LOW TEMPERATURE & HUMIDITY CHAMBER

SE-70 to +100°C(+180°C) / 20 to 98%RH

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temp. / Humid. range
Fluctuation
Variation in space
Temperature rise time
Temperature fall time
Lowest temperature
CONFIGURATIONController
Internal material
External material
Insulating material
REFRIGERATING SYSTEM
Refrigerator
Out put
Power ability changing
Cooling coil
DRY HEATERWET HEATERBLOWER FANACCESSORIESSAFETY UNITSWATER SUPPLYING SYSTEM
Water tank
OPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)Silvery Emperor seriesSE(SSE)-73KT
SE(SSE)-74KT
SE(SSE)-75KT
SE(SSE)-77KT
SE(SSE)-78KT
Highest temperature of SE-type is +100°C (SSE-type is +180°C)
AC 200V ±10% 3φ 50 / 60Hz
40A
40A
50A
60A
75A
Balanced Temperature and Humidity Control system (BTHC system)
Ambient: +10 to +30°C, 20 to 80%RH-70 to +100°C(+180°C) / 20 to 98%RH±0.3°C(-70 to +100°C), ±1.0°C(+100 to +180°C) / ±2.5 %RH
2.0°C(-70 to +100°C), 4.0°C(+100 to +180°C) / 5.0 %RH-70 to +100°C within 60 min. (-70 to +180°C within 90 min.) at refrigerator OFF+20 to -70°C within 90 min.
-70°C
KATO KT instrumentation
18Cr-8Ni stainless steel plate, grinding finished
18Cr stainless steel plate, line finished
Glass wool / Silicone packing
Dual condensing refrigerating system : Air cooling or Water coolingComplete closed type compressor
1.5 / 1.5 kW
2.2 / 2.2 kW
2.2 / 2.2 kW
2.2 / 2.2 kW
3.0 / 3.0 kW
Multi step automatic switching system by micro computer control
Plate fin cooler : Corrosion prevention coating finishes
stainless steel sheath heater
stainless steel plug heater
Sirocco fan
Observation window : W250×H350mm with Heating wire, Cable hole 30φ×1pc,
Adjustable shelf ×2 sets, Halogen room lamp, Caster, Adjuster
Earth leakage circuit breaker, Boiling without water prevention unit,
MIxing motor overload relay, Overload and short circuit protector for SSR,
Overload and short circuit protecting fuse for operating circuit,
Phase reverse relay, Water pressure switch for water cooling type,
Refrigerator overload relay, Alarm of temperature limit error,Overheat prevention unit, Refrigerator pressure switch
Water pump-up with natural gravitation method or Supply from waterworks
fixed tank on the chamber 1 set
Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS-232C,
Excessive cooling preventive unit, Deionizer for humidifying water, Gauze wick,
Tank with automatic pump-up function, See-through door, Time signal unit, Defroster
175288441
800
1000
500× 700× 500

600× 800× 600

700× 900× 700 1000×1000× 800 1000×1000×1000

1110×1600× 975 1260×1760×1075 1330×1860×1175 1730×2075×1275 1730×2075×1475
400450*The performance values are based on JTM K-09, IEC60068-3-6:2001,
and no specimen inside the test area.
*The rising and falling performances are given at an ambient temperature of+20°C, or with a cooling water temperature of +25°C.
*Humidifying water quality should be PH 6.0 to 7.2, electricalconducting rate is less than 0.02mS/cm.
*The descriptions, inside the( ) in the table, are for the +180°C
specification SSE model.
*The above specifications may be changed for any improvement of
product without giving any preliminary notification.

530700760

LOW

TEMPERATURE

SE&HUMIDITYCHAMBER-40 to +100°C(+180°C) / 20 to 98%RH

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temp. / Humid. range
Fluctuation
Variation in space
Temperature rise time
Temperature fall time
Lowest temperature
CONFIGURATION
Controller
Internal material
External material
Insulating material
REFRIGERATING SYSTEM
Refrigerator
Out put
Power ability changing
Cooling coil
DRY HEATERWET HEATERBLOWER FANACCESSORIESSAFETY UNITSWATER SUPPLYING SYSTEM
Water tank
OPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)Silvery Emperor seriesSE(SSE)-43KT
SE(SSE)-44KT
SE(SSE)-45KT
SE(SSE)-47KT
SE(SSE)-48KT
Highest temperature of SE-type is +100°C (SSE-type is +180°C)
AC 200V ±10% 3φ 50 / 60Hz
30A
30A
30A
50A
50A
Balanced Temperature and Humidity Control system (BTHC system)
Ambient: +10 to +30°C, 20 to 80%RH-40 to +100°C(+180°C) / 20 to 98%RH±0.3°C(-40 to +100°C), ±1.0°C(+100 to +180°C) / ±2.5 %RH
2.0°C(-40 to +100°C), 4.0°C(+100 to +180°C) / 5.0 %RH-40 to +100°C within 45 min. (-40 to +180°C within 80 min.) at refrigerator OFF+20 to -40°C within 60 min.
-40°C
KATO KT instrumentation
18Cr-8Ni stainless steel plate, grinding finished
18Cr stainless steel plate, ine finished
Glass wool / Silicone packing
Single condensing refrigerating system : Air cooling or Water coolingComplete closed type compressor
1.5 kW
1.5 kW
1.5 kW
2.2 kW
2.2 kW
Multi step automatic switching system by micro computer control
Plate fin cooler : Corrosion prevention coating finishes
stainless steel sheath heater
stainless steel plug heater
Sirocco fanObservation window : W250×H350mm with Heating wire, Cable hole 30φ×1pc,
Adjustable shelf ×2 sets, Halogen room lamp, Caster, Adjuster
Earth leakage circuit breaker, Boiling without water prevention unit,
MIxing motor overload relay, Overload and short c ircuit protector for SSR,
Overload and short circuit protecting fuse for operating circuit,
Phase reverse relay, Water pressure switch for water cooling type,
Refrigerator overload relay, Alarm of temperature limit error,
Overheat prevention unit, Refrigerator pressure switch
Water pump-up with natural gravitation method or Supply from waterworks
fixed tank on the chamber 1 set
Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS-232C,
Excessive cooling preventive unit, Deionizer for humidifying water, Gauze wick,
Tank with automatic pump-up function, See-through door, Time signal unit, Defroster
175
288
441
800
1000
500× 700× 500

600× 800× 600

700× 900× 700 1000×1000× 800 1000×1000×1000

1020×1650× 950 1120×1650×1050 1220×1750×1150 1535×1910×1250 1535×1910×1450
320370*The performance values are based on JTM K-09, IEC60068-3-6:2001,
and no specimen inside the test area.
*The rising and falling performances are given at an ambient temperature of+20°C, or with a cooling water temperature of +25°C.
*Humidifying water quality should be PH 6.0 to 7.2, electricalconducting rate is less than 0.02mS/cm.
*The descriptions, inside the( ) in the table, are for the +180°C
specification SSE model.
*The above specifications may be changed for any improve ment ofproduct without giving any preliminary notification.

420670720

LOW

TEMPERATURE

SE&HUMIDITYCHAMBER-25 to +100°C(+180°C) / 20 to 98%RH

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temp. / Humid. range
Fluctuation
Variation in space
Temperature rise time
Temperature fall time
Lowest temperature
CONFIGURATIONController
Internal material
External material
Insulating material
REFRIGERATING SYSTEMRefrigerator
Out put
Power ability changing
Cooling coil
DRY HEATERWET HEATERBLOWER FANACCESSORIESSAFETY UNITSWATER SUPPLYING SYSTEM
Water tank
OPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)Silvery Emperor seriesSE(SSE)-23KT
SE(SSE)-24KT
SE(SSE)-25KT
SE(SSE)-27KT
SE(SSE)-28KT
Highest temperature of SE-type is +100°C (SSE-type is +180°C)
AC 200V ±10% 3φ 50 / 60Hz
20A
30A
30A
40A
50A
Balanced Temperature and Humidity Control system (BTHC system)
Ambient: +10 to +30°C, 20 to 80%RH-25 to +100°C(+180°C) / 20 to 98%RH±0.3°C(-25 to +100°C), ±1.0°C(+100 to +180°C) / ±2.5 %RH
2.0°C(-25 to +100°C), 4.0°C(+100 to +180°C) / 5.0 %RH-25 to +100°C within 45 min. (-25 to +180°C within 80 min.) at refrigerator OFF+20 to -25°C within 60 min.
-25°C
KATO KT instrumentation
18Cr-8Ni stainless steel plate, grinding finished
18Cr stainless steel plate, line finished
Glass wool / Silicone packing
Single condensing refrigerating system : Air cooling or Water coolingComplete closed type compressor
0.75 kW0.75 kW
1.1 kW
1.1 kW
1.5 kW
Multi step automatic switching system by micro computer control
Plate fin cooler : Corrosion prevention coating finishes
stainless steel sheath heater
stainless steel plug heater
Sirocco fanObservation window : W250×H350mm with Heating wire, Cable hole 30φ×1pc,
Adjustable shelf ×2 sets, Halogen room lamp, Caster, Adjuster
Earth leakage circuit breaker, Boiling without water prevention unit,
MIxing motor overload relay, Overload and short circuit protector for SSR,
Overload and short circuit protecting fuse for operating circuit,
Phase reverse relay, Water pressure switch for water cooling type,
Refrigerator overload relay, Alarm of temperature lim it error,
Overheat prevention unit, Refrigerator pressure switch
Water pump-up with natural gravitation method or Supply from waterworks
fixed tank on the chamber 1 set
Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS-232C,
Excessive cooling preventive unit, Deionizer for humidifying water, Gauze wick,
Tank with automatic pump-up function, See-through door, Time signal unit, Defroster
175288441
800
1000
500× 700× 500

600× 800× 600

700× 900× 700 1000×1000× 800 1000×1000×1000

1020×1650× 950 1120×1650×1050 1220×1750×1150 1520×1910×1250 1520×1910×1450
310370*The performance values are based on JTM K-09, IEC60068-3-6:2001,
and no specimen inside the test area.
*The rising and falling performances are given at an ambient temperature of+20°C, or with a cooling water temperature of +25°C.
*Humidifying water quality should be PH 6.0 to 7.2, electricalconducting rate is less than 0.02mS/cm.
*The descriptions, inside ( ) in the table, are for the +180°C
specification SSE model.
*The above specifications may be changed for any improvement of
product without giving any preliminary notification.

400600650