November 12, 2020
Ver.1.04

USB3.0
Near Infrared (InGaAs) Camera
ARTCAM-130SWIR-USB3
INSTRUCTION BOOKLETARTRAY CO., LTD.

- 1 -Contents
1. Attention...............................................................................................................................................................................- 3 2. Introduction..........................................................................................................................................................................- 6 3. Main Features.....................................................................................................................................................................- 6 4. The Product.........................................................................................................................................................................- 6 5. Setup....................................................................................................................................................................................- 7 5.1.
Camera connection method (connection example)..............................................................................................- 7 6. Specifications ......................................................................................................................................................................- 8 6.1.
Camera Specification................................................................................................................................................- 8 6.1.1.
External Dimensions ........................................................................................................................................- 9 6.1.2.
Sensor Surface Perimeter Dimension ...........................................................................................................- 9 6.2.
Sensor specification (based on sensor manufacturer's data)............................................................................- 10 6.2.1.
List of sensor specifications ...........................................................................................................................- 10 6.2.2.
Spectral sensitivity characteristics ................................................................................................................- 10 6.3.
Camera function ......................................................................................................................................................- 11 6.4.
External Trigger Function........................................................................................................................................- 12 7. Recommended PC ..........................................................................................................................................................- 14 7.1.
Recommended System Requirements................................................................................................................- 14 -- 2 -1. Attention■ About this manual
1．Before using the camera, please read this manual thoroughly.
2．Please keep this manual reachable and always refer to the contents when needed.
3．Please contact us if the manual is lost or damaged. We will provide a replacement.
4．We cannot guarantee the safety of improper uses of the camera.
5．For your safety, please follow the directions of this manual.
6．All contents are subject to change.
7．Images in this manual may have been simplified to be easier comprehended.
8．Please contact us if you find any unclear points or mistakes in this manual.
9．Quoting, copying or altering some or all parts of the manual without our permission is prohibited.
10．We are not responsible for any lost or damages on your profits due to the use of our products.
11．Please understand that our oversea branches do not provide maintenance or repair services.
■ About the Icons
To keep the safety of the user, other people and their properties, please pay attention to the following icons.

Warning
If the user fails to follow the instruction, serious injury or death may occur.

CautionIf the user fails to follow the instruction, physical injury on human or damages on hardware may occur.

- 3 -■ For Safe Use

Warning● In following circumstances, please stop using the product and turn off the power immediately to prevent the risks offires and electric shocks. If the product is defective, please contact us for repair or exchange. For your safety, please do
not disassemble, modify or repair the camera on your own.
Please stop using the product and turn off the power immediately when:
・The product becomes smoky or gets extremely hot on the surface, or makes unusual smells or sounds.
・Foreign material or water gets into the product.
・The product falls and becomes damaged.
● Do not place the product on unstable surfaces. The product may be fallen and people may get hurt.

Caution● Do not expose the product to steam or fumes to avoid electric shocks and fires.
● Do not leave the product in high temperature places such as inside of vehicles or under direct sunlight.
High temperature may cause damages to the camera, or even cause fires.
● Do not cover the product with cloth or other materials. The product may get extremely hot and the heat may cause
deformations on the parts or even cause fires.
● Please avoid dropping or shocking the product as the product may be damaged.
● Do not touch the cable with a wet hand. Such action may cause electric shocks.
● Please avoid continuously contacting the surface of the camera to your skin when the camera is being used. Thesurface temperature of the camera may cause burns.
■ Other Notices● Please do not use the camera under strong lights such as sun light for a long period. Also please do not expose thecamera under strong lights even when the product is not being used because the sensor might be damaged.
Maintenance● Wipe the dirt on surface with soft cloth or tissue paper. Do not use alcohol, thinner or benzene to avoid damaging thesurface paints.
Electro Magnetic Interference● The camera may interference with electronic devices such as TV and radio. Please do not place the camera next tosuch equipment.
- 4 -■ Export Control
This product is a List Control item subject to the Foreign Exchange and Foreign Trade Act and its relevant legislations
No.1 10 (2) and (4). To export this product from Japan, obtaining export licenses from the Ministry of Economy, Tradeand Industry is required. It is also necessary for our company to manage the information of users and the purposes ofuse. If the end users or purposes of use change after the purchase and thus cause the need to apply export licenses
from Ministry of Economy, Trade and Industry, please contact the sales representative in advance.

■ Guarantee
To save the environment, we do not issue warranty in printed format. Instead, all records of the warranty periods, delivery
dates and the customer information are well kept in our system.
For more details, please refer to the sites below:
Hardware Warranty: http: //www.artray.us/download/artray_warranty.pdf■ Dispose
To dispose this product, please return the camera to us. If you decide to dispose the camera without returning it us,
please follow related regulations and have the camera disposed as an industrial waste. Please always keep records ofthe dispose, and make sure the disposed camera cannot be accessed or used by any 3rd party.

■ Limited Responsibility
We do not guarantee the functions of this product or the descriptions on this manual to be completely adapted to theusers’ end applications or marketing purposes. We are also not held responsible for any direct or indirect damages
caused by our products under any circumstances.
Please do not use this product on applications that require high reliability. This product is not manufactured to be used
as medical, nuclear, aerospace, transportation equipment or equipment that is highly related to human safety. We are
not held responsible for any damages on the users’ property, equipment or personal safety caused by this product.

- 5 -2. Introduction
This manual is to identify the specification of ARTCAM-130SWIR-USB3, NIR camera adopts InGaAs (Indium galliumarsenide) sensor with USB3.0 interface.

3. Main Features⚫ NIR Detection with High Sensitivity
The InGaAs sensor can detect radiation invisible to human eyes and CCD/CMOS cameras in the wavelength range
between 400 nm to 1700 nm bond.
Otherwise, in spite of InGaAs sensor, it has 35% of quantum efficiency between 600 nm to 900 nm, so it can be applied
to detection a wide range of wavelength.
⚫ High Resolution, High Frame Rate
1280 x 1024 pixel and delivers image at 30 frame/second.
⚫ USB3.0 interface
We adopt high reliability USB3.0 interface as industial application.

4. The Product
1)
2)
3)
4)

CameraUSB3.0 cable TYPE A – micro B connecter Approx. 3M
AC adapter (AC100-240V -> DC12V)
Software CD<Options>
1) C Mount lens

- 6 -5. Setup
5.1. Camera connection method (connection example)- 7 -6. Specifications
6.1. Camera Specification
ITEM
Sensor Type

1.3 M pixels InGaAs Image Sensor

InterfaceUSB3.0 (Bulk transfer)

Number of Output Bits

13bit

Frame Rate

30 fps

Shutter Speed

1/10,000 ~ 1sec

Synchronization System

Internal Synchronization / External Trigger Synchronization

Lens Mount

C Mount

External I/O terminal

HR10A-7R-6S(73) (HIROSE)PowerDC12V External Input

Power Consumption

Approx.3W (During normal operation)

Ambient Conditions

Operating Temperature / Humidity ： 10 ~ 35°C / 10 ~ 80%（Non-condensing）
Storage Temperature / Humidity ： 0 ~ 60°C / 10 ~ 95%（Non-condensing）

External Dimensions
Weight71.6(W) × 61.5(H) × 59.0(D)mm* Lens, tripod mount and cables not included
Approx. 360g
* Lens, tripod mount and cables not included- 8 -6.1.1.

External Dimensions

Drawing of the base without a tripod mount6.1.2.

Sensor Surface Perimeter Dimension- 9 -6.2. Sensor specification (based on sensor manufacturer's data)
6.2.1.

List of sensor specifications

ITEM
Image pickup device

SCD InGaAs sensor Cardinal1280

Number of effective pixels

1280(W) x 1024(H)

Number of output pixels

1280(W) x 1024(H)Pixel Size

10[μm] x 10[μm]Imaging Area
Detection frequency band
Saturation capacity /readout noise

12.8[mm] x 10.24[mm](Diagonal 17.173 mm, equivalent to approximately 1 inch)
400 to 1700 nm (whole)
800 to 1600 nm (quantum efficiency 35% or more)
Peak wavelength: 1550 nm (quantum efficiency 80% or more)High Gain: 10Ke / 35e
Mid Gain:
500Ke / 170e
Low Gain: 1Me / 350e

A/D Resolution

6.2.2.

13bit

Spectral sensitivity characteristics- 10 -6.3. Camera function
The functions of the camera are as follows.
SETTINGNO
ITEM
RANGE

DEFAULTVALUE

REMARKS

1Shutter Speed

1 ~ 10,000

100

Changing the shutter speed
Shutter speed [μ sec] = set value * 100

2Global Gain0 ~ 255(0 ~ 7.97times)

128

Gain magnification = set value * 1/32

3RAW image

ON/OFFOFF

4Filter compensation

ON/OFFOFF

5HistogramCompensation

ON/OFFOFF- 11 -6.4. External Trigger Function
This camera has an external trigger electronic circuit, which is insulated by a photocoupler, and thus enables
synchronized shooting by the input signal received from the external circuit.

HR10A-7R-6S(73)(HIROSE)

R1 = 2.4kΩ (5～12V)
R2 = 4.7kΩ

The Pin Assignment of the External I/O Terminal

Pin NumberFunction

1GND

2(Option)3(Option)

4

External Input(-) (Trigger Reception -)5

External Input(+) (Trigger Reception +)6+12V※ The connector pin assignment is subject to change.

- 12 -Details of trigger timing

Item
Numbers
t1
t2

t3
t4
t5
t6
t7

Items

Required Time

Effective Trigger Input Pulse≧10μsec.

Time required from the starting of the External
Trigger to the starting of the Internal Trigger Signals(may delay caused by the photocoupler)
Time required from the starting of the Internal
Trigger Input to the Exposure Time
Strobe Time

≦1μsec.

Time required for data transfer(from the sensor to the FPGA)Time required for the transfer(from the FPGA to USB)
Trigger interval36msec≦1μsec.

Depends on PC specifications and software.
<0.1sec- 13 -7. Recommended PC
7.1. Recommended System Requirements●Host Controller
This camera is applicable to USB 3.0.
Connecting to USB 2.0 host controller may cause low-speed or failure to function properly.
●CPUThe driver of this camera is applicable to computer architecture “x86” or “amd64”.
The specification of CPU effects directly the imaging process speed, therefore it is highly recommended to utilize
a high-end CPU if possible.
●Memory
In the viewer software, there is a data buffer which can store 4 to 8 frames.
Therefore, it is necessary to keep spare space at least for 8 frames in the memory.
(For example, when using 1.3MP color camera, 1280 x 1024 x 3 x 8 [byte] = 30[MB] is necessary.)
It is highly recommended to keep enough memory space especially when using high resolution camera.
●OSThis camera is applicable only to the architecture of Windows NT (32bit/64bit).
Standard functions are confirmed with OS after Windows 7.
In addition, it is recommended to use Windows 10.

Caution■Please refer the restrictions below when you use ARTCAM series.
(1) Recommended System Requirements
If the system specifications do not meet the requirements recommended above, it may be difficult to run at themaximum frame rate.
(2) Use of other USB3.0 Hardware
The data on our camera/converter is transferred in bulk mode. For this reason, when using our camera/converter,
please avoid using other bulk transferred USB3.0 hardware such as Memory stick, External HDD, External DVD,
CDROM etc.
As a solution, we recommend installing a PCI USB host card to the PC and connect external USB hardware tothis port only.
(3) USB3.0 Cable Extension
We cannot guarantee the functionality of the USB3.0 camera if the user adopts USB3.0 extension cables orrepeaters which are not confirmed by us. With the extension cables or repeaters, the bandwidth of transfer may
differ, and thus caused malfunctions such as a low frame rate or recognition failure on the camera.
What may cause the problem is that the regulation of the power lines becomes not enough, and so causes
impudence mismatch on data signals.
*For the recommended extension cable, please contact our sales department.
(TEL: +81-3389-5488)

ARTRAY CO., LTD. Obtained ISO 9001: 2015 2018/9/17
1-17-5 Kouenjikita, Suginami-ku, Tokyo 166-0002 Japan
Tel: +81-3-3389-5488

Fax: +81-3-3389-5486

Email: sales@artray.us

URL: www.artray.us- 14 -