Self-Adhesive & Disposable Electrodes
Data Sheet

ELECTRODES 18032021

IMPORTANT NOTESThe present document provides the information about the technical characteristics of the SelfAdhesive & Disposable Electrodes provides with or sold for PLUX products incl. BITalino,biosignalsplux and physioplux kits.
All self-adhesive & disposable electrodes are based on the Covidien ArboTM H124SG
Electrodes. Based on supplier availability, these electrodes may also be provided withdifferent branding names such as: KendalTM ECG Electrodes or Cardinal Health. Images ofthe electrodes with different brandings are displayed below.

Fig. 1: Electrodes with Covidien
branding.

Fig. 2: Electrodes with Arbobranding.

Fig. 3: Electrodes with Cardinal
Health branding.

For further details on the H125SG Electrodes, please refer to the original datasheet:https://biosignalsplux.com/downloads/docs/datasheets/Covidien_Disposable_Electrodes_(H124SG).pdf

Fig. 4: H124SG disposable electrodes.

PLUX – Wireless Biosignals, S.A.
Av. 5 de Outubro, n. 70 – 8.
1050-059 Lisbon, Portugal
plux@plux.info
http://biosignalsplux.com/REV A© 2021 PLUXThis information is provided "as is," and we make no express or implied warranties whatsoever with respect to functionality, operability, use,fitness for a particular purpose, or infringement of rights. We expressly disclaim any liability whatsoever for any direct, indirect, consequential,incidental or special damages, including, without limitation, lost revenues, lost profits, losses resulting from business interruption or loss of data,
regardless of the form of action or legal theory under which the liability may be asserted, even if advised of the possibility of such damages.Self-Adhesive & Disposable Electrodes
Data Sheet

ELECTRODES 18032021

BASIC PRODUCT INFORMATION> Electrode Size: Round / Ø24mm> Total Product surface: 425mm2> Gel area: 201mm2> Adhesive Area: 251mm2> Sensor Area: 80 mm2> Product Thickness (foam): 1mm> Connector: Stud> Color (Foam): White
MATERIALS INFORMATION> Backing material: Polyethylene foam (PE)> Adhesive Characteristics: Medical grade pressure sensitive adhesive> Gel Characteristics: Conductive and adhesive hydrogel> Supporting / black label: Polypropylene film (PP) transparent> Release liner: Polyethylene film (PE), one side siliconized> Sensor (Stud Connector): Polymer Ag/AgCl coated> Adapter / Connector: Stainless steel> Biocompatibility: Test according to DIN ISO 10993 passed (tested by the originalmanufacturer)> Latex Content: None
> Halogenated Hydrocarbon Content (e.g. PVC): None
> Phthalate Derivatives Content (e.g. DEHP): None
> RoHS Directive Compliant: Yes
COMPATIBLE WITH THE FOLLOWING SENSORS
These electrodes are compatible with the following PLUX sensors of the BITalino,biosignalsplux, and physioplux product lines:> Electromyography (EMG)> Electrocardiography (ECG)> Electrodermal Activity (EDA)> Electroencephalography (EEG)> Eletrooculography (EOG)> Electrogastrography (EGG)
APPLICATION NOTESGeneral application notes are provided here; please refer to the individual sensor
documentation for more detailed information about the correct sensor-specific electrodeplacements.
> Always use new electrodes; do not reuse these disposable electrodes;> Do not use the same electrodes on different persons;> Clean the skin surface with an alcohol solution before applying the electrodes to remove
dead skin cells, oils and to improve skin conductivity;> Do not apply the electrodes over damages skin
> Do not apply the electrodes on theDISCLAIMER
Specifications and other information presented in this document are for information purpose
only and do not provide any recommendation or advise of any kind. Data may differ from
actual values quoted here and Tyco / Covidien does not make any warranties orrepresentations to any characteristics. Subject to changes without notice.
ORDERING GUIDE
Reference
870202113

Package Description
Gelled Self-Adhesive Disposable Ag/AgCl Electrodes (Pack of 50)

PAGE 2 OF 2