PAGE 1 OF 2 – DS1091 R2

DATA SHEET DS1091

CONVEYOR LABORATORY UNIT
GEW’s Conveyor Laboratory Unit can be specified with any combination of GEW’s production UV
curing systems (arc, LED or Excimer) and customisation options such as nitrogen inerting, heightadjustment, and quartz window options. A high precision motor allows samples and measurement
devices to pass under the UV unit(s) at variable speed up to 100m/min. The height adjustment function
allows the unit to accurately mimic almost any press configuration.
Multiple lamps can be arranged in series and with RHINO ArcLED control; LED and arc lamp curing canthen be rapidly interchanged as required. Appropriate cooling systems are specified according tocustomer requirements and the floor standing units can be set on wheels or permanently located.
Whatever your laboratory requirements, GEW can custom design a solution to accurately reflect theproduction environment you serve.

The control panel enables individual switchingfor each lamphead, as standard.

GEW (EC) Limited Crompton Way, Crawley, West Sussex RH10 9QR
T: +44 (0) 1737 824 500
F: +44 (0) 1737 823 822E: sales@gewuv.com
W: gewuv.com

Bed height is manually adjustable inincrements of 0.01mm.

DATA SHEET DS1091

PAGE 2 OF 2 – DS1091 R2Key features:• Fully customisable Conveyor Laboratory Unit
• Can be equipped with any combination of GEW UV, Excimer or LED lamps• Long infeed to ensure samples reach belt speed prior to first lamp
• All safety interlocks, emergency stops• Perfect for R&D or light production• Option for nitrogen inertion, height adjustment and almost any other option GEW manufacture• Manual height adjustment with customisable lamphead height up to 150mm (inerting significantly• reduces height adjustment).
Compatibility
Arc/array lengthInterface plate
Speed
Inert nitrogen atmosphere
Cooling
Power supply unit

WaterAir
Nitrogen
TypeSupply

Additional services
Overall dimensions LxWxH
Working area required LxWxH

E2C, E4C, NUVA2, AeroLED, LeoLED and ExciRay.
38cm as standard.*
125, 250 and 500mm (specification dependent).
Up to 100m/min.
Optional (required for ExciRay). Supplied at 8 bar at 20°C.
Oxygen-free (purity 0.99999). Nitrogen supply required.
All (water-cooled roller & water-cooled bed).
E2C, E4C, NUVA2 and AeroLED.
ExciRay only.
GEW RLT or RHINO.
380 Vac – 480 Vac 50/60 Hz 3P + N + E.
Compressed air supply, minimum 6 bar.
3312 x 1015 x 1429mm. Height includes ExciRay lamphead.
4000 x 2000 x 2000mm, approx.

*Other widths may be available upon request

Further Specifications:
Speed range
Conveyor belt width
Lamphead width
Effective curing width
Maximum substrate thickness

4-100m/min – digital readout, continuously variable.
400mm.
380mm.
330mm (ExciRay 300mm).
35mm.
Modular sections of 125mm, 250mm & 500mm; customers
1625mm space for lampheads
can use up to 1625mm in any combination of below:
Nitrogen inlet.
125mm
Blank.
1x lamphead of any type.
250mm
Blank.
2x lampheads of any type/combination.
500mm
Blank.
Up to 35mm: from 5mm to 40mm, measured from belt toOverall bed height adjustment
underside of plate(s). Can be increased to 150mm when using
interchangeable booster plates.
Optional nitrogen inerting
Adjustable down to below 50ppm.
Additional external components (e.g. fan, chiller and cabinet)
Power supplies and coolingalso require space.

GEW (EC) Limited Crompton Way, Crawley, West Sussex RH10 9QR
T: +44 (0) 1737 824 500
F: +44 (0) 1737 823 822
E: sales@gewuv.com
W: gewuv.com