NeonatalVital Signs
Simulator IILM-098
Instruction Manual
Thank you for purchasing the LM-098 Neonatal Vital Signs Simulator II.
Please read this instruction manual carefully to ensure correct use of the product,and store it in a safe place for easy access.

KOKEN CO., LTD.
1-4-14 Koraku, Bunkyo-ku, Tokyo 112-0004 JAPANURL:http://www.kokenmpc.co.jp
E-mail:customer@kokenmpc.co.jp©2012-2021 KOKEN CO., LTD. C-1-992-1-01-08

Handling and Safety Precautions
These precautions should be strictly observed in order to ensure safe,
long-term use of the product.
The following precautions should be observed particularly strictly:
1. The simulator is made with special silicone rubber to simulate the texture of human skin, so werecommend that it be handled carefully in the same manner as an actual infant.
2. Refrain from pressing or pulling the simulator with excessive force to prevent dents anddeformities. The simulator is also susceptible to damage from sharp instruments or fingernails.
3. This simulator cannot be used for bathing practice. Do not attempt to bathe the simulatoror place it in water under any circumstances. In addition, do not put water or tubing into thesimulator, as doing so may cause it to malfunction.
4. Be careful not to bend, twist or step on the connecting cord or power supply cord in order toprevent disconnection.
5. Be careful not to drop the simulator and the control box or subject them to strong impacts, asdoing so may cause malfunction of the simulator.
6. Do not use oil-based inks or dyes on the simulator, as they may cause permanent staining.
7. Do not place objects on the simulator and do not place it face-down, as doing so may change
the shape of the simulator.
8. Do not attempt to open the Magictape™ on the simulator's back or remove the contents, asdoing so may cause malfunction of the simulator.
9. Only use the AC adapter and AC cord provided with the simulator.
If necessary, prepare an appropriate plug adapter according to the wall sockets in your country.
10. This product is intended for use in standard operating environments. Do not use it in heavy
industrial settings.

1

Handling and Safety Precautions
These precautions should be strictly observed in order to ensure safe,
long-term use of the product.
Caring for the simulator after use
1. To clean the simulator's skin, wipe with a clean piece of gauze soaked in water or a mild detergent
diluted with water. Do not use thinners, benzene or similar solvents, as they may cause damage
to the simulator.
2. When not in use, store the simulator in the carrying case away from direct sunlight or ultraviolet
light.

1. The contents of this instruction manual are subject to change without notice.
2. This instruction manual may not be reproduced in part or in its entirety without permission.
3. Please contact the manufacturer in the event that any errors or omissions are found in the contents ofthis instruction manual.
4. This product should be used only as described in this instruction manual. In particular, the productshould not be used in any way that contravenes the precautions noted in the instruction manual.

2

Table of Contents

1. Outline and Features

2. Components and Configuration

3. Handling the Simulator

4. Simulator Configuration Charts

5. Specifications3

1. Description & features
DescriptionWith this LM-098 simulator, infant vital signs can be measured and the entire body can bemonitored. A crying-sound function reproduces the clinical setting more realistically, while heartand breathing sounds can be checked, enabling hands-on training of measurement and monitoringclose to those of the actual clinical site.

Features
Control box
Heart and respiratory rates
・ Heart rate and respiratory rate are displayed digitally over a normal range, and the status of
the infantcan be easily set by turning a rotary dial.
Heart sounds and breathing sounds
・ An external speaker provided on the control box make these sounds audible in the nearby
environment.
・ The sounds can be set independently, so the balance can be adjusted to match the statussettings.
・ Highly realistic sounds are produced with clear sound quality.
Crying-sound function
・ Four different crying patterns can be reproduced.
・ Four buttons are provided to enable direct operation.
* The LM-098's four crying sound buttons are intended to simulate the following meanings:
1)hungry; 2) sleepy; 3) "hold me"; 4) comfortable.
Please note, however, that crying sounds differ among infants and the crying sounds produced by theLM-098 are intended solely as a reference.

Neonatal simulator
Full-body monitoring・ Silicone rubber is used to produce an extremely realistic texture.
・ Monitoring of the head area (palpation of skull molding) includes locating the anterior
fontanel, posterior fontanel, sagittal suture, and coronal suture.
・ A rectal thermometer can be inserted.
Auscultation
・ Vital sign measurement includes auscultation of heart and breathing sounds.

4Visual inspection
・ Thoracoabdominal breathing synchronized to the set respiratory rate can be observed
visually.
Crying-sound function
・ The sound of crying can be heard coming from the mouth.

52. Components and Configurationý View of simulator components

ƤƠè72´ŗeýƬ <ƭýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýý
ýýýýýýýý ý ýƤý
ýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýý
ýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýý ƥýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýƣýý ý ýƦý
ýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýý
ýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýýý
1. Neonatal Simulator (withcord)ƣýconnectingyªZýƝpÀŴƜƃ ƞýýƤýŴƚƂƙƜƗƎſűŹý
ý 2. Control box
ý 3. AC Adapter (with DC cord)ƥýċč ŭŽƌżƜý ƝĎč ŴƜƃ ƞý
ý 4. AC cordƦýċč ŴƜƃý
※ If necessary, prepare an appropriate plug adapter according to the wall sockets in yourý
ýcountry.
Īċč ŴƜƃŝ/=ŜØ Ś0œő£ūň«ńŒŉĽĹý
ýýýýýýýý
6ýýý

Control box front panel

54

1

62
3

10

1.

Monitor2.

Crying sound buttons

3.

Thoracoabdominal movement On/Off button

4.

Heart rate adjustment dial

5.

Respiratory rate adjustment dial

6.

Heart sound volume dial

7.

Breathing sound volume dial

8.

Crying sound volume dial

9.

External speaker volume dial

10. External speaker

78

79

＜コントロールボックス リア＞

Control box rear view

2２
3３
１1

1 . Cable connector
2 . Power switch

１接続コネクタ２ 電源スイッチ
３ 電源コネクタ

3 . Power inlet＜モニター概要＞
Monitor132
4

1 . Heart rate display
2 . Respiratory rate display
3 . Crying sound setting display

１ movement
心拍数表示
4 . Thoracoabdominal
setting display (On/Off)
２
３
４

呼吸数表示
泣き声種類表示
胸腹部可動ＯＮ/ＯＦＦ状態表示-87 -１２
３
４

Neonatal simulator＜新生児人形＞

2２3３

4４

１1

1 . Connecting cord１

接続コード※人形からは外れません
２
呼吸音スピーカー内蔵部
2 . Breathing sound internal speaker３
心音スピーカー内蔵部
3 . Heart sound internal speaker
４ movable
胸腹部可動部
4 . Thoracoabdominal
part*Cannot be disconnected from the simulator

9３．モデルの取扱方法

3. Handlingthe Simulator＜３－１モデル使用方法＞
3-1 How to operate the simulator① 人形についている接続コードをコントロールボックスの接続コネクタに接続します。
1. Connect the cable attached to the simulator to the cable connector on the control box.

② DC コードをコントロールボックスの電源コネクタに接続します。
2. Connect the DC cord to the power inlet of the control box.

③ AC コードを AC アダプターに接続します。
3. Connect the AC cord to the AC Adapter.

④ AC コードをコンセントに差し込みます。

4. Plug the AC cord into a power outlet.

5. Turn the power switch on the control box to the ‘ON' position.
⑤ コントロールボックスの電源スイッチをＯＮにします。
6. When the power is turned on, the monitor displays the heart rate, respiratory rate, crying⑥ 電源が入ると、モニターに心拍数、呼吸数、泣き声の種類、胸腹部可動ＯＮ／ＯＦＦがsound setting and thoracoabdominal movement setting, and each function starts operating.
表示され、各機能が作動します。

7. Adjust the heart and respiratory rates and volume to the desired level by turning therespective dials.
⑦ 心拍数、呼吸数及び音量は、各調節ダイヤルを回して調節してください。
8. Thoracoabdominal movement is controlled by pressing the corresponding button to ON or⑧ 胸腹部の可動は、胸腹部可動ＯＮ／ＯＦＦボタンを押して切り替えてください。
OFF.
9. Press buttons 1-4 to select the desired crying sound and adjust the volume by turning the⑨ 泣き声は、１～４の泣き声種類選択ボタンを押して切り替え、音量を調節ダイヤルで調節
volume dial.
してください。

10. Heart and breathing sounds are also audible from the external speaker. Use the external
speaker volume dial to adjust the volume to the desired level.
⑩ 心音、呼吸音は、外部スピーカーからも聞くことができます。外部スピーカー音量ダイヤルで調節してください。
* Crying sounds are emitted from the simulator's mouth, rather than from the external※泣き声は外部スピーカーからは聞こえません。人形の口元から聞こえます。
speaker.

* Refer to the table on the following page for details on each of the control box buttons and		dials.
※コントロールボックスの各ボタン、ダイヤルにつきましては、次頁の表をご確認ください。

-9-10

3-2 How to operate the control box

43567

12

1 Select crying soundBaby Sound

8These 4 buttons are used to select the desiredcrying sound.

1 □2 □3 □4□* The LM-098's 4 crying sound buttons are designed
to simulate the following meanings:
1) hungry; 2) sleepy; 3) "hold me"; and 4) comfortable.
Crying sounds differ among infants and the crying
sounds made by the LM-098 are intended solely
as a reference.

2 		 Configure		 thoracoabdominal		 movement

Abdominal move
ON/OFFPressing this button turns the thoracoabdominal
movement on & off.
* Auscultation is possible while thoracoabdominal
movement is on, but a slight buzzing noise may
be present.
Turn off the thoracoabdominal movement tobetter distinguish heart and breathing sounds.

3 Adjust heart rate

Heart Rate

Turn the dial to adjust the heart rate within therange of 100-180 beats per minute.

4 Adjust respiratory rate

Respiration Rate

Turn the dial to adjust the respiratory rate within therange of 30-80 breaths per minute.

5 Adjust heart sound volume

Heart Sound

Turn the dial to adjust heart sound volume duringauscultation.

6 Adjust breathing sound volume

Respiratory Sound

Turn the dial to adjust breathing sound volume
during auscultation.

7 Adjust crying sound volume

Baby Sound

Turn the dial to adjust crying sound volume.

8 Adjust external speaker volume

Speaker Volume

Turn the dial to adjust external speaker volume.

11

3-3 After Use
1. Turn the power switch to the ‘OFF' position.
2. Unplug the AC cord from the power outlet.
3. Disconnect the DC cord from the control box.
4. Disconnect the connecting cord from the control box.

4. Simulator Configuration Chart
ItemQuantityNeonatal simulator (with connecting cord)

1

Control box

1

AC adapter (with DC cord)

1AC cord

1

Storage bag

1

Instruction manual (this document)

15. Specifications
Neonatal simulator
HeightWeightHead circumference

Approx. 48 cm

Approx. 2,700 g

Approx. 34 cm

Control box
LengthWidth

HeightWeightPower supply

Approx. 16 cm

Approx. 22 cm

Approx. 11 cm

Approx. 1 kg

100V - 240V ACCaution
・Only use the AC adapter and AC cord provided with the simulator.
If necessary, prepare an appropriate plug adapter according to the wall sockets in your country.

12