We test, You produce

D620

HIGH SPEED COMPACT FLOW TESTER
After 45 years, and 150,000 testers installed in more than 5,000 customers worldwide, ATEQ
presents its revolutionnary ﬂow tester: D620.
New electronic module, new measurement module, new interface, new accessories… All has
been made to improve your quality control.
And yet, inside can be found the latest leak testing technics and technology that will garantee
you the fastest and most accurate leak testing cycle on the market.

Highlights
LARGE COLOR DISPLAY
MAXIMUM PERFORMANCE
MULTIPLE COMMUNICATION

Applications
FLOW CONTROL
cookers, ovens, valve, gas taps, gas burners, lighters...
LEAK DETECTION
Valve seats, gear boxes, hydraulic components, motors,,,
CONTROLof aerosols, injectors...

D620

HIGH SPEED COMPACT FLOW TESTER
Measurement ranges

Technical Speciﬁcations

FLOW MEASUREMENT
Range l/h

AccuracyMax. resolution l/h+-(2.5%F +0.001l/h)+-(2.5%F +0.005l/h)

0.0001+-(2.5%F +0.03l/h)+-(2.5%F +0.08l/h)

0.001+- (2.5%F +0.15l/h)+-(2.5%F +0.5l/h)

0.01
0.1

4.000

+- (2.5%F +1.5l/h)+-(2.5%F +4l/h)

10.000+-(2.5%F +10l/h)

1153080
1505001.500

Physical

Weight: 7 Kg
Dimensions: (WxHxD) 250x150x360 mm

Interface

Graphic color screen with simple and intuitivemenu-driven operator controls
SMARTKEY: programmable key with user’s functions

0.0001

External electrical supply 24 VDC/2A or 100 - 240 VAC/50W

0.001

Air supply

Clean and dry air required
Air quality standard to be applied (ISO 8573-1)

0.01

Temperature

Operating: +5°C to +45°C
Storage: 0°C to +60°C

0.1

F =FlowMain featuresOption• Flow measurement in direct or indirect testing• Continuous measuring mode
• Wide ﬂow measurement range• USA, S.I. and customer units• Measurement in standard or customised conditions• Recalculated ﬂow at the nominal pressure• Measurement element fully integrated into one module• 128 programs• Mechanical or electronic regulator from vacuum to 10 bar• setting on line (via USB)• 2 languages (English + other one)
• Manual calibration on front panel with calibrated leak
• Data storage on internal memory or USB key• 9 inputs / 5 local outputs compatible with 5 series instruments• 6 inputs/ 6 outputs 24 V programmable for external automation• External capillary: closed to test port
• Electronic regulator• Shut oﬀ / dumping / by-pass valve• 2 pneumatic outputs for control of sealing connectors• Memory card for archiving results (400 000)

Accessories• Simple remote control TLC 60 connexion I/O• Advanced remote control TLC 600 connexion master USB• USB key for parameters access• Y valve• WinAteq 300 software• Leak calibrator

Environment• ROHS standard• Master jets• RS232: Printer, modbus RTU• Slave USB: PC supervision (WinAteq 300, sesame)• Master USB: USB key to transfert (parameters, results, ...)• Option Fieldbus: Proﬁbus, devicenet, proﬁnet, Ethernet/IP, Modbus RTU• Vacuum pump
• Sealing connectors
Please refer to ATEQ accessories list

Additional:
Time & date, language & units choice, test customisation & more...

TLC 60
Remote control

Master jet

D670
19”3U case

Pneumatic coupler

TLC 600
Remote control

15, rue des Dames - ZI des Dames - 78340 Les Clayes-sous-Bois - FranceT.: +33 (0) 1 30 80 10 20 - F.: +33 (0) 1 30 54 11 00 - info@ateq.com - ateq.com

Speciﬁcations subjects to change without notice, non-contractual photos. August 2020.

Flexible communication format