X-ray Line-Scan Camera Series

Crane XIRH High Energy Detector
The X-Scan Imaging XIRH8800 series of linear
array x-ray cameras offer high performancefor high-energy x-ray and gamma-ray
scanning applications in a compact formfactor. A heavy-metal housing shields diodearrays and electronics ensuring long-life
reliability under extreme radiation conditions.
A wide selection of scintillation material
converts high-energy photons into visible
light, fiber optics convey the visible light to ashielded, off-axis CMOS imaging linear diodearray (LDA) while providing a wide dynamicrange, optimization of sensitivity andresolution, and solid-state reliability. The close
proximity of the analog-to-digital converters(ADC) to the detector chips and the use of
low-voltage-differential-signal
technology
minimize interference noise. A collection of
hardware for interfacing to computers andsoftware including drivers, an intuitive
application programming interface (API),and example code software expedite
developments of x-ray scanning systems.

Key Features
Off-axis, fiber-optic design for high-energy reliability in a compact form factor
50 KeV to 15 MeV energy range
Choice of scintillators: GOS:Tb, CsI:Tl, CWO
Wide range of resolutions & selection of lengths
Incorporates X-Scan Imaging’s proprietary photodiode arrays•
Selectable resolution•Low noise, wide dynamic range, high sensitivity•High MTF
16-bit analog-to-digital conversion
Supports variable scan speed with position synchronization
Software development kit
Device drivers, libraries, standard API
GigE/Camera Link/USB3 interface

Applications
Industrial non-destructive testing (NDT)
Weld and corrosion inspection
Fan-beam computed tomography (CT)1

X-Ray Line-Scan Camera Series: Crane XIRH High Energy

Model: XIRH88˽˽-[LLL]i
Model series

XIRH8850

XIRH8801

XIRH8802

XIRH8804

XIRH8808

XIRH8816

Resolution

50 µm

0.1 mm

0.2 mm

0.4 mm

0.8 mm

1.6 mmNumber of pixels

LLL × 512

LLL × 256

LLL × 128

LLL × 64

LLL × 32

LLL × 16

Maximum line rate

550 Hz1500 Hz

3 KHz

6 KHz

12 KHz23 KHzi Active Length is (25.6 mm × LLL) where LLL is a multiple of 6 and LLL ≥ 12 (minimum length is 308 mm andno maximum length limit).
The maximum line rate is available for LLL≤18 (461 mm). Also depending in scintillator choice, image quality
may be degraded at line rates greater than 1 KHz.

Standard Options
Part Numbering:
Example: XIRH8802W15/600-024-GX-FGE
Definitions:XI
RH 88 02 W15[1] [2] [3] [4] [5] [6] [7]/600 –
[8]

024
[9]

–

DS –
[10]

GX [11]FGE[12]Position

Description

Position

Description[1]

Product Family[7]

Scintillator Code[2]Array Type

[8]

Energy Rating

[3]

Shape

[9]

Detector Length (Inches)[4]

Energy Option
H= With FOP[10]

Housing Aspect[5]

Array Series[11]Interface
G=GigEC=CameraLink
U=USB

[6]

Pixel Pitch
02=200um, 04=400um, etc[12]

PC Frame Grabber Card2

X-Ray Line-Scan Camera Series: Crane XIRH High Energy

Setup
The XIRH8800 series camera system includes a camera unit, a software development kit, power
adapter and cabling. The frame-grabber to be installed in the computer is provided optionally.
The objects to be scanned should be passed between the x-ray source and the camera.

Information furnished by X-Scan Imaging is believed to be accurate and reliable. However, noresponsibility is assumed by X-Scan Imaging Corporation for its use. Users are responsible for their products
and applications using X-Scan Imaging components. To minimize the risks associated with users’ products
and applications, users should provide adequate design and operating safeguards. No responsibility is
assumed by X-Scan Imaging Corporation for any infringements of patents or other rights of third parties
that may result from the use of the information. No license is granted by implication or otherwise under
any patent or patent rights of X-Scan Imaging Corporation.
© 2018 X-Scan Imaging Corp.

3