User’s ManualBluetooth Digital Internal Caliper Gauge
Type : SSIM Series
Contact Point

Battery cover
Display
ButtonButtonButtonMovable Arm

Contact Point

Grip

Fixed Arm

LeverPartsFeatures

Battery Cover
Display

for exchange the Battery(CR2032)
Display Measuring value, Setting mode, etc.

ButtonBasic function by short press, Advanced function by long press

ButtonChange each mode, digit, etc.

ButtonSend button on Measuring, Enter button on setting

Lever

Lifting lever for measuring internal dimension in the work

Fixed Arm

Flame for measuring base

Movable Arm

Lifting flame on measuring internal dimension in the work

Contact Point

Point for Measuring

GripGrip*1) Refer to Indicator manual for detail operation.
*2) Refer to Chapter 6 about Probe shape.

1Q-160-1-E Ver1.1

Descriptions
ＳＳIM－1
ＳＳＳ－５４０

ＳＳIM－２

ＳＳIM－５ＳＳIM－４

Measuring range

Dimensions

Contact Point Hight
MAX Measuring depth

Unit:mm

Measuring Depth by type for a internal dimension in a work, 10mm, 15mm, 20mm and 30mm or more
Measuring Depth
TypeInternal
dimension

Internal dimension

Unit:mm2

１．Power on/off operation（１）Power on
Press any button, then Power on.
（２）Power off① Long pressbutton(2 sec. or more), then it is standby and display is disappeared.
② Long press
button(4 sec. or more), then 「ｏｆｆ」displayed and Power off.
※）In the case of no using for a long time, do ② operation.
２．Preparation for Measuring（１）Setting for Bluetooth① Long press
button, then Advanced mode.（「Unit」displayed）② Short press
button 5 times, then 「bt CFG」displayed.（Bluetooth setting mode）③ Short press
button any times, then select 「HID」 or 「PAIr」 and press
button.
④ Short press
button any times, then 「bt」displayed.（Bluetooth setting mode）⑤ Short pressbutton, then select 「on」 and press
button.（Bluetooth on mode）⑥ Short press
button any times, then 「bt」displayed.（Bluetooth setting mode）⑦ Short press
button, then 「rESEt」displayed and press
button.（Bluetooth Reset）⑧ Pair connection with PC.
⑨ Display on Bluetooth mode
Display status
off

Operating mode
Bluetooth disconnected

blinkinng Bluetooth advertisingon

Pairing mode

Bluetooth connected
reset : clear pairing information

Instrument name on PC display
Mini Indicator

PairSY303

HID

S Dial Mini HID

MAC : display the MAC address
Simple : profile without pairing
Pair : paired and secured profile
HID : virtual keyboard

３．Measuring（１）Setting Zero point① Contact the contact point by using a Ring gauge or a Outside micrometer.
② Short press
button, then 「Zero」displayed and reset.
*）In case of doing Zero reset by using Outside micrometer, keep micrometer for vertical position
so that micrometer head can be downward. Operate micrometer without clamping.
（２）Setting ＭＩＮ mode
① Short press
button 2 times, then 「Mode」displayed.
② Short press
button any times, then 「Min」displayed.
③ Press
button, then ＭＩＮ mode is fixed.
（３）Measuring① Contact the contact point inside a work by Lever.
② Short press
button, then ＭＩＮ value is reset.
③ The MIN value is fixed by holding a work with Arms and Shaking Arms to left and right (orup and down).
④ Press
button, then measuring data(MIN value) is sent to PC.

33

4.Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity

5．Installing and replacing the battery
Mini Indicator （SSI-150S）

６．Specifications
TypeSSIM-1

SSIM-4

0.1Resolution(mm)Measuring range(mm)SSIM-20.01

10-10010-100

10-30

±0.1

Indication error(mm)SSGM-5

20-40
±0.02

5 or less

Measuring force (N)MAX Measuring depth(mm)130180100150

Contact point hight(mm)2224Contact point Tip shape(mm)

Φ2.0

Φ2.0

Φ2.0

R1.5/Φ2.0

Weight(g)

465

585

465

565

Built-in Indicator

SSI-150S

Working temperature(℃)

5～40Strage temperature(℃)-10～60IP protection

IP67Battery life

about 6 months(under general using)Battery

CR2032

７．Certification
contains bluetooth module ISP091201DRegionCertification

USAFCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea South
MexicoMSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan

001-A06167

TaiwanCCAH18LP2040T6EU
India

WPC : ETA-1003/2-17-/RLO(WR)4

CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration has
traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. This
product is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp