User’s ManualBluetooth Digital Constant Pressured Thickness Gauge

Type : SSPG Series
Lift Lever

WeightPole
Display

ButtonButtonButtonBattery CoverHolder Dial
HolderSpindle
Probe

Granite BaseParts

Dimensions

Features

Battery Cover
Display

for exchange the Battery(CR2032)
Display Measuring value, Setting mode, etc.

ButtonBasic function by short press, Advanced function by long press

ButtonChange each mode, digit, etc.

ButtonSend button on Measuring, Enter button on setting

Lift Lever

Lifting lever for putting in the work

WeightWeight for constant pressure

SpindleMeasuring axis

Probe

Contact pointGranite Base

Base for measuring

Unit:mm*1) Refer to Indicator manual for detail operation.

Q-155-1-E Ver1.3

11．Power on/off operation（１）Power on
Press any button, then Power on.
（２）Power off① Long press
button(2 sec. or more), then it is standby and display is disappeared.
② Long press
button(4 sec. or more), then 「ｏｆｆ」displayed and Power off.
※）In the case of no using for a long time, do ② operation.
2．Preparation for Measuring（１）Setting for Bluetooth① Long press
button, then Advanced mode.（「Unit」displayed）② Short press
button 5 times, then 「bt CFG」displayed.（Bluetooth setting mode）③ Short press
button any times, then select 「HID」 or 「PAIr」 and press
button.
④ Short press
button any times, then 「bt」displayed.（Bluetooth setting mode）⑤ Short press
button, then select 「on」 and press
button.（Bluetooth on mode）⑥ Short press
button any times, then 「bt」displayed.（Bluetooth setting mode）⑦ Short press
button, then 「rESEt」displayed and press
button.（Bluetooth Reset）⑧ Pair connection with PC.
Display status
Operating mode
⑨ Display on Bluetooth modeoffPairing mode

Instrument name on PC displayblinkinng Bluetooth advertisingonStandard Indicator

Bluetooth disconnectedBluetooth connected

PairSY289

reset : clear pairing information

HID

S Dial work HID

MAC : display the MAC address
Simple : profile without pairing

3．Measuring（１）Setting zero point① Hold this Probe in standing position.
② Short press
button, 「Zero」displayed and reset.
（２）Measuring a thickness① Place a work between contact point and anvil by Lift lever.
② Press
button, then measuring data is sent to PC.

Pair : paired and secured profile
HID : virtual keyboard

4.Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity
5．Installing and replacing the battery

2

6．Specifications

Measured ItemJIS
SSPG-01

K6732-1996
K6783-1994

SSPG-02

SSPG-11

SSPG-12

Measuring

Reference StandardModel

Poly Vinyl Chloride Films for Agriculture

Resolution

Range(mm)

(mm)

0.001

12for Agriculture
Polyethylene Films for Packaging

0.8N(80±5gf)

ContactPointΦD(mm)
Φ50.00112(125±15gf)
1226±147mN

Φ5(125±15gf)

Z1709-1995

Heat Shrinkable Plastic Films for Packaging

K6400-1-2004

Flexible Polyurethane Foam

K6402-1976

Flexible Urethane Foam
（Repeal of Standard in 2000）

0.01

120.363N(37gf)

Φ35.7

K6301-1995

Vulcanized Rubber（Repeal of Standard in 1998）

0.01

120.785N(80gf)

Φ50.01120.785N(80gf)

Φ100.01

120.394N(40gf)

Φ160.01

122.35N(240gf)

Φ11.3

0.01

120.343N(35gf)

Φ25.2

K6404-2-3-1999 Rubber Cloth / Plastic Cloth
SSPG-13SSPG-14
SSPG-15

SSPG-16

K6328-1999

Rubber Coated Fabrics

K6250-2006

Rubber / For A Method（less than 35 IRHD)

L1086-2007

Fusible Interlining Fabrics(Non Woven Textile)

L1086-2007

Fusible Interlining Fabrics(Ordinary Textile)

L1096-1999

Woven Textile(Ordinary Textile)

L1018-1999

Knitted Fabrics(Ordinary Textile)

L1086-2007

Fusible Interlining Fabrics(Ordinary Knitting Fabric)

L1096-1999

Woven Fabrics(Crinose Textile)

K6505-1995

Man made Upper Material of Shoes

K6550-1994

Leathers

23.5kPa(240gf/㎠)
0.7kPaΦ1012
3.85±0.1N(390±10gf)

0.01

130.431N(44±10gf)※）Instrument weight = Stand / Indicator etc (about 3.7kg) + Weight

7．Certification
contains bluetooth module ISP091201DRegionCertificationUSA

FCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534
MSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan

001-A06167

TaiwanCCAH18LP2040T6EU
India

23.6kPa

49.03±1.177ｋPa(500±12gf/㎠)

3.854±0.098N(393±10gf)

A Method(35 IRHD and over)

Mexico2kPa

0.7kPa(7gf/㎠)

Rubber

Korea South10±1ｋPa

0.7kPa(7gf/㎠)

0.01

K6250-2006

0.363kＰa(3.7gf/㎠)

10±２ｋPa

SSPG-17

SSPG-18

Measured Pressure

1.22±0.14N

Ethylene / Vinyi Acetate Copolymer Film

Z1702-1994

Load（MeasuringForce）

WPC : ETA-1003/2-17-/RLO(WR)3Φ535 IRHD以上22±5ｋPa2(2.24±0.51gf/mm )CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration
has traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. Thisproduct is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp