LED

L-0995AZ

LED Operating microscope (with X-Y)

Wide Angle Viewing System

Monitor Option

SDI 4 / BIOM 5®

®

1720±501490±50700

www.inami.co.jp/en/680

L-0990A(with X-Y)

1620±501490±50700±50

L-09901390±501490±50700±50

LED

L-0955XD
Factory Option

5x-15x3Dview

Lithium Ion
Battery

With optional pole and base (K-1422)

1050
165

400

20060

With the optional pole and base(K-1422),
L-0955 series can be used as a stand type
operating microscope.

190

470

930

L-0955 series can be securely contained in ahigh quality lockable carrying case with TSA
lock system and viberation-resistant casters
that ensure safe and smooth transportation of
the microscope during travel.

660
127

540

LED

L-0955SD
5x-15x
Lithium Ion
Battery

3Dview
With the optional pole and base(K-1422),
L-0955 series can be used as a stand type
operating microscope.

L-0955 series can be securely contained in ahigh quality lockable carrying case with TSA
lock system and viberation-resistant casters
that ensure safe and smooth transportation of
the microscope during travel.

1140
165

400

200

150

280

470

930

660

540

HalogenL-0940SD

6x-18x3Dview
L-0955 series can be securely contained in ahigh quality lockable carrying case with TSA
lock system and viberation-resistant casters
that ensure safe and smooth transportation of
the microscope during travel.

With the optional pole and base(K-1422),
L-0955 series can be used as a stand typeoperating microscope.

1090
165

400

200

185

360

470

930

660

127

540

L-09701390±501540±50700±50L-0950D

1500
165

400

600

1155

1425

1260

3Dview

600

Specifications
L-0995AZ

L-0990A
MEGA-80

L-0990
MEGA-21

L-0970
MEGA-10

L-0955SD

L-0955XD

L-0940SD
DECA-21

L-0950D
EASIO smart

Type

Galilean type

Galilean type

Galilean type

Galilean type

Galilean type

Galilean type

Galilean type

Galilean type

Magnification Changer

Continuous
auto zoom

5 Stepsby drum rotation

5 Stepsby drum rotation

5 Stepsby drum rotation

Continuous zoomby drum rotation

Continuous zoomby drum rotation

Continuous zoomby drum rotation

Continuous zoomby drum rotation

12.5x

12.5 x

12.5x

12.5x

12.5x

12.5x

15x

15x

Total Magnification

4.4x - 26.4x

3.2x, 4.8x, 8x,
12.8x, 20x

3.2x, 4.8x, 8x,12.8x, 20x

2x, 3x, 5x,8.1x, 12.7x

5x - 15x

5x - 15x

6x - 18x

6x - 18x

Field of View

Φ48-8mm

Φ51-8mm

Φ51-8mm

Φ103-17 ㎜

Φ35-10mm

Φ35-10mm

Φ45mm

Φ50-14mm

Pupilary adjustment

54-70mm50-75mm50-75mm50-75mm

50 - 70 mm

50 - 70 mm

54 - 70 mm

50-75mm

F=200mm(Apocromatic)

F=175mm

F=175mmm

F=250mm

F=175mm

F=175mm

F=175mm

F=250mm-7D - + 7D

-6D - + 5D

-6D - + 5D

-6D - + 5D

-6D - + 6D

-6D - + 6D

-6D - + 6D-6D - + 6D

Cobalt, Red free,
Heat absorption,
1/2ND

Cobalt, Red free,
Heat absorption,
1/2ND

Cobalt, Red free,
Heat absorption,
1/2ND

Cobalt, Red free,
Heat absorption,
1/2ND

Retina protection,
Red-Free, Plain,
Bulb color

Retina protection,
Red-Free, Plain,
Bulb color

Cobalt, Red free,
Heat absorption,
1/2ND

Yellow, Red free,
Heat absorption,
1/2ND

LED

Halogen

Halogen

Halogen

LED

LED

Halogen

HalogenMaximum intencity

60,000Lux

110,000Lux

110,000Lux

110,000Lux

80,000Lux

80,000Lux

80,000Lux

80,000Lux

Co-axialIllumination Field

Φ45mm

Φ50mm

Φ50mm

Φ50mm

Φ45mm

Φ45mm

Φ45mm

Φ40mm

Mortorised

Mortorised

Mortorised

Mortorised

Manual slider

Mortorised

Manual slider

Manual slider

40mm

40mm

40mm

40mm

30mm

40mm

30mm

31mm

1.3mm/sec

1.3mm/sec

1.3mm/sec

1.3mm/sec-

1.3mm/sec--

0°- 30°0°- 30°0°- 30°0°- 30°0°- 90°-2.7-27.5°

0°- 90°0°- 90°

Head Swing Angle---

Left-Right : -30° - +30°---

Left-Right : -30°- +30°
Back-Forth : -30°- +30°

Objective Lens
Swing Angle-------

Left-Right : -30°- +30°
Back-Forth : -30°- +30°

Type

Floor Type

Floor Type

Floor Type

Floor Type-

Floor Type--Weight

4.0kg

1.3kg

1.3kg

1.3kg

-

1.3kg

--

Built-in(Swichable)

Detachable(Option)

Detachable(Option)---

Detachable(Option)

-

92kg

67kg

63kg

64kg

17kg

18kg

15kg

36kg

Microscope

Eyepiece

Objective Lens

Diopter Adjustment
Illumination
FilterLight Source

Focal Adjustment
Type
Adjustable Distance

Focusing Speed
Head Inclination
Head Inclination Angle

Foot Remorte

Red Reflex Unit
Type
WeightTotal WeightOptional Accessories
CodeDescription

L-541Beam Splitter Device

L-542

Observation Tube
with Image Rotation (Monocular)

L-542-CB

Coaxial Observation Tube(Binocular)

L-0544

Video Camera Adapter
with C-mount (f=100mm)

L-0568-231S2-N

CCD Camera (540TV Lines) (NTSC)

L-0568-231S2-P

CCD Camera (540TV Lines) (PAL)

L-0541DC-FR

Foot Remote Controller
for L-0541DC-1.3

L-0500

Red Reflex Unit forOperation Microscope"Red Reflex Booster"L-1422Stand for L-0940
Portable Microscope, H-Style Base

OBJ-002

Objective Lens (F=250mm)(Ent Surgery, Dental Surgery)

OBJ-003

Objective Lens (F=300mm)(Neurosurgery)

OBJ-A008

Objective Lens (F=175mm)(Ophthalmic Surgery)

OBJ-012

Objective Lens (F=350mm)(larygeal Surgery)

OBJ-014

Objective Lens (F=400mm)(larygeal Surgery)

OBJ-024-AP

Objective Lens (F=200mm)
Apochromat (Ophthalmic Surgery)

L-054845 degree adaptor

L-0995AZ

L-0990A
MEGA-80

L-0990
MEGA-21

L-0970
MEGA-10

L-0955SD

L-0955XD

L-0940SD
DECA-21

L-0950D
EASIO smart

EquipedEquipedEquipedEquipedEquipedEquipedEquipedEquipedv1.0