User’s ManualBluetooth Digital Hight Gauge

Type : SSH-R1400/SSH-R1600Features➢Measuring Range：400mm/600mm➢Resolution：0.01mm➢Ｗｉｒｅｌｅｓｓ Communication ：Ｂｌｕｅｔｏｏｔｈ

Q-162-1-E ver1.21

Description

14101113122120262728292223

２０．Automatic reference mode indicator
２１．Data sending
２２，Diameter / groove mode indicator
２３．Preset Mode

１．Base２，Post３．Carriage
４．Fast Movement

２６．Active reference indicator
２７．Unit(mm)２８．+/-Indicator
２９．Low battery

５．Carriage Lock
６．Fine Adjustment
７．Probe Folder
８．Probe９．Display
１０．ＭＯＤＥ Button
１１．ＳＥＴ Button１２．SEND Button
１３．Slot for battery

Dimensions

１４．Slot for interface cable

Unit :mm2月

★ Installing and replacing the battery★Changing the probe＊）Comply with the probe orientation !

★Using the carriage➢Rapid carriage movement :Turn the lever (4) and Updown➢fine adjustment :Tighten the carriage locking knob (5)and rotate the knob (6).

3

1. Operating features of the instrument①Button
The instrument has two operating modes: basic functions (direct access) and advanced functions.
In addition to the configuration functions, you can access to MIN, MAX and DELTA mode, plus
input of multiplication factor other than 1:1 (see chaps. 3 and 4).
②ButtonThe <favorite> button gives direct access to the function used most often (see chap. 7).
③ButtonSets a Preset value, resets the MIN/MAX mode, verifies a selection, and controls switching off theinstrument. By default, SIS mode enables automatic switch-off with no loss of origin (see chap. 8).
④Serial communication（Bluetooth/RS232/USB）
It is possible to activate or de-activate certain functions of the instrument via serial communication(see chap. 11).
Data transmission parameters are 4800bps,7bits,even parity, and 2 stop bits.

2. Start
Press any button.
For a Bluetooth connection (see chap. 7).

4

3. Basic functions
Each short press ongives direct access to the basic functions :・Choice of reference：
REF1/REF2/AutoREF
rEF

rEF1

rEF2

Auto

・Inputting a preset value：：Next：Digit 0-9

PrE

diAbt(Bluetooth）

・Selection of Measurement mode：
OFF：Surface measurement ON：Diameter/GrooveOnOFFOn

OFF：SaverESEt(Reset)

MAC・Bluetooth ON/OFF・Reset Pairing mode・Display the MAC add.

4. Advanced functions
Prolonged pressure (>2s) on
gives access to the advanced functions. Then, each short press
onaccesses the required function

Assign measurement unit(mm/inch)
Unit

Assign measurement direction (+/-)

dir

Assign the constant value (Usually the value is “0.00”)：next digit：0-9：saveCSt(Constant)
OFF

Auto

Activates/deactivates the automatic switch-off
Auto : Activates / Man : De-activates

MAn

Assign Bluetooth profile(Config.)
SIMPLE/Pair/HID
bt.CFGSIMPLE

PAIr

HId5

5. Taking a constant with a calibration block

Block Gauge“CSt” ：Constant value for reference measurementNote) The probe constant must
be re-entered after change ofprobe or after modifying theprobe’s position.

Press the CST key and touch one of theground surfaces of the cylindrical gauge.
Release the probe and touch the opposite
surface. The probe constant value is displayed
and memorized until a new constant value istaken. The value of the constant is stored andthen displayed for a few seconds.

6. Diameter / groove measuring mode
To use this mode, you must first enter the touch value constant using the CSt menu or a calibration
block (see chap. 4)

・Reference measurement
・Surface measurement (low probe)
・Activation of the diameter/ groove mode・Groove measurement (high probe)
・Deactivation of the diameter/ groove mode
・Height measurement (low probe)

67. Operation via Bluetooth
7.1. HID mode（External Key board mode）

7.3. Bluetooth configuration①Set HID mode by Advanced function.

Display status②Set BT On mode by Advanced function.

Operating mode

③Reset Bluetooth mode by Basic function.

off④Pairing connection the instrument to the PC.
（Instrument name：Hi Gage HID）
⑤Send the measured data by the DATA button.
7.2. Pair mode
①Set Pair mode by Advanced function.

blinkinng Bluetooth advertising
onBluetooth disconnectedBluetooth connected
reset : clear pairing information
MAC : display the MAC address②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：SY258）⑤Send the measured data by the DATA button.

Simple : profile without pairing
Pair : paired and secured profile
HID : virtual keyboard

7.4. Bluetooth Connection :1° Activate Bluetooth compatible software and hardware (Master: PC, Display Unit).
2° Start the instrument. By default the Bluetooth® module is active and the instrument is available forconnection (advertising mode).
3° If no connection is established during the advertisement period reactivate the Bluetooth® moduleusing the bt / On menu.
4° Instrument is ready to communicate (connected mode.)
7.5. Only with paired profile:Pairing with master is automatically done at first connection.
To connect the instrument to a new master (new pairing), pairing information on the instrument must becleared using the bt/ rESEt menu.
7.6. Bluetooth Specifications:

ItemsSpecification

Frequency band

2.4GHzModulation

GFSK

Max output power

Class3(1mW)

Range≦15m（open space），1-5m(industrial environment)

VersionBluetooth4.*

7

８. Favorite key
The «favorite» key gives direct access to a predefined function, and can be configured according to theneeds of the user. In order to assign a function to the «favorite» key, give a prolonged press on,
and then select the required function :Validation of selection: By a prolonged press on

or a short press onor.

Note :- A function can also be assigned via RS232 or Bluetooth using the command.

９. Switching off
The dial gauge goes automatically into stand-by if not used for 10 minutes, unless automatic switch-off
mode has been turned off (see Chap. 4, advanced functions).
Stand-by mode can be forced by a prolonged press (> 2 sec) on
In stand-by mode, the value of the origin is retained by the sensor (SIS mode), and the instrument
automatically restarts with any movement of the measurement probe, RS command, Bluetooth® requestor press on button.
The instrument can be switched off completely for a long period of non-use, but this will necessitate azero reset on restart (the origin will be lost) :- Prolonged press (>4 sec) on

10. Re-initializing the instrument
The initial instrument settings can be restored at any time by a prolonged press (>4 sec) simultaneouslyonanduntil the message rESEt is displayed.

11. Connecting the instrument
The instrument can be connected to a peripheral via a Proximity (RS or USB), Power (RS or USB) cableor Bluetooth®. See page 3 for connecting the Power cable.
Measured values can be transmitted and the instrument driven using predefined commands (see chap. 10
for a list of the main commands).

812. Serial communication commands
Selection and configuration

Interrogation

CHA+ / CHA-

Assign measurement direction
CHA+:positive sense / CHA-:negative sense

CHA?

FCT0 / FCT1 / … / FCTA / … / FCTF

Assign «favourite» function

FCT?

MM / IN
KEY0 / KEY1

Assign measurement unit
MM:mm/IN:inch
Assign Keypad Lock
KEY0:Lock/KEY1:Unlock

UNI?
KEY?

MUL +/-xx.xxxx

Assign the multiplier value

MUL?

PRE +/-xxx.xxx

Assign preset value

PRE?

STO1 / STO0
TOL1 / TOL0REF1 / REF2
ECO1 / ECO 0
INTE1 / INTE0

Assign Hold mode
STO1:ON / STO0:OFF
Assign Tolerance mode
TOL1:ON /TOL0:OFF
Change active reference
Two tolerance values are REF1 or REF2
Assign Economic mode
ECO1:ON / ECO0:OFF
Assign 2 points measurement mode
INTE1:ON / INTE0:OFF

STO?
TOL?
REF?
ECO?
INTE ?

LCAL dd.mm.yy

Modify last calibration date

LCAL?

NCAL dd.mm.yy

Modify next calibration dateNCAL?

NUM x...x (up to 20 chars)

Modify the instrument number

NUM?

MIN /MAX /DEL /NOR
AOFF1 /AOFF0
CFGBAR NOR / CFGBAR MAX

FACT1 / FACT2 / FACT5 / FACT10
RES1 / RES2 / RES3

Assign MIN, MAX, Delta, Normal mode
MIN:Minimum/MAX:Maximum/DEL:Delta=MAX-MIN/
NOR:Normal=Current value
Activates/deactivates the automatic switch-off
AOFF1:Activate/AOFF0:De-activate
Assign Bargraph display
CFGBAR NOR:Normal bargraph/
CFGBAR MAX:Keep Bargraph on Max value
Assingn analogue scale factor
FACT1:1scale=1digit/FACT2:1scale=2digits/
FACT5:1scale=5digits/FACT10:1scale=10digits
Change of resolution
RES1:0.0001mm/RES2:0.001mm/RES3:0.01mmMOD?

Active mode (MIN, MAX, Delta or Normal)?
Response : MIN/MAX/DEL/NOR

AOFF?

Status of the automatic switch-off
Response:AOFF1/AOFF0

CFGBAR?

Bargraph configuration?
CFGBAR NOR/CFGBAR MAX

FACT?

Status of the analogue scale factor?
Response : FACT1/FACT2/FACT5/FACT10RES?

TOL +/-xxx.xxx +/-yyy.yyy

Inputting current tolerance limits
x:lower tolerance limit/y:upper tolerance limit?

CLE

Reset(Clear) of MIN, MAX or Delta

SET?

UNI1 / UNI0
OUT1 /OUT0

Activate / de-activate UNIT command(MM/IN)
UNI1:ON/UNI0:OFF
Activate / de-activate continued data transmission
OUT1:ON/OUT0:OFF

ID?
BAT?

PRE ON / PRE OFF

Activate / de-activate Preset function(PRE command)VER?

ANA ON / ANA OFF

Activate / de-activate the analogue scale

MAC?

PRE
SET
SBY xxBTRST

Recall Preset value
Zero reset
xx number of minutes before stand-by
Activate/de-activate Bluetooth® module
BT1:ON/BT0:OFF
Reset Bluetooth pairing information

OFF
RST
SBY

Switch-off (wake up using a button or RS)
Reset the instrument
Put instrument in stand-by mode(SIS)

FAC RST

Reset (Restores the factory parameters)
Inputting current tolerance limits
n :nominal value /x :lower tolerance limit/y :upper tolerance limit

BT1 / BT0TOL +/-nnn.nnn +/-xxx.xxx +/-yyy.yyy(In the case of SSI-650）

9Measurement direction?
Response : CHA+ / CHA«favourite» function ?
Response : FCT0～FCTF
Measurement unit active?
Response : MM/IN
Keypad locked?
Response : KEY0/KEY1
Multiplier value?
Response : +/-xx.xxxx
Preset value?
Response : +/-xxx.xxx
Status of HOLD function?
Response : STO1/STO0
Status of Tolerance mode?
Response : TOL1/TOL0
Active Reference ?
Response : REF1/REF2
Current economic mode?
Response : ECO1/ECO0
2 points mode ?
Response : INTE1/INTE0
Date of last calibration?
Response : dd.mm.yyyy
Date of next calibration?
Response : dd.mm.yyyy
Instrument number?
Response : NUM x...x

Status of the current resolution?
Response : RES1/RES2/RES3
Current value (the displayed value)?
Response : +/-zzz.zzz ⇒current value
in the case of Tol mode=+/-zzz.zzz ⇒current value<+/-xxx.xxx ⇒lower tolerance limit>+/-yyy.yyy ⇒upper tolerance limit
Main instrument parameters?
Response : CHA+/CHA-,MM/IN,X1/X2/X5,
RES1/RES2/RES3,MIN/MAX/DEL/NOR,
STO0/STO1,KEY0/KEY1,BAT1/BAT0
Instrument identification code?
Response : SYxxx
Status of Battery?
Response : BAT1: OK/ BAT0: low battery
Version No. and date of firmware
Response : Vx.x DD.MM.YYYY
Bluetooth® MAC address?
Response :XXX…XXX(up to 12 chars)

13.Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure properoperation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity.

14. Specifications
Type

SSH-R1400

SSH-R1600
0.01

Resolution(mm)Measuring range(mm)

0-400

0-600

Indication error(mm)±40±5010

Repeatability(μm)
Weight(g)

4970

5230

Working temperature(℃)

5～40Strage temperature(℃)-10～60IP protection

IP51

Battery life

About 6 months（under general using）

Battery

CR2032

15.Description of Bluetooth module:This module is based on Nordic Semiconductor nRF8001 μBlue Bluetooth Low Energy Platform. ThenRF8001 is a single chip transceiver with an embedded baseband protocol engine, suitable for ultra-lowpower wireless applications conforming to the Bluetooth Low Energy Specification contained within v4.0of the overall Bluetooth specification. The nRF8001, used in the current revision of ISP091201, is a
product using a ROM for the baseband protocol engine.

16.Certification
contains bluetooth module ISP091201DRegion

CertificationUSA

FCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea SouthMexico

MSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)10CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration has
traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. This
product is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp