June 3, 2013
Ver1.30

USB2.0 CCD CAMERAARTCAM-WOM Series
INSTRUCTION BOOKLET
0.3M pixels

ARTCAM-424KY-WOM Series

1.25M pixels

ARTCAM-445KY2-WOM Series

1.5 M pixels

ARTCAM-150P5-WOM Series

1.5 M pixels

ARTCAM-285CX-WOM Series

2.0 M pixels

ARTCAM-274KY-WOM Series

5.0 M pixels

ARTCAM-500P2-WOM Series

5.0 M pixels

ARTCAM-655KY-WOM Series

ARTRAY CO., LTD.

Contents
1. Attention………………………………….3
2. Introduction………………………………6
3. Main Features……………………………6
4. The Product………………………………7
5. Connect the Camera…………………….8
6. Specifications……………………………..97. External Dimensions…………………….13

8. Spectral Sensitivity Characteristics……...19
9. AD Conversion…………. ……. ……. …..26
10. Optional Functions………………………27
11. Recommended PC……………………..29

2

1. Attention■About this manual
1. Before using the camera, please read this manual thoroughly.
2. Please keep this manual reachable and always refer to the contents when needed.
3. Please contact us if the manual is lost or damaged. We will provide an replacement.
4. We cannot guarantee the safety of improper uses of the camera.
5. For your safety, please follow the directions of this manual.
6. All contents are subject to change.
7. Images in this manual may have been simplified to be easier comprehended.
8. Please contact us if you find any unclear points or mistakes in this manual.
9. Quoting, copying or altering some or all parts of the manual without our permission is prohibited.
10. We are not responsible for any lost or damages on your profits due to the use of our products.
11. Please understand that our oversea branches do not provide maintenance or repair services.
■About the Icons
To keep the safety of the user, other people and their properties, please pay attention to the following icons.

Warning
If the user fails to follow the instruction, physical injury on human or damages on hardware may occur.

CautionIf the user fails to follow the instruction, serious injury or death may occur.

3

■For Safe Use

Warning●In following circumstances, please stop using the product and turn off the power immediately to prevent the risks offires and electric shocks. If the product is defective, please contact us for repair or exchange. For your safety, please do
not disassemble, modify or repair the camera on your own.
Please stop using the product and turn off the power immediately when:
・The product becomes smoky or gets extremely hot on the surface, or makes unusual smells or sounds.
・Foreign material or water gets into the product.
・The product falls and becomes damaged.

●Do not place the product on unstable surfaces. The product may be fallen and people may get hurt.

Caution●Do not expose the product to steam or fumes to avoid electric shocks and fires.
●Do not leave the product in high temperature places such as inside of vehicles or under direct sunlight.
High temperature may cause damages to the camera, or even cause fires.
●Do not cover the product with cloth or other materials. The product may get extremely hot and the heat may cause
deformations on the parts or even cause fires.
●Please avoid dropping or shocking the product as the product may be damaged.
●Do not touch the cable with a wet hand. Such action may cause electric shocks.
●Please avoid continuously contacting the surface of the camera to your skin when the camera is being used. The
surface temperature of the camera may cause burns.
■Other Notices●Please do not use the camera under strong lights such as sun light for a long period. Also please do not expose thecamera under strong lights even when the product is not being used because the sensor might be damaged.
Maintenance● Wipe the dirt on surface with soft cloth or tissue paper. Do not use alcohol, thinner or benzene to avoid damaging
the surface paints.
Electro Magnetic Interference●The camera may interference with electronic devices such as TV and radio. Please do not place the camera next tosuch equipment.
4

■Export Control
The Government of Japan controls military sensitive goods and technologies including relevant dual-use goods andtechnologies.
This product is a Catch-all Control item subject to the Foreign Exchange and Foreign Trade Act and its relevant
legislations. Except for exporting to the 27 white countries designated in the Cabinet Order, export licenses are required
if the products are going to be used for military use or if the end user of the product is related to all kinds of militaryactivities. If your circumstances cause the need to apply export licenses, please notify us before you place orders. Also,
please notify us in advance if the end users or purposes of use change after the purchase and thus cause the need toapply export licenses.

*The 27 white countries not under the UNSC Arms Embargo:
Argentina, Australia, Austria, Belgium, Bulgaria, Canada, Czech Republic, Denmark, Finland, France, Germany,
Greece, Hungary, Ireland, Italy, Republic of Korea, Luxemburg, Netherlands, New Zealand, Norway, Poland, Portugal,
Spain, Sweden, Switzerland, United Kingdom, and United States
About the Japanese Security Export Controls, please refer to the webpage for Security Export Control Policy, The
Ministry of Economy, Trade and Industry:
http://www.meti.go.jp/policy/anpo/englishpage.html■Guarantee
To save the environment, we do not issue warranty in printed format. Instead, all records of the warranty periods,
delivery dates and the customer information are well kept in our system.
For more details, please refer to the sites below:
Hardware Warranty : http://www.artray.us/download/artray_warranty.pdf

5

2. Introduction
ARTCAM-WOM Series is CCD cameras for industrial use.
This series adopt USB2.0 interface, so the data can be delivered to PC directly without a capture board.
Cameras are equipped with on board memory which avoid data missing during transfer.
We provide a variety including cased cameras, board cameras which can be easily incorporated to other equipment,
and so on.

3. Main Features
・High Resolution, High Frame Rate
We provide high speed CCD cameras of a wide range of resolutions from 0.3 mega pixels to 5 mega pixels.
・USB2.0 Interface
Our cameras adopt USB2.0 interface, so the data can be delivered to PC directly and no capture board is needed.
・Small and Light
Our cameras are small and light, thus can be easily installed according to your desires.
Also, the board type camera with small type board lens is easy to be incorporated with other machines.
・Varieties to Your Needs
We provide a variety including cased cameras, board type cameras, C-Mount board type cameras.
Also, we provide color and monochrome models to meet customers’ needs.
・Bounded Viewer Software
We provide the Viewer Software - ART-Viewer (Windows XP/Vista/7).
With this software, users can easily preview, save images and change camera settings.
・2D Measurement & Filing Software(Optional)
Useful 2D measuring functions are available with the optional 2D Measuring & Filing Software - ART-MEASURE.
・Recording Software(Optional)
The optional recording software - ART-REC allows the user to save movies in AVI format.
・Software Development Kit(Optional)
For customer who prefers to create their own image processing software or use our camera with other applications, we
provide SDK upon requests.

6

4. The Product1) Camera2) USB Cable
Type A-B connector cable, 3m approx.
*1 A USB cable with auxiliary power supply is attached for ARTCAM-424KY-WOM/ARTCAM-655KY-WOM series
and customized model of external power supply is also available.
*2 For open frame type and model without USB connector, a Type A- connector of 3m will be attached.
3) Introduction Booklet
4) Viewer software/Device Driver CD<Options>
1) C Mount lens2) 2 Measurement & Filing Software [ART-MEASURE]
3) Recording Software [ART-REC]
4) Software Development Kit [ARTCCM-SDK]5) 5m extension cable with USB2.0 repeater

75. Connect the Camera
5-1 How to connect a camera

5-2 Shutter Speed
Shutter speed can be changed on the standard viewer software ART-Viewer.
The shutter speed is set directly on the sensor register, so please refer to the following formulas for the real exposure
time.
Model
424KY-WOM Series

MethodRange : 1~525(Exposure Time : 1/96332~1/100 sec)
Exposure Time=((525 – Shutter Value)×780)+425)×0.0244 [μsec]*at 40.94MHz

445KY2-WOM Series

Range : 2~1003(Exposure Time : 1/94488~1/22.47 sec)
Exposure Time=(((1003 – Shutter Value)×1600)+381)×0.0278 [μsec]

150P5-WOM Series

Range : 7~1074(Exposure Time : 1/58824~1/12.5 sec)
Exposure Time= ((1074 – Shutter Value)×1790+424)×0.042 [μsec]

285CX-WOM Series

Range : 3~1070(Exposure Time : 1/39152~1/12.6 sec)
Exposure Time= ((1070 – Shutter Value)×1790+613)×0.042 [μsec]*at 36MHz*at 24MHz*at 24MHz
274KY-WOM Series

Range : 2~1252(Exposure Time : 1/47493~1/15 sec)
Exposure Time= (((1252 – Shutter Value)×1920)+758)×0.0278 [μsec]*at 36MHz

500P2-WOM Series

Range : 1~1014(Exposure Time : 1/8293~1/4 sec)
Exposure Time=(Shutter Value×2894)×0.042 [μsec]*Only when the shutter speed is set as 1014,
exposure time will become 2028×2894×0.042 [μsec]*at 24MHz

655KY-WOM Series

Range : 2~2078(Exposure Time : 1/12531~1/6 sec)
Exposure Time=(2079 – Shutter Value) ×3192×0.025 [μsec]*at 40MHz

5-3 Operation clock settings
5-3-1 ARTCAM-150P5-WOM series
Users can switch the operation clock between 24MHz / 12MHz with the standard viewer software or SDK.
*40MHz may also be supported. For further details, please contact our sales representatives.
5-3-2 ARTCAM-285CX-WOM series
Users can switch the operation clock between 40MHz / 24MHz / 12MHz with the standard viewer software or SDK.

8

6. Specifications
6-1 Color Model
ARTCAM-

ARTCAM-

ARTCAM-

424KY-WOM

445KY2-WOM

150P5-WOM

424KY-WOM-OP

445KY2-WOM-OP

150P5-WOM-OP

424KY-WOM-OP-CM

445KY2-WOM-OP-CM

150P5-WOM-OP-CM

Sensor Type
Total Number

ARTCAM285CX-WOM
285CX-WOM-OP-CM

CCD Color Image Sensor
692(H)×504(V)1348(H)×976(V)1434(H)×1050(V)1434(H)×1050(V)

659(H)×494(V)

1296(H)×966(V)1392(H)×1040(V)1392(H)×1040(V)Resolution

656(H)×494(V)

1280(H)×960(V)

1360(H)×1024(V)

1360(H)×1024(V)Active Imager

4.88(H)×3.66(V)mm

4.86(H)×3.62(V)mm

6.47(H)×4.83(V)mm

8.98(H)×6.71(V)mm

Size

1/3”

1/3”

1/2”

2/3”

7.4(H)×7.4(V)μm

3.75(H)×3.75(V)μm

4.65(H)×4.65(V)μm

6.45(H)×6.45(V)μm

Of Pixels
Number Of
Effective Pixels

Scan Mode
Pixel Size

Progressive Scan

Shutter Type

Global Shutter

Color Filter

RGB Bayer Pattern

Array
Frame Rate

100fps

22fps

12fps

12.6fps

Shutter Speed

1/96332~1/100

1/94988~1/22.47s

1/58824~1/12.5s

1/39152~1/12.6s

InterfaceUSB2.0 Bulk Transfer

Synchronization

Internal Synchronization

SystemLens Mount

C Mount(424KY-WOM、424KY-WOM-OP-CM)(445KY2-WOM、445KY2-WOM-OP-CM)

Board lens Mount M12 P0.5

Board lens Mount M12(424KY-WOM-OP)

P0.5(445KY2-WOM-OP)

Power
Power
Consumption

Under Approx.3W

Ambient

External

Weight(150P5-WOM、

C Mount

150P5-WOM-OP-CM)(285CX-WOM、Board lens Mount M12 P0.5

285CX-WOM-OP-CM)(150P5-WOM-OP)

DC5V(Supplied by USB port)
Under Approx.2W

Operating Temperature/Humidity：0~35°C/10~80%(Non condensing)

Conditions

Dimensions

C Mount

C Mount

50(W)×47(H)×49.7(D)mm

50(W)×47(H)×59.6(D)mm

50(W)×47(H)×49.7(D)mm(424KY-WOM)(445KY2-WOM)

43.5(W)×43.5(H)×41.4(D)mm

43.5(W)×43.5(H)×33.3(D)mm(424KY-WOM-OP)(445KY2-WOM-OP)

50(W)×47(H)×43.8(D)mm

50(W)×47(H)×36.7(D)mm(424KY-WOM-OP-CM)(445KY2-WOM-OP-CM)*Lens and tripod not included*Lens and tripod not included

Approx.150g(424KY-WOM)

Approx.120g(445KY2-WOM)

Approx.120g(150P5-WOM)

Approx.70g(424KY-WOM-OP)

Approx.50g(445KY2-WOM-OP)

Approx.50g(150P5-WOM-OP)

Approx.110g(4424Y-WOM-OP-CM)

Approx.80g(445KY2-WOM-OP-CM)

Approx.80g(150P5-WOM-OP-CM)*Lens, tripod and cable not*Lens, tripod and cable not*Lens, tripod and cable notincludedincludedincluded

9(150P5-WOM)

50(W)×47(H)×49.7(D)mm

43.5(W)×43.5(H)×33.3(D)mm(285CX-WOM)(150P5-WOM-OP)

50(W)×47(H)×36.7(D)mm

50(W)×47(H)×36.7(D)mm(285CX-WOM-OP-CM)(150P5-WOM-OP-CM)*Lens and tripod not*Lens and tripod notincludedincludedApprox.120g(285CX-WOM)
Approx.80g(285CX-WOM-OP-CM)*Lens, tripod and cable notincluded6-2 Color Model
ARTCAM274KY-WOM
274KY-WOM-OP
274KY-WOM-OP-CM
Sensor TypeARTCAM-

ARTCAM-

500PⅡ-WOM

655KY-WOM

500PⅡ-WOM-OP-CM

655KY-WOM-OP-CM

CCD Color Image Sensor

Total Number Of Pixels

1688(H)×1248(V)2658(H)×1970(V)2536(H)×2068(V)

Number Of Effective Pixels

1628(H)×1236(V)

2588(H)×1960(V)2456(H)×2058(V)Resolution

1600(H)×1200(V)

2560(H)×1920(V)2456(H)×2058(V)

Active Imager

7.16(H)×5.44(V)mm

8.80(H)×6.66(V)mm

8.47(H)×7.10(V)mm

Size

1/1.8”

2/3”

2/3”

Scan Mode

Progressive Scan

Interlace

Progressive Scan

Pixel Size

4.4(H)×4.4(V)μm

3.4(H)×3.4(V)μm

3.45(H)×3.45(V)μm

Shutter Type

Global Shutter

Color Filter Array

RGB Bayer Pattern

Frame Rate

15fps

4fps

6.1fps

Shutter Speed

1/47493~1/15s

1/8293~1/4s

1/12531~1/6s

InterfaceUSB2.0 Bulk Transfer

Synchronization System

Internal Synchronization
C Mount

Lens Mount(274KY-WOM、274KY-WOM-OP-CM)

C Mount

Board lens Mount M12 P0.5

(500PⅡ-WOM、500PⅡ-WOM-OP-CM)

C Mount(655KY-WOM、655KY-WOM-OP-CM)(274KY-WOM-OP)
Power

DC5V(Supplied by USB port)

Power Consumption

Under Approx.2W

Ambient Conditions

Under Approx.3W

Operating Temperature/Humidity：0~35°C/10~80%(Non condensing)50(W)×47(H)×49.7(D)mm

External Dimensions(274KY-WOM)

50(W)×47(H)×49.7(D)mm

50(W)×47(H)×59.6(D)mm

43.5(W)×43.5(H)×33.3(D)mm(500PⅡ-WOM)(655KY-WOM)(274KY-WOM-OP)

50(W)×47(H)×36.7(D)mm

50(W)×47(H)×43.8(D)mm

50(W)×47(H)×36.7(D)mm(500PⅡ-WOM-OP-CM)(655KY-WOM-OP-CM)(274KY-WOM-OP-CM)*Lens and tripod not included*Lens and tripod not included*Lens and tripod not included
Approx.120g(274KY-WOM)
WeightApprox.120g(500PⅡ-WOM)

Approx.150g(655KY-WOM)

Approx.80g(500PⅡ-WOM-OP-CM)

Approx.110g(655KY-WOM-OP-CM)*Lens, tripod and cable not included*Lens, tripod and cable not included

Approx.50g(274KY-WOM-OP)
Approx.80g(274KY-WOM-OP-CM)*Lens, tripod and cable not included

106-3 Monochrome Model
ARTCAM-

ARTCAM-

ARTCAM-

424KY-BW-WOM

445KY2-BW-WOM

150P5-BW-WOM

424KY-BW-WOM-OP

445KY2-BW-WOM-OP

150P5-BW-WOM-OP

424KY-BW-WOM-OP-CM

445KY2-BW-WOM-OP-CM

150P5-BW-WOM-OP-CM

Sensor Type
Total Number

ARTCAM285CX-BW-WOM
285CX-BW-WOM-OP-CM

CCD Monochrome Image Sensor
692(H)×504(V)1348(H)×976(V)1434(H)×1050(V)1434(H)×1050(V)

659(H)×494(V)

1296(H)×966(V)1392(H)×1040(V)1392(H)×1040(V)Resolution

656(H)×494(V)

1280(H)×960(V)

1360(H)×1024(V)

1360(H)×1024(V)Active Imager

4.88(H)×3.66(V)mm

4.86(H)×3.62(V)mm

6.47(H)×4.83(V)mm

8.98(H)×6.71(V)mm

Size

1/3”

1/3”

1/2”

2/3”

4.65(H)×4.65(V)μm

6.45(H)×6.45(V)μm

Of Pixels
Number Of
Effective
PixelsScan Mode
Pixel Size

Progressive Scan
7.4(H)×7.4(V)μm

3.75(H)×3.75(V)μm

100fps

22fps

12fps

12.6fps

1/96332~1/100s

1/94988~1/22.47s

1/58824~1/12.5s

1/39152~1/12.6s

Shutter Type
Frame Rate
Shutter
Speed

Global ShutterInterfaceUSB2.0 Bulk Transfer

Synchronizati

Internal Synchronization

on System

Lens Mount

C Mount

C Mount

C Mount(424KY-BW-WOM、(445KY2-BW-WOM、(150P5-BW-WOM、

C Mount

424KY-BW-WOM-OP-CM)

445KY2-BW-WOM-OP-CM)

150P5-BW-WOM-OP-CM)(285CX-BW-WOM、Board lens Mount M12

Board lens Mount M12

Board lens Mount M12 P0.5

285CX-BW-WOM-OP-CM)

P0.5(424KY-BW-WOM-OP)

P0.5(445KY2-BW-WOM-OP)(150P5-BW-WOM-OP)

Power
Power
Consumption

DC5V(Supplied by USB port)
Under Approx.3W

Under Approx.2W

Ambient
Operating Temperature/Humidity：0~35°C/10~80%(Non condensing)

Conditions

External
Dimensions

Weight50(W)×47(H)×59.6(D)mm

50(W)×47(H)×49.7(D)mm

50(W)×47(H)×49.7(D)mm(424KY-BW-WOM)(445KY2-BW-WOM)(150P5-BW-WOM)

43.5(W)×43.5(H)×41.4(D)mm

43.5(W)×43.5(H)×33.3(D)mm

43.5(W)×43.5(H)×33.3(D)mm(285CX-BW-WOM)(424KY-BW-WOM-OP)(445KY2-BW-WOM-OP)(150P5-BW-WOM-OP)

50(W)×47(H)×36.7(D)mm

50(W)×47(H)×43.8(D)mm

50(W)×47(H)×36.7(D)mm

50(W)×47(H)×36.7(D)mm(285CX-BW-WOM-OP-CM)*Lens and tripod not included(424KY-BW-WOM-OP-CM)(445KY2-BW-WOM-OP-CM)(150P5-BW-WOM-OP-CM)*Lens and tripod not included*Lens and tripod not included*Lens and tripod not included

Approx.150g(424KY-BW-WOM)

Approx.120g(445KY2-BW-WOM)

Approx.120g(150P5-BW-WOM)

Approx.70g(424KY-BW-WOM-OP)

Approx.50g(445KY2-BW-WOM-OP)

Approx.50g(150P5-BW-WOM-OP)

Approx.110g(424KY-BW-WOM-OP-CM)

Approx.80g(445KY2-BW-WOM-OP-CM)

Approx.80g(150P5-BW-WOM-OP-CM)*Lens, tripod and cable not included*Lens, tripod and cable not included*Lens, tripod and cable not included

1150(W)×47(H)×49.7(D)mm

Approx.120g(285CX-BW-WOM)
Approx.80g(285CX-BW-WOM-OP-CM)*Lens, tripod and cable not included6-4 Monochrome Model
ARTCAM-

ARTCAM-

274KY-BW-WOM

655KY-BW-WOM

274KY-BW-WOM-OP

655KY-BW-WOM-OP-CM

274KY-BW-WOM-OP-CM
Sensor TypeCCD Monochrome Image Sensor

Total Number Of Pixels

1688(H)×1248(V)

2536(H)×2068(V)

Number Of Effective Pixels

1628(H)×1236(V)

2456(H)×2058(V)Resolution

1600(H)×1200(V)2456(H)×2058(V)

Active Imager

7.16(H)×5.44(V)mm

8.47(H)×7.10(V)mm

Size

1/1.8”

2/3”

Scan Mode
Pixel Size

Progressive Scan
4.4(H)×4.4(V)μm

3.45(H)×3.45(V)μm

Shutter Type

Global Shutter

Frame Rate

15fps

6.1fps

Shutter Speed

1/47493~1/15s

1/12531~1/6s

InterfaceUSB2.0 Bulk Transfer

Synchronization System

Internal Synchronization
C Mount

Lens Mount(274KY-BW-WOM、274KY-BW-WOM-OP-CM)

C Mount

Board lens Mount M12 P0.5

(655KY-BW-WOM、655KY-BW-WOM-OP-CM)(274KY-BW-WOM-OP)
Power
Power Consumption
Ambient Conditions

DC5V(Supplied by USB port)
Under Approx.2W

Under Approx.3W

Operating Temperature/Humidity：0~35°C/10~80%(Non condensing)50(W)×47(H)×49.7(D)mm(274KY-BW-WOM)

External Dimensions

50(W)×47(H)×59.6(D)mm

43.5(W)×43.5(H)×33.3(D)mm(655KY-BW-WOM)(274KY-BW-WOM-OP)

50(W)×47(H)×43.8(D)mm

50(W)×47(H)×36.7(D)mm(655KY-BW-WOM-OP-CM)(274KY-BW-WOM-OP-CM)*Lens and tripod not included*Lens and tripod not included
Approx.120g(274KY-BW-WOM)
WeightApprox.150g(655KY-BW-WOM)

Approx.50g(274KY-BW-WOM-OP)
Approx.80g(274KY-BW-WOM-OP-CM)*Lens, tripod and cable not included

12

Approx.110g(655KY-BW-WOM-OP-CM)*Lens, tripod and cable not included

7. External Dimensions
7-1 C-MOUNT, CASED, USB CABLE TYPEARTCAM-445KY2-WOM / 445KY2-BW-WOM
ARTCAM-150P5-WOM / 150P5-BW-WOM
ARTCAM-285CX-WOM / 285CX-BW-WOM
ARTCAM-274KY-WOM / 274KY-BW-WOM
ARTCAM-500PⅡ-WOMC Mount

USB B Connector

Drawing

of

the

base

without a tripod panel137-2 C-MOUNT, CASED, USB CABLE TYPEARTCAM-424KY-WOM / 424KY-BW-WOM
ARTCAM-655KY-WOM / 655KY-BW-WOM

C Mount

USB B Connector

Drawing

of

the

base

without a tripod panel

147-3 BOARD LENS, BOARD TYPEARTCAM-445KY2-WOM-OP / 445KY2-BW-WOM-OP
ARTCAM-150P5-WOM-OP / 150P5-BW-WOM-OP
ARTCAM-274KY-WOM-OP / 274KY-BW-WOM-OP

15

7-4 BOARD LENS, BOARD TYPEARTCAM-424KY-WOM-OP / 424KY-BW-WOM-OP

16

7-5 C-MOUNT, BOARD TYPEARTCAM-445KY2-WOM-OP-CM / 445KY2-BW-WOM-OP-CM
ARTCAM-150P5-WOM-OP-CM / 150P5-BW-WOM-OP-CM
ARTCAM-285CX-WOM-OP-CM / 285CX-BW-WOM-OP-CM
ARTCAM-274KY-WOM-OP-CM / 274KY-BW-WOM-OP-CM
ARTCAM-500PⅡ-WOM-OP-CM

17

7-5 C-MOUNT, BOARD TYPEARTCAM-424KY-WOM-OP-CM / 424KY-BW-WOM-OP-CM
ARTCAM-655KY-WOM-OP-CM / 655KY-BW-WOM-OP-CM

18

8. Spectral sensitivity characteristics
8.1 ARTCAM-424KY-WOM series(Color model)(Color model)

8.2 ARTCAM-424KY-BW-WOM series(Monochrome model)(Monochrome model)

198.3 ARTCAM-445KY2-WOM series(Color model)(Color model)

8.4 ARTCAM-445KY2-BW-WOM series(Monochrome model)(Monochrome model)

208.5 ARTCAM-150P5-WOM series(Color model)(Color model)

8.6 ARTCAM-150P5-BW-WOM series(Monochrome model)(Monochrome model)

218.7 ARTCAM-285CX-WOM series(Color model)(Color model)

8.8 ARTCAM-285CX-BW-WOM series(Monochrome model)(Monochrome model)

8.8 ARTCAM-285CX-BW-WOM series(Monochrome model)

228.9 ARTCAM-274KY-WOM series(Color model)(Color model)

8.10 ARTCAM-274KY-BW-WOM series(Monochrome model)(Monochrome model)

238.11 ARTCAM-500PⅡ-WOM series(Color model)(Color model)

248.12 ARTCAM-655KY-WOM series(Color model)(Color model)

8.13 ARTCAM-655KY-BW-WOM series(Monochrome model)(Monochrome model)

25

9. AD Conversion (ARTCAM-WOM series)*AD conversion to 16 bit and thenoutput the upper 10 bit to digital gain
9-1 Standard functions (configurable in standard viewer/SDK)・Digital gain
Global gain (digital gain) can be configured with the standard viewer software or SDK.
Setting range on standard viewer software: 0 – 1023
Default value on standard viewer software: 512
Digital gain = gain value x 0.03125 – 6.0 (dB)
Example 1: When default value is 512,
Analog gain = 512 x 0.03125 – 6.0 ＝10 (dB)
9-2 Optional functions (please contact us for further details.)
・Analog gain:
Can be set to -3dB, 0dB, 6dB, 12dB or 18dB・Black level correction
By default, black level correction is set automatically, and the Pedestal level is set to 32 at 10bit.
(When output is at 8bit, the Pedestal level is approximately 32 / 4 = 8.)
The target value of Pedestal level (32) can also be modified.

26

10. Optional Functions
10-1 Trigger Input
ARTCAM-WOM series are designed with external trigger.
Trigger Input Circuit

Camera

3.3V
BNC

R1

TRIG

PC814

R2

R1 = 2.4kΩ
R2 = 10kΩ

27

FPGA

10-2 GPIO
GPIO option is available upon request before the delivery of the ARTCAM-WOM series.
Pin assign
No.
1
2
3
4
5
6
7
8

Function
GPIO_0
GPIO_1
GPIO_2
VCC(3.3V) Maximum output 50mA
GPIO_3
GPIO_4
GPIO_5
Gnd

・Each of the six GPIO pins is pulled up to 3.3V with 1kΩ on the circuit board and is connected to the IO pins on
FPGA.
・When GPIO pins are used as output pins, the FPGA output is driven at high impedance and Low level.
(At the time of power-up: high impedance)
To control GPIO, please use the ArtCam_SetIOPort / GetIOPort functions in SDK.
To control GPIO output, please specify the corresponding OR calculated values of each bit as per the table below to thesecond argument of the SetIOPort function.
IO pinGPIO_0
GPIO_1
GPIO_2
GPIO_3
GPIO_4
GPIO_5

Bit
0x01
0x02
0x04
0x08
0x10
0x20

For example, to have GPIO_0, GPIO_1, GPIO_4 output High(Z) and GPIO_2, GPIO_3, GPIO_5 output Low,
please call this as follows:
ArtCam_SetIOPort(hACam, (0x01 | 0x02 | 0x10), 0, 0);
To read the GPIO pins, BYTE-type values can be gotten with the GetIOPort function.
For example, to get the status on GPIO_4 pin, please call the followings:
BYTE data = 0x00;
ArtCam_GetIOPort(hACam, &data, 0, 0);
BOOL gpio4 = (data & 0x10) ? TRUE : FALSE;
And the BOOL-type variable gpio4 of GPIO_4 pin is gotten
・Extensions
Monitoring the vertical synchronization signals, horizontal synchronization signals, strobe signals,
and trigger signals on the sensor are possible as extensions of the GPIO option. (Optional)

28

11. System Requirements
11-1 Recommended System Requirements●Intel chip set, Southbridge above ICH6
●CPU: Intel Core2duo 2.4GHz or higher●Memory: Above 512MB●OS: Windows XP or above
11-2 Minimum System Requirements●USB2.0 Enhanced Host Controller●CPU: Intel Pentium4 1.7GHz or higher●Memory: Above 256MB●OS: Windows XP or above

Caution■Please refer the restrictions below when you use ARTCAM.
(1) Recommended System Requirements
If the system specifications do not meet the requirements recommended above, it may be difficult to run at the
maximum frame rate, especially at 48MHz high-speed mode.
(2) Connection of Multiple USB Devices
Please be aware that depending on the specifications of the user's PC, it may be difficult to supply the power to theUSB devices when multiple USB devices are connected.
(3) USB2.0 cable Extension
We cannot guarantee the functionality of the USB2 camera if the user adopts USB2.0 extension cables or repeaterswhich are not confirmed by us. With the extension cables or repeaters, the bandwidth of transfer may differ, and thus
caused malfunctions such as a low frame rate or recognition failure on the camera.
What may cause the problem is that the regulation of the power lines becomes not enough, and so causes impudencemismatch on data signals.
*For the recommended extension cable, please contact our sales department. (TEL: +81-3-3389-5488)

ARTRAY CO., LTD.
1-17-5 Kouenjikita, Suginami-ku, Tokyo 166-0002 Japan
Tel : +81-3-3389-5488
Fax : +81-3-3389-5486
Email : sales@artray.us URL : www.artray. us29