…Clearly a Better Value

http://www.aandd.jp

Contents
Load Cell Terminology

3Single Point

4
5
6
7
8
9
10
11
12

LCB03/04 Series
LCB05 Series
LC-4001
LC-4101 Series
LC-4102 Series
LC-4103 Series
LC-4204 Series
LCB22 Series

Beam TypeLCM13 Series
LCM13-M Series
LCM19 Series
LCM19-M Series
LC-4221 Series
LC-5206 Series
LC-5223 Series
LBP Series

Tension Type
LC-1205 Series
LC-1216 Series
LCS15 SeriesCompression TypeLCC07 Series
LCC11/N Series
LCC12 Series
LC2224 Series
CP/TP Series
C2F1/C2Z1/C2X1 Series
CMX/CM Series
TM/UM Series

13
14
15
16
17
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34

Button Load Cell

2

LCC21 Series
LCCA21 Series

35
36

Flameproof TypeLCB01/02/06 Series
LCM17/LCM17-M Series

37
38
39

Flameproof Type Equipment

40

Weighing Platform

41

Equipment

42

Accessories

44

Load Cell Terminology
Each country has its own standard for the performance and quality of load cells.
In Japan, there is no standard stipulated uniquely for load cells, although JIS(Load Cell - Test Method JIS B7602) and JMIF (Japan Measuring Instruments
Federation) prescribe methods for performance tests.
Meanwhile, the OIML (International Organization of Legal Metrology) sets theinternational recommendation titled “Metrological Regulation for Load Cells,”
with which each member country tries to make their local regulation consistent.
The terms described below are commonly used by the OIML and its member
countries.
Rated Capacity (RC)The maximum load that a load cell can measure while meeting its specifications.

Rated Output (RO)The difference when there is no load and when there is a load of rated capacity. It is
generally expressed in output per excitation voltage (mV/V). Alternatively called“Span.”

Safe load limit
The maximum load that can be applied beyond the rated capacity without causing
any permanent damage. Expressed as a percentage of rated capacity.

Compensated temperature range
The temperature range within which the rated output and the zero balance arecompensated to meet the specifications.

Temperature effect on zero balance
Drifting of the zero balance caused by changes in the ambient temperature.
Expressed as a percentage of rated output.

Temperature effect on rated output
Drifting of the rated output caused by changes in the ambient temperature.

Nonlinearity
The maximum deviation in output from the linear calibration
curve linking the zero balance and the rated output, which ismeasured only when the load increases. Expressed as apercentage of rated output.

OutputHysteresis error
The maximum difference in output between when the loadincreases and when the load decreases.

Combined error
The maximum deviation of output from the linear calibration
curve linking the zero balance and the rated output, including
when the load increases and decreases.

Recommended/maximum excitation voltage

Hysteresis
Nonlinearity

Combined
error

Load

The voltage applied to the input terminals of load cell.

Zero balance
The electrical output when the rated excitation voltage is applied without any loadon the cell. Generally expressed as a percentage of rated output.

Input terminal resistance
Measured while the output terminals are open, with no load on the cell.

Output terminal resistance
Measured while the input terminals are open, with no load on the cell.

Insulation resistanceThe direct current resistance between the load cell unit and its circuit.

3

Single Point
Single-point load cells are ideal to be used alone andfor relatively small loading.
They are already adjusted to correct four-corner errors.
Example Applications●Electronic balances●Desktop scales●Small/medium-sized platform scales●Conveyer scales, etc.

Example of Installation
The LCB04

200(MAX)

200(MAX)

Corner point

200(MAX)

4200(MAX)

Platform

Platform spacer

Platform

Corner lineof platform

Corner stopper

Base
Corner stopper

Unit = mm

Single Point

LCB03/04 Series

Single-point Aluminum Load Cell

LCB03 Series

LCB04 Series

LCB03M/L Series
12

130

80

6

18

LCB03K025L

25.49kg (250N)

LCB03K030L

30.59kg (300N)

LCB03K035L

35.69kg (350N)

LCB03K060L

61.18kg (600N)

LCB04K060M

61.18kg (600N)

LCB04K100M

102.0kg (1kN)

LCB04K150L

153.0kg (1.5kN)

LCB04K250L

254.9kg (2.5kN)

Safe load limit

150% of R.C.

Combined error

0.02% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±5% of R.O.

Input terminal resistance

420±30Ω

Output terminal resistance

350±5Ω

Insulation resistance

2000MΩ/DC50V(B/E)

Compensated temperature range–10ºC to 40ºC

SIG–

Net Weight : 0.23kg(L Series : 0.3kg)
18

20.39kg (200N)

BLUE

25

150
102

15.30kg (150N)

LCB03K020M

2mV/V±10%

M6 34mm deep40

18

10.20kg (100N)

LCB03K015M

Rated output

EXC+

YELLOW SHIELD

18

6

6.118kg (60N)

LCB03K010M

WHITE EXC–

22

LCB04M Series

LCB03K006M

Specifications(40) : L Series
25

Capacity

Unit = mm

GREEN SIG+
15 30
(40)

3.059kg (30N)

CABLE COLOR CODE
RED

4-M6

12

106

Model No.
LCB03K003M

Temperature effect on
Zero balance

0.023% of R.O./10ºC

Span

0.014% of Load/10ºC Typ.

Cable diameter/lengthø3.5/0.4m (LCB03)
ø3.5/1.5m (LCB04)

4-M6 16mm deep

30

20

150
98

20

20

IP54

6

4-M8 16mm deep

32

34

4-M8 16mm deep

Dustproof/waterproof

Net Weight : 0.5kg

20

6300✕300mm (LCB03)

40

LCB04L Series

Maximum platform size

400✕400mm (LCB04)3430

2-M6 16mm deep

32

Net Weight : 0.5kg

5Single Point

LCB05 Series

Hermetically Sealed Single-point Stainless Steel Load Cell
Ideal for measurements in harsh environments
Model No.
LCB05K030

30.59kg (300N)

Capacity

LCB05K060

61.18kg (600N)

LCB05K150

153.0kg (1.5kN)

Specifications

40

150

10

18

LCB05K030/060

Unit = mm

37

8.5

2-M6 Depth 15mm

6

M6 Pierced

32

19

30

6.5

4-M4 Bolt with Hexagonal Hole18

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

86

6 18

102

Net Weight : 1.2kg
18

18

4-M6 Depth 12mm

32

150

10

37

8.5

4-M8 Depth 15mm

20

6

32

19

30

6.5

4-M4 Bolt with Hexagonal Hole

40

20

LCB05K150

86

6 20

98

20

20

4-M8 Depth 12mm

32

Net Weight : 1.2kg

6

Rated output

2mV/V+15% –0%

Safe load limit

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±5% of R.O.

Input terminal resistanceApproximately 390Ω

Output terminal resistance

350±5Ω

Insulation resistance

500MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on
Zero balance

0.04% of R.O./10ºC

Span

0.014% of Load/10ºC Typ.

Cable diameter/lengthø5.6/3m

Maximum platform size

400✕400mm

Dustproof/waterproof

IP67

Single Point

LC-4001

Single-point Aluminum Load Cell with Ultra-high Sensitivity
Resolution as high as 10mg
Built-in metal overload stopper
Model No.
LC-4001-G120

Capacity
122.4g (1.2N)

Specifications
Rated output

0.4079mV/V or greater

Safe load limit

300% of R.C.

Combined error

0.015% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance

40±25% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±5Ω

Insulation resistance

500MΩ/DC50V

Compensated temperature range–5ºC to 35ºC

Temperature effect on
Zero balance

0.14% of R.O./10ºC

Span

0.02% of Load/10ºC Typ.

Cable diameter/length

F.P.C.75mm

Maximum platform size120✕120mm

Dustproof/waterproof

IP22

F.P.C. connection code
1. Excitation –2. Signal –3. Excitation +4. Signal +5. N.C.

Included connector

Approx. 66

Depth 8

Unit = mm
Net Weight : 0.26kg

7Single Point

LC-4101 Series

Single-point Aluminum Load Cell Ideal for Small
Capacity Weighing
Model No.
LC-4101-G600

611.8g (6N)

Capacity

LC-4101-K1.5

1.530kg (15N)

LC-4101-K003

3.059kg (30N)

LC-4101-K006

6.118kg (60N)

LC-4101-K015

15.30kg (150N)

Specifications
Rated output

1.0197mV/V+15% –0%

Safe load limit

300% of R.C.

Combined error

0.015% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance

20±5% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±5Ω

Insulation resistance

500MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

6-M4 Depth 10

Approx.1.5m ø4Unit = mm
Net Weight : 0.12kg

8

Zero balance

0.04% of R.O./10ºC

Span

0.014% of Load/10ºC Typ.

Cable diameter/lengthø4/1.5m

Maximum platform size

300✕300mm

Dustproof/waterproof

IP22

Single Point

LC-4102 Series

Single-point Aluminum Load Cell Ideal for Small/Medium
Capacity Weighing
Model No.
LC-4102-K010

10.20kg (100N)

Capacity

LC-4102-K015

15.30kg (150N)

LC-4102-K030

30.59kg (300N)

LC-4102-K060

61.18kg (600N)

LC-4102-K150

153.0kg (1.5kN)

Specifications
Rated output

1.0197mV/V+15% –0%

Safe load limit

300% of R.C.

Combined error

0.015% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance

20±5% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±5Ω

Insulation resistance

500MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on
Zero balance

0.04% of R.O./10ºC

Span

0.014% of Load/10ºC Typ.

Cable diameter/lengthø4/1.5m

Maximum platform size

400✕400mm

Dustproof/waterproof

IP54

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD
4-M6 Depth 15
Rear

2-M6 Depth 15
Protective coverCable

M6 Pierced

Load

Loading area

Unit = mm
Net Weight : 0.6kg

Width of shelf

9Single Point

LC-4103 Series

Single-point Aluminum Load Cell Ideal for Medium
Capacity Weighing
Model No.
LC-4103-K060

61.18kg (600N)

Capacity

LC-4103-K100

102.0kg (1kN)

LC-4103-K150

153.0kg (1.5kN)

Specifications
Rated output

1.0197mV/V+15% –0%

Safe load limit

300% of R.C.

Combined error

0.015% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance

20±5% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±5Ω

Insulation resistance

500MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on

2-M8
Depth 22

Loading area

Load

Protective cover

M8 Pierced

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

10

Width of shelf

Unit = mm
Net Weight : 1.2kg

Zero balance

0.04% of R.O./10ºC

Span

0.014% of Load/10ºC Typ.

Cable diameter/lengthø6/2m

Maximum platform size

400✕600mm

Dustproof/waterproof

IP54

Single Point

LC-4204 Series

Single-point / Multi-point Aluminum Load Cell

Model No.
LC-4204-K300

Capacity
305.9kg (3kN)

LC-4204-K600

611.8kg (6kN)

Specifications
Rated output

1.5296mV/V±0.2%

Safe load limit

200% of R.C.

Combined error

0.015% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance

20±5% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±5Ω

Insulation resistance

500MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on

Width of shelf

0.014% of Load/10ºC Typ.

Cable diameter/lengthø6/5m

Maximum platform size

600✕700mm

Dustproof/waterproof

IP54

4-M12 Pierced

Protective cover

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

0.04% of R.O./10ºC

Span

Loading area

Load

4-ø13 Pierced

Zero balance

Unit = mm
Net Weight : 2kg

11Single Point

LCB22 Series

Single-point Aluminum Load Cell with Built-in
Overload Stopper
Extremely durable structure built to withstand excessive loads and impacts
without the need for additional stoppers
Model No.
LCB22K006

6kg (58.84N)

Capacity

LCB22K010

10kg (98.07N)

LCB22K015

15kg (147.1N)

LCB22K020

20kg (196.1N)

LCB22K030

30kg (294.2N)

Specifications
Rated output

1.5mV/V±5%

Safe load limit

500% of R.C.

Ultimate safe overload

1000% of R.C.

Combined error

0.02% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±5% of R.O.

Input terminal resistance

1.13kΩ±100Ω

Output terminal resistance1kΩ ±10ΩInsulation resistance

2000MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

150
126.5

2-M8 Depth 16 (through hole)
31

47.3

54

Direction of Load

17

160

31

9

28

19

Mounting surface

20 12.4
47

17

103

17

6.5

2-M8 Depth 16

Unit = mm
Net Weight : 0.5kg

12

Zero balance

0.014% of R.O./10ºC Typ.

Span

0.011% of Load/10ºC Typ.

Cable diameter/lengthø4/1.5m

Maximum platform size

400✕400mm

Dustproof/waterproof

IP65 ** Although this load cell is rated IP65 it can besubmerged up to 1m in water for up to 30 minutes
similar to the IP67 rating.

Beam TypeBeam type (multi-point) load cells are suitable
for large capacity weighing, andideal for low-profile scales.
Example Applications●Platform scales●Hopper scales●Tank scales●Conveyer scales, etc.

Example of Installation
The LCM13
Platform
Load cell
Leveling footSocketPlatform

Ball
SocketLoad cell
Frame

13

Beam TypeLCM13 Series

Beam Type Stainless Steel Load Cell

LCM13T002

Model No.
LCM13K100

102.0kg (1kN)

Capacity

LCM13K200

203.9kg (2kN)

LCM13K300

305.9kg (3kN)

LCM13K500

509.9kg (5kN)

LCM13T001

1.020t (10kN)

LCM13T1.5

1.530t (15kN)

LCM13T002

2.039t (20kN)

LCM13T003

3.059t (30kN)

LCM13T005

5.099t (50kN)

Specifications

LCM13K200

A

D

E

F
(36)

C

I

L

M
Mounting Surface

(1)

K

J

G
H
Direction of Load (Positive output)

Model No.
LCM13K100
LCM13K200
LCM13K300
LCM13K500
LCM13T001
LCM13T1.5
LCM13T002
LCM13T003
LCM13T005

Unit = mm

LCM13K500 / LCM13T001 / LCM13T1.5 / LCM13T002/ LCM13T003
AC

D

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

380±20Ω

Output terminal resistance

350±3.5Ω

Insulation resistanceGreater than 5000MΩ

E

A

B

C

D

E

F

G

H

I

J

Zero balance

0.016% of R.O./10ºC Typ.

Span

0.013% of Load/10ºC Typ.

Cable diameter/lengthø4/3m

Dustproof/waterproof

IP67

K

L

32 15 2 581

130 16 25 76 13 32 2-ø14 M12 ø21

32 15 2 581

171 19 38 95 19 38 2-ø21 M20 ø30.2 38 18 2 76

1.5

171 19 38 95 19 43 2-ø21 M20 ø30.2 43 20.5 2 76

2Net Weight = kg

B

A

C

D

E

F

K

K

14

L

M
Mounting Surface

I

J

H
G
Direction of Load (Positive output)

M
Mounting Surface

L

I

J

G
H
Direction of Load (Positive output)

M Net Weight130 16 25 76 13 32 2-ø14 M12 ø21

LCM13T005–10ºC to 40ºC

Temperature effect onF

B

Safe load limit

Compensated temperature range

Swivel Foot LCZAFT01S* See accessories pagefor more details.

B

2mV/V±0.1%

at DC50V

Weighing Module Kit
LCZAP1413 / LCZAP1414 / LCZAP1415

LCM13K100 / LCM13K200 / LCM13K300

Rated output

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Beam TypeLCM13-M Series

Stainless Steel Beam Load Cell with Built-in Vibration
Controllers for Lift Prevention
Ideal for tank and hopper scales when corrosion resistance is required
Built-in stopper with pre-adjusted stop cap for easy installation
Model No.
LCM13K100-M

102.0kg (1kN)

Capacity

LCM13K200-M

203.9kg (2kN)

LCM13K300-M

305.9kg (3kN)

LCM13K500-M

509.9kg (5kN)

LCM13T001-M

1.020t (10kN)

LCM13T1.5-M

1.530t (15kN)

LCM13T002-M

2.039t (20kN)

LCM13T003-M

3.059t (30kN)

Specifications
Rated output

2mV/V±0.1%

Safe load limit

150% of R.C.

Permissible horizontal force

Maximum 6kN (611kg)
LCM13K100-M ~ T1.5-M
Maximum 12kN (1223kg)
LCM13T002-M ~ T003-M

Permissible lifting force

Maximum 20kN (2039kg)
LCM13K100-M ~ T1.5-M
Maximum 30kN (3059kg)
LCM13T002-M ~ T003-M

N

B

E

L

Additional : 4 washers
Use washers to secure
the base.
H

M

F

H

6

C

8
ø2
ø13

D

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

380±20Ω

Output terminal resistance

350±3.5Ω

Insulation resistanceGreater than 5000MΩ
at DC50V

4–ø14(for upper plate installation)

Compensated temperature range

G

Temperature effect on
0.016% of R.O./10ºC Typ.

Span

0.013% of Load/10ºC Typ.

Cable diameter/lengthø4/3m

Dustproof/waterproof

IP67

I

A

–10ºC to 40ºC

Zero balanceJ

4–ø18(for lower base installation)

Combined error

P

Model No.
LCM13K100-M
LCM13K200-M
LCM13K300-M
LCM13K500-M
LCM13T001-M
LCM13T1.5-M
LCM13T002-M
LCM13T003-M

Unit = mm

O

K

O

P

Net Weght

104 70 55 136

138 17 20 23 25 70 15 17 63

118

5.6

118

145 17 24 26 25 84 15 19 63 152

8.6

A

B

C

97

208

116

242

D

E

F

84 48 175

G

H

I

J

K

L

M

N

Net Weight = kg

15

Beam TypeLCM19 Series

Beam Load Cells for Tank and Floor Scales

Model No.
LCM19K500

Capacity
509.9kg (5kN)

LCM19T001

1.020t (10kN)

LCM19T1.5

1.530t (15kN)

LCM19T002

2.039t (20kN)

Specifications
Rated output

2.0394mV/V±0.1%

Safe load limit

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

380±20Ω

Output terminal resistance

350±3.5Ω

Insulation resistanceGreater than 5000MΩ
at DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on

Weighing Module Kit
LCZAP1413 / LCZAP1414 / LCZAP1415
Swivel Foot LCZAFT01S* See accessories pagefor more details.

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

A

C

D

E

F

B

G

I

H

K

J

Direction of load

Model No.
LCM19K500
LCM19T001
LCM19T1.5
LCM19T002

Unit = mm

16

A

B

C

D

E

L

M

Mounting surface

F

G

H

I

J

K

L

M Net Weight

32 15 2 581

171 19 38 95 19 38 2-ø21 M20 ø30.2 38 18 2 76

1.5

130 16 25 76 13 32 2-ø14 M12 ø21

Net Weight = kg

Zero balance

0.011% of R.O./10ºC Typ.

Span

0.013% of Load/10ºC Typ.

Cable diameter/lengthø4/3m

Dustproof/waterproof

IP67

Beam TypeLCM19-M Series

Beam Load Cell with Built-in Vibration Controllers
for Lift Prevention
Excellent for tank and hopper scales
Built-in stopper with pre-adjusted stop cap for easy installation
Model No.
LCM19K500-M

Capacity
509.9kg (5kN)

LCM19T001-M

1.020t (10kN)

LCM19T1.5-M

1.530t (15kN)

LCM19T002-M

2.039t (20kN)

Specifications
Rated output

2.0394mV/V±0.1%

Safe load limit

150% of R.C.

Permissible horizontal force

Maximum 6kN (611kg)
LCM19K500-M ~ T1.5-M
Maximum 12kN (1223kg)
LCM19T002-M

Permissible lifting force

Maximum 20kN (2039kg)
LCM19K500M ~ T1.5-M
Maximum 30kN (3059kg)
LCM19T002-M

N

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

380±20Ω

Output terminal resistance

350±3.5Ω

Insulation resistanceGreater than 5000MΩ
at DC50V

B

E

L

Additional : 4 washers
Use washers to secure
the base.
H

M

F

Compensated temperature range

6

C

8
ø2
ø13

H
4–ø18
(for lower base installation)–10ºC to 40ºC

Temperature effect on

D

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Combined error

Zero balance

0.011% of R.O./10ºC Typ.

Span

0.013% of Load/10ºC Typ.

Cable diameter/lengthø4/3m

Dustproof/waterproof

IP67

4–ø14
(for upper plate installation)

I

A

J

G

P

Model No.
LCM19K500-M
LCM19T001-M
LCM19T1.5-M
LCM19T002

Unit = mm

O

K

A

B

C

F

G

H

O

P

Net Weight

97

208

104 70 55 136

138

17 20 23 25 70 15 17 63

118

5.6

116

242

118

145

17 24 26 25 84 15 19 63 152

8.6

D

E

84 48 175

I

J

K

L

M

N

Net Weight = kg17

Beam Type

LC-4221 Series

Hermetically Sealed, Beam Type Stainless Steel Load Cell
Ideal for measurements in harsh environments
Model No.
LC-4221-K010

10.20kg (100N)

Capacity

LC-4221-K020

20.39kg (200N)

LC-4221-K050

50.99kg (500N)

LC-4221-K100

102.0kg (1kN)

LC-4221-K200

203.9kg (2kN)

LC-4221-K300

305.9kg (3kN)

Specifications
Rated output

2.0394mV/V±0.2%

Safe load limit

200% of R.C.

Combined error

0.05% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±3.5Ω

Insulation resistanceGreater than 5000MΩ
at DC50V

Compensated temperature range–10ºC to 60ºC

Temperature effect on
120
C

D

ø26

B

ø38

A

Two face
width 12CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

20

F

2-E

12

60

36

24

20

3

20

10

M8(only K010-K050)

Unit = mm
Net Weight : 0.8kg

Model No.

LC4221-K010/020/050
LC4221-K100/200/300

18

A

27
25

B

18
20

C

65
65

D

10
10

E

ø6.6
ø8.4

F

ø8.1

ø10.1

Zero balance

0.07% of R.O./10ºC

Span

0.02% of Load/10ºC Typ.

Cable diameter/length

ø6/5m

Dustproof/waterproof

IP67

Beam Type

LC-5206 Series

Beam Type Tool Steel Load Cell

Model No.
LC-5206-K300

305.9kg (3kN)

Capacity

LC-5206-K500

509.9kg (5kN)

LC-5206-T001

1.020t (10kN)

LC-5206-T002

2.039t (20kN)

Specifications
Rated output

2.0394mV/V±0.2%

Safe load limit

200% of R.C.

Combined error

0.02% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±3.5Ω

Insulation resistanceGreater than 5000MΩ
at DC50V

Compensated temperature range–10ºC to 60ºC

Temperature effect on

Model No.

LC5206-T001/T002
Unit = mm

0.07% of R.O./10ºC

Span

0.02% of Load/10ºC Typ.

Cable diameter/lengthø6/3m (K300/K500)
ø6/5m (T001/T002)

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

LC5206-K300/K500

Zero balance

Dustproof/waterproof

A

B*

185

10

137

10

C

52

D

55

E

30

F

31

G

19

88 57.5 39.5 43.2 35

H

16

I

25

J

K

L

N

P

Net Weight (kg)

17 72.5 5.5

0.8

1.7

83 10.5 105

19 44.5 102.5 17

M
–

–

–

IP67

0.7

B* : Approximately

19

Beam Type

LC-5223 Series

Beam Type Tool Steel Load Cell

Model No.
LC-5223-K500

509.9kg (5kN)

Capacity

LC-5223-T001

1.020t (10kN)

LC-5223-T1.5

1.530t (15kN)

LC-5223-T002

2.039t (20kN)

LC-5223-T003

3.059t (30kN)

LC-5223-T005

5.099t (50kN)

Specifications
Rated output

2.0394mV/V±0.2%

Safe load limit

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±3.5Ω

Insulation resistance

5000MΩ/DC50V

Compensated temperature range–10ºC to 60ºC

Temperature effect on

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Zero balance

0.07% of R.O./10ºC

Span

0.02% of Load/10ºC

Cable diameter/lengthø4/3m

Dustproof/waterproof

IP67

SpacerModel No.

A

B

C

D

E

F

H

J

130

16

25

76

13

32

52

171

19

38

95

19

38

76

ø21

LC5223-K500
LC5223-T001
LC5223-T1.5
LC5223-T002
LC5223-T003
LC5223-T005
Unit = mm

20

K

L

Net Weight (kg)M12
ø14 P=1.75
Depth16

26

1

M20
P=1.5
Depth1938

1.5

Beam TypeLBP Series

Hermetically Sealed Beam Type Load Cell

Model No.
LBP-10L

10.20kg (100N)

Capacity

LBP-20L

20.39kg (200N)

LBP-30L

30.59kg (300N)

LBP-50L

50.99kg (500N)

LBP-100L

102.0kg (1kN)

LBP-200L

203.9kg (2kN)

LBP-300L

305.9kg (3kN)

LBP-500L

509.9kg (5kN)

LBP-1

1.020t (10kN)

LBP-2

2.039t (20kN)

LBP-5

5.099t (50kN)

LBP-1010.20t (100kN)

Specifications

øQH

J Drill(✕2)

HøG

Cable out location for
200L to 10L

Metal bellows(positive-bellows-protection)
I

D

C

øF
øP

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD(E)

B

Rated output

2.0394mV/V+0.2% –0%

Safe load limit

150% of R.C.

Combined error

0.04% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC20V

Zero balance±1% of R.O.

Input terminal resistance

420±40Ω

Output terminal resistance

347.5±3Ω

Insulation resistance

2000MΩ/DC50V

Compensated temperature range–10ºC to 75ºC

Temperature effect on

Cable out location for 10L to 100L

N

M

ø7 Cable

K

Load direction+ Polarity

Zero balance

0.02% of R.O./10ºC

Span

0.02% of Load/10ºC

Cable diameter/lengthø7/3m

Dustproof/waterproof

IP67

(L)
A

Model No.

A

B

LBP-10L ~ 100L

125 72

C

D
8

E

10

øF øG

LBP-200L ~ 500L

200 120 45

2042

12 8 Reamer 8.5 Drill(✕2) 53

20

15

20

30

52

14 12 Reamer 14 Drill(✕2) 80

40

LBP-1

210 125 50

15

20

40

52

15 16 Reamer 18 Drill(✕2) 90

LBP-2

215 125 50

20

20

43

58

16 ø26 Spotface 22 Drill(✕2) 95

35

H

I

J

K

L

M

Unit = mm

35

N

15

øP øQ Net Weight (kg)40

25

52 50.5

2

40

40

30

68

65

2.8

40

50

34

68

65

4

20 ø32 Spotface 22 Drill(✕2) 170 50

60

50

92

90

8

70

56 108 90

1242 38.5

0.9

20 Drill
3 Depth
26 Drill
LBP-5

300 135 120 20

25

61

72

3 Depth
20 Reamer
LBP-10300 118 135 22

25

68

82

+0.2
25 ø38+0.1
Spotface 32 Drill(✕2) 187 52

10 Depth21

Tension TypeTension type load cells are ideal for tensile
measurements (compression measurements are also possible).
Installation is easy (although safety precautions must be observed).
The structure is less susceptible to biased loading.
Example Applications●Hopper scales●Tension measurements, etc.
Example ofInstallation

Load cell bracket(stationary side)
Protective hardware

The LC-1205

Load cell
Rod-end bearing
Joint

Load cell bracket(load side)22

Tension Type

LC-1205 Series

S-Type Tool Steel Load Cell

Model No.
LC1205-K020

20.39kg (200N)

Capacity

LC1205-K050

50.99kg (500N)

LC1205-K100

102.0kg (1kN)

LC1205-K200

203.9kg (2kN)

LC1205-K500

509.9kg (5kN)

LC1205-T001A

1.020t (10kN)

LC1205-T002

2.039t (20kN)

LC1205-T005

5.099t (50kN)

Specifications
Rated output

2.0394mV/V±0.5%1.5296mV/V±0.5% (K020/K050)

Safe load limit

200% of R.C.
250% of R.C. (K020/K050)

Rod-end bearings are sold separately.

CABLE COLOR CODE (Tension)
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

A

B

C

D

M✕P✕Depth

16M6✕1✕1019

23

M12✕1.75✕10

0.5

24

28

M18✕1.5✕22.5

1.3

64

19

23

LC1205-K200/K500

50
75

64

100

LC1205-T002

75

100

LC1205-T005

DC5 to12V

Maximum excitation voltage

DC15V

Zero balance±2% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±3.5Ω

Insulation resistance

5000MΩ/DC50V

Compensated temperature range–10ºC to 60ºC

Zero balance

0.07% of R.O./10ºC

Span

0.02% of Load/10ºC Typ.

Cable diameter/length

ø6/3m
ø6/5m (T001A/T002/T005)

50

LC1205-T001A

Recommended excitation voltage

Dustproof/waterproof

LC1205-K020/K050
LC1205-K100

0.02% of R.O.

Temperature effect on

CABLE COLOR CODE (Compression)
RED
EXC +
WHITE EXC –GREEN SIG –
BLUE
SIG +
YELLOW SHIELD

Model No.

Combined error

50

75

64

100

12
24
36

28
40

M6✕1✕11M18✕1.5✕25
M24✕2✕21

IP54

Net Weight (kg)
0.3
0.4
1.3
1.9

Unit = mm

23

Tension Type

LC-1216 Series

Hermetically Sealed Tool Steel Load Cell
Ideal for measurements in harsh environments
Model No.
LC1216-K100

102.0kg (1kN)

Capacity

LC1216-K200

203.9kg (2kN)

LC1216-K500

509.9kg (5kN)

LC1216-T001A

1.020t (10kN)

LC1216-T002A

2.039t (20kN)

LC1216-T005

5.099t (50kN)

Specifications
Rated output

2.0394mV/V±0.5%

Safe load limit

200% of R.C.

Combined error

0.05% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±2% of R.O.

Input terminal resistanceApproximately 400Ω

Output terminal resistance

350±3.5Ω

Insulation resistance

5000MΩ/DC50V

Compensated temperature range–10ºC to 60ºC

Temperature effect on

CABLE COLOR CODE (Tension)
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Zero balance

0.07% of R.O./10ºC

Span

0.02% of Load/10ºC Typ.

Cable diameter/length

ø6/3m
ø6/5m (T001A/T002A/T005)

Dustproof/waterproof

CABLE COLOR CODE (Compression)
RED
EXC +
WHITE EXC –GREEN SIG –
BLUE
SIG +
YELLOW SHIELD

Model No.

B

øC

62

77

H

M✕P✕Depth

16M12✕1.75✕10

31

20
20

M18✕1.5✕13

M18✕1.5✕13

4.0

41

34

M24✕2✕25

5.6

19

16

80

102

75

63

27

80

102

90

63

36

LC1216-T002A/005

180

Unit = mm

G

16

150
150

F

45

LC1216-T001A(LC1216-T002)

E

50

62

104

D

77

104

LC1216-K200/K500

24

A

LC1216-K100

80

102

50
75

45
63

16
27

19
31

M6✕1✕9Net Weight (kg)
1.7
1.8
4.0

IP67

Tension TypeLCS15 Series

S-Type Stainless Steel Load Cell

Model No.
LCS15K500

509.9kg (5kN)

Capacity

LCS15T001

1.020t (10kN)

LCS15T002

2.039t (20kN)

Specifications
Rated output

2mV/V±0.1%

Safe load limit

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

380±20Ω

Output terminal resistance

350±3.5Ω

Insulation resistance

5000MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on
Zero balance

0.016% of R.O./10ºC

Span

0.013% of Load/10ºC Typ.

Cable diameter/lengthø5/4.5m

Dustproof/waterproof

IP67*When using a stainless steel rod end bearing,
set the normal load to 1/2 of the rated capacity.

Stainless steel rod end bearing

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD(Sold separately)

Protective tube

Model No.

Model No.

A

B

C

øD

M✕P✕Depth

Net Weight (kg)88

M12✕1.75✕1680

88

M18✕1.5✕15

1

100

108

M24✕2✕24

2

4

20

80

LCS15T001

4.5

25

LCS15T002

5.5

35

LCS15K500

0.8

Intended
StaticStatic
Net Weightload cells radial load fracture load(g)

LCZARE12S LCS15K500

5.88kN 23.53kN

85

LCZARE18S LCS15T001 11.52kN 46.09kN

258

LCZARE24S LCS15T002

730

17.4kN 69.62kN

Unit = mm

25

Compression Type
Compression type load cells are used for compression
measurements and are suitable for large capacity
weighing.
Example Applications●Truck scales●Weigh bridges●Hopper scales●Tank scales●Compression measurements, etc.

Example of Installation
Truck scale

StopperMaximum clearance : 5mm
Recommended clearance : 2 to 3mm
Reinforced plate
Shunt wire
Base plate
Mutual level error : within 3mm
Self level : within 1/500
Slope of bottom : more than 1/100

26

Use steel conduit
and flexible tube forload cell cableCompression TypeLCC07 Series

Hermetically Sealed Stainless Steel Load Cell
Equipped with a transverse load stopper
Model No.
LCC07K500

509.9kg (5kN)

Capacity

LCC07T001

1.020t (10kN)

LCC07T002

2.039t (20kN)

LCC07T003

3.059t (30kN)

LCC07T005

5.099t (50kN)

Specifications
Rated output

2mV/V±0.1%

Safe load limit

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

780±20Ω

Output terminal resistance

700±10Ω

Insulation resistance

500MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on
CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

100±1

8-ø12

0.04% of R.O./10ºC

Span

0.014% of Load/10ºC Typ.

Cable diameter/lengthø5.6/5m

Dustproof/waterproof

IP67

77

100±1

77

Zero balance

3

95

16

SUS BallR1/4-1912

ø78

Unit = mm
Net Weight : 4kg

27

Compression TypeLCC11/N Series

Hermetically Sealed Stainless Steel Load Cellwith Very Large Weighing Capacity
High resistance to adverse environmental conditions and corrosion (IP68)
Available with or without mounting kit
Model No.
LCC11T010/LCC11T010N

Capacity
10.20t (100kN)

LCC11T020/LCC11T020N

20.39t (200kN)

LCC11T030/LCC11T030N

30.59t (300kN)

Specifications
Rated output

2mV/V±0.1%

Safe load limit

200% of R.C.

Combined error

0.016% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

800±80Ω

Output terminal resistance

2200±10ΩInsulation resistance

5000MΩ/DC50V

Compensated temperature range–20ºC to 60ºC

Temperature effect on

200±1
140

8-ø20
(Upper/lower shared)

100
150±1

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Load

36 12

Ground terminal M4
12
84
ø72

28

47

42

103

199±1

Four-conductor
shielded cable

Upper dust cover

Under boots
108

12 36

100

PT-1/2

ø72
ø94
Load

28

ø99
Unit = mm
Net Weight : 7.5kg

Zero balance

0.019% of R.O./10ºC Typ.

Span

0.010% of Load/10ºC Typ.

Cable diameter/lengthø8/12mDustproof/waterproof

IP68 (100h at 1.5m immersion)

Compression TypeLCC12 Series

Hermetically Sealed Stainless Steel Weighing Modulewith Very Large Weighing Capacity
High resistance to adverse environmental conditions and corrosion (IP68)
Equipped with mount assembly (transverse movement control + lift-off prevention)
Model No.
LCC12T010

10.20t (100kN)

Capacity

LCC12T020

20.39t (200kN)

LCC12T030

30.59t (300kN)

Specifications
Rated output

2mV/V±0.1%

Safe load limit

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

800±80Ω

Output terminal resistance

2200±10ΩInsulation resistance

5000MΩ/DC50V

Compensated temperature range–20ºC to 60ºC

Temperature effect on

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

8-ø18

120±0.5

Unit = mm
Net Weight : 23kg

Zero balance

0.019% of R.O./10ºC Typ.

Span

0.010% of Load/10ºC Typ.

Cable diameter/lengthø8/12mDustproof/waterproof

IP68 (100h at 1.5m immersion)

Mount fixture
Max. transverse force Max. force for lift
75kN
80kN

160±1.8120±0.5130±1.8

LCC12 Series

270±2.2225±118Fourconductor
shielded
cablePT-1/218

127

GroundterminalM4

300±2.2

29

Compression Type

LC-2224 Series

Column Type Load Cell Ideal for Truck Scales and Hoppers

Model No.
LC2224-T010

10t (98kN)

Capacity

LC2224-T020

20t (196kN)

LC2224-T030

30t (294kN)

LC2224-T040

40t (392kN)

LC2224-T050

50t (490kN)

Specifications
Rated output

2mV/V±0.1%

Safe load limit

200% of R.C.

Combined error

0.02% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±1% of R.O.

Input terminal resistance

410±30Ω

Output terminal resistance355±3.5Ω

Insulation resistance

5000MΩ/DC50V

Compensated temperature range–20ºC to 60ºC

Temperature effect on

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Model No.
LC2224-T010
LC2224-T020
LC2224-T030
LC2224-T040
LC2224-T050

Unit = mm

30

Zero balance

0.009% of R.O./10ºC

Span

0.012% of Load/10ºC Typ.

Cable diameter/lengthø8/10mDustproof/waterproof

IP67

The top and bottom plate
dimensions are the same.

A

B

C

D

E

F

G

H

J

K

L

Net Weight (kg)

200

140

58

92

157

33

29

199±1

131

22

104

12

200

140

78

92

186

43

39

228±1

158

23

126

15

220

170

89

112

205

49

45

247±1

171

26

138

18

Compression TypeCP(Compression) / TP(Tension) Series

Hermetically Sealed Tension and Compression Load Cell
Ideal for measurements in harsh environments
Capacity
20.39kg (200N)
50.99kg (500N)
102.0kg (1kN)
203.9kg (2kN)
305.9kg (3kN)
509.9kg (5kN)
1.020t (10kN)
2.039t (20kN)
3.059t (30kN)
5.099t (50kN)
10.20t (100kN)
20.39t (200kN)
30.59t (300kN)
50.99t (500kN)
102.0t (1MN)

Model No.
TP (CP)-20L
TP (CP)-50L
TP (CP)-100L
TP (CP)-200L
TP (CP)-300L
TP (CP)-500L
TP (CP)-1
TP (CP)-2
TP (CP)-3
TP (CP)-5
TP (CP)-10
TP (CP)-20
TP (CP)-30
TP (CP)-50
TP (CP)-100

Specifications
Rated output

3.0592mV/V±0.1%

Safe load limit

150% of R.C.
500% of R.C.(20L to 100L)

Combined error

0.05% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC20V

Zero balance±1% of R.O.

Input terminal resistance

350±3.5Ω

Output terminal resistance

350±5Ω

Insulation resistance

5000MΩ/DC50V

Compensated temperature range–10ºC to 75ºC

Temperature effect on

CP

36
75

Model No.
PT 1/2 14 threads per inch

TP

TP-20L ~ 100L

CP-20L ~ 100L
TP-200L ~ 1

CP-200L ~ 1

ø9.7 Cable

TP-2 ~ 5(G)
Adapter

TP-10

J

TP-20

(E)

F

øI

CP-2 ~ 5

øA

42

B

l radiu
herica
H (sp
s)
J

1.5

C

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG –
BLUE
SIG +
YELLOW SHIELD

0.03% of R.O./10ºC

Span

0.03% of Load/10ºC

Cable diameter/lengthø9.7/3m

Dustproof/waterproof

IP67

For the 200L model and higher, we can also manufacture two-bridge, low-temperature, andhigh-temperature models.

(16)

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Zero balance

øD

CP-10
CP-20
TP-30

CP-30
TP-50

CP-50

TP-100CP-100

Unit = mm

øA

B

C

øD

E

F

Spherical(G) Radius øI
H

89

115

18

32

6

7

108

50

12

M10✕1.25 Depth 14

3.1

89

115

32

7

9

108

50

14

M12✕1.25 Depth 14

3.2

89

150

44

65

10

17

108 100

26

M24✕2 Depth 30

4.8

125 215

88

100

12

28

150 20042

M39✕2 Depth 50

10.2

165 290

78

145

15

28

194 20058

M50✕2 Depth 65

27.0

203 360 145 1722340

233 30070

M64✕3 Depth 90

50.0

232 452 145 19530

48

263 300

82

M76✕3 Depth 115

88.0

310 610 180 262

30

70

343 300 108 M100✕3 Depth 150

200.0

23
28

J

Net Weight (kg)31Compression Type

C2F1/C2Z1/C2X1 Series

Hermetically Sealed Compression Load Cell
Ideal for measurements in harsh environments

Model No.
C2F1 (C2Z1)-500L
C2F1 (C2Z1)-1
C2F1 (C2Z1)-2
C2F1 (C2Z1)-5
C2F1 (C2Z1)-10
C2F1 (C2Z1)-20
C2X1-30
C2X1-50
C2X1-100
C2X1-200
C2X1-300
C2X1-500

C2F1 Specifications

C2Z1 Specifications

C2X1 Specifications

Rated output

2.0394mV/V±0.4%

Rated output

2.0394mV/V±0.4%

Rated output

2.0394mV/V±1%

Safe load limit

150% of R.C.

Safe load limit

500% of R.C.

Safe load limit

150% of R.C.

Combined error

0.3% of R.O.

Combined error

0.2% of R.O.

Combined error

0.4% of R.O.

Recommended excitation voltage

DC5 to 12V

Recommended excitation voltage

DC5 to 12V

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC20V

Maximum excitation voltage

DC20V

Maximum excitation voltage

DC20V

Zero balance±1% of R.O.

Zero balance±1% of R.O.

Zero balance±1% of R.O.

Input terminal resistance

350±3.5Ω

Input terminal resistance

350±3.5Ω

Input terminal resistance

350±3.5Ω

Output terminal resistance

350±5Ω

Output terminal resistance

350±5Ω

Output terminal resistance

350±5Ω

Insulation resistance

1000MΩ/DC50V

Insulation resistance

1000MΩ/DC50V

Insulation resistance

1000MΩ/DC50V

Compensated temperature range–10ºC to 75ºC

Compensated temperature range–10ºC to 75ºC

Compensated temperature range–10ºC to 75ºC

Temperature effect on

Temperature effect on

Temperature effect on

Zero balance

0.05% of R.O./10ºC

Zero balance

0.05% of R.O./10ºC

Zero balance

0.05% of R.O./10ºC

Span

0.1% of Load/10ºC

Span

0.1% of Load/10ºC

Span

0.1% of Load/10ºC

Cable diameter/lengthø9.7/3m

Cable diameter/lengthø9.7/3m

Cable diameter/lengthø9.7/3m

Dustproof/waterproof

IP67

Dustproof/waterproof

IP67

Dustproof/waterproof

IP67

36

(F)

Unit = mm

75(16)

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

ø9.7 Cable

PT 1/2 14 threads per inch

(H)
B

)

J
L

øA

B

C2Z1-500L ~ 2

113

85

50

C2Z1-5 ~ 10

138 110 100

C2Z1-20164 145 200

C2X1-30, 50

138 160 200

C2F1-5 ~ 10
C2F1-20

C2X1-100
P.C.D.

DøG

C2X1-200
4-E

Spherical P.C.D
RadiusD
C

Model No.
C2F1-500L ~ 2

ical radius

C (spher

42

øA
øI

32

Capacity
509.9kg (5kN)
1.020t (10kN)
2.039t (20kN)
5.099t (50kN)
10.20t (100kN)
20.39t (200kN)
30.59t (300kN)
50.99t (500kN)
102.0t (1MN)
203.9t (2MN)
305.9t (3MN)
509.9t (5MN)

C2X1-300
C2X1-500

98

105

50

98

100 100

120 120 150

F

øG

H

J

L

50

4-M6 Depth 10 138

80

10 M16✕1.5 27

1.5

60

4-M8 Depth 15 165 100

15

1.5

80 4-M12 Depth 20 193 120

20

50
60

E

4-M6 Depth 10 120
4-M8 Depth 15 120

76
76

80 4-M12 Depth 20 145 100
80 4-M12 Depth 20 165 120

164 190 200 100 4-M16 Depth 20 193 140
214 230 300 130 4-M16 Depth 25 245 180
214 320 600 130 4-M24 Depth 35 245 180
316 400 800 200 6-M30 Depth 50 349 260

9

10
10
20
30
30
30
30

øI

M16✕1.0 36
35

44

45

66

24
36
50
70

32

1.5

170

200

8
3

10

2

34

2

150

2

2

80

100

4

14

1.5

72

Approximately

2

45

95

120

1.5

Net Weight (kg)

2
2

6

18
50

140

Compression Type

CMX/CM Series

Hermetically Sealed Compression Load Cells
Miniature size and waterproof light weight construction
Model No.
CMX-50L

50.99kg (500N)

Capacity

CMX-100L

102.0kg (1kN)

CMX-200L

203.9kg (2kN)

CMX-500L

509.9kg (5kN)

CMX-1

1.020t (10kN)

CMX-2

2.039t (20kN)

CM-5

5.099t (50kN)

CM-10

10.20t (100kN)

CM-20

20.39t (200kN)

Specifications
Rated output

2.0394mV/V±0.4% (CMX)
2.0394mV/V±0.5% (CM)150% of R.C.

Combined error

0.2% of R.O. (CMX)
0.25% of R.O. (CM)

Unit = mm

3

19
83

20

dius 50

1.5

25
22

CMX

spherical ra

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V (CMX), 18V (CM)

Zero balance±2% of R.O.

Input terminal resistance425±50Ω

Output terminal resistance

350±5Ω

Insulation resistance

1000MΩ/DC50V

Compensated temperature range–10ºC to 60ºC

Temperature effect on
40

P.C.D.42
M5✕4, 6 Depth

Zero balance

0.1% of R.O./10ºC

Span

0.1% of Load/10ºC

Cable diameter/lengthø5/3m (CMX)
ø8.6/3m (CM)Net Weight : 0.8kg

Dustproof/waterproof

IP67

ø5 Cable

CM
5

ø114

(47)

(16)

ø25

PT 1/2 14 threads per inch

20

100

al radius

spheric

ø22

ø8.6 Cable
50

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

ø62
ø12

Safe load limit

P.C.D. 90
M8✕4, 15 Depth

Net Weight : 3.7kg

33

Compression Type

TM/UM Series

Hermetically Sealed Load Cells for Multiple Applications
Tension-TM, Tension and Compression-UM
Model No.
TM (UM)-50L-A

50.99kg (500N)

Capacity

TM (UM)-100L-A

102.0kg (1kN)

TM (UM)-200L

203.9kg (2kN)

TM (UM)-500L

509.9kg (5kN)

TM (UM)-1

1.020t (10kN)

TM (UM)-2

2.039t (20kN)

TM (-A) Specifications

UM

TM

TM

UM

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG –
BLUE
SIG +
YELLOW SHIELD

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Rated output

2.0394mV/V±0.5%

Safe load limit

150% of R.C.

Combined error

0.2% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC18V

Zero balance±10% of R.O.

Input terminal resistance425±50Ω

Output terminal resistance

350±5Ω

Insulation resistance

1000MΩ/DC50V

Compensated temperature range–10ºC to 60ºC

Temperature effect on
(62)

øA
øG
1.6 Slot

Zero balance

0.1% of R.O./10ºC

Span

0.1% of Load/10ºC

Cable diameter/lengthø5/3m

Dustproof/waterproof

IP67

UM (-A) Specifications
D

Rated output

E
C

2-H
1.6 SlotøGModel No.
TM-50L-A, 100L-A
UM-50L-A, 100L-A
TM-200L
UM-200L
TM-500L ~ 2
UM-500L ~ 2

Unit = mm

34

øA

B

C

D

E

F

øG

H

Net Weight (kg)

50

70

43

27

23

10

10

M12✕1.75

0.3

50
62

70
85

40
45

30
40

20
20

13
18

10
13.5

M12✕1.75
M16✕2

2.0394mV/V±1% (Tension)
2.0394mV/V±3% (Compression)

(ø12)

B

F

Cable shield ø5 Cable

0.5
0.9

Safe load limit

150% of R.C.

Combined error

0.3% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC18V

Zero balance±10% of R.O.

Input terminal resistance425±50Ω

Output terminal resistance

350±5Ω

Insulation resistance

1000MΩ/DC50V

Compensated temperature range–10ºC to 60ºC

Temperature effect on
Zero balance

0.1% of R.O./10ºC

Span

0.1% of Load/10ºC

Cable diameter/lengthø5/3m

Dustproof/waterproof

IP67

Button Load Cell

LCC21 Series

Stainless Steel Compact Compression Load CellCan be used for load distribution and press pressure measurement
Model No.
LCC21N100

10.20kg (100N)

Capacity

LCC21N200

20.39kg (200N)

LCC21N500

50.99kg (500N)

LCC21KN001

102.0kg (1kN)

Specifications
Rated output

1mV/V or more

Safe load limit

150% of R.C.

Combined error

0.5% of R.O.

Recommended excitation voltage

DC5V

Maximum excitation voltage

DC5V

Zero balance±100% of R.O.

LCC21N100/N200

Input terminal resistance1kΩ ±0.1kΩ

LCC21N500/KN001

Output terminal resistance1kΩ ±0.01kΩ

Insulation resistance

500MΩ/DC50V

Compensated temperature range

0ºC to 70ºC

Permissible temperature range–10ºC to 80ºC

Standard attachment holder

Temperature effect on

Stainless steel load cell
B

Attachment holder
G
H

Attachment surface

Model No.
LCC21N100/N200
LCC21N500/KN001

Unit = mm

A
B C
ø10 18 4
ø16 24 7

Span

0.5% of Load/10ºC Typ.

Cable diameter/length

ø2mm/2m

Dustproof/waterproof

IP64

Include standard attachment holder 1 piece
Resonance frequency
LCC21N100

LCC21N200

LCC21N500

LCC21KN001

45kHz

55kHz

30kHz

35kHz

C

F

E

D

I

Force direction

2-ø3.5

0.5% of R.O./10ºC Typ.

J

A

ø2

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
YELLOW SHIELD

Zero balance

D
3.6
6

E
1.6
3

Stainless steel
thickness : 1.5

F
G H I J Net Weight (g)SR3 16 8 28 20 Approx. 15
SR6 22 14 32 24 Approx. 20

35

Button Load Cell

LCCA21 Series

Stainless Steel Compact Compression Load CellCan be used for load distribution and press pressure measurement
Model No.
LCCA21N100

10.20kg (100N)

Capacity

LCCA21N200

20.39kg (200N)

LCCA21N500

50.99kg (500N)

LCCA21KN001

102.0kg (1kN)

Specifications
Safe load limit

150% of R.C.

Combined error

0.5% of R.O.

Supply voltage

DC5V±0.25V less than 16mA

Load resistancemore than 5kΩ

Output voltage

1V to 4V *1

Zero voltage

1V±0.04V

Span voltage

2V±0.01V (1V to 3V)

Insulation resistance

500MΩ/DC50V
between signal wires and load cell

Compensated temperature range

0ºC to 70ºC

Permissible temperature range–10ºC to 80ºC

Temperature effect onAGH

2-ø3.5I
J

Force direction

Attachment holder

Stainless steel
thickness : 1.5

C

B

F

D

Stainlesssteelload cell

E

CABLE COLOR CODE
RED
Power +WHITE Power –GREEN Output
BLUE
GND
YELLOW SHIELD

Attachment surface

0.6% of R.O./10ºC Typ.

Span

0.6% of Load/10ºC Typ.

Frequency characteristics

100Hz (–3dB)

Cable diameter/length
From load cell to amplifier

ø2mm/2m

From amplifier

ø6mm/0.5m

Dustproof/waterproof

Load cell IP64

Include standard attachment holder 1 piece andamplifier case holder 1 set

99

32

Amplifier

Zero balance

Resonance frequency
LCCA21N100 LCCA21N200 LCCA21N500 LCCA21KN001
45kHz
13

Nickel plating

Amplifier case holder

Attachment Diagram
14

77

35

45

10

2-ø3.5Model No.
LCCA21N100/N200
LCCA21N500/KN001

Unit = mm

36

A
B C
ø10 18 4
ø16 24 7

D
3.6
6

55kHz

30kHz

35kHz

*1

E
1.6
3

F
G H I J Net Weight (g)SR3 16 8 28 20 Approx. 65
SR6 22 14 32 24 Approx. 70* Net Weight includes the amplifier and cables

Load [% of R.C.]

0%

100%

150%

Output voltage

1V

3V

4V

Standard attachment holder

Flameproof Type
Flameproof load cell series with TIIS certificates
In factories and locations where explosive fumes, oils, and other dangerous flammable items are handled,
products with flameproof constructions must be used to prevent fires from electrical sparks and heat.
A&D load cells have passed type approval of electrical equipment for use in explosive atmospheresby the Technology Institution of Industrial Safety (TIIS).

Flameproof and intrinsically safe flameproof enclosures●Flameproof enclosureThis enclosure is totally enclosed. If gas or vapor enters the interior of this enclosure and causes an explosion,
the enclosure withstands the force of the explosion, and thereby prevents fires caused by the explosion from
igniting gas or vapor outside the enclosure.
●Intrinsically safe flameproof enclosureThis enclosure is equipped with technical measures to prevent electrical sparks and heat in the electrical circuit of
the equipment from igniting fires in explosive atmospheres.
●Example of displayed ratings
and their meanings

Indicates an flameproof productTypes of flameproof constructions(d: flameproof; i: intrinsically safe flameproof)
1. Electrical equipment group (or explosion class)
2. Temperature class (or ignition temperature)
1. Electrical equipment group in TIIS

Types and explosion classes of gases and vapors

Group I indicates flameproof equipment for coalmines, and Group II indicates equipment for otherlocations. Groups are further classified as A, B, orC based on applicable gases or vapors.

Range of safety gaps for gases and vapors
0.9 mm or more
More than 0.5 mm and less than 0.9 mm
0.5 mm or less

2. Temperature class in TIIS

Temperature class Maximum surface temperature of flameproof equipment Temperature class Ignition temperature of gases and vapors
T1
Items of 450°C or less
G1
Items exceeding 450°C
T2
Items of 300°C or less
G2
Items exceeding 300°C
T3
Items of 200°C or less
G3
Items exceeding 200°C
T4
Items of 135°C or less
G4
Items exceeding 135°C
T5
Items of 100°C or less
G5
Items exceeding 100°C
T6
Items of 85°C or less
G6
Items exceeding 85°C

In the past, ignition temperatures were classified
by the gas or vapor ignition temperature,but international standards now classify themby the maximum surface temperature of theflameproof equipment and are noted in thefollowing manner.

3. Typical explosion classes and ignition temperatures
Ignition
temperatures
Explosionclasses

1

2
3

G1

G2

G3

G3

AcetoneEthanol
Gasoline
Acetaldehyde
Ammonia
Butanol
HexaneEthyl ether
EthaneButanePentaneToluene Carbon dioxide
Acetic acid
MethanePropaneCoal gas
Ethylene
Isoprene
Ethylene oxide Hydrogen sulfide
Hydrogen
Acetylene

Class
A
B
C

Group
IIA
IIB
IIC

Explosion class1
2
3●Classification by location of use in TIIS
Locations (zones) that require flameproof measures areclassified from 0 to 2 by the degree of hazardous
atmosphere. It is necessary to select the equipment
according to these location classes.
Zone 0: A location where a dangerous atmosphere iscontinually present or present for a long period
under normal conditions.
Zone 1: A location where a dangerous atmospheremay be generated under normal conditions.
Zone 2: A location where a dangerous atmospheremay be generated under abnormal conditions.
37

Flameproof TypeLCB01/02/06 Series

Flameproof Single-point Load Cells

Flameproof class TIIS ExdIIBT4 compliant
Model No.
LCB01K030E

30.59kg (300N)

Capacity

LCB01K060E

61.18kg (600N)

LCB02K150E

153.0kg (1.5kN)

LCB06K300E

305.9kg (3kN)

LCB06K600E

611.8kg (6kN)

Specifications
Rated output

0.3059mV/V±10% (LCB01/02)
0.55mV/V±10% (LCB06)

LCB01/02 Series

Safe load limit

150% of R.C.

Combined error

0.03% of R.O.

Recommended excitation voltage

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance

20±5% of R.O.

Input terminal resistance418±20Ω (LCB01/02)
410±20Ω (LCB06)

Output terminal resistance

348.5±5Ω (LCB01/02)
350±5Ω (LCB06)

Insulation resistance

5000MΩ/DC50V

Compensated temperature range–10ºC to 40ºC

Temperature effect on
Zero balance

0.05% of R.O./10ºC

Span

0.03% of Load/10ºC Typ.

Maximum platform size

400✕400mm (LCB01)400✕600mm (LCB02)

LCM06 Series

Double-sided
width 24

M27×1

40

287
165

40 18

124
150

24

Female screw PF 1/2(Nominal diameter ofcompatible conduits: 16)

F

4-ø13

75

K

J

L
O

Model No.
LCB01K030E/060E
LCB02K150E

38

Unit = mm

A
165
210

B
65
76

C
M6
M8

D
46
56

E
60
70

M

N

F
G H
I
193 75 65 81
238 82.5 72.5 89

M

J
10
7

106

130.6

22

82.3
J

40.3

G

I

H

22

ø38

4-M12

13

D

C

44

B

ø57

52~56

LCB06 Series

A

IP54

E

LCB01/02 Series

ø7/6m

Dustproof/waterproof

13

600✕700mm (LCB06)
Cable diameter/length

Earth terminal M4

K
50
61

L
70
80

M
6
7

N
28
61

100

O Net Weight (kg)
180
2.2
4.9
230

LCB06 Series Net Weight : 10kg
CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –

ORANGE Sensing +BLACK Sensing –YELLOW SHIELD

Flameproof TypeLCM17/LCM17-M Series

Flameproof Beam Type Stainless Steel Beam Load Cells
Flameproof class TIIS ExdIIBT4 compliant

LCM17 Series

Model No.
LCM17K100E

102.0kg (1kN)

Capacity

LCM17K200E

203.9kg (2kN)

LCM17K300E

305.9kg (3kN)

LCM17K500E

509.9kg (5kN)

LCM17T001E

1.020t (10kN)

LCM17T002E

2.039t (20kN)

Steady-arm models with anti-uplift mechanism(LCM17 unit and steady arm)

Model No.
LCM17K100E-M

102.0kg (1kN)

Capacity

LCM17K200E-M

203.9kg (2kN)

LCM17K300E-M

305.9kg (3kN)

LCM17K500E-M

509.9kg (5kN)

LCM17T001E-M

1.020t (10kN)

LCM17T002E-M

2.039t (20kN)

Specifications
Rated output

C

D

B

F

E

O
Female screw G1/2 (PF1/2)(Nominal diameter of compatible conduits: 16)(0.5mV/V±0.1% for 1kN)

CABLE COLOR CODE
RED
EXC +
WHITE
EXC –
GREEN
SIG +
BLUE
SIG –
YELLOWSHIELDGREEN / YELLOW EARTH

M

+0.2

N0

Q±2

R±2

P

K
J

ø38

Direction of force

Earth terminal M4

DC5 to 12V

Maximum excitation voltage

DC15V

Zero balance±2% of R.O.

Input terminal resistance390±30Ω

Output terminal resistance

350±5Ω

Insulation resistanceGreater than 5000MΩ
at DC50V

Compensated temperature range
Zero balance

0.028% of R.O./10ºC

Span

0.011% of Load/10ºC Typ.

Cable diameter/lengthø7/5m

Dustproof/waterproof

IP54

LCM17-M
Permissible horizontal force

Maximum 10kN (1019kg)

Permissible lifting force

Maximum 20kN (2039kg)

4.5±2

CABLE COLOR CODE
RED
EXC +
WHITE EXC –GREEN SIG +
BLUE
SIG –
GREEN / YELLOW EARTH

B

C

D

E

F

G

H

J

K

L

M

N

O

P

Q

R Net Weight

170

56

25

76

13

43

32

43

41

94

M12

ø21

ø14

20.5

88

54

2.8

212

60

38

95

19

48

38

48

46

117

M20

ø30.2

ø21

23

94

59

3.9

K

L

P

Q

R

S

Net Weight–10ºC to 40ºC

Temperature effect on

LCM17-M Series

(S)

N

M

B
F
L

Additional : 4 washers
Use washers to secure
the base.

E

8
ø2 3
ø1
6

B

C

118

238

124 90 42 140

180 17 24 16 25 125 15 56 54 157

54.5

38 324132274

124 90 42 172

180 17 24 16 25 125 15 60 54 193

57

43 360 11.2
Net Weight = kg

F

G

H

I

J

M

N

O

H
G

I

8.7

Q

A

øR

E

4–ø14 (for upper plate installation)

4–ø18(for lower base installation)

Net Weight = kgA

D

Female screw G1/2(PF1/2)(Normal diameter ofcompatible conduits 16)

J

Unit = mm

Model No.
LCM17K100E-M
LCM17K200E-M
LCM17K300E-M
LCM17K500E-M
LCM17T001E-M
LCM17T002E-M
Unit = mm

0.03% of R.O.

Recommended excitation voltage

D
CA
Model No.
LCM17K100E
LCM17K200E
LCM17K300E (256)
LCM17K500E
LCM17T001E
LCM17T002E (298)L

150% of R.C.

Combined error

HA

1.0mV/V±0.1%

Load limit

Double-sized width 24

LCM17 Series

H
G

Double-sided width 24

LCM17-M Series

P

O

K

39

Flameproof Type Equipment

AD-4386 Series of Flameproof Summing Boxes
Two to four flameproof load cells can be summed in parallel.
Boxes include an embedded potentiometer to adjust
the output of load cells.
●AD-4386-2, AD-4386-3, and AD-4386-4(for summing 2, 3, and 4 load cells, respectively)●Complies with TIIS EXdIIBT4 specifications for explosion class andignition temperature and is suitable for Zones 1 and 2●Certified under the Labor Safety and Health Law (TC13430 ) in Japan

AD-4387 Flameproof Junction Box
This extension connection box is used when an flameproof
load cell and indicator are separated.
Its compact size and light weight make operation easy.
●Complies with TIIS EXdIIBT4 specifications for explosion class andignition temperature and is suitable for Zones 1 and 2*There is no Labor Safety and Health Number because there is no electrical contact.

SB-FP Series Flameproof Platforms
Uses lightweight and compact single-point load cell●Complies with TIIS EXdIIBT4 specifications for explosion class andignition temperature and is suitable for Zones 1 and 2●Optional roller conveyor canModel No. Capacity Dimensions (D✕W✕Hmm) Platform Weight
be prepared. Conforms to
SB30K05-FP
30kg 290✕375✕140 13kg
i2G4 flameproof standard for SB60K05-FP 60kg 290✕375✕140 13kg
SB150K06-FP 150kg 390✕530✕170 15kg
use in Zones 0,1 and 2
SB300K07-FP 300kg 600✕700✕210 57kg
●Compliant with
SB600K07-FP 600kg 600✕700✕210 57kg
LCB01/LCB02/LCB06
Cable length : 6m Measurement Accuracy : 1/3000

AD-4403-FP Flameproof Weighing Indicator
Built-in sequences for normal batching and loss-in-weightweighing.
Complies with TIIS EXdIIBT5X specifications for explosion classand ignition temperature and is suitable for Zones 1 and 2●High-speed sampling (100 times/second) ensures sensitive response
to weight changes●Equipped with a main and secondary display
The main display shows the mass value and the secondary
display shows settings and total weight●Calculates net weight automatically after weighing●Equipped with a control I/O with 6 inputs and 9 outputs as standard●Equipped with 20 mA current loop output as standard(for connections to A&D peripherals)●Digital calibration function
40

Weighing Platform

Waterproof Weighing Platforms SB Series
Easy-to-use, complete platform equipped withstainless steel weighing pan and load cell
●Water and dust proof (IP-65) for sanitary,
safe use in the toughest industrial environment
Model No.
SB-15K10
SB-60K11
SB-100K12
SB-200K12

Capacity Dimensions (D✕W✕Hmm) Platform Weight
4.5kg
15kg
250✕250✕110
8kg
60kg
330✕424✕110
14kg
100kg
390✕530✕135
14kg
220kg
390✕530✕135

Cable length : 3m
Linearity & Hysteresis : 0.03% of R.O.
Pan material : Stainless steel

SB-SC Series Dustproof/Waterproof Platforms
IP68 compliant body resists water from heavy duty cleaning
Can be fully submerged in water for cleaning(Conditions : Water depth at most 1m continuously for 24 hours)
Model No.
Capacity Dimensions (D✕W✕Hmm) Platform Weight
SB30K17SC
30kg
300✕380✕117
9.3kg
SB60K17SC
60kg
300✕380✕117
9.3kg
SB150K17SC 150kg
300✕380✕117
9.3kg
SB60K18SC
60kg
390✕530✕117
13kg
SB150K18SC 150kg
390✕530✕117
13kg

Cable length : 1.5m
Linearity & Hysteresis : 0.03% of R.O.
Base unit and pan material : Stainless steel 304

Skeleton frame structure
Easy-to-clean design won’t accumulate stains or grime41Equipment

AD-4379 Stainless Steel Summing Box
Stainless steel,
dustproof/waterproof (IP67).
Up to 4 connections possible.
Equipped with load cellsensitivity adjustment trimmer.
For load cell cable diameters
between 4 to 12mm.

40

166
184
200

40

Unit = mm

67

ø8
2-

30
166

Net Weight =1.5kg

AD-4382 Summing Box
For connecting 2 load cells.
Dustproof, waterproof IP56.
Equipped with load cellsensitivity adjustment trimmer.

Max outer diameter ø8
For load cell (2)Max outer diameter ø8
For indicator

Cross
screw

Unit = mm

AD-4380SUS/AD-4380 Junction Box
AD-4380SUS

AD-4380

AD-4380SUS
Extends cables from load cell to indicator.
Stainless steel, dustproof/waterproof (IP67).
Does not contain load cell sensitivity adjustment trimmer.
AD-4380
Extends cables from load cell to indicator.
Dustproof, waterproof IP56.
Does not contain load cell sensitivity adjustment trimmer.

165 MAX30

AD-4380SUS

AD-4380

160
140
ø135

2-ø8

42

25

3

65 MAX

4-M6 Cross hole withhexagonal bolt

Unit = mm

Equipment

Anti-vibration Mounting Fixture LCZ Series
Model No.

LCZAP0405
LCZAP0408

Unit = mm

C
G

20kN

(Only in X direction)B

LCZAP0411

Max.transverse load

200L~1LCZAP0411

10

F

E

D

4-øJ
H

2~5
15

LCZAP0408

M16

6-øJ

D

CP/CP-FP Model

K

Model No.

LCZAP0405

E

Applicable range chart

A

B

C

D

E

F

G

H

(I)

J

K

LCZAP0405

154

170

300 11.5

77

200

100

15

124

8

25

250

LCZAP0408

213

200

315 12.5 105

185

130

23

167

12

40

265

LCZAP0411

303

230

330 17.5 125

170

160

30

243

14

55

I

X-Load Direction

L

H

Model No.

A

* Intended load cells for CP/CP-FP

280

L

Weighing Module Kit LCZAP1413/1414/1415
Unit = mm

4-ø
G

Load cell(sold separately)

Model No. LCZAP1413 LCZAP1414 LCZAP1415
LCM13K100 LCM13T002 LCM13T005
LCM13K200
Intended LCM13K300 LCM13T003
LCM13K500
load cells LCM13T001 LCM19T002

L

C

K
E

øG

4-

Max. horizontal force
20kN
12kN
6kN
Max. lifting force
40kN
30kN
20kN
Max. displacement 2mm(All directions) 2mm(All directions) 2mm(All directions)
Net Weight
6.0kg
12.0kg
3.4kg

H

Model No.

A

B

C

136

F

øG

14

15

20

23

50

120

84

18

175

14

18

24

26

60

84

150

100

18

190

14

25

24

29

75

114

100

LCZAP1414

116

212

LCZAP1415

121

227

E

H

J

K

H

L

15

168

D

I

70

97

D
C

* Net weight does not include load cell

LCZAP1413

B

F

LCM13T1.5
LCM19K500
LCM19T001
LCM19T1.5

70
I

J
A

Swivel Foot LCZAFT01S
ø26

Unit = mm
M12

28

10

ø20.6

M12 Hex nut

10

20

55 to 60
20 to 25

ø58

LCM13K500
LCM13T001

10

20

Intended load cells

LCM13K100
LCM13K200
LCM13K300
LCM13K500
LCM13T001
LCM13T1.5
LCM19K500
LCM19T001
LCM19T1.5

ø58

Net Weight : 0.25kg

43

Accessories

Mounting Fixtures
Rod-end bearings
Order No.
LCB-A6LCB-A12LCB-A18
LCB-18
LCB-24-2T
LCB-24-5T

Intended load cells
LC1205-K020/K050/K100, LC1216-K100, LC1122-K050/K100
LC1205-K200/K500, LC1216-K200/K500, LC1122-K250/K500
LC1205-T001A, LC1216-T001A
LC1205-T002
LC1216-T002A
LC1205-T005, LC1216-T005

Remarks
M6✕1
M12✕1.75
M18✕1.5
M18✕1.5
M24✕2
M24✕2

Mounting illustration

Unit = mm

Rod-end bearings A
Model No.
B
LCB-A6
LC1205-K020/K050/K100
116 134
LCB-A12
LC1205-K200/K500
152 182
LCB-A18
LC1205-T001A
194 236
LCB-18LC1205-T002
214 257
LCB-24-5T
LC1205-T005
268 338

C øDH7 E
6
718
30 12 1242 18 1743 18 16.3
70 25 25.3

F
9
16
23
20
35

Unit = mm

Rod-end bearings A
Model No.
B
LCB-A6
LC1216-K100
160 178
LCB-A12
LC1216-K200/K500
194 224
LCB-A18
LC1216-T001A
270 312
LCB-18(LC1216-T002)
285 328
LCB-24-2T
LC1216-T002A
320 376
LCB-24-5T
LC1216-T005
342 412Stainless-steel rod end bearing (for LCS15 load cell)
0

LCZARE12S
Applicable load cell: LCS15K500
Static radial load rating: 5.88kN
Static fracture load: 23.53kN
Mass: 85g

16−0.13

Unit = mm
12±0.3

)
.81
23

(69)

(Sø

33±0.354±0.5(ø17.6)

ø12 H730±0.5M12✕1.75

0

1.7
5)
ø3

72±0.544±0.7(ø21.9)

ø18 H716.5±0.3(93)

Applicable load cell: LCS15T001
Static radial load rating: 11.52kN
Static fracture load: 46.09kN
Mass: 258g

23−0.1342±0.5(S

LCZARE18S

M18✕1.5

57±0.744(ø29.6)

.86
(S
ø4
2

94±0.5(122))

Applicable load cell: LCS15T002
Static radial load rating: 17.4kN
Static fracture load: 69.62kN
Mass: 730g

031−0.1322±0.356±0.5

ø25 H7LCZARE24S

M24✕2.0

Approximately

Approximately

Approximately

Mounting illustration
LC-1216

Approximately

Mounting illustration
LC-1205*When using a stainless steel rod end bearing,
set the normal load to 1/2 of the rated capacity.
*One stainless steel nut (Type 1) is included as standard.

C øDH7 E
6
718
30 12 1242 18 1743 18 16.3
56 25 2270 25 25.3

F
9
16
23
20
31
35

Accessories

Compression fixtures
Components - Single convex
Mounting plate
Load button
LCMP-1
LCLB-1
LCMP-2
LCLB-2
LCMP-4
LCLB-4
LCMP-5
LCLB-5

Bearing plate
LCBP-1LCBP-1LCBP-2
LCBP-2

LC1205-K020/K050/K100
LC1205-K200/K500
LC1205-T001A/T002
LC1205-T005

Single convex

Components - Double convex
Button-hole plate Load button
Bolt
LCBHP-1
LCLB-1M6
LCBHP-1
LCLB-2M6
LCBHP-2
LCLB-4M10
LCBHP-3
LCLB-5
M10

Bolt
M6
M6
M10
M10

*A bolt is included to attach the load cellto the mounting plate.

Double convex
Mounting illustration

ø60

E

E

Load cell LC-1205

ø60

Button-hole plate (LCBHP)
Load button (LCLB)ALoad button protector (LCLB)

H

Bearing plate (LCBP)

Load cell LC-1205

H

Mounting illustration

A

Intended load cells

B
8-øF

B
C

A
115
115
164
167

C

B

C

Button-hole plate (LCBHP)

8-øF

Model No.
LC1205-K020/K050/K100
LC1205-K200/K500
LC1205-T001A/T002
LC1205-T005

D

D

Load button (LCLB)
Mounting plate (LCMP)

B
77
77
105
105

B
C

Unit = mm

C
100
100
130
130

D
16
16
22
22

E
16
16
19
19

F
8
8
12
12

H
72
72
115
118

A
120
120
178
190

Model No.
LC1205-K020/K050/K100
LC1205-K200/K500
LC1205-T001A/T002
LC1205-T005

B
77
77
105
105

Unit = mm

C
100
100
130
130

D
16
16
19
19

E
16
16
19
19

F
8
8
12
12

H
70
70
110
116

Compression fixtures
Components - Single convex
Mounting plate
Load button
LCMP-1
LCLB-1
LCMP-2
LCLB-2
LCMP-4
LCLB-4
LCMP-5
LCLB-5Single convex

Bolt
M6
M6
M10
M10

Components - Double convex
Button-hole plate Load button
Bolt
LCBHP-1
LCLB-1M6
LCBHP-1
LCLB-2M6
LCBHP-2
LCLB-4M10
LCBHP-3
LCLB-5
M10*A bolt is included to attach the load cellto the mounting plate.

Double convex
Mounting illustration

ø60

Load button (LCLB)JLoad cell LC-1216

øGO ring(Attachment of LCBHP)øG

DLoad button (LCLB)

Mounting plate (LCMP)

B

B

C

Button-hole plate (LCBHP)

8-øF

8-øF

B
C

Model No.
LC1216-K100
LC1216-K200/K500
LC1216-T001A/(T002)
LC1216-T002A/T005

H

A

J

H

Load cell LC-1216

E

Load button protector (LCLB)

ø60

Button-hole plate (LCBHP)

E

Bearing plate (LCBP)AMounting illustration

D

LC1216-K100
LC1216-K200/K500
LC1216-T001A/(T002)
LC1216-T002A/T005

Bearing plate
LCBP-1LCBP-1LCBP-2
LCBP-2C

Intended load cells

A
B C
155 77 100
155 77 100
214 105 130
247 105 130

BC

Unit = mm

D
16
16
22
22

E
16
16
19
19

øF øG HJ
8 77 114 62
8 77 114 62
12 102 165 80
12 102 198 80

Model No.
LC1216-K100
LC1216-K200/K500
LC1216-T001A/(T002)
LC1216-T002A/T005

A
B C
160 77 100
160 77 100
228 105 130
270 105 130

Unit = mm

D
16
16
19
19

E
16
16
19
19

øF øG HJ
8 77 110 62
8 77 110 62
12 102 160 80
12 102 196 80

45

Accessories

Mounting Fixtures
For TPFor TM

E

D
C

øEH7

Universal shackle

A±1.0

Min B
(C)

øA

øF

D hexagonal nut

øF1.6 split pin
øHH7

G
F

øB

Unit = mm

Intended load cells Model No. øA B C D hexagonal nut E
TP-20L ~ 100L S-1 89 183 209 M10✕1.25 26
TP-200L ~ 1 S-2 89 195 229 M12✕1.25 34
TP-2 ~ 5 S-3A 89 280 342 M24✕2 62
S-4 125 461 561 M39✕2 100
TP-10
S-5 165 622 742 M50✕2 120
TP-20

F G øHH7 Net Weight (kg)0.2
14 10.5 10
0.3
16 13.3 12
1.4
28 19.3 22943 39 40
17
53 49 50

Note: The Min B dimension is the attachment minimum dimension when
the shackle is screwed in completely. The attachment is attached in
the range of B to B+0.2 mm. (The shackle screw is rotated backabout half a turn to line up with the shackle position.)

46

Model No. øAIntended load cells
TM-50L-A
SM-1 138
TM-100L-A
TM-200L-A
TM-500L ~ 1 SM-2A 170
SM-2 170
TM-2

Unit = mm

øB

C

D

30

12+0.1

16–0.10

øEH7 øF Net Weight (kg)
12

50

0.2

15
21
16
38
0
39 15.25+0.13 19–0.13 16

62
62

0.4
0.4

Note: When the TM series is used, we recommend use of a universal
shackle (for tension) to prevent bending load on the load cell.

Bearing plate (BP), mounting plate (MP), expansion assembly and mounting plate (EA)
Use the bearing plate and mounting plate
when attaching the load cell.

If you anticipate bending load on the load cell from heat expansion,
use the EA.

Mounting illustration - Bearing plate, Mounting plate
C

C

4-øF
Bearing plate (BP)

D

4-øF
Mounting plate (MP)

A

(K)
A

(K)

G

E

D

Mounting illustration - Expansion assembly (EA), Mounting plate

E

M

øL

50

D
C

20 100

10

G

Mounting plate (MP)4-øFExpansion assembly (EA)
4-øF
10

Weld

Unit = mm50

20 100

D
C

Weld

Unit = mm

Note: The BP-100, MP-100, and EA-100 and higher have handles attachedto their plates.
Intendedload cells
CP-20L ~ 100L
CP-200L ~ 1
CP-2 ~ 5
CP-10
CP-20
CP-30
CP-50
CP-100C2Z1-500L ~ 2
C2F1-500L ~ 2
C2Z1-5, 10
C2F1-5, 10
C2Z1-20
C2F1-20
C2X1-30, 50
C2X1-100
C2X1-200
C2X1-300
C2X1-500
CMX-50L ~ 2
CMX-5 ~ 20

Expansion Assembly (EA) and Mounting Plate Dimensions
Bearing Plate and Mounting Plate Dimensions
A  C  D E øF G K Net Weight (kg) Model No.
A  C  D E øF G øL K M ±N Net Weight (kg)Model No.
BP-1, MP-1 152 100 77 15 8 15 122
4EA-1, MP-1 177 100 77 15 8 15 60 122 25 ±83
BP-1, MP-1 154 100 77 15 8 15 124
4EA-1, MP-1 179 100 77 15 8 15 60 124 25 ±83
BP-2, MP-2 213 130 105 23 12 23 167
8
EA-2, MP-2 248 130 105 23 12 23 90 167 35 ±8
6BP-10, MP-10 303 160 125 30 14 30 243
EA-10, MP-10 348 160 125 30 14 30 110 243 45 ±1016
12BP-20, MP-20 378 200 150 30 14 30 318
EA-20, MP-20 443 200 150 30 14 30 130 318 65 ±1026
18BP-30, MP-30 480 240 180 40 19 40 400
EA-30, MP-30 545 240 180 40 19 40 160 400 65 ±15
4636BP-100, MP-110 620 260 200 60 23 60 500
EA-100, MP-110 710 260 200 60 23 60 210 500 90 ±15
9665BP-210, MP-210 800 360 280 60 27 60 680
EA-200, MP-210 905 360 280 60 27 60 300 680 105 ±15180130131 130 105 23 12 23 8586
166 130 105 23 12 23 90 85 35 ±8
BP-2, MP-2
EA-2, MP-2
151 130 105 23 12 23 10586
186 130 105 23 12 23 90 105 35 ±8
170 160 125 30 14 30 110
16
12
215 160 125 30 14 30 110 110 45 ±10
BP-10, MP-10
EA-10, MP-10
160 160 125 30 14 30 100
16
12
205 160 125 30 14 30 110 100 45 ±10
205 200 150 30 14 30 1452618
270 200 150 30 14 30 130 145 65 ±10
BP-20, MP-20
EA-20, MP-20
180 200 150 30 14 30 1202618
245 200 150 30 14 30 130 120 65 ±10
BP-50, MP-50 240 220 160 40 19 40 160
EA-50, MP-50 305 220 160 40 19 40 160 160 65 ±154030BP-100, MP-100 310 260 200 60 23 60 190
EA-100, MP-100 400 260 200 60 23 60 210 190 90 ±158664360 (EA) 280 (EA)
BP-200, MP-200 390 280 200 80 27 80 230
EA-200, MP-200 475
80 27 60 300 230 105 ±15170100
280 (MP) 200 (MP)400 (EA) 320 (EA)
BP-300, MP-300 500 280 200 80 35 100 320
EA-300, MP-300 585
80 35 60 350 320 125 ±15222110280 (MP) 200 (MP)500 (EA) 400 (EA)
BP-500, MP-500 630 380 280 100 40 130 400
EA-500, MP-500 745
100 40 80 430 400 165 ±15457260
380 (MP) 280 (MP)EA-2, MP-2-A 98 130 105 15 12 23 90 25 35 ±8
BP-2-A, MP-2-A 55 130 105 15 12 15 25
7
4
EA-10, MP-10-A 155 160 125 30 14 30 110 50 45 ±10
BP-10, MP-10-A 110 160 125 30 14 30 50
1612
N : Permitted movement range

47

Appearance and/or specifications subject to change for improvement without notice.

✽ Load Cells-ADCC-03-ZW3-16103