Shad-o-Scan 1501/3001/4501 Datasheet™

CMOS X-Ray Detectors

Shad-o-Scan Scanning X-Ray Detectors:
Ultimate Sensitivity and ResolutionOverview
With the release of the Shad-o-Scan 1501/3001/4501, Teledyne DALSA introduces
a new family of scanning x-ray detectors specifically designed for the challenging
requirements of high-performance industrial and scientific x-ray scanningapplications. The Shad-o-Scan detectors leverage Teledyne DALSA’s advanced
CMOS image sensing technology, which delivers high-speed, low-dose x-ray
images and yields higher image quality than other scanning devices.

Key Features» Radiation hardened CMOStechnology ensures long lifetime» 14-bit digital real-time output» 99 μm pixel size
» 5G Ethernet interface» Energy range 10-225 kV» High speed (up to 50 m/min)» Low power consumption(no active cooling required)» On-board pixel correction» D-TDI (Digital Time DelayIntegration)» Ready-to-run software and drivers

Typical Applications» Non-destructive testing (NDT)» Electronics inspection (2D/3D)» Food inspection» Scientific imaging

www.teledynedalsa.com/ndt

The Shad-o-Scan xx01 product family consists of 3 models up to 45 cm, a 99 μm
pixel size, a high-speed, real-time image transfer via a 5G Ethernet interface, 14 bits
digitization and SDK’s, drivers and programming support. The camera interface
allows easy access to features such as adjusting the frame rate, single and multiple
frame acquisitions, and control of advanced timing modes.

SoftwareEach Shad-o-Scan detector ships with Teledyne DALSA’s CamExpert software andan Ethernet driver. The software is compatible with Windows 7, 8 and 10. Check
with your sales representative for compatibility with other Windows versions orwith the Linux operating system. The detector can be connected on a network,
but for optimal performance a dedicated network adapter is highly recommended.
For writing custom applications to acquire images from the camera, we recommend
using Teledyne DALSA’s Sapera Essential or the Sapera LT SDK (free download at
http://www.teledynedalsa.com).

Shad-o-Scan 1501/3001/4501 Datasheet™

CMOS X-Ray Detectors

Specifications
Detector Specifications

Shad-o-Scan 1501 Shad-o-Scan 3001 Shad-o-Scan 4501

Active Area LengthResolution

152 mm

304 mm

456 mm1536 x 64 pixels

3072 x 64 pixels

4608 x 64 pixels

Pixel Size

99 μm

Pixel Binning Mode

1x1 / 1x2 / 2x2 / 4x4

ROI Readout

Programmable

Maximum Scanning speed

Up to 50 m/min

1Digitalization

14 bits (16384 levels)

Image Lag

<0.1%

Non-Linearity (10..90%FS)<1.5%

Typical Dynamic Range

72 dB

X-Ray Sensor Lifetime Dose> 10 kGy

Digital TDI Functionality

Digital TDI Functionality
TimeCamera Specifications
Supply Voltage (DC)+15 V

Power Consumption (max)

21 W

Data and Control Interface

5GigE Ethernet

Trigger Connector

TTL

On-board Calibration

Yes

Signal intensity

YesShad-o-Scan Pre-Datasheet

Digital Time Delay Integration (D-TDI)

General Specifications
Operating Temperature
Storage Temperature
Humidity (non-condensing)
Weight 2
Ingress Protection Class(1) 8 lines read-out mode(2) depends on detector model

0 to 40°C

Resolution &10 toSensitivity
80% R.H.
-10 to +55°C< 5 kgThe Shad-o-Scan detectors are designed to work with x-ray sources operating at a wide range of beam
DALSA Confidential and Proprietary
IP..... X-ray energies as low as 10-15
IP69KkeV can Teledyneenergy settings.
be detected.
The cameras can be used with x-ray
energies as high as 225kV, although we recommend the use of additional shielding and/or collimation at
higher energies in order to protect the sensor element and electronics from damage.

The pixel spacing determines the limiting resolution of the sensor. The actual Modulation Transfer Function(MTF) of the detector depends on the type of scintillator that is installed, as well as the binning mode
selected. A thicker phosphor screen will produce more signal, but at the expense of high-frequency contrast.
Typical MTF curves for the standard scintillator option is shown in the graph below.

Resolution & Sensitivity(1)
The Shad-o-Scan detectors are designed
to workwith x-ray
sources operating at
TypicalSensitivity
50kVp
80kVp
a wide range of beam energy settings. X-ray[ADU/µGy]energies as low as 10-15 kV can bedetected. The cameras can be used with x-ray
energies as high as 225
DRZ-Std
9.7 kV, although
13.7
we recommend the use of additional(1) shielding and/or collimation at higher energies
W target, 2mm glass window, no filtrationin order to protect the sensor element and electronics from damage.

MTF for Shad-o-Scan Detectors

100%
90%
80%70%60%50%

The pixel spacing determines the limiting resolution of the sensor. The actual
Modulation Transfer Function (MTF) of the detector depends on the type ofscintillator that is installed, as well as the binning mode selected. A thicker phosphor
screen will produce more signal, but at the expense of high-frequency contrast.
Typical MTF curves for the standard scintillator option is shown in the graph below.
Typical Sensitivity 1 [ADU/μGy]

Low SettingHigh Setting

DRZ-Std

16.2

77.5

(1) W target, 80kV, no filtration

www.teledynedalsa.com/ndt

99µm + DRZ

40%
30%
20%
10%0%

00.5

11.522.5

3Spatial Frequency [lp/mm]

3.544.55

Shad-o-Scan 1501/3001/4501 Datasheet™

CMOS X-Ray Detectors

21Mechanical Drawing 1501 Model
26

B

175190ATeledyne Rad-icon Imaging Proprietary Information

6Mechanical5 Drawing 3001
Model
4
TITLE:
SIZE

6A

5

2

23.9 23.9

16.38

A

DWG. NO.

1501 LC Detector Assy_Mechanical_01

P/N

4
SB1680SHEETREV

1 of 101

3213211DDCBC

183

6.63 6.63

A

EAR99183

B

146.5

C

33.3

28.7
146.5

C

33.3

28.7

19.4 19.4

STATUS LEDSTATUS LED

TRIGGER
TRIGGER

POWER
POWER

COMMUNICATION INTERFACE

18.4 18.4

48.5 48.5

148

148

16.38

SCALE

337
304.24337
304.24Shad-o-Scan 1501 LC

ECCN

COMMUNICATIONINTERFACEHOLE  6 mm
4x M5*0.8 MOUNTING

B

4x M5*0.8 MOUNTING HOLE  6 mm118118

B

UNLESS OTHERWISE SPECIFIED

151517150

17150

ENGINEER

DATE

1. UNITS: MMRCD
1/11/2019
2. DRAWING PERUNLESS OTHERWISE SPECIFIED ENGINEERAPPROVED BYDATEASME Y14.5 2009
1. UNITS: MM
RCD
1/11/2019
3. TOLERANCES:
2. DRAWING PERLINEAR±0.15
APPROVED BY
ASME Y14.5 2009APPLICATION/PROJECTANGLES ±0.5
3. TOLERANCES:
SURFACE FINISH 1.6
LINEAR±0.15
APPLICATION/PROJECTANGLES ±0.5
1.6 IT CONTAINS ARE PROPRIETARY TO TELEDYNE DALSA INC..
THIS DOCUMENTFINISH
AND THE INFORMATIONSURFACEPROPRIETARY

654654www.teledynedalsa.com/ndt

33

TELEDYNE DALSATHIRD ANGLE PROJECTION

TITLE

Everywhereyoulook

TM

THIRD ANGLE PROJECTIONTITLE
DWG NO.

Shad-o-scan

SCALE
DWGNO.

THIS DOCUMENT AND THE INFORMATION IT CONTAINS ARE PROPRIETARY TO TELEDYNE DALSA INC..
THIS DOCUMENT MUST NOT BE DUPLICATED, OR ITS CONTENT USED OR DISCLOSED(IN WHOLE OR IN PART) WITHOUT PERMISSION FROM TELEDYNE DALSA INC.

Everywhereyoulook

TM

Shad-o-scan

THIS DOCUMENT MUST NOT BE DUPLICATED, OR ITS CONTENT USED OR DISCLOSED(IN WHOLE OR IN PART) WITHOUT PERMISSION FROM TELEDYNE DALSA INC.

PROPRIETARYTELEDYNE DALSAMA 3001 SHAD-O-SCAN

High Tech Campus 27 (M/S 14)5656 AE Eindhoven
THE NETHERLANDS
Phone: +31 (0)40 2599000
Fax: +31 (0)40 2599005
High Tech
Campus 27 (M/S 14)
www.teledynedalsa.com
5656 AE Eindhoven
THE NETHERLANDS
Phone: +31 (0)40 2599000
Fax: +31 (0)40 2599005
www.teledynedalsa.com

REV

SUFFIX3322-925-32623 MA 3001 SHAD-O-SCAN
191 - 1 - 1
NTS

3322-925-32623

SHEET
SUFFIXVALID WHEN PRINTED ON A3 SIZE PAPER

SCALE

2NTS

VALID WHEN PRINTED ON A3 SIZE PAPER

2SHEET1 OF191- 1 - 11
1 OF 13SIZE
REV11A33SIZE

A3AAShad-o-Scan 1501/3001/4501 Datasheet™

CMOS X-Ray Detectors

MechanicalDrawing
4501 Model
6542

1489
456.41

16.38
23.9

A

148
6.63

D3C

31183

BC298.5
28.7

STATUS LED

COMMUNICATION INTERFACE
POWER

TRIGGER

18.4

19.4

48.5

185.4

4x M5*0.8 MOUNTING HOLE  6 mm

15118

B

169.5

150

A

Ordering Information
654

All Shad-o-Scan detectors are shipped with a universal input
power supply (90-264V, 50-60Hz) and a power cable. Pleasecontact your nearest sales representative for additional options.
P/N

Teledyne Rad-icon Imaging
765 Sycamore Drive, Milpitas, CA 95035 USA

Description

SB1681-02

Shad-o-Scan 1501 with DRZ-Std Scintillator

SB1661-02

Shad-o-Scan 3001 with DRZ-Std Scintillator

SB1671-02

Shad-o-Scan 4501 with DRZ-Std Scintillator

Americas
EuropeAsia PacificContact Information

Milpitas, CA, USA | +1 408-736-6000 | TDI_sales.rad-icon@teledynedalsa.com
Eindhoven, The Netherlands | +31 40-259-9000 | TDI_sales.sensors@teledynedalsa.com
Tokyo, Japan | +81 3-5960-6353 | TDI_sales.asia@teledynedalsa.com
Shanghai, China | +86 21-3368-0027 | TDI_sales.asia@teledynedalsa.com

www.teledynedalsa.com/ndt

Teledyne DALSA has its corporate offices
in Waterloo, Canada. Teledyne DALSA
reserves the right to make changes at any
time without notice.
© Teledyne DALSA 20220222

1