LARGE SIZE TEMPERATURE CHAMBER (INDUSTRIAL OVEN)

JMB(Ambient temp. +30°C) to +300°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)JMB series
JMB-31DPA

JMB-32DPA

30A

40A

JMB-33DPA JMB-34DPA JMB-35DPA
Highest temperature is +300°C
AC 200V ±10% 3φ 50/60Hz
50A
50A
100A
Hot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

JMB-36DPA

JMB-37DPA

125A

150A

(Ambient temp. +30°C) to +300°C±0.6°C(at +150°C) ±0.8°C(at +300°C)
2.3°C(at +150°C), 4.5°C(at +300°C)(Ambient temp. +30°C) to +300°C within 60 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
6.0 kW
9.0 kW
12.0 kW12.0 kW24.0 kW30.0 kW42.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
640
1000
1440
1500
2700
3600
4800
1000×800× 1000×1000× 1200×1200× 1500×1000× 1500×1500× 2000×1500× 2000×2000×
800
1000
1000
1000
1200
1200
1200
1775×1200× 1775×1400× 2000×1600× 2350×1450× 2500×1950× 3100×1950× 3100×2450×
1000
1200
1200
1200
1400
1400
1400
600
600
700
820
900
1200
1500

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

LARGE SIZE TEMPERATURE CHAMBER (INDUSTRIAL OVEN)

JMBBB(Ambient temp. +20°C) to +200°C

NAME

MODEL

POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)JMB series
JMB-21DPA

JMB-22DPA

20A

30A

JMB-23DPA JMB-24DPA JMB-25DPA
Highest temperature is +200°C
AC 200 V 10% 3φ 50 / 60 Hz
40A
40A
75A
Hot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

JMB-26DPA

JMB-27DPA

100A

125A

(Ambient temp. +20°C) to +200°C±0.5°C(at +100°C), ±0.7°C(at +200°C)
1.5°C(at +100°C), 3.0°C(at +200°C)(Ambient temp. +20°C) to +200°C within 45 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
4.5 kW
6.0 kW
9.0 kW
9.0 kW
18.0 kW
21.0 kW
27.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface R S-232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
640
1000
1440
1500
2700
3600
4800
1000×800× 1000×1000× 1200×1200× 1500×1000× 1500×1500× 2000×1500× 2000×2000×
800
1000
1000
1000
1200
1200
1200
1775×1200× 1775×1400× 2000×1600× 2350×1450× 2500×1950× 3100×1950× 3100×2450×
1000
1200
1200
1200
1400
1400
1400
600
600
700
820
900
1200
1500

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

M I D D L E SI Z E TE M P E R AT U R E C H A M B E R (I N D U S T R I A L O V E N )RO(Ambient temp. +30°C) to +400°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)RO series
RO-41DPA

RO-42DPA

30A

40A

RO-43DPA
RO-44DPA
RO-45DPA
Highest temperature is +400°C
AC 200V ±10% 3φ 50/60Hz
40A
40A
50A
Hot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

RO-46DPA

60A(Ambient temp. +30°C) to +400°C±0.7°C(at +200°C), ±0.8°C(at +400°C)
3.0°C(at +200°C), 6.0°C(at +400°C)(Ambient temp. +30°C) to +400°C within 90 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
6.0 kW
9.0 kW
9.0 kW
9.0 kW
12.0 kW
16.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
288
512
640
1000
1200
1500
600×800×600 800×800×800 800×1000×800 1000×1000× 1000×1200× 1000×1500×
1000
1000
1000
1200×1600× 1450×1600× 1450×1800× 1650×1800× 1650×2000× 1650×2300×
1350
1600
1600
1800
1800
1800
400
500
650
730
800
850

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

M I D D L E SI Z E TE M P E R AT U R E C H A M B E R (I N D U S T R I A L O V E N )RO(Ambient temp. +30°C) to +300°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)RO series
RO-31DPA

RO-32DPA

20A

30A

RO-33DPA
RO-34DPA
RO-35DPA
Highest temperature is +300°C
AC 200V ±10% 3φ 50/60Hz
30A
40A
40A
Hot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

RO-36DPA

50A(Ambient temp. +30°C) to +300°C±0.6°C(at +150°C), ±0.8°C(at +300°C)
2.3°C(at +150°C), 4.5°C(at +300°C)(Ambient temp. +30°C) to +300°C within 60 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
4.0 kW
6.0 kW
6.0 kW
9.0 kW
9.0 kW
12.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
288
512
640
1000
1200
1500
600×800×600 800×800×800 800×1000×800 1000×1000× 1000×1200× 1000×1500×
1000
1000
1000
1150×1500× 1350×1500× 1350×1700× 1550×1700× 1550×1900× 1550×2220×
1250
1500
1500
1700
1700
1700
350
450
600
680
750
800

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

M I D D L E SI Z E TE M P E R AT U R E C H A M B E R (I N D U S T R I A L O V E N )RO(Ambient temp. +20℃ ) to +200℃

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)RO series
RO-21DPA

RO-22DPA

15A

20A

RO-23DPA
RO-24DPA
RO-25DPA
Highest temperature is +200℃
AC 200V ±10% 3φ 50/60Hz
20A
30A
30A
Hot air circulation system
Ambient: +10 to +30℃, 20 to 80%RH

RO-26DPA

40A(Ambient temp. +20℃) to +200℃±0.5℃(at +100℃), ±0.7℃(at +200℃)
1.5℃(at +100℃), 3.0℃(at +200℃)(Ambient temp. +20℃) to +200℃ within 50 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
3.0 kW
4.0 kW
4.5 kW
6.0 kW
6.0 kW
9.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
288
512
640
1000
1200
1500
600×800×600 800×800×800 800×1000×800 1000×1000× 1000×1200× 1000×1500×
1000
1000
1000
1150×1500× 1350×1500× 1350×1700× 1550×1700× 1550×1900× 1550×2220×
1250
1500
1500
1700
1700
1700
350
450
600
680
750
800

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

SMALL SIZE TEMPERATURE CHAMBER (INDUSTRIAL OVEN )TRO(Ambient temp. +80°C) to +500°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)

TRO series
TRO-52DPA
Highest temperature is +500°C
AC 200V ±10% 3φ 50/60Hz
30A
Hot air circulation method
Ambient: +10 to +30°C, 20 to 80%RH

TRO-51DPA

20A

TRO-53DPA

40A(Ambient temp. +80°C) to +500°C±0.6°C(at +150°C), ±0.8°C(at +300°C)
2.3°C(at +150°C), 4.5°C(at +300°C)(Ambient temp. +30°C) to +300°C within 60 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
1.5 kW
3.0 kW
4.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
72
150
288
450×400×400
600×500×500
800×600×600
1000×650×515

1150×750×610

1400×900×800

60

100

200

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

SMALL SIZE TEMPERATURE CHAMBER (INDUSTRIAL OVEN )TRO(Ambient temp. +30°C) to +300°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)

TRO series
TRO-32DPA
Highest temperature is +300°C
AC 200V ±10% 3φ 50/60Hz
15A
Hot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

TRO-31DPA

15A

TRO-33DPA

20A(Ambient temp. +30°C) to +300°C±0.6°C(at +150°C), ±0.8°C(at +300°C)
2.3°C(at +150°C), 4.5°C(at +300°C)(Ambient temp. +30°C) to +300°C within 60 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
1.5 kW
3.0 kW
4.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
72
150
288
450×400×400
600×500×500
800×600×600
1000×650×515

1150×750×610

1400×900×800

60

100

200

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

SMALL SIZE TEMPERATURE CHAMBER (INDUSTRIAL OVEN )TRO(Ambient temp. +20°C) to +200°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)

TRO series
TRO-22DPA
Highest temperature is +200°C
AC 200V ±10% 3φ 50/60Hz
15A
Hot air circulation system
Ambient: +10 to +30°C, 20 to 80%RH

TRO-21DPA

15A

TRO-23DPA

15A(Ambient temp. +20°C) to +200°C±0.5°C(at +100°C), ±0.7°C(at +200°C)
1.5°C(at +100°C), 3.0°C(at +200°C)(Ambient temp. +20°C) to +200°C within 45 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
1.2 kW
2.25 kW
3.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
72
150
288
450×400×400
600×500×500
800×600×600
1000×650×515

1150×750×610

1400×900×800

60

100

200

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be changed for any improvement of product without giving any preliminary notification.

CLEAN TYPE TEMPERATURE CHAMBER (INDUSTRIAL OVEN )CLN(Ambient temp. +80°C) to +300°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)

CLN series
CLN-32DPA
Highest temperature is +300°C
AC 200 V 10% 3φ 50 / 60 Hz
40A
Hot air circulation method
Ambient: +10 to +30°C, 20 to 80%RH

CLN-31DPA

20A

CLN-33DPA

50A(Ambient temp. +80°C) to +300°C±0.6°C(at +150°C), ±0.8°C(at +300°C)
2.3°C(at +150°C), 4.5°C(at +300°C)(Ambient temp. +80°C) to +300°C within 60 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
4.5 kW
9.0 kW
12.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet、Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
205
432
864
600×600×570
600×900×800
900×1200×800
1080×1640×1310

1080×1940×1555

1380×2350×1555

340

370

470

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be change d for any improvement of product without giving any preliminary notification.

CLEAN TYPE TEMPERATURE CHAMBER (INDUSTRIAL OVEN)

CLN(Ambient temp. +30°C) to +200°C

NAME
MODEL
POWER SUPPLYELBCONTROL SYSTEMOPERATING CONDITIONSPERFORMANCE
Temperature range
Fluctuation
Variation in space
Temperature rise time
CONFIGURATION
Controller
Internal material
External material
Insulating material
DRY HEATER
Out put
BLOWING / CHUMINGACCESSORIESSAFETY UNITSOPTIONSINTERNAL CAPACITY (L)INTERNAL DIMENSIONS :W.H.D.mm
EXTERNAL DIMENSIONS :W.H.D.mm
WEIGHT (kg)

CLN series
CLN-22DPA
Highest temperature is +200°C
AC 200 V 10% 3φ 50 / 60 Hz
15A
Hot air circulation method
Ambient : +10 to +30 °C, 20 to 80 %RH

CLN-21DPA

15A

CLN-23DPA

15A(Ambient temp. +20°C) to +200°C±0.5°C(at +100°C), ±0.7°C(at +200°C)
1.5°C(at +100°C), 3.0°C(at +200°C)(Ambient temp. +20°C) to +200°C within 45 min.
KATO DPA instrumentation
Stainless steel plate
Cold-rolled steel plate, Painting finishedRock wool, Silicon packing
Stainless steel sheath heater
4.0 kW
6.0 kW
9.0 kW
Ventilation and circulation system with Pressure fan
Adjustable shelf ×2 sets, Air inlet, Exhaust pipe
Earth leakage circuit breaker, Blowing motor overload relay, Overheat prevention unit
Signal tower, Recorder, Cable hole 30, 50, 100φ, Shelf, Operation holes, Interface RS -232C,Time signal unit, Overheating preventing unit for heater unit, Floor Reinforcement
205
432
864
600×600×570
600×900×800
900×1200×800
1080×1640×1155

1080×1940×1390

1380×2300×1400

320

350

450

*The performance values are based on JTM K-07, IEC60068-3-5:2001, and no specimen inside the test area.
*The above specifications may be change d for any improvement of product without giving any preliminary notification.