MANOSTAR DIGITAL SENSORNEWIncredible “MANOSTAR”

high quality
compact bodyclear display
30mm × 30mm

Equipping the capacitive sensor ofin-house development

Highly sensitive and precise
Correspondence to various micro diﬀerential pressure ranges

The body size: 30mm width and 30mm height
Capability of close adherence installation

Sophisticated design and visibility【External view 】

Main
display

30mm

One of the smallest micro diﬀerential
pressure sensors【Installation example 】

Subdisplay

30mm

High-quality white LCD (main display)
Alphanumeric indication in 12 segments

YAMAMOTO ELECTRIC WORKS CO.,LTD.JAPANSPECIFICATIONS
Product code

Pressure range code

ＱＤＰ３３ Ｎ １ Ｄ ２００

Pressure range

Unit

Analog output

Comparison output

Pressure range code

Pressure measuring Measuring diﬀerential pressuremethod（example）

Gas to be measured Air or non-corrosive gas (not liquid)

Max. value of scale(Indicate “±” for ± range)

Sensor method

Electrostatic capacitance type

ＤＥＰａ
ｋＰａ

Installation position On vertical wall only

１
４

4 to 20 mA
1 to 5 V

Ambient humidity

ＮＰ

NPN transistor
PNP transistor

Rated pressure range

Medium and ambient 0 to 60 ℃ (no freezing)
temperature
35 to 85 % RH (no dewing)Withstanding pressure 10kPaof instrument bodyWithstanding pressure 10kPaof receiving element

Main display 12 segments LCD (White or Red) 4 digits
Sub display 12 segments LCD (Orange) 4 digits
Accuracy± 1.0 % FS ± 1 digit (at 23 ℃)
Temperature characteristic ± 0.15 % FS/℃ (zero + span)

Display

LCD display

D 500 ～

50Pa

0. 0～

50. 0

D 1000 ～100Pa0 .0～

100. 0Insulation resistance Between each terminal and case : 10 MΩ or more (500 V DC megger)

D 2000 ～200

Pa0～200

D 3000 ～300

Pa0～300
500

Setting method
Output display
Comparison output
Output type

D 5000 ～

500

Pa0～

D 10000 ～1000

Pa0～ 1000

E 10 ～

1kPa

0 .00 ～

1 .00

E 20 ～

2kPa

0 .00 ～

2 . 00

D +-50- 50 ～+50

Pa- 50 . 0～

D +-100- 100 ～

+100

Pa

- 100～

D +-200- 200 ～

+200

Pa

- 200

D +-300- 300 ～

+300

Pa

- 300

D +-500- 500 ～

+500

Pa

- 1000 ～ +1000

Pa

D +-1000

Push type of digital setting
Red LCD×2
NPN transistor open collector : Two channels
PNP transistor open collector : Two channels

Analog output

Accuracy ± 1.0 % FS (at 23℃ )Temperature characteristic ± 0.15 % FS/℃ (zero + span)

50 . 0Pressure inlet

M5 internal thread
Product with metal barb connector

100Mass

30g approx.

～

200

Connector

RITS Connector 6P (TE Connectivity)～

300

Power supply voltage 12 to 24 V DC ± 10 % (with a ripple 10 % or less)

- 500

～500

- 1000～ 1000Connector polarity Pressure side mark indicated on the pressure inletthat high side is H and low side is L.

Applicable piping : I.D.4

Protection degree

Standard IEC60529 IP-code IP 40

2 .00

Zero adjustment

Push type auto zero
(both display scale and analog output adjust zero point at the same time)0 . 00 ～10.00

Material of the outer case PBT and polyamidePa0 . 00 ～

25.00

+10Pa

- 10 . 00 ～10.00

Conformed standards RoHS Directive, UL Standards

+25

Pa

- 25 . 0

25. 0

E +-1-1 ～+1kPa- 1. 00 ～1.00

E +-2-2 ～+2kPa- 2 . 00 ～＊ D 100 ～10Pa＊ D 250 ～25＊ D +-10- 10 ～＊ D +-25- 25 ～～Accessories

Nil(wiring side connector and attaching parts are sold separately)＊ Expected to release in 2020.

OUTLINE※Refer to instruction manual for other sizes.

322930RITS Connector 6P65432 130

φ4.7

PaH

L

MODE

High-pressure side
piping connector

(1.5)

12

Low-pressure side
piping connector■The speciﬁcations and description of the product explained in this leaﬂet may be subject to change without prior notice because of modiﬁcation and the like.
■The contents are as of September 2019.

Contact UsTEL078-621-7000

http://www.manostar.co.jp/

eigyou@manostar.co.jp

Sole Agent
1-2-3,Nishi-shiriike-cho,Nagata-ku,Kobe,Hyogo 653-0031 JAPAN
TEL.+81(78)-621-7000 FAX.+81(78)-621-7788

QDP33-SEP.2019-E1

Our website.

E-MAIL

Manufacture
1-2-3,Nishi-shiriike-cho,Nagata-ku,Kobe,Hyogo 653-0031 JAPAN
TEL.+81(78)-631-6000 FAX.+81(78)-631-6020

YAMAMOTO ELECTRIC WORKS CO.,LTD.JAPAN