E110D-i

E-SERIES

E110D-i

DUAL-ENERGYCOMPACTUP TO 30 KW PEAK
POWER
FULLY COMPUTERIZEDMINIMIZEDCONSUMABLES

E-GUN MODULATOR UP TO 30 KW
ScandiNova's E110D-i is a compact solid-state
pulse Modulator designed for system integration. The
Modulator is designed to operate all kinds of diodeE-guns with a peak power up to 30 kW and a voltage
level up to 30 kV. To facilitate system integration, theModulator has a split design with a Tank Unit anda Pulse Unit, and can be placed in a very compact
enclosure. The Modulator is well protected against any
stress from the E-gun. The E100D-i is available withsingle or dual energy. The dual-energy model is forapplications where there is a need to switch between
two different energy levels.
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers both a user-friendly interface and a capability forboth manual and automatic pulse shape modification.
The pulse length, amplitude and repetition rate can all
be changed.

THE E-SERIES
ScandiNova's E-series
contains a range of E-gun
pulse modulators with highreliability and performance,
a compact design and low
energy consumption.
Our E-gun pulse modulators
are used for driving E-gunsin a wide range of linac
applications including highenergy physics, radiation
therapy, cargo inspection,industrial X-ray andsterilization.

Different levels of service and support programmes areavailable.

Excellence in pulsed powerwww.scandinovasystems.com

E110D-i

UNIT

VALUE

NOTES

Peak power to E-gun

kW

0-30

Dual energy option available

Average power to E-gun

W0-40

Dual energy option available

Operational voltage rangekV0-30

Higher voltage on request

Operational current range

A0-1

Higher current on request

Pulse length, topμs

0.5-4Longer pulse length on request

Pulse repetition rate

Hz0-400

According to external trig generator
Single shot operation possible
Higher frequency option available

Cooling-Air

INTERFACECONNECTOR

DEFAULT

Mains power, single-phase

C1495-255 VAC, 50/60 Hz

Control interface

RJ45

Modbus TCP

Trig input

BNC

Electrical

Diagnostics

BNC

Pulse voltage & current signals

SIZE AND WEIGHT

UNIT

VALUE

Weight, Pulse Unitkg

16Weight, Tank Unit (incl. oil)kg29

OPTION

OpticalThe Standard Modulator Includes
Filament power supply
Control system
Control via Modbus TCP

For more information, visit www.scandinovasystems.com/E110D-i

Pulse sensors
Internal trig generator

178 mmDigitizer

28.7 mm
408.7 mm440.7 mm
12.7 mmCable kit between Pulse Unit and Tank Unit,
standard length 2 m, option up to 10 m

497 mm

HV pulse cable, 0.5m
Over temperature protection
Factory acceptance test certificate
Manuals

470.8 mm

Services

Pulse Unit

Training in handling, operationand maintenance
Shipping515 mm
256 mmInstallation and start-up
Service contract

270 mm

297 mm
242 mm

DOC-011738-02 / 2024-04-09

SPECIFICATIONS

304 mmØ 8.5 mm
17 mm

43 mm

Tank Unit

Contact ScandiNova for any requirements not covered by this specification.
Information contained in this document is subject to change without notice.

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power