User’s ManualBluetooth Digital Depth Gauge
Type : SSD Series

Battery cover
Display

Display

ButtonButtonButtonButtonButtonBase

Battery cover

ButtonBaseProbe

Probe

PartsFeatures

Battery cover for exchange the Battery(CR2032)
Display

Display Measuring value, Setting mode, etc.

Button Basic function by short press, Advanced function by long press
Button Change each mode, digit, etc.
Button Send button on Measuring, Enter button on setting
Base

Measuring Base

ProbeContact point*1) Refer to Indicator manual for detail operation.
*2) Refer to Chapter 6 about Probe shape.

Q-152-1-E Ver1.41

17(94.7)435(12.3)

Discription

16a

Type

Base Shapea x 16

Hole diameterin Base bottom plane

SSD-21075ｘ162.5

SSD-211
SSD-213

75ｘ16
75ｘ16

2.55.2

SSD-214

75ｘ16

5.2

SSD-215100x165.2

SSD-2100M

75x16

2.5

SSD-2110M
SSD-2130M

75x16
75x16

2.5
5.2
Unit：mm

7.5(107.9)

Φ59.5(20.1)

25.4

16

aType

Base Shapea x 16

Hole diameterin Base bottom plane

SSD-2100
SSD-2150

75ｘ16
100x162.5
5.2
Unit：mm2

SSD-293

SSD-250
SSD-2500M

SSD-252
SSD-2520M

SSD-251

SSD-283

SSD-280SSD-2410
SSD-2400

Detail3

１．Power on/off operation（１）Power on
Press any button, then Power on.
（２）Power off① Long press
button(2 sec. or more), then it is standby and display is disappeared.
② Long press
button(4 sec. or more), then 「ｏｆｆ」displayed and Power off.
※）In the case of no using for a long time, do ② operation.
２．Preparation for Measuring（１）Setting positive value in a measuring value① Long press
button, then Advanced mode.
② Short press
button, 「dir」displayed.
※）
button press times：1/100mmtype is one time, 1/1000mmtype are two times.
③ Short press
button 2 times, then negative displayed, and press
button.

（２）Setting for Bluetooth① Long press② Short press③ Short press④ Short press

button, then Advanced mode.（「Unit」displayed）
button 5 times, then 「bt CFG」displayed.（Bluetooth setting mode）
button any times, then select 「HID」 or 「PAIr」 and press
button.
button any times, then 「bt」displayed.（Bluetooth setting mode）⑤ Short press
button, then select 「on」 and press
button.（Bluetooth on mode）⑥ Short press
button any times, then 「bt」displayed.（Bluetooth setting mode）⑦ Short press
button, then 「rESEt」displayed and press
button.（Bluetooth Reset）⑧ Pair connection with PC.
Pairing mode

Instrument name on PC display
Mini IndicatorStandard Indicator

Pairモード

SY303

SY289

HIDモードS Dial Mini HID

S Dial work HID⑨ Display on Bluetooth mode
Display status
off

Operating mode
Bluetooth disconnected

blinkinng Bluetooth advertisingonBluetooth connected
reset : clear pairing information
MAC : display the MAC address
Simple : profile without pairing
Pair : paired and secured profile
HID : virtual keyboard

4３．Measuring（１）Setting zero point① As shown in the Figure.1、bring the base surface into firm contact with the plane being thereference surface of measurement.
② Short press
button, then 「Zero」displayed and reset.（Figure.1）*）Don’t slide this one right and left, back and forth, leaving the base contacting to thereference surface.(Figure.2)（２）Depth measuring
As shown in the Figure.3, bring the base surface into firm contact with the plane being thereference surface of measurement and put the contact point at the measuring position of theconcave part where the depth is measured. Press
button, then measuring data is sent toPC.

Zero Reset

Press gently.

No spacing norlilting.

Reference

Reference

Figure.1：ZeroReset

Press gentlyby hands.

Don’t slide

Figure.2：Sliding prohibited

Figure.3：Depth Measuring

4.Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity

5．Installing and replacing the battery（１）In case Mini Indicator（SSI-150S、SSI-250）（２）In case Standard Indicator（SSI-550、SSI-560）

5

６．Specifications
Type

SSD-210

SSD-211

SSD-213

SSD-214

SSD-215

SSD-2100M SSD-2110M

SSD-2130M

0.01

Resplution(mm)12220（5pcs extension rods）

12

Measuring range(mm)±20Indication error(μm)
Contact point(mm)NeedleΦ2FlatΦ3.2Steel ball Φ3.2Steel ball Φ3.2Steel ball

None

Extension rod

75x16
Φ2.5

Φ2.5

12220（5pcs extension rods）±6±5
Φ3.2Steel ball

NeedleΦ5.2

Φ5.2239

270

5pcs

75x16
Φ2.5

Φ2.5

Φ5.2

214

17x16

100x16Φ2.5Φ5.2262

SSI-250

323
SSI-550

0.5～0.9±20%

Measuring force(N)Φ3.2Steel ball

None

SSI-150S

Built-in Indicator

SSD-215012Φ2Flat

100x16
Φ5.2214Weight(g)Needle5pcs

Base shape(mm)Base hole diameter(mm)SSD-21000.001

0.65～0.9±20%
5～40Working temperature(℃)-10～60Strage temperature(℃)
IP protection

IP67

Battery life

about 6 months(under general using)Battery

CR2032Type

SSD-293

Resplution(mm)0.01

Measuring range(mm)12220（5pcs extension rods）

SSD-250

SSD-251

SSD-252

IP54

SSD-280

SSD-28355凹5凸4Needle

Spesial needle

NeedleSSD-2400

0.001
12125Needle

Φ3.2Steel ball

Needle±20±20

Contact point(mm)Φ3.2Steel ball

Extension rod

5pcs

Base shape(mm)150x8050x16Special
Φ16Cylinder

50x16

Base hole diameter(mm)Φ11.0

Φ2.0SpecialSpecial

Φ3.0

Weight(g)

450

160

130160140

Built-in Indicator

SSI-150S

None

凹5凸425Needle

Needle25±6None
Φ16Cylinder Φ25Cylinder

SSD-2410
0.001±6Indication error(μm)Φ3.2Steel ball
None

50x16

50x16

100x16100x16

Φ8.0

Φ2.5

Φ5.2

Φ2.0

Φ8.0150

149

144

327

SSI-150S

SSI-250

0.5～0.9±20%

Measuring force(N)SSD-2500M SSD-2520M

0.01

307
SSI-560

0.65～1.15±20%
5～40Working temperature(℃)-10～60Strage temperature(℃)
IP protection

IP67

IP54

Battery life

about 6 months(under general using)Battery

CR2032

７．Certification
contains bluetooth module ISP091201DRegion

CertificationUSA

FCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea South
MexicoMSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan001-A06167

TaiwanCCAH18LP2040T6EUIndia

WPC : ETA-1003/2-17-/RLO(WR)6CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration has
traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. This
product is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp