EA01282D/06/A2/01.20
71462038
2020-07-31

Products

Solutions

Installation Instructions
Flow conditioner
t-mass I 300, 500-digital

EN Installation Instructions: EA01282D →  3
DE Einbauanleitung: EA01282D →  11

Services

EA01282D

Flow conditioner
t-mass I 300, 500-digital
Table of contents
1

Overview of accessories . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

3Authorized installation personnel . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

5Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

6Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

7Flow conditioner . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

8Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

Endress+Hauser

3

EA01282D

1

Overview of accessories

The Installation Instructions apply to the following accessories:
Order number

Device component

DK6004-

1 × flow conditioner, 35-hole• The order number of the accessory (on the product label on the package) can differ
from the production number (on the label directly on the accessory)!
• You can find the order number of the relevant accessory by entering the production
number of the accessory in the spare parts search tool.
• We recommend that you keep the Installation Instructions and packaging together at
all times.

2

Designated use

The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type. Only original parts from Endress+Hauser may be used. Only
spare parts kits designed by Endress+Hauser for the measuring device can be used at any
time.
Inspection is performed using the W@M Device Viewer. The procedure involved is describedas follows.
A spare parts overview is located in the interior of some measuring devices. If the spareparts kit is specified on this list, no inspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for productdata and click Spare Parts.

3.

All spare parts for the measuring device aredisplayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order Thenumber of the spare parts kit corresponds to anorder number on the spare parts list.

4

Endress+Hauser

EA01282D

3

Authorized installation personnel

Authorization to carry out installation depends on the measuring device's approval type. The
table below shows the authorized group of people in each case.
Whoever carries out the installation has full responsibility to ensure that work is carriedout safely and to the required quality standard. He/she must also guarantee the safety ofthe device following installation.

1)

4

Measuring device approval

Personnel authorized to carry out installation 1)

No approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Safety instructions• Check whether the spare part matches the labeling on the measuring device as described on
the cover page.
• The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type.
Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.
• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:• Specialized technical staff must be trained in instrument safety.
• They must be familiar with the individual operating conditions of the devices.
• In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is de-energized.
• For measuring devices intended for use in hazardous locations, please observe theguidelines in the Ex documentation (XA).
• In the case of measuring devices in safety-related applications in accordance withIEC 61508 or IEC 61511: Commission in accordance with Operating Instructions after
repair. Document the repair procedure.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Hot surfaces! Risk of injury! Before commencing work: allow the system and measuring
device to cool down to a touchable temperature.
Endress+Hauser

5

EA01282D• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removing the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn the measuring device off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open the housing for a brief period. Avoid the penetration of foreign bodies, moisture
or contaminants.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.
• If spacing is reduced or the dielectric strength of the measuring device cannot be
guaranteed during repair work, perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service plug:• Do not connect in explosive atmospheres.
• Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your Endress+Hauser service organization.

5

Symbols used

5.1

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

6

Series of steps

Endress+Hauser

EA01282D

6

Tools list

7

Flow conditioner

Use flow conditioners if it is not possible to observe the required inlet runs. Flow conditioners
improve the flow profile and therefore reduce the necessary inlet runs.
Mount the flow conditioner in flow direction in front of the measuring device.
Available in the following nominal diameters and flange standards:• ASME B16.5 Cl. 150/l. 300• EN 1092-1 PN10/PN16/PN25/PN40• JIS B2220 10K/20K• DN 80 (3")• DN 100 (4")• DN 150 (6")• DN 200 (8")• DN 250 (10")• DN 300 (12")

2
1
2

1

2

Flow conditioner
Seal

Endress+Hauser

7

EA01282D

5...10 x DN

≥8 x DN

5 x DN

1
1
1

Recommended inlet and outlet runs when using a flow conditioner
Flow conditioner

In the case of bidirectional sensors, also observe the inlet run in the opposite direction.

%

5
4
A

3

1
2

2
3
1
4
0
0

2

6

4

8

10

DN

B
2
A
B
1
2
3
4
8

The additional measured error to be expected depends on the type of interference and inlet run
Additional measured error (%)
Inlet runs (DN)
2 × 90 ° elbow, 3-dimensional
Expansion
2 × 90° elbow
Reduction or 90° elbow
Endress+Hauser

EA01282D

8

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize the disposalof WEEE as unsorted municipal waste. Do not dispose of products bearing this marking as
unsorted municipal waste. Instead, return them to Endress+Hauser for disposal under theapplicable conditions.

Endress+Hauser

9EA01282D

Strömungsgleichrichtert-mass I 300, 500-digital
Inhaltsverzeichnis
1Übersicht Zubehörteile . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12

2

Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123

Einbauberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13

5

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

6

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

7

Strömungsgleichrichter . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

8

Entsorgung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17

Endress+Hauser

11

EA01282D

1Übersicht Zubehörteile

Die Einbauanleitung ist für folgende Zubehörteile gültig:
Bestellnummer

Gerätekomponente

DK6004-

1 × Strömungsgleichrichter 35-Loch• Die Bestellnummer des Zubehörteils (auf dem Produktaufkleber der Verpackung) kannsich von der Produktionsnummer (auf dem Aufkleber direkt auf dem Zubehörteil)unterscheiden!
• Durch Eingabe der Produktionsnummer des Zubehörteils im Ersatzteilfindetool kanndie Bestellnummer des entsprechenden Zubehörteils ermittelt werden.
• Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

2

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

12

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

Endress+Hauser

EA01282D

3

Einbauberechtigte Personen

Die Berechtigung zur Durchführung eines Einbaus ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die einen Einbau vornimmt, übernimmt die Verantwortung für die Sicherheit
während der Arbeiten, die Qualität der Ausführung und die Sicherheit des Geräts nachdem Einbau.

1)

4

Zulassung des Messgeräts

Einbauberechtigter Personenkreis 1)

Ohne Zulassung

1, 2, 3Mit Zulassung (z.B. IECEx)

1, 2, 3Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:• In Gerätesicherheit ausgebildet.
• Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
• Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Messgeräten für den explosionsgefährdeten Bereich: Hinweise in der ExDokumentation (XA) beachten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
Endress+Hauser

13

EA01282D• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:• Nicht in explosionsfähiger Atmosphäre anschließen.
• Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

5

Verwendete Symbole

5.1

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

14

Handlungsschritte

Endress+Hauser

EA01282D

6

Werkzeugliste

7

Strömungsgleichrichter

Wenn die erforderlichen Einlaufstrecken nicht eingehalten werden können,
Strömungsgleichrichter verwenden. Strömungsgleichrichter verbessern das Strömungsprofil
und verkürzen dadurch die erforderlichen Einlaufstrecken.
Den Strömungsgleichrichter in Durchflussrichtung vor dem Messgerät montieren.
Verfügbar in folgenden Flanschnormen:• ASME B16.5 Cl. 150/Cl. 300• EN 1092-1 PN10/PN16/PN25/PN40• JIS B2220 10K/20K
Verfügbar in folgenden Nennweiten:• DN 80 (3")• DN 100 (4")• DN 150 (6")• DN 200 (8")• DN 250 (10")• DN 300 (12")

2
1
2

1

2

Strömungsgleichrichter
Dichtung

Endress+Hauser

15

EA01282D

5...10 x DN

≥8 x DN

5 x DN1
3
1

Empfohlene Einlauf- und Auslaufstrecken bei Verwendung eines Strömungsgleichrichters
Strömungsgleichrichter

Beim bidirektionalen Sensor Einlaufstrecke auch in Gegenrichtung einhalten.

16

Endress+Hauser

EA01282D

%

5
4
A

3

1
2

2
3
1
4
0
0

2

6

4

8

10

DN

B
4

Zu erwartender zusätzlicher Messfehler mit Strömungsgleichrichter in Abhängigkeit von
Störungsart und Einlaufstrecke

A
B
1
2
3
4

Zusätzlicher Messfehler (%)
Einlaufstrecken (DN)
2 × 90°-Krümmer dreidimensional
Erweiterung
2 × 90°-Krümmer
Reduktion oder 90°-Krümmer

8

Entsorgung

Gemäß der Richtlinie 2012/19/EG über Elektro- und Elektronik-Altgeräte (WEEE) ist das
Produkt mit dem abgebildeten Symbol gekennzeichnet, um die Entsorgung von WEEE als
unsortierten Hausmüll zu minimieren. Gekennzeichnete Produkte nicht als unsortierter
Hausmüll entsorgen, sondern zu den gültigen Bedingungen an Endress+Hauser
zurückgeben.

Endress+Hauser

17

*71462038*
71462038

www.addresses.endress.com