
Waterlines, air/nitrogen lines■ STANDARD SPECIFICATIONS

Material

Small ﬂow rate type / Body
Large ﬂow rate type / Body
code：Ａ
code：BGases(air / nitrogen)Measuring ﬂuid
Liquids(Equivalent to water: Density 1.0g/cm3 Viscosity 1.0mPa・s)
Min. 5 to 50 L/min（nor）
Min. 0.2 to 2 L/min（nor）
AirAirMax. 30 to 300 L/min（nor）Max. 5 to 50 L/min（nor）
Min. 0.3 to 3 L/min
Min. 10 to 100 mL/min
Measuring range
WaterWater
Max. 1 to 10 L/min
Max. 0.3 to 3 L/min
Flow rate range,please select the format code
Range ability
10：1（10:2 for some models） 10：1
Indication accuracy ±5％ F.S.
Fluid pressure Max. 0.5MPa
Fluid temperature Max. 50℃
Wetted part PMMAFloat
SUS304, PTFE, Glass, Ruby
GasketNitrile rubber（NBR）
CapPOM
Joint
SUS304
Scale plate Polycarbonate
Lock nut
Not attached
POM
Process
Rc : 1/8"（Standard）、1/4"
Rc : 3/8"（Standard）、1/4"
Connection
Screw mount on the panel front Lock nut mount on the panel front
Mounting(Standard)(Standard)
Massapprox.0.095ｋ
ｇapprox.0.12ｋ
ｇ

Small ﬂow rate type

Large ﬂow rate type

■ BASIC MODEL CODE（Small ﬂow rate type / Body code:A）
Model CodeP-062

Fluid□ □□□□ ̶ □ □̶

LWater̶

AAir̶NN2̶

Z

Measurinng
range

̶

R □ A

Type
Reed switch alarm
UL-approved
PAU optical alarm unit

Water：100mL/min0004〃 ：400mL/min

0010〃：0020〃 ：2L/min

0025

〃：

2.5L/minFor CE approval, refer to "Index & Quick Reference for P Series
Purgemeters" (TG-S0001).

■ STANDARD FLOW RATE TABLE

〃：

0020

Gas ：0050〃 ：5L/min(nor)0100〃 ：10L/min(nor)

0200

〃 ：

20L/min(nor)

0300

〃 ：

30L/min(nor)〃 ：50L/min(nor)

0500
Valve

̶0̶

L

Air or N220ºC, 0 MPa(1 atm)Not provided
Inlet

WaterAlarm setting
range

Water

Alarm setting
range

0.1 to 1L/min

0.2 to 0.8L/min

2 to 20L/min（nor） 0.2 to 2L/min

0.2 to 2L/min

0.4 to 1.6L/min

3 to 30L/min（nor） 0.5 to 2.5L/min

0.5 to 2.5L/min

0.5 to 2L/min

5 to 50L/min（nor） 10 to 40L/min（nor） 0.3 to 3L/min

Water

(HO)
(HC)

̶R

1/8"（Standard）

21/4"
Small ﬂow rate type（22mmWide、105mm long）

Fluidrange

Water

LWater

̶

AAir

̶

NN2̶Z

Measurinng
range

10 to 100L/min（nor） 0.3 to 3L/min

10 to 100L/min（nor） 20 to 80L/min（nor） 0.3 to 3L/min

0.6 to 2.4L/min

20 to 200L/min（nor） 0.5 to 5L/min

20 to 200L/min（nor） 40 to 160L/min（nor） 0.5 to 5L/min

1 to 4L/min

30 to 300L/min（nor） 60 to 240L/min（nor） 1 to 10L/min

2 to 8L/min

̶

R □ B

Water：0050〃 ：5L/min

0100〃 ：10L/min

0500

Gas ：1000〃： 100L/min(nor)

2000

〃

： 200L/min(nor)

3000

〃

： 300L/min(nor)

Alarm

̶0̶

L

Process Connection type
Process Connection size

3L/min

50L/min(nor)Not provided
Inlet0Not provided

AReed switch

(LO)

B

Reed switch

(LC)

C

Reed switch(HO)

Reed switch

(HC)

D

Body code

Notes

Other liquids / gases
0030

Alarm setting
range

Descriptions

̶ValveAlarm setting

Refer to "Index &Quick Reference
for P Series
Purgemeters"(TG-S0001).

Rc
1□ □□□□ ̶ □ □5 to 50L/min（nor）

30 to 300L/min（nor） 1 to 10L/min

Reed switch
Reed switchA

For alarm output codes of A to D
Air or N220ºC, 0 MPa (1atm)

CD̶

0.6 to 2.4L/min

Large ﬂow rate type / Bodycode: B
Air or N2
20ºC, 0 MPa(1 atm)

(LC)

Body code

P-062

0.1 to 1L/min

For alarm output codes of 0(LO)Reed switch

Process Connection type

0.5 to 5L/min（nor） 40 to 400mL/min

5 to 50L/min（nor） 0.3 to 3L/min

Reed switch

B

Process Connection size

0.2 to 2L/min（nor） 10 to 100mL/min

1 to 10L/min（nor）

Not provided

AModel CodeFor alarm output codes of A to D

Air or N220ºC, 0 MPa (1atm)

3L/min2L/min(nor)■ BASIC MODEL CODE（Large ﬂow rate type / Body code:B）

Small ﬂow rate type / Bodycode: AFor alarm output codes of 0

1L/min0030

Alarm

Availability○×Notes

Other liquids / gases
00010■ ALARM CONTACT AND OUTPUT

Descriptions

̶̶R

Refer to "Index &Quick Reference
for P Series
Purgemeters"(TG-S0001).

Rc
2

1/4"33/8"（Standard）
B

Large ﬂow rate type（30mmWide、114mm long）

TG-F1107-2E
RevisedOct 2015 K
1st edition Jun 2014 K

P-060Purgemeter■ ORDERING INFORMATION
Basic model code

Items to be speciﬁed

P-062- □□□□□ - □□ - □□□No designation（Select in the model code table.）■ DIMENSIONS● Body code A: Standard type (Connection size Rc1/8)● Body code B: Standard type (Connection size Rc3/8)■ Screw mounting on the panel front■ Valve at the inlet■ Lock nut mounting on the panel front■ Valve at the inlet

Panel

2-φ25

TOKYO

2-Rc3/8

2-φ15

2-Rc1/8

2-φ4
OUT

OUT

KEISO
JAPAN

2-M2384

84

Panel（t=3 to 10）

114

63

84

1484

105

2-M3
D63

（MAX.28）

22

10.5

1226

IN
15

10.5

IN

30

Panel cut

● Body code A: Reed switch alarm type (Connection size Rc1/8)（MAX.29）

33

Panel cut20● Body code B: Reed switch alarm type (Connection size Rc3/8)■ Lock nut mounting on the panel front■ Valve at the inlet■ Screw mounting on the panel front■ Valve at the inlet
Panel

Panel (t = 3 to 10)

2-φ15

2-Rc1/8

2-φ25
2-Rc3/8

2-φ42φ

OUT

15

15

OUT
2-M23（MAX.28）

26

54

84

φ1284

114

63

15
10.5

22

32

1215

IN

12IN

10.5

13.5

2-M3
D6(1)

48

57

14
1284

（31）
84

105

2-M3
D6（1）

30（MAX.29）

33

20

Panel cutPanel cut

Note: Use non-magnetic material for the panel when the alarm
output code is one of A to D.
For the case of mounting multiple sets in parallel (with alarm):
A magnet is set in the float. If the distance between two
sets is short, they may interfere with each other and theﬂow rate indication may not be accurate. Contact the factory
if the mounting distance is less than 50 mm.

● Standard material
Parts name
Body
Float ＊
Valve body
NeedleCap
GasketFitting
Scale plate

Standard material○ PMMA○ SUS304/PTFE/Glass/Ruby○ SUS304○ SUS304○ POM○ Nitrile rubber (NBR)○ SUS304
PolycarbonateParts marked ○ come in contact with the measuring ﬂuid.
＊ We will select a proper material for your application.

· SUS316 may be used instead of SUS304 due to production reasons.
· ASTM or AISI materials equivalent to JIS materials may be used dueto production reasons.

* Specification is subject to change without notice.

Head Office : Shiba Toho Building, 1 – 7 – 24 Shibakoen, Minato-ku, Tokyo 105 – 8558
Tel : +81-3 – 3431 – 1625 (KEY) ; Fax : +81-3 – 3433 – 4922
e-mail : overseas.sales@tokyokeiso.co.jp ; URL : http://www.tokyokeiso.co.jp
2

TOKYO KEISO CO., LTD.

TG-F1107-2E