Capture thefastest processes

ZEISS Axiocam 807 monoYour Fast, 7 Megapixel Microscope Camera forLive Cell Imaging with Large Fields of View
zeiss.com/axiocam807-monoZEISS Axiocam 807 monoYour Fast, 7 Megapixel Microscope Camera forLive Cell Imaging with Large Fields of View
In fluorescence microscopy, the imaging
detector has a significant influence on
the resulting image quality. Improving
signal-to-noise ratio at high imaging
speeds is crucial, particularly whenobserving delicate living cells ororganisms. ZEISS Axiocam 807 monowas developed for exactly this purpose.
With its 17.6 mm diagonal CMOS sensor,
ZEISS Axiocam 807 mono acquires
large fields of view with a single shot.
Capture the surrounding environment of
your sample for additional information
or improve throughput when scanninglarge areas of your sample. The 7
megapixel sensor will resolve the finest

Murine lung tissue with tumor metastasis fixed with 4% PFA and stained for: tumor cells (RFP),

details of your specimen.

macrophages (siglecH-GFP), T-cells (Ly6-G647) and DNA (DAPI). Sample Courtesy of H. IshikawaAnkerhold, Walter-Brendel-Zentrum für Experimentelle Medizin München, Germany

The frame rate of 73 images per second

Like other ZEISS Axiocam cameras

at full sensor resolution resolves evenwith actively stabilized sensors,

the fastest processes and can be

ZEISS Axiocam 807 mono reaches afurther improved by binning or using astable temperature within seconds

sub-array of the sensor. Distortion-free

after microscope start-up, delivering

imaging is guaranteed by the global

reproducible results immediately.

shutter technology. In combination

Because of its hardware triggeringwith its high peak quantum efficiency

capabilities, the camera can be used inof 78 % and low readout-noise, highcomplex setups with many accessories.

signal-to-noise ratios even at low light

ZEISS Axiocam 807 mono is the ideal

conditions are ensured.

choice for life science applications
requiring fast, low light imaging ofZEISS Axiocam 807 mono uses a dual

large fields of view or efficient scanningUSB 3.0 interface for data transfer.

of large sample areas.

In contrast to other proprietaryconnections, this standardized
interface provides stable and fast data
rates for reliable image acquisition.

2

Recommended for:• Fast imaging of dim fluorescent signals
with good signal-to-noise ratio• Fast tile scanning applications• Dynamic events in live cell imaging• Flexible setups with varyingapplications

Optical section of mitotic cells created with ZEISS Apotome. Cells have DNA stained with Hoest33342,
Aurora B with Alexa 488 und Tubulin with Alexa 568

Simpler. More intelligent. More integrated.
• 7 megapixel CMOS sensor with global shutter technology• 4.5 micron pixels for optimal resolution• Large sensor with 17.6 mm diagonal for extended• Global shutter architecture for distortion-free images

field of view

• Reproducible image quality due to active thermal• Wide sensitivity spectrum from 350 nm to 1000 nm

stabilization of the sensor• 73 full-resolution images per second• Robust, very fast and easy-to-use dual USB 3.0 connection• High-quality noise inhibition technology and 78 % sensor• Hardware triggering

quantum efficiency for lowlight imaging

Quantum Efficiency ZEISS Axiocam 807 mono
100,000
90,00070,00060,000
50,00040,00030,00020,00010,000
0,000
400
420
440
460
480
500
520
540
560
580
600
620
640
660
680
700
720
740
760
780
800
820
840
860
880
900
920
940
960
980
1000

Quantum Eﬃciency [%]80,000

Wavelength [nm]3Technical Data and Conformity

FeatureValue

Sensor TypeSony CMOS image sensor, Global Shutter architecture
Image Diagonal 17.6 mm, equivalent to 1.1" Sensor Format
Image Field (14.5 mm × 9.9 mm)

Sensor Pixel Count

3216 (H) × 2208 (V) = 7.1 Megapixel

HW Subsampling 2×1608 (H) × 1104) (V) = 1.8 Megapixel, high speed full view mode

Pixel Size

4.5 μm × 4.5 μm

Bit Depth

14 bit / 12 bit or 8 bit/pixel

Exposure Range

0.1 ms up to 60 s

Gain

1×, 2×, 4×, 8×, 16×Binning

1 × 1, 2 × 2, 3 × 3, 4 × 4, 5 × 5 (combined analog and digital binning)

Dark Current Signal

0.3e- / p / s at 25 °C sensor temperature

HDR

Reduced readout noise at Gain 1× for best combination of sensitivity and high intensity levels in one frame.

Cooling System

Active cooling, regulated sensor temperature 25 °C

Spectral Sensitivity

Approx. 350 nm – 1000 nm, protection glass (coated)

Interfaces

Dual USB 3.0

Trigger Port

Connector for trigger cable: Trigger-in, trigger-out, readyPower Supply

By USB 3.0 connections, power consumption 7 W max.

Operation System

Win 10 ×64 Enterprise

Software

ZEN 3.6 (blue edition) and higher, ZEN core 3.5 and higher

Image Enhancement Functions

Denoising, sharpening, shading correction, dark current compensation

Automatic Feature

Optional automatic exposure time adaption

Optical Interface

C-Mount

Dimensions and Weight

10.8 cm × 7.8 cm × 6.1 cm / 580 g

Order Number

ZEISS Axiocam 807 mono: 426560-9160-000

Frame RateFPSLive Image> 303216 × 2208

731602 × 1104

260 (2 × 2 subsampling)

1920 × 1080

145

1024 × 1024151

512 × 512

282

1920 × 256

487

1920 × 128

506

Read Noise (gain)

Full Well Capacity

Dynamic Range

5.7 e- (1×)

25,000 e-

4,420:1< 4.6 e- (2×)

12,500 e-

2,730:1< 3.9 e- (4×)

6,250 e-1,610:1< 3.4 e- (8×)

3,125 e-930:1

2.9 e- (2.9×)

1,560 e-530:1

4.0 e- (HDR mode)

25,000 e-

6,230:1

ISO 13485:2016

Carl Zeiss Microscopy GmbH07745 Jena, Germany
microscopy@zeiss.com
www.zeiss.com/axiocam807-mono

Not for therapeutic use, treatment or medical diagnostic evidence. Not all products are available in every country. Contact your local ZEISS representative for more information.
EN_40_012_140 | CZ 09-2022 | Design, scope of delivery, and technical progress subject to change without notice. | © Carl Zeiss Microscopy GmbH

Mono and color versions
Sensor Size