Video NV RecorderSpecification
Accessories

Video IN
Camera I/F
Number of Channels
Frame Rate

Gigabit Ethernet (Camera is required to use specified model)
2ch
100fps × 1ch（ QVGA resolution ：240p ）30fps × 1ch（ HD resolution：720p ）30fps × 1ch（ HD resolution：720p ）* p：progressive

Analog IN
Number of Channels
Coupling
A/D converter
Bandwidth[Hz]Sampling
Sampling[Hz]Frequency Bandwidth[Hz]Sampling[Hz]Analog Output
A/D resolutions
Input Range
Dynamic Range
High Pass Filter
Low Pass Filter
Absolute maximum
input voltage
Inter-channel
phase diﬀerence
ICPTM Curent4ch（ TEDS applicable) Selectable for each ch.
DC / AC / ICP TM
delta-sigma (simultaneous sampling)
40k I 20k I 10k I 5k I 2.5k I 1.25k I 630 I 100（ Bandwidth × 2.4 ）96k I 48k I 24k I 12k I 6k I 3k I 1.5k I 240
40k I 20k I 10k I 5k I 2k I 1k I 500 I 100
102.4k I 51.2k I 25.6k I 12.8k I 5.12k I 2.56k I 1.28k I 256（ Bandwidth × 2.56 ）None
16bit / 24bit selectable
0.01 / 0.0316 / 0.1 / 0.316 / 1 / 2 / 3.16 / 10 [V]
100dB（20kHz bandwidth, 1V rrange, 24bit mode ）OFF / 5Hz（‒12dB/oct）
OFF / 200 / 500 / 1k / 2k [ Hz ]（‒12dB/oct）

AC adapter
Microphone (1 pc.)
Ear-phone (1 pc.)
Installation Manual (1 pc.)
Operation Manual (included in CD)

http://www.teac.eu

Options■

Camera Basler ace series (specified model)■

Recording Media CFast / SDHC card

±50 V
20kHz band and below : within 1 degree
40kHz band and below : within 3 degrees
4mA / 24V
Digital IN / Pulse INGPSCANPulse■■■■■

CFast is a flash memory card with the SATA interface
which has enhanced format of the Compact Flash,and supports a higher maximum transfer rate.

Position data, Time data
All packets data recorded (2 signals monitor available)1ch (Either Pulse or Ext. Trigger)■

Remote Control ER-VRRC

OtherVideo & Analog sync.
Accuracy
View while Recording
Recording START/STOPTrigger
Voice Memo
Recording Media
Synchronization
LCD
Operation
Dimensions / Weight
Power supply

Operating conditions< +/- 1fps ( @ 30fps)
Video (1ch) / Analog Data / CAN
Manual mode / Trigger mode
Level, Timer, Repeat, External
Pre Trigger, Post Trigger
Available
CFast(up to 64 GB) / SDHC( up to 32 GB) *CFast media is required to record video
LX-100 series / WX-7000 series, VR-24 x 2
5.7 inch
Touch Panel + Button
W260 x D186 x H77 [mm] / Approx. 2.3[kg]DC 12 ̶ 16 V
AC 100 ̶ 240 V (when using AC adapter)
Operating temperature/humidity 0 to 40°
C/10 to 80% (no condensation)-20 to 60°
C/5 to 90% (no condensation)
Storage temperature/humidity
Operating air pressure
860 ̶ 1060 hPa
Vibration resistance
MIL-STD-810E Figure 514.4-1, 2, 3

Wired Remote Control Unit ER-VRRC■Battery unit

Battery unit is attached.
* Battery and Battery Charger are not included.

■

Carrying Bag

Recording Time (@ CFast 64 GB)
CameraHD(1280x720 / 30fps)
VGA(640x480 / 30fps)
QVGA(320x240 / 100fps)
̶Analog * 4ch [hours : minutes]
96kHz / 24bit

1ch
1ch
2ch
1ch
̶

48kHz / 24bit

24kHz / 24bit

2:57
6:12
3:34
6:58
23:08

3:09
7:09
3:52
8:12
46:17

2:37
4:53
3:06
5:21
11:34

This table shows approximate times, and record time is different depending on video data.

Other company names and product names in this document are the trademarks or registered trademarks of their respective owners.
Features and specifications are subject to change without notice.
Precaution : To ensure safe handling and operation, read the Instruction Manual before use.

TEAC EUROPE GmbH.

Something is happening in test site that cannot find
only by analog data analysis.

Bahnstr.12, 65205 Wiesbaden-Erbenheim, Germany
Phone : +49-(0)611-7158-0
Fax : +49-(0)611-7158-393
http://www.teac.eu
Email: ipd-sales@teac.eu

Records 2ch video, 4ch analog, CAN, GPS and Pulse in perfect sync.
An All-In-One Data Recorder

Copyright © 2018 TEAC CORPORATION. All rights reserved.

PRINTED IN JAPAN 0318 PDF・ISD-061D-4P-TE

Display Example

The VR-24 answers When, Where, Why and How events heppen.
TEAC has innovated the analog and video recording technologies that have been used for decades in the field of testing environments.

Analog signal can beshown as waveform

CAN, GPS and Pulse

FFT analysis

The VR-24 is a data recorder that can record traditional video/analog signals along with CAN, GPS and Pulse data simultaneously in perfect sync.
Support for wide bandwidth (40kHz). Because of its small lightweight design with battery operation capability, the VR-24 is an ideal stand-alone data recorder for on-site measurements.
These additional data can reveal "hidden details" in your recorded data that you never knew was there.

All-in-one & Stand-alone
An all-in-one unit that records analog, video, CAN, GPSand Pulse data simultaneously in perfect sync.
Recorded video and data can be reviewed
on this screen.

Portable for field use
Small & lightweight design makes it easier to carry.
Superior portability for field use.
*Can be operated on battery power (option).

Smaller than A4/Letter size

Small footprint
Compact design ( W 10 1/4 x D 7 3/8 x H 3 1/8 in)

Light Weight
Approximately 2.3kg /5lbs.

Touch panel

Large 5.7 inch touch panel allows intuitive operation.

Graphically designed top screen andsetting menu provide intuitive
operation. Highly visible display allows
users to check the recorded data
immediately after the measurements.
With a variety of display contents andfull of useful functions, VR-24 is the data
recorder for on-site measurement.

Wide Bandwidth 40kHz

Fulfilling increasing demands for high frequency
noise measurements on EV and turbo chargers
required when downsizing engines.

Optional Software

Optional Software

Displays of synchronous recording data
in a single window

VR-24 now has exciting
new real-time analysis functions!

For Viewing and Analysis

Real-time analysis viewer

Optional VR-View allows you to view signal waveform, video and position data
recorded by VR-24, on a single display. The function of watching synchronized video
and waveforms would greatly help you analyze the phenomena you recorded.

To meet the needs of various industries, the VR-24
now can provide the following analysis screens inaddition to its existing functions:■■■

System configurations

Connected devices and cables are not included with the product.

■■■

Time diagram of four analog channels
Power spectrum and Power spectral density
1/1, 1/3 Octave analysis
FFT spectrogram
Frequency response function
Auto/Cross correlation

Analog device input/ 4chCANGPS[ NMEA0183 ]3-axis accelerometerwith a built-in pre-amplifier

Microphone*Support for ICP TM sensors

Synchronized recording is possible with other TEAC data recorders such as
WX-7000 and LX-100 series, as well as between two VR-24 units.

Pulse

CamerasramF

ctrog
FT spe