User’s ManualBluetooth Digital Pitch Caliper

Type : SSC-650P/650E/650C

Features➢Measuring easily Pitch hole to hole or Pitch edge to hole
➢Resolution：0.01mm➢Wireless communication：Bluetooth

Q-148-1-E ver1.1

Flat plate Cone1

Description

Dimensions

Unit : mm

Flat plane cone
■Measuring Pitch hole to hole or Pitch edge to hole

Edge to Holeplane cone
Dimensions

Cone

Unit : mm

Flat plane■Measuring Pitch edge to hole

Dimensions

Unit : mmCone■Measuring Pitch hole to hole2

Description
DATA★Measuring Pitch hole to hole
MODESET

Flat plane cone
ワークWork★Operation①Button MODE
There are two functions, Basic one and Advanced one.
②Button DATAIn the case of measuring, Data transmission. In the case of setting, Data enter.
SET③ButtonPreset by long press. In the case of setting, Data change.
④Serial communication（Bluetooth/RS232/USB）
It is able to transmit the measuring data and set the mode data by serial communication.
Transmission specification are 4800bps, 7 bits, even parity and 2 stop bits.

[1] Basic functions : MODE button⇒Short press( < 1 sec )Short press
Long press
Preset value

set the preset value

SSC-650P

10.00mm

SSC-650E

10.00mm

SSC-650C

20.00mmEnter the presetvalue

Bluetooth mode

reset pairing information

3display the MAC address

[2] Advanced functions : MODE button⇒Long press( ≻ 1 sec)

mminchSimplePaired and Secure

Unit

Bluetooth Profile

HID(2) not available with Simple profile(3) Active configuration displays first[3] Switching off[4]Favorite setting : DATA button⇒Long Press( ≻ 1 sec)

SET button⇒Long Press( ≻ 2 sec)

(1)Activation/Deactivation
Favorite setting
Set DATA ModeAct.

Set “0.00”

Set HOLD ModeStandby
Set SET Mode
Power OffSet PRESET Mode[5] Replacing the battery

4Deact.

[6] Operation via Bluetooth
6.1. HID mode（External Key board mode）

6.3.Bluetooth configuration①Set HID mode by Advanced function.
Display status②Set BT On mode by Advanced function.

Operating mode

③Reset Bluetooth mode by Basic function.

off④Pairing connection the instrument to the PC.
（Instrument name：S_Cal EVO HID）

blinking Bluetooth advertisingon⑤Send the measured data by the DATA button.
6.2.Pair mode
①Set Pair mode by Advanced function.

Bluetooth disconnectedBluetooth connected
reset : clear pairing information
MAC : display the MAC address
Simple : profile without pairing②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：SY295）⑤Send the measured data by the DATA button.

Pair : paired and secured profile
HID : virtual keyboard

6.4. Bluetooth Connection :1° Activate Bluetooth compatible software and hardware (Master: PC, Display Unit).
2° Start the instrument. By default the Bluetooth® module is active and the instrument is available forconnection (advertising mode).
3° If no connection is established during the advertisement period reactivate the Bluetooth® moduleusing the bt / On menu.
4° Instrument is ready to communicate (connected mode.)
6.5. Only with paired profile:Pairing with master is automatically done at first connection.
To connect the instrument to a new master (new pairing), pairing information on the instrument must be
cleared using the bt/ rESEt menu.
6.6. Bluetooth® Specifications:

Items

Specification

Frequency band

2.4GHzModulation

GFSK

Max output power

Class3(1mW)

Range≦15m（open space），1-5m(industrial environment)

VersionBluetooth4.*5[7]仕様
ItemsSSC-650P

SSC-650E

Resolution(mm)SSC-650C

0.01

Measuring Range(mm)5～150/10～150

10～15020～150Indication Error(mm)

±0.03

±0.03

±0.03

Repeatabirity(mm)±0.01

Max slider speed(m/sec)

2.5

Display refresh rate(times/sec)

min 10

Data output

BluetoothMean power consumption(μA)

45Battery life

about 6 months(general using)

Woking temperature(℃)

5～40

Storage tempaerature(℃)-10～60Weight(g)

235

265

BatteryCR2032

Span(mm)258

300

Fixed jaw

Flat plane Cone

Flat planeConeSlider jaw

Flat plane ConeConeCone[8]MaintenanceCarefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity.

6

[9]Description of Bluetooth® module:This module is based on Nordic Semiconductor nRF8001 μBlue Bluetooth Low Energy Platform. ThenRF8001 is a single chip transceiver with an embedded baseband protocol engine, suitable for ultra-low
power wireless applications conforming to the Bluetooth Low Energy Specification contained within v4.0of the overall Bluetooth specification. The nRF8001, used in the current revision of ISP091201, is aproduct using a ROM for the baseband protocol engine.

[10]Certification
contains bluetooth module

ISP091201DRegion

CertificationUSA

FCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea South
MexicoMSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan

001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)7

CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration
has traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. This
product is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.
TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp