User’s ManualBluetooth Digital Caliper 150/200/300

Type : SSC-650/750/850

Features➢Measuring Range➢Resolution➢Protection➢Wireless communication：150mm/200mm/300mm：0.01mm：IP67：Ｂｌｕｅｔｏｏｔｈ

Q-139-1-E ver1.4

1Description

SSC-650
Measuring Range：0-150mm。

SSC-750
Measuring Range：0-200mm

SSC-850
Measuring Range：0-300mm

Unit：mm
SSC-650
SSC-750
SSC-850

A234.5290.5
388B405064

C
111414D
1619192E
16x3.516x3.516x4

Depth rod
4x1.4
4x1.4
-Description
Internal Jaw
DATA

Depth RodMODESET

External Jaw★Operation①Button MODE
There are two functions, Basic one and Advanced one.
②Button DATAIn the case of measuring, Data transmission. In the case of setting, Data enter.
SET③ButtonPreset by long press. In the case of setting, Data change.
④Serial communication（Bluetooth/RS232/USB）
It is able to transmit the measuring data and set the mode data by serial communication.
Transmission specification are 4800bps, 7 bits, even parity and 2 stop bits.

[1] Basic functions : MODE button⇒Short press( < 1 sec )Short press
Long press

set the preset valueEnter the presetvalue

Bluetooth mode

reset pairing information

3display the MAC address

[2] Advanced functions : MODE button⇒Long press( ≻ 1 sec)

UnitBluetooth Profile

mminchSimple

Paired and SecureHID(2) not available with Simple profile(3) Active configuration displays first[3] Switching off[4]Favorite setting : DATA button⇒Long Press( ≻ 1 sec)

SET button⇒Long Press( ≻ 2 sec)

(1)Activation/Deactivation
Favorite setting

Set DATA ModeAct.
Deact.

Set “0.00”
StandbySet HOLD Mode

Set SET Mode

Power OffSet PRESET Mode[5] Replacing the battery4[6] Operation via Bluetooth
6.1. HID mode（External Key board mode）6.3.Bluetooth configuration①Set HID mode by Advanced function.
Display status②Set BT On mode by Advanced function.

Operating mode

③Reset Bluetooth mode by Basic function.

off④Pairing connection the instrument to the PC.
（Instrument name：S_Cal EVO HID）

blinking Bluetooth advertisingon⑤Send the measured data by the DATA button.
6.2.Pair mode
①Set Pair mode by Advanced function.

Bluetooth disconnectedBluetooth connected
reset : clear pairing information
MAC : display the MAC address
Simple : profile without pairing②Set BT On mode by Advanced function.
③Reset Bluetooth mode by Basic function.
④Pairing connection the instrument to the PC.
（Instrument name：SY295）⑤Send the measured data by the DATA button.

Pair : paired and secured profile
HID : virtual keyboard

6.4. Bluetooth Connection :1° Activate Bluetooth compatible software and hardware (Master: PC, Display Unit).
2° Start the instrument. By default the Bluetooth® module is active and the instrument is available forconnection (advertising mode).
3° If no connection is established during the advertisement period reactivate the Bluetooth® moduleusing the bt / On menu.
4° Instrument is ready to communicate (connected mode.)
6.5. Only with paired profile:Pairing with master is automatically done at first connection.
To connect the instrument to a new master (new pairing), pairing information on the instrument must be
cleared using the bt/ rESEt menu.
6.6. Bluetooth Specifications:

Items

Specification

Frequency band

2.4GHzModulation

GFSK

Max output power

Class3(1mW)

Range≦15m（open space），1-5m(industrial environment)

VersionBluetooth4.*

5

[7]仕様Items

SSC-650

SSC-750

Resolution(mm)SSC-850

0.01

Measuring Range(mm)

0～150

0～200

0～300

Indication Error(mm)±0.02(0-100mm)±0.03(100-150mm)±0.02(0-100mm)±0.03(100-200mm)±0.02(0-100mm)±0.03(100-300mm)

Repeatabirity(mm)±0.01

Max slider speed(m/sec)

2.5

Display refresh rate(times/sec)

min 10

Data output

BluetoothMean power consumption(μA)

45Battery life

about 6 months(general using)

Woking temperature(℃)

5～40

Storage tempaerature(℃)-10～60Weight(g)175

IP Protection

IP67

BatteryCR2032

Span（without depth rod）(mm)

234.5

290.5

388

External jaw(mm)405064

Internal jaw(mm)161919Depth rod

レレ-[8]Maintenance
Carefully dry all mechanical parts of the instrument after contact with liquids to ensure proper
operation and avoid corrosion.
Don’t use aggressive products (alcohol, trichloroethylene or others) to clean plastic parts.
Don’t expose the instrument to direct sunlight, heat or humidity.

6

[9]Description of Bluetooth® module:This module is based on Nordic Semiconductor nRF8001 μBlue Bluetooth Low Energy Platform. ThenRF8001 is a single chip transceiver with an embedded baseband protocol engine, suitable for ultra-low
power wireless applications conforming to the Bluetooth Low Energy Specification contained within v4.0of the overall Bluetooth specification. The nRF8001, used in the current revision of ISP091201, is a
product using a ROM for the baseband protocol engine.

[10]Certification
contains bluetooth module

ISP091201DRegion

CertificationUSA

FCC ID : 2AAQS-ISP091201

CanadaIC : 11306A-ISP091201

BrazilAnatel : 0516-14-4534

Korea South
MexicoMSIP-CRM-iNs-ISP091201
IFT : RCPSYIS14-0655

Japan

001-A06167

TaiwanCCAH18LP2040T6

EUIndia

WPC : ETA-1003/2-17-/RLO(WR)7

CERTIFICATE OF CALIBRATIONWe hereby certify that this product has been calibrated and found to be in accordance with theapplicable NATIONAL STANDARDS and TECLOCK STANDARDS, Equipment used in this calibration
has traceable accuracy to the NATIONAL LENGTH and FORCE STANDARD.

Notice for use
Be sure to conduct a routine check for this product according to the purpose of use before use. This
product is precision instrument, periodically considering frequency of use, environmental conditions andmethod of use.
It is not guaranteed for the performance of this product, which has been repaired or disassembled byother than TECLOCK.

For appearance and other design improvement, this products
subject to change without advance notice.
TECLOCK Corporation
TECLOCK SmartSolutions Corporation
http://www.teclock.co.jp
HEAD OFFICE
2-10-3 MARUTA-CHO, OKAYA-SHI, NAGANO-KEN 394-0042
PHONE:81-266-22-4912, FACSIMILE:81-266-22-4914
E-mail:teclock@teclock.co.jp