PICOXEA feeatures:

High‐qquality picose
econd laser < 50 ps,
High‐Q
QE Single Phooton Detectoor,
Fast timming TDC measurementts 60ps,
Remotte control USSB and wirelless interfacee,Graphiical User Inteerface,
LabVIEEW and C++ DLL librariess

The PICOXEEA from AU
UREA Techno
ology is the first “all inn one” and easy‐to‐use Time Correlated Single
Photon Cou
unting TCSPCC ANALYSER..
Engineered with “high performancce, and easee‐of‐use” innnovative minndset, the PICOXEA
P
fullyy integrates
t world’s mostmadvancced Geiger‐mmode avalannche photodiode (Geigerr‐SPAD) andwithin the same box; thend pulsed lasser source and Time to Digital
DConveerter (TDC) technologies
ts. Moreover,high efficient picosecon
he most intu
uitive Graphical User Inteerfaces compatible with
it includes tthe fast dataa processingg and also thLabVIEW annd most proggramming languages.
This newly design stand‐alone PICOOXEA TCSPCC ANALYSER allows any very‐low‐levvel of light down
dto thequired for fastfevent fluorescencee, photolumminescence oro Photon Soource Time‐single photton level req
Resolved mmeasurementts.

Application
ns :Quantu
um Optics,Quantu
um Dots charracterization
nSpectrooscopy, Ramaan
spectrooscopy
Photo‐luminescence
Singlet Oxygen meaasurement
Eye‐saffe Laser Rangging (LIDAR)
Fiber opptics charactterization(OTDR)
meFluoresscence life tim
Thin film
m characteriization (OLEDD)

AUREA Tec
chnology SAS
S 18 rue Alain Savary 250
000 Besançonn www.aureate
echnology.com
mJaan. 2015 V1.0

SPECIFICATTIONSLASER EMISSSION
WavelengthhOptical pulsse width
Peak powerr
Repetition Rate
CW
Optical output
Laser Modee

405 too 1550 nm (ssee table 1)
35 to 80 ps (see taable 1)
Adjusttable to 250 mW max
120 MHzM max
Continnuous laser emissioneavaailable in opttion (typ 30 mW)
mO
Other
waveleength and pulse width available on requestrSinglee mode fiber , FC/PC Connector
Manu
ual or AutoManuual: the user choose the peakppower

PHOTON DEETECTION
Detector
Spectral rannge
Detection efficiency
eDark Count rate

Timing Jitteer
Dead Time range
After pulsin
ng Probabilityy

Detection outputosignall
Delay
TRIGGER SIIGNAL (Gatee)
External triggger
Internal triggger generattor
Effective gaate width
CORRELATIION
Resolution
Measuremeent range
Correlation Data rate

Auto: the peak power is auto
omatically adjusted
ato keepka goodd andshort pulse shape.
@ 15550 nm@7750 nm
InGaA
As APD in Geiger mode
SI APDA
900 too 1700 nm
4000 to 900nm
10 to 25 % (adjusttable)Upp to 75 %< 5000 cps @ 10%% Efficiency
2 cps gradeeC< 250< 10000 cps @ 10% Effficiency < 1001 cps gradeeB(chammpion “AC” optiono)< 505 cps gradee A< 200 ps @ 20% QE
Q< 2802 ps, (<50pps in option))
Adjusttable 500nss to 1 ms. (100ns 600 ns typincremments)< 0.1%% at 100 kHz @ 10 ns gategand < 0,5%0à 10MHHz @ 500 ns gate
g10% QQE< 5% aat 1 MHz @ 10 ns gate anda 10%
QE
TTL siggnal [20 ns width]w
TTTL or NIM
Variabble from 0 to 128ns [0.5ns Variable from 0 to 128ns [0.5nsincremments]
inccrements]
Variabble up to 20 MHz, TTL
Variable up to 20 MHz, TTLL
Variabble up to 20
2 MHz (geenerator Variable up to 20 MHz, TTLL
providded)
Adjusttable from 1 to 100 nss [0.5ns Adjjustable fro
om 1 to 100 ns
incremments][0.5ns incremeents]
60 ps (time bin)0 to 500
5 nsup to 0.5 million correlation/scsec

AUREA Tec
chnology SAS
S 18 rue Alain Savary 250
000 Besançonn www.aureate
echnology.com
mJaan. 2015 V1.0

Table 1 : Laser sp
pecificatioons by wa
avelength
Las‐xxxx

Wavelength (nm)Pullse width (ps))

Peak powwer (mW)

SSpectral(nm)LAS‐4440
LAS‐4470
LAS‐5510
LAS‐6635
LAS‐6665
LAS‐6690
LAS‐7770
LAS‐8830
LAS‐8850
LAS‐9905
LAS‐1106
LAS‐1131
LAS‐1155440< 60> 150
<5470
< 60> 150
<5510
< 80> 100
< 10635< 454> 200<7665< 454> 200<7690< 50> 200
<7770
< 50> 100
<7830
< 454> 100< 10850< 50> 100
< 10905< 50> 100
< 101060< 60> 100
< 151310< 35> 50
< 151550< 35> 50< 15
Other wavelengths, pulsee widths or poower availablle on request

width50 ps

M
Measure of picosecondplaser excita
ation @ 650n
nm for 80 MHzM
repetitioon rateAUREA Tec
chnology SAS
S 18 rue Alain Savary 250
000 Besançonn www.aureate
echnology.com
mJaan. 2015 V1.0

QE vs Waveleng
gth for VIS deetection (3 diffferent optionns)

QE vs Wavelength
Wf NIR detecttion
for

Scree
en-shots of the softwarre

AUREA Tec
chnology SAS
S 18 rue Alain Savary 250
000 Besançonn www.aureate
echnology.com
mJaan. 2015 V1.0Connnector :CTLL_USB
Opt IN
Opt OUT
Dettection OUTT
Triggger (Clock IN
I & OUT)Electrical, Mechhanical and Environmen
ntal
wer supply
Pow
Pow
wer consump
ption
Dimmension (LxWWxH)
We
eight
Opeerating temp
perature
Stoorage temperratureMini USSB 2.0 type B
FC/PC optical connectorquest)(other types on req
FC/PC optical connectorquest)(other types on req
SMA feemale type
SMA feemale type:

110 – 230
2 VAC
< 10 Watts @ 5 VDC
C330 x 280
2 x 80 mm334 kg+ 10°C to + 30°C‐ 40°C tot + 70°C

Quaality controll
A coomplete testt sheet and useru manual is provided forf each systtem.

Orddering Informmation
PIC
COXEA_LASXXXX_YYY_ZZ__00
YYY
Y: VIS for visible range orr NIR for Infrra‐red range
ZZ : BL for blue,, RD for red anda LJ for loww jitter
00 ffor FC/PC lasser output annd detector input
01 ffor FC/PC lasser output an
nd FC/APC foor detector input
10 FC/APC laserr output andd FC/PC for detector inpu
ut
11 FC/APC laserr output andd for detector input

Conntact Inform
mationPleaase contact support@ausureatechnolo
ogy.com

wwww.aureatech
hnology.com
mAUREA Tecchnology SAS
S 18 rue Alain Savary 250
000 Besançonn www.aureateechnology.com
mJaan. 2015 V1.0