MIDDLE SIZE TEMPERATURE CHAMBER (INDUSTRIAL OVEN)

RO(Ambient temp. +30°C) to +400°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)RO series
RO-41DPA

RO-42DPA

30A

40A

RO-43DPA
RO-44DPARO-45DPA
Highest temperature is +400°C
AC 200V ±10% 3φ 50/60Hz40A
40A50AHot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

RO-46DPA

60A(Ambient temp. +30°C) to +400°C±0.7°C(at +200°C), ±0.8°C(at +400°C)
3.0°C(at +200°C), 6.0°C(at +400°C)(Ambient temp. +30°C) to +400°C within 90 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
6.0 kW
9.0 kW
9.0 kW
9.0 kW
12.0 kW
16.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
288
512
640
1000
1200
1500
600×800×600 800×800×800 800×1000×800 1000×1000× 1000×1200× 1000×1500×1000
1000
1000
1200×1600× 1450×1600× 1450×1800× 1650×1800× 1650×2000× 1650×2300×1350
1600
1600
1800
1800
1800
400
500
650
730
800
850

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

M I D D L E SI Z E TE M P E R AT U R E C H A M B E R (I N D U S T R I A L O V E N )RO(Ambient temp. +30°C) to +300°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATIONController
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)RO series
RO-31DPA

RO-32DPA

20A

30A

RO-33DPA
RO-34DPA
RO-35DPA
Highest temperature is +300°C
AC 200V ±10% 3φ 50/60Hz
30A
40A
40A
Hot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

RO-36DPA

50A(Ambient temp. +30°C) to +300°C±0.6°C(at +150°C), ±0.8°C(at +300°C)
2.3°C(at +150°C), 4.5°C(at +300°C)(Ambient temp. +30°C) to +300°C within 60 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
4.0 kW
6.0 kW
6.0 kW
9.0 kW
9.0 kW
12.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement2885126401000
1200
1500
600×800×600 800×800×800 800×1000×800 1000×1000× 1000×1200× 1000×1500×1000
1000
10001150×1500× 1350×1500× 1350×1700× 1550×1700× 1550×1900× 1550×2220×1250
1500
1500
1700
1700
1700
350
450
600
680
750
800

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

M I D D L E SI Z E TE M P E R AT U R E C H A M B E R (I N D U S T R I A L O V E N )RO(Ambient temp. +20℃ ) to +200℃

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)RO series
RO-21DPA

RO-22DPA

15A

20A

RO-23DPA
RO-24DPA
RO-25DPA
Highest temperature is +200℃
AC 200V ±10% 3φ 50/60Hz
20A30A30A
Hot air circulation system
Ambient: +10 to +30℃, 20 to 80%RH

RO-26DPA

40A(Ambient temp. +20℃) to +200℃±0.5℃(at +100℃), ±0.7℃(at +200℃)
1.5℃(at +100℃), 3.0℃(at +200℃)(Ambient temp. +20℃) to +200℃ within 50 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
3.0 kW
4.0 kW
4.5 kW
6.0 kW
6.0 kW
9.0 kWVentilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
288
512
640
1000
1200
1500
600×800×600 800×800×800 800×1000×800 1000×1000× 1000×1200× 1000×1500×1000
1000
1000
1150×1500× 1350×1500× 1350×1700× 1550×1700× 1550×1900× 1550×2220×1250
1500
1500
1700
1700
1700
350
450
600
680
750
800

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.