KR2000/3000 SERIESGRAPHIC RECORDER

with measured
data protection

KR2000/3000 series are paperless recordersthat prevent falsification of data to meet therequirements of FDA 21CFR Part11 formedicinal chemical manufacturing. Employs
high visibility display and high operating
function. Also it realize data recording andmanagement by easy operation.
*FDA 21CFR Part 11: The U.S. Food and Drug Administration rule on electronic records and electronic signatures. It isa requirement when replacing the paper-based records toelectronic media and enacted in 1997.
KR2000KR3000144 x 144 mm Size
5.6" TFT color LCD Display

288 x 288 mm Size
12.1" TFT color LCD Display■ FEATURES

Easy operation
Touch screen (KR3000)
High speed sampling 100ms

Export data to USB flash drive
LAN network capability
Various functions such as calculation■ MODELS●KR2000●KR3000

KR2 P □□M□□□K R 3 P □ □- □ □ □Measuring points/sampling rate*
20: 12 points/100ms
40: 24 points/100ms
60: 36 points/100ms
80: 48 points/100ms
21: 12 points/1s
41: 24 points/1s
61: 36 points/1s
81: 48 points/1s
Communications interface (option)
N : None
R : High-order (RS232C)
S : High-order (RS422A/RS485)
Digital input/ alarm output (option)
0: None
1: Alarm output 12 points (a contact)
2: Alarm output 6 points (c contact)
3: Alarm output 24 points (a contact)
4: Alarm output 12 points (c contact)
5: Alarm output 12 points (a contact)+ 6 points (c contact)
A: Digital input 8 points
B: Digital input 8 points+ alarm output 12 points (a contact)
C: Digital input 8 points+ alarm output 6 points (c contact)
D: Digital input 8 points+ alarm output 24 points (a contact)
E: Digital input 8 points+ alarm output 12 points (c contact)
F: Digital input 8 points+ alarm output 12 points (a contact)+ alarm output 6 points (c contact)
Carrying handle & feet (option)
A : None
T : With carrying handle & feet

Measuring points/sampling rate*
60 : 6 points/100ms
20 : 12 points/100ms
61 : 6 points/1s
21 : 12 points/1s
Communications interface (option)
N : None
R : High-order (RS232C/RS485)
Q : High-order (RS232C/RS485)+ Low-order (RS485)
Digital input/ alarm output (option)
0 : None
1 : Mechanical relay output - 12 points(a contact)
2 : Mechanical relay output - 6 points(c contact)
7 : Digital input - 8 points+ MOS relay output 8 points
Carrying handle & feet (option)
A : None
T : With carrying handle & feet
* 1 to 4 channels input (4 points) when setting faster than
500ms sampling rate with model of 1sec sampling rate.

1PSE-365AKR2000/3000 SERIES with measured data protection■ PREVENTING FALSIFICATION ■ AUDIT TRAILOF DATA
Automatically record the login record, changeStore the file in binary format and restrainfalisification. Only able to opened by
ZAILA-P exclusive application softwareand display a message if falsified.

●Binary file example* Example when opened by word pad.
Unable to check the contents.

Display audit operation.
setting and other various operation record.
Display audit trail by recorder itself.
Audit trail can be also be referred by
ZAILA-P exclusive application software.

●Audit trail screen■ Electronic signature
Replay and confirm the record file by recorderand able to electronic signature to that file.
Confirm the signature infomration by fileinformation of recorder or ZAILA-P exclusive
application software.

●File information display■ LOGIN FUNCTION
Register up to 5 administrators and 100 general users and only registered users can access.
Set 10 kinds of access authority and signature level to general users.
ID, Full name, authority, password is settable per each user.

●Login user registration●Authority & signature level●Login screen

Smooth Operation by touch screen!
Input / Computation setting
Easy setting and display

KR3000 SERIES

Login operation
Easy-to-input the letter and value by touching.

2Data replay, CSV conversion
ZAILA-P Exclusive application software (standard attached)
By using exclusive application software, each file can data replay, confirm audit trail,signature, print, convert to PDF and convert to CSV file.
Each file recorded in KR2000 & KR 3000 can be taken out by USB flash drive or
fowarding to networkl server.

●ZAILA-P

KR3000KR2000USB flash drive■ INPUT SPECIFICATIONS
Measuring points: KR2000 --- 6 points, 12 points
KR3000 --- 12 points, 24 points, 36 points, 48 points
Input types:
Universal (refer to the table of measuring range)
Accuracy ratings: 0.1% digit (exceptions) * Measurement range conversion
accuracy
Reference junction compensation accuracy:
K, E, J, T, N, Platinel II --- 0.5 or less
Sampling rate:
100ms --- Approximately 100ms for all points
1 ms --- Approximately 300ms for all points*Burnout:
Disconnection of input signal is detected on thermocouple
and resistance input.
* When sampling rate is set below 0.5s at KR2P61/ KR2P21, then input willautomatically becomes 4 points and sampling rate will be 100ms.

■ COMMUNICATION SPECIFICATIONS●NetworkCommunication type:
Ethernet (10BASE-T/100BASE-TX)
FTP client:Transfer a data file to a network server
SNTP client:
The time can be synchronized to the time of SNTP server
E-Mail:E-Mail notification at specified time for alarm activation
Report data at specified time is selectable from all
registered data
Notification address --- Maximum 8 contacts■ ALARM SPECIFICATIONS
Setups:Alarm types:■ RECORDING SPECIFICATIONSInternal memory: 512MB
Exterior memory: Store the data file to USB flash drive
Recording cycle: 100, 200, 500ms, 1, 2, 3, 5, 10, 15, 20, 30s
1, 2, 3, 5, 10, 15, 20, 30, 60minLogging data:
Measured data --- Time of day, month and year of recording
start, tag, measured data, alarm status/types, maker text,etc
Setting parameter --- All setting parameter
Computation result data
Store types:
Binary type
Storing methods: Manual start / stop
Schedule (designation for time of day and date)
Trigger signal (alarm event, digital input)
Data logging of before and after trigger points* Pre-trigger is selectable
Measuring numbers of pre-trigger --- Max 950 data

Delay function:
Alarm settings:■ GENERAL SPECIFICATIONS
Rated power voltage:
100 to 240V AC (universal power supply) 50/60Hz
Maximum power consumption:
KR2000 --- 50VA
KR3000 --- 65VA
Normal operating condition:
Ambient temperature & humidity --- 0 to 50ºC, 20 to 80%RH
Power voltage --- 90 to 264V AC
Power frequency --- 50/60Hz2%
Attitude --- left/right/forward tilting 0ºC, backward tilting 0 to
20%
Weight:
KR2000 --- About 2.2kg (max)KR3000 --- About 7.2kg (max)Mounting:
Panel mounting■ COMPUTATION SPECIFICATIONS
Computation points:
KR2000 --- Maximum 44 points
KR3000 --- Maximum 128 points
Computation types:
Arithmetic operation, comparison operations, logicaloperations, integration operations, channel data operations,dew point, relative humidity, wind direction, 16 direction
display, increment per time, remaining amount of internal
memory, abnormality judgment, user lockout judgment■ DISPLAY SPECIFICATIONS
Display types:

Measured data display(Trend screen, Data screen, Bar-graph screen)
Historical trend display(Simultaneous display with Real-time trend is available)
Information display(alarm display, marker list, file list, audit trail)
Setting screen
Display points:
KR2000 --- Max 44 points
KR3000 --- Max 56 points*The LCD display may contain some pixels that always or never illuminate, andthe brightness of some areas of the display may appear uneven. There are
typical LCD performance characteristics and do not constitute malfunctions.

Up to 4 alarms can be programmed per channel
Upper limit, lower limit, differential upper limit, differential
lower limit (deadband is selectable), abnormal data
Setup range of alarm delay --- 0 to 3600 seconds
AND/OR selectable■ STANDARDS
Protection:

KR2000 --- IEC529 IP65 compliance (front part)
KR3000 --- IEC529 IP54 compliance (front part)
CE:
KR2000 (approved)
KR3000 (approved)
EMC directive --- EN61326-1 Class A
EN61000-3-2
EN61000-3-3
Low voltage directive --- EN61010-1
II, pollution level II
II,Over voltage (installation) category IImeasuring category II*The indication equivalent to 1mV may vary under the test environment byEMC directives.

■ OPTION SPECIFICATIONS
Please see standard version's PS sheets.

* Other detail specfications are common to stanard versions.

3

■ DIMENSIONS■ MEASURING RANGES-200.0-200.0-200-200.0-200.0-200-200.0-200.0-200-200.0-200.00000tototototototototototototototo300.0℃
600.0℃
1370℃
200.0℃
350.0℃
900℃
250.0℃
500.0℃
1200℃
250.0℃
400.0℃
1200℃
1760℃
1300℃
1760℃

B

0to1820℃

N-200.0-200.0-200

tototo400.0℃
750.0℃
1300℃

W-WRe26

0to2315℃

EJ
T
R
S

T/C

WRe5-WRe26

0to18

63

24.5

204

5.000V
10.00V
20.00V
50.00V

0.1%1digit*-200 to 0℃：
0.2%1digit

14430

200.5

●KR300025

0.1%1digit*0 to 400℃：
0.2%1digit
0.1%1digit*0 to 400℃：Out of
accuracy ratings*400 to 800℃：
0.15%1digit
0.15%1digit*-200 to 0℃：
0.3%1digit

288

K

63

0.1%1digit

137
161

totototo●KR2000

0.15%1digit*0 to 100℃：
4%1digit*100 to 400℃：
0.5%1digit

2315℃

0.2%1digit

0.2%1digit*0 to 300℃：
1.5%1digit*300 to 800℃：
0.8%1digit278-5.000-10.00-20.00-50.00(with built-in
voltage divider)

Accuracy ratings

88.8

DC voltage

Messureing range
13.80mV-13.80to27.60mV-27.60to69.00mV-69.00to200.0mV-200.0to500.0mV-500.0
to2.000V-2.000 to

144

Input type

33 38.8288297.6

251

Unit: mm

RTD

PtRh40-PtRh20

0to1888℃

NiMo-Ni-50.0-50.0-50

tototo290.0℃
600.0℃
1310℃

0.2%1digit

CR-AuFe

0.0

to280.0K

0.2%1digit*0 to 20K：
0.5%1digit*20 to 50K：
0.3%1digit

Platinel2

0.0
0.00

tototoU-200.0-200.0-200.0

tototo350.0℃
650.0℃
1395℃
250.0℃
500.0℃
600.0℃

L-200.0-200.0-200tototo250.0℃
500.0℃
900℃

0.1%1digit*−200 to 0℃：
0.2%1digit

Pt100-140.0-200.0-200.0

tototo150.0℃
300.0℃
850.0℃

0.1%1digit*-140.0 to 150.0℃
700 to 850℃：
0.15%1digit-140.0-200.0-200.0

tototo150.0℃
300.0℃
649.0℃

0.1%1digit*-140.0 to 150.0℃：
0.15%1digit

Pt50-200.0to

649.0℃

0.1%1digit

Pt-Co

4.0

to374.0K

0.15%1digit*4 to 50K：
0.3%1digit

JPt100■ AVAILABLE OPTIONSNameValidation Document
Traceability Certificate
Installation Qualification (IQ) Certificate
Operational Qualification (OQ) Certificate

0.15%1digit■ SOFTWARE (ZAILA-P) ENVIRONMENT

0.15%1digit*-200 to 0℃：
0.3%1digit

The accuracy ratings are converted into the measuring range under reference
operatingcondition. Thermocouple input does not contain reference junction compensation
accuracy.
K,E,J,T,R,S,B,N : IEC584,JIS C1602-1995
W-WRe26,WRe5-WRe26,PtRh40-PtRh20,Platinel II
II,NiMo-Ni,Cr-AuFe : ASTM Vol14.03
U(Cu-CuNi),L(Fe-CuNi) : DIN43710

CPU

1GHz or faster

OSWindows XP/Vista/7*Internet Explorer 6.0 or later

Memory256MB or more
(512MB or more recommended)

Disk drive

CD-ROM drive: 1 drive or more
Hard disk drive: Disk space of 1 drive or more for100MB or more

Language

Japanese, English

Specifications subject to change without notice. Printed in Japan (I) 2017. 3

32-8 KUMANO-CHO,ITABASHI-KU,TOKYO 173-8632
Telephone : +81-3-3956-2171
Facsimile : +81-3-3956-0915
E-mail : inter@chino.co.jp
Website : www.chino.co.jp/4