X-ray Line-Scan Camera Series

Hawk XID Dual Energy
The X-Scan Imaging XID8800 series of dualenergy linear array x-ray cameras offer highperformance and capability to differentiatematerials in a variety of applications. At theheart of a XID8800 camera are X-Scan
Imaging’s CMOS silicon imaging detector
diode array chips providing wide dynamic
range and solid-state reliability.
A wideselection of filter and scintillation materials
select and convert x-rays for detection by
the diode array and optimizes x-ray energy

discrimination, sensitivity, and resolution. The
proximity of the analog-to-digital converters(ADC) to the detector chips and the use of
low-voltage-differential-signal(LVDS)
technology minimize interference noise. A
collection of hardware for interfacing tocomputers and software including drivers, anintuitive application programming interface(API), and example code software expedite
developments of x-ray scanning systems.

Key Features
Extract material of a target from dual-energy image data
Incorporates X-Scan Imaging’s proprietary XB8800 Photodiode Detectors•High resolution with varieties of filters and scintillators•Low noise, wide dynamic range, high sensitivity, high x-ray
energy contrast•
X-ray energy range options for:oLow x-ray energy range (25 – 100 KeV)oHigh x-ray energy range (45 – 160 KeV)
Variable scan speed with position synchronization
16-bit analog-to-digital conversion
Software development kit•
Device drivers, libraries, standard API

X-Ray Line-Scan Camera Series: Hawk Dual Energy

Applications
Food and industrial inspection requiring high contrast
Composite material sorting and inspection
Security and cargo screening
Waste sorting and recycling
Rare metal or mineral detection
Drug detection and control

Filter Material
Copper 0.250, 0,400, 0.800mm standard
Filter material can be customized.
Adjustable low energy/high energy integration times

ModelActive lengthi

XID88˽˽-12XID88˽˽-18

Number of pixels
XID8804 Series

XID8808 Series

XID8816 Series

307 mm

768 × 2384 × 2

192 × 2461 mm1152 × 2

576 × 2288 × 2

XID88˽˽-24614 mm

1536 × 2

768 × 2384 × 2

XID88˽˽-36922 mm

2304 × 2

1152 × 2

576 × 2

XID88˽˽-48

1229 mm3072 × 2

1536 × 2768 × 2i Other detector lengths are available upon request. Minimum active length is 154mm.

X-Ray Line-Scan Camera Series: Hawk Dual Energy

Mechanical Configurations
X-Scan Imaging housings are available in two form factors. The DR housing is a low profile,wider detector to fit under conveyor systems or other tight spaces. The DS housing is a taller,narrower profile. The standard X-Scan Imaging detectors, Single Energy, Dual Energy, andCMOS TDI all share the same mounting hole pattern.

DR:DS:

X-Ray Line-Scan Camera Series: Hawk Dual Energy

Setup
The XID8800 series camera system includes a camera unit, a software development kit, power
adapter and cabling. The frame-grabber to be installed in the computer is providedoptionally. The objects to be scanned should be passed between the x-ray source and thecamera.

Information furnished by X-Scan Imaging is believed to be accurate and reliable. However, noresponsibility is assumed by X-Scan Imaging Corporation for its use. Users are responsible for theirproducts and applications using X-Scan Imaging components. To minimize the risks associated withusers’ products and applications, users should provide adequate design and operating safeguards. No
responsibility is assumed by X-Scan Imaging Corporation for any infringements of patents or other rightsof third parties that may result from the use of the information. No license is granted by implication orotherwise under any patent or patent rights of X-Scan Imaging Corporation.
© 2018 X-Scan Imaging Corp.