Maternity Model
Type IILM-043N-CE
Instruction Manual
Thank you for purchasing the Maternity Model Type II.
Please read this instruction manual carefully to ensure correct use of the product,and store it in a safe place for easy access.

Manufacturer:

KOKEN CO., LTD.
1-4-14 Koraku, Bunkyo-ku, Tokyo 112-0004 JAPANwww.kokenmpc.co.jp

EU Authorized Representative for KOKEN CO., LTD.

EC REPMDSS GmbH
Schiffgraben 4130175 Hannover, Germany
www.mdssar.com© 2015-2023 KOKEN CO.,LTD. C-1-1014-2-02-01

Handling and Safety Precautions
These precautions should be strictly observed in order to ensure safe, long-term
use of the product.
The following precautions should be observed particularly strictly:
1. The skin is made of a special silicone with a texture that is similar to human skin, and the abdominal
wall piece is made with a special material that is soft to the touch. Please handle these models withthe same care that you would exercise when working with a human body. Applying excessive force
may damage the model.
2. The material of the abdominal wall piece has a distinct odor. This odor is not harmful to the humanbody and does not indicate poor quality. The distinct odor fades over time. There may also be a smellof baby powder remaining on the model, as baby powder is applied during fabrication.
3. The abdominal wall piece may become sticky due to the oil component. Always place it correctly inthe protective sheet (provided as an accessory) before storage. If it becomes too sticky after use orafter washing, it may be difficult to set the skin in place. In these cases, apply baby powder (providedas an accessory) liberally to the abdominal wall piece.
*See “Caring for the Model” on page 10.
4. The skin may become sticky from the oil component in the abdominal wall piece. If this happens,wash the skin in water to remove the oil before use.
To prevent the oil from getting onto the skin, always place the abdominal wall piece correctly in theprotective sheet (provided as an accessory) before storage.
*See “Caring for the Model” on page 10.
5. The abdominal wall piece may change its shape over a prolonged period. Check the shape of thepart about three years after purchase. If it is deformed to the extent that its function is affected,
consider replacing it.
6. Air bubbles or impressions in the abdominal wall piece that may remain from the fabrication process
do not affect the strength of the material, as the material used is tear-resistant.
7. Do not use oil-based inks, pencil, chalk, or paints on the model because they may cause permanent
staining.
8. Please do not use oily lubricant (baby oil, olive oil etc.), alcohol lubricant, and lubricant containing
organic solvent. These may cause breakage of the model.
9. In caring for the model, do not use rubbing alcohol, thinners, or benzin or other organic solventson the model, as doing so may cause damage.

1

10. Do not use the model outdoors or in damp places; doing so may damage the model.
11. Placing objects directly on top of the model or placing it on uneven surfaces may cause
deformation.
12. An excessive amount of air injected into the amnion may damage the amnion. When injecting air,palpate the abdomen of the model to check the tension in the amnion.
13. When setting the skin on the body, the skin may have wrinkles or sags due to the construction of
the model. Please note that it may not always be possible to completely remove these wrinkles andsags.
14. Storing the model for an extended period in a high or low temperature environment or under
exposure to direct sunlight may cause the oil component in the abdominal wall piece to ooze out,
causing deformation or discoloration of the model. If possible, store the model at 10-50 °C, awayfrom direct sunlight.
15. Take care to avoid touching the internal parts through the gap between the panel and the body;doing so may cause the internal electronic device to malfunction.

16. This product is intended for use in standard operating environments. Do not use it in heavy industrial
settings.
17. This model uses urethane foam for some of its parts, and may be susceptible to damage from pests
that are attracted to urethane foam (such as insects in the ant family).
When storing this model, ensure that thorough measures are taken to prevent pest damage before
storing. The company will not be involved with or held liable for pest damage that occurs duringstorage.

18. About the AC Adapter
This model does not come with an AC adapter.
Please use an AC adapter with the following specifications.		

EMC Directive			

EMS: EN61000-6-1 or EN61000-6-2
EMI: EN61000-6-3				

or EN61000-3-2 and EN61000-3-3				

and EN55032 class B or EN55011 group1 class B

2

LV Directive:

EN62368

RoHS Directive:

EN IEC 63000

Rated Output :15VDC 1.2A

Rated Input :AC100 - 240V 50/60Hz

Plug Shape :Inner diameter ø2.1mm/Outer diameter ø5.5mm
Center minus

1. The contents of this instruction manual are subject to change without notice.
2. This instruction manual may not be reproduced in part or in its entirety without permission.
3. Please contact the manufacturer in the event that any errors or omissions are found in the contents ofthis instruction manual.
4. This product should be used only as described in this instruction manual. In particular, the productshould not be used in any way that contravenes the precautions noted in the instruction manual.

3

Table of Contents
1. Outline and Features
2. Components and Configuration
3. Setting up the Model
3-1. Attach the fetus in amnion
3-2. Attach the abdominal wall piece
3-3. Attach the skin3-4. Turn on the power
4. Caring for the Model
5. Specifications

41. Outline and Features
Outline
The four types of Leopold’s Maneuvers, fetal heartbeat monitoring, and abdominal
measurement can be practiced. The abdominal wall piece is made of soft, stretchy material,
and the abdomen can be inflated with air as a substitute for amniotic fluid; thus, reproducing ahighly realistic sensation. An internal synthesizer reproduces a realistic fetal heartbeat sound.
Volume and speed are adjustable.

Features
1. The trunk skin and fetus material are made of a special silicone rubber with an appearance andtexture that closely simulates human skin, for realistic practice.
2. The model contains a pelvis that enables realistic palpation to check the pubic symphysis andthe ilium during uterine fundal height measurement and external pelvimetry, with a sensationthat closely resembles an actual human body.
3. The amnion can be filled with air to make handling and control easier.
4. An internal synthesizer enables auscultation of a realistic fetal heartbeat sound. The volume andspeed are freely adjustable.
5. Traube stethoscope monitoring of the fetal heartbeat can be practiced. The heartbeat sound canalso be emitted from a speaker on the side of the panel.

52. Components and Configuration①②③④⑤⑥⑦⑧smalllarge①②Components
B ody

QuantitySkin

1
1Fetus in amnion

1③

Abdominal wall piece⑤

Two-way stopcock (with Luer ﬁtting)1⑦

Birth canal pads (1 small, 1 large)

2

④
⑥⑧

Rubber bladder
Baby powder

Abdominal wall protective sheet
Storage bag

Instruction Manual (this document)This model does not come with an AC adapter.
Please see page 2 about the AC adapter.

6111

1

1

1

3. Setting up the Model
3-1. Attach the fetus in amnion
1) Remove the main body from the storage bag and remove the skin.

Abdominal wall
protective sheet
Abdominal wall
protective sheet
Abdominal wall
protective sheet2) Remove the abdominal wall piece that is wrapped in the abdominal wall protective sheet andthe fetus in amnion from the body; then, place a birth canal pad as shown below. 			
No birth canal pad ... Seitz method (-)
		* Birth canal pad (small) ... normal		Birth canal pad (large) ... Seitz method (+)
Fetal heartbeat cable
Fetal heartbeat cable
Fetal heartbeat cable

Birth canal pad
Birth canal pad
Birth canal pad3) As shown in the illustration, thread the air feed tube through the panel from the inside of thebody to outside the body, and set the fetus in amnion in the body.		 *The position and orientation of the fetus in amnion can be readily adjusted. When setting
it in the breech presentation, place the air feed tube in the groove on the inside of the		 body so that it does not interfere.

GrooveAir feed tube

GrooveAir feed tube

GrooveAir feed tube74) Attach the two-way stopcock to the air feed tube. Then, attach the rubber bladder to thetwo-way stopcock.

3-2. Attach the abdominal wall piece1) Take the abdominal wall piece out of the abdominal wall protective sheet and place it on thebody with the fetus in amnion in place. Place it in the body so that the “Upper part” and“Lower part” marks on the abdominal
wall piece aresidepositioned as shown below.
Chest

Chest side
Chest side

Leg sidewall piece

Leg side
Leg side

Abdominal wall pieceAbdominal wall piece

Upper part
Upper part
Lower part
Lower part2) Anchor the abdominal wall piece onto the fixation pins on the body. Four fixation pins aresupplied.

Near the left and right armpsFixation
pin
NearNear
the thesolar
plexussolar
plexus

Near
the
Near
theleft
leftandandright
rightarmpit
armpitFixation
pinpinFixation8- 8 -

Near
the groin
groin
Near the3-3. Attach the skin1) After anchoring the abdominal wall piece to the fixation pins, place the skin over the mainbody.		 *Do not secure the hook and loop fasteners on the sides at this time.

2) Open the two-way stopcock.

3) Twist the valve on the rubber bladder clockwise to close it, and fill the rubber bladder with air.

[Caution]
When an excessive amount of air is injected into the amnion, the amnion may burst. Wheninjecting air, feel the abdomen of the model to check the tension in the amnion.
[REFERENCE]
As the volume of air in the amnion becomes larger, the situation where palpation is moredifficult can be simulated. Using a smaller volume of air simulates the situation wherepalpation is easier. Adjust to suit the situation.

9C adapter jack5) After adjusting the volume of air in the amnion, close the two-way stopcock to prevent air
from leaking out of the amnion.
*If this step is skipped, air will slowly leak out of the amnion over time.

6) Secure the hook and loop fasteners in four places on each side of the skin.
Smooth down the skin vertically and horizontally to push wrinkles and sags on the abdomen
outwards. Pull the skin taut and secure the hook and loop fasteners.		 *Depending on the volume of air injected, it may not always be possible to completely 		
remove these wrinkles and sags.

3-4. Turn on the power
1) Connect the AC adapter *connector to the AC adapter jack on the panel. Plug in the cord, andpress the power switch on the panel.
Power switch

Power switch* See page 2Power switch

AC adapter jackAC adapter jack2) Adjust the fetal heartbeat volume with the VOLUME dial on the panel, and the speed of thefetal heartbeat with the HEART RATE dial. The SPEAKER switch is used to switch emissionof the fetal heartbeat sound to an external speaker.

104. Caring for the Model(1) Remove the air from the amnion before storing.
To remove air, open the two-way stopcock, and twist the valve on the rubber bladdercounterclockwise to loosen it.

(2) Wrap the abdominal wall piece in the abdominal wall protective sheet provided as anaccessory, then place it back on the main body and place the skin over the top to store.
*1. The abdominal wall protective sheet prevents the oil component from oozing out of theabdominal wall piece to the skin and other parts.
*2. If the oil component contained in the abdominal wall piece oozes out to the skin and		 other parts, it will promote deterioration of the product. Be sure to wrap the abdominal		 wall piece in the abdominal wall protective sheet for storage.
*3. If the abdominal wall piece becomes sticky, apply the baby powder provided as an 		accessory generously to the piece.

(3) If the model becomes dirty, wipe it with a moist gauze or similar soft material using a neutral
solvent diluted with water.
*Do not clean it with rubbing alcohol, thinners, benzene, or other organic solvents.
(4) Place the model in the proper orientation when storing to prevent deformation.
(5) Storing the model for an extended time in a high or low temperature environment, or under
exposure to direct sunlight, may cause the oil component in the abdominal wall piece to ooze
out, causing deformation or discoloration of the model. If possible, store the model at 10-30°C,
away from direct sunlight.
(6) Place the model in the storage bag provided as an accessory when storing.

11

5. Specifications
Product name

LengthWidth

HeightWeightMaternity Model Type II Main Body

Approx. 35 cm

Approx. 52 cm

Approx. 23 cm

Approx. 11 Kg

Storage bag

Approx. 35 cm

Approx. 63 cm

Approx. 35 cm

Approx. 1 Kg

12