1500

SERIES

N95 HEALTHCAREPARTICU LATE RESPIRATORS& SURGICAL MASKS
FI T TING INSTRUC TIONS

1

2

Hold respirator in hand with molded nose contour (narrowend) at finger tips, allowing headstraps to fall below hand.

3

Place respirator under chin with molded nose contour(narrow end) up. Nose cushion must not be creased insiderespirator. Raise top strap to top back of head. Pull shorterbottom strap over head, below ears, to around neck. Do notwear with only one strap because it may affect fit.

4Adjust respirator for comfortable fit.

Each time user enters contaminated area seal should be fit
checked. Cover front of respirator by cupping both hands.
INHALE SHARPLY. A negative pressure should be felt insiderespirator. If any leakage is detected at respirator edges,
adjust straps by pulling back along the sides and/or reposition respirator. Repeat until sealed properly, otherwise see
your supervisor. Entry into a contaminated area with an
improper fit may result in sickness or death.

! ALERT
Failure to follow all instructions and warnings on the proper use of this respirator may reduce
respirator effectiveness, wearer protection and may result in sickness and death.
For additional information consult your supervisor, read all information on box, and phone+1 (800) 421-0668 ext. 512/550.

MOLDEX-METRIC, INC.
10111 W. Jefferson Blvd., Culver City, CA 90232
Tel: +1 (800) 421-0668 or +1 (310) 837-6500 Fax: +1 (310) 837-9563
Email: sales@moldex.com www.moldex.com
Moldex Technical Service Department: +1 (800) 421-0668
or +1 (310) 837-6500 ext. 512/550
U.S. Patents #D285,374 and #4,850,347.
Moldex, Ideas that wear well, Softspun and Dura-Mesh are registered trademarks.
The mesh design is a trademark of Moldex-Metric, Inc. Made in U.S.A.
9700V-776 REV F 07/09

USER INSTRUCTION
Users must follow instructions each time respirator is worn. It's the wearer's responsibility to obtain aproper fit. To choose the correct model/size Moldex Respirator you must fit test and user seal check. If
you cannot obtain a proper fit do not enter the contaminated area. OSHA regulation 29 CFR 1910.134 (f)requires that the user must be fit tested before being assigned a respirator and annually thereafter. UseMoldex Qualitative Fit Test Kit. User must be clean-shaven. Any facial hair, such as beards or long sideburns, may prevent the respirator from fitting properly. Eyewear must not interfere with face to facepiece seal. Re-fit test when there are any changes to the wearers physical condition that may affect fit.

WARNING
This product contains no components made from natural rubber latex. However, individuals highly sensitive to natural rubber latex may have an allergic reaction. This product does not eliminate the risk ofcontracting any disease or infection. Change immediately if the respirator comes in contact with bloodor fluids, including body fluids. OSHA has not set a permissible exposure level for airborne biohazards.
This product has not been sold with warnings or use instructions for personnel involved in industrial orrelated situations. If you are considering such uses, first contact the Moldex Technical Services
Department at +1 (800) 421-0668 or +1 (310) 837-6500 ext. 512/550.

3200
SERIES

N95 HEALTHCAREPARTICU LATE RESPIRATORS& SURGICAL MASKS
FI T TING INSTRUC TIONS

1

2

Hold respirator in hand with molded nose contour (narrowend) at finger tips, allowing headstrap to fall below hand.

3

Place respirator under chin with molded nose contour(narrow end) up. Nose cushion must not be creased
inside respirator. Pull head halo to top of head so it
rests on the crown.

4

Adjust halo on crown of head to provide a comfortable
and secure fit.

Each time before entering a contaminated area, perform auser seal check. Cover front of respirator by cupping bothhands. INHALE SHARPLY. A negative pressure should be felt
inside respirator. If any leakage is detected at respirator
edges, reposition respirator and/or select another size.
Repeat until sealed properly, otherwise see your supervisor.
Entry into a contaminated area with an improper fit may
result in sickness or death.

! ALERT
Failure to follow all instructions and warnings on the proper use of this respirator may
reduce respirator effectiveness, wearer protection and may result in sickness and death.
For additional information consult your supervisor, read all information on box, and phone+1 (800) 421-0668 ext. 512/550.

MOLDEX-METRIC, INC.
10111 W. Jefferson Blvd., Culver City, CA 90232
Tel: +1 (800) 421-0668 or +1 (310) 837-6500 Fax: +1 (310) 837-9563
Email: sales@moldex.com www.moldex.com
Moldex Technical Service Department: +1 (800) 421-0668
or +1 (310) 837-6500 ext. 512/550
U.S. Patents #D285,374 and #4,850,347.
Moldex, Ideas that wear well, Softspun and Dura-Mesh are registered trademarks.
The mesh design is a trademark of Moldex-Metric, Inc. Made in U.S.A.
9700V-776 REV F 07/09

USER INSTRUCTION
Users must follow instructions each time respirator is worn. It's the wearer's responsibility to obtain aproper fit. To choose the correct model/size Moldex Respirator you must fit test and user seal check. If
you cannot obtain a proper fit do not enter the contaminated area. OSHA regulation 29 CFR 1910.134 (f)requires that the user must be fit tested before being assigned a respirator and annually thereafter. UseMoldex Qualitative Fit Test Kit. User must be clean-shaven. Any facial hair, such as beards or long sideburns, may prevent the respirator from fitting properly. Eyewear must not interfere with face to facepiece seal. Re-fit test when there are any changes to the wearers physical condition that may affect fit.

WARNING
This product contains no components made from natural rubber latex. However, individuals highly sensitive to natural rubber latex may have an allergic reaction. This product does not eliminate the risk ofcontracting any disease or infection. Change immediately if the respirator comes in contact with bloodor fluids, including body fluids. OSHA has not set a permissible exposure level for airborne biohazards.
This product has not been sold with warnings or use instructions for personnel involved in industrial orrelated situations. If you are considering such uses, first contact the Moldex Technical Services
Department at +1 (800) 421-0668 or +1 (310) 837-6500 ext. 512/550.