MicroW XL

MicroW XL is a temperature data logger from -40°C to250°C (standard calibration from 25°C to 140°C, calibrablefrom -40°C to 250°C, only the probe will resist above
140°C. For calibrations above 140°C probe minimum lengthis 50 mm) with 20, 50, 100, 150 mm or on demand length
external probe (probes cannot be switched) on a 3 mmbase (base height is not counted for probe length),managed with Windows software and USB interface(DiskInterface HS, Multibay). Battery is user
replaceable and the data logger is provided with a
National Reference traceable cetiﬁcate on 6 points.
Note: if used below -30°C, battery life will be highly
reduced.

Main features
With diﬀerent lengths rigid probe for penetration
Completely food grade and waterproof
All software calculate lethality value (F0, PU, A0 ecc.)
Low battery consumption for an extended battery life
User replaceable battery (software shows battery status)
Very easy to deploy in any type of package
Reference traceable calibration certiﬁcate included
Available extended calibration from -40°C to 250°C (order extra calibration points; in case of widecalibration range the accuracy might be worse. Request the possible calibration ranges in advance)
For calibration over 140°C the minimum probe length is 50 mm
Heat protections available for use above 140°C

Applications

Validation

Sterilisation

Pharmaceutical

Healthcare

MedicalCosmetics

Laboratories

Plus
Extremely high accuracy and precision: with an accuracy of ± 0,1°C these devices can be employed in any
application involving pharmaceuticals, validation, laboratory and medical ﬁeldHigh accuracy even outside the calibration range
Fast response time thanks to the 3 mm diameter probe
Printed reports compliant with health regulations and ISO (data are not editable in the software)

The system
The system is made up by:
MicroW XL temperature data logger
DiskInterface HS or Universal Multibay
SPD software or TS Manager software (compatible with the FDA 21 CFR Part 11 regulation)

Food & Beverages

Pasteurisation

Accessories
SPDTS Manager
Tecno Calib
DiskInterface HS
Universal multibay
Locking bolt
Fastening system
S-Micro XL / MicroW XL heat protection 1
S-Micro XL / MicroW XL heat protection 2
MicroW XL battery kit

Technical speciﬁcations
Dimensions64 h X 20 Ø (mm)Probe dimensions

Probe base dimensions 3 h X 14 Ø (mm) - Probe 20/50/100/150/on demand l X 3 Ø (mm) (lon demand: min. 12 mm / max. 175 mm. For longer probes ask for quotation)

Weight61 gMaterials

Stainless steel AISI316L, PEEK

Temperature range-40°C ÷ +140°C

Sensor temperature range-40 °C ÷ +250 °C

Standard calibration points (temperature)

25/50/75/100/121/140°C

Extra calibration points (temperature)

Within the range -40 °C ÷ +250 °C

Temperature resolution

0,01 °C

Temperature accuracy± 0,1 °C (within the calibration range) / ± 0,05 °C with ultra accuracy option

Memory (n. of acquisitions)20.224

Acquisition step

From 1 every second up, with 1 second steps

Protection degree

IP68

Battery life

+17.000.000 acquisitions at 1 second step continuously (calculated time @25°C. Battery life
may be shorter if used in low temperatures)

Software&Mobile App

SPD, TS Manager

Accessories

DiskInterface HS, Multibay universale