User’s ManualBluetooth Digital Indicator 1/1000

Type : SSI-550/560/570/580
Features➢Resolution➢Protection➢Ｗｉｒｅｌｅｓｓ Communication：0.001mm：IP54：ＢｌｕｅｔｏｏｔｈQ-141-1-E ver2.01

Description① MODE button② SET button③ “Favorite” button④ Clamping shaft Ø8 or 3/8”
⑤ Lifting cap⑥ Contact point Ø2 / M2.5 or 4-48-UNF⑦ Slot for Proximity cable⑧ Slot for battery or Power cable⑨ Measurement units (mm / INCH)⑩ + / - Indicator⑪ Low battery⑫ Mode menu display⑬ MIN/MAX/DELTA mode
⑭ Preset mode
⑮ Tolerance mode
⑯ 7-digit display⑰ Hold measured value⑱ Keypad lock⑲ Send data
⑳ Bluetooth® active㉑ Multiplication factor㉒ Tolerance indicators㉓ Active reference2Dimensions
Diagram for rear fixings

単位：mmA

B

CD

SSI-550

56.7

66.1

50.5

36.6

SSI-560

69.4

78.5

50.5

49.4

単位：mmA

B

C

SSI-570

121

142

88

SSI-580

172

244

139

3

★Installing and replacing the battery

1. Operating features of the instrument①Button
The instrument has two operating modes: basic functions (direct access) and advanced functions.
In addition to the configuration functions, you can access to MIN, MAX and DELTA mode, plus
input of multiplication factor other than 1:1 (see chaps. 3 and 4).
②ButtonThe <favorite> button gives direct access to the function used most often (see chap. 7).
③ButtonSets a Preset value, resets the MIN/MAX mode, verifies a selection, and controls switching off theinstrument. By default, SIS mode enables automatic switch-off with no loss of origin (see chap. 8).
④Serial communication（Bluetooth/RS232/USB）
It is possible to activate or de-activate certain functions of the instrument via serial communication(see chap. 11).
Data transmission parameters are 4800bps,7bits,even parity, and 2 stop bits.

2. Start
Press any button.
For a Bluetooth connection (see chap. 6).

4

3. Basic functions
Each short press on

gives direct access to the basic functions :・Choice of reference(REF1 or REF2)
rEF

rEF1

rEF2

・Inputting a Preset value
nextdigit 0...9
save PresetPrE

ModE

norMin

toL

OFFOn

Onbt

MAX

OFF

rESEt

dELtA

MAC

・Assign NORMAL/
MIN/ MAX/DELTA
mode・Tolerance display(inputting tolerance limits,see chap. 5)・Bluetooth ON/OFF・Reset Pairing mode・Display the MAC add.

Bluetooth・Note : - It is possible to assign a different preset value to each of the 2 References. Similarly,
different tolerance limits can be assigned to References 1 and 2.

4. Advanced functions
Prolonged pressure (>2s) on
gives access to the advanced functions. Then, each short press
onaccesses the required function:
Assign measurement unit(mm/inch)
Unit

MMInchAssign resolution (0.001mm/0.01mm)
rES

Assign measurement direction (+/-)
dir

Assign multiplier mode OFF/ONMuLt

OFFOn

Assign the multiplier value
next digit

0～9

Save Mult.

Activates/deactivates the automatic switch-off
Auto(Automatic):Activates/Man:(Manual)De-activates
OFFAutoMAn

Assign Bluetooth profile(Config.)
SIMPLE/Pair/HID
bt.CFGSIMPLEPAIrLoc

OFFOnHId

4

Assign Button lock mode(OFF/ON)
Lock mode lockand.
To unlock, Pressfor 5 sec.

5. Inputting tolerance limits
To input or modify the tolerance limits, toL On mode should be selected, followed by a short press on:

・Input the lower tolerance limit
・Input the upper tolerance limit
next digit

0～9

Save ＴＯＬ.

Note :- For measuring internal dimensions, the red and yellow indicators can be switched over by reversing
the order in which the tolerance limits are input (lower limit > upper limit).
- It is possible to input different tolerances on REF1 and REF2.
- It is also possible to display the tolerance limits when the instrument is operating in MIN, MAX or
DELTA mode.
- If no tolerance limits have been defined by the user, the instrument will only display the tolerance limit
indicators
but will not turn on the indicator lights (red - green - yellow).

6. Operation via Bluetooth
6.1. HID mode（External Key board mode）①Set HID mode by Advanced function.
②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：S_Dial work HID）
⑤Send the measured data by the DATA button.
6.2. Pair mode
①Set Pair mode by Advanced function.
②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：SY289）⑤Send the measured data by the DATA button.

66.3 Bluetooth configuration

Display statusoffOperating mode
Bluetooth disconnected

blinking Bluetooth advertising
onBluetooth connected
reset : clear pairing information
MAC : display the MAC address
Simple : profile without pairing
Pair : paired and secured profile
HID : virtual keyboard

6.4. Bluetooth Connection :1° Activate Bluetooth compatible software and hardware (Master: PC, Display Unit).
2° Start the instrument. By default the Bluetooth® module is active and the instrument is available forconnection (advertising mode).
3° If no connection is established during the advertisement period reactivate the Bluetooth® moduleusing the bt / On menu.
4° Instrument is ready to communicate (connected mode.)
6.5. Only with paired profile:Pairing with master is automatically done at first connection.
To connect the instrument to a new master (new pairing), pairing information on the instrument must be
cleared using the bt/ rESEt menu.
6.6. Bluetooth Specifications

Items

Specification

Frequency band

2.4GHzModulation

GFSK

Max output power

Class3(1mW)

Range≦15m（open space），1-5m(industrial environment)

VersionBluetooth4.*77. Favorite key
The «favorite» key gives direct access to a predefined function, and can be configured according to theneeds of the user. In order to assign a function to the «favorite» key, give a prolonged press on,
and then select the required function :Validation of selection: By a prolonged press on

or a short press onor.

Note : - A function can also be assigned via RS232 using the command <FCT + Function No.>(FCT 0..9 A..F)

example : Toggle unit = <FCT7>, multiplication factor = <FCTA>.

8. Switching off
The dial gauge goes automatically into stand-by if not used for 10 minutes, unless automatic switch-off
mode has been turned off (see Chap. 4, advanced functions).
Stand-by mode can be forced by a prolonged press (> 2 sec) on
In stand-by mode, the value of the origin is retained by the sensor (SIS mode), and the instrument
automatically restarts with any movement of the measurement probe, RS command, Bluetooth® requestor press on button.
The instrument can be switched off completely for a long period of non-use, but this will necessitate azero reset on restart (the origin will be lost) :- Prolonged press (>4 sec) on

9. Re-initializing the instrument
The initial instrument settings can be restored at any time by a prolonged press (>4 sec) simultaneouslyonanduntil the message rESEt is displayed.

10. Connecting the instrument
The instrument can be connected to a peripheral via a Proximity (RS or USB), Power (RS or USB) cableor Bluetooth®. See page 3 for connecting the Power cable.
Measured values can be transmitted and the instrument driven using predefined commands (see chap. 10
for a list of the main commands).
Note : - In Tolerance mode, the tolerance limit lights remain lit only for a few seconds while themeasurement stabilizes. On the other hand, they will remain lit continuously if the instrument isconnected to, and powered by, the Power RS (USB) cable.

811. Serial communication commands
Selection and configuration

Interrogation

CHA+ / CHA-

Assign measurement direction
CHA+:positive sense / CHA-:negative sense

CHA?

FCT0 / FCT1 / … / FCTA / … / FCTF

Assign «favourite» function

FCT?

MM / IN
KEY0 / KEY1

Assign measurement unit
MM:mm/IN:inch
Assign Keypad Lock
KEY0:Lock/KEY1:Unlock

UNI?
KEY?

MUL +/-xx.xxxx

Assign the multiplier value

MUL?

PRE +/-xxx.xxx

Assign preset value

PRE?

STO1 / STO0
TOL1 / TOL0
REF1 / REF2
ECO1 / ECO 0
INTE1 / INTE0

Assign Hold mode
STO1:ON / STO0:OFF
Assign Tolerance mode
TOL1:ON /TOL0:OFF
Change active reference
Two tolerance values are REF1 or REF2
Assign Economic mode
ECO1:ON / ECO0:OFF
Assign 2 points measurement mode
INTE1:ON / INTE0:OFF

STO?
TOL?
REF?
ECO?
INTE ?

LCAL dd.mm.yy

Modify last calibration date

LCAL?

NCAL dd.mm.yy

Modify next calibration dateNCAL?

NUM x...x (up to 20 chars)

Modify the instrument number

NUM?

MIN /MAX /DEL /NOR
AOFF1 /AOFF0
CFGBAR NOR / CFGBAR MAX

FACT1 / FACT2 / FACT5 / FACT10
RES1 / RES2 / RES3

Assign MIN, MAX, Delta, Normal mode
MIN:Minimum/MAX:Maximum/DEL:Delta=MAX-MIN/
NOR:Normal=Current value
Activates/deactivates the automatic switch-off
AOFF1:Activate/AOFF0:De-activate
Assign Bargraph display
CFGBAR NOR:Normal bargraph/
CFGBAR MAX:Keep Bargraph on Max value
Assingn analogue scale factor
FACT1:1scale=1digit/FACT2:1scale=2digits/
FACT5:1scale=5digits/FACT10:1scale=10digits
Change of resolution
RES1:0.0001mm/RES2:0.001mm/RES3:0.01mm

MOD?

Active mode (MIN, MAX, Delta or Normal)?
Response : MIN/MAX/DEL/NOR

AOFF?

Status of the automatic switch-off
Response:AOFF1/AOFF0

CFGBAR?

Bargraph configuration?
CFGBAR NOR/CFGBAR MAX

FACT?

Status of the analogue scale factor?
Response : FACT1/FACT2/FACT5/FACT10RES?

TOL +/-xxx.xxx +/-yyy.yyy

Inputting current tolerance limits
x:lower tolerance limit/y:upper tolerance limit?

CLEReset(Clear) of MIN, MAX or Delta

SET?

UNI1 / UNI0
OUT1 /OUT0

Activate / de-activate UNIT command(MM/IN)
UNI1:ON/UNI0:OFF
Activate / de-activate continued data transmission
OUT1:ON/OUT0:OFF

ID?
BAT?

PRE ON / PRE OFF

Activate / de-activate Preset function(PRE command)VER?

ANA ON / ANA OFF

Activate / de-activate the analogue scale

MAC?

PRE
SET
SBY xxBTRST

Recall Preset value
Zero reset
xx number of minutes before stand-by
Activate/de-activate Bluetooth® module
BT1:ON/BT0:OFF
Reset Bluetooth pairing information

OFFRST
SBY

Switch-off (wake up using a button or RS)
Reset the instrument
Put instrument in stand-by mode(SIS)

FAC RST

Reset (Restores the factory parameters)
Inputting current tolerance limits
n :nominal value /x :lower tolerance limit/y :upper tolerance limit

BT1 / BT0TOL +/-nnn.nnn +/-xxx.xxx +/-yyy.yyy(In the case of SSI-650）

9Measurement direction?
Response : CHA+ / CHA«favourite» function ?
Response : FCT0～FCTF
Measurement unit active?
Response : MM/IN
Keypad locked?
Response : KEY0/KEY1
Multiplier value?
Response : +/-xx.xxxxPreset value?
Response : +/-xxx.xxx
Status of HOLD function?
Response : STO1/STO0
Status of Tolerance mode?
Response : TOL1/TOL0
Active Reference ?
Response : REF1/REF2
Current economic mode?
Response : ECO1/ECO0
2 points mode ?
Response : INTE1/INTE0
Date of last calibration?
Response : dd.mm.yyyy
Date of next calibration?
Response : dd.mm.yyyy
Instrument number?
Response : NUM x...x

Status of the current resolution?
Response : RES1/RES2/RES3
Current value (the displayed value)?
Response : +/-zzz.zzz ⇒current value
in the case of Tol mode=+/-zzz.zzz ⇒current value<+/-xxx.xxx ⇒lower tolerance limit>+/-yyy.yyy ⇒upper tolerance limit
Main instrument parameters?
Response : CHA+/CHA-,MM/IN,X1/X2/X5,
RES1/RES2/RES3,MIN/MAX/DEL/NOR,
STO0/STO1,KEY0/KEY1,BAT1/BAT0
Instrument identification code?
Response : SYxxx
Status of Battery?
Response : BAT1: OK/ BAT0: low battery
Version No. and date of firmware
Response : Vx.x DD.MM.YYYYBluetooth® MAC address?
Response :XXX…XXX(up to 12 chars)12. Specifications
Items

Specification

Type

SSI-550

SSI-560

SSI-570

SSI-580

Measuring range

12.5mm

25mm

50mm

100mm

Resolution

0.001mm

Measureing force

0.65-0.90N

0.65-1.15N

1.1-2.6N

1.8-3.8N

Indication Error

3µm

4µm

5µm

6µm

Repeatabirity

2µmMAX. speed of travel

1.7m/sNo. of measurements /second

Normal mode : up to 10 meas/s , MIN/MAX mode : 20 meas/s

Data output

Bluetooth/USB/RS232

Data output parameter

4800bauds,7bits,parity,2stop bits

Battery life

about 6 months(general using)Working temperature

5～40℃

Storage temperature-10～60℃

Weight

119g

123g

EMC

161g
EN61326-1

IP specification

IP54

BatteryCR2032

13. Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity.

10208

14. Description of Bluetooth® module:This module is based on Nordic Semiconductor nRF8001 μBlue Bluetooth Low Energy Platform. ThenRF8001 is a single chip transceiver with an embedded baseband protocol engine, suitable for ultra-low
power wireless applications conforming to the Bluetooth Low Energy Specification contained within v4.0of the overall Bluetooth specification. The nRF8001, used in the current revision of ISP091201, is a
product using a ROM for the baseband protocol engine.

15.Certification
contains bluetooth module ISP091201DRegion

Certification

USAFCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea SouthMexico

MSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)11

CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration
has traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. Thisproduct is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp