Bulletin No.MIJ-B6112D

取扱説明書
EA10S 形バルブポジショナ（HART 通信編）

Model EA10S【目 次】
1 概要 ........................................................................................................................................................................ 2
1.1 適用範囲 .......................................................................................................................................................... 2
1.2 接続の概要 ...................................................................................................................................................... 2
2 準備 ........................................................................................................................................................................ 3
2.1 PC の準備 ......................................................................................................................................................... 3
2.2 DD ファイルのダウンロード ................................................................................................................................ 33 接続 ........................................................................................................................................................................ 4
3.1 FieldMate の起動 ............................................................................................................................................ 4
3.2 DD ファイルのインストール ................................................................................................................................ 6
3.3 DTM Works の起動 ......................................................................................................................................... 8
3.4 DTM Works の終了 ....................................................................................................................................... 10
3.5 FieldMate の終了 .......................................................................................................................................... 11
4 メニューの説明 ...................................................................................................................................................... 12
4.1 Drive Information ......................................................................................................................................... 12
4.2 Detail ............................................................................................................................................................. 13
4.3 Device Information ....................................................................................................................................... 14
4.4 HART Information ........................................................................................................................................ 15
4.5 Calibration .................................................................................................................................................... 16
4.6 Hand Calibration .......................................................................................................................................... 17
4.7 Parameters .................................................................................................................................................... 18
4.8 Valve Setting ................................................................................................................................................. 19
4.9 User Set Character ....................................................................................................................................... 20
4.10 Diagnostic .................................................................................................................................................... 21
4.11 Cycle Count（診断機能…反転動作回数アラーム）......................................................................................... 224.12 Travel Accumulator（診断機能…作動距離積算アラーム） ............................................................................ 23
4.13 Full Close Count（診断機能…全閉回数アラーム） ........................................................................................ 24
4.14 Deviation（診断機能…開度偏差アラーム） .................................................................................................... 25
4.15 Review ......................................................................................................................................................... 26
5 操作方法 ............................................................................................................................................................... 27
5.1 数値入力、リスト選択タイプ ............................................................................................................................. 27
5.2 実行タイプ ...................................................................................................................................................... 30
5.3 設定データのレポート化 .................................................................................................................................. 32
5.4 設定パラメータの保存/ロード ........................................................................................................................... 35
5.5 トレンドチャート ............................................................................................................................................... 39
5.6 操作履歴・パラメータ変更履歴 ........................................................................................................................ 42
5.7 機器アイコン設定 ............................................................................................................................................ 44

1 / 45Model EA10S

1 概要
1.1 適用範囲
本書はEA10S形式バルブポジショナのHART®(※1)通信機能を使用する際の説明書です。当説明書では通信ソフト事例として、横河電機株式会社のFieldMate®(※2) Rev3.01.10 の画面を使って説明をします。
※1）HARTはFieldComm Groupの登録商標です。
※2）FieldMate とは、横河電機株式会社の機器調整・設定・管理ソフトウェアであり、横河電機株式会社の登録商標です。 (マニュアルや、最新の対応 OS は横河電機株式会社のホームページを参照ください。)

1.2

接続の概要
EA10S形電空ポジショナでHART通信を行うときの、接続の概要を下図に示します。

±の極性は
ありません

FieldMate®
インストール済み PC

・PCにはHARTモデムのドライバ、通信ソフト（FieldMate®）、EA10S用DDのインストールが必要です。
･電流源（電流発生器・IOカードなど）は、電源電圧が少なくとも10Ｖ以上出力できるものを使用して下さい。
・HARTモデム(※3)は、図のA、Ｂのポイントにクリップで接続して下さい。極性はありません。
※3）HART モデムは別途、ご入手ください。参考までに以下の製品を紹介いたします。
-

横河電機株式会社 USB FieldMate Modem(FiledMate 購入時、オプションにて付属)-

株式会社エム・システム技研 USB 対応 BELL202 モデル(別売り)-

ピーアンドエフ社 VIATOR USB HART Interface(別売り)

2 / 45Model EA10S

2 準備
2.1 PC の準備
PCに、HARTモデムのドライバ、および通信用ソフトウェア（FieldMate）をインストールします。
FieldMateの詳細については、横河電機HPのFieldMateマニュアルをダウンロードし参照して下さい。
①

FieldMateインストール後、初回のみDTMカタログの更新をしてください。

②（Windows10の場合）スタートメニューより、「YOKOGAWA FieldMate」のプルダウンより、「DTM Setup」をクリックします。

③

上図の.exeフォルダを起動後に表示される「DTMカタログを更新しますか？」のダイアログで「はい」を④クリックすることでDTMカタログの更新が自動で行われます。

⑤

更新完了後、DTM Setup画面を終了してください。

※ この操作は初回のみ行えばよいでしょう。

2.2 DD ファイルのダウンロード
FieldMate で EA10S の詳細なパラメータ調整を HART 通信上で利用するために、DD（Device Description）ファ
イルのダウンロードとインストールが必要です。
①

弊社の HP 上 EA10S のページ(https://www.motoyama-cp.co.jp/product/ea10s.html)にある「通信用 DD」をクリックし、任意の場所にダウンロードします。

②

ダウンロードした「EA10S_DD.zip」を展開します。

③

フォルダ内の「006044」→「e184」以下が当ポジショナの DD ファイルです。

※ この操作は初回のみ行えばよいでしょう。
※ DD のインストール方法については後述します。

3 / 45Model EA10S

3 接続
3.1 FieldMate の起動① 電流源からEA10Sに4mA以上の電流を入れます。
② HARTモデムをPCへ接続し、HARTモデムのクリップをライン上に接続します。
③ FieldMateを立ち上げます。以下のようなログイン画面が表示されます。
通信設定は「HART」を選択し、「ログイン」をクリックします。

4 / 45Model EA10S④ 起動時にメイン画面が以下のような機器ナビゲータ画面になっていた場合、画面左下のセグメントビュ
ーワアイコンをクリックしてセグメントビューワ画面に移行してください。

セグメントビューワ
アイコン⑤ セグメントビューワにて、画面左上の「更新」をクリックし、以下のような画面が出れば接続OKです。

接続されている HART 機器（EA10S）が認識されると、機器がリストアップされる。

5 / 45Model EA10S
3.2 DD ファイルのインストール
EA10S用のDD（Ｄｅｖｉｃｅ Ｄｅｓｃｒｉｐｔｉｏｎ）ファイルをインストールします。
① 接続が完了した画面で以下の○部を右クリックまたは「操作」をクリックし、「DDファイルのインストール」
をクリックします。

DD ファイルのインストール
をクリックする。

② 「この機器のEDDファイルを指定してインストール」をクリックします。

③ DDファイルを保管している場所を選びます。(本山製作所のDDファイルフォルダは「006044」→「e184」です。)

6 / 45Model EA10S④ 「機器タイプと異なるDDファイルを選択した可能性があります。このまま続けますか？」という警告文が出ることがありますが「はい」をクリックしてください。
⑤ 以下の画面が出たら「OK」をクリックします。

注意： 上記は2023年2月時点のDDファイルです。DDリビジョンが異なる場合（例：0205、0206など）がありま
すが、随時アップグレードしているため、そのままインストールいただいて問題ありません。

⑥ 「処理に成功しました」と出ればDDファイルインストールの完了です。「OK」をクリックしてください。

7 / 45Model EA10S
DTM Works の起動
DD ファイルを読み込み、内部詳細パラメータを表示・書き込みするためのアプリケーション「DTM
Works」を起動します。
① 接続が完了した画面で以下の○部を右クリックするか、右側にある「操作」というアイコンをクリックします。

DTM を選択をクリックする。

② 「DTM を選択」をクリックします。

③ 「HART Built-in DTM(一番下にあります。)」を選択し、「OK」をクリックします。

8 / 45Model EA10S④ 「選択した DTM を起動しますか？」と出ますので「はい」をクリックします。

⑤ 「DTM データロードオプション」というタブが出ますので「デフォルトデータをロード」を選択
し、「OK」をクリックします。

⑥ DD ファイル画面が表示されます。画面左側にメニューツリーが表示され、メニューツリーのメニ
ューの左側にある「+」マークをクリックするとメニュー階層が展開表示されます。
この後の状態確認や調整、設定の変更は、このメニューを選択することで行えるようになります。

ツールバー

右側にアプリケーションエリア

左側にメニューツリー
（ナビゲーションエリア）

9 / 45Model EA10S
3.3 DTM Works の終了①

画面左上の「ファイル」をクリックし、「終了」をクリックします。(画面右上の「閉じる」のボタンでも同様の動作が行えます。)② 「DTM Works を終了しますか」と出るので「はい」をクリックします。

③ 「DTM データをデータベースに保存しますか」と出ますが、「いいえ」クリックしてください。

10 / 45

Model EA10S
3.4 FieldMate の終了① 画面左上の「ファイル」をクリックし、「終了」をクリックします。(画面右上の「閉じる」のボタンでも同様の動作が行えます。)② 「終了しますか」というタブが出ますので、「はい」をクリックして終了です。

11 / 45

Model EA10S

4 メニューの説明
ここでは、DD 画面左側に表示されるメニューツリー階層と、その中の各パラメータ項目の説明を記載します。
通常のパラメータ項目は参照のみ可能です。太字のパラメータ項目は設定変更ができます。

4.1 Drive Information

現在のポジショナの運転状況が確認できます。表示される項目は、・Loop current：ポジショナに入力されている電流値

・Valve Position(%)：現在の調節弁の開度（％）

・Error (%)：入力信号と調節弁開度の偏差（％）

・Control Mode

：入力信号の区分を示します。
Normal（入力電流に追従する）と、
Fixed Control（HART 上の設定信号に追従する）の切換えができます。

・Fixed Control Value(%)：HART 上から開度指示ができます。
Control Mode を Fixed Control にし、このパラメータの数値を設定する
ことで、入力電流に関係なく開度指示ができます。

12 / 45

Model EA10S
4.2 Detail

現在のポジショナの運転に関わる内部データが参照できます。表示される項目は、・BIAS LOW (25%)：オートキャリブレーションで設定された LOW 側のバイアス値。

・BIAS HIGH(75%)：オートキャリブレーションで設定された HIGH 側のバイアス値。

・I Value：現在の積分値。

・I Max：オートキャリブレーションで設定された、積分値の上限。

・I Min：オートキャリブレーションで設定された、積分値の下限。

・Mv：現在の、トルクモータへのデジタル出力値（0～4095）。

13 / 45

Model EA10S
4.3 Device Information

ポジショナの Tag などの情報を設定できます。表示される項目は、・Write Protect：ポジショナの書込み保護の有効／無効が設定できます。

・Tag：Tag を入力できます（８文字以内）。

・Long Tag：Long Tag を入力できます（３２文字以内）

・Descriptor：調整者名などを入力できます。

・Message：メッセージを入力できます。
（３２文字以内）

・Date：設定した年月日を入力できます。
（m/d/yyyy：月/日/西暦年の書式）

・Final asmbly number：任意の Final assembly number を入力できます。

・Valve Serial number：調節弁のシリアルナンバー等を入力できます。

・Cfg chng count：パラメータ変更が何回行われたかを確認できます。

HART 通信上で、１つのパラメータ変更で＋１回、複数パラメータ同時変更で変更数分＋されます。
また、キャリブレーション 1 回に＋１回カウントされます。
本体操作で、パラメータを 1 回変更、またはキャリブレーション 1 回に＋１回カウントされます。
・Device Display mode

：本体ＬＣＤの表示を変更できます。
Normal（通常表示）
Reverse（0-100%を逆に表示）
Digital（内部データ表示）

・Loop current mode

：入力電流の値を設定値として有効に／無効に設定できます。※
Enabled（有効）／Disabled（無効）※Disabled（無効）に設定すると、入力電流に追従しなくなりますので、通常は Enabled（有効）にして下さい。

14 / 45

Model EA10S
4.4 HART Information

ポジショナの HART に関する情報を確認できます。表示される項目は、・Universal rev：HART のバージョンを確認できます。

・Fld dev rev：EA10S のデバイスリビジョンを確認できます。

・Software rev：ソフトウェアのリビジョンを確認できます。

・Num req preams：メッセージ要求時の最小プリアンブル数を確認できます。

・Num resp preams：メッセージ応答時の最小プリアンブル数を確認できます。

・Poll addr

：HART ポーリングアドレスを確認、変更できます。

・HT Feedback：HART 通信時の開度情報を正／逆方向に設定できます。
HT NORMAL （通常表示）
HT REVERSE（0-100%を逆に表示）

15 / 45

Model EA10S
4.5 Calibration

ポジショナのキャリブレーションを実行できます。表示される項目は、・AUTO1 Calibration：オート１キャリブレーション（ストローク調整のみ）を実行できます。

・AUTO2 Calibration：オート２キャリブレーション（全て調整）を実行できます。

・AUTO3 Calibration：オート３キャリブレーション（ストローク調整除き全て）を実行できます。

・BIAS Calibration：バイアスキャリブレーション（トルクモータのバイアス）を実行できます。

ストローク調整

制御パラメータ

駆動部の作動方向

バイアス調整

AUTO1

○

×

○

×

AUTO2

○

○

○

○

AUTO3

×

○

○

○

BIAS

×

×

×

○

16 / 45

Model EA10S
4.6 Hand Calibration

ポジショナのストローク、アナログ弁開度出力に関する内部設定値が確認、変更できます。表示される項目は、・PV Zero：調節弁の０％位置のデータです。

・PV End：調節弁の１００％位置のデータです。

・TR Zero：開度出力機能（ｵﾌﾟｼｮﾝ）の 0%点（4mA 入力時）設定、
デジタル 12bit 値 0～4095。

・TR End：開度出力機能（ｵﾌﾟｼｮﾝ）の 100%点（20mA 入力時）設定、
デジタル 12bit 値 0～4095。

・TR Direction：開度出力機能（ｵﾌﾟｼｮﾝ）の正・逆方向設定。
NORMAL （正方向）REVERSE（逆方向）

17 / 45

Model EA10S
4.7 Parameters

ポジショナの PID 制御パラメータの確認、変更ができます。

・Deadzone：不感帯。目標値に対して積分制御を停止する帯域（%）を設定します。
グランドパッキンの摩擦力が大きく、ハンチングを生じる場合など、不感帯を設定することで、ハンチングを防止することが出来ます。

・KP：比例ゲイン。大きくすると、目標値と弁開度の偏差を速やかに小さくすることができます。しかし、大きくしすぎるとハンチングが生じます。

・KI：積分定数。大きくすると、目標値と弁開度の偏差が補正される時間が短くなります。
しかし、大きくしすぎるとハンチングが生じます。

・KD：微分定数。大きくすると、目標値に到達するまでの時間を短くできます。
しかし、大きくしすぎるとハンチングが生じます。

・KP_, KI_, KD_：目標値と弁開度の偏差が 1%以内の時に各定数の強さを変更するためのパラメータです。標準値は 10 ですが、例えば KP_のパラメータを 12 と変更することで、目標値と弁開度の偏差が 1%以内の時には、比例ゲインは設定されたパラメータの120%とすることができます。

・KI Latency：バルブがシートしている状態から、動き出すまでの秒数です。通常はオート
キャリブレーションで設定されますので、変更する必要はありません。

18 / 45

Model EA10S
4.8 Valve Setting

調節弁に関するパラメータの確認、変更ができます。表示される項目は、・Actuator Direction：駆動部の動作方向です。Direct／Reverse に変更できます（通常はオートキャリブレーションで設定されます）

・Positioner Direction：ポジショナの動作方向です。
Normal （入力信号増加で、出力空気圧増加）
Reverse（入力信号増加で、出力空気圧減少）

・Valve Direction：調節弁の全閉方向です。
Direct （ステムが下降して閉）
Reverse（ステムが上昇して閉）・Tight-shut Open

：強制的に調節弁を全開にする信号％です。通常は無効（１００％）です。

・Tight-shut Close：強制的に調節弁を全閉にする信号％です。通常は０．５％に設定されています。

・Signal Zero

：０％指示の入力電流です。通常は４ｍＡに設定されています。

・Signal End：１００％指示の入力電流です。通常は２０ｍＡに設定されています。

・Full Open Time

：調節弁の全閉→全開の作動時間です。

・Full Close Time

：調節弁の全開→全閉の作動時間です。

・Control characteristic：制御時の特性が変更できます。
リニア（Linear）／ユーザ定義（User set character)／
クイックオープン(Sq root)／イコールパーセント(Square)／イコールパーセント２(Equal Percent2(92A))

19 / 45を選択設定可能Model EA10S
4.9 User Set Character

ユーザ定義の特性を確認、変更ができます。
P0 (at 0.00%)～ P16 (at 100.00%)までの 6.25%刻みの１７点の制御特性を確認、変更ができます。
P17 (at 106.25%)は、100％以上の信号入力に対する傾きを設定できます。
なお、各ポイントの数値はなだらかになるように設定して下さい。
このユーザ特性は、前項の Control characteristic で User set character を選択すると有効になります。

20 / 45

Model EA10S
4.10 Diagnostic

診断機能の設定、ポジショナ内部のエラーの確認、リセットができます。表示される項目は、・Diagnostic Config：診断機能メニューに入ります。

・Master reset：ポジショナをリセット…電源 OFF→ON と同じ状態にします。

・Self Test

：ポジショナ内部に問題がないか Self Test を実行します。
※

バルブ診断機能ではありません。

・Reset Configuration Changed ：Device Status の設定変更ビットをクリアします。
・Reset More Status Available : ERROR/WARNING/ALARM 等によって生じた Device Status の「More Status
Available」ビットをクリアします。
・Device Status Check: Device Status（HART 仕様準拠）を確認します。

・Adjust 4mA Signal:入力電流基準値(4mA 側)を校正・設定します。操作時は 4mA 入力必須です。

・Adjust 20mA Signal：入力電流基準値(20mA 側)を校正・設定します。20mA 入力必須です。

・Device status：Device Status（HART 仕様準拠）。

・ERROR：動作上のエラーを表示します。（Command48 にて実装）

・WARNING：警告を表示します。
（Command48 にて実装）

・INFORMATION：通知を表示します。
（Command48 にて実装）

・ALARM：診断機能のアラームを表示します。
（Command48 にて実装）※

上記 Device Status と Command48（ERROR/WARNING/INFORMATION/ALARM）は、別途 HART パラメータリスト

説明ドキュメントにて詳細を説明します。パラメータリストの入手は、お近くの支店・営業所にお問い合わせください。

21 / 45

Model EA10S
4.11 Cycle Count（診断機能…反転動作回数アラーム）

診断機能の反転動作回数の設定・確認ができます。表示される項目は以下の通りです。
・Cyc Cnt Function：カウンタの ON/OFF です。初期値は OFF です。

・Cyc Cnt Limit Alarm：アラーム発報の ON/OFF 設定です。初期値は OFF です。
・Cyc Cnt Deadzone：バルブが反転したと判断する、反転してからのステム移動距離（開度％）のしきい値設定です。
・Cyc Cnt Limit Number：アラーム発報するしきい値（回数）の設定です。

・Cyc Cnt Number：現在のカウンタ値です。自動更新されないため、以下の Read Diag Value

ボタンをクリックして更新してください。
・Reset Cyc Cnt Number：カウンタ値を０リセットします。

・Read Diag Value：現在のカウンタ値を読み込みます。

【アラーム発報画面】カウンタ値がしきい値以上になると、Configuration→Diagnostic 画面の ALARM で「Cycle
Count Limit」が表示されます。

22 / 45

Model EA10S
4.12 Travel Accumulator（診断機能…作動距離積算アラーム）

診断機能の作動距離積算の設定・確認ができます。表示される項目は以下の通りです。
・Trv Acc Function：カウンタの ON/OFF です。初期値は OFF です。

・Trv Acc Limit Alarm：アラーム発報の ON/OFF 設定です。初期値は OFF です。

・Trv Acc Deadzone：作動したと判断するための最少ステム作動距離（開度％）のしきい値設定です。
・Trv Acc Limit Number：アラーム発報するしきい値（％）の設定です。

・Trv Acc Value：現在のカウンタ値です。自動更新されないため、以下の Read Diag Value

ボタンをクリックして更新してください。
・Reset Trv Acc Number：カウンタ値を０リセットします。

・Read Diag Value：現在のカウンタ値を読み込みます。

【アラーム発報画面】カウンタ値がしきい値以上になると、Configuration→Diagnostic 画面の ALARM で「Travel
Accumulator Limit」が表示されます。

23 / 45

Model EA10S
4.13 Full Close Count（診断機能…全閉回数アラーム）

診断機能の全閉回数の設定・確認ができます。表示される項目は以下の通りです。
・FC Cnt Function：カウンタの ON/OFF です。初期値は OFF です。

・FC Cnt Limit Alarm：アラーム発報の ON/OFF 設定です。初期値は OFF です。

・FC Cnt Deadzone：バルブが全閉したと判断する、ステム位置（開度％）のしきい値設定です。

・FC Cnt Limit Number：アラーム発報するしきい値（回数）の設定です。

・FC Cnt Number：現在のカウンタ値です。自動更新されないため、以下の Read Diag Value

ボタンをクリックして更新してください。
・Reset Cyc Cnt Number：カウンタ値を０リセットします。

・Read Diag Value：現在のカウンタ値を読み込みます。

【アラーム発報画面】カウンタ値がしきい値以上になると、Configuration→Diagnostic 画面の ALARM で「FullClose Count Limit」が表示されます。

24 / 45

Model EA10S
4.14 Deviation（診断機能…開度偏差アラーム）

診断機能の開度偏差アラームの設定・確認ができます。表示される項目は以下の通りです。
・Deviation Mode

：アラーム発報の ON/OFF です。初期値は ON です。

・Deviation Deadzone：偏差（％）のアラーム発報しきい値設定です。

・Deviation Time

：偏差が継続した時間（秒）のアラーム発報しきい値設定です。

・Read Diag Value：現在の各設定値を読み込みます。通常は使用しません。

【アラーム発報画面】偏差が Deviation Deadzone(%)以上の状態が、Deviation Time(秒)以上続くと、
Configuration→Diagnostic 画面の ERROR で「Deviation Limit」が表示されます。

偏差が長く続き、積分値が貯まっている状態なので、積分値飽和ウォーニングも発報されることがあります。

25 / 45

Model EA10S
4.15 Review

ポジショナのモデル名、メーカー名、Tag などの一般情報が確認できます。パラメータ項目は、他の各メニュ
ーと重複しており、同じパラメータです。

26 / 45

Model EA10S

5 操作方法
この章では、操作方法の例をあげ、設定の変更方法を解説します。

5.1

数値入力、リスト選択タイプ
数値入力、リスト選択の操作例として、Drive Information のメニューから、Control Mode を変更し、リストと数値を変更する方法を解説します。
① まずは、Control Mode の▼をクリックします（リスト操作１）。
② 「Fixed Value」を選択します（リスト操作２）。

③ 変更された箇所がピンク色に表示されます。
（これは、設定が変更されたが、ポジショナには転送されていない状態です。
）

27 / 45

Model EA10S④ 次に、Fixed Control Value(%)をクリックし、変更したい数値を入力します（数値入力操作）。
この時、前述同様、変更された箇所がピンク色になります。

⑤ 2 箇所の変更が終わったら、「機器にダウンロード」のボタンを押します。

28 / 45

Model EA10S⑥ これで設定変更が完了し、パラメータが EA10S に書き込まれました。

機器ダウンロードが完了すると、元の白色に戻ります。

29 / 45

Model EA10S
5.2 実行タイプ
実行タイプの操作例として、Configuration→Auto Calibration メニューの、Auto2 Calibration（フルキャリブレーション）を実行する方法を解説します。
①

Auto2 Calibration をクリックします。

② 自動制御ループから外れる旨のダイアログが表示されるので、「OK」を押します。

30 / 45

Model EA10S③ さらにキャリブレーション実行ダイアログで「OK」を押します。

④ Auto2 Calibration が開始されます。本体側もキャリブレーション開始されます。

⑤ Auto2 Calibration が終了すると、ダイアログの表示が以下のように Completed 表示になるので、「OK」を押して終了です。本体液晶でも COMPLETE の表示が出て完了します。

31 / 45

Model EA10S
5.3 設定データのレポート化
設定データのレポート化保存の方法を解説します。テキスト形式・HTML 形式・エクセル形式で保存可
能です。
① 設定が終了した後、FieldMate セグメントビューワ画面上の認識された EA10S リスト上の「全パラメータ」をクリックします。

32 / 45

Model EA10S② 以下のようなダイアログが現れますので、しばらく待ちます。

③ 読み込み完了後、以下のような画面になりますので、「Web ブラウザ」をクリックします。

33 / 45

Model EA10S④ WEB ブラウザが起動し、読み込んだデータが html ファイルで表示されます。

① （Google Chrome の場合）プラウザ内で右クリックし、「名前を付けて保存」をクリックします。

② 任意の保存場所を選び、ファイル名を決定後、「保存」をクリックします。
③ 同様にテキスト（メモ帳）形式、表（エクセル）形式で保存可能です。

34 / 45

Model EA10S
5.4 設定パラメータの保存/ロード
前回調整した全パラメータをデータベース/ファイルで保存し、再び EA10S ポジショナに取り込む方法をご説明します。EA10S 交換時などに便利な機能です。
【保存】① 接続完了画面の○部を右クリックし、「Parameter Manager」を開きます。起動にしばらく時間がかかります。

Parameter Manager をクリックします。

② Parameter Manager 画面が開いたら「保存/エクスポート」ボタンをクリックします。

35 / 45

Model EA10S③ 下記ダイアログが表示されますので、理由を書き、「データベースに保存」または「ファイルにでクスポート」のどちらかを任意にクリックして保存/エクスポートします。

④ これで設定パラメータの保存が完了です。

【ロード】① 保存時と同様に、「Parameter Manager」を開きます。

② 「ワークエリア」の「ロード/インポート」をクリックします。

36 / 45

Model EA10S③ 「データベースからロード」、あるいは「ファイルからインポート」をクリックします。

④ 「データベースからロード」では以下のようにデータベースが表示されますので、取り込みたいデータをクリックし、「OK」をクリックします。

37 / 45

Model EA10S⑤ 画面右側(○部)がパラメータになります。ロード/インポートにより変更された書き込み可能パラメ
ータは紫色になり書き込みチェックが自動的に入ります。

変更された読み込みのみのパラメータは黄色になります。書き込まれる内容を確認後、「チェックしたデータのダウンロード」をクリックします。

⑥ データベースに保存してあったデータがポジショナに書き込まれます。しばらく待ちます。

① これで設定パラメータのロードは完了です。Parameter Manager を終了します。

38 / 45

Model EA10S
5.5 トレンドチャート
FieldMate の Device Viewer を使用して、HART Command3 によるトレンドチャートを表示する方法を説明します。
① 接続画面の○部を右クリックし、「Device Viewer」をクリックします。

Device Viewer をク
リックします。

② Device Viewer 画面が起動します。
※ 最初に表示される「診断情報」タブは、HART 仕様の Device Status と Extend Device Status が表示されます。

39 / 45

Model EA10S③ 「トレンド情報」タブをクリックするとトレンドチャート画面が表示されます。

トレンド情報タブをク
リックします。

トレンドは、
PV が開度(%)で赤線、SV が偏差(%)で緑線、TV がドライブ値で青線、QV が現在の積分値(%)で黄線です。

表示されるトレンドは HART 仕様 Command3 で EA10S で割り付けた下表の４種類です。
【EA10S Command3 表示項目】
パラメータ

詳細説明

PV

バルブの開度(%)

ポテンショメータからのフィードバック値。

SV

入力信号%と実開度%間の偏差。(%)

入力電流と実開度 PV 値との偏差

TV

Maniulate Value。ドライブ値＝モータ制

偏差から PID 制御式により計算された現在のドラ御量(デジタル 12bit 値 0～4095)

イブ値

現在の I=内部積分値

PID 制御計算における現在の積分値。

QV④ トレンドチャートの更新速度を速くしたい場合は、ツールバーの「更新時間を設定します」ボタンをクリックします。

40 / 45

Model EA10S⑤ 「更新時間の設定」ダイアログが開きますので、更新時間を設定します。最短で 5 秒まで設定可能です。

⑥ 画面下の PV,SV,TV,QV をダブルクリックすると、下図のように設定ダイアログが開き、それぞれのトレンドのスケール設定・最大値・最小値・色、表示/非表示、などが設定可能です。
下図は、PV（開度％、赤線）をダブルクリックした事例です。

この表でダブルクリックします。

41 / 45

Model EA10S
5.6 操作履歴・パラメータ変更履歴
FieldMate の操作ログ機能を使用して、過去に FieldMate 上で行った操作履歴を表示する 1 例を説明します。より詳しい使い方は FieldMate マニュアルを参照ください。
① FieldMate 画面左下のビューワ切り替え部で、「操作ログ」ボタンを左クリックします。

② FieldMate 画面が「操作ログ」画面に切り替わります。

履歴リスト

カレンダーや期間指定ができます。

42 / 45

Model EA10S③ 「操作ログ」画面上の詳細を表示したい履歴リストの上で右クリック、または▼ボタンを左クリックし、出てきたサブ
メニューから「履歴情報を開く」を左クリックします。

履歴情報を開くを左クリックします。

④ 右側に詳細画面が出ますので、ここで過去にどのツールでどのような操作、パラメータ変更を行ったかの履歴が表示されます。

詳細画面

この例では、2023/1/31 に、DTM Works
ツールで、BIAS Calibration を実行した履歴を示しています。
⑤ 履歴リストを左クリックすると、「詳細」内容も切り替わります。

43 / 45

Model EA10S
5.7 機器アイコン設定
接続している機器のアイコンを変更することができます。
① 接続画面の○部を右クリックし、「機器アイコン設定」をクリックします。

機器アイコン設定 をクリックします。

② 「この機器のアイコンをファイルを指定してインストールする。」をクリックします。

③ 任意の画像を選択し、「開く」クリックします。これで機器のアイコンを任意の画像に切り替えることができます。

44 / 45

Bulletin No.MIJ-B6112D

http://www.motoyama-cp.co.jp
本 社 工 場

Main Office & Factory

TEL（022）344-4511（代表） / FAX（022）344-4522
E-mail : info@motoyama-cp.co.jp
5-2, Ohira Aza Kameoka, Ohira-mura, Kurokawa-gun, Miyagi, 981-3697, Japan
TEL +81-22-344-4511 / FAX +81-22-344-4522
E-mail : info-overseas@motoyama-cp.co.jp〒981-3697 宮城県黒川郡大衡村大衡字亀岡５－２●販売・サービスネットワーク

東

京

支

店

〒210-0007 神奈川県川崎市川崎区駅前本町 10-5 クリエ川崎 11F

TEL（044）381-8770（代表） FAX（044）381-8772

大

阪

支

店

〒550-0014 大阪市西区北堀江 1 丁目 12-19 クリモトビル 3F

TEL（06）6535-8111（代表） FAX（06）6535-8655

国

際

営

業

部

〒210-0007 神奈川県川崎市川崎区駅前本町 10-5 クリエ川崎 11F

TEL（044）381-8771

FAX（044）381-8773

札

幌

営

業

所

〒001-0912 札幌市北区新琴似12条7丁目1-47
リバティタウンP棟101号

TEL（011）766-1520

FAX（011）766-1521

東

北

営

業

所

〒981-3697 宮城県黒川郡大衡村大衡字亀岡 5-2

TEL（022）344-1761

FAX（022）344-1762

上

越

営

業

所

〒942-0036 新潟県上越市大字東中島2393番地

TEL（025）542-5151

FAX（025）542-5152

関

東

営

業

所

〒290-0046 千葉県市原市岩崎西1丁目5-19

TEL（0436）21-4400

FAX（0436）21-3540

静

岡

営

業

所

〒422-8033 静岡市駿河区登呂4丁目28-5

TEL（054）269-5377

FAX（054）237-5855

名 古 屋 営 業 所〒481-0012 愛知県北名古屋市久地野安田36番地

TEL（0568）26-6681

FAX（0568）26-6631

水〒481-0012 岡山県倉敷市二日市358

TEL（086）486-3381

FAX（086）486-3382

阪 神 サー ビ スセンター〒560-0894 大阪府豊中市勝部2丁目18-3

TEL（06）6854-7511

FAX（06）6854-7512

姫

路

出

張

所

〒672-8014 兵庫県姫路市東山276番地1

TEL（079）263-8640

FAX（079）246-2130

徳

山

営

業

所

〒745-0861 山口県周南市新地1丁目6-11

TEL（0834）21-5012

FAX（0834）31-0450

四

国

営

業

所

〒792-0896 愛媛県新居浜市阿島1丁目6-26

TEL（0897）47-8993

FAX（0897）47-8995

大

分

営

業

所

〒870-0108 大分市大字三佐字山ノ神980-1

TEL（097）527-3704

FAX（097）522-2352

TEL +86-411-6589-1277

FAX +86-411-6589-1278

大 衡 サービスセンター
上 越 サービスセンター
千 葉 サービスセンター

中 部 サービスセンター

島

営

業

所

周 南 サービスセンター
新 居 浜 サービスセンター
大 分 サービスセンター●海外関連会社本山阀门（大連）有限公司 〒116601 中国 遼寧省大連市金州新区港興大街 39 号 14-7
●海外販売代理店

中国、韓国、台湾、シンガポール、インドネシア、マレーシア、サウジアラビア●SALES AND SUPPORT NETWORK

Overseas Marketing & Sales Dept. : 11th Floor, Clie Kawasaki, 10-5, Ekimae-honcho, Kawasaki-ku, Kawasaki-city, Kanagawa, 210-0007 Japan
TEL: +81-44-381-8771
FAX : +81-44-381-8773
Domestic Sales Branches : Tokyo, Osaka, Sapporo, Tohoku, Joetsu, Kanto, Shizuoka, Nagoya, Hanshin, Tokuyama, Shikoku, Oita

●OVERSEAS AFFILIATED COMPANY

MOTOYAMA VALVE （DALIAN） CO., LTD
Gangxing Street 39-14-7, Jinzhou New District, Dalian-city, 116601 China

TEL: +86-411-6589-1277

FAX: +86-411-6589-1278●OVERSEAS NETWORK

China, Korea, Taiwan, Singapore, Indonesia, Malaysia, Saudi Arabia◆当社は保証期間を原則として製品納入後１年としています。
◆The product will be warranted for one year after delivering this product in principle.
■本取扱説明書の記載内容は、商品の改良等のため予告なく変更することがありますので予めご了承下さい。
■MOTOYAMA is continuously improving and upgrading its product design, specifications and/or dimensions. Information included herein issubject to change without notice.

■本取扱説明書は正しい情報の提供を目的としたものであり、本製品の市場性または適合性の保証を証明するものではありません。
■This instruction manual is supplied for information purpose only and should not be considered certified marketability and conformabilityof this product.

202402 MSP