User’s ManualBluetooth Digital Indicator 1/10000

Type : SSI-650
Features➢Resolution➢Protection➢Ｗｉｒｅｌｅｓｓ Communication：0.0001mm：IP54：Ｂｌｕｅｔｏｏｔｈ

Q-143-1-J ver1.21

Description⑪ Low battery⑫ + / - Indicator⑬ Keypad lock
⑭ 7-digit display⑮ Hold measured value⑯ Send data
⑰ Measurement units (mm / INCH)⑱ Preset mode
⑲ Analogue scale⑳ Lower limit exceeded㉑ Upper limit exceeded㉒ Tolerance mode

① MODE button② SET button③ “Favorite” button④ ◁ button⑤ ▷ button⑥ Clamping shaft Ø8 or 3/8”
⑦ Contact point Ø2 / M2.5 or 4-48-UNF⑧ Lifting cap⑨ Slot for Proximity cable⑩ Slot for battery or Power cable㉓ Analogue scale lower limit㉔ Analogue scale upper limit㉕ Unit and value of analogue scale㉖ Tolerance mode
㉗ MIN/MAX/DELTA mode
㉘ Multiplier㉙ Bluetooth connection㉚ Symmetrical tolerances

Diagram for rear fixings

56.7
36.6

50.5
66.1

2
月

★Installing and replacing the battery

1. Operating features of the instrument①Button
The instrument has two operating modes: basic functions (direct access) and advanced functions.
In addition to the configuration functions, available access to the MIN, MAX and DELTA mode, ordisplay of tolerances (see chaps. 3 and 4)②ButtonThe <favorite> button gives direct access to the function used most often (see chap. 9).
③ButtonSets a Preset value, resets the MIN/MAX mode, verifies a selection, and controls switching off theinstrument. By default, SIS mode enables automatic switch-off with no loss of origin (see chap. 11).
④ButtonWith the «Left and Right» buttons, the user may change the extent of the analogue scale as well asenter the values of the nominal dimension and tolerance limits. (see chap. 6.2)

⑤Serial communication（Bluetooth/RS232/USB）
It is possible to activate or de-activate certain functions of the instrument via serial communication(see chap. 13).
Data transmission parameters are 4800bps,7bits,even parity, and 2 stop bits.

2. Start
Press any button.
For a Bluetooth connection (see chap. 5).

3

3. Basic functions
Each short press on

gives direct access to the basic functions :・Normal mode
: Zero reset
・Min (Max) mode : Preset on the measuredmin (max) value

SEt・Inputting a Preset value(see chap. 7)

PrEModEnorMin

MAXdELtA

IntE

OFF-AMin+AMin+AMAX

toL

OFFOnrESEt

MACbt(Bluetooth） On

OFF・Assign NORMAL/MIN/MAX/DELTA mode(see chap. 5)・2 points measurement (Dynamic preset)(see chapter 11)
・Tolerance display(inputting tolerance limits, see chap. 6)
・Bluetooth ON/OFF・Reset Pairing mode
・Display the MAC add.

4. Advanced functions
Prolonged pressure (>2s) on
gives access to the advanced functions. Then, each short press onaccesses the required function:

Assign measurement unit(mm/inch)
Unit

MMInchAssign resolution (0.0001mm/0.001mm)
rES

Assign measurement direction (+/-)
dir

Assign multiplier mode OFF/ON
MuLt

OFFOn

Assign the multiplier value
next digit

OFFAutoMAnbt.CFG

SIMPLEPAIrLocOFFOn

0～9Save Mult.

Activates/deactivates the automatic switch-off
Auto(Automatic):Activates/Man:(Manual)De-activates

HId

4

Assign Bluetooth profile(Config.)
SIMPLE/Pair/HID
Assign Button lock mode(OFF/ON)
Lock mode lockand.
To unlock, Pressfor 5 sec.

5. Work in MINIMUM, MAXIMUM and DELTA mode :In this operating mode, the digital display stores the MIN, MAX or DELTA value. On the other hand, theanalogue scale always indicates the current measured value.
- A short press on
resets the MIN/MAX stored value (CLEAr)- A long press (>2s) on
assigns the Preset value to the position of current measurement.

6. Inputting (or modifying) tolerance limits
The tolerance limits are defined by introducing lower and upper intervals, compared to the nominal rating.
Two methods are available :6.1 Using the TOL mode

Input the nominal value (see chap. 7)

Input the lower tolerance limit (see chap. 7)
Input the upper tolerance limit (see chap. 7)

6.2 Use of the buttons
and:
It is also possible to enter or modify the nominal value or the tolerance limits by prolongedly pressing(>2s) theand / orbutton :Note :- It is also possible to display the tolerance limits when the instrument is operating in MIN, MAX or DELTA
mode.
- If no tolerance limits have been defined by the user, the instrument will only display the tolerance limit
indicators but will not turn on the indicator lights (red - green - yellow).
- In case of measuring internal ratings, you can cross the indicators (red and yellow) by reversing theorder of entering the tolerance intervals (upper interval < lower interval).

57. Entering a numeric value
To enter or modify a numeric value (Preset, nominal ratings, tolerance intervals or multiplication factor),
you can proceed in two ways :7.1 Increases / decreases the current value :Decreases the current value
Increases the current value
Note: - the scanning speed increases with a long press on the button- save the value with one of the three lower buttons
7.2 Entering digit by digit :Increase of the selected digit
Selection of the next digit
Save the value

8. Scale the analog display
The user may change the analogue display’s scale value by shortly pressingor.

8.1 Centering function of the analogue scale
When the tolerance limits are asymmetric with respect to the nominal ratings, it is possible to realign themto the analogue scale, by a short simultaneous press onand.

Asymmetric tolerance limits

Realigned tolerance limits

69. Operation via Bluetooth
9.1. HID mode（External Key board mode）

9.3. Bluetooth configuration①Set HID mode by Advanced function.

Display status②Set BT On mode by Advanced function.

Operating mode

③Reset Bluetooth mode by Basic function.

off④Pairing connection the instrument to the PC.
（Instrument name：S_Dial Work HID）
⑤Send the measured data by the DATA button.
9.2. Pair mode
①Set Pair mode by Advanced function.

blinkinng Bluetooth advertisingonBluetooth disconnectedBluetooth connected
reset : clear pairing information
MAC : display the MAC address②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：SY289）⑤Send the measured data by the DATA button.

Simple : profile without pairing
Pair : paired and secured profile
HID : virtual keyboard

9.4. Bluetooth Connection :1° Activate Bluetooth compatible software and hardware (Master: PC, Display Unit).
2° Start the instrument. By default the Bluetooth® module is active and the instrument is available forconnection (advertising mode).
3° If no connection is established during the advertisement period reactivate the Bluetooth® moduleusing the bt / On menu.
4° Instrument is ready to communicate (connected mode.)
9.5. Only with paired profile:Pairing with master is automatically done at first connection.
To connect the instrument to a new master (new pairing), pairing information on the instrument must becleared using the bt/ rESEt menu.
9.6. Bluetooth Specifications:

Items

Specification

Frequency band

2.4GHzModulation

GFSK

Max output power

Class3(1mW)

Range≦15m（open space），1-5m(industrial environment)

VersionBluetooth4.*

7

10. Favorite key
The «favorite» key gives direct access to a predefined function, and can be configured according to theneeds of the user. In order to assign a function to the «favorite» key, give a prolonged press on,
and then select the required function :Validation of selection: By a prolonged press on
or a short press onor.
Note : - A function can also be assigned via RS232 using the command <FCT + Function No.>(FCT 0..9 A..F)
example : Toggle unit = <FCT7>, multiplication factor = <FCTA>.

11. Adjustment and use of IntE dynamic measuring mode
Certain applications need to adjust the instrument to the MIN (or MAX) measured value. In this case,proceed as follows :11.1 Adjustment of the instrument- Enter a Preset value corresponding to the actual size of the standard (see chap. 3)- In the IntE menu, select the mode -AMIN (automatic selection of DIR- and mode MIN) or +AMIN(automatic selection of DIR+ and mode MIN) or +AMAX (automatic selection of DIR+ and mode MAX)
depending on application- Make a standard measurement (going through the turnaround point)- Adjust the instrument by selecting the SEt mode and pressing on the button(see chap. 3)- The instrument is adjusted and ready to measure.
11.2 Measure- Make the measurements going through the turnaround point. The digital display stores and displays
the MIN (or MAX) measured value.
- Before each new measurement, reset the measured value by a short press on
Examples :8.

12. Switching off
The dial gauge goes automatically into stand-by if not used for 10 minutes, unless automatic switch-off
mode has been turned off (see Chap. 4, advanced functions).
Stand-by mode can be forced by a prolonged press (> 2 sec) on
In stand-by mode, the value of the origin is retained by the sensor (SIS mode), and the instrument
automatically restarts with any movement of the measurement probe, RS command, Bluetooth® requestor press on button.
The instrument can be switched off completely for a long period of non-use, but this will necessitate azero reset on restart (the origin will be lost) :- Prolonged press (>4 sec) on

13. Re-initializing the instrument
The initial instrument settings can be restored at any time by a prolonged press (>4 sec) simultaneouslyonanduntil the message rESEt is displayed.

14. Connecting the instrument
The instrument can be connected to a peripheral via a Proximity (RS or USB), Power (RS or USB) cableor Bluetooth®. See page 3 for connecting the Power cable.
Measured values can be transmitted and the instrument driven using predefined commands (see chap. 10for a list of the main commands).
Note : - In Tolerance mode, the tolerance limit lights remain lit only for a few seconds while themeasurement stabilities. On the other hand, they will remain lit continuously if the instrument isconnected to, and powered by, the Power RS (USB) cable.

15.Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity.

916. Serial communication commands
Selection and configuration

Interrogation

CHA+ / CHA-

Assign measurement direction
CHA+:positive sense / CHA-:negative sense

CHA?

FCT0 / FCT1 / … / FCTA / … / FCTF

Assign «favourite» function

FCT?

MM / IN
KEY0 / KEY1

Assign measurement unit
MM:mm/IN:inch
Assign Keypad Lock
KEY0:Lock/KEY1:Unlock

UNI?
KEY?

MUL +/-xx.xxxx

Assign the multiplier value

MUL?

PRE +/-xxx.xxx

Assign preset value

PRE?

STO1 / STO0
TOL1 / TOL0
REF1 / REF2
ECO1 / ECO 0
INTE1 / INTE0

Assign Hold mode
STO1:ON / STO0:OFF
Assign Tolerance mode
TOL1:ON /TOL0:OFF
Change active reference
Two tolerance values are REF1 or REF2
Assign Economic mode
ECO1:ON / ECO0:OFF
Assign 2 points measurement mode
INTE1:ON / INTE0:OFF

STO?
TOL?
REF?
ECO?
INTE ?

LCAL dd.mm.yy

Modify last calibration date

LCAL?

NCAL dd.mm.yy

Modify next calibration dateNCAL?

NUM x...x (up to 20 chars)

Modify the instrument number

NUM?

MIN /MAX /DEL /NOR
AOFF1 /AOFF0
CFGBAR NOR / CFGBAR MAX

FACT1 / FACT2 / FACT5 / FACT10
RES1 / RES2 / RES3

Assign MIN, MAX, Delta, Normal mode
MIN:Minimum/MAX:Maximum/DEL:Delta=MAX-MIN/
NOR:Normal=Current value
Activates/deactivates the automatic switch-off
AOFF1:Activate/AOFF0:De-activate
Assign Bargraph display
CFGBAR NOR:Normal bargraph/
CFGBAR MAX:Keep Bargraph on Max value
Assingn analogue scale factorFACT1:1scale=1digit/FACT2:1scale=2digits/
FACT5:1scale=5digits/FACT10:1scale=10digits
Change of resolution
RES1:0.0001mm/RES2:0.001mm/RES3:0.01mm

MOD?

Active mode (MIN, MAX, Delta or Normal)?
Response : MIN/MAX/DEL/NOR

AOFF?

Status of the automatic switch-off
Response:AOFF1/AOFF0

CFGBAR?

Bargraph configuration?
CFGBAR NOR/CFGBAR MAX

FACT?

Status of the analogue scale factor?
Response : FACT1/FACT2/FACT5/FACT10RES?

TOL +/-xxx.xxx +/-yyy.yyy

Inputting current tolerance limits
x:lower tolerance limit/y:upper tolerance limit?

CLEReset(Clear) of MIN, MAX or Delta

SET?

UNI1 / UNI0
OUT1 /OUT0

Activate / de-activate UNIT command(MM/IN)
UNI1:ON/UNI0:OFF
Activate / de-activate continued data transmission
OUT1:ON/OUT0:OFF

ID?
BAT?

PRE ON / PRE OFF

Activate / de-activate Preset function(PRE command)VER?

ANA ON / ANA OFF

Activate / de-activate the analogue scale

MAC?

PRE
SET
SBY xxBTRST

Recall Preset value
Zero reset
xx number of minutes before stand-by
Activate/de-activate Bluetooth® module
BT1:ON/BT0:OFF
Reset Bluetooth pairing informationOFFRSTSBY

Switch-off (wake up using a button or RS)
Reset the instrument
Put instrument in stand-by mode(SIS)

FAC RST

Reset (Restores the factory parameters)
Inputting current tolerance limits
n :nominal value /x :lower tolerance limit/y :upper tolerance limit

BT1 / BT0TOL +/-nnn.nnn +/-xxx.xxx +/-yyy.yyy(In the case of SSI-650）

10Measurement direction?
Response : CHA+ / CHA«favourite» function ?
Response : FCT0～FCTF
Measurement unit active?
Response : MM/IN
Keypad locked?
Response : KEY0/KEY1
Multiplier value?
Response : +/-xx.xxxx
Preset value?
Response : +/-xxx.xxx
Status of HOLD function?
Response : STO1/STO0
Status of Tolerance mode?
Response : TOL1/TOL0
Active Reference ?
Response : REF1/REF2
Current economic mode?
Response : ECO1/ECO0
2 points mode ?
Response : INTE1/INTE0
Date of last calibration?
Response : dd.mm.yyyy
Date of next calibration?
Response : dd.mm.yyyy
Instrument number?
Response : NUM x...x

Status of the current resolution?
Response : RES1/RES2/RES3
Current value (the displayed value)?
Response : +/-zzz.zzz ⇒current value
in the case of Tol mode=+/-zzz.zzz ⇒current value<+/-xxx.xxx ⇒lower tolerance limit>+/-yyy.yyy ⇒upper tolerance limit
Main instrument parameters?
Response : CHA+/CHA-,MM/IN,X1/X2/X5,
RES1/RES2/RES3,MIN/MAX/DEL/NOR,
STO0/STO1,KEY0/KEY1,BAT1/BAT0
Instrument identification code?
Response : SYxxx
Status of Battery?
Response : BAT1: OK/ BAT0: low battery
Version No. and date of firmware
Response : Vx.x DD.MM.YYYYBluetooth® MAC address?
Response :XXX…XXX(up to 12 chars)

17. Specifications
Items

Specification

Measuring range

12.5 mm

Resolution

0.0001 mm

Measureing force

0.65～0.9 N

Indication Error

1.8 µmRepeatabirity

0.5 µmMAX. speed of travel

1.7 m/s

No. of measurements /second

Normal mode : up to 10 meas/s , MIN/MAX mode : 20 meas/s

Data output

Bluetooth/USB/RS232

Data output parameter

4800bauds,7bits,parity,2stop bits

Battery life

about 6 months(general using)Working temperature

5～40℃

Storage temperature-10～60℃

Weight119g

EMCEN61326-1

IP specification

IP54

Battery

CR2032

18.Description of Bluetooth® module:This module is based on Nordic Semiconductor nRF8001 μBlue Bluetooth Low Energy Platform. ThenRF8001 is a single chip transceiver with an embedded baseband protocol engine, suitable for ultra-low
power wireless applications conforming to the Bluetooth Low Energy Specification contained within v4.0of the overall Bluetooth specification. The nRF8001, used in the current revision of ISP091201, is a
product using a ROM for the baseband protocol engine.

19.Certification
contains bluetooth module ISP091201DRegion

Certification

USAFCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea SouthMexico

MSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)11

CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration
has traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. Thisproduct is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp