Fully Automated Urine Analyzer

AUTION MAXAX-4060

Solution to Your Needs

Related products

User friendly

Uriﬂet S 11UA

Color Touch Panel 7 inch color touchscreen displays
measurement results, instrument conditions and operating
instructions.

Uriflet S 11UA is a urinalysis
reagent for machine reading
which can measure 11items.
Albumin/Creatinine ratio andProtein/Creatinine ratio canalso be calculated.
-Measurement itemsGlu, Pro, Bld, pH, Bil, Uro,Ket, Nit, Leu, Cre, Alb,
*P/C and A/C ratio will becalculated.

"Sample delivery system" enablesmicro volume samples to be measured
As the sample is drawn through the nozzle and delivered
correctly and quantitatively to each pad of the strip, the micro
volume samples (minimum 1ml) can be measured easily. No
cross contaminations are found on the reagent portions despitehigh sample volume.

Specifications
Urine
AUTION Sticks/ Uriflet S
GLU (Glucose), PRO (Protein), BIL (Bilirubin), URO(Urobilinogen), PH (pH), BLD (Blood), KET (Ketones),NIT (Nitrite), LEU (Leukocytes), CRE (Creatinine), ALB(Microalbumin), P/C (Ratio of Protein to Creatinine),
A/C (Ratio of Albumin to Creatinine), S.G. (Specific gravity),
turbidity and color tone
Measurement range
Test strip: For rank table, see page 1-12. S.G.: 1.000 to 1.050(Resolution: 0.001)
Color tone: For color classification diagram, see page 1-10.
Turbidity: Three levels including "clear", "turbid" and "denseturbid"
Measurement principle
Test strip: Dual-wavelength reflectance measurement (Singlewavelength for BLD) S.G.: Reflection refractometry
Color tone: Light-transmission measurement Turbidity:Light-scattering measurement
Measurement wavelength 5 LED wavelengths (450, 525, 562, 635 and 760 nm)
Test strip reaction time
Approximately 60 seconds
Test strip storage
Two test strip storage compartments, each of which cancontain different test strips
Test strip storage capacity Maximum 200 test strips x two storage compartments
Processing speed
Normal measurement: Maximum 225 samples/hour
Small volume measurement: Maximum 112 samples/hour
Required sample volume Normal, STAT measurement: Minimum 2 mL
Small volume measurement: Minimum 1 mL
Measurement object
Reagent
Measurement items

Sample tubes (length: 95 - 110 mm, diameter: 014.0 - 15.8mm)
Sample loading capacity
One-way transportation (default): Maximum 50 samples
Loop transportation: Maximum 100 samples
Warm-up time
Maximum 4 minutes
Display
Large color liquid crystal display (800 × 480 dots)
Built-in printer
58-mm width printer paper (24 digits)
Memory capacity
Normal, STAT, small volume measurement: 2500 tests
Control measurement: 200 tests
Check measurement: 50 tests
Trouble list: 100 tests
External output
1 port (Replaceable with the optional Ethernet terminal)
Communication system
RS-232C compliant (Switchable between one-way andtwo-way)
Transmission speed
RS-232C: Selectable from 300, 600, 1200, 2400, 4800,9600 and 19200 bps
Ethernet: 10BASE-T
Measurement environment Temperature: 10 - 30°C
Humidity: 30 - 60% RH (No condensation)
Dimensions
530(W) × 530(D)× 530(H) mm(including the sampler)
WeightMain body: Approximately 40 kg
Sampler: Approximately 4 kg
Power requirements
100 - 240 V AC (Maximum power line fluctuation of±10%)50/60 HzPower input
Maximum 150 VA
Sample container

Legal manufacturer
1480 Koji, Konan-cho, Koka-shi, Shiga 520-3306, JAPANEuropean representative

Yousuien-nai, 59 Gansuin-cho, Kamigyo-ku,
Kyoto 602-0008, JAPANTEL +81-75-662-8979 FAX +81-75-431-1202

Prof. J.H. Bavincklaan 2, 1183 AT Amstelveen, THE NETHERLANDS
https://www.arkray.eu/english/*Designs and specifications may be changed without prior notice.

AG220523-01

reviewed