Product Specifications

The Thermo Scientific
SpectraCAM scientific grade
imaging system is based on thenew random access Charge
Injection Device (RACID) imager
architecture featuring low noise,
arbitrary pixel selection and readout,
Non Destructive Read-out
capabilities, and collective pixel
readout or binning, and is available
in either 1024 x 1024, or 540 x
540 pixel format.

Thermo Scientific SpectraCAM™

Designed for Versatility
Designed for applications requiringextremly wide linear dynamic range,or adaptive exposure control, theThermo Scientific SpectraCAM imaging
system uses unique random
access integration techniques
employing true pixel addressability,Non-Destructive ReadOut (NDRO),
collective read or binning, and selective
Region of Interest (ROI) clearing toprovide high dynamic range that may
exceed 10(8) or 28bits allowing theuser to read very dim signals in thepresence of highly illuminated ones.
Additional unique camera features
included in the system are artificial
intelligence (AI) algorithms for thereadout or injection of ROIs based on
the experimentally observed signal.
The 16bit thermo electrically cooledSpectraCAM™ system is available ineither a purged, or hermeticallysealed configuiration for single use orOEM applications.

processing. The camera software supports
export of data in various imaging(.FITS, .TIF, .JPEG, and .CID) andspreadsheet formats. Along with
programmable position and size, each
ROI may be individually programmed
for binning and NDRO.
Features:. CID (Charge Injection Device)
Imager technology. High Linear Dynamic Range, up to28bits (with 100 NDRO’ s). True Random Pixel Addressability

Performance Options
The random access CID is sensitivefrom 165nm - 1100nm, and is capableof direct X-Ray imaging.
Application specific coatings such as
GdO2S for imaging of mid to highlevel X-Rays, as well as LUMOGEN
coating for extended UV range below
165nm are available. The camera
controller is based on PENTIUM®
CPU/PC104+ architecture enabling
dynamic control of ROI data acquisition
algorithms and real-time video. Non-Destructive Pixel Readout. Wide Wavelength Response(165 - 1000nm). Collective Non-Destructive Readoutof selectable Binned Regions. Virtually No Blooming Under SevereLight conditions. Contiguous Pixel Structure with High
Fill Factor

Applications:. Spectroscopy. Biological Imaging. Other Scientific Applications

Product Specifications

The SpectraCAM imaging system is part of a line of scientific grade cameras and imagers with applications spanning a full spectrumof industries and applications. Thermo Scientific CIDTEC Cameras & Imagers has been in business for over 25 years with imaging
products in scientific, machine vision, aerospace, medical, and radiation hardened markets.

SpectraCAM86, and SpectraCAM84 scientific imaging systems
Imager:
CID86:
CID84:

Pixel Full Well
>500,000 e>500,000 e-540 x 540 pixels
1024 x 1024 pixels

Readout Noise<240e- single read @ 50KHz<24e- with 128 NDRO’ sBinning
Independent horizontal and verticalcolumns and 4 rows may be read collectively andNon destructively (NDRO).

Pixel Size:
27microns by 27 microns

Dark Current:<7 e- per second per pixel at -45

Linearity+/- 3% from 3% to 80% of saturation

ROI<20usec. delay between ROI selection

Quantum Efficiency>45% @ 550nm and >30% @ 200nm*Deep UV and X-Ray converter coatings available foruse below 165nm

Cooling:
3 stage TE cooler with water recirculation

Readout Speed Selectable:Slow Scan 50 kHz
Fast Scan 200kHz

Sealed, (SpectraCAM84 only):
Hermetic sealed 3 years, Integrated water jacket
Purge:
Purified Argon or Nitrogen at 60 - 80 mL/min

RAI (Random Access Integration) Software Drivers
Fixed Exposure Time: User programmed ROIs integrated for user-selected exposure time analogous to CCD exposure.
Random Access Integration: Integration time automatically varied at ROI site based on experimentally observed signal.
Using non-Destructive read, collective read, and ROI clear features, intensely illuminated ROIs are integrated for multiple short
integration times while weakly illuminated ROIs are simultaneously allowed to integrate for longer periods, allowing for the extensionof the dynamic range up to 108 or 28-bits.
TRS (Time Resolve) Software Drivers (OPTIONAL)TRS Drivers allow for the observation of ROI signal profile with respect to time for multiple ROIs. The algorithms allow user programmabletime resolution as short as 100 microseconds for a single ROI. The algorithms are specifically designed for time-resolved techniques suchas gas and liquid chromatography, flow injection analysis, and laser or spark ablation.
AutoIntegrate Drivers (OPTIONAL)
AutoIntegrate drivers allow the user to automatically combine the complete field of view offered by a full-frame “Fixed-Time” exposure withthe extended dynamic range of the “Random Access Integration” algorithms described above, allowing for 107 dynamic range (24-bits) ona full-frame image without compromising precision.

©2007 Thermo Fisher Scientific Inc. All rights reserved. All other trademarks are the property of Thermo Fisher Scientific Inc. and its subsidiaries. Specifications,terms and pricing are subject to change. For additional specifications visit the Microanalysis and Imaging resource center at: www.thermo.com/cidtec඾ུ࿶ව์คయၑഝ‫ڼ‬৆ٛ২ɧĳĸĴĮıııĶġ୷ဩࡇ஑‫ޘ‬ঌུ಴ĳĮĳĮĸ΍ϋΞΛ·ΫσŕņōĻıĵĸĮĵĴĵĮĳıĺıȁŇłřĻıĵĸĮĵĴĵĮĳıĺĸũŵŵűĻİİġŸŸŸįŢťŴĮŪŮŨįŤŰįūű