ESD (ENDOSCOPIC SUBMUCOSAL DISSECTION)
TRAINING MODEL
with storage caseLM-083

Instruction Manual

Thank you very much for purchasing our ESD Training Model.
Please carefully read this instruction manual for a proper use andkeep this instruction manual at hand.

 Table of Contents 
1.
2.
3.
4.

OutlineCharacteristics
Composition
How to Fix Dissected Stomach
How to fix the belt
5. Precaution for Use
6. How to Clean up
7. Specifications

C-1-961-1-01-04

1. Outline
This resinoid model respresent almost similar figure of human stomach, and can be usedfor the training of ESD. Considering the operability of endoscope, the transition part,from esophagus to stomach is made of soft resin. By setting a real dissected stomachof porcine at the area where ESD is supposed to be executed, very similar feeling of
human stomach wall can be recognized using endoscope.

2. Characteristics(1) The training can be started simply with the combination of this resinoid model anddissected porcine stomach.
(2) Very realistic and similar feeling to the actual treatment.
(3) Perforatoin procedure which never be performed during the actual treatment can beexperienced while recognizing the risk of treatment with realistic feeling.

3. Name of part and structure
Stomach Model

Model Case

Lid (9kinds)

Furniture Case

Metal platesfor fitting
dissectedstomach(９kinds)

Electrode

Fixing belt

-1-4. How to Fix Dissected Stomach
Caution: Since plate for fixation have sharp-edged part, please pay your attention not
to be stuck fingers.
4-1. Stick the electrode on a metal plate for fitting dissected stomach.

4-2. Get out the fixing band through the metal plate for fitting dissected stomach.

4-3. Put the front side (marked) of the dissected stomach on the metal plate for fitting
dissected stomach. * Please make sure the electrode adheres to the dissectedstomach.

MarkFront side (marked)

Back side（mucous membrane）-2-4-4. In order to fix the dissected stomach with the metal plate for fitting dissected stomach,push it against the needle of the metal fitting plate. Using the hollow of the injectionneedle case.

Pull the dissected stomach and fix the dissected stomach with the metal fitting plate tothe needle on the diagonal line in the same manner as above.

There are eight needles in total. Fix them all.

Make sure that the dissected stomach adheres to the metal plates for fitting
dissected stomach.

-3-

4-5. Put the metal plate fitting the dissected stomach of 4-4. with the stomach model,using the fixing belt.

How to fix the belt

In case of No.1 through 2, and 5 through 8

In case of No.3

In case of No.4

In case of No.9-4-

4-6. Install the stomach model with the model case.

-5-5. Precaution for Use
When you insert the endoscope, please be sure to use lubricant such as silicone sprayand glycerin etc.

6. How to Clean up
・Please wipe the inside of case with a wet cloth.
・When you wash the stomach model, please take it out and wash it with a neutral
detergent. Please dry it enough after washing.
・Please wash the metal fittings for dissected stomach in the same manner as thestomach model. Be careful not to stick your finger etc. with the needle when you
wash them.

7. Specifications
Model case (aluminum-made）

1 unitStomach model (silicone rubber, polyurethane resin)

1 unitMetal plates for fitting dissected stomach (stainless steel)1 pc for each kind (9 pcs)

Fixing belt (1cm width x 50cm length, both-sided Velcro)

2 pcs

Circle-cutting electrode1 pc

Instruction manual

1 copyDVD for installation method of resected porcine stomach

1 copyStorage case240 (L) x 315 (W) x 225 (H) mm

Weight2.5 kg

KOKEN CO.,LTD http://www.kokenmpc.co.jp
1-4-14 Koraku, Bunkyo-ku, Tokyo 112-0004 JAPAN
Email: customer@kokenmpc.co.jp-6-Electrode used for ESD Training Model
One (1) pc of exclusive electrode for ESD Training Model (applicable for electrode codewith clip (P code)) is attached with the model but this electrode is not sold separately.
Please use the electrode currently used at your site by processing for ESD Training
Model. However reusable type electrode cannot be used since it does not fit with the
opening shape of training model. Please use disposable type without fail.

Processing Example①Mono-polar type without code(Ex.: 1149 manufactured by 3M）
7cm
7cm

Cut

theelectrodeattheconnecting side into 7x7cm②Mono-polar type with code （Ex.: 1149C manufactured by 3M, CONMED made, etc.）
7cm
7cm

Cut

theelectrodeat

theconnecting side with code into
7x7cm same as above＊ ③Partible type without code
（Ex： NESSYΩ manufactured by ERBE、MAJ-897 manufactured by Olympus, etc.）
7cm

7cm

Cut

theelectrode

at

the

connecting side into 7x7cm
same as above*Basically “Partible type” is not recommendable since the electricity may not be turn on stably.
(There will be some possibility of error sound). It is desirable to use “mono-polar type” but if you need touse “Partible type”, it may be used by holding gauze soaked with saline solution between porcine slice and anelectrode.