Product Specifications

The Thermo Scientific CID8725D
radiation hardened Monochrome
camera features new Low Noise,Preamplifier Per Pixel Radiation
Hardened Charge Injection Device(CID) imager technology for use inradiation environments.
Monochrome RS-170 video isavailable via Coax, and digital
video is available via USB2.0.
The compact remote head isconnected to a camera control
unit (CCU) with a flexible cablesupplied to length.

Thermo Scientific CID8725D
Designed for Versatility
The CID8725D camera is designed
specifically for Monochrome imagingin radiation environments.
Camera versions include models foruse in air applications with typicalambient operating temp. conditions,and OEM models designed for use incustomer specific enclosures such asa 40mm diameter tube housing.
Our unique radiation hardened CID
based cameras feature a small
detachable remote head with radiation
tolerance to at least 3 x 106 rads
total dose and low noise operation influx rates up to 1 x 105 rads/hr.
Video output is standard RS-170 format
via the Camera Control Unit (CCU)
BNC connector, or digitally via theUSB2.0 port.
Camera models include CID8725DX6
for operation up to 50 meters remote
distance between the rad. hard headand CCU while the CID8725DX7 offers
remote operation up to 150 meters.

loss due to SETI’ s (single event transferinefficiencies) is minimized.
CID based cameras allow at least anorder of magnitude improvement inoperation when compared to CCD andCMOS based cameras and imagers.
Options include customized packagingand Color versions, as well as
partnered programs for complete systems
for Air or Underwater operation.
Features:. CID (Charge Injection Device)
Radiation Hardened Imager. Exceptional signal to noise and sensitivity.
. 3 x 106 Rads Total Dose (gamma). Excellent image at 1 x 105 rads/hr

Uniqueness of the CID
The radiation hard PPP (PreamplifierPer Pixel) CID based imager technology
allows exceptional signal to noisewith sensitivity never before available
with radiation hardened cameras.
The CID8725D camera has been tested
and proven tolerant to high levelsof gamma radiation, and since charge
readout is within the pixel, charge. Replaceable Remote Head
. USB2.0 digital output. Automatic White Balance

Applications:. Inspection and measurement. Process monitoring, Robotic handling. Hot cell monitoring, Laser beam profiling. Research

Product Specifications

ImagerImage Format
Total Pixels
Pixel Size
Full Well Capacity
Active Area
Optical Format
Electrical
Scanning Format
Resolution
S/N Ratio

Sensitivity
Composite Video
Black Level
White Level
Sync Level
Geometric Distortion
Input Power
Input Voltage
CameraLine Adapter
Input Current
Spectral Response
Gain

Interface
Outputs

Thermo Scientific CID8725D Radiation Hard camera module
730H x 512V
10H x 484V
18.0 x 16.4 micron>100,000 electrons
14.5 mm diagonal
1”

RS-170,30FPS,Interlace>400 TVL (horizontal)-45db typ. signal/RMS
10KHz - 4.2MHz, with3.58MHz trap
.1 lux (for min. video)
1 lux (for maximumoutput) with AGC in,illumination T=2850K
1V p-p, terminatedinto 75 ohm+50mV (Auto Clamp)+700mV-300mV0%
18 Watts (max.)+15VDC Nominal+7/+10VDC(TE)110 - 220 VAC +/10%, 50/60 Hz
Camera 1.2A avg.
SW3 - X2/X4

J1000 Head Interface
J1001 Adv. Features
J1002 Raw Video
P1000 Power Input

Digital Interface

J5 USB2.0 type “B”

Analog Video

J2 “Main” Video Out

MechanicalWeightCable Length
Lens Mount
Camera head caseEnvironmental
Temperature Range
OperatingStorage
Humidity
Shock

The CID8725D Monochrome solid state video camera is part of a proven line ofradiation hardened cameras and sensors whose applications span a full spectrumof industries and applications. Thermo Scientific CIDTEC Cameras &Imagers has been in business for over 25 years with imaging products in scientific,
machine vision, aerospace, medical, and radiation hardened markets.

CCU 0.86 kg. (30 oz.)
Head 0.45 kg. (16 oz.)P/S 1.81 kg. (64 oz.)
DX1 option to 50M
DX2 option to 150M
Standard “C” Mount(1.0” - 32 Thread)Standard Al housing
or TEST/SHIP case

0C to 55C case(DX6)-25C to 85C
0-95% noncondensing
50G (1/2 Sinewave at10ms duration)©2007 Thermo Fisher Scientific Inc. All rights reserved. All other trademarks are the property of Thermo Fisher Scientific Inc. and its subsidiaries. Specifications,terms and pricing are subject to change. For additional specifications visit the Microanalysis and Imaging resource center at: www.thermo.com/cidtec඾ུ࿶ව์คయၑഝ‫ڼ‬৆ٛ২ɧĳĸĴĮıııĶġ୷ဩࡇ஑‫ޘ‬ঌུ಴ĳĮĳĮĸ΍ϋΞΛ·ΫσŕņōĻıĵĸĮĵĴĵĮĳıĺıȁŇłřĻıĵĸĮĵĴĵĮĳıĺĸũŵŵűĻİİġŸŸŸįŢťŴĮŪŮŨįŤŰįūű