Save resources, Save nature

THERMO JETTERULG-9000series

FEATURES●No need of LN2. Just supply electric power and dry air to realize very low temperature!
●Even a small rate of 100L/min of air can realize -120°C in approximately 60 minutes.

(250L/min max.)●Not like using LN2, it allows all-night operation without attendance of personnel for watching.
●It enables stable temperature control as it doesn’t make condensation in the piping.
●It realizes remarkable reduction of running cost with a small initial cost.
●Adopting our unique single compression multi-cascaded condensation refrigerating system.
●Use of a special dryer for controlling dew point temperature.
●ULG-9300 reaches 200°C by micro-heater control.
●The vibration free system is good to use in combination with precision equipment.
●Energy saving design of small, lightweight, quiet equipment needs no help of LN2.
●Safe, simple and maintenance free pump less design●Optional air supply chamber enables local temperature control (DUT).

APPLICATIONSULG-9100series

ULG-9300series

Airflow control (no temperature control)

Environmental Test Type●For connection with analyzers/testers●For analyzers/testers (cooling and heating)●Heat cycle tests of mounted device●Environmental test (big change of temp.)●Degradation process tests●Can interconnect with various equipment(option)●Frozen food manufacturer●Polymer manufacturer●Mold cooling●Air cooled grinding (semi-dry process)●Storage of bio material (blood, cell, etc.)●Can interconnect with various equipment(option)

DIMENSIONS & CONNECTION●Air jet outlet can be connected to a chamber section of a tester directly to get efficiency of coolingspecimens.
●Air supply required: Industrial dry air●Power supply required: 3-ph 200V 50/60Hz740510
ULG-1110

9200*Feel free to ask themanufacturer forconnection, test use.

Side view

Front view

PRINCIPLEoutMulti level cascade condensers

Low temp airinDry air

Heat exchanger
Compressor
Air cooled condenser●Originally blended refrigerants of different boiling points are condensed sequentially in cascadedcondensation process to achieve very low temperature.
●Not like a conventional two-stage refrigerating system, it has only one compressor, saving energy andspace, making light weight equipment. The simple structure reduces cost of maintenance.

PERFORMANCE●Arrival time example to setting temperature
at 200ℓ/min. of air flow.
(Test example of 9ℓ-SUS-chamber )●Reachable to low-temperature range bycontrolling air flow.
(Test example of 9ℓ-SUS-chamber )APPLICATION EXAMPLE<Option>●Reachable to low-temperature range by
chamber content.
(It depends on content or material)●Heat cycle test example repeating cooling/heating.
(Required related instrument and software
separately)SPECIFICATIONSULG-9100series

MODEL
Weight & Dimensions

ULG-9300series

Approx. 200kg・510(W)×740(D)×1,110(H)mm

Power Supply

3φ 200V 50/60Hz

Rated Current

9.8A

12.3A

Circuit Breaker

30ACompressor

2.2kw

Heatern/a

Cooling Performance

Approx.500W(200ℓ/min. at -95℃ output

Air Flow

1kw
30～250ℓ/min (Variable)

Startup Time

Approx. 60min (Outlet temperature -120ºC，at Air Flow 100ℓ/min)

Lowest Temperature-120 ºC (at Air Flow 100ℓ/min

Condensation Method

Air cooledPort

IN: Rc3/8” female

Temperature Control

n/a

Utility*depends on environment)

OUT: 3/8” Stainless Steel-120 ºC ~ +200ºC Arbitrary setting available

Compressed Dry Air (0.4～0.6Mpa

more than 150ℓ/min air supply required for ≧100ℓ/min)

OPTION●Dew indicator (for monitoring heatless dryer)●Air Supply Unit (Heat resistant)●Chamber Unit (Consult us for Design)●Heat cycle test (depend on specifications)●For maintenance1)Defrost a week and change dried wood a year
. We recommend that heatless dryer be defrosted a week and dried wood be changed a year.
2) Strict adherence to operation procedure when stopping
When stopping low-temperature output, first stop the cooling machine and confirm the temperature of output gas is over 0℃,
then stop the compressed air, finally stop the main SW of the Thermo Jetter to prevent from malfunction of freeze and dew
drop in the pipe.
3)Regular maintenance service
Feel free to contact us about regular maintenance service for remuneration.
.

UESHIMA SEISAKUSHO CO., LTD.
■For inquiries, request for quotations on our products:<Manufacturer>

UESHIMA SEISAKUSHO CO., LTD.
6-5-22, Yaho, Kunitachi-shi, Tokyo 186-0011, Japan
TEL: +81 42 572 1397
FAX: +81 42 573 1520
Email: sales@ueshima-seisakusho.com

https://www.ueshima-seisakusho.co.jp
Subject to alteration without notice027-466540-7