H350R（B）-C
Instruction Manual Ver1.3

Suruga Seiki Co.,Ltd.
OST Department

Index*******************************************************************************************************************
1. Introduction ........................................................................ 3
1.1 Preface ........................................................................ 3
1.2 Components and Accessories ..................................................... 4
1.3 Introduction of Options............................................................ 6
1.4 Safety Instruction ................................................................ 7
1.5 Notification...................................................................... 8
2. System overview ................................................................... 9
2.1 Main feature .................................................................... 9
2.2 Components name and function .................................................. 10
2.3 Connection of Components configuration and instrument.............................. 11
3. Operation method.................................................................. 13
3.1 Measurement method ........................................................... 13
3.1.1 Installation of auto collimator .................................................. 13
3.1.2 Auto Collimator setting position ................................................ 13
3.1.3 Angle correction............................................................. 14
3.2 Measurement example .......................................................... 15
4. Others ........................................................................... 17
4.1 Trouble shooting................................................................ 17
4.1.1 Symptom and action......................................................... 17
4.1.2 The action again at appearing the ghost......................................... 18
4.2 Basic Specification .............................................................. 19
4.2.1 Specification................................................................ 19
4.2.2 H350R(B) - C050 ........................................................... 19
4.2.3 H350R(B) - C100 ........................................................... 19
4.2.4 H350R- C175 .............................................................. 20
4.2.5 Overveiw Size .............................................................. 20
4.3 Warranty and Service ........................................................... 22********************************************************************************************************************<Inquiry/Contact> ．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．
．Page 242

1. Introduction
1.1 Preface
Thank you for choosing SURUGA’S production in this time.
Please read this instruction manual and use this system correctly. And keep this manual carefully
and read again in case you need this manual.
Regarding peripherals equipments like image processing unit, please refer to manuals that attachedwith each system.
In case of missing or damaging this manual, please inquiry to us on referring <Inquiry/Contact>(refer to Page24)
Please be careful, configuration and accessories depend on the purchasing contract.

Caution
Unauthorized reproduction or duplication of this manual in part or in all is strictly prohibited.
Information in this manual is subject to change without prior notice.
This manual has been prepared carefully; however, please contact us if you have questions orsuggestions.
In case of paging disorder and missing page, we would like to replace it.

31.2 Components and Accessories
This production consists of bellow components and accessories. Please check them.

■Laser Auto Collimator

Configuration in case purchasing only system

Optical Head Main Body

1 Qty

AC Adaptor（6V 2A）・Image output cable (BNS-P)
Beam diameter switching plate*1Qty(Length 1.5m)
1Qty*）Switching the output beam diameter fromφ6 to φ1ｍｍ or φ2ｍｍis available.
In caseφ3 orφ4 is required, please purchase a variable aperture .
Instruction Manual

1Qty

Fig. AC Adaptor 1Qty (length 1.5m)

4Laser Auto Collimator Configuration in case of set purchasing
Set Configuration Table
Set name
Main body＋
LCD Monitor
Half set
IP set 500
IP set 1000
Full set 500
Full set 1000

MonitorMonitorbracket

Stand

Tilt

Image
Processing

Switching
Plate

H350R-C xxx/D

VCPM-5656P

－

－

－

－

H350-AP*1

VCPM-5656P－－
VCPM-5656W
VCPM-5656W

HA10-MB－－
HA10-MB
HA10-MB

HA10
－
－
HA10
HA10

HB10
－
－
HB10
HB10

－
HIP-500-P
HIP1000-P
HIP-500
HIP-1000

H350-AP
H350-AP
H350-AP
H350-AP
H350-APSet model*1

H350R-C xxx/D-H*1
H350R-C xxx/IP5*1
H350R-C xxx/IP10*1
H350R-C xxx/F5*1
H350R-C xxx/F10*1）xxx means measurement range (050 means for±0.5deg,100 means for ±1deg)
Configuration/Accessory
Optical head
Optical head main body

1 Qty

AC Adaptor (DC6V 850mA*1(DC12V 300mA*2), Length 1.5m)

1 Qty*1) for H350R(B)-C050&H350R(B)-C100*2) for H350R-C175
Beam diameter switching plate (Switching to φ1,φ2 available)

1 Qty

Accessory
LCD Monitor
LCD Monitor body

1 Qty

AC Adaptor for LCD Monitor (DC12V 1.25A)

1 Qty

）

1 Qty

Converting connector*）Model Suffix is P: RCA Plug-BNC Plug
Model Suffix is W: RCA Plug-BNC Jack
Monitor Bracket

1 Qty

HA10（In case half / Full set）

1 Qty

Stand
Screw M3-12（For installing stand-tilt）

4 Qty

2 Axis tilting stage
HB10

1 Qty
Screw M3-8 (For installing tilt head)

4 Qty

Conversion Plate
H350-AP

1 Qty

5

1.3 Introduction of Options
We line up following option and please purchase as your requirements.
Optical Components
Name

Angle
Unit: deg (Min)

Optical Parallel

0(0)

Wedge Substrate

1
0.5
0.25
0.2
0.1

(60)
(30)
(15)
(12)
(6)

Conversion Plate
Name
Conversion Plate

Tolerance
Unit::sec

Model

Remark+/-5HS-0φ30,t=10
One face is
AL coating+/-10

HS-100HS-050HS-025HS-020HS-01040*40
t=10Non Coating

Thickness
6mm

Beam Diameter Switching Plate
Name
Pinhole diameter
Beam Diameterφ3 andφ4ｍｍ
Switching Plate

Model
H350-AP

RemarkModel

Remark

H350-P

12-M3

Fig Conversion Plate

6

8-3.5

6 depth3.5

1.4 Safety Instruction
Please make sure that read following safety precaution before using.
Symbol means prohibit.

CautionPlease make sure that keep safety precautions in this manual. In case of negligence, it may cause
injury or damage physically.
*Precaution on safety and usage
This production corresponds to class2 laser equipment. This system should be installed and run
by the people who have knowledge of safety for laser equipments.
Don’t look in the beam directly or observe bean by optical methods.
Don’t touch to beam output by fingers or hit an object. Negligence may cause measurement
precise lower.
*Cable connection
AC Adaptor is attached with H350-C.This adaptor should connect before turning power on andoperation should be done under correct connection.
*Don’t use adaptor without attached AC adaptor.
*Environment
Avoid using this system in following environment- Many dust and powder dust (especially medal powder- Direct sunlight- Fire and heated place- Vibrating place- Splash water and oil- Inclining and stable place*Safekeeping/Storage
In case of no usage for long time or transferring this system,
make sure unplug from the outlet to prevent unexpected fireor electric shock.

7

Caution*Power Source
The standard attached AC adaptor is for only AC100 voltages therefore don’t connect AC
Adaptor to without AC 100voltage plug outlet. If you would like to run this system without AC100
voltage then please contact our sale person.
*Disassembly/Modification
Make sure that don’t disassemble, modify and irregular repair the system.
Negligence may cause the electric shock and dangerous.
If your find something wrong thenplease contact Our OST department sales Group.(0120-789-446 tall free)*Repair
In following case, unplug the power cable promptly
After that, please ask the repair for OST department sales Group
After that going on running the system, it may cause fire, electric shock and injury.
- In case of irregular situation like irregular sound, smell bad, smoking- In case of power code is injured- Drop the water on this system and foreign objects come into.
- Fall down this system or break the cabinet*Inquiry/Contact us refer to Page24

1.5 Notification*Semiconductor laser
The built-in semiconductor laser has own life.
If you find the output light become extremely weak then we recommend replacement of laser andplease inquiry and contact us.
Refer to 4.3 Warranty and Service (Page23)8

2. System overview
2.1 Main feature
Laser auto collimator is an angle displacement measuring equipment.
It outputs the parallel light that collimates built-in light source. If illuminating the light toreflection mirror that is located small distance from the collimated light, then in case of inclining
surface, returned image is misaligned. This distance that the image transfer is proportional tofocus distance of built-in lens and inclined angle eventually angle measurement is completed.
Main Feature* Detecting small angle displacement by the built-in CCD Camera and video output the input
angle displacement from BNC connector of the main body.
* Due to semiconductor laser, excellent for orientation and stability.
* Switching aperture mechanism allows being variable output beam diameter therefore it ispossible to select the beam diameter to measurement objects.
*LD power and CCD shutter speed is adjustable. Please refer to basic specification about
built-in light source wavelength and sight range.

92.2 Components name and function
LD adjustment Volume

Cable Connector

LD switch

Rotary Switch

Variable Aperture

Fig1-1 H350-C Main Body
*Cable Connector
Input power source and output image signal
Connect the attached cable*Rotary Switch
Switching CCD camera speed（1/100、1/250、1/500、1/1000、1/2000、1/5000、1/10000、1/100000 sec）
・LD volume
Adjust the output light (Maximum output is less than 1mW)
・Variable Aperture
Switching output beam diameter（φ1,2,3,4,6）・LD switch
Turn on/off the LD output (Built-in CCD power don’t shut down)

102.3 Connection of Components configuration and instrument
As requirement, it is possible to measure with connecting your stand, monitor etc5

NTSC (EIA)
Image process

Monitor
CRT Monitor CD Monitor

AC Adaptor

Variable ApertureStandard Set

Fig2．
．Components configuration and connection example

Stand
HA10

Install the rod on the M8 tap hole of stand base.

2Axis tilt stage
Fix the rod clamp attached with stand by attached screws to black knob side. Install the
auto collimator on the stage of the other side of black knob. Please install the auto
collimator by using attached screws and in this time the black knob should be located under
the auto collimator, the opening side for measurement.
Monitor
Connect the conversion connector to attached connector and RCA plug, yellow frommonitor.

11

Monitor bracket
Install the bracket that attached with monitor to the monitor.
Sandwich the lowlet of the monitor bracket as like bellow figure.
Install the monitor bracket on the rod and clamp it.

AC Adaptor
Connect AC adaptor to auto collimator and LCD monitor.
Please use the attached AC adaptor,DC6V and 2A.
Please use the attached AC adaptor,DC12V and 1.25Amp,to LCD monitor, red connector,

Others and accessaries
Regarding to instruction of the image processing units, please refer to each instruction
manual respectively

123. Operation method
3.1 Measurement method
Regarding to contents of instruction here, it is the method under using SURUGA optional 2axis tilt
stage (HB10), optical parallel and wedge substrate. In case of purchasing only main body then referto this method and set up the systems.

3.1.1 Installation of auto collimator
Install the autocollimator on 2axis tilt stage (HB10)

Fig3. Setup (example)

3.1.2 Auto Collimator setting position
The distance from laser output of main body to the optical parallel (HS-0) should be within
measurement distance.
Adjusting 2 axis tilt stage (HB-10) as like the laser light reflected on the optical parallel superimpose
with the center on the reference (center) from the monitor film or image processing equipment.
Refer to 4.2 Basic specifications (Page21) about measurement distance13

3.1.3 Angle correction
Put on the wedge substrate and illuminate the measurement light(In case of non-coating wedge substrate, put the optical parallel on the wedge substrate)
Against the measurement center (3.1.1 – 3.1.2) that set by using optical parallel, measurement
angle that the reflect light point indicates is the wedge substrate angle.
Please measure on this point as reference angel (correction value)
In case of purchasing the half set or main body/LCD monitor, the corrected LCD film is attached with
monitor and refers to this film as visual inspection.
In case of purchasing the full set or high rigidity set, please refer image processing instruction
manual.

Wedge substrate

Fig4. Wedge substrate setup

Situation
that
laser
diameter transfer by angle
of wedge substrate
Reference point

Cross line

Screen at fine adjustment

Screen at setting wedge substrate

Fig5. Measurement by CCD

14Circle display(Max angle within
measurement range)Line display

3.2 Measurement example
The following are some of applications on using this production. Please measure on referring them.
Mounting optical parts
For example in optical parts assembling process of
optical pick up. Inclining inspection at assembly
prism or mirror/lens is available to chassis.
。

HIP-500/HIP-1000 also increases measurement
effectiveness with its travel distance measuringfunction and pass-fall grading function.

Inclination adjustment of LD optical axis
Optical axis adjustment is available in LD
mounting process. By direct insertion of the lightfrom LD, measuring inserted light can beachieved. Power off the light from Auto collimatorand Auto collimator needs to be placed as baseline. During measuring internal light source needs to be off*When operation parallelism adjustment or intensity distribution checkbesides measuring inclination of LD optical axis, Please use “Proco” Profile andcollimation checker.
Assembling adjustment of Optical Pickup lensDuring a mounting and curing process, itevaluates inclination of pickup lens against
collimation-out from a chassis. Suruga’s auto
collimator can change a diameter of output beam
between Φ1 to 4 (Non slit case,Φ6) and canproperly illuminate with light volume adjustment function.
Inclination of Deflection test for Optical Disk or Spindle Motor
Checking the amount of linearity & deflection of optical diskor spindle motor. By using a featured HIP-1000’s “Orbital
measurement” function, the central point of rotation
deflection can be observed with real time measuring oforbital tracking.

15

HDD Head Inclination Check
Enable to measure inclination of HDD disk,
such as GMR head

Measuring Parallelism between CCD Device and Cover Glass
Measuring derivation between CCD reflected light andCover Glass reflected light. The label measurement
function of HIP-1000 enable numerical value display ofangle derivation (Parallelism)

Inclination & parallelism measurement of various items
Available for parallelism checking of various items, suchas Optical Device, Various metallic parts. By using
additional jig instrument, it can achieve measuring
rectangular angle of rectangular mirror. *Performance
depends on surface coarseness of metal surface

Adjustment of Factory Instrument
LAN ONLINE OBSERVE

Availabletomeasuretheinclination

andparallelism of surface table, parts etc.
Auto collimator supports increase of production
yield rate. It provides surveillance on the
production line and adjustment in assemblingprocess.

16

4. Others
4.1 Trouble shooting
4.1.1 Symptom and action
Symptom

No laser output

Cause
Unplug power code
Output power is weak

ActionPlug the power code properly
Adjust power volume

Laser’s life is completed

Contact our sales person

Refer page―
Page 10
Page 8Page 23

Set the distance that to measurement
The distance to measurement object within measurement distance forPage 20
object is too far
each production
No laser indication on the Due to measuring the angle
Measure the object within measurement
screen and monitor－
displacement over measuring
range.
range.
Page 8Laser‘s life
Contact our sales person
Page 23
Reflection
rateof Adjust by variable aperture or attach the
Difficult to read laser light－measurement object is low
mirror on the measurement object
Adjust the shutter speed
Light volume on
monitor is strong

the

CCD Sensitivity is highLaser power is too strong

Page 10

Refer to image
In case using image processing, adjust processing
the binarization level.
instruction
manualPage 10Adjust laser power level
Page 20
Adjust the shutter speed

Page 10Ghost appears
Refer to image
In case using image processing, adjust processing
the binarization level.
instruction
manualCCD Sensitivity is high17

4.1.2 The action again at appearing the ghost
In case of using internal light source to this system, the ghost might appear around origin point.
This ghost is the reflected light from surface of LD device and it is not defective.
To remove this ghost, please adjust following steps.
1. Reduce the laser power and complete adjustment that ghosts doesn’t appear.
2.Rise up shutter speed and complete adjustment that ghosts doesn’t appear.
3. In case of using image-processing unit, rise up binarization level and complete adjustmentthat ghosts doesn’t appear.

Measuring transfe distance

Click then finish

Fig6. The ghost on LD Cover Glass (on using HIP-1000)
Internal optical system is adjusted as refer to the internal light source of auto collimator.
In case that auto collimator is used against external input, ghost appearance might depends on
the laser deviation direction
If the ghost appears, change deviation direction of laser or adjust laser power.
And in case of adjusting on auto collimator side, adjust the CCD shutter speed.

184.2 Basic Specification
To improve this production, the specification or overviews are subject to change without prior notice.

4.2.1 Specification
Usage environment: 0 – 40 deg.C

20 - 80 %RH (non condense)

Storage environment: 0 – 40 deg.C

20 - 80 %RH (non condense)

Electric Power:: Input AC100 to 240 Voltage ± 10 % 50/60Hz: Output DC 6V, 850mA (DC12V, 300mA*)*) for H350R-C175 only

Light source: Red semiconductor laser: Blue perple semiconductor laser (Bule LD type)

Laser wavelength: 650 nm

Inside light source delivation: Strate delivation (Orientation is Vertical as base surface is refered)
Laser output: Less than 1.0 mW (Class2)

Aperture:φ1,2,3,4,6 mm (Variable)

Measuring Wavelength : 650+/-10 nm: 405+/-10 nm (Blue LD Type)4.2.2 H350R(B) - C050
Size (H350R): 40x50x80 mm (Only main body) refer to Page20(H350B): 40x50x85 mm (Only main body) refer to Page20

Weight:: 0.22 kg (Only main body)

Measurement distance: 0 to 200 mm (φ4,6) 0 to 300 mm（φ1,2,3）

Measurement range: +/-0.5 deg

4.2.3 H350R(B) - C100
Size (H350R): 40x50x80 mm (Only main body) refer to Page21(H350B): 40x50x85 mm (Only main body) refer to Page21

Weight:: 0.36 kg (Only main body)

Measurement distance: 0 to 240 mm

Measurement range: +/-1.0 deg

19

4.2.4 H350R- C175
Size

: 40x50x99 mm (Only main body) refer to Page21

Weight:: 0.28 kg (Only main body)

Measurement distance: 0 to 240 mm

Measurement range: +/-1.75 deg

4.2.5 Overveiw Size

H350-C050

20

Fig. H350-C100

Fig. H350-C175

21

4.3 Warranty and Service
Warramty
Please inform us the 8 digit serial number that marked on certification or rear side of production atinquirying/contacting us. We record the serial number and deliver date. The warranty period is one
year after deliver date.However,in case of following, out of warranty and the repair is chargerable.
-

Misusage, due to modification, repair by suruga’s engineer, mulfunction and damege-

Fall down the system during transportration or transfering.Or irregular operation
cause malfunction or damage.

-Fire, Brine Damage, Gas Damage, Abnormal voltage and Act of providence such as
earthquake, thunder, wind and flood damage and the others cause malfunction anddamage.

-Negligence for instruction manual or caution cause malfunction and damege.

Service
Please chek Page20 before repair request. If you have questions and concern then please contact
us, Sales Group of OST department<During under warranty>
In case that malfunction happen under normal usage following instruction and caution,the repair isfree. In case of above malfunction out of warraty,the repair is chargerable.
<Expired warranty period>
In case the performance is maintained by repair, we accept chargerable repair depending on
custorem’s request.

<Repair possible period>
The minumam period of spare parts (the parts to maintain the function) of this productoin is one year
after discontinueing this production. This period is “repair possible period” After this period somecase that repair might be possible therefore please contact us.
Notification: About the malfunction on this production, we don’t accept respponsibility without
warranty repair base on the certification.

22

23

For More Information Call to us:

SURUGA SEIKI CO., LTD.
Overseas Section, OST Division
3F KonanYKBldg,2,4,12,Konan,Minato-ku, Tokyo, 108-0075 Japan
TEL : +81-3-6711-5014
FAX : +81-3-6711-5021
URL : http://www.suruga-ost.com/
E-mail : e-ost@suruga-g.co.jp
OST-D3282-1

24