TEST REPORTSTANDARD: RTCA DO-160G:2010 Section 21.5

Applicant
FUJITA ELECTRIC WORKS,LTD
945, Yamanishi,Ninomiya-machi,Naka-gun,Kanagawa 259-0124 Japan
Tel : +81 463 71 2541

Equipment Type
TrademarkModel (s)Serial No.
Test Result
Report Number
Original Issue Date

Testing Laboratory
Intertek Japan K.K. Matsuda Laboratory(Open area test site)1283 Yadoriki, Matsuda-machi, Ashigarakami-gun,
Kanagawa-ken, 258-0001 Japan
Tel.: +81 465 89 2316 Fax.: +81 465 89 2160
(Anechoic chamber)
1386 Yadoriki, Matsuda-machi, Ashigarakami-gun
Kanagawa-ken, 258-0001 Japan
URL: http://www.japan.intertek-etlsemko.com

WATCH LOGGER
FUJITA ELECTRIC WORKS,LTD
KT-295F / KT-295DX
03FE001401008376 / 03FE0014FB000131
Complied
23060135JMA-001
August 4, 2023

This report is for the exclusive use of Intertek's Client and is provided pursuant to the agreement between Intertek and its
Client. Intertek's responsibility and liability are limited to the terms and conditions of the agreement. Intertek assumes noliability to any party, other than to the Client in accordance with the agreement, for any loss, expense or damage
occasioned by the use of this report. Only the Client is authorized to permit copying or distribution of this report and then
only in its entirety. (This test report shall not be reproduced except in full, without written approval of Intertek Japan K. K.).
Any use of the Intertek name or one of its marks for the sale or advertisement of the tested material, product or service
must first be approved in writing by Intertek. The observations and test results in this report are relevant only to thesample tested. This report by itself does not imply that the material, product, or service is or has ever been under anIntertek certification program.

Approved by

Tested by

Hideaki Kosemura[ Reviewer ]Hidetoshi Sasaki[ Engineer ]©2022 Intertek Japan K.K.

Page 1 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

Original :August 4, 2023

1

GENERAL INFORMATION ...................................................................................................................... 3

2

TEST SUMMARY ..................................................................................................................................... 4
2.1
Test standards & Test results .......................................................................................................... 4
2.2
Test application ................................................................................................................................ 4

3

SYSTEM INFORMATION ......................................................................................................................... 5
3.1
System configuration ........................................................................................................................ 5
3.2
Support Equipment specification...................................................................................................... 5
3.3
Cable specification ........................................................................................................................... 5
3.4
Variation of Models........................................................................................................................... 6
3.5
EUT setup Block diagram................................................................................................................. 7

4

OPERATIONAL CONDITION................................................................................................................... 8
4.1
Operating Flow ................................................................................................................................. 8

5

MEASUREMENT UNCERTAINTY ........................................................................................................... 9

6

TEST CONDITION AND TEST RESULTS............................................................................................. 10

7

TEST EQUIPMENTS .............................................................................................................................. 118

PHOTOGRAPHS OF TEST SET-UP ..................................................................................................... 12
ANNEX :
CHART OF TEST RESULTS ................................................................................................ 15
APPENDIS : WATCHLOGGER LINEUP©2022 Intertek Japan K.K.

Page 2 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

1

Original :August 4, 2023

GENERAL INFORMATIONTEST PERFORMED
Location

Matsuda No.8 Test Site

EUT Received

July 12, 2023

Date of Test

July 12, 2023 and July 13, 2023

Qualifications of Testing Laboratory
Accreditation/Recognition
Scope
VLAC
Wireless / EMC Testing
NVLAP
Wireless / EMC Testing
FCC
Wireless / EMC Testing
ISED
Wireless Testing
BSMI
EMC Testing
SABS
EMC Testing
FilingVCCI
EMC Testing
ABBREVIATIONSESAElectronic subassembly
EUT
Device Under Test
AE
Associated Equipment
AN
Artificial Network
LCD
Liquid-Crystal Display©2022 Intertek Japan K.K.

Cal
Q-P
PK
AVG
N/A

Page 3 of 54

Lab. Code
VLAC-008-3
600234-0
JP0009JP0004 (CABID)
SL2-IN-E-6009
N/A

Remarks
JAPAN
USA
USA
CANADATAIWANSouth Africa

A-0127

JAPAN

Calibration
Quasi-peak
Peak
Average
Not applicable or Not available

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

Original :2

TEST SUMMARY

2.1

Test standards & Test results

August 4, 2023

EMC-tests were performed according to measurement methods specified in the following reference standards:

Test requirements:
RTCA DO-160G:2010

Environmental Conditions and Test Procedures for Airborne Equipment

Test methods:

Result

RTCA DO-160G:2010
Section 21.5

Radiated RF Emissions

Remarks

Refer to Annex for the chart of the test result.

2.2

Pass

Test application

The tests were carried out on the following points.
Test method

Limit

Test application
KT-295F

RTCA DO-160G:2010
Section 21.5

Super multi mode
Front, Rear, Left, Right, Top, Bottom

Category H

KT-295DX©2022 Intertek Japan K.K.

Remarks

Super multi mode
Front, Rear, Left, Right, Top, Bottom

Page 4 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

3

Original :August 4, 2023

SYSTEM INFORMATIONThe equipment under test (EUT) consisted of the following apparatus.
The information of this section is provided by the Applicant or customer. Intertek doesn't take any
responsibility for the information.

3.1 System configuration
Symbol

Item

Model

SerialTrade Mark

Note

A1

WATCH LOGGER

KT-295F

03FE001401008376

FUJITA ELECTRIC
WORKS,LTD

Non
Bluetooth
model

A2

WATCH LOGGER

KT-295DX

03FE0014FB000131

FUJITA ELECTRIC
WORKS,LTD

BluetoothmodelPower Ratings of EUT :DC3.0 VPower Supply :DC3.0 V (Battery)

Condition of Equipment :Production

EUT Photo

See Section 8

3.2 Support Equipment specification
No support equipment was provided for the test.

3.3 Cable specification
No cable was provided for the test.

©2022 Intertek Japan K.K.

Page 5 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

Original :August 4, 2023

3.4 Variation of Models
No
1
2
3
4
5
6
7
8
9
10

Difference
KT-155F
Refer to Appendix A Table A.1
KT-155U
Refer to Appendix A Table A.1
KT-175F
Refer to Appendix A Table A.1
KT-115XLF
Refer to Appendix A Table A.1
KT-255F
Refer to Appendix A Table A.1
KT-255U
Refer to Appendix A Table A.1
KT-275F
Refer to Appendix A Table A.1
KT-215XLF
Refer to Appendix A Table A.1
KT-195F
Refer to Appendix A Table A.1
KT-195U
Refer to Appendix A Table A.1
Tested model
11
KT-295F
Refer to Appendix A Table A.1
12
KT-295U
Refer to Appendix A Table A.1
13
KT-215LF
Refer to Appendix A Table A.1
14
KT-155F/EX
Refer to Appendix A Table A.1
15
KT-165F
Refer to Appendix A Table A.2
16
KT-265F
Refer to Appendix A Table A.2
17
KT-195U/GX
Refer to Appendix A Table A.3
18
KT-295U/GX
Refer to Appendix A Table A.3
19
KT-195F/GX
Refer to Appendix A Table A.3
20
KT-295F/GX
Refer to Appendix A Table A.3
21
KT-155FP
Refer to Appendix A Table A.4
22
KT-115LFP
Refer to Appendix A Table A.4
23
KT-155F/EX(LED)
Refer to Appendix A Table A.4
24
KT-255F/32
Refer to Appendix A Table A.4
25
KT-115LFP/A
Refer to Appendix A Table A.4
26
KT-155DX
Refer to Appendix A Table A.5
27
KT-255DX
Refer to Appendix A Table A.5
28
KT-195DX
Refer to Appendix A Table A.5
Tested model29
KT-295DX
Refer to Appendix A Table A.5
30
KT-115XDXRefer to Appendix A Table A.5
31
KT-215XDXRefer to Appendix A Table A.5
32
KT-155DXP
Refer to Appendix A Table A.5
33
KT-115DXP
Refer to Appendix A Table A.5
KT-295F and KT-295DX were tested on behalf of all models by applicant's judgment.

©2022 Intertek Japan K.K.

Model No.

Page 6 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

Original :August 4, 2023

3.5 EUT setup Block diagram
3.5.1 Super multi mode

 : EUT
A1 or A2

The symbols and numbers assigned to the equipments and cables on this diagram correspond to the ones infrom Sections 3.1 to 3.3.

©2022 Intertek Japan K.K.

Page 7 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

4

Original :August 4, 2023

OPERATIONAL CONDITION

The test was carried out under the following mode.
The information of this section is provided by the Applicant or customer. Intertek doesn't take any
responsibility for the information.

4.1 Operating Flow
4.1.1 Super multi mode
Cycle time for operation: 60 sec

EUT measures temperature, humidity and impact.

EUT store data.

EUT display data.

Note: KT-295DX has Bluetooth function. Bluetooth function is OFF during the test.

©2022 Intertek Japan K.K.

Page 8 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

5

Original :August 4, 2023

MEASUREMENT UNCERTAINTY

Radiated RF Emissions
100MHz – 200MHz

4.85 dB(k=2)

200MHz – 1GHz

5.84 dB(k=2)

1GHz – 6GHz

5.58 dB(k=2)

Traceability to national standard in SI units is ensured with these values.
Compliance with the limits in this standard are determined without in consideration of the measurement
uncertainty of the measurement instrumentation.

©2022 Intertek Japan K.K.

Page 9 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

6

Original :August 4, 2023

TEST CONDITION AND TEST RESULTS

See ANNEX for the detailed result.

6.1 Radiated RF Emissions
Location

Matsuda No.8 Test Site

Test Engineer

Hidetoshi Sasaki

Test Specification
Item

SectionCategory

Test Procedure

Radiated RF Emissions

21.5

H

LEN-RJP-EM035

Frequency Range of Measurements
Model

Operating mode

Tested side

Required Frequency
Range

Measured Frequency
Range

KT-295F

Super multi mode

Front, Rear, Left,
Right, Top, Bottom

100 MHz – 6000 MHz

100 MHz – 6000 MHz

KT-295DX

Super multi mode

Front, Rear, Left,
Right, Top, Bottom

100 MHz – 6000 MHz

100 MHz – 6000 MHz

Environmental Condition
Date
Temperature Variation
July 12, 2023

21 oC–24 oCooJuly 13, 2023

22 C–24 C

Test Results
Model

Operating mode

Humidity Variation

Atmospheric Pressure

44 %–56 %97.7 kPa–97.8 kPa

37 %–48 %97.7 kPa–97.9 kPa

Tested side
Front, Rear, Left,
KT-295F
Super multi mode
Right, Top, Bottom
Front, Rear, Left,
KT-295DX
Super multi mode
Right, Top, Bottom
Note: Refer to Annex for the chart of the test result.

©2022 Intertek Japan K.K.

Page 10 of 54

ResultRemarks

Pass

See Note

Pass

See Note

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001
7

Original :August 4, 2023

TEST EQUIPMENTS

Measurement equipment
Equipment

Model number

Serial number

Manufacturer

Calibration
date

Calibrationdue date

Radiated RF Emissions
Attenuator

6803.17.B

None(DML412)

HUBER+
SUHNER

Mar.30, 2023

Mar.29, 2024

Biconical
antenna

BBA9106

None(D02083002)

Schwarzbeck

Mar.22, 2023

Mar.21, 2024

Logperiodic
antenna

3146

1672

EMCO

Mar.22, 2023

Mar.21, 2024

Double Ridged
Guide antenna

3115

2567

EMCO

Dec.09, 2022

Dec.08, 2023

Pre-amplifier

ZFL-500/ZX60-3018G

2007040601

Mini-Circuits

Jan.13, 2023

Jan.12, 2024

Pre-amplifier

8449B

3008A00455

HEWLETT
PACKARDFeb.13, 2023

Feb.12, 2024

Spectrum
Analyzer

E7403A

MY42000061

Agilent

Jul. 04, 2023

Jul.03, 2024

©2022 Intertek Japan K.K.

Page 11 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

88.1

Original :August 4, 2023

PHOTOGRAPHS OF TEST SET-UP
Equipment Under Test
KT-295F
Antenna Direction:Top

Z axis (Front)

Front
Right
X axis (Right)

Antenna
AntennaDirection:
Direction:
Left
Rear

Rear
Y axis (Top)

Left
Bottom©2022 Intertek Japan K.K.

Page 12 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

Original :August 4, 2023

KT-295DX
Antenna Direction:Top

Z axis (Front)

Front
Right
X axis (Right)

Antenna
AntennaDirection:
Direction:
Left
Rear

Rear
Y axis (Top)

Left
Bottom©2022 Intertek Japan K.K.

Page 13 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

8.2

Original :August 4, 2023

Test Setup

8.2.1 Section 21.5 - Radiated RF Emission
8.2.1.1 Super multi mode (KT-295F and KT-295DX)

100 MHz-200 MHz (Horizontal)

100 MHz-200 MHz (Vertical)

200 MHz-1000 MHz (Horizontal)

200 MHz-1000 MHz (Vertical)

1000 MHz-6000 MHz (Horizontal)

1000 MHz-6000 MHz (Vertical)©2022 Intertek Japan K.K.

Page 14 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Report No. 23060135JMA-001

ANNEX :

©2022 Intertek Japan K.K.

Original :August 4, 2023

CHART OF TEST RESULTS

Page 15 of 54

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 16 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H: RTCA/DO-160G Sec21.5 Cat.H -6dB

190M

: RTCA/DO-160G Sec21.5 Cat.H: RTCA/DO-160G Sec21.5 Cat.H -6dB

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: AmbientPower source : File number : Engineer: Hidetoshi Sasaki
Note:

Section 21.5
Ambient Noise (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : Serial number : -Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 17 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H: RTCA/DO-160G Sec21.5 Cat.H -6dB

800M

: RTCA/DO-160G Sec21.5 Cat.H: RTCA/DO-160G Sec21.5 Cat.H -6dB

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: AmbientPower source : File number : Engineer: Hidetoshi Sasaki
Note:

Section 21.5
Ambient Noise (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : Serial number : -Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 18 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H: RTCA/DO-160G Sec21.5 Cat.H -6dB

5G

: RTCA/DO-160G Sec21.5 Cat.H: RTCA/DO-160G Sec21.5 Cat.H -6dB

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: AmbientPower source : File number : Engineer: Hidetoshi Sasaki
Note:

Section 21.5
Ambient Noise (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : Serial number : -Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 19 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Front
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Front (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 20 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Front
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Front (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 21 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Front
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Front (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 22 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Rear
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Rear (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 23 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Rear
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Rear (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 24 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Rear
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Rear (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 25 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Left
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Left (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 26 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Left
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Left (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 27 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Left
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Left (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 28 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Right
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Right (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 29 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Right
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Right (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 30 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Right
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Right (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 31 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Top
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Top (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 32 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Top
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Top (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 33 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Top
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Top (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 34 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Bottom
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Bottom (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 35 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Bottom
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Bottom (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 36 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Bottom
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295F: Bottom (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 13 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295F
Serial number : 03FE001401008376

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 37 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Front
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Front (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 38 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Front
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Front (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 39 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Front
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Front (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 40 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Rear
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Rear (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 41 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Rear
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Rear (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 42 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Rear
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Rear (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 43 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Left
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Left (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 44 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Left
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Left (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 45 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Left
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Left (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 46 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Right
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Right (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 47 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Right
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Right (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 48 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Right
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Right (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 49 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Top
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Top (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 50 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Top
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Top (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 51 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Top
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Top (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 52 of 54

110M

110M

120M

120M

130M

130M

140M

140M

150M
Frequency [Hz]150M
Frequency [Hz]

160M

160M

170M

170M

180M

180M

200M

190M

200M

: RTCA/DO-160G Sec21.5 Cat.H

190M: RTCA/DO-160G Sec21.5 Cat.H

-10
100M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
100M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Bottom
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Bottom (100MHz-200MHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 53 of 54
300M

300M

400M

400M

Frequency [Hz]Frequency [Hz]

500M

500M

600M

600M

700M

700M

900M

1G

800M

900M

1G

: RTCA/DO-160G Sec21.5 Cat.H

800M: RTCA/DO-160G Sec21.5 Cat.H

-10
200M

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
200M

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Bottom
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Bottom (200MHz-1GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

Page 54 of 54
2G

2G

Frequency [Hz]Frequency [Hz]

3G

3G

4G

4G

6G

5G

6G

: RTCA/DO-160G Sec21.5 Cat.H

5G: RTCA/DO-160G Sec21.5 Cat.H

-10
1G

0

10

20

40

50

-1/1-

Copyright(c)2007 Intertek Japan K.K.

30

: Data(Vert): Data(Hori)

Distance 1.00 m

August 4, 2023

60

70

80 Vert

-10
1G

0

10

20

30

40

50

60

70

80 Hori

Test mode: Super multi mode
Power source : Battery (DC3 V)
File number : Bottom
Engineer: Hidetoshi Sasaki
Note:

Section 21.5
KT-295DX: Bottom (1GHz-6GHz)

Original :emiT 3, 0, 0, 0

Emission Level [dBuV/m]©2022 Intertek Japan K.K.

Emission Level [dBuV/m]

Date tested : Jul 12 2023
Company: FUJITA ELECTRIC WORKS,LTD
EUT Name: WATCH LOGGER
Model number : KT-295DX
Serial number : 03FE0014FB000131

Matsuda No.8 Test Site

RTCA DO-160G Section21.5< Graph number # >Report No. 23060135JMA-001

LFT-FJP-EM011 / Effective Date: 31 May 2022

APPENDIX A: WATCHLOGGER LINEUP

Table A.1䝉䞁䝃䞊
Sensorᗘ䞉‵ᗘ/ Humidity
TemperatureᗘTemperature

Temperatureᗘ⾪ᧁImpactTemperature/ Humidityᗘ䞉‵ᗘTemperatureᗘ䞉‵ᗘ
Humidity䝕䝄䜲䞁
External
Model
ᆺ␒No.

Tempeᗘ
ratureImpact⾪ᧁ㻷㼀㻙㻝㻡㻡㼁㼁㻿㻮䕿䠉䠉㻷㼀㻙㻝㻣㻡㻲
NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䠉䠉㻷㼀㻙㻝㻝㻡㼄㻸㻲NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䠉䠉㻷㼀㻙㻞㻡㻡㻲
NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䕿䠉㻷㼀㻙㻞㻡㻡㼁㼁㻿㻮䕿䕿䠉㻷㼀㻙㻞㻣㻡㻲
NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䕿䠉㻷㼀㻙㻞㻝㻡䠴㻸㻲NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䕿䠉㻷㼀㻙㻝㻥㻡㻲
NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䠉䕿㻷㼀㻙㻝㻥㻡㼁㼁㻿㻮䕿䠉䕿㻷㼀㻙㻞㻥㻡㻲
NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䕿䕿㻷㼀㻙㻞㻥㻡㼁㼁㻿㻮䕿䕿䕿㻷㼀㻙㻞㻝㻡㻸㻲 㻷㼀㻙㻝㻡㻡㻲㻛㻱㼄
NFC
NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕
䕿
䕿
䕿
䠉
䠉
䠉

Range
⠊ᅖ

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻠㻜䡚㻗㻤㻜䉝

㻙㻤㻜䡚㻗㻟㻜䉝
-80‫ع‬+80ႏ

Precision⢭ᗘ㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻟䉝㻔㻙㻝㻜䡚㻡㻜䉝㻕㼼㻜㻚㻡䉝㻔እഃ⠊ᅖ㻕㼼㻜㻚㻡䉝㻔㻙㻤㻜䡚㻟㻜䉝㻕㻜㻚㻝䉝㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻟㻞㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
㻜䡚㻥㻥䠂
㼼㻡䠂

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻟㻞㻘㻜㻜㻜
㻜䡚㻥㻥䠂
㼼㻡䠂

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
㻜䡚㻥㻥䠂
㼼㻡䠂

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
㻜䡚㻥㻥䠂
㼼㻡䠂

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻟㻞㻘㻜㻜㻜
㻜䡚㻥㻥䠂
㼼㻡䠂

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
㻜䡚㻥㻥䠂㼼㻡䠂㻝䡚㻞㻡㻡ศ
㻟㻞㻘㻜㻜㻜
䠉䠉䠉䠉䠉㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉䠉䠉䠉䠉㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻟㻞㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉
㻡䡚㻣㻡㻳
㼼㻝㻡䠂
㻝㻳
When
percepting
ឤ▱䛾㒔ᗘ
㻞㻘㻜㻜㻜

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻠㻘㻜㻜㻜
㻜䡚㻥㻥䠂㼼㻡䠂㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉

㻜㻚㻝䉝
㻝䡚㻞㻡㻡ศ
㻝㻠㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉
㻡䡚㻣㻡㻳
㼼㻝㻡䠂
㻝㻳
When
percepting
ឤ▱䛾㒔ᗘ
㻞㻘㻜㻜㻜

㻝䡚㻞㻡㻡ศ
㻝㻠㻘㻜㻜㻜
㻡䡚㻣㻡㻳
㼼㻝㻡䠂
㻝㻳
When
percepting
ឤ▱䛾㒔ᗘ
㻞㻘㻜㻜㻜

㻝䡚㻞㻡㻡ศ
㻟㻞㻘㻜㻜㻜
㻡䡚㻣㻡㻳
㼼㻝㻡䠂
㻝㻳
When
percepting
ឤ▱䛾㒔ᗘ
㻞㻘㻜㻜㻜

㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉䠉䠉䠉䠉Temperatureᗘ
Humidity‵ᗘ(Outside range)
Resolutionศゎ⬟
Period
(min)
ᐃ࿘ᮇ
Data
capacity
䝕䞊䝍
Range
⠊ᅖ
Precision⢭ᗘ

Humidity‵ᗘ

Resolutionศゎ⬟
Period
(min)
ᐃ࿘ᮇ
Data
capacity
䝕䞊䝍
Range
⠊ᅖ
Precision⢭ᗘImpact⾪ᧁ

155F/EX㻷㼀㻙㻝㻡㻡㻲
NFC
㻺㻲㻯㻔㻲㼑㼘㼕㻯㼍㻕䕿䠉䠉Communication
Systemㄞ㎸᪉ἲ
Measurementᐃᑐ㇟
ObjectLowᗘ䠄ప 䠅
Temperature

Resolutionศゎ⬟
Period
(min)
ᐃ࿘ᮇ
Data
capacity䝕䞊䝍ᵵᵟᵲᵡᵦᵪᵭᵥᵥᵣᵰᴾἻỶὅỴἕἩ㻜㻚㻝䉝㻝䡚㻞㻡㻡ศ
㻝㻢㻘㻜㻜㻜
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉
䠉

Table A.2
ೞᆔൔ᠋ᘙCommuPPXQL
nication
System\Circuit.7)ध৊঳
Same asKT175F

1R

Model
0RGHO
No.



KT-175F

ആ২
Temperature)H/L&Dৢਦ
NFC

KT-165F

ആ২
Temperatur)H/L&Dৢਦ
NFC

KT-275F

ആ২‫؞‬ဒ২
Temperature/ Humidity)H/L&Dৢਦ
NFC

KT-265F

Temperatureആ২‫؞‬ဒ২
/ Humidity)H/L&Dৢਦ
NFC

External([WHUQ

Sensor.7)ध৊঳
Same asKT275F

Enclosure
Dimension

WeightBattery¼¼PP

J&5

¼¼PP

J&5

¼¼PP

J&5

¼¼PP

J&5

.7)ध.7)म‫ؚ‬৐৚भ૥ୡद‫ؚ‬घदपఃஈੋभਃரदघ‫؛‬
KT-175F
and KT-275F have been already registered on last test.
.7)ध.7)ऋ‫ؚ‬০৚୯ਸఃஈखथःञटऌञःਃரदघ‫؛‬
KT-165F
and KT-265F are the models to be added.
KT-165F is changed the enclosure size to smaller than KT-175F, and has the same function as KT-175F with five color panels.
.7)म‫ؚ‬.7)भᲕ৬१ॖ६॑৵औऎखञुभद‫ؚ‬ਃચम.7)ध৊गदघऋ‫ؚ‬ঃॿঝऋ‫ڱ‬౦ँॉऽघ‫؛‬KT-265F is changed the enclosure size to smaller than KT-275F, and has the same function as KT-275F with five color panels.
.7)म‫ؚ‬.7)भᲕ৬१ॖ६॑৵औऎखञुभद‫ؚ‬ਃચम.7)ध৊गदघऋ‫ؚ‬ঃॿঝऋ‫ڱ‬౦ँॉऽघ‫؛‬
KT-165F
circuit is entirely the same as KT-175F's one except for the different battery capacity.
.7)भ৚ଡ଼म‫ؚ‬ਗ਼౱ઍ୤ऋ౮ऩॊਰਗम‫ؚ‬.7)भ৚ଡ଼ध৸ऎ৊ग৚ଡ଼पऩॉऽघ‫؛‬KT-265F
circuit is entirely the same as KT-275F's one except for the different battery capacity.
.7)भ৚ଡ଼म‫ؚ‬ਗ਼౱ઍ୤ऋ౮ऩॊਰਗम‫ؚ‬.7)भ৚ଡ଼ध৸ऎ৊ग৚ଡ଼पऩॉऽघ‫؛‬Batteryਗ਼౱&5म‫ؚ‬ઍ୤ऋ‫ۗڶۜڬڬگ‬भজॳक़঒॥ॖথਗ਼౱दघ‫؛‬
CR2430 is the lithium coin battery with 300mAh capacity.
Batteryਗ਼౱&5म‫ؚ‬ઍ୤ऋ‫ۗڶۜڬڲڭ‬भজॳक़঒॥ॖথਗ਼౱दघ‫؛‬
CR2025 is the lithium coin battery with 160mAh capacity.
84F02930-01ᵏTable A.3
ೞᆔൔ᠋ᘙ
ImpactMeasured೾৒ৌ଴ӂӡҿҴ
Sensor&RPPXৢਦ্ૄQLFDWLRQ
6\VWHP

Impact้ொॹ‫ॱش‬Record੶ஈઍ୤
Capacity

Measure้ொ೾৒෇೧

㻷㼀㻙㻝㻥㻡㼁

Temperatureആ২‫้؞‬ொ/ Impact

USB
86%ৢਦ

2,000
events੯

‫*ع‬ᾁ㻷㼀㻙㻝㻥㻡㼁㻛㻳㼄ആ২‫้؞‬ொ/ Impact

Temperature

USB
86%ৢਦ

16,000events੯

‫*ع‬ᾂ㻷㼀㻙㻞㻥㻡㼁

Temperatureആ২‫؞‬ဒ২‫้؞‬ொ

USB
86%ৢਦ

2,000
events੯

‫*ع‬ᾃ㻷㼀㻙㻞㻥㻡㼁㻛㻳㼄Temperatureആ২‫؞‬ဒ২‫้؞‬ொ/ Humidity / Impact

USB
86%ৢਦ

16,000events੯

‫*ع‬ᾄ㻷㼀㻙㻝㻥㻡㻲

Temperatureആ২‫้؞‬ொ/ Impact

NFC)H/L&Dৢਦ

2,000
events੯

‫*ع‬ᾅ㻷㼀㻙㻝㻥㻡㻲㻛㻳㼄ആ২‫้؞‬ொ/ Impact

Temperature

NFC)H/L&Dৢਦ

16,000events੯

‫*ع‬ᾆ㻷㼀㻙㻞㻥㻡㻲

ആ২‫؞‬ဒ২‫้؞‬ொ/ Humidity / Impact

Temperature

NFC)H/L&Dৢਦ

2,000
events੯

‫*ع‬ᾇ㻷㼀㻙㻞㻥㻡㻲㻛㻳㼄Temperatureആ২‫؞‬ဒ২‫้؞‬ொ/ Humidity / Impact

NFC)H/L&Dৢਦ

16,000events੯

‫*ع‬

‫۞ۃ‬

Model
ਃர੡
Name

ᾀ

ਗ௴
Eternal/ Humidity / Impact

Range

Other
Circuitजभ౎भ৚ଡ଼

Same
as KT-195U.78ध৊঳

Same as KT-295U.78ध৊঳

Same as KT-195F.7)ध৊঳

Same
as KT-295F.7)ध৊঳

EnclosureᲕ৬࿡১‫كۜۜق‬੎୤
WeightDimensionઞ৷ਗ਼౱Battery¼¼

J&5

¼¼

J&5

¼¼

J&5

¼¼

J&5

¼¼

J&5

¼¼

J&5

¼¼

J&5

¼¼

J&5

Recordingcapacities of KT-195U/GX, KT-295U/GX, KT-195F/GX and KT-295F/GX are higher than KT-195U, KT-295U, KT-195F and KT-295F.
.78‫؞‬.78‫؞‬.7)‫؞‬.7)पৌखथ‫ؚ‬.78*;‫؞‬.78*;‫؞‬.7)*;‫؞‬.7)*;म‫੶ؚ‬ஈઍ୤ऋੜइथःऽघ‫؛‬
Impact measurement
ranges of KT-195U/GX, KT-295U/GX, KT-195F/GX and KT-295F/GX are wider than KT-195U, KT-295U, KT-195F and KT-295F.
.78‫؞‬.78‫؞‬.7)‫؞‬.7)पৌखथ‫ؚ‬.78*;‫؞‬.78*;‫؞‬.7)*;‫؞‬.7)*;म‫ؚ้‬ொ೾৒෇೧ऋઁऎऩढथःऽघ‫؛‬The memories
and impact sensors for KT-195U_KT-295U and KT-195U/GX_KT-295U/GX are different. The other circuits are same.
.78‫؞‬.78ध.78*;‫؞‬.78*;म‫ؚ‬૚‫ر‬ऋওঔজ‫ش‬ध้ொ७থ१‫ش‬भ૗ಌਰਗम৊ग৚ଡ଼पऩॉऽघ‫؛‬The memories
and impact sensors for KT-195F_KT-295F and KT-195F/GX_KT-295F/GX are different. The other circuits are same.
.7)‫؞‬.7)ध.7)*;‫؞‬.7)*;म‫ؚ‬૚‫ر‬ऋওঔজ‫ش‬ध้ொ७থ१‫ش‬भ૗ಌਰਗम৊ग৚ଡ଼पऩॉऽघ‫؛‬ᵖᵒᵤᵎᵑᵐᵏᵗᵋᵎᵏᵏ

7DEOH$

6HQVRU&RPPXQLFDWLRQ
6\VWHP

5HFRUG&DSDFLW\

0HDVXUHPHQW
5DQJH

2WKHU&LUFXLW(QFORVXUH'LPHQVLRQ PP:HLJKW%DWWHU\㻷㼀㻙㻝㻡㻡㻲㻼

7HPSHUDWXUH

NFC)H/L&DHYHQWVf٦

6DPHDV.7)¼¼

J&5

ᾁ㻷㼀㻙㻝㻝㻡㻸㻲㻼

7HPSHUDWXUH

NFC)H/L&DHYHQWVf٦

6DPHDV.7;/)¼¼

J&5

ᾂ㻷㼀㻙䠍䠑䠑㻲㻛㻱㼄㻔㻸㻱㻰㻕

7HPSHUDWXUH

NFC)H/L&DHYHQWVf٦

6DPHDV.7)(;¼¼

J(5

ᾃ㻷㼀㻙㻞㻡㻡㻲㻛㻟㻞

7HPSHUDWXUH
+XPLGLW\

NFC)H/L&D

7HPSHUDWXUH
HYHQWV+XPLGLW\HYHQWV

7HPSHUDWXUHf٦
+XPLGLW\f٫

6DPHDV.7)¼¼

J&5

ᾄ㻷㼀㻙㻝㻝㻡㻸㻲㻼㻛㻭

7HPSHUDWXUH

NFC)H/L&DHYHQWVf٦

6DPHDV.7;/)¼¼ۖ&5‫۞ۃ‬

0RGHO1R

ᾀ([WHUQDO

7KHFLUFXLWRIWKHVHPRGHOVDUHWKHVDPHDVWKHPRGHOVOLVWHGLQn2WKHU&LUFXLW|
7KHFLUFXLWRI.7)3.7/)3DQG.7/)3$DUHWKHVDPHDQGWHPSHUDWXUHVHQVRULVSUREHW\SH
7KHFLUFXLWRI.7)(; /(' LVWKHVDPHDV.7)(;DQGWKH/('WXUQV21ZKHQUHFRUGLQJWHPSHUDWXUHRXWRIVHWWLQJUDQJH
7KHFLUFXLWRI.7)LVWKHVDPHDV.7)2QO\WKHPHPRU\FDSDFLW\RI.7)LVGLIIHUHQW

ᵖᵒᵤᵎᵑᵈᵈᵈᵋᵎᵏᵏ

7DEOH$

6HQVRU&RPPXQLFDWLRQ
6\VWHP

5HFRUG&DSDFLW\

0HDVXUHPHQW
5DQJH

2WKHU&LUFXLW(QFORVXUH'LPHQVLRQ PP:HLJKW%DWWHU\㻷㼀㻙㻝㻡㻡㻰㼄

7HPSHUDWXUH

NFC)H/L&D%/(HYHQWVf٦

%/(DGGHGRQ.7)¼¼

J&5

ᾁ㻷㼀㻙㻞㻡㻡㻰㼄

7HPSHUDWXUH
+XPLGLW\

NFC)H/L&D%/(%/(DGGHGRQ.7)¼¼

J&5

ᾂ㻷㼀㻙㻝㻥㻡㻰㼄

7HPSHUDWXUH,PSDFWNFC)H/L&D%/(%/(DGGHGRQ.7)¼¼

J&5

ᾃ㻷㼀㻙㻞㻥㻡㻰㼄

7HPSHUDWXUH
+XPLGLW\,PSDFW

NFC)H/L&D%/(%/(DGGHGRQ.7)¼¼

J&5

ᾄ㻷㼀㻙㻝㻝㻡㼄㻰㼄

7HPSHUDWXUH

NFC)H/L&D%/(HYHQWVf٦

%/(DGGHGRQ.7;/)¼¼

J&5

ᾅ㻷㼀㻙㻞㻝㻡㼄㻰㼄

7HPSHUDWXUH
+XPLGLW\

NFC)H/L&D%/(

7HPSHUDWXUH HYHQWV+XPLGLW\
HYHQWV

7HPSHUDWXUHf٦
+XPLGLW\f٫%/(DGGHGRQ.7;/)¼¼

‫ڵ‬J&5

ᾆ㻷㼀㻙㻝㻡㻡㻰㼄㻼

7HPSHUDWXUH

NFC)H/L&D%/(HYHQWVf٦

%/(DGGHGRQ.7)3¼¼

J&5

ᾇ㻷㼀㻙㻝㻝㻡㻰㼄㻼

7HPSHUDWXUH

NFC)H/L&D%/(HYHQWVf٦

%/(DGGHGRQ.7/)3¼¼

‫ڵ‬J&5‫۞ۃ‬

0RGHO1R

ᾀ([WHUQDO

7HPSHUDWXUH
7HPSHUDWXUH HYHQWVf٦
+XPLGLW\
+XPLGLW\f٫
HYHQWV
7HPSHUDWXUH
7HPSHUDWXUHf٦
 HYHQWV,PSDFWHYHQWV ,PSDFW‫*ع‬7HPSHUDWXUH
7HPSHUDWXUH HYHQWVf٦
+XPLGLW\
+XPLGLW\f٫
HYHQWV,PSDFW‫*ع‬,PSDFWHYHQWV$OOWKHPRGHOVDERYHDUHWKH%OXHWRRWKIXQFWLRQDGGLWLRQDOPRGHOVRIWKHPRGHOVOLVWHGLQn2WKHU&LUFXLW|$OOWKHPRGHOVDERYHDUHXVHGLQDLUERUQHPRGH %OXHWRRWKIXQFWLRQLV2)) LQWKHDLUFUDIW
7KHFLUFXLWRI.7';3DQG.7';3DUHWKHVDPHDVWKHPRGHOVOLVWHGLQn2WKHU&LUFXLW|DQGWHPSHUDWXUHVHQVRULVSUREHW\SH

ᵖᵒᵤᵎᵑᵈᵈᵈᵋᵎᵏᵐ