UF シリーズ
超音波流量計
IM-F2208-J01

取扱説明書

2 版 2023 06 E
初版 2021 11 E

＜目

次＞

UF シリーズ
超音波流量計目次

はじめにお読みください 本書で使用しているマークについて ............................................................................................................ Ⅰ 一般的な注意事項 ......................................................................................................................................... Ⅰ 電気的接続について ..................................................................................................................................... Ⅱ 材質について ................................................................................................................................................ Ⅱ ガラス、樹脂を使用している製品について ................................................................................................. Ⅱ ガラス管・樹脂管面積流量計の使用について ............................................................................................. Ⅲ 防爆仕様で納入された製品について ............................................................................................................ Ⅲ 保守、点検について ..................................................................................................................................... Ⅲ
1. 概要 ....................................................................................................................................................................... 1
2. 製品内容 ................................................................................................................................................................ 1
3. 一般的注意事項 ..................................................................................................................................................... 2
3.1 電源 .................................................................................................................................................................................... 2
3.2 測定可能流体 ..................................................................................................................................................................... 2

4. 設置 ....................................................................................................................................................................... 3
4.1 設置 .................................................................................................................................................................................... 3
4.1.1 設置場所 .................................................................................................................................................................................. 3
4.1.2 配管上の取付位置 ................................................................................................................................................................... 4
4.1.3 UF シリーズ外形図 .................................................................................................................................................................. 5

4.2 配線 .................................................................................................................................................................................... 6
4.2.1 使用ケーブル ........................................................................................................................................................................... 6

5. 運転 ....................................................................................................................................................................... 7
5.1 運転準備............................................................................................................................................................................. 7
5.2 通電 .................................................................................................................................................................................... 7
5.3 ゼロ点調整 ......................................................................................................................................................................... 8
5.4 運転中の表示 ..................................................................................................................................................................... 8
5.5 運転中のエラー.................................................................................................................................................................. 8

6. パネル操作 ............................................................................................................................................................ 9
6.1 操作・表示箇所.................................................................................................................................................................. 9
6.2 モードの切替 ................................................................................................................................................................... 10
6.3 ゼロ点調整 ....................................................................................................................................................................... 11
6.3.1 ゼロ点調整方法 ..................................................................................................................................................................... 11

6.4 パラメータの設定方法..................................................................................................................................................... 12
6.4.1 設定モードのパラメータ構成............................................................................................................................................... 12
6.4.2 設定項目の選択 ..................................................................................................................................................................... 13
6.4.3 パラメータ一覧表 ................................................................................................................................................................. 14
6.4.4 パラメータの確認・変更 ...................................................................................................................................................... 15

6.5 パラメータ詳細................................................................................................................................................................ 17

＜目

次＞

7. 機能説明 .............................................................................................................................................................. 20
7.1 基本項目の設定................................................................................................................................................................ 20

8. 日常点検 .............................................................................................................................................................. 24
8.1 チューブ継手、接続部の点検.......................................................................................................................................... 24
8.2 接続配管の点検................................................................................................................................................................ 24
8.3 防水性の点検 ................................................................................................................................................................... 24
8.4 検出器の点検 ................................................................................................................................................................... 24

9. トラブルシューティング .................................................................................................................................... 25

＜はじめにお読みください＞ Ⅰ

はじめにお読みください
このたびは弊社製品をご採用いただき、まことにありがとうございます。
この取扱説明書には本製品の設置方法、取扱い上の注意事項等が記載されていますので、ご使用前に必ずご一読ください。

■ 本書で使用しているマークについて
本書は、弊社製品のご使用に際しお客様にご注意いただきたい内容について記載しています。
この記載内容は弊社全製品に共通する事項となります。
次の表示の区分は、表示内容を守らずに誤って使用をした場合に生じる危害や損害の程度を説明しています。

警告 この表示は、取り扱いを誤った場合に「死亡または重傷を負う可能性が想定される」内容です。
注意

この表示は、取り扱いを誤った場合に「軽傷を負う可能性または物的損害の発生が想定される」内容です。

注記

弊社製品を安全かつ正しくご使用いただくための内容です。

■ 一般的な注意事項

警告●製品は工業計器としての用途にのみ使用し、その他の用途には使用しないでください。
●製品は工業計器として最適な品質管理のもとに製造、調整、検査を行い納入しております。みだりに改造や変更を行うと本来の性能を発揮できないばかりか、不具合や事故の原因となります。改造や変更は絶対に行わないでください。改造や変更の必要がある場合は弊社までご連絡ください。
●仕様書に記載された仕様範囲内での使用を厳守してください。この範囲を超えた条件での使用は故障、破損の原因となります。
●設置作業の際は必ず安全靴、手袋、保護メガネなどの防護手段を講じてください。
●プロセスへの設置・接続の際は必要に応じてプラントあるいは装置の停止を行ってください。
●重量の大きな製品の設置は落下による人体・器物などへの損傷または過大な衝撃、破損などが生じないよう吊下方法を含めた安全措置を講じてください。また、製品設置箇所では必要に応じて配管サポート等の処置を行ってください。

注意●製品の運搬は納入時の梱包状態で行ってください。
運搬作業時は製品の落下による人体・器物などへの損傷または過大な衝撃による破損などが生じな
いよう安全措置を講じてください。
●開梱後、製品の中には、水、埃、砂などを入れないでください。
●プロセスへの設置・接続に必要な締結部品のボルト、ナット、ガスケット（パッキン）は、原則としてお客様の所掌となります。圧力、温度などの仕様や耐食性を確認して適切なものを選定してください。
●プロセスへの設置・接続の際は、接続継手の規格・寸法合わせが正しいか確認し、接続配管との偏芯、フランジの倒れがないように設置してください。正しく行われない場合は製品の故障、誤動作、破損などの原因となります。

＜はじめにお読みください＞ Ⅱ

注記●保管の際は納入時の梱包状態で保管してください。保管の環境については本書を参照ください。
●設置後、製品を「足場」として使用するなど、荷重を掛けないでください。故障、破損の原因となります。
●製品に貼付されているラベルに表示されている注意事項は、必ず守ってください。
●製品は最適な品質管理のもとに製造、調整、検査を行い納入しておりますが、不測の要因で故障が発生する可能性もあります。運転・安全上の重大な問題が発生するプロセスにおいては、万が一に備えて同様な機能を果たす機器を併設、二重化を行うなど、より一層の安全性の確保を推奨します。

■ 電気的接続について警告●電気配線（結線）に際しては仕様書、本書などに記載されている内容を確認のうえ、正しく配線（結線）してください。誤配線（結線）は機器の故障の原因となるばかりでなく、事故の原因となることがあります。また、配線（結線）作業の際は電源が遮断されていることを確認し感電に注意してください。
●電源を接続する製品の場合は、仕様書、本書を参照して電圧および消費電力を確認して適合する電源を接続してください。適合する電源以外の電圧の電源に接続した場合、機器の破損や作動の不具合、事故につながる恐れがあります。
●通電中は、感電事故防止のため内部の機器には絶対に触れないでください。

注意●設置工事から電気配線作業完了にいたる間、雨水などが製品内に入らないよう注意してください。
また、配線完了後は遅滞なく正しく防水措置を実施してください。

■ 材質について注意●材質の指定がない場合には使用条件・運転条件から最適な材質選定に努めておりますが、実際のプロセスにおける使用条件・運転条件につきましては知見できないこともあります。最終的な材質の決定および耐食性や適合性の確認はお客様の責任で行ってください。製品の材質は仕様書に記載されています。

■ ガラス、樹脂を使用している製品について警告

注意●製品の接液部または測定部、表示部の材質にガラス、樹脂を使用している場合、過度の加圧、温度
衝撃、急激な流体の流入の衝撃圧などによりガラス、樹脂が破損する場合があります。
万が一破損した場合、ガラス、樹脂などの破片が飛散するなどして二次災害および作業者に危険が及ぶ恐れがあります。破損の原因となるような運転条件にならないように注意してください。
また、飛散防止の措置を行ってください。

●運搬、保管および運転に際しては、ガラス部、樹脂部に機械的衝撃を与えないように注意してください。
●ガラスはアルカリ系溶剤で侵食されます。アルカリ系溶剤は使用しないでください。
●樹脂は溶剤系の液体で破損することがあります。仕様書、本書などに記載されている流体以外には使用しないでください。
●樹脂は使用環境により劣化が早まることがあります。設置ならびに運転にあたっては、樹脂の耐食性、紫外線耐性などの耐環境性に考慮してください。

＜はじめにお読みください＞ Ⅲ■ ガラス管・樹脂管面積流量計の使用について
ガラス管・樹脂管面積流量計は以下の事項に配慮して使用してください。

警告

注意●以下の流体条件および使用環境では、ガラス管・樹脂管面積流量計は不適ですので設置しないでください。
・衝撃圧力がある、あるいは衝撃圧力が予想されるプロセス
・万が一ガラス管/樹脂管が破損した場合、二次的な災害が予想されるプロセス－毒性（刺激性、麻酔性などを含む）のある流体－引火性のある流体－爆発性のある流体・ガラスが破損した時にガラス片が飛散し、人身事故などが考えられる場合
・設置場所が、外部からの飛散してきた異物などでガラスの破損が考えられる場合・運転が ON／OFF 運転で、フロートが急上昇し、その衝撃でガラスが破損すると考えられる場合
・流量計に温度衝撃（急冷／急騰）が加わる、あるいは温度衝撃が予想されるプロセス●接液部または測定部にガラスおよび樹脂を使用している製品において、運転停止に伴い流れが停止
して測定液体が測定管内に残留した場合、周囲温度が氷点下になると液体が凍結してガラス、樹脂を破損する恐れがあります。
（一般的には冬期に運転停止して液抜きをしないなど）運転停止中に測定液体が凍結する恐れがある場合は、液体を完全に抜き取ってください。
●樹脂は一般的に金属に比較して機械強度が低く、取扱いには注意が必要です。設置の際は接続配管・継手の寸法違い、偏芯、過大な締結トルクでねじ込むことなどによる機械的応力が加わらないよう注意してください。

■ 防爆仕様で納入された製品について警告●該当する法規・規則・指針に適合した配線、接地工事を確実に実施してください。また、構造の改造、電気回路の変更などは法令違反であり規則・指針に適合しなくなるので絶対に行わないでください。保守・点検については法令・規則・指針に従い、作業を実施してください。

注意●製品の防爆等級は仕様書、製品の銘板に記載されています。対象ガスおよび設置場所が防爆関連法
規・規則・指針に準拠するか確認してください。

■ 保守、点検について警告●製品を保守、点検などでプロセスから取外す際は、測定対象の危険性・毒性に留意して作業を行ってください。関連する配管・機器類からの漏れおよび残留などにより人体・機器類への損傷が生じないよう注意してください。
●電気を使用している製品では感電事故防止のため、電源が遮断されていることを確認してください。

注記●製品の保守、点検については使用条件・運転条件などによりその周期、内容が異なります。
本書を参照の上、お客様にて実際の運転状況を確認して判断してください。

1
1. 概要
UF シリーズ超音波流量計は、超純水や純水、水などの流量測定用に設計された流量計です。接液部はすべてストレートの PFA チューブで構成され、圧力損失はほぼゼロで可動部がなく、液溜まりを生じやすい O リングなどの機械的シール部分もありません。理想的なクリーン構造で、半導体装置などの高度なクリーン度を要求されるプロセスに最適な流量計です。４桁流量表示と流量出力、警報出力および通信機能を搭載しており、現場指示と通信を用いた集中管理が可能です。

2. 製品内容
UF シリーズは以下の内容で納入されます。不足・欠損などがあった場合は速やかにお買い求め先へご連絡ください。
番号

内

容

数量

1

UF シリーズ本体

1

2

専用ケーブル

1

3

パラメータリスト

1

4

検査成績書（注 1）

（注 1）

5

取扱説明書（注 1）（注 1）

注 1：ご要望に合わせて添付します。

IM-F2208-J01

2
3. 一般的注意事項
3.1 電源注記
接続する電源は下記内容にご注意ください。
● 電圧
DC12〜24V±10%の電源を使用してください。
● 他の電源との共用
電源は必ず計装用電源を使用し、動力用電源との共用は避けてください。
● ノイズ
近傍にインバータなどのノイズ源がある場合は、ノイズフィルタにより発生源側でノイズを確実に除去してください。
● 瞬時停電瞬時停電の無いようご注意ください。

3.2 測定可能流体

注記
本器では液体全般を測定できますが、下記の制限があります。
● 流量
UF シリーズのサイズによって測定できる流量範囲が決まっております。下記の表をご参照ください。
また、フルスケール流量以上の測定（100%を超える測定）は保証しておりません。ご注意ください。

・ 適合検出器・流量範囲(L/min の場合)
形式

流量レンジ(L/min)

接続チューブサイズ

UF-10

0 ~ 201/2’’ (12.7×9.52mm)

UF-15

0 ~ 503/4’’ (19.05×15.88mm)

UF-20

0 ~ 901’’ (25.4×22.22mm)

UF-21

0 ~ 150

1’’ (25.4×22.22mm)● 音速測定可能な流体音速は 1000〜2000m/s です。この範囲外の流体は測定できません。ご注意ください。
● 動粘度
設定可能な動粘度範囲は 0.3〜40mm2/s となっております。この範囲外の流体では正しい測定結果を表示できません。ご注意ください。
● 固形物、気泡
固形物、スラリーおよび気泡等が、多量に含まれるものは測定に適しません。ご注意ください。

IM-F2208-J01

3
4. 設置
4.1 設置
正しい測定を行うため、検出器の設置は下記の条件を考慮して下さい。

4.1.1 設置場所

注記
設置場所は下記の条件を考慮してください。
● 周囲温度が 10℃〜50℃で、直射日光の当たらない場所。
● 周囲湿度 30〜80％RH 以内（結露しないこと）。
● 誘導障害を受ける恐れの無い場所。動力機器の近くなどは避けてください。
● 水滴や、腐食性ガスのない所。
● 水没する恐れのない場所。
● 保守点検が容易にできる場所。

IM-F2208-J01

4
4.1.2 配管上の取付位置

注記
正しい測定を行うために、次の項目について考慮して取付位置の選定および取り付けを行ってください。
● 測定管内が常に流体で満たされていること。
水平、垂直、斜めの配管いずれにも取り付けできますが、気泡がたまりにくい取付姿勢をお奨めします。また、堆積や沈殿しやすい液体の場合は、液抜けのしやすい取付姿勢にしてください。
● 開放配管に取り付ける場合は配管の低い部分に設置してください。
配管の高い部分に設置した場合満液にならない場合があります。
● 検出器には流れ方向が矢印で示してありますので、流れ方向は必ず矢印の向きに合うように取り付けてください。
● 配管内の圧力が、大気圧（正圧）以上になるような位置に取り付けてください。
● 流量調整バルブは検出器の二次側に設置することをお奨めします。
流量調整バルブを検出器の一次側に設置すると減圧により、気泡が発生する場合があります。測定管路内の気泡は超音波信号の減衰要素であり、測定不能となる場合があります。
● ゼロ点の確認や保守点検を容易にするため、バイパス配管を設置してください。
● 検出器の固定では入口、出口側接続配管の応力影響が無いようにしてください。
● チューブ継手の接続については各継手メーカーの施工要領書に従ってください。
● 表示器は 180 度回転できますが、3.5N･m 以上のトルクで回転させると回転止めが破損する恐れがありますのでご注意ください。
● 取り付け姿勢は超音波送受信面が空気にふれないように垂直配管を推奨します。
水平配管の場合は表示器を上面になるように設置してください。

注意

超音波送受信方向

配管内に気体の層が有ったり、超音波の通り道を空気で遮ることがないようにご注意く
ださい。

IM-F2208-J01

5
4.1.3 UFシリーズ外形図

UF series

ULTRASONIC FLOWMETER

ERROR

℃
L/min

OUT1
OUT2
TEMP

ENT
DISP

ZERO

形式
UF10
UF15
UF20/21

ΦOD
12.7
19.05
25.4

ΦID
9.52
15.88
22.22

L
230
280
300

ケーブル外形図

IM-F2208-J01

64.2 配線
UF シリーズは３種類の出力タイプがあり、アナログ出力、警報／周波数出力、通信入出力の組合せが異なります。
ご注文時は形式コードでご選択ください。
・結線図
出力タイプ１出力タイプ２出力タイプ３4.2.1 使用ケーブル
付属の専用ケーブルを使用してください。
専用ケーブル配線色割り当て
ケーブル色
赤
茶
緑
青
橙

タイプ１
電源+
電源 0V
アナログ出力+デジタル出力 1(DO1)+
デジタル出力 2(DO2)+

タイプ２
電源+
電源 0V
アナログ出力+出力 com
RS485+

タイプ３
電源+
電源 0V
出力 com
デジタル出力 1(DO1)+
RS485+

黄

出力 com

RS485—

RS485—

IM-F2208-J01

7
5. 運転
本器は納入に先立ち、ご指定の仕様に基づいてデータ設定・調整がなされております。
取り付けおよび配線が完了した後、本章の手順に従って操作していただければ、流量表示と電流およびパルスの流量信号が得られます。万一、運転開始時に不具合が生じた場合は本章と「7. 機能説明」を参照の上、設定パラメータの確認を行ってください。
また、特にご指定のない機能については標準設定値に設定されておりますので、必要に応じて設定パラメータの変更を行ってください。

5.1 運転準備
取り付けおよび配線が完了しましたら、電源投入前に次の点をいま一度必ず確認してください。

注意1）配線● 電源および出力端子の配線に誤りの無いこと。
● ケーブルが確実に端子に接続されていること。
● 接地が確実に行われていること。
● 電源電圧が仕様と合っていること。

注記2）配管取付● 流れ方向と流れ表示矢印が一致していること。
● 配管との接続が確実にされていること。

注記
3）通水● 検出器測定管路内を満液状態にして、流体を静止させてください。
● この際、バルブにリークがなく、完全に流体が静止していることを確認してください。
● また、気泡の発生や溜まり部がないことも確認してください。
● 気泡が溜まっている場合はしばらくブローして、完全に気泡を除去してください。

5.2 通電1) 仕様の電源電圧を投入してください。
2) 規定の性能を満足させるため、通電後約 15 分間ウォームアップしてください。

IM-F2208-J01

8
5.3 ゼロ点調整
UF 形超音波流量計は初めて計測を行う前に必ずゼロ点調整を実施する必要があります。ゼロ点調整を行うには、フロントパネルの ZERO キーを 3 秒間長押ししてください。詳細は「6.3.ゼロ点調整」で説明します。

注記
ゼロ点調整の前に下記の内容を満たしていることを確認してください。
● 正しく設置されていること● 流量計配管内に気泡がなく満液状態であること● 流体が完全に静止していること

5.4 運転中の表示
UF シリーズは運転中、LED に瞬時流量を表示します。
電源投入後は瞬時流量表示となります。
流量計配管内が空または気泡がある場合は LOSS 表示となります。
配管内が満水になるまで通水してください。

5.5 運転中のエラー
運転中何らかのエラーが発生した場合、LED 表示器にエラーメッセージが表示されます。異常内容を確認後、エラー対応方法を実施してください。
・運転中のエラーメッセージ
No.

エラー表示
(内容)

原因と対応方法

超音波波形が検出できません
1
LOSS（波形無し）

・管路が満液でない可能性があります
・管路が満液状態か確認してください

IM-F2208-J01

9
6. パネル操作
本流量計は納入に先立ち、ご指定の仕様に基づいてデータ設定・調整を行っておりますが、特にご指定のない機能については標準設定値に設定されております。
流量レンジや流量計パラメータを変更する場合はこの章を参照して各種パラメータの調整を行ってください。

6.1 操作・表示箇所

1

2

3

4

5

No.

名

6

称

7

機

①

表示器

7 セグメント LED4 桁②

ERROR ランプ

測定エラー時に点灯。点灯色：赤③

OUT1 ランプ

上限警報出力時に点灯。点灯色：黄④

OUT2 ランプ

下限警報出力時に点灯。点灯色：黄

能

⑤

キー

測定モード：温度表示切替
設定モード：メニューの移行、値の移行⑥

キー

測定モード：3 秒長押しでゼロ点調整
設定モード：メニューの移行、桁の移行⑦

キー

測定モード：3 秒長押しで設定モードへ移行
設定モード：3 秒長押しで設定の確定、１ショットでキャンセル

IM-F2208-J01

10
6.2 モードの切替
UF シリーズは通常時の「測定モード」、設定を確認・変更する「設定モード」、ゼロ点調整を行う「ゼロ点調整モード」の 3 つのモードがあり、キー操作でモードを切り替えられます。

自動終了

長押し
電源投入

測定モード

ゼロ点調整モード

前の設定で測定継続
設定モード
長押し

長押し

更新の確認

1) 測定モード
通常時の動作モードで、あらかじめ変換器に設定されたデータ（条件）により流量を測定し、その結果を表示・出力します。表示および出力内容は設定モードで変更できます。
2) ゼロ点調整モード
キーを 3 秒長押しすることでこのモードに切り替わります。
ゼロ点調整モードでは超音波測定の最適化と流量のゼロ点調整を行います。初めて計測を行う時、設置場所を変えた
時、測定対象を変えた時などはゼロ点調整を行ってください。ゼロ点調整モードは自動終了し、更新の確認画面が表示されます。ゼロ点調整中の出力は通常時と異なります。設備側で出力を監視している場合などは異常検出されないようにした上でゼロ点調整を行ってください。
3) 設定モード
キーを 3 秒長押しすることでこのモードに切り替わります。
設定モードでは変換器の各種パラメータを確認・設定することができます。
設定モード内のどこからでもキーを長押しすることで設定モードは終了し、更新の確認画面が表示されます。

IM-F2208-J01

11
6.3 ゼロ点調整

注記
UF シリーズでは初めて計測を行う前に、必ずゼロ点調整を実施する必要があります。調整の前に下記の内容を満たしていることを確認してください。
● 正しく超音波センサが設置されていること● 超音波センサが満液状態であること● 測定管路の流体が完全に静止していること

6.3.1 ゼロ点調整方法
・ゼロ点調整開始
ゼロ点調整はフロントパネルのキーを 3 秒間長押ししてください。
LED 表示内容が右図のように表示されます。
ドットがリレー点滅し約 16 秒で完了し自動的に測定モードに戻ります。

ゼロ点調整が失敗した場合は右図のようにエラー表示となります。
ゼロ点調整が失敗した場合には、
再度ゼロ点調整を実施してください。
ゼロ点調整が連続で 3 回以上失敗する場合は、センサ管路内が気泡などの影響で不安定な状態であると考えられます。通液や脱泡処理によるセンサ管路内の安定化や変換器を再起動して、再度ゼロ点調整を実施してください。

IM-F2208-J01

12
6.4 パラメータの設定方法
測定モードでを 3 秒間長押しすると設定モードに移行します。

6.4.1 設定モードのパラメータ構成
設定メニューを選択するときはを押してください。

設定メニュー表示でパラメータ設定に移行するには
設定変更は設定の確定はまたはを長押ししてください。

で値の選択および数値設定を行ってください。

を長押ししてください。

測定モードに戻るときはを短押ししてください。

測定モード
長押し

設定モード
長長押し
パラメータの確認・変更

設定項目の選択
決定：長押しキャンセル：短押し

長短押し
測定モード

IM-F2208-J01

13
6.4.2 設定項目の選択
パラメータ設定モードでは、始めに設定メニューの選択画面が表示されますので、設定したいメニュー項目を
で選択し、

キー

キーを長押ししてください。パラメータ入力メニューに変わります。また、設定を決定したい場合はキーを長押ししてください。

キーを押して設定したいメニュー項目を選択
してください。メニュー項目は循環表示するようになっています。
設定したいメニュー項目の選択後、キーを長く押すことでパラメータの変更画面に移行します。
また、キーを長押しすることで更新の確認に移行します。

長押しパラメータ変更画面に移行する

決定：長押しキャンセル：短押し選択できる設定項目は次の 20 項目に分かれています。

IM-F2208-J01

14
6.4.3 パラメータ一覧表
No.

内

容

表

示

説

明

1

動粘度

動粘度補正設定2

Ｋファクタ

補正係数設定3

フルスケール

フルスケール流量設定(アナログ出力周波数出力連動)

4

ダンピング

時定数設定

5

ローカット選択

ローカットオフの有効無効選択

6

ローカット値

ローカットオフ値の設定

7

デジタル出力 1
ファンクション

上下限警報出力、積算出力、エラー時出力、流れ方向、周波数出力の選択

8

デジタル出力 1
接点選択

ノーマルオープン、ノーマルクローズの選択

9

デジタル出力 2
ファンクション

上下限警報出力、積算出力、エラー時出力、流れ方向、周波数出力の選択

10

デジタル出力 2
接点選択

ノーマルオープン、ノーマルクローズ
の選択

11

上限警報値

瞬時流量上限警報値設定(%)

12

下限警報値

瞬時流量下限警報値設定(%)

13

上限警報ヒステリシス

警報動作から復帰させる流量幅設定(%)

14

下限警報ヒステリシス

警報動作から復帰させる流量幅設定(%)

15

移動平均サンプル数

移動平均パラメータ設定
1 サンプル 35ms

16

移動平均シフト量

移動平均パラメータ
35ms×シフト量

17

ボーレート

RS485 通信仕様設定

18

パリティ

RS485 通信仕様設定

19

スレーブアドレス

RS485 通信仕様設定

20

流れの方向

流れ方向の設定

IM-F2208-J01

15
6.4.4 パラメータの確認・変更
設定項目が表示されている状態でで選択または、キーを長押しすると数値入力またはパラメータ選択表示になります。

キー

キーで桁シフトして設定してください。

キーを長押しするとパラメータが点滅し設定が確定されます。
キーを短押しするとパラメータは変更されず設定項目表示に戻ります。
1) パラメータを選択する場合例）ＤＯ１ファンクションの設定〔DISITAL OUTPUT 1〕
上限警報から周波数出力に変更する場合

長押し

5 回押し

長押し

キーを長押しして現在の設定値を表示させる。

キーを 5 回押して周波数出力を選択する。

キーを長押しして確定する。
全桁が点滅した後、設定項目表示へ戻る。

IM-F2208-J01

162) パラメータ値を入力する場合例）動粘度の設定〔KINEM VISCOSITY〕
1.00mm2/s を 2.00mm2/s に変更する場合

キーを長押しして現在の設定値を表示させる。

点滅している桁をキーを押して右の桁へ移行する。

キーで２に変更する。

長押しキーを長押しして確定する。
全桁が点滅した後、設定項目表示へ戻る。

IM-F2208-J01

17
6.5 パラメータ詳細
1. 設定パラメータ
パラメータメニュー
LED 表示

選択内容および設定範囲
LED 表示動粘度
KINEMATIC VISCOSITY
00.30〜40.00 (mm2/s)

K ファクタ
K_FACTOR
0.45〜2.20

フルスケール流量
FULL SCALE
1〜9999 (L/min)

時定数
TIME CONSTANT
00.0〜25.0(s)

低流量カットオフの使用可否
LOW CUTOFF
ON, OFF

低流量カットオフの設定
LOW CUTOFF
00.0〜25.0 %FS

IM-F2208-J01

18
パラメータメニュー
LED 表示

選択内容および設定範囲
LED 表示NONE

FLOW ALARM HI

FLOW ALARM LO
デジタル出力１の設定
FUNCTION DO1
TOTAL OUT 1L/P

FLOW DIRECTION

SIGNAL LOSSPULSE OUT 1kHz
デジタル出力１の出力論理
NO/NC
NO、NC

デジタル出力 2 の設定
FUNCTION DO2
デジタル出力１の設定同様

デジタル出力 2 の出力論理
NO/NC
NO、NC

上限警報値
HIGH ALARM
00.0〜150.0 %FS

下限警報値
LOW ALARM
00.0〜99.9 %FS

上限警報ヒステリシス
HIGH ALARM HYSTERESIS
00.0〜30.0 %FS

IM-F2208-J01

19パラメータメニュー
LED 表示

選択内容および設定範囲
LED 表示

下限警報ヒステリシス
LOW ALARM HYSTERESIS
00.0〜30.0 %FS

移動平均サンプル数
MOVING AVERAGE SAMPLES
0〜64

移動平均シフト量
MOVING AVERAGE SHIFT
0〜63

ボーレートの設定
BAUD RATE
19K(19200), 38K(38400), 57K(57600) (bps)

パリティの設定PARITYEven, Odd, None

アドレスの設定
ADDRESS
01〜99

FORWARD

流れの方向の設定
FLOW DIRECTIONFORWARD/REVERSEFORWARD（INVERTED）
FORWARD/REVERSE（INVERTED）

IM-F2208-J01

20
7. 機能説明
7.1 基本項目の設定
1) 動粘度の設定(KINEM VISCOSITY)
測定流体の動粘度を入力します。
本流量計はお客様ご指定の内容に従い動粘度のパラメータを設定し、その設定において調整しております。しかし、実際測定する流体の動粘度が異なる場合、このパラメータを調整することで自動的に出力リニアライズすることができます。

注記
本パラメータを変更した場合、出荷調整時と条件が変わりますので流量保証が出来かねます。ご了承ください。

2) 補正係数の設定(K_FACTOR)
本パラメータはセンサ個体差を調整するための機能で、調整を行った後納品しております。
流量計の測定ラインで精度の高い校正が可能な場合は、補正係数(K ファクタ)を使用することで流量を合わせこむことが可能です。合わせこみはどの点でも可能ですので、非ニュートン流体など偏差の大きい流体の流量を 1 点だけ合わせることも可能です。
補正係数は掛算の係数で、表示、出力流量は次のようになります。
表示・出力流量 = 内部測定流量 × 補正係数(K)・補正係数による表示、出力流量の変化

・補正係数による一点合わせこみの例

IM-F2208-J01

213) 時定数(63%応答)の設定(TIME CONSTANT)
流量変化に対する表示と出力の応答性(追従性)を設定します。
設定した数字が大きいほど、応答時間が長くなります。流量の変化が大きく、表示がふらついて見にくい時は、大きな値に変更してください。また、吐出量の測定など流量の急変化を捉える必要がある場合は小さな値に変更してください。
・時定数変更による流量変動減少例

・ステップ応答の場合の例
時定数増加

流量

時定数 小

流量

時定数 大

時間

時間

4) 低流量カットオフ機能の設定(LOW CUTOFF)
ゼロ点を安定させるため低流量カットオフを設定します。
設定したカットオフ値(%：フルスケールに対する割合)以下の流量は 0 になります。
始めに確認メニューが表示されますので、低流量カットオフを使用するときは“ON”を、使用しない場合は“OFF”
を選択してください。
“ON”を選択した場合、カットオフ値を入力してください。
・低流量カットオフ無し

・流量カットオフ有り

IM-F2208-J01

225) 移動平均の設定(MOVING AVERAGE SAMPLES / SHIFT)
気泡やノイズによる測定流量の振れをとる効果があります。
“ON”を選択した場合、カットオフ値を入力してください。
・ステップ応答の例

・脈流等の乱れがある場合の例
実流量

流量

時間
出力流量
流量

時間
35ms サンプリングデータを移動平均サンプル数で設定したデータの平均を取り出力します。
シフト量 通常は 1 個ずつシフトする移動平均をシフト量で設定したデータ数分の出力でホールドして次の平均を出力します。

IM-F2208-J01236) 温度補正の設定方法
UF シリーズ超音波流量計は水温の影響により測定誤差が生じます。ご使用の水温に応じた設定変更を行う事により、±2%FS 精度にてご使用いただけます。出荷時は 20℃にて校正されています。ご使用になる水温による補正係数を用いて、水温補正後のＫファクタを計算して頂き、本体のキー操作より設定を変更してください。出荷維持に設定されている K ファクタはパラメータリストか本体の設定モードで確認できます。
UF15 を水温 85℃でご使用になる例
出荷時の K ファクタ 1.090、水温 85℃補正係数 1.114 の場合
1.090 × 1.114 ≒ 1.214
温度補正を掛けた K ファクタ 1.214 を設定してください。
UF10 温度補正係数
温度℃

0

1

2

3

4

5

6

7

8

9

20

1.000

1.004

1.007

1.010

1.013

1.017

1.020

1.023

1.026

1.029

30

1.032

1.035

1.038

1.041

1.044

1.047

1.050

1.053

1.056

1.059

40

1.062

1.065

1.068

1.071

1.074

1.077

1.080

1.083

1.086

1.088

50

1.091

1.094

1.097

1.100

1.103

1.106

1.108

1.111

1.114

1.117

60

1.120

1.122

1.125

1.128

1.131

1.133

1.136

1.139

1.141

1.144

70

1.147

1.149

1.152

1.155

1.157

1.160

1.163

1.165

1.168

1.171

80

1.173

1.176

1.178

1.181

1.183

1.186

1.188

1.191

1.193

1.196

90

1.198

-

-

-

-

-

-

-

-

-

UF15 温度補正係数
温度℃
20

0
1.000

1
1.003

2
1.005

3
1.008

4
1.010

5
1.012

6
1.015

7
1.017

8
1.019

9
1.021

30

1.024

1.026

1.028

1.030

1.032

1.034

1.036

1.038

1.040

1.042

40

1.045

1.046

1.048

1.050

1.052

1.054

1.056

1.058

1.060

1.062

50

1.064

1.065

1.067

1.069

1.071

1.072

1.074

1.076

1.077

1.079

60

1.081

1.082

1.084

1.085

1.087

1.088

1.090

1.091

1.093

1.094

70

1.096

1.097

1.098

1.100

1.101

1.102

1.104

1.105

1.106

1.107

80

1.109

1.110

1.111

1.112

1.113

1.114

1.115

1.116

1.117

1.118

90

1.120

-

-

-

-

-

-

-

-

UF20 温度補正係数
温度℃
20

0
1.000

1
1.003

2
1.006

3
1.009

4
1.012

5
1.015

6
1.017

7
1.020

8
1.023

9
1.025

30

1.028

1.030

1.032

1.034

1.037

1.039

1.041

1.043

1.045

1.047

40

1.049

1.050

1.052

1.054

1.056

1.057

1.059

1.060

1.062

1.063

50

1.065

1.066

1.068

1.069

1.071

1.072

1.073

1.074

1.076

1.077

60

1.078

1.079

1.081

1.082

1.083

1.084

1.085

1.087

1.088

1.089

70

1.090

1.091

1.093

1.094

1.095

1.096

1.097

1.099

1.100

1.101

80

1.103

1.104

1.105

1.107

1.108

1.109

1.111

1.112

1.114

1.116

90

1.117

-

-

-

-

-

-

-

-

-

IM-F2208-J01

24
8. 日常点検注記
超音波流量計は可動部や消耗部品がなく、ほとんどメンテナンスフリーでご使用いただけますが、長期に渡って安定してご使用いただくために、以下の日常点検を実施することをお奨めします。

8.1 チューブ継手、接続部の点検● 液漏れ、浸透はないか。
● ナットの緩みはないか。
● テフロンの性質により、一度締め付けても時間が経つと緩むことがありますので定期的に増締めしてください。

8.2 接続配管の点検● 配管に曲がりが生じて検出部に過大な応力が加わっていないか。
● 配管振動は大きくないか。

8.3 防水性の点検● ケーブル接続口は完全に最後まで差し込まれているか。

8.4 検出器の点検● 継手接続部および測定管内に異物、または気泡だまりがないか、目視により確認する。

IM-F2208-J01

25
9. トラブルシューティング
超音波流量計のトラブルは配線や取り付けなど設置に起因するもの、計器自体の故障など、さまざまな原因が考えられます。
原因探索にはトラブルの現象を正確に把握し、それぞれに応じた対応を取ることが近道です。
ここでは、一般的に考えられるトラブル現象別に、トラブルシューティングフローを記載しています。トラブル現象を確認し、対応する項目を参照してください。
トラブルの現象1

表示（液晶表示）が点灯しない

2

表示（液晶表示）が正常でない

3

表示がロックして変化しない

4

エラー表示が出る

5

キー操作を受け付けない、データ設定ができない

6

ゼロ点調整でエラー表示が出る

7

流体を流しても指示がゼロのままである

8

表示は出るが出力がでない

9

ゼロ点が不安定10

ゼロ点で指示が出る、振り切れる

11

流体を流すと指示が不安定

12

実流と指示が合わない

13

実流と出力が合わない

14

流体を流すと指示が振り切れる

参照するトラブルシューティング項目

A：表示が出ないまたは表示が正常でない時

B：ゼロ点調整がうまくいかない時
C：流体を流しても指示が出ない時
D：ゼロ点が不安定な時
E：指示が不安定な時
F：実流と指示が合わない時

IM-F2208-J01

26
A：表示が出ない、または表示が正常でない時
START A

LED 表示は出ているかNo

電源が正しく供給されているか調べる

Yes
エラー表示は
出ているか

Yes
電源接続・電圧は
正しいか

No

電源を
正しく入れるNo
Yes
エラー表示を調べ、内容に従って調査する
基板の故障が考えられる
電源電圧は規定範囲内か

No
サービスへ

Yes
電源を
正しく供給する
電源を一度切り、
10 秒以上待ってから再投入する

正常な表示は
出るか

No

Yes
データ設定
操作ができるかNo

Yes
設定データをチェックする

正常動作するか

END

No

サービスへ

IM-F2208-J01

27
B：ゼロ点調整がうまくいかない時
START B

電源投入

電源をいったん切り
10 秒以上待ってから再投入する

VX.X.Xと表示する

Yes

計測表示となるか
No

No

表示に
Loss
が出るか
Yes

No

検出器に気泡が混入しているか
Yes

流量計内の気泡を除去する

ゼロ点調整を実施

ゼロ点調整後0.0 が表示されるか

No

Yes
END

IM-F2208-J01

28
C：流体を流しても指示が出ない時
START CLED 表示は正常に出るかYes

流量出力は出るか

No

Yes

No

受信計器の結線を外し、出力端子に直接電流計を接続する
ループチェックを行う

・配線断絶
・受信計器異常・入出力仕様のミスマッチ

正しいデータを設定する

Yes

流量出力は出るかNo

No

流れ方向および
電流出力関係の設定データは正しいか
Yes
出力回路故障流れ方向は合っているかNo

サービスへ

流れ方向を正しくする

Yes
設定データは正しいか(流量レンジ等)

No
正しいデータを設定する

Yes

配管内に液は充満しているかNo

液が充満して流れるよう
流量計の設置位置を変更する

No

流れを止め、
正しくゼロ点調整を行う

Yes
ゼロ点は合っているか
Yes
1

IM-F2208-J01

29

1

Yes

コネクタの
接続は確かかコネクタを正しく
接続し直す

No
Yes

異物が付着しているか

異物を取り除く

No

流量計配管内の点検をする

流量計配管内は正常か

Yes

原因が特定できない時はサービスまでご連絡ください

No

電源をいったん切り、10 秒以上待って再投入してください。
その後 START からやり直してください。

IM-F2208-J01

30
D：ゼロ点が不安定な時
START DNo

測定管内は完全に液で充満されているか

完全に充満するよう配管上の取付位置を変更する
下→上の流れが望ましい。
上→下または水平配管の場合は要注意

Yes
Yes

バルブに
漏れがあるかバルブの交換等を行い、漏れを止めるNo
ポンプ等による
圧力変動や振動はあるか
No

Yes

センサの下流側だけでなく上流側のバルブも閉め、液を完全に制止させる（特に小口径形の場合）

Yes

フラッシング等により気泡を抜く
脱泡装置を取り付ける

Yes

誘導ノイズの影響がないか調べる
配線のシールド、隔離を行う

No
液に気泡が
含まれているかNo
ポンプ・モータ・変圧器などの動力機器があるか
No
コネクタの
1
接続は確かかYes

コネクタを正しく接続し直す

No異物が付着しているか

Yes

異物を取り除く

No
検出部の点検をする

検出部は正常かYes

原因が特定できない時はサービスまでご連絡ください

No

電源をいったん切り、10 秒以上待って再投入してください。
その後 START からやり直してください。

IM-F2208-J01

31
E：指示が不安定な時
START E

ゼロ点の指示は安定しているかNo

ゼロ点でも、流体を流しても指示が不安定な時は、最初に「D:ゼロ点が不安定な時」を参照して原因を調べる

No

完全に充満するよう配管上の取付位置を変更する
下→上の流れが望ましい。
上→下または水平配管の場合は要注意

Yes

脈動をなくす脈動の影響のない場所に移設する応答性を要求しなければ、時定数を大きくする

Yes

測定管内を液が完全に充満して流れているか
Yes

流れに脈動があるか
No

Yes

液に気泡が
含まれているか

上流側に脱泡装置を取り付けるNo

スラリーが多量に混入しているかYes

時定数を大きく設定する(改善されない場合はご相談ください)Yes

キャビテーションにより指示が不安定になる場合がある
バルブを全開で使用するか、バルブの下流側に移設する

Yes

粘度、密度が不均一になると指示が不安定となるので上流側での異種液の注入は避け、下流側で行う混合の場合は充分攪拌する

No

上流側に全開できない
バルブが設置されているかNo上流側で液の混合や異種液の注入を行っているかNo

1

C：流体を流しても指示が出ない時を参照

IM-F2208-J01

32
F：実流と指示が合わない時
START FYes

LED 表示は正しいか

アナログ出力は
正しいか

No

Yes

パルス出力は正しいか
No

No

流れを止めゼロ点のチェックをする

流量レンジの設定は正しいかNo「C：流体を流しても指示が出ない時」を参照して出力関係のチェックを行う

Yes
正しい流量レンジに設定する

No

ゼロ点は
合っているか

流体を満液状態で完全に静止させ、ゼロ点調整を行う

Yes
配管径、流量レンジ、K ファクタなどの設定データのチェックを行う

No

設定データは正しいか

正しいデータを設定する

Yes
接地は完全かNo

アース端子から第 3 種接地(接地抵抗 100 以下)を確実に行う

No

完全に充満するよう配管上の取付位置を変更する
下→上の流れが望ましい。
上→下または水平配管の場合は要注意

Yes

混入した気泡の体積分、指示は大きめとなる
上流側に脱泡装置を取り付ける

Yes
測定管内を液が完全に充満して流れているか
Yes
液に気泡が
含まれているか
No
No

測定流体が
仕様と合っているか

流体仕様を確認する

Yes
流れに脈動があるか

Yes

脈動周期が短いと指示が低めになる場合がある
脈動をなくす脈動の影響のない場所に移設する

No
1

C:流体を流しても指示が出ない時を参照

IM-F2208-J01

■ サービスネット
製品の不具合などの際は弊社営業担当か、弊社営業所までご連絡ください。
営業所については弊社ホームページをご覧ください。

■ 製品保証弊社ホームページをご覧ください。

All right Reserved Copyright © 2023 TOKYO KEISO CO., LTD.
本書からの無断の複製はかたくお断りします。

https://www.tokyokeiso.co.jp