”View areasthat are out of reach””Lightweightand fully portable”

RADCAMLoki®

”For use in various
application areas”

Radiation tolerant inspection camera platform.
Designed to allow the operator to view areasthat are normally out of reach.

Reaching new heights with Loki
RADCAM® Loki is ISEC’s newest radiation tolerant inspection camera system,
which is designed to allow the operator to view areas that are normally outof reach without having to erect scaffolding.
This is done by incorporating a pole that is extendable up to 6 meters (20 ft)
as well as adding tilt functionality to the camera head via its tilt motor.
The on-board display tablet allows the operator to view and record live
video in HD.
Gamma dose rate measurement probe can also be modularly added to theRADCAM Loki which further expand the range of application by enabling
operators to capture simultaneously visual and gamma radiation data.
The operator can then add his or her own notes (or even draw) directly
onto the tablet thanks to LOKI´s touchscreen display. All video and/or still
images can then be transferred to other media devices allowing an easytransfer of information between departments or for job briefings. With thenew LOKI from ISEC, nothing is out of reach.

Find out more at www.isec.se

Strengthen ALARA
Increase SAFETY
Reduce COSTS
Save TIME

RADCAM®
Loki

See the differencewith RADCAM® Loki
• Designed for safe, efficient andcost-effective inspections• Ready-to-go inspection tool
• User-friendly and intuitiveinterface• High-quality radiation tolerant
camera• Clean and contaminated environment compatible• Easy metadata generation andexport• Qualified optional gamma
dose-rate measurement

Simplicity••••Easily extendable pole
Suitable for all inspections
Easy media transfer
User-friendly documentation
management software

Excellence• 180° view, HDTV 720P camera• LED lights for perfect viewing
in dark places• Lightweight, 2.5 Kg, design• Industrial grade tablet

Endurance••••

Gamma Radiation tolerant
6 hours continuous operation
Robust construction
Contamination protectionaccessories

RADCAM®
Loki
TECHNICAL SPECIFICATIONS
CameraSensor

1/4” CMOS color sensor - HD (1280x720p)

Minimum lighting

0.5 Lux @ F2.5

Optical zoom

NoDigital zoom

Yes

Angle of view45°

Focus / Iris

Manual / fixedPhysical features
Pan camera head

Tilt – 180° view

Lamps

2 LED lights following the tilt

Dimentions

H: 12 cm (4.7”) L: 210 – 610 cm (6ft 11" – 20ft) W: 22 cm (8.7")

Weight

2.5 Kg (5.5 Lb.)

Pole reach

From 210 up to 610 cm (6ft 11" – 20ft),
several pole lengths possible

Power supply 8” rugged tablet

Rechargeable battery, 8500mAh, 8 hours continuous
operation

Power supply battery pack

Rechargeable battery, 6 hours continuous operation

Communication
Media format

H264, JPEG

Frame rate

Max 25/30 fps

Media storageTablet, USB, SD card

Environment
Operating temperature-20 °C to 50 °C (-4 °F to 122 °F)

Humidity

10–85% RH (non-condensing)

Radiation
Protection typeGamma resistance (Co60)

Dose/dose rate

> 1000 Gray | > 100 Gray/h

Other
Visual inspection

Fulfills reqirements for VT-3 / VT-1

ExportUSB, SD card, WiFi or Bluetooth 4.0

ACCESSORIES
Dose rate detector
Measurement range H*(10)(gamma, X-ray)Medium flux:
10 µSv/h to 99.9 mSv/h (1 mRem/h to 9.99 Rem/h)
Low flux:
0.5 µSv/h to 9.99 mSv/h (50 µRem/h to 0.99 Rem/h)

TypeCompensated GM-tube

Mounting

Modular mount – Loki adapted

Display/recording

Live overlay display/recording on Loki Tablet GUI

Contamination protection
TypeDisposable protection sleeve and special camera moduleprotection

Material

Made of nuclear approved material (PMUC or equivalent)
ISEC Monitoring Systems AB
Diabasgatan 12 | SE-254 68 | Helsingborg | SWEDEN
SB: +46 42 334 800 | www.isec.se

The innovative,
cost-efficient choicein nuclear monitoring
systems