We test, You produce

ERD620NEW COMPACT VALVES TESTER
After 45 years, and 150,000 testers installed in more than 5,000 customers worldwide, ATEQ
presents its revolutionnary valves tester ERD620.
New electronic module, new measurement module, new interface, new accessories… All has
been made to improve your quality control.
The purpose of the instrument is to generate a pressure ramp and is waiting for an event as aﬂow appears (ex: valve opening). A ramp down can also detect when valve is closed. Theinstrument will display the event pressures and compare them to reject levels. As an option
the ﬂow event can be replaced by a dry contact event.

Highlights
LARGE COLOR DISPL AY
MAXIMUM PERFORMANCE
MULTIPL E COMMUNICATION

Applications
FLOW CONTROL
Opening valves, closing valves with ﬂow detection...
LEAK DETECTION
Leaking levels with ﬂow detection...
CONTROL
Pressure contact elements...

ERD620

NEW COMPACT VALVES TESTER
Measurement ranges

Technical Speciﬁcations

FLOW MEASUREMENT
Range l/h
53080
1505001.500

AccuracyMax. resolution l/h+- (2.5%F + 0.005 l/h)+- (2.5%F + 0.03 l/h)

0.0001+- (2.5%F + 0.08 l/h)+- (2.5%F + 0.15l/h)

0.001+- (2.5%F + 0.5 l/h)+- (2.5%F + 1.5 l/h)

0.01

Physical

Weight: 8 Kg
Dimensions: (WxHxD) 250x150x360 mm

Interface

Graphic color screen with simple and intuitivemenu-driven operator controls
SMARTKEY: programmable key with user’s functions

0.001
0.01
0.1

F =Flow

External electrical supply 24 VDC/2A or 100-240 VAC/50W
Air supply

Clean and dry air required
Air quality standard to be applied (ISO 8573-1)

Temperature

Operating: +5°C to +45°C (+41 °F to 113 °F)
Storage: 0°C to +60°C (32 °F to 140 °F)

Main features• Flow measurement in direct or indirect testing• Wide ﬂow measurement range• USA, S.I. and customer units• Measurement in standard or customised conditions• Recalculated ﬂow at the nominal pressure• Measurement element fully integrated into one module• 128 programs• Electronic regulator from vacuum to 10 bar• setting on line (via USB)• 2 languages (English + other one)
• Data storage on internal memory or USB keyEnvironmentOption• ROHS standardFlexible communication format• RS232: Printer, modbus RTU• Slave USB: PC supervision (WinAteq 300)
• Master USB: USB key to transfert (parameters, results, ...)• Option Fieldbus: Proﬁbus, devicenet, proﬁnet, Ethernet/IP,Modbus RTU• 9 inputs / 5 local outputs compatible with 5 series instruments• 6 inputs/ 6 outputs 24 V programmable for external automation• External capillary: Close to the latest part• 2 pneumatic outputs for control of sealing connectors• Memory card for archiving results (400 000)

Additional:
Time & date, language & units choice, test customisation & more...

• Simple remote control TLC 60 connexion I/O• Advanced remote control TLC 600 connexion master USB• USB key for parameters access• Y valve• WinAteq 300 software• Leak calibrator• Master jets
• Vacuum pump
• Sealing connectors
Please refer to ATEQ accessories list

TLC 600
Remote control

Master jet

TLC 60
Remote control
Pneumatic coupler

15, rue des Dames - ZI des Dames - 78340 Les Clayes-sous-Bois - France
T.: +33 (0) 1 30 80 10 20 - F.: +33 (0) 1 30 54 11 00 - info@ateq.com - ateq.com

Speciﬁcations subjects to change without notice, non-contractual photos. August 2020.

Accessories