HM-TD2037T-25/X
Thermographic Network Box Camera

The Thermographic Network Box Camera with small size, low power consumption and simple integration realizes real-timehigh-accuracy object temperature measurement.

Key Feature● 384 × 288 resolution, high sensitivity sensor● Object temperature range: -20°C to 150°C (-4°F to 302°F), 0°C to 550°C (32°F to 1022°F); Max. (± 2°C, ± 2%)● Temperature measurement at user-defined points, lines, and areas● A highlighted cross at the max. temperature in the thermal image● Thermal image frame rate: 50 fps● Color alarm (isotherm) available for Above, Between, and Below object temperatures

Specification

Thermal Module
Image Sensor

Vanadium Oxide Uncooled Focal Plane Arrays

Resolution

384 × 288

Pixel Pitch

17 μm

Spectral Range

8 μm to 14 μm

NETD< 35 mK (25°C, F# = 1.0)

Focal Length

25 mm

IFOV

0.68 mrad

Field of View

14.8° × 11.2° (H × V)

Min. Focusing Distance

5m

Aperture

F1.0

Measurement and Analysis
Object Temperature Range-20°C to 150°C (-4°F to 302°F), 0°C to 550°C (32°F to 1022°F)

Temperature Accuracy

Max. (± 2°C, ± 2%)

Measurement Presets

Hot spot, cold spot in real time

Point

10 User Presets

Line

1 User Presets

Area

7 User Presets

Color Alarm (Isotherm)

Above/Between/Below

Thermographic Integration
Temperature Metadata

Continuous output data of preset thermographic rules and measured temperature

Thermographic Data Encoding

4 fps, 32-bit 384 × 288 per frame

Thermographic Data Upload
Raw Data Upload

Real-time full-screen thermographic data matrix (16-bit or 32-bit 384 × 288) via RTSP,ISAPI, and SDK
Real-time full-screen YUV data via RTSP and SDK

Video and Image
Main Stream

50 fps (384 × 288)

Video Compression

H.264/MJPEG

Color Palettes

15 options: White Hot, Black Hot, Fusion 1, Rainbow, Fusion 2, Ironbow 1, Ironbow 2,Speia, Color 1, Color 2, Ice Fire, Rain, Red Hot, Green Hot, Dark Blue

Network
Protocols

IPv4/IPv6, HTTP, HTTPS, 802.1x, QoS, FTP, SMTP, UPnP, SNMP, DNS, DDNS, NTP, RTSP,RTP, TCP, UDP, IGMP, ICMP, DHCP

Simultaneous Live View

Up to 3 channels

User/Host level

Up to 32 users, 3 levels: Administrator, Operator, User

Security
ClientUser authentication (ID and PW), HTTPS encryption, IEEE
802.1x (EAP-MD5, EAP-TLS), access control, IP address filtering
HIKMICRO StudioInterface
Alarm Input

1, alarm input (0 to 3.3 VDC)

Alarm Output

1, alarm output; NC, level quantity

Analog Output

1, CVBS output

Communication Interface

1, RJ45 10 M/100 M self-adaptive Ethernet interface
1, RS-485 interface (ModBus available)

GeneralMenu Language

EnglishPower Supply

10 to 30 VDC, Max. 2.3 W

Working Temperature/Humidity

-20°C to 50°C (-4°F to 122°F)
90% or lessProtection LevelIP40MountM3WeightApprox. 140 g (0.33 lb)

Dimensions

74 mm × 45 mm × 45 mm (2.91" × 1.77" × 1.77")

Available Model

HM-TD2037T-25/X

Dimension

COMPLIANCE NOTICE: The thermal series products might be subject to export controls in various countries or regions, including
without limitation, the United States, European Union, United Kingdom and/or other member countries of the Wassenaar
Arrangement. Please consult your professional legal or compliance expert or local government authorities for any necessary
export license requirements if you intend to transfer, export, re-export the thermal series products between different countries.