M200

M-SERIES

M200

COMPACTUP TO 5 MW RF PEAK
POWER
FULLY COMPUTERIZEDMINIMIZEDCONSUMABLESMAGNETRON MODULATOR UP TO 5 MW
ScandiNova's M200 is a stand-alone pulse Modulator
in a very compact enclosure. The Modulator is optimized
for a wide range of different magnetrons with an RF peak
power ranging from 2.5 MW to 5 MW, and can also power
klystrons. The M200 is also available as a dual version(see M200D).
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers a user-friendly interface. The pulse length,
amplitude and repetition rate can all be changed.
The M200 can be delivered as a pure high-power pulse
modulator or as part of an RF Unit including magnetronand related components. Different levels of service andsupport programmes are available.

THE M-SERIES
ScandiNova's M-series
contains a range of magnetronpulse modulators with highreliability and performance,
a compact design and low
energy consumption.
Our magnetron pulse
modulators require about
one third of the space ofalternative solutions yet still
offer around 30% higher
efficiency, two operating
advantages that especiallyappeal to cancer treatment
providers.
Their high reliability andperformance also attract manyindustry applications.

Excellence in pulsed powerwww.scandinovasystems.com

M200

UNIT

VALUE

NOTES

Magnetron RF peak power

MW

2.5-5

Depends on choice of magnetron

Magnetron RF average power

kW

3.5

Maximum, depends on choice of magnetron

Modulator peak power

MW

14

Maximum

Modulator average power

kW

16

Maximum

Pulse voltage

kV

40-75

Pulse current

A30-250

RF pulse length, topμs

0.5-5

Depends on max average power

Pulse repetition rate

Hz0-500

Depends on max average power

Modulator Voltage Stability, RMS%0.4

Verified on resistive load (see options)

Water coolingl/min818-40 °C, non-condensing

INTERFACECONNECTOR

DEFAULTOPTION

Mains power, three-phase

Terminal block (4-wire)

400±10 % VAC, 50/60 Hz

208/380/480 VAC

Mains power, single-phaseC20

230±10 % VAC, 50/60 Hz

115 VAC

Control interface

RJ45

Modbus TCP

Water cooling interface

12 mm

Legris Push-inSwagelock

Trig input

BNC

Electrical

OpticalDiagnostics

BNC

Pulse voltage & current signals

SIZE AND WEIGHT
Weight (incl. oil)

UNITkg

VALUE
200

The Standard Modulator Includes

Services

Filament power supply

Training in handling, operationand maintenanceControl system
For more information, visit www.scandinovasystems.com/M200

Control via Modbus TCP
Pulse sensors
HV pulse cable

544 mm

DOC-004508-02 / 2024-01-19

SPECIFICATIONS

1267 mm

647 mmService contract

Typical Magnetron Loads

Factory acceptance test certificate(M200 can also power klystrons)

Manuals

BVERIOptions

VE2116

Enhanced PRF Range (1000–2000 Hz)

Information contained in this document is subject to change
without notice.

Installation and start-up

Internal trig generator

Pulse/RF diagnostics

Contact ScandiNova for any requirements not covered bythis specification.

Shipping

Enhanced Stability (down to 0.1 %)Digitizer

E2V
M5028
MG8076

Gunport

GLVAC

Additional System Components

GLM8028

Circulator & RF loads
Directional coupler
Magnet power supply
Waveguide windows
Magnetron

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power