User’s ManualBluetooth Digital Indicator Mini 1/1000 , 1/100
Type : SSI-250/150SFeatures➢Resolution➢Protection➢Ｗｉｒｅｌｅｓｓ Communication：0.001/0.01mm：IP67：ＢｌｕｅｔｏｏｔｈQ-142-1-E ver1.2

1Description⑨Measurement units (mm / INCH)⑩ + / - Indicator⑪ Low battery⑫ Preset Mode
⑬ MIN/MAX/DELTA mode
⑭ Multiplication factor⑮ Hold measured value①MODE button②SET button③“ Favorite ” button④Clamping shaft Ø8 or 3/8”
⑤Lifting cap (depends on version)⑥Contact point Ø2 / M2.5 or 4-48-UNF⑦Slot for battery or Power cable⑧Dovetail fixing⑯ Analog scale range⑰ Keypad lock⑱ 6 digits display⑲ Bluetooth® active⑳ Send data
㉑ Analog display

2★Installing and replacing the battery

1. Operating features of the instrument①Button
The instrument has two operating modes: basic functions (direct access) and advanced functions.
In addition to the configuration functions, you can access to MIN, MAX and DELTA (TIR) mode,
plus input of multiplication factor other than 1:1 (see chaps. 3 and 4).
②ButtonThe <favorite> key gives direct access to the function used most often (see chap. 6).
③ButtonSets a Preset value, resets the MIN/MAX mode, verifies a selection, and controls switching off theinstrument. By default, SIS mode enables automatic switch-off with no loss of origin (see chap. 7).
④Serial communication（Bluetooth/RS232/USB）
It is possible to activate or de-activate certain functions of the instrument via serial communication(see chap. 9).
Data transmission parameters are 4800bps,7bits,even parity, and 2 stop bits.

2. Start
Press a button.
For a Bluetooth® connection (see chap. 5).

Short Press

3. Basic functions
Each short press on

Long Pressgives direct access to the basic functions :PrEbt
Bluetooth

MAXMinnorModE

OndELtArESEt

OFF・Assign NORMAL/
MIN/ MAX/DELTA
mode・Bluetooth ON/OFF・Reset Pairing mode
・Display the MAC add.

MAC

4. Advanced functions
Prolonged pressure (>2s) on
gives access to the advanced functions. Then, each short press
onaccesses the required function:

Assign measurement unit(mm/inch)
Unit

MMInchAssign resolution (0.001mm/0.01mm)
rES

Assign measurement direction (+/-)
dir

Assign multiplier mode OFF/ON
MuLt

OnOFF

Assign the multiplier value
next digit

0～9Save Mult.

Activates/deactivates the automatic switch-off
Auto : Activates / Man : De-activates
OFFAutoMAnbt.CFG

SIMPLEPAIrLoc

OFFOn

HId

Assign Bluetooth profile(Config.)
SIMPLE/Pair/HID
Assign Button lock mode(OFF/ON)
Lock mode lockand.
To unlock, Pressfor 5 sec.
Assign Analogue scale.

AnA

1scale=1digit

1scale=5digits 1scale=10digits

4Of

5. Operation via Bluetooth
5.1. HID mode（External Key board mode）

5.3. Bluetooth configuration①Set HID mode by Advanced function.

Display status②Set BT On mode by Advanced function.

Operating mode

③Reset Bluetooth mode by Basic function.

off④Pairing connection the instrument to the PC.
（Instrument name：S_Dial Mini HID）
⑤Send the measured data by the DATA button.
5.2. Pair mode
①Set Pair mode by Advanced function.

blinkinng Bluetooth advertisingonBluetooth disconnectedBluetooth connected
reset : clear pairing information
MAC : display the MAC address②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：SY303）

Simple : profile without pairing
Pair : paired and secured profile⑤Send the measured data by the DATA button.

HID : virtual keyboard

5.4. Bluetooth Connection :1° Activate Bluetooth compatible software and hardware (Master: PC, Display Unit).
2° Start the instrument. By default the Bluetooth® module is active and the instrument is available forconnection (advertising mode).
3° If no connection is established during the advertisement period reactivate the Bluetooth® moduleusing the bt / On menu.
4° Instrument is ready to communicate (connected mode.)
5.5. Only with paired profile:Pairing with master is automatically done at first connection.
To connect the instrument to a new master (new pairing), pairing information on the instrument must becleared using the bt/ rESEt menu.
5.6. Bluetooth Specifications:

Items

Specification

Frequency band

2.4GHzModulation

GFSK

Max output power

Class3(1mW)

Range≦15m（open space），1-5m(industrial environment)

VersionBluetooth4.*

5

6. Favorite key
The «favorite» key gives direct access to a predefined function, and can be configured according to theneeds of the user. In order to assign a function to the «favorite» key, give a prolonged press on,and then select the required function :Validation of selection: By a prolonged press on

or a short press onor.

Note :- A function can also be assigned via RS232 using the command <FCT + Function No.> (FCT 0..9 A..F)
example : Toggle unit = <FCT7>, multiplication factor = <FCTA>.

7. Switching off
The dial gauge goes automatically into stand-by if not used for 10 minutes, unless automatic switch-off
mode has been turned off (see Chap. 4, advanced functions).
Stand-by mode can be forced by a prolonged press (> 2 sec) on
In stand-by mode, the value of the origin is retained by the sensor (SIS mode), and the instrument
automatically restarts with any movement of the measurement probe, RS command, Bluetooth® requestor press on button.
The instrument can be switched off completely for a long period of non-use, but this will necessitate azero reset on restart (the origin will be lost) :- Prolonged press (>4 sec) on

8. Re-initializing the instrument
The initial instrument settings can be restored at any time by a prolonged press (>4 sec) simultaneouslyonanduntil the message rESEt is displayed.

9. Connecting the instrument
The instrument can be connected to a peripheral via a Proximity (RS or USB), Power (RS or USB) cableor Bluetooth®. See page 3 for connecting the Power cable.
Measured values can be transmitted and the instrument driven using predefined commands (see chap. 10for a list of the main commands).

610. Serial communication commands
Selection and configuration

Interrogation

CHA+ / CHA-

Assign measurement direction
CHA+:positive sense / CHA-:negative sense

CHA?

FCT0 / FCT1 / … / FCTA / … / FCTF

Assign «favourite» function

FCT?

MM / IN
KEY0 / KEY1

Assign measurement unit
MM:mm/IN:inch
Assign Keypad Lock
KEY0:Lock/KEY1:Unlock

UNI?
KEY?

MUL +/-xx.xxxx

Assign the multiplier value

MUL?

PRE +/-xxx.xxx

Assign preset value

PRE?

STO1 / STO0
TOL1 / TOL0
REF1 / REF2
ECO1 / ECO 0
INTE1 / INTE0

Assign Hold mode
STO1:ON / STO0:OFF
Assign Tolerance mode
TOL1:ON /TOL0:OFF
Change active reference
Two tolerance values are REF1 or REF2
Assign Economic mode
ECO1:ON / ECO0:OFF
Assign 2 points measurement mode
INTE1:ON / INTE0:OFF

STO?
TOL?
REF?
ECO?
INTE ?

LCAL dd.mm.yy

Modify last calibration date

LCAL?

NCAL dd.mm.yy

Modify next calibration dateNCAL?

NUM x...x (up to 20 chars)

Modify the instrument number

NUM?

MIN /MAX /DEL /NOR
AOFF1 /AOFF0
CFGBAR NOR / CFGBAR MAX

FACT1 / FACT2 / FACT5 / FACT10RES1 / RES2 / RES3

Assign MIN, MAX, Delta, Normal mode
MIN:Minimum/MAX:Maximum/DEL:Delta=MAX-MIN/
NOR:Normal=Current value
Activates/deactivates the automatic switch-off
AOFF1:Activate/AOFF0:De-activate
Assign Bargraph display
CFGBAR NOR:Normal bargraph/
CFGBAR MAX:Keep Bargraph on Max value
Assingn analogue scale factor
FACT1:1scale=1digit/FACT2:1scale=2digits/
FACT5:1scale=5digits/FACT10:1scale=10digits
Change of resolution
RES1:0.0001mm/RES2:0.001mm/RES3:0.01mm

MOD?

Active mode (MIN, MAX, Delta or Normal)?
Response : MIN/MAX/DEL/NOR

AOFF?

Status of the automatic switch-off
Response:AOFF1/AOFF0

CFGBAR?

Bargraph configuration?
CFGBAR NOR/CFGBAR MAX

FACT?

Status of the analogue scale factor?
Response : FACT1/FACT2/FACT5/FACT10RES?

TOL +/-xxx.xxx +/-yyy.yyy

Inputting current tolerance limits
x:lower tolerance limit/y:upper tolerance limit?

CLEReset(Clear) of MIN, MAX or Delta

SET?

[8]Specifications
UNI1 / UNI0
OUT1 /OUT0

Activate / de-activate UNIT command(MM/IN)
UNI1:ON/UNI0:OFF
Activate / de-activate continued data transmission
OUT1:ON/OUT0:OFF

ID?
BAT?

PRE ON / PRE OFF

Activate / de-activate Preset function(PRE command)VER?

ANA ON / ANA OFFActivate / de-activate the analogue scale

MAC?

PRESETSBY xxBTRST

Recall Preset value
Zero reset
xx number of minutes before stand-by
Activate/de-activate Bluetooth® module
BT1:ON/BT0:OFF
Reset Bluetooth pairing information

OFFRSTSBY

Switch-off (wake up using a button or RS)
Reset the instrument
Put instrument in stand-by mode(SIS)

FAC RST

Reset (Restores the factory parameters)
Inputting current tolerance limits
n :nominal value /x :lower tolerance limit/y :upper tolerance limit

BT1 / BT0TOL +/-nnn.nnn +/-xxx.xxx +/-yyy.yyy(In the case of SSI-650）

7

Measurement direction?
Response : CHA+ / CHA«favourite» function ?
Response : FCT0～FCTF
Measurement unit active?
Response : MM/IN
Keypad locked?
Response : KEY0/KEY1
Multiplier value?
Response : +/-xx.xxxx
Preset value?
Response : +/-xxx.xxx
Status of HOLD function?
Response : STO1/STO0
Status of Tolerance mode?
Response : TOL1/TOL0
Active Reference ?
Response : REF1/REF2
Current economic mode?
Response : ECO1/ECO0
2 points mode ?
Response : INTE1/INTE0
Date of last calibration?
Response : dd.mm.yyyy
Date of next calibration?
Response : dd.mm.yyyyInstrument number?
Response : NUM x...x

Status of the current resolution?
Response : RES1/RES2/RES3
Current value (the displayed value)?
Response : +/-zzz.zzz ⇒current value
in the case of Tol mode=+/-zzz.zzz ⇒current value<+/-xxx.xxx ⇒lower tolerance limit>+/-yyy.yyy ⇒upper tolerance limit
Main instrument parameters?
Response : CHA+/CHA-,MM/IN,X1/X2/X5,
RES1/RES2/RES3,MIN/MAX/DEL/NOR,
STO0/STO1,KEY0/KEY1,BAT1/BAT0
Instrument identification code?
Response : SYxxx
Status of Battery?
Response : BAT1: OK/ BAT0: low battery
Version No. and date of firmware
Response : Vx.x DD.MM.YYYYBluetooth® MAC address?
Response :XXX…XXX(up to 12 chars)11. Specifications
TypeSSI-250

Measuring range
Resolution

12.5mm0.001mm

Measuring force
Indication error

SSI-150S

0.01mm
0.5～0.9N ±20%

4µm

10µm

Repeatability

2µmMAX. speed of travel

1.7m/sNo. of measurements/sec

Normal mode : up to 10 meas./s , MIN/MAX mode : 20 meas./s

Data output

Bluetooth/USB/RS232

Data output format

4800bauds,7bits,parity,2stop bits

Battery life

about 6 months(general using)Working temperature

5～40℃

Strage temperature-10～60℃

Weight75g

IP specification

IP67

BatteryCR2032

12.Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity.

8

13.Description of Bluetooth module:This module is based on Nordic Semiconductor nRF8001 μBlue Bluetooth Low Energy Platform. ThenRF8001 is a single chip transceiver with an embedded baseband protocol engine, suitable for ultra-low
power wireless applications conforming to the Bluetooth Low Energy Specification contained within v4.0of the overall Bluetooth specification. The nRF8001, used in the current revision of ISP091201, is a
product using a ROM for the baseband protocol engine.

14.Certification
contains bluetooth module ISP091201D
RegionCertification

USAFCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea SouthMexico

MSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)9CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration
has traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. Thisproduct is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp