August 26, 2022
Ver1.03

USB2.0 Ultraviolet CCD CAMERAARTCAM-407UV-WOM Series
INSTRUCTION BOOKLET

1.5 M pixels

ARTCAM-407UV-WOM Series

ARTRAY CO., LTD.

- 1 -Contents
1.

Attention ................................................................................................................................................... - 3 -2.

Introduction .............................................................................................................................................. - 6 -3.

Features .................................................................................................................................................... - 6 -4.

The Product .............................................................................................................................................. - 7 -5.

Connections.............................................................................................................................................. - 7 5.1.

How to connect a camera ............................................................................................................... - 7 -5.2.

Shutter Speed................................................................................................................................... - 7 -6.

Specification............................................................................................................................................. - 8 6.1.

7.

Specifications ................................................................................................................................... - 8 -External Dimensions ............................................................................................................................... - 9 7.1.

C-MOUNT, CASED, USB CABLE TYPE ........................................................................................ - 9 -7.2.

BOARD LENS, BOARD TYPE........................................................................................................ - 10 -7.3.

C-MOUNT, BOARD TYPE ............................................................................................................. - 11 -8.

Spectral Sensitivity Characteristics .................................................................................................... - 12 8.1.

ARTCAM-407UV-WOM series ..................................................................................................... - 12 -9.

Ultraviolet (UV) Transmitting, Visible Absorbing Filter Characteristics ........................................ - 12 -10.

AD Conversion (ARTCAM-WOM series)......................................................................................... - 13 -10.1. Standard functions (configurable in standard viewer/SDK).................................................... - 13 10.2. Optional functions (please contact us for further details.) ...................................................... - 13 11.

Optional Functions ............................................................................................................................ - 14 -11.1. Trigger Input ................................................................................................................................... - 14 11.2. GPIO................................................................................................................................................. - 14 11.3. Trigger Timing Chart...................................................................................................................... - 16 12.

System Requirements ....................................................................................................................... - 17 -12.1. Recommended System Requirements ........................................................................................ - 17 -- 2 -1. Attention■About this manual
1. Before using the camera, please read this manual thoroughly.
2. Please keep this manual reachable and always refer to the contents when needed.
3. Please contact us if the manual is lost or damaged. We will provide a replacement.
4. We cannot guarantee the safety of improper uses of the camera.
5. For your safety, please follow the directions of this manual.
6. All contents are subject to change.
7. Images in this manual may have been simplified to be easier comprehended.
8. Please contact us if you find any unclear points or mistakes in this manual.
9. Quoting, copying or altering some or all parts of the manual without our permission is prohibited.
10. We are not responsible for any lost or damages on your profits due to the use of our products.
11. Please understand that our oversea branches do not provide maintenance or repair services.
■About the Icons
To keep the safety of the user, other people and their properties, please pay attention to the following
icons.

W

Warning

If the user fails to follow the instruction, serious injury or death may occur.

CautionIf the user fails to follow the instruction, physical injury on human or damages on hardware may occur.

- 3 -■For Safe UseWarning●In following circumstances, please stop using the product and turn off the power immediately toprevent the risks of fires and electric shocks. If the product is defective, please contact us for repair orexchange. For your safety, please do not disassemble, modify or repair the camera on your own.
Please stop using the product and turn off the power immediately when:
・The product becomes smoky or gets extremely hot on the surface, or makes unusual smells or sounds.
・Foreign material or water gets into the product.
・The product falls and becomes damaged.
●Do not place the product on unstable surfaces. The product may be fallen and people may get hurt.

Caution●Do not expose the product to steam or fumes to avoid electric shocks and fires.
●Do not leave the product in high temperature places such as inside of vehicles or under direct
sunlight. High temperature may cause damages to the camera, or even cause fires.
●Do not cover the product with cloth or other materials. The product may get extremely hot and theheat may cause deformations on the parts or even cause fires.
●Please avoid dropping or shocking the product as the product may be damaged.
●Do not touch the cable with a wet hand. Such action may cause electric shocks.
●Please avoid continuously contacting the surface of the camera to your skin when the camera isbeing used. The surface temperature of the camera may cause burns.
■Other Notices●Please do not use the camera under strong lights such as sun light for a long period. Also please do
not expose the camera under strong lights even when the product is not being used because the sensor
might be damaged.
Maintenance● Wipe the dirt on surface with soft cloth or tissue paper. Do not use alcohol, thinner or benzene toavoid damaging the surface paints.
- 4 -Electro Magnetic Interference●The camera may interference with electronic devices such as TV and radio. Please do not place thecamera next to such equipment.
■Export Control
This product is a Catch-all Control item subject to the Foreign Exchange and Foreign Trade Act and its
relevant legislations. Except for exporting to the 26 white countries designated in the Cabinet Order,
export licenses are required if the products are going to be used for military use or if the end user of theproduct is related to all kinds of military activities. If your circumstances cause the need to apply export
licenses, please notify us before you place orders. Also, please notify us in advance if the end users or
purposes of use change after the purchase and thus cause the need to apply export licenses.
About the Japanese Security Export Controls, please refer to the webpage for Security Export Control
Policy, the Ministry of Economy, Trade and Industry:
www.meti.go.jp/policy/anpo/englishpage.html
The above is based on the enforcement laws and regulations at the time of issuance of this document.
Please be sure to check the latest laws and regulations before actually exporting this product.
■Guarantee
To save the environment, we do not issue warranty in printed format. Instead, all records of thewarranty periods, delivery dates and the customer information are well kept in our system.
For more details, please refer to the following link:
Hardware Warranty: http://www.artray.us/download/artray_warranty.pdf●We do not guarantee the functions of this product or the descriptions on this manual to becompletely adapted to the users’ end applications or marketing purposes. We are also not heldresponsibility for any direct or indirect damages caused by our products under any circumstances.
● Please do not use this product for a task that require high reliability. This product is not
manufactured to be used as medical, nuclear, aerospace, transportation equipment or equipment orthat is highly related to human safety. We are not held responsibility for any damages on the users’
property, equipment or personal safety caused by this product.
■Disposal●To dispose this product, please return the camera to us. If you decide to dispose the camera without
returning it to us, please follow related regulations and have the camera disposed as an industrial
waste. Please always keep track of the disposal, and make sure the disposed camera cannot beaccessed or used by any third party.

- 5 -2. Introduction
The ARTCAM-407UV-WOM is camera adopting a CCD sensor equipped sensitivity of ultraviolet.
The area of visible light is hardly recognized based on its combination with UV lighting equipment and iteasily reflects blur, stain and scratch on the surface of objects.
This series adopt USB2.0 interface, so the data can be delivered to PC directly without a capture board.
Cameras are equipped with on board memory which avoid data missing during transfer.
We provide a variety including cased cameras, board cameras which can be easily incorporated to otherequipment, and so on.

3. Features⚫

High Resolution, High Frame Rate

With high resolution of 1360(H) x 1024(V) pixels, the frame rate can reach 12fps.
⚫

USB2.0 Interface

Our cameras adopt USB2.0 interface, so the data can be delivered to PC directly and no capture board
is needed.
⚫

Small and Light

Our cameras are small and light, thus can be easily installed according to your desires.
Also, the board type camera with small type board lens is easy to be incorporated with other machines.
⚫

Varieties to Your Needs

We provide a variety including cased cameras, board type cameras, C-Mount board type cameras.
Also, we provide color and monochrome models to meet customers’ needs.
⚫ Included Viewer Software
ART-Viewer (Windows 7/8/10).
Users can easily preview, save images and change camera settings by ART-Viewer.
⚫ Included Software Development Kit
When developing image processing software, a software development kit that enables you to acquire
Camera image acquisition and camera control are possible with other applications is included asstandard.
You can also download it from the website below.
http://www.artray.us/download_sdk.html⚫ 2D Measurement & Filing Software(Optional)
Useful 2D measuring functions are available with the optional 2D Measuring & Filing Software ART-MEASURE.
- 6 -4. The Product1) Camera2) USB2.0 Cable (Type A-B connector cable, 3m approx.)*A type-A connector (3m) is attached for an open frame model without USB connector.
3) Viewer software and Device driver CD<Option>
1) C Mount Lens
2) 2D Measurement & Filling Software <ART-MEASURE>
3) LabView DLL
4) USB extension cable with repeater function (5.0m)

5. Connections
5.1. How to connect a camera

5.2. Shutter Speed
Shutter speed can be changed on the standard viewer software ART-Viewer.
The shutter speed is set directly on the sensor register, so please refer to the following formulas for thereal exposure time.
ModelMethod
Range : 7~1074(Exposure Time : 1/58824~1/12.5 sec)

407UV-WOM Series

Exposure Time= ((1074 – Shutter Value)×1790+424)×0.042[μsec]*at 24MHz- 7 -6. Specification
6.1. Specifications
ARTCAM-407UV-WOM
407UV-WOM-OP
407UV-WOM-OP-CM
Sensor TypeCCD Monochrome Image Sensor

Total Number Of Pixels

1434(H)×1050(V)

Number Of Effective Pixels

1392(H)×1040(V)Resolution

1360(H)×1024(V)Active Imager

6.47(H)×4.83(V)mm

Size1/2”

Scan Mode

Progressive ScanPixel Size

4.65(H)×4.65(V)μm

Shutter Type

Global Shutter

Frame Rate

12fps

Shutter Speed

1/58824~1/12.5s

InterfaceUSB2.0 Bulk Transfer

Synchronization System

Internal Synchronization

Lens Mount

C Mount(407UV-WOM、407UV-WOM-OP-CM)
Board lens Mount M12 P0.5(407UV-WOM-OP)

Power

DC5V(Supplied by USB port)

Power Consumption

Under Approx.2W

Ambient Conditions

Operating Temperature/Humidity：0~35°C/10~80%(Non condensing)50(W)×47(H)×49.7(D)mm(407UV-WOM)

External Dimensions

43.5(W)×43.5(H)×33.3(D)mm(407UV-WOM-OP)
50(W)×47(H)×36.7(D)mm(407UV-WOM-OP-CM)*Lens and tripod mount not included
Approx.120g(407UV-WOM)

WeightApprox.50g(407UV-WOM-OP)
Approx.80g(407UV-WOM-OP-CM)*Lens, tripod mount and cable not included

.

- 8 -

7. External Dimensions
7.1. C-MOUNT, CASED, USB CABLE TYPEARTCAM-407UV-WOM

Drawing of the base without a tripod mount- 9 -7.2. BOARD LENS, BOARD TYPEARTCAM-407UV-WOM-OP- 10 -

7.3. C-MOUNT, BOARD TYPEARTCAM-407UV-WOM-OP-CM- 11 -

8. Spectral Sensitivity Characteristics
8.1. ARTCAM-407UV-WOM series

9. Ultraviolet (UV) Transmitting, Visible Absorbing Filter Characteristics
ARTCAM-407UV-WOM is adopting UV transmitting, visible absorbing filter U340.
Please review the image sensor graph of wave length characteristic together for detailed information.
*A model without filter is available.

- 12 -10.

AD Conversion (ARTCAM-WOM series)*AD conversion to 16 bit and thenoutput the upper 10 bit to digital gain

10.1.

Standard functions (configurable in standard viewer/SDK)・Digital gain
Global gain (digital gain) can be configured with the standard viewer software or SDK.
Setting range on standard viewer software: 0 – 1023
Default value on standard viewer software: 512
Digital gain = gain value x 0.03125 – 6.0 (dB)
Example 1: When default value is 512,
Analog gain = 512 x 0.03125 – 6.0 ＝10(dB)

10.2.

Optional functions (please contact us for further details.)

・Analog gain:
Can be set to 3dB, 0dB, 6dB, 12dB or 18dB・Black level correction
By default, black level correction is set automatically, and the Pedestal level is set to 32 at 10bit.
(When output is at 8bit, the Pedestal level is approximately 32 / 4 = 8.)
The target value of Pedestal level (32) can also be modified.

- 13 -11.

Optional Functions

11.1.

Trigger Input

ARTCAM-WOM series are designed with external trigger.

Trigger Input Circuit

Camera

3.3V
BNC

R1

TRIG

PC814

FPGA

R2

R1 = 2.4kΩ
R2 = 10kΩ

11.2.

GPIO

GPIO option is available upon request before the delivery of the ARTCAM-WOM series.
Pin assign
No.

Function

1

GPIO_02

GPIO_13GPIO_2

4

VCC(3.3V) Maximum output 50mA5GPIO_36GPIO_47GPIO_58Gnd- 14 -・Each of the six GPIO pins is pulled up to 3.3V with 1kΩ on the circuit board and is connected to the IO
pins on FPGA.
・When GPIO pins are used as output pins, the FPGA output is driven at high impedance and Low level.
(At the time of power-up: high impedance)
To control GPIO, please use the ArtCam_SetIOPort / GetIOPort functions in SDK.
To control GPIO output, please specify the corresponding OR calculated values of each bit as per thetable below to the second argument of the SetIOPort function.
IO pinBitGPIO_00x01

GPIO_10x02

GPIO_20x04

GPIO_30x08

GPIO_40x10

GPIO_50x20

For example, to have GPIO_0, GPIO_1, GPIO_4 output High(Z) and GPIO_2, GPIO_3, GPIO_5 output
Low, please call this as follows:
ArtCam_SetIOPort(hACam, (0x01 | 0x02 | 0x10), 0, 0);
To read the GPIO pins, BYTE-type values can be gotten with the GetIOPort function.
For example, to get the status on GPIO_4 pin, please call the followings:
BYTE data = 0x00;ArtCam_GetIOPort(hACam, &data, 0, 0);
BOOL gpio4 = (data & 0x10) ? TRUE : FALSE;
And the BOOL-type variable gpio4 of GPIO_4 pin is gotten
・Extensions
Monitoring the vertical synchronization signals, horizontal synchronization signals, strobe signals,
and trigger signals on the sensor are possible as extensions of the GPIO option. (Optional)- 15 -11.3.

Trigger Timing Chart

t1

Trigger input pulse widtht2

From the start of the trigger
Internal trigger signal riset3

Internal trigger signal pulse widtht4

Sensor pixel reset periodt5Sensor exposure periodt6

From exposure completion to datatransfer

t7

Data transfer (sensor-FPGA)

t8

Data transfer (FPGA-USB → PC)

t9

Software processing time

t10

Valid trigger acceptance interval

10 μs or more
About 2 μs or less
About 10 μs (fixed value)
Shutter setting value * 1 line time
(1 line time = 1790 * 1/24 [μsec])(1074 - Shutter setting value) * 1 line time
About 8 line hours
1024 line time ≒ 76 ms
It depends on the USB transfer speed.
(Approximately 40 msec at standard speed)
About 2 msec
(depending on CPU speed and processing content)
About 160 msec

- 16 -12.

System Requirements

12.1. Recommended System Requirements●Host Controller
This camera is applicable to USB 2.0.
●CPUThe driver of this camera is applicable to computer architecture “x86” or “amd64”.
The specification of CPU effects directly the imaging process speed, therefore it is highly
recommended to utilize a high-end CPU if possible.
●Memory
In the viewer software, there is a data buffer which can store 4 to 8 frames.
Therefore, it is necessary to keep spare space at least for 8 frames in the memory.
(For example, when using 1.3MP color camera, 1280 x 1024 x 3 x 8 [byte] = 30[MB] isnecessary.)
It is highly recommended to keep enough memory space especially when using high resolution
camera.
●OSThis camera is applicable only to the architecture of Windows NT (32bit/64bit).
Standard functions are confirmed with OS after Windows 7.
In addition, it is recommended to use Windows 10.

Caution■Please refer the restrictions below when you use ARTCAM series.
(1) Recommended System Requirements
If the system specifications do not meet the requirements recommended above, it may be difficult torun at the maximum frame rate.
(2) Connection of Multiple USB Devices
When connecting multiple USB devices to one USB host controller, some USB device may fail tofunction stably depending on the specifications of the host controller.
(Problem of power supply, data transfer bandwidth, etc.)
To avoid power supply problem, it is recommended to use powered USB hub.
(3) USB2.0 Cable Extension
We cannot guarantee the functionality of the USB2.0 camera if the user adopts USB2.0 extension
cables or repeaters which are not confirmed by us. With the extension cables or repeaters, thebandwidth of transfer may differ, and thus caused malfunctions such as a low frame rate or recognition
failure on the camera.
What may cause the problem is that the regulation of the power lines becomes not enough, and socauses impudence mismatch on data signals.
*For the recommended extension cable, please contact our sales department.
(TEL: +81-3389-5488)- 17 -ARTRAY CO., LTD. Obtained ISO 9001: 2015 2018/9/17
1-17-5 Kouenjikita, Suginami-ku, Tokyo 166-0002 Japan
Tel: +81-3-3389-5488Fax: +81-3-3389-5486

Email: sales@artray.us

URL: www.artray. us- 18 -