Medical TechnologyAnalysis

Mass Spectrometry

SmartVane
Revolutionary vacuum pump formass spectrometers withhermetically sealed pump housingSmartVane
Revolutionary vacuum pump formass spectrometers withhermetically sealed pump housing

No oil leakage

Your added value

Low noise level

Hermetically sealed pump housing
The SmartVane is the first rotary vane pump for mass
spectrometry with a hermetically sealed pump housing. By
omitting shaft seals, it eliminates the main cause of oilleakage. As a result, the SmartVane combines the advantagesof the proven rotary vane pump principle with a revolutionary
sealed design. This makes it ideal as a backing pump for mass
spectrometers in the fields of environmental and food analysis
as well as pharmaceutical and clinical analysis.
Ideal for laboratory use
With its low noise level, the SmartVane ensures optimal
conditions in the laboratory. At its typical operating pressureof <10 hPA, it is quieter than other pumps in this application
area. This means that even challenging tasks can becompleted in a pleasant working environment. Anotheradvanatge point is that the SmartVane‘s compact design
makes it easy to install in existing systems. As a plug andplay solution, it can be easily connected to already installed
equipment. Thanks to intelligent communication options,
the SmartVane conveniently corresponds with your system.

l3

Lower power consumption

Compact designIntelligent
communication options

Long maintenanceintervals

Environmentally friendly and low maintenance
The SmartVane incorporates an energy-efficient IPM motorwith standby function. Its low power consumption
automatically lowers operating costs and reduces the carbon
footprint. The integrated motor also enables the special
design that makes a conventional seal with shaft seals
obsolete. As a result, you benefit from maintenance intervalsof up to two years.
Applications■ ICP-MS■ LC/MS■ Backing pump for turbopumps

SmartVane
Revolutionary vacuum pump formass spectrometers withhermetically sealed pump housing

Pump stage

Fan

Inlet

FeedthroughMotorCoupling

Conventional pump design, before SmartVane

Pump stage

Inlet
Motor with encapsulatedwindings

VacuumOil

DC fan with own motor

Pump design SmartVane

Motor rotor

l5

We drive sustainablesolutions
Environmentally friendly
By using novel IPM1) synchronous motors with sensorless
INFORM® control2), our SmartVane pumps achieveenvironmentally friendly efficiency. Reducing power
requirements while maintaining high pumping capacity resultsin lower operating costs for the operator while the motor
achieves premium efficiency and significantly exceeds theminimum values of efficiency level IE3 as applicable tostandard asynchronous motors. Everyone benefits from this.
In addition, the use of the new IPM motor technology resultsin significantly smaller product dimensions compared toproducts using asynchronous motors.
Conserving resources
However, the pump‘s performance and installation dimensions
are not the only environmental benefits. Our SmartVane
already scores points in production through conscious
material savings. For example, less copper is used andadvanced technologies eliminate the need for additional
sensors. The pump housing, like the hood and the housing of
the electronics, is manufactured by casting, which avoids
excessive production waste.
Sustainable
Last but not least, man and machine benefit equally fromsustainable use of state-of-the-art technologies: The newgeneration rotary vane pumps emit less waste heat and also
have a longer service life thanks to innovative standby
operation. They run extremely quietly and with low vibration,
making their working environment safe and comfortable.
1)
2)

Interior Permanent Magnet
INFORM® process according to Prof. Schrödl

SmartVane
Technical data, dimensions,order numbers, accessories

Pumping speed m3/h

Pumping speed [m3/h]

Pumping speed

70

60

50

4030

20

1000.01
SmartVane 55
SmartVane 70

0.1

1

10100

1000

Inlet
pressure
Intakepressure in[hPa]mbar

Dimensions

470 (SmartVane 70)
415
361

262
167
11.5

95

177
144

68.5

IN

148260

OUT

235

309

70

9160312Dimensions in mml7

Technical data
SmartVane 55
Nominal pumping speed
Ultimate pressure without gas ballast
Input voltage
Motor protection
Motor typeI/O interfaces
Leakage rate of check valve
Dimensions (L x W x H)
Rated power at maximum speed
Protection class
Certifications
Cooling type
Ambient temperatureContinuous gas inlet temperature, max.
Connection flange (inlet)
Connection flange (outlet)
Continuous inlet pressure, max.
Exhaust pressure, min./max.
WeightInput voltage: tolerance (%)
Mains cable included
Mains cable
Gas ballast
Operating medium
Operating medium quantity
Operating height, max.
Speed

SmartVane 70

31 – 50 m³/h

31 – 70 m³/h
2 · 10-1 hPa
100 – 127 V / 200 – 240 V AC (±10), 50/60 Hz
Internal
1-ph motor
RS-485; I/O port
1 · 10-2 Pa m3/s
415 x 312 x 308 mm
470 x 312 x 308 mm
0.75 kW
1 kW
IP20
CE, cTUVus
Air
15 – 45 °C
40 °C
DN 40 ISO-KF
DN 25 ISO-KF / G1/2
15 hPaAtmospheric pressure
36 kg
37 kg
10 %
No
No, C16 connector in terminal box
No
H1 / F4
1.75 l4,000 m
800 – 1350 rpm
800 – 1800 rpm

Order numbers
With operating medium H1With equipment F4

SmartVane 55

SmartVane 70

PK D80 000
PK D80 001

PK D81 000
PK D81 001

Accessories
Accessories
FAK 040, activated carbon filter
SAS 40, dust separator
Mains cable 230 V AC, CEE 7/7 to C15, 3 m
Mains cable 110 V AC, NEMA 5-15 to C15, 3 m
Y-distributor M12 for RS-485
Adapter RS-485 to M12
OmniControl 001 mobile, control units
OmniControl 001 mobile, rack unit without integrated power supply unit
Interface cable RJ 45 to M12
USB RS-485 converter
Interface cable, M12 m straight / M12 m straight, 3 m
Operating fluid
H1, 1 lF4, perflour polyether, 0.25 l
F4, perflour polyether, 0.5 l
F4, Perflour polyether, 1 lOrder number
PK Z30 008
PK Z60 510
PK 050 343
PK 050 344
P 4723 010
PE 100 150-X
PE D20 000 0
PE D40 000 0
PM 051 726-T
PM 061 207-T
PM 061 283-T
Order number
PK 001 210-T
PK 005 885-T
PK 005 886-T
PK 005 887-T

Further accessories can be found on our website at www.pfeiffer-vacuum.com.

SmartVane

4.000 mOperating altitude

100 - 127 V /200 - 240 V
Input voltage

Pfeiffer Vacuum GmbHGermany
T +49 6441 802-0

Errors excepted. All data subject to change without prior notice. PD 0065 PEN (November 2022/PoD)Are you looking for an
optimal vacuum solution?
Please contact us: