High-output UV-LED Light
LNSP-UV-FN Series

Ideal for UV-Excitation Applications

Select the Best Directional Characteristics for the Application
LNSP-UV365-FNNR

LNSP-UV365-FN

Wide Models

Narrow Models

Emitting Surface Lengths of 100, 200, and 300 mm

CCS Inc.

LN S P- U V- FN

S er i e s

Wide Range of Application with Both Wide Irradiation and Narrow Irradiation Models
Wide Models
Provides diffused illumination over a wide area.

Output Comparison

LNSP-UV365-FNNR
Previous model(LDL-74x27UV365)
LNSP-UV365-FNNR

Approx. 40 x

*The brightness demends on the spectral sensitivity.

Application Examples
White Light

Image of Foreign Matter on Paper

LNSP-300UV365-FNNR
Dust and other foreign
matter on the paper canbe captured. The paperabsorbs ultraviolet
lightwaves, so only theforeign matter disperses
the light to provide
high-contrast images.

White Tissue Paper

Blue Light

Image of Contact Lenses

LNSP-300UV365-FNNR
The presence of packaged
contact lenses can bedetected. Some types ofcontact lenses absorb
ultraviolet lightwaves,which produces
high-contrast images ofthe packed contact lenses.

Packed Contact Lenses

Technical Data

*Actual measurement values. (These values are for reference only.)

Brightness Distribution-50-200-150-100-500(mm)50LWD =50 mm

10015050403020100-10-20-30-400-50-200200Spectral Distribution-150-100-500(mm)50LWD = 100 mm(mm)
100

100

150

50
4030
20100-10-20-30-400-50-200200

Distance (LWD) Characteristic for Perpendicular Irradiation

-150

355

360

365

370375380385390395400

100500550100150LWD (mm)LED illuminators with a 385 nm wavelength are also available as custom products.
*The data provided here is for reference only. Results for individual Units may vary.

200-500(mm)501001500200

LNSP-300UV365-FNNR

150
Relative output (%)

Relative energy (%)

Wavelength (nm)Irradiance strength (mW/cm2)

350-100

Output Changes Over Time

LNSP-300UV365-FNNR
10090807060
50403020100345

100

Output level (%)

(mm)
100
Output level (%)

50403020100-10-20-30-40

Output level (%)

LWD = 30 mm(mm)

250

300

10090807060504030201000102030Time (min)405060LNSP-UV-FN Series of UV-LED Lights
LNSP-UV365-FN
Output Comparison

Narrow Models
Convergent irradiation in a narrow area.
Intensity loss is minimal, enabling long irradiation distances.

Previous model(LDL-74x27UV365)

Approx. 150 x

LNSP-UV365-FN*Output will vary based on the camera’s spectral response.

Application Examples
Image of Alignment of Transparent Film Blue Light

LNSP-300UV365-FN

The alignment oftransparent film on atransparent board can bechecked.
Only the transparent film
disperses the light, so theedges can be picked up ina high-contrast image ofthe transparent film.

Transparent Board on
Bottom with Film on Top

White Light

Image of Invisible Codes

LNSP-300UV365-FN

A code that is printed withinvisible ink can beconfirmed.
The invisible ink reacts towavelengths in theultraviolet range. Irradiating
the printed section
produces a high-contrast
image of the code.

Plastic Plate with CodePrinted in Invisible Ink

Technical Data

*Actual measurement values. (These values are for reference only.)

Brightness Distribution-50-200-150-100-500(mm)50LWD =50 mm

100150200Spectral Distribution

50403020100-10-20-30-400-50-200-150-100-500(mm)

LWD = 100 mm(mm)10050100

15020050403020100-10-20-30-400-50-200

Distance (LWD) Characteristic for Perpendicular Irradiation

-150

355

360

365

370

375

380

385

390

395

400

400
300
200
1000550

100150200

LWD (mm)

LED illuminators with a 385 nm wavelength are also available as custom products.
*The data provided here is for reference only. Results for individual Irradiators may vary.

-500(mm)

501001500200LNSP-300UV365-FN

500
Relative output (%)

Relative energy (%)

Wavelength (nm)Irradiance strength (mW/cm2)

350

-100

Output Changes Over Time

LNSP-300UV365-FN
10090807060
50403020100345

100
Output level (%)

(mm)
100

Output level (%)

50403020100-10-20-30-40

Output level (%)

LWD = 30 mm(mm)

250

300

10090807060
504030201000102030Time (min)405060Dimension Diagrams (mm)(300 mmConnector

Metal Connector (PRC04-12A26S-37M18)

Wide Models

n+36.3: Overall lengthn: Emitting surface

34

Safety standards: Conforms to EN 62471, EMC standard: Conforms to EN61000-6-2 and EN 61000-6-4.

Environmental regulation

RoHS compliantCooling method

Forced air cooling

Accessories

Frame nuts (4), 2-m FG line (1), M3 set screw (1)For mounting
Four nut slots
Detail View A

30
031 W

800 g

200 mm61 W

1,100 g1006244

LNSP-300UV365-FNNR

300 mm92 W

1,400 g

LNSP(UV365-FN: Emitting surface length)

Narrow Models(6.7)

n+39: Overall lengthn: Emitting surface

341006251

LNSP-100UV365-FN100 mm

31 W

1,000 g1006252

LNSP-200UV365-FN

200 mm61 W

1,400 g1006167

LNSP-300UV365-FN

300 mm92 W

1,800 g16(74.6)

6For mounting
Four nut slots
Detail View A

25.8Weight(max.)(13.6 dia.)

35

Model

30
0

Power
Light-emitting
consumption
surface(max., includinglengthfans)

DirectnumberNarrow Models LNSP-UV365-FN

62

LNSP-UV365-FN

Detail View A

Modeln

LNSP-100UV365-FN
LNSP-200UV365-FN
LNSP-300UV365-FN

100200300

19.5100 mm

LNSP-200UV365-FNNR

Fan air inlet

No. ofcooling fans12341.5

16LNSP-100UV365-FNNR

1006254

No. ofcooling fans1234

2×M3 (To attach FG line, same on opposite side.)
9.4 (emitting surface)

1006253

Weight(max.)

n100200300

1.5(79.8)

PowerLight-emitting
consumption
surface(max., includinglengthfans)

Model

27.3

(13.6 dia.)

LNSP-UV365-FNNR

DirectnumberModel
LNSP-100UV365-FNNR
LNSP-200UV365-FNNR
LNSP-300UV365-FNNR

14.3

15.5

Temperature: −20 to 60°C, Humidity: 20% to 85% (with no condensation)

CE Marking

Wide Models LNSP-UV365-FNNR

62

Storage environment

Narrow Models(6.7)

Temperature: 0 to 40°C, Humidity: 20% to 85% (with no condensation)

Operating environment

Wide Models

UV365-FNNR: Emitting surface length)

12

LNSP-

Aluminum alloy, steel plate, and quartz glass (on Narrow Models only)

Cable length

12Ultraviolet: 365 nm typical

Case material

6

Peak wavelength

30 (emitting surface)

Specifications

Fan air inlet
Detail View A

2×M3 (To attach FG line, same on opposite side.)

Options
LED Light Unit Cables
These cables are used to connect
LED Light Units to Control Units. Use
the Cable that is suitable for yourinstallation site.

Ultraviolet Transmission Filters

QCB Series
Direct number

3000815

3000816

3000817

3000818

3000819

Model

QCB-2

QCB-3

QCB-5

QCB-10QCB-20

Cable length

2m

3m

5m

10 m

20 m

Weight (max.)
Cable diameter: 16.5 mm
Allowable cable bending radius: 99 mm

1.1 kg

1.5 kg

2.4 kg

4.6 kg

8.9 kg

U340 Series

Ultraviolet Cutting Filters
L42 Series

ModelSize
U340-25 M25.5 P0.5
U340-27 M27.0 P0.5
U340-30 M30.5 P0.5
U340-40 M40.5 P0.5
U340-46 M46.0 P0.75

Model
L42-25
L42-27
L42-30L42-40
L42-46Size
M25.5 P0.5
M27.0 P0.5
M30.5 P0.5
M40.5 P0.5
M46.0 P0.75

Control Units for LNSP-UV-FN Series
Dimension Diagram (mm)Features

Digital display

371

131

Key lock

EIA-485 communications connector

Fan exhaust outlet

Output connector(37-pin metal connector)

Four, M4 bottom mounting screws(Insertion depth of 5 mm max.)

192

AC inlet

16.5

Power switch

18

Optional External Control Cables
Dimension Diagrams (mm)335

These Cables are used for parallel or EIA-485 communications.
Select the right cable for the required control method.

Parallel Communications Cable

ON/OFF Input Cable

EIA-485 Communications Cable

Direct number: 3000683
Model name: EXCB2-M20-3

Direct number: 3000682
Model: EXCB2-M10-3

Direct number: 3000685
Model: EXCB2-E3-3

3000

21920

1 26.1 dia.

9 1019 2020-pin MIL connector

Cut off
on one end.

3000

215 dia.

110910-pin MIL connector

123Cut off
on one end.

123e-CON connector, 3 pins

3.9 dia.

3000
1 2

EIA-485 Communications Junction Cable

Parallel Communications and ON/OFF Input Branch Cable

Direct number: 3000721
Model name: EXCB2-E3-E3-0.2

Direct number: 3000684
Model name: EXCB2-M10M20-3

200e-CON connector, 3 pins

300013.812920-pin MIL Connector
1101 29 10Relay Connector
Direct number: 3000720
Model: ECNR-E3CN4

1 230

Cut off
on one end.

10010-pin MIL connector

123

3.9 dia.

1232e-CON relay connector,3-pin, 4×4

13PSCC-60048
2000846
Constant lighting
Constant-current system
Variable current control
1 channel
43 VDC max., 582 W max. (including 30 W max. for fans)
Manual and external intensity control
Front manual/external switch (MODE)
Set any of 256 levels via the setting switch. Press and hold the switch for 2 seconds to lock the intensity value.
Parallel communications
8-bit intensity value setting (B0 to B7) and write signal (WR)EIA-485 communications
Command input through EIA-485 communications
External
Ethernet communications
Command input via TCP/IP or UDP/IP communications
External control mode can be selected by pushing the setting switch while turning ON the power to the Control Unit.
Parallel bit input
OFF signal (ON/OFF)
Lighting control
EIA-485 communications
Command input via EIA-485 communications
Ethernet communications
Command input through TCP/IP UDP/IP communications
EIA-485 communications ID
Set via the front ID switch (00 to 03). Maximum of 4 connected Units.
settings
Terminating resistanceSet via the front ID switch. (Terminating resistance is connected only when ID is set to 00.)
Front digital “E01” display
Error detection display LED burnout detection, open circuit
LED burnout detection, short circuit
Front digital “E02” display
Light Unit fan slowdown or stoppage
Front-panel digital “F01” to “F15” display
Communications error detection
Front digital “E04” displayConnector unconnected detection
Front digital “E04” display
Internal Control Unit error detection
Front digital “E05” display
Error detection output Parallel communications
Output to pins 19 and 20. Photocoupler isolation. Open-collector output. Closed for alarm (Load current: 10 mA max.)
EIA-485 communications Confirmed with status command via EIA-485 communications. (Command sent at error occurrence.)
Ethernet communications Confirmed with status command via TCP/IP or UDP/IP communications. (Command sent at error occurrence.)
100 to 240 VAC (+10%, −15%), 50/60 Hz
Input power
Power consumption (typ.) 750 VA
Operating temperature and humidity Temperature: 0 to 40°C, Humidity: 20% to 85% RH (with no condensation)
Storage temperature and humidity Temperature: −20 to 60°C, Humidity: 20% to 85% RH (with no condensation)
Forced air cooling
Cooling method
Safety standard: Conforms to EN 61010-1, EMC standard: Conforms to EN 61326-1, Class A.
CE Marking
Environmental regulation RoHS compliant
Material, coating, and surface processing Steel plate, thickness of cover: 1.0, thickness of chassis: 2.0, N3 leather tone finish
7,000 g max.
Weight2 meter long 3-prong power cord with ground terminal (1), keys (2)Accessories

ID switch

Parallel communications
connector

Specifications
Model
Direct numberLighting method
Drive method
Lighting method
The number of channels
Applicable Light Unit rating
Light control
Manual control

Air inlet

225

ON/OFF input
connector▪ Constant-current system, Control to 256 light intensities, 582-W output.
▪ 1 channel/1 connector (37-pin metal connector)▪ Ethernet, parallel, or EIA-485 communications for external control.
▪ External controls (Dimming control and ON/OFF Control)▪ Error detection for cooling fan error, LED open circuit, LED short circuit, etc.
▪ Interlock with key switch or external control via parallel communications

Ethernetcommunications
connector

Manual/external
mode selector

Setting switch

7.2 dia.

Analog Control Unit for LED Light Unit: PSCC-60048

19 20

19

Cut off
on one end.

20

CCS and LIGHTING SOLUTION are all registered trademarks or trademarks of CCS, Inc.

CautionTo ensure proper and safe use of the product, please read the Instruction Guide completely before using the product.
For product improvement, specifications and designs are subject to change without notice.

Headquarters
Shimodachiuri-agaru, Karasuma-dori, Kamigyo-ku, Kyoto 602-8011 Japan
Phone: +81-75-415-8284 / Fax: +81-75-415-8278
URL: http://www.ccs-grp.com

E-mail: intlsales@ccs-inc.co.jp

Copyright(c) 2012 CCS Inc. All Rights Reserved.
Descriptions in this catalog are based on information available as of December 2012. 02002-00-1212-LNSP-UV