TP SERIES

Thermal Imaging Sensor
New addition to TP series, a compact infrared thermal
imaging sensor, 6Hz analog output model !
TP-H series is an installation type, compact andhighly versatile infrared thermal imaging sensor
utilizing a 2000 pixel resolution infrared detector.
New addition of 6Hz, analog output type realizes
usage in various fields utilizing thermal data such as
intra-area hot spot detection, temperature variations
in various line and facilities and trend monitoring.
■ Model●Frame rate 6Hz specifications
Measurement range -20 to 300°C

TP-H02□□AN
Viewable Angle
60 : 60°
x 60°
25 : 25°x 25°
Interface
A : Ethernet with analog output●Standard specifications
Measurement range -20 to 300°C

Measurement range 100 to 800°C

TP-L02□□□N

TP-L0225EK
Viewable Angle
60 : 60°
x 60°
25 : 25°x 25°
Interface
E : Ethernet
U : USB

Viewable Angle
25 : 25°x 25°
Interface
E : Ethernet■ Standard Configuration. Compact infrared thermal imaging sensor
Frame rate

. Compact infrared thermal imaging sensor
Standardspecifications unit

6Hzunit
specifications .
Ferrite core
. Exclusive power cable(Ø3.7mm, O terminal, 2mm)
. Contact input/output, analog output
exclusive cable (2m)
. LAN cable (straight cable, CAT5e). Universal head
. Universal head fixing screws (3 pieces). Carl plugs for mounting the universal headfor concrete (3 pieces). Lens cap. Quick manual. Application software. Instruction manual (CD-ROM). Ferrite core
. Exclusive power/alarm output cable(Ø3.7 mm/O terminal, 2.5m). Exclusive communication cable(LAN or USB)

. Universal head
. Screws for universal head (3 pieces). Curl plugs for mounting the universal headfor concrete (3 pieces). Lens cap. Connector cap. Fixing screw (attached to bottom of thermal
image sensor). Quick manual. Application software. Instruction manual*Power supply (24V DC) is sold separately.

*Power supply (12V DC) is sold separately.

1PSE-708

TP SERIES■ Models
ModelsNEW

NEW

TP-H0260AN

TP-H0225AN

Specification

TP-L0260EN

6Hz (5Hz at alarm output)

Frame rate

TP-L0225EN

TP-L0225EK

TP-L0260UN

0.5Hz

3Hz (1Hz at alarm output)-20 to 300ºC

TP-L0225UN

100 to 800ºC-20 to 300ºC800ºCMeasurement
range
300ºC

300ºC
100ºC

0ºC
-20ºC-20ºC

Measurement spotsize and distance

60°
x 60°

25°
x 25°

60°
x 60°

25°
x 25°

Radius resolution

21.8mrad9.1mrad21.8mrad9.1mradInterface
4 to 20mA DC

Power consumption

Max 2.5VA (at 24V DC)

25°
x 25°

21.8mrad9.1mradUSB2.0-compliant
communication speed fixed at 115kbps

Ethernet (10BASE-T/100BASE-TX)

Analog output

60°
x 60°

Max 2.5VA (at 12V DC)

Max1VA (at 12V DC)■ Measurement spot size and distance● 25º x 25º specification
D (View width) = 0.5 x L (Measurement distance)● 60º x 60º specification
D (View width) = 1.4 x L (Measurement distance)
7.0m

2.5m

4.2m
1.4m
(About 30mm)

0.5m(About 10mm)
1m(About 90mm)(About 146mm)

1m

1.5m

(About 31mm)(About 52mm)
3m

3m

5m
5m

Distance 500mm
Example of measured object 180 x 180mm

Distance 500mm
Example of measured object 180 x 180mm■ Exclusive application software (Standard accessory)
Offers configuration of the compact thermal imaging
sensor, thermal image, temperature data storage,
trend graph display and image processing areavailable for the compact thermal image sensor.
Correspond to multiple languages
Japanese, English, Chinese (simplified), Korean, German
and ItalianConnectivity

Ethernet specification--- 4 sensors
USB specification--1 sensor2

■ Alarm detecting functione.g.) Using Alarm 1, output contact when temperaturerises 10ºC or more within 60sec.

About rate of change alarm. High limit. Low limit. Sensor abnormal. Rate of change alarm NEW. Output contact when set area

Reference temperature

Alarm setting contents

temperature exceeds setting
temperature within set time(max. 60 seconds).
. Effective when relative value
management (change from thenormal state) is needed.

(Only for TP-H series). Output pattern(Auto reset/Manual reset). Output logic selectable(A contact/ B contact)

T : Quantity of reference
temperature change(e.g. 10ºC)

T : Changed value time unit
(e.g. 60sec.)
Auto reset

Alarm
Alarm

Manual reset● Alert an alarm when T (Quantity of reference temperature change)> rate of change alarm setting value■ Specifications
Main unit specifications
Models

TP-L0225EK

Temperature
measurement range
Communicationinterface

100 to 800ºCTP-L0260EN,TP-L0225EN-20 to 300ºC
USB

6Hz (5Hz at alarm output)

Temperature
resolution

0.5Hz

3Hz (1Hz at alarm output)
0.5ºC (at 100ºC black body)±1% or ±3°C of measured
value, whichever is greater.

±2% or ±3°C of measured value, whichever is greater.

Repeatability
Detectingelement
Measurement
wavelength
Measurement
view angle
Radiusresolution

TP-L0260UN,TP-L0225UN

EthernetFrame rate

Accuracy ratings(Under ambient
temperature
25 ± 2°C)

TP-H0260AN,TP-H0225AN

0.3ºC
Thermopile array with 2000 pixels
Center wavelength 10µm
25º x 25ºSpecify from 60º x 60º or 25º x 25º

9.1mrad60º x 60º... 21.8mrad, 25º x 25º... 9.1mrad

Focus

Fixed focus

Emissivity
correction

0.10 to1.00 (0.01 increments)

InterfaceUSB2.0-compliant communication
speed fixed at 115kbps

Ethernet (10BASE-T/100BASE-TX)

Analog output

4 to 20mA DC

Number ofcontact output
Number ofcontact input

2 points (Non-voltage contact output)
1 point (For digital contact
output for reset)

Power supply

12 to 24V DC

24V DC

12 to 24V DCPowerconsumption
Working temperature
range
Working humidity
range

Max 2.5VA (at 12V DC)

Max 2.5VA (at 24V DC)

Max 1VA (at 12V DC)-10 to 50ºC
10 to 80%RH (no dew condensation)

Material

Polycarbonate Resin black

WeightAbout 150g (sensor main unit)

Dust and water
proof structure

IP 65 (when using exclusive cable and fixing screw)

Conforming
standard

CE (EN61326-1)

Function of the main unit

CE (EN61326-1) *Excluded when
LAN cable is connected.

IP 65 (when using exclusive cableand fixing screw)
CE (EN61326-1) * Excluded whenUSB cable is connected.

Set alarm conditions from personal computer and if the set value is exceeded, digital contact is output.

Monitor mode (Using exclusive application software)
Output temperature data continuously from command of the personal computer.

Capture mode (Using without besides exclusive application software)
Output temperature data per one row from command of High-order instrument (such as PC or PLC).
*Communication command is released for Ethernet specifications.

3

■ Application software specifications

Function(1) Display of thermal image(2) Settings of communication environment(3) Settings of thermal image sensor. Emissivity settings. Area designation (one area). Alarm settings (two contact) within the designated area
(e.g. high limit, low limit, self-diagnostic and rate of change*). Analog output settings**Only available for TP-H
(4) Image temperature data saving (CSV)(5) Thermal image saving (JPEG)(6) Maximum/minimum value indications(7) Max. 8 areas alarm settings
Alarm settings in max of eight areas(8) Temperature scale range switching(9) Image processing. Averaging. Mirror-reversed rotation. Upside-down rotation . Make the image out of temperature range
transparent** Only available for TP-H
(10) Trend graph (Manual, Auto save)(11) Language selection. Japanese. English. Chinese (Simplified). Korean. German. Italian

Hardware requirements
Windows XP (32bit)/Vista (32bit)/7 (32bit/64bit)*1: XP or later version is recommended,.NET.
Framework 2.0 or later version is required*2: USB specification is not supported by 64bit
OS*3: Multiple contacts connecting software is notsupported by 64bit OSOSMemory

2GB or more is recommended

CPU TP-L series

2GHz or more is recommended

CPU TP-H series

Connecting with one unit---3GHz or more is recommended
Connecting with two units or more---Dual core is recommended 3GHz or more

■ For high temperature environment
Water-cooling case

■ For oil mist and dusty enviroment
Air purge CaseStores the compact infrared thermal imaging sensor.
Water-cooling and air purge function are provided.

MODEL : TP-ZCC1
The air purage case isused to disperse dust
and fume for keeping
the light path.

■ For fire detection
Bandpass filter for fire detection.
Front

Put the filter above the lens to detect
existence of fire.
(While bandpass filter is used for firedetection,
TP series cannot be used astemperature sensor.)

Rear

For window, BaF2 is used. Assembles TP-L series tothe model TP-ZCC3 and performs adjustment.

■ Dimentions
Frame rate 6Hz
Viewable Angle 60°
x 60°

Viewable Angle 25°
x 25°

6862

79

62

1362

62

UNIVERSAL HEAD

Viewable Angle 60°
x 60°376280Standard
Viewable Angle 25°
x 25°
2248

62

22

φ2062
CONNECTOR

SCREW
1/4-20 UNC

10SCREW
1/4-20 UNC

1062

φ58

φ20

CONNECTORSpecifications subject to change without notice. Printed in Japan ( I ) 2012. 632-8 KUMANO-CHO,ITABASHI-KU,TOKYO 173-8632
Telephone : +81-3-3956-2171
Facsimile : +81-3-3956-0915
E-mail : inter@chino.co.jp
Website : http://www.chino.co.jp/"