USB 2.0 3648-Pixel 16bit CCD Line Camera with External Trigger(Part Numbers: TCE-1304-U, TCE-1304-UW )FEATURES
 USB2.0 interface
 No external power supply
required Optical integration time
adjustable from 100μs to6.5s
 3648 pixel silicon linear
CCD array 8μm x 200μm pixel size 16-Bit A/D converter forhigh intensity resolution
 Scan rate up to 138 scans/second External trigger capability 4 GPIOs
 SDK for user applications
 Demo software with GUI

PRODUCT DESCRIPTION
Mightex’s TCE-1304-U line camera is a cost-effective high-performance B/W enclosed line camera, based on a single-line, 3648-pixel CCD chip with USB2.0 (480 Mb/s) interface. CCD linecameras have several advantages over their area-array counterparts, including high optical linear
resolution that allows systems developers to use the cameras to capture two-dimensional (2-D)
images by moving the object or the CCD perpendicularly to the scan line. TCE-1304-U is a compact line-scan camera ideal for a variety of OEM applications such as industry process control,
optical spectroscopy and bio-medical imaging etc. Setting up the TCE-1304-U line camera is very
easy: one simply needs to install the camera’s application software into any PC, and then connect
the line camera to the PC using a USB cable. There is no need to install a DAC card or to use an
external power supply.
TCE-1304-UW camera is the window-less version of TCE-1304-U, with the glass window removed from the image sensor. Therefore, the former is more sensitive to UV and is more suitable
for applications that involve coherent light sources.

PERFORMANCE SPECIFICATIONSParametersCCDAPPLICATIONS Industrial Process Control
 Optical Spectroscopy

TCE-1304-U and TCE-1304-UW

UnitToshiba TCD1304DG

Number of pixels

3648

pixelsPixel size

8 x 200μm

Spectral range

TCE-1304-U: 350 to 1000
TCE-1304-UW: 200 to 1000

nm

Pixel output clock

0.5

MHz

Data storage on camera

4frames

ADC resolution

16bitsExternal trigger

Yes

Exposure time range

0.1 ~ 6,500

Number of GPIOs

4 programmable I/O’s

Frame rate

138

Host interface

USB2.0msscans/second** Frame Rate is dependent on exposure time. When exposure time is set to 0.1 ms, the camera can achieve
138 scans/second.

SDK Features
Operation Systems

Windows 2000, XP, Vista and Windows 7

Minimum Requirement

RAM > 64M, hard disk space used > 10M

USB Port
Multiple Cameras
Device Driver
Demo Application
Library Files

2.0
Supported
Yes
Yes
Yes (DLL files and static library file)

Example Codes

Yes (VC++ and Delphi)

Frame Attributes*

Exposure Time, Time Stamp, Trigger Event Count, Over-Exposure Detection.

* SDK will provide call back, which will send user frame data and the related attributes of the frame. The
attributes include: Exposure Time, Time Stamp, Trigger Event Count and Over-Exposure Flag.

Mightex Systems, 2343 Brimley Road, Suite 868, Toronto, Ontario M1S 3L6, Canada
Tel: 1-416-840-4991 or 1-925-218-1885, Fax: 1-416-840-6541, Email: sales@mightex.com
www.mightex.com or www.mightexsystems.com

Mightex
Simply Brighter

USB 2.0 3648-Pixel 16bit CCD Line Camera with External Trigger(Part Numbers: TCE-1304-U, TCE-1304-UW )SPECTRAL RESPONSE (for TCE-1304-U Only)

OPERATION CONDITION
Operating Temperature Range-10°C ~ 50°C

Relative Humidity, Non-condensing

5% ~ 95%

EXAMPLE OF GRAPHICAL USER INTERFACE

Mightex Systems, 2343 Brimley Road, Suite 868, Toronto, Ontario M1S 3L6, Canada
Tel: 1-416-840-4991 or 1-925-218-1885, Fax: 1-416-840-6541, Email: sales@mightex.com
www.mightex.com or www.mightexsystems.com

Mightex
Simply Brighter

USB 2.0 3648-Pixel 16bit CCD Line Camera with External Trigger(Part Numbers: TCE-1304-U, TCE-1304-UW )MECHANICAL DIMENSIONS

Length x Width x Height: 78(mm) x 42(mm) x 57(mm)Weight: 200g
With a world-class OEM design team, Mightex offers a broad range of customized solutions in order to meet individual customer’s unique requirements. Please call 1-416-840 4991 or email sales@mightex.com for details.

【お問い合せ先】〒���-����
東京都千代田区富士見�-�-�ステージビルディング��階TEL ��-����-���� https://www.ads-img.co.jp

Mightex Systems, 2343 Brimley Road, Suite 868, Toronto, Ontario M1S 3L6, Canada
Tel: 1-416-840-4991 or 1-925-218-1885, Fax: 1-416-840-6541, Email: sales@mightex.com
www.mightex.com or www.mightexsystems.com

Mightex
Simply Brighter