ＭＯＤＥＬ：PI-Series

Nissin Pulse Electronics Co., Ltd.

Overview
Our High-Voltage Test System [PI-Series] has been developed for use in school education
to suit experiments and practical training.
AC, DC, and impulse voltage tests can be performed with this test system.
The test system is used for destructive testing of samples (insulators, cables, etc.)
and withstand voltage testing.
This test system incorporates many features that could not be achieved with conventional
test systems, emphasizing speed-up of experiments and accuracy of data.
This test system is also equipped with safety devices to ensure safety.
The test system can be operated from the operation panel or remotely from a PC.

Features

Operation from touch panel and remote operation from PC.
Various tests can be controlled from the operation panel or PC.

Both control screens have

The communication control method between the PC and operation panel uses opticalEthernet.
This maintains high insulation and allows safe remote control.
In addition, this PC can be used with general-purpose products, so it is very flexible.

 User interface・High operability is achieved using digital technology.

・The impulse generator uses a self-destructive starting method to achieve stabledischarge.

・AC chart mode allows you to display the output voltage in real time（For AC MODE）

・

Measured data can be saved as a log file on the touch panel’s SD card.

・Oscilloscope waveforms can be displayed and saved on the PC display.



Various test modes Impulse voltage generation test
・

SINGLE Mode: Charge and discharge for one shot.

・MULTI

Mode: Charge and discharge for multiple time(1 to 10 times) AC voltage generation test
・MANUAL

・PROGRAM Mode: Automatic AC voltage generation test(1 to 10 times multiple steps)

Mode: Manual AC voltage generation test

 DC voltage generation test
・MANUAL

Mode: Manual AC voltage generation test

・

PROGRAM Mode: Automatic AC voltage generation test(1 to 10 times multiple steps)

MULTI Mode for Impulse test

PROGRAM Mode for AC test

PROGRAM Mode for DC test

Configuration contents ＜PI-Series for standard＞
AC test system

0～50kV・3kVA

DC test system

0～±50kV・5mA
200kV

Impulse test

300kV
400kV

High voltage probe

DC.100kV,IMP120kV.p,

TYPE: EP-100K

AC35kV rmsSphere gap
Impulse Voltage GeneratorΦ100mm, Max. 170kV.p

TYPE: G-100HS

Sphere Gap

Highly safe

・ Equipped with various interlock circuits
・ A revolving light warns the surrounding area of high voltage conditions.

Emergency stop

Key switch

Light sensor

Discharge device

Revolving lightClassification

Unit Name

Specification

200

Type of PI-Series
300

400

○
○
○
▲
○
○
○
▲

○
○
○
▲
○
○
○
▲

○
○
○
▲
○
○
○
▲

○
○
○
○
○
○

○
○
○
○
○
○

○
○
○
○
○
○

○－－○
○
▲
▲
▲
○
▲
▲
▲
○
▲
○
○
○
○
○
○○▲
▲：Option

－○－○
○
▲
▲
▲○▲
▲▲○▲
○○
○○○○
○▲－－○○
○▲
▲▲○▲
▲
▲○▲
○○○
○
○
○○
▲

Control system
Control Unit
Measuring
Equipment

Control Rack

PLC Control Unit
DC Generator Unit
Oscilloscope
Peak
Ethernet Module
Storage Rack andPower Supply

PHM-1023

PLC Control, touch panel
DC.±50kV, ±5mA
1GS/s
1.25GS/s±0.1kV～1999.9kV

CB-200F-3W
CB-200F-5W

Remote system

Computer SystemPC
MonitorPrinter
Control Software
Simulation SoftDesk

18.5inch Monitor
A4 size Ink jet printerTest SystemImpulse Voltage
Generator

Impulse Voltage
Generator

IVG-200K025W
IVG-300K025W
IVG-400K025W

Discharge device
AC Voltage
Generator

High Voltage
ProbesTest Transformer
High Voltage Probe(cable lengs:10m)
Polarity Change Unit
Video Camera
Sphere Gap

Standard Electrode
OthersInsulator Tester
Suspension Insulator
Dummy Doll
Insulator Material

TT-50K60S
TT-50K100S
TT-100K50S
EP-50K
EP-100K
EP-150KP
PX-50W
G-100HS
G-150HS
Plate
NeedleStick
Oil Cup
ZSK-50W
5pc

200kV 1.25kJ
300kV 1.87kJ
400kV 2.5kJ
Built into the main body
50kV, 3kVA
50kV, 5kVA
100kV, 5kVA
DC.50kV IMP.70kV.p
DC.100kV IMP.120kV.p
DC.100kV IMP.145kV.p
DC.±50kV
Video Camera
φ100mm 170kV
φ150mm 249kV
φ100mm 2pc
φ20×120mm 2pc
φ12.5×120mm 2pc
Oil Cup 64×68×Depth63mm
Air / Oil combined unit
5pc
Hight about 180cm
Glass/FRP/Bake/Kraft,other○：Standard■Caution●The contents of this catalog is as of November, 2023.
●The company name, product name and logo are registered trademark of the company.
●The detailed contract scope will be determined individually.

Address

・Head office & Main Factory2744-3, Yamazaki, Noda, Chiba 278-0022, Japan
Tel+81 4 7123 0611 Fax+81 4 7123 8831

・Kansai Office
47, Umezu-Takase-cho, Ukyo-ku, Kyoto 615-8686, Japan
Tel +81 75 864 8912 Fax+81 75 864 8934