MIchrome
Revolutionary industrial and biological microscope camera
Two Core Technology: Real-time image stitching & depth-of-field fusion
USB3.0 interface, Support OEM/ODM development

ISO9001

www.tucsen.com

Revolutionary PC Computing Imaging Software Mosaic V2

Unique from the cumbersome process of traditional technology to obtain images after processing, therevolutionary computing imaging software Mosaic V2 provides real-time image stitching and real-time depthof field fusion. This can automatically complete the image while the operator moves the stage - productivity
at its best.

Real-time image stitching
After

Within a few seconds of moving the stage,
Mosaic V2 can complete the whole process ofpanoramic stitching in real time, and it can be
accurately and quickly stitched under different
magnifications and arbitrary angles.

Sample: Mouse ovary section

BeforeMagnification:10X

Real-time depth-of-field fusion
Rotating the focus ring to image different depth of field points, Mosaic V2 can realize the depth of field expansion and full-length details at a glance, no more blurred images!
BeforeSample: Circuit board pin
Magnification: 4.5X

AfterMinimalist Operation Mode, Work More Efficiently and Effortlessly

Fast, efficient and worry-free is the core design concept of Mosaic V2 software. It adopts a new image"shooting-processing-measurement-reporting" function, modular design, and integrates various intelligent
image processing algorithms from Tucsen. It is dedicated to providing users with more simple operation
modes and improve work efficiency in all circumstances.

Main function menu Image workspace

Submenu area

Image library

Mosaic V21

Modular function configuration
Function configuration box

Users can adjust all functions including
exposure, processing and measurementaccording to different applications, andcustomize the exclusive working interface!

Function module display area

2

Efficient measurement flow

3

Visual property editing

The measurement stream can be

During the measurement process,used to record repetitive measurement

the user can modify the properties

steps, making it easier for users toof lines, fonts, colors, etc. very

perform measurement tasks faster.

intuitively.

Statistical area

4

Create an experiment report

Support for project information input,
then automatically generate
experimental reports containing image,measurement and counting information.

Choose the Right MIchrome Camera for Your Application

High-speed global shutter for fluorescence imaging
MIchrome 5 Pro
The MIchrome 5 Pro has many outstanding performance capabilities
beyond those of CCD cameras. It not only has an obvious advantages influorescent applications, but its global shutter technology can help
users get better and faster operation experience when performing

300x240 pixels, RGB:281K

300x240 pixels, RGB:281K"real-time image stitching " for example.

7087

3849

Note: Global shutters are ideal for capturing dynamic
samples more accurately, avoiding the distortion of theGlobal shutter

Rolling shutter

moving object caused by non synchronized pixel exposure.

Count:2079
Mean:5746.380
StdDev:447.027
Bins:2563564

1266
Count:990
Mean:2401.105
StdDev:375.157
Bins:256

Min:3849
Max:7087
Mode:5476(31)Bin Width:12.648

Min:1266
Max:3564
Mode:2197(19)Bin Width:8.977

Camera: MIchrome 5 Pro

Camera: 694CCD

Exposure time: 200ms

Exposure time: 200ms

vsLarge area array camera for high resolution imaging
MIchrome 20The MIchrome 20's 1-inch sensor can achieve up to 20 megapixel
resolution, giving users a better sensory effect and better image quality.
Sample name: Pathological section of chronic leukemia

The economically popular 6 megapixel color microscope camera
MIchrome 6The 6 MP meets the needs of most microscopic imaging applications,and its economical price positioning combined with the many features
provided by Mosaic V2 gives users with a truly value-for-moneyexperience.
Sample: circuit board

Sample: pine stem cross section

Quality within reach·

Meticulous CNC arc processing·

Environmental protection spraying·Unique side decorative patterns·

Stainless steel C ring with no debris

Fluorescencemicroscope

Biologicalmicroscope

Plant stem section, 10X

Blood vessel, 20X

Polarizing
microscope

Metallographic
microscope

Stereoscopic
microscope

Rock section, 20X

PCB board,5X

Screw hole, 4.5X

Sample >>Applications

Camera specification

Software functions

Product Model

MIchrome 5 Pro

MIchrome 20MIchrome 6· Real-time EDF

Sensor Model

IMX264LQR-C

IMX183CQJ-J

IMX178LQJ-C· Real-time image stitching

Sensor Type

CMOS

CMOS

CMOS

· Real-time sharpening

Sensor Size

2/3"

1"

1/1.8"

Modular function configuration

Color/MonoColor

ColorColorIntelligent 12-bit ISP color reproduction

Pixel Size

3.45x3.45(μm)

2.4x2.4(μm)

2.4x2.4(μm)

Real-time fluorescence image synthesis and editing

Resolution

2448(H)x2048(V)5472(H)x3648(V)3072(H)x2048(V)HDR image synthesis

Frame

35fps ( 2448x2048 )88fps ( 1224x1024 )Micro-imaging-based intelligent automatic exposure

15fps ( 5472x3648 )53fps ( 2736x1824 )41fps ( 3072x2048 )Intelligent flat field correction based on dynamic calculation

67fps ( 1824x1216 )Smart measurement workflow

Shutter Mode

GlobalRolling

Rolling

Exposure Time0.13ms-15s

0.13ms-15s

0.13ms-12s

Auto Setting

Exposure Time, Color scale, White Balance

Manual Setting

Exposure Time, Gain, Noise, Gamma, FFC

Color Temperature

2000-15000K

PC Software

Mosaic V2

Picture Format

JPG/PNG/TIFF/DICOM

Operating System

Windows , Mac

PC Requirements

CPU: Intel Core i5 or better(Quad or more Core), RAM: 8G or more

Multiple Cameras

Supports 4 Cameras Simultaneously in SDK

Data Interface

USB3.0

Implements drawing: points, lines, rectangles,polygons, circles, arcs, angles

Optical Interface

Standard C Mount

Data export as TXT or Excel

Camera Size

68*68*47mm

Report generation and printing

Camera Weight327g

Implements multiple iterations of workflow execution
Supports single shot, delayed camera
Automatic video and delay video generation
Output format selection

Tucsen Photonics Co., Ltd.
Add: 5# Wanwushe Smart Industrial Park , No.2 Yangqi Branch Rd,
Gaishan Town, Cangshan Area, Fuzhou, Fujian, PRC, China.
Tel: +86-591-28055080
Web: www.tucsen.com
E-mail: support@tucsen.com

User parameter group save and load
Dynamic\static measurement
Layered measurement
Customize measuring gauges, layers, precision
Customize image naming, style, save location