/1,6/1,6)1VHULHV

Oblique Angled Lights for Line Sensor

LNIS/LNIS-FN Series/1,6VHULHV/1,6)1VHULHVStreak Inspection
Best for Finding
Moving-direction Scratches

Applications

Visual inspection 6WUHDNLQVSHFWLRQRQVKHHWV
 6FUDWFKLQVSHFWLRQRQWUDQVSDUHQWILOP 6FUDWFKLQVSHFWLRQRQSODWHJODVV 6FUDWFKLQVSHFWLRQRQVKHHWPHWDO

CCS Inc.

LNIS/LNIS-FN series

Bi-directional angled light withunique light-focusing technology
Best for finding moving-direction scratches
Crossing angled lighting

Conventional lighting

Recommended
irradiation range

Imaging example
Imaging moving-direction scratches on a film
Conventional light-focusing Line Light: LNSP series

LNIS series
Line scan camera

Line scan camera
Line Light (LNIS-300SW)

Line Light (LNSP-300SW)

Moving direction

Moving direction

Imaging condition:
100% intensity

Only a scratch is selectively highlighted. Brightness andnoises of the background do not increase even in the highintensity.
1ScratchMoving direction

Moving direction

ScratchImaging condition:
50% intensity

Imaging condition:
100% intensity

It is hard to highlight only a scratch with ordinary lighting.
Brightness and noises of the background increase in the highintensity, so that a contrast ratio is not enhanced.

Finding moving-direction
scratches
Designed for detecting moving-direction scratches
Under a brand-new concept, the main purpose of the LNIS series is to find moving-direction scratches,
which are difficult to find using conventional Line Lights.
For higher intensity, CCS provides the LNIS-FN series to meet more applications.
1,000,000
900,000

LNIS series

Illuminance (lx)800,000
700,000

LNIS-FN series

600,000

LNSP series

500,000
400,000
300,000
200,000
100,000
00204060

80100120140

Illuminating distance (mm)Characteristics

LNIS-FN series(with fans)
Irradiation
uniformity

Brightness

180200* Actual measurement values at the center of the emitting surface,
100% intensity (This does not guarantee product quality.)

LNIS series(fan-less model)

Illuminance

160

The LNIS series are:1) Fan-less (natural air-cooling)2) Space-saving3) 1,000 mm max. in length (standard products)4) Driven by the constant-voltage system
If you need higher intensity, use the LNIS-FN
series which are equipped with cooling fans.

Space saving

Lengthvariation

The LNIS-FN series support more than 1,000 mm
length with standard product.
Uniformity of the emitting surface is good dueto the constant-current driven system.
SeriesQuietness

Illuminance

LNIS-FN series

678,000 lx

LWD = 50 mm

LNIS series

310,000 lx

LWD = 50 mm

Length variation
LNIS
Min.
Fan-less model

Min. 100 mm

Max.
Max. 1,000 mm Specify the emitting surface length.

LNIS-FN

COMING SOONMin.

With fans

Select the emitting surface lengthfrom 100 to 1,500 mm with a 100mm pitch.
(1,000 mm max. for the LNIS series)
Please contact your CCS sales
representative for details.

Min. 100 mm

COMING SOONMax.
Max.

1,500 mm You can specify the emitting surface length more than 1,000 mm.

* You may find a joint portion of the optical component on the emitting surface whose length is 1,300 mm and more.
However, there is no problem for use in dark-field imaging.

2Conceptual image of the irradiation

Example: LNIS-500SW

Direction of measurement
Fan-less(Natural air-cooling)

Long side500 mm

Long side

Short sideWhite

Graph of effective illumination range
Illuminating distance: 30 mm
Recommended
irradiation range
400 mmRight90

LWD:30 mm

LWD:50 mm

350 mmLong side

100
Relative illuminance (%)(Refer to the table on p.4)

* The values are based on the simulation.
This does not guarantee product quality.

80Left

70
60
50
40
30
20
100-350

-300

-250

-200

-150

-100

-50

050100

150

200

250

300

350

Lateral position (mm)

LWD:
70 mm

300 mmIlluminating distance: 50 mm* The values are based on the simulation.
This does not guarantee product quality.
Long side

LWD:
100 mm

240 mm

Right

Left

Relative illuminance (%)

1009080

Right

Left

70
60504030
20100-350-300-250-200-150-100-50050

100150200250300

350

Lateral position (mm)Illuminating distance: 70 mm

Relative illuminance (%)

10090* The values are based on the simulation.
This does not guarantee product quality.
Long side

Right

80

Left

7060
50
4030
20
100-350-300-250-200-150-100-50050100150200

250300350

Lateral position (mm)Illuminating distance: 100 mm* The values are based on the simulation.
This does not guarantee product quality.
Long side

Relative illuminance (%)

1009080Right70Left

6050403020
100-350-300-250-200-150-100-50050

100

150

200

250300350

Lateral position (mm)The section on the graph where " Left " and " Right " overlap is the section where light fromthe left and right sides overlaps. The recommended illumination range is the range in thisoverlapping section where each illuminance is ensured for 80% or higher of the peak.

* LWD is the distance from the Line Light to the workpiece.

3* These graphs are for reference only and do not guarantee product quality.

Graph of the change in illuminanceLight Unit used: LNIS-500SW, LNIS-400SW-FN* Actual measurement values at the center of the emitting surface, 100% intensity(This does not guarantee product quality.)

1,000,000
900,000

Illuminance (lx)800,000
700,000

LNIS series

600,000

LNIS-FN series

500,000
400,000
300,000
200,000
100,000
0020

40

60

80

100

120

140

160

180

200

Irradiation distance (mm)Characteristics of the irradiation distribution

Light Unit used: LNIS-400SW* This graph is for reference only and does not guarantee product quality.

Measuring direction: long side

Measuring direction: short side
Irradiation distribution:short side

40

2030

10-10
1-20-30-40-50

50
60

0.5

70Angle (deg)
-60
-70

80
-1-80
0-0.5

00.5

1Relative irradiation strength

Graph of the correlation between intensity and light output
LNIS series

LNIS-FN series

Light output characteristics of each intensity range
when using Analog Control Unit (constant voltage) PSB3-30024

100100909060Intensity rangeIrradiation amount (%)

80
Irradiation amount (%)

70

Perfect for
the LNIS series

80

18 to 24 V

50
40

Intensity range

30

15 to 24 V

Intensity range

20

12 to 24 V

100

Light output characteristics
when using Analog Control Unit (constant current) PSCC-60048

75

95

115

135155
175
195
Intensity scales (Steps)

215

235

70
60
50
40
30
20
100

255

051102153
Intensity scales (Steps)

204

255

Light Unit used: LNIS-1500SW-FN* Actual measurement values when using Analog Control Unit PSCC-60048(This does not guarantee product quality).

Light Unit used: LNIS-500SW* Actual measurement values when using Analog Control Unit PSB3-30024(This does not guarantee product quality).

Table of the recommended irradiation range (Where illuminance of the left/right beam is 80% of the peak value or more.)(mm)

Emitting surface
length
LWD

100

200

300

400

500

600

700

800

900

1,000

1,100

1,200

1,300

1,400

1,500

10

40

140

240

340

440

540

640

740

840

940

1,040

1,140

1,240

1,340

1,440

30

100

200

300

400

500

600

700

800

900

1,000

1,100

1,200

1,300

1,400

50

50

150

250

350

450

550

650

750

850

950

1,050

1,150

1,2501,350

70

100

200

300

400

500

600

700

800

900

1,000

1,100

1,200

1,300

100

40140

240

340

440

540

640

740

840

940

1,040

1,140

1,240

LNIS/LNIS-FN series

LNIS-FN series* The values are based on the simulation. Actual range of the effective irradiation depends on your imaging environment.
* LWD is the distance from the Line Light to the workpiece.

4

LNIS series
Fan-less (Natural air-cooling)Specifications
WhiteLED color

Case material

Aluminum alloy, Resin

Cable length

300 mmConnector

Metal connector SRCN1A16-7P (JAE)

Operating environment

0 to 40°C, Humidity: 20 to 85%RH (with no condensation)

Storage environment-20 to 60°C, Humidity: 20 to 85%RH (with no condensation)

A: Emittingsurface(mm)

B: Totallength(mm)Powerconsumption(W)

Weight(g)(max.)

LNIS-100SW

100

126

21430

LNIS-200SW

200

226

41760

LNIS-300SW

300

326

61

1,090

LNIS-400SW

400

426

81

1,420

LNIS-500SW

500

526

101

1,740

LNIS-600SW

600

626

121

2,070

LNIS-700SW

700

726

142

2,400

LNIS-800SW

800

826

162

2,730

LNIS-900SW

900

926

182

3,050

LNIS-1000SW

1,000

1,026

202

3,380

Safety standard: EN 62471 compliant

CE marking

RoHS compliant

Cooling method

Natural air-cooling
Relative irradiation strength (%)

Environmental regulations

Light spectrum

Model

5,800 K (typ.)Correlated color temperature

10080
60
40
200

380

430

480

530

580

630

680

730

780

Wavelength (nm)Dimensions (mm)7.5

B: Total length (not including the cable)
A: Emitting surface

1.5(18)

4.2

5.4

88

: Emitting surface length

42SW

21 (Emitting surface)

LNIS-322425

70

42(Ø6.9)

555

10.5

Detail Diagram C300

Mounting slot

4, M5 nut slots
See “Detail Diagram C”.

Options
Light Unit cables

These cables are used to connect the Light Unit and the Control Unit. You can choose from 2 m, 3 m, 5 m, 10 m, and 20 m.
FCB-3-1.25SQ-ME7

FCB-5-1.25SQ-ME7

FCB-10-1.25SQ-ME7

2m

3m

5m

10 m

20 m

Weight430 g

580 g

1,000 g2,000 g5,000 gCable permitted bending radius: 63.6 mm

20,000
Ø10.6

FCB-20-2.0SQ-ME7

Ø11.7

n X 1,000

Ø10.6

FCB-n-1.25SQ-ME7(n = 2, 3, 5, 10)Ø10.6

Dimensions (mm)5FCB-20-2.0SQ-ME7

FCB-2-1.25SQ-ME7

Cable length

Model

Cable permitted bending radius: 63.6 mm

LNIS - FN seriesWith fans

Specifications
WhiteLED color

5,800 K (typ.)Correlated color temperature

Model

A: Emittingsurface(mm)

B: Totallength
(mm)

Power
consumption(W)(including fans)

Weight(g max.)

Numberof coolingfanCase material

Aluminum alloy, Steel sheet, Resin

Cable length

300 mm

LNIS-100SW-FN

100

144

41900

1Connector

Metal connector (PRC04-12A26S-37M18)

LNIS-200SW-FN

200

244

81

1,400

2

Operating environment

0 to 40°C, Humidity: 20 to 85%RH (with no condensation)

LNIS-300SW-FN

300

344

117

1,900

3

Storage environment-20 to 60°C, Humidity: 20 to 85%RH (with no condensation)

LNIS-400SW-FN

400444157

2,400

4

Safety standard: EN 62471 compliant EMC standard: Conforms to EN61000-6-2, EN61000-6-4

LNIS-500SW-FN

500

544

192

2,900

5

Environmental regulations

RoHS compliant

LNIS-600SW-FN

600

644

233

3,400

6Cooling method

Forced air-cooling

LNIS-700SW-FN

700

744

268

3,900

7Frame nuts (four for emitting surface length up to 1,000 mm,
seven for emitting surface length over 1,100 mm),
FG line (2 m) x1, M3 Mounting screw x1

LNIS-800SW-FN

800844309

4,4008

LNIS-900SW-FN

900944345

4,900

9

LNIS-1000SW-FN

1,000

1,044

384

5,500

10

LNIS-1100SW-FN

1,100

1,144

425

6,000

11

LNIS-1200SW-FN

1,200

1,244

460

6,500

12

LNIS-1300SW-FN

1,300

1,344

501

7,000

13

LNIS-1400SW-FN

1,400

1,444

536

7,500

14

LNIS-1500SW-FN

1,500

1,544

576

8,000

15

CE marking

Relative irradiation strength (%)

Accessories

Light spectrum

1009080706050403020100300

380460540620700780860

Wavelength (nm)Dimensions (mm)4

B: Total length (not including the cable)
34

A: Emitting surface

Detail Diagram C

1962(6.7)

12: Emitting surface length

1.5

6SW-FN23 (Emitting surface)

LNIS-(101.3)

41

32

16(Ø13.6)

30

0

56.5

14.5

Mounting slot

Fan air inlets

4, nut slots

2xM3 (For FG line installation, Same for opposite side)

See “Detail Diagram C”.

Options
Light Unit cables

These cables are used to connect the Light Unit and the Control Unit. Use the cable that is suitable for your installation site.

QCBM-2

QCBM-3

QCBM-5

QCBM-10

QCBM-20

Model

QCB-2

QCB-3

QCB-5

QCB-10

Cable length

2m

3m

5m

10 m

20 m

Cable length

2m

3m

5m

10 m

20 m

Weight800 g

1,000 g1,500 g2,700 g5,000 gWeight1,100 g1,500 g2,400 g4,600 g8,900 gDimensions (mm)Applicable Control Unit: PSCC-30048(n = 2, 3, 5, 10, 20)

QCB-n(Ø12.6)

QCBM-n(n = 2, 3, 5, 10, 20)
n x 1000

Cable permitted bending radius: 75.6 mm

QCB-20n x 1000Ø16.5Model

Applicable Control Unit: PSCC-60048

Cable permitted bending radius: 99 mm6

Refer to our website for product details.

Applicable Analog Control Unit for the LNIS series

You can also use
your smartphoneor cell phone.

SearchCCS PSB3-30024
Use a search engine.

PSB3-30024

Characteristics

Constant-voltage system

Variable voltage control

1 channel

Light output is 1 channel with 4 connectors (Metal connectors x 2, EL connectors x 2).
Equipped for parallel, serial, and analog external control all in a single Unit.
Select the appropriate voltage range for the Light Unit with Light intensity
range selector to set the optimum intensity.

[300 W capacity]

Dimensions (mm)Setting switch(Pushbutton dial)

MODE

110
ON/OFF inputconnector

L1-4

Fan air inlets

AC inlet

7Four, M3 bottom mounting screw holes(Insertion depth of 5 mm max.)ID
ANALOG/SERIAL

POWER

EIA-485 communications/analog input connector

Side view

Power switch
Bottom view

Rear view

Refer to our website for product details.

Applicable Analog Control Unit for the LNIS-FN series
Constant-current system

Fan exhaust outlet

L1-3
L1-2

ID switch

ON/OFFFront view

Characteristics

L1-1(EL connectors)

Fan exhaust outlets

MANUAL
EXTERNAL
RANGE
HIGH
MIDDLE
LOW

196

17.5

(Metal connectors)

Output connectors

Light intensity
range selector

PSB3-30024

INTENSITY
PARALLEL

7

Output connectors

245

16

Manual/external
mode selector

9765

Digital display
Parallel communications
connector

Use a search engine.

1 channel

Variable current control

You can also use
your smartphoneor cell phone.

SearchCCS PSCC

Intensity control is performed by varying the current.
Equipped for parallel, EIA-485, and Ethernet communications external control all in a single Unit.
Error detection function notifies insufficient speed or stoppage of the cooling fans in the Light Unit, andalso notifies LED burnout errors due to an open or shorted LED circuit.

PSCC-30048 [300 W capacity]
Parallel communications
connector

Dimensions (mm)Bottom mounting screw holes 4-M3

Output connector(19-pin metal connector)

Ethernet connector
270(Insertion depth 5 mm)
15.8

Fan exhaust outlet

238.4

15.8

Air inlets

65

110

1697

Air inlets

EIA-485 communications
connector

ON/OFF input connector

Air inlets

Power switch

AC inlet

PSCC-60048 [600 W capacity]
Manual/external
mode selector Ethernet connector

Setting switch

Air inlets

Fan exhaust
outletDigital display
225

Dimensions (mm)Output connector(37-pin metal connector)

371
Bottom mounting screw holes 4-M4

P S C C - 60048(Insertion depth 5 mm)

MODE
MANUAL
EXTERNAL

RESET

192

131

INTENSITY

PARALLEL

ON/OFFSERIALID

POWER

ETHERNETKEYLOCKLOCKFG-+Key lock switch
ID switch

Parallel communications
connector

EIA-485 communications
connector

Power switch

16.5

UNLOCK

ON/OFF InputConnector

18

AC inlet335● “CCS”, “LIGHTING SOLUTION”, “LNIS”, “LNIS-FN”, “PSB”, and “PSCC” are registered trademarks or trademarks of CCS Inc.

CAUTION● To ensure proper and safe use of the product, please read the Instruction Guide completely before using the product. ● The design and specifications of this product are subject to change without notification for product improvement.
● The workpiece imaging examples included in this pamphlet are intended to serve only as references to help you select a suitable Light Unit. Please verify the functionality and conditions required for your particularapplication before you make a final selection. The sample workpieces used in this pamphlet have been processed specifically for sample imaging. They are not intended to represent product quality and performance.

CCS Inc.
Headquarters

CCS America, Inc

CCS Europe NV/SACCS Asia PTE LTD

CCS Inc. Shanghai Office

CCS Inc. Shenzhen office

Shimodachiuri-agaru, karasuma-dori, kamigyo-ku,
Kyoto 602-8011 JAPANTEL : +81-75-415-8284 / FAX : +81-75-415-8278
URL : http://www.ccs-grp.com/
E-mail : sales@ccs-inc.co.jp
63 Hillview Avenue #07-10, Lam Soon Industrial
Building, Singapore 669569
TEL : +65-6769-1669 / FAX : +65-6769-3422
URL : http://www.ccs-asia.com.sg/
Email : sales@ccs-asia.com.sg

5 Burlington Woods Suite 204 Burlington, MA 01803 USA
TEL : +1-781-272-6900 / FAX : +1-781-272-6902
URL : http://www.ccsamerica.com/
Email : info@ccsamerica.com

Room 308B-309, CIMIC Tower No.1090 Century Avenue,
Pu Dong New Area, Shanghai 200120, P.R. China
TEL : +86-21-5835-8728 / FAX : +86-21-5835-8928
Email : ccschina@ccs-inc.co.jp

Bergensesteenweg 423, Bus 13
1600 Sint-Pieters-Leeuw, Belgium
TEL : +32-(0)2-333-0080 / FAX : +32-(0)2-333-0081
Email : info@ccseu.com

17B,China Economic Trade Building, 7Rd Zizhu, Zhuzilin,
Futian District, Shenzhen 518040 P.R.China
TEL : +86-755-8279-0477 / FAX : +86-755-8279-0478
Email : ccschina@ccs-inc.co.jp
Copyright © 2015 CCS Inc. All Rights Reserved.
Content current as of Junuary 2015.

02002-02-1407-LNIS