User’s ManualBluetooth Digital Micrometer 0-30/30-66
型名 : SSM-750/850
Features➢Resolution：➢Protection：IP67➢Ｗｉｒｅｌｅｓｓ Communication：Ｂｌｕｅｔｏｏｔｈ

SSM-750(0-30mm)

SSM-850(30-66mm)

Q-140-1-E ver1.4

1Description
1. Measuring spindle
2. Anvil

8. Battery cover
9. Battery low Indicator

3. Isolation plate
4. MODE button
5. SET button
6. Rotating thimble

10. Hold indicator
11. Data send indicator
12. Locking indicator
13. Bluetooth® indicator

SSM-750(0-30mm)

SSM-850(30-66mm)

A

37.3mm

73.3mmB

27.5mm

43mm

C68mm

74mm

D12.5mm

13.5mm

E63mm

86mm

F181mm

230mm

G

7.2mm

9mm

2

★Installing and replacing the battery
The display of the symbol «B»⑨ indicates the end of the battery life. However there remain still someworking hours.
1. Open the battery cover ⑧ using the accessory(opener) provided
2. Change the Battery (Lithium CR2032 type)
3. Check the rubber protection position
4. Close the battery cover ⑧★Measuring force adjustment

1. Operation①Button
There are two functions, Basic one and Advanced one.
In the case of setting, Data enter.
②ButtonPreset by long press. In the case of setting, Data change.
In the case of measuring, Data transmission.
③Serial communication（Bluetooth/RS232/USB）
It is able to transmit the measuring data and set the mode data by serial communication.
Transmission specification are 4800bps, 7 bits, even parity and 2 stop bits.

2. Start
Push any button.
Refer to Page15 about Bluetooth.

3

３．Basic functions : MODE button⇒Short press( < 1 sec )Short press
HoldLong press
Presetbt
Bluetooth

OnrESEt
ResetOFF

MAC

４．Advanced functions : Mode button Long press( ≻ 1 sec)

UnitMMInchbt.CFGSIMPLE

PAIr4HId５．Maintenance
Keep the micrometer in a dry environment when not using it for a longer period of time to avoid rustformation of the metallic parts.
Do not close the measuring spindle with the anvil when not in use. Keep a distance of 1-2 mm.
Do not use aggressive products (alcohol, trichloroethylene or others) to clean the plastic parts. Do notkeep the micrometer in places which are exposed to sun, heat or humidity.
Important : dry carefully all metal parts of the instrument after effect of moisture to guarantee a perfectmechanical functioning and to avoid rust formation.

６．Switching ON, initializationAfter having pressed on a button, introduced the measuring unit, the instrument shows «SET» for theinitialization of the reference point. Bring back the measuring spindle on the anvil (or on a Gauge block).
Press then [SET]. The instrument is now ready (Check the Preset)

７．Standby
Mode of reduced consumption without loss the reference position. The standby mode is activatedautomaticallyafter 10 minutes of no use. It can also be activated by pressing the [SET] key until extinction of display.
The instrument awakes automatically in case of movement detection on thimble, by a pressure on a button
or when requesting Data.

８．Full Switch off
In order to fully switch off the instrument (minimum power consumption), press the [SET] button until
the message «OFF» appears. At this state no data is maintained and therefore the Reference position islost. The «SET» message will be displayed again during the next use (see “Switching ON”).

９．Description of the menu system
The [MODE] key enables the selection of the different menus (each key stroke activates the followingmenu). The [SET] key enables the activation of a function assigned to the corresponding selected menu.
To avoid a wrong action, each menu is active only during 5s. This period passed, the instrument switches
automatically back to «Measuring Mode» as long as no action has been performed to the [MODE] or[SET] key.

5

10．Preset mode
Preset makes it possible to introduce a reference value different from zero.
Press many times the [MODE] button until menu «PRE» appears. Then press the [SET] button to activate
the function Introduction of Preset. The display shows 00.000 or the last stored value of Preset. The first
digit blinks.
Each push on the [SET] button modifies the value of the digit above the cursor (±, 0..9).
Each push on the [MODE] button moves the cursor to the right. A long press on the [MODE] button
validates the value of Preset and leaves the Preset menu. The instrument automatically returns to«Measuring Mode» when there are no further actions regarding the buttons [MODE] or [SET] for a periodof 15s.

11．Reset, recall of the presetA push of 2s on the [SET] button assigns the Preset-value (or zero) to the current position of themeasuring spindles.

12. Changing unit (mm/inch)
Repeatedly press the [MODE] button to display the [UNIT] menu. Preset then the [SET] button tochange the unit «MM» or «INCH».

13. Locking of the instrument
Press [MODE] until the display shows « LOC». Then press [SET] to lock the instrument.
If the instrument is locked, the function send data (button [SET]) only is active. A pressure of 5s on abutton, unlocks the instrument.

14. HOLD Mode
Press many times the [MODE] key until the menu «HOLD» is displayed. Then press the [SET] key toswitch «ON» or switch «OFF». Press [MODE] key to validate.
During measurement , press the [SET] key to freeze the value. A «H» appears in the display. A 2ndpression on the [SET] key reactivate the dynamic value on display.
This function is useful when the display is not clearly in the field of vision.

6

15. Operation via Bluetooth15.1. HID mode（External Key board mode）

15.3. Bluetooth configuration①Set HID mode by Advanced function.

Display status②Set BT On mode by Advanced function.

off③Reset Bluetooth mode by Basic function.

Operating mode
Bluetooth disconnected

blinking Bluetooth advertising④Pairing connection the instrument to the PC.
（Instrument name：S_Mike PRO HID）
⑤Send the measured data by the DATA button.
15.2. Pair mode
①Set Pair mode by Advanced function.

onBluetooth connected
reset : clear pairing information
MAC : display the MAC address
Simple : profile without pairing②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：SY276）⑤Send the measured data by the DATA button.

Pair : paired and secured profile
HID : virtual keyboard

15.4. Bluetooth Connection :1° Activate Bluetooth compatible software and hardware (Master: PC, Display Unit).
2° Start the instrument. By default the Bluetooth® module is active and the instrument is available forconnection (advertising mode).
3° If no connection is established during the advertisement period reactivate the Bluetooth® moduleusing the bt / On menu.
4° Instrument is ready to communicate (connected mode.)
15.5. Only with paired profile:Pairing with master is automatically done at first connection.
To connect the instrument to a new master (new pairing), pairing information on the instrument must becleared using the bt/ rESEt menu.
15.6. Bluetooth Specifications:

Items

Specification

Frequency band

2.4GHzModulation

GFSK

Max output power

Class3(1mW)

Range≦15m（open space），1-5m(industrial environment)

VersionBluetooth4.*

7

16. Error messages
In case of problems, the following error messages will be sent on the data connection and on the display :- ERR0: Error of the reading sensor- ERR1: Error of parity- ERR2: Error in the received remote command- ERR3: Overflow of the measuring range- ERR8: Memory error- ERR9: Critical error, the instrument must be revised- No Data: Bluetooth® transmission error
The errors 0 and 3 must be confirmed by the button [MODE] or [SET], and then the instrument must bereinitialized.
The Error ‘No Data’ must be confirmed by button [MODE] or [SET]17. Technical specifications
Specification
Items
SSM-750

SSM-850

Measuring range

0-30mm30-66mm

Resolution

1µm

1µm

Measureing force

Ajustable 5N/10N

10N

Indication error±2µm±2µmRepeatabirity

±1µm

±1µm

Probe/Anvil Flatness

0.6µm

0.6µm

Probe/Anvil Parallelism

2µm

2µm

Advance

10mm/rotation

10mm/rotation

Number of refreshments display

8 times/s

8 times/s

Data output

BluetoothBluetooth

Data output parameter

4800bauds,7bits,parity,2stop bits 4800bauds,7bits,parity,2stop bits

Battery life

about 6 months(general using)about 6 months(general using)Working temperature

5～40℃

5～40℃

Storage temperature-10～60℃-10～60℃

Weight270g

425g

IP specification

IP67

IP67

BatteryCR2032

CR2032

818. Description of Bluetooth® moduleThis module is based on Nordic Semiconductor nRF8001 μBlue Bluetooth Low Energy Platform. ThenRF8001 is a single chip transceiver with an embedded baseband protocol engine, suitable for ultra-low
power wireless applications conforming to the Bluetooth Low Energy Specification contained within v4.0of the overall Bluetooth specification.

19.Certification
contains bluetooth module

ISP091201DRegion

CertificationUSA

FCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea SouthMexico

MSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)9CERTIFICATE OF CALIBRATION
We hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration
has traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. Thisproduct is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.

TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp