We test, You produce

V570

MEASUREMENT BY VOLUME COMPARISON
Without any doubt the most advanced high performance automated volume tester on themarket.
Specially adapted to automatic and semi-automatic control, the V570 has a very complete
range of equipment and beneﬁts from the use of the latest technological innovations of
the 5 series of instruments.
The V570 is composed of two equal volumes (T1 and T2) which are constantly maintainedat test pressure between two measurements. Following the start of a cycle, T1 and T2 are
dumped into the volume to be tested (VT) and the known reference volume (VR).
After a short stabilisation time, the new diﬀerential pressure measurement is proportional
to the diﬀerence between VT and VR. Through its user friendly interface, the V570 thendisplays the results of the test either in Pa or in a chosen unit of volume if the calibration
procedure has taken place.
Highlights
NEW IMPROVED PERFORMANCEOPTIMISED MEASUREMENTHIGH SPECIFICATIONS COMPETITIVELY PRICEEASY SET-UP

Applications
Cylinder heads, cast parts, combustion chambers,piston heads, Carburettor ﬂoats, liquid levels (bottling),
measurement of ﬁlling levels (oil levels)...

V570

MEASUREMENT BY VOLUME COMPARISON
Measurement ranges

Technical Speciﬁcations

Range

AccuracyMax. resolution+- 500 Pa+- 2.5 % of the full scale +- 1 Pa

1 Pa

Physical

Weight: 15 kg(WxHxD) 537x157x380 mm

Interface

Navigation keys
14 digit displays
4 line LCD display
Results indicators

PRESSURE MEASUREMENT
Range

AccuracyMax. resolution

5 - 75 hPa+- 1 % of the full scale

0.1% F.S.

MECHANICAL PRESSURE REGULATION
from 0,5 kPa to 7,5 kPa
The base unit is the Pa. The resolution in terms of volume is inferior to 1% of the test volume. The accuracy depends on several
calibration parameters. Volume range: from a few cm3 to 10 dm3 with accuracies reaching a few mm3 on small volumes of several cm3

Main features• Measurement accuracy of a few mm3 in several litres• Choice of operation in air or liquid volumes• Software adjustment of the test and reference
circuits through a volume auto zero• Leak test and simultaneous measurment of volume• High resolution mode
• Single calibration mode
• Estimation of the volume of the test part• 16 programs

Environment• ROHS standard

Flexible communication format• RS232: printer, P.C. and parameter save modules
Additional:
Time and date display, choice of languages and units,
customisation of tests for your requierement...

External electrical supply 90 - 260 V AC / 50 W
Air supply

Clean and dry air required
Air quality standard to be applied (ISO 8573-1)

Temperature

Operating: +10°C to +45°C
Storage: 0°C to +60°COption• 7 local inputs / 5 local outputs• 1 quick connector for calibration checking• 2 pneumatic outputs for control of sealing connectors• 6 x 24 V programmable outputs for external connection• 2 analogue current or voltage outputs

Accessories• 2 functions remote control• Printer• ATEQ software (WINATEQ, S5 Demo)• Needle valve• Calibrated master leaks• Filtration kit• Sealing connectors
Please refer to ATEQ accessories list

Master jet

Measurement principles
Pneumatic coupler

15, rue des Dames - ZI des Dames - 78340 Les Clayes-sous-Bois - France
T.: +33 (0) 1 30 80 10 20 - F.: +33 (0) 1 30 54 11 00 - info@ateq.com - ateq.com

Speciﬁcations subjects to change without notice, non-contractual photos. February 2021.

VOLUME MEASUREMENT