Technical Specifications
Test menu........................................................ C lotting, chromogenic andimmunological assays
System configurations
Base......................................................... Open-Tube Sampling
CTS........................................................... Closed-Tube Sampling (cap-piercing)
LAS........................................................... L aboratory Automation Systemcompatibility (offboard sampling)
Continuous sample andreagent loading............................................ YES
Continuous operation............................... YES
Samples and reagent
transport system.......................................... Racks
Samples onboard
Base, CTS............................................. 120LAS........................................................... C ontinuous from LAS trackor 90 front-loaded
Reagents onboard....................................... 60 (44 refrig. + 16 room temp.)
Reagents barcode reader........................ YES (integrated)
Barcoded reagents..................................... YES (barcode includes lot,exp. date and vial size)
Cuvettes onboard........................................ 800
Applications onboard................................ 500 (250 user-definable)
Tests/sample.................................................. 30
Throughput
Base	 PT......................................................... up to 360 tests/hr (360 samples/hr)	 APTT................................................... up to 320 tests/hr (320 samples/hr)
PT and APTT ................................. up to 330 tests/hr (165 samples/hr)
CTS	 PT......................................................... up to 270 tests/hr (270 samples/hr)	 APTT................................................... up to 270 tests/hr (270 samples/hr)
PT and APTT.................................. up to 260 tests/hr (130 samples/hr)

Sample predilution.................................... YES
Calibration curve predilution............... YES
Factor parallelism....................................... YES
STAT capability............................................ YES (at any time, in any position)
Rerun and reflex testing......................... YES (configurable)
Reaction-curves display.......................... YES
Quality Control program.......................... YES (with configurable multi-rules)
Results autovalidation.............................. YES
Patient samples results database..... 20,000 samples (configurable)
Security system............................................ YES (configurable)
Events log system....................................... YES
Bidirectional interface............................. YES (with host query)
PC........................................................................ External (dedicated)
Operating system........................................ Windows® XPMonitor.............................................................. 17-inch color LCD touchscreen(external)
Keyboard.......................................................... YES (external)
Mouse................................................................ YES (external)
Printer............................................................... YES (external)
User interface............................................... Windows XP-based
Unit dimensions (w x d x h)Base, CTS............................................. 151 x 76 x 73 cm (64 x 30 x 29 in)
LAS........................................................... 1
 88 x 86 x 162 cm (74 x 34 x 64 in)(includes integrated table and arm)
Weight (analyzer)
Base......................................................... 150 kg (330 lbs)
CTS........................................................... 160 kg (352 lbs)
LAS........................................................... 184 kg (404 lbs)