Product Specifications

The Thermo Scientific CID8825D
radiation hardened COLOR camerafeatures new Low Noise,Preamplifier Per Pixel Radiation
Hardened Charge Injection Device(CID) imager technology for use inradiation environments.
Color NTSC video is available via
Coax, or RGB connectors. Digital
video is available via USB2.0.
The compact remote radiation
hardened head is connected to acamera control unit (CCU) with aflexible cable supplied to length.

Thermo Scientific CID8825D
Designed for Versatility
The CID8825D is the only rad. hardsolid state camera capable of Colorimaging in radiation environments.
Cameras for use in air applications
with typical ambient operating temp.
conditions, and OEM models for use
in customer specific enclosures suchas 40mm diameter tube housing areavailable.
The unique radiation hardened CID
based cameras feature a small
detachable remote head with radiation
tolerance to at least 3 x 106 rads
total dose and low noise operation influx rates up to 1 x 105 rads/hr.
Video output is standard NTSC format
via the Camera Control Unit (CCU)
BNC connector, R,G,B connectors, ordigitally via the USB2.0 port.
Camera models include CID8825DX6
for operation up to 50 meters remote
distance between the rad. hard head
and CCU while the CID8825DX7 offers
remote operation up to 150 meters.

transfer inefficiencies) is minimized.
CID based cameras allow at least anorder of magnitude improvement inoperation when compared to CCD and
CMOS based cameras and imagers.
Options include monochrome andcustomized packaging, as well aspartnered programs for complete systems
for Air or Underwater operation.

Features:. COLOR CID (Charge Injection Device)
Radiation Hardened Imager. Exceptional signal to noise and sensitivity.
. 3 x 106 Rads Total Dose (gamma). Excellent image at 1 x 105 rads/hr

Uniqueness of the CID
The radiation hard PPP (PreamplifierPer Pixel) CID imager technology allow
exceptional signal to noise with sensitivity
never before available with radiation
hardened cameras.
These cameras have been tested andproven in high levels of gamma radiation,and since readout is within thepixel, loss due to SETI’ s (single event. Replaceable Remote Head. USB2.0 digital output. Automatic White Balance

Applications:. Inspection and measurement. Process monitoring, Robotic handling. Hot cell monitoring, Laser beam profiling. Research

Product Specifications

ImagerImage Format
Total Pixels
Pixel Size
Full Well Capacity
Active Area
Optical Format
Electrical
Scanning Format
Resolution
S/N Ratio

Sensitivity

Composite Video
Black Level
White Level
Sync Level
Geometric Distortion
Input Power
Input Voltage
CameraLine Adapter
Input Current
GainInterface
Outputs

Digital Interface
Analog Video

Mechanical
WeightCable Length
Lens Mount
Camera head caseEnvironmental
Temperature Range
OperatingStorage
Humidity
Shock

Thermo Scientific CID8825D Color Radiation Hard camera module
730H x 512V
710H x 484V
18.0 x 16.4 micron>100,000 electrons
14.5 mm diagonal1”

The CID8825D COLOR solid state video camera is part of a proven line of radiation
hardened cameras and sensors whose applications span a full spectrum ofindustries and applications. Thermo Scientific CIDTEC Cameras & Imagers has
been in business for over 25 years with imaging products in scientific, machine
vision, aerospace, medical, and radiation hardened markets.

NTSC, 30FPS, Interlace>380 TVL (horizontal)-45db typ. signal/RMS
10KHz - 4.2MHz, with
3.58MHz trap
10 lux (for min. video)
20 lux (for maximumoutput) with AGC in,illumination T=2850K
1V p-p, terminatedinto 75 ohm+50mV (Auto Clamp)+700mV-300mV0%
18 Watts (max.)+15VDC Nominal+7/+10VDC(TE)110 - 220 VAC +/10%, 50/60 Hz
Camera 1.2A avg
X2/X4

J1000 Head Interface
J1001 Adv. Features
J1002 Mono Video
P1000 Power Input
J5 USB2.0 type “B”
J4 (R) Composite Video
J3 (G)J2 (B)

Accumulated Dose: 30,060 Gy (3.06 MRad) in 279kRad flux rate.
Irradiation test results at University of Maryland

CCU 0.86 kg. (30 oz.)
Head 0.45 kg. (16 oz.)P/S 1.81 kg. (64 oz.)DX6 option to 50M
DX7 option to 150M
Standard “C” Mount(1.0” - 32 Thread)Standard Al housing
or TEST/SHIP case

0C to 55C case (DX6)-25C to 85C
0-95% noncondensing
50G (1/2 Sinewave at10ms duration)©2007 Thermo Fisher Scientific Inc. All rights reserved. All other trademarks are the property of Thermo Fisher Scientific Inc. and its subsidiaries. Specifications,terms and pricing are subject to change. For additional specifications visit the Microanalysis and Imaging resource center at: www.thermo.com/cidtec඾ུ࿶ව์คయၑഝ‫ڼ‬৆ٛ২ɧĳĸĴĮıııĶġ୷ဩࡇ஑‫ޘ‬ঌུ಴ĳĮĳĮĸ΍ϋΞΛ·ΫσŕņōĻıĵĸĮĵĴĵĮĳıĺıȁŇłřĻıĵĸĮĵĴĵĮĳıĺĸũŵŵűĻİİġŸŸŸįŢťŴĮŪŮŨįŤŰįūű