User’s ManualBluetooth Digital Thickness Gauge
Type : SSS Series
Lift Lever
Battery Cover

Display

Button

Display

Lift Lever
ButtonButtonButtonButtonButtonBattery CoverSpindleSpindle
Probe

Probe

Anvil

Anvil

Flame

Flame

ＳＳＳ－５６５０

ＳＳＳ－５４０
Parts

Features

Battery Cover
Display

for exchange the Battery(CR2032)
Display Measuring value, Setting mode, etc.

ButtonBasic function by short press, Advanced function by long press

ButtonChange each mode, digit, etc.

ButtonSend button on Measuring, Enter button on setting

Lift Lever

Lifting lever for putting in work

Flame

Flame

SpindleMeasuring axis

Probe

Contact point

Anvil

Base of measuring*1) Refer to Indicator manual for detail operation.
*2) Refer to Chapter 6 about Probe shape.

1Q-150-1-E Ver1.4

Descriptions
ＳＳＳ－５４０

Unit:mm

176.5

Φ103

30.5

101

(95.5)

43(11.2)

930

14

772.5

ＳＳＳ－５５０

7.5

25.4

101

49.6

Φ59.5

46.8

ＳＳＳ－５６５０2823

Φ10(Φ15)

2

7.5

30
72.5

Anvilmoving１．Power on/off operation（１）Power on
Press any button, then Power on.
（２）Power off① Long press
button(2 sec. or more), then it is standby and display is disappeared.
② Long press
button(4 sec. or more), then 「ｏｆｆ」displayed and Power off.
※）In the case of no using for a long time, do ② operation.
２．Preparation for Measuring（１）Setting for Bluetooth① Long press
button, then Advanced mode.（「Unit」displayed）② Short press
button 5 times, then 「bt CFG」displayed.（Bluetooth setting mode）③ Short press
button any times, then select 「HID」 or 「PAIr」 and press
button.
④ Short press
button any times, then 「bt」displayed.（Bluetooth setting mode）⑤ Short press
button, then select 「on」 and press
button.（Bluetooth on mode）⑥ Short press
button any times, then 「bt」displayed.（Bluetooth setting mode）⑦ Short press
button, then 「rESEt」displayed and press
button.（Bluetooth Reset）⑧ Pair connection with PC.
⑨ Display on Bluetooth mode
Display status
off

Operating mode

Pairing modeonStandard Indicator

Pairモード

SY303

SY289

HIDモードS Dial Mini HID

S Dial work HID

Bluetooth disconnected

blinkinng Bluetooth advertisingBluetooth connected
reset : clear pairing information
MAC : display the MAC address
Simple : profile without pairing
Pair : paired and secured profile
HID : virtual keyboard

3Instrument name on PC display
Mini Indicator３．Measuring（１）Setting zero point① Hold this Moving arm in standing position.
② Short press
button, 「Zero」displayed and reset.
（２）Measuring a thickness① Place a work between contact point and anvil by Lift lever.
② Press
button, then measuring data is sent to PC.

4.Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity

5．Installing and replacing the battery（１）In case Mini Indicator（SSI-150S）（２）In case Standard Indicator（SSI-550, SSI-560）

６．Contact point and Anvil for Thickness Gauge

Unit：mm◆LS type◆LW type

Contact
point

Contact
point

Contactpoint

Contact
point

Anvil

Anvil

Anvil

Anvil

◆Standard type

Suitable for ahigh parallelism
sheet-like work

ceramic

Suitable for amoon lens etc.

◆3A type

Suitable for aconcave lenslike shape work

Suitable for aconvex lens anda small blockΦ3.2steel ball

Φ3.2steel ballSK4 temperingceramic

Φ3.2steel ballSK4 tempering

4７．Specifications
TypeSSS-540 SSS-540-LS SSS-540-LW SSS-540-3A SSS-550 SSS-550-LS SSS-550-LW SSS-550-3A SSS-5650
0.01Resolution(mm)0.001
12Measuring range(mm)SSS-130
0.0150(25)

Indication error(μm)±20±3±20

Measuring force(N)

1.2 or less

1.2 or less

1.2 or lessContact point(mm)Φ10FlatΦ3.2BallΦ3.2BallΦ5FlatΦ10FlatΦ3.2BallΦ3.2BallΦ5FlatΦ10FlatΦ10FlatAnvil(mm)Φ10FlatΦ10FlatΦ3.2BallΦ5FlatΦ10FlatΦ10FlatΦ3.2BallΦ5FlatΦ10FlatΦ10Flat

Insertion depth(mm)301203045

Weight(g)

176

344

380

543

SSI-550

SSI-560

IP54

IP54

SSI-150S

Built-in Indicator

5～40

Working tempaerature(℃)-10～60Strage tempaerature(℃)
IP67

IP protection
Battery life

about 6 months（under general using）

Battery

CR2032８．Certification
contains bluetooth module ISP091201DRegion

CertificationUSA

FCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea South
MexicoMSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)5CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration has
traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. This
product is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp