E10 SERIESELECTRONIC PRESET COUNTERDIN 72 x 72
LARGE LED DISPLAY
6 DIGIT MODEL
1 LEVELHIGH COUNT SPEED
ADD / SUBTR COUNTERMODELS

SPECIFICATIONS
Display

NO. OF
DIGITS

MODEL
E10-166M
E10-166MR

6PRESETMEMORYLEVEL○1FRONTRESETPower Source

4.3 VA

Preset Level

1 LevelSetting Range

0 - 999999 ※Input Mode
Count Mode

Open Collector Input: 50 μs / Contact Input: 16.6 ms

Duty

1:1Output

Relay Type 1C, 250 VAC, 2A (125VA), 220 VDC 2A (60W) cosф = 1Reset Input
Reset Mode
Operation Mode

Operation ModeSelection SwitchRESET

1 2 3 4 5 6

0.1

1SOutput Time
Adjustment Knob

Memorylatch output
Contact (100 ms min.)Open Collector (sink current 10mA max.)
Remote Reset, Auto-Reset, Front Panel Reset (E10-166MR)
Auto-reset (display resets when preset value is reached)
Overrun (counting continues even after preset value is reached)
EEPROM Data Retention: approximately 20 years
Memory Frequency: 100,000 times max.

Operating Temperature -10°C to + 50°C (should not be frozen)
Hi-pot Test
Insulation Resistance

Digital Switch for Setting

Red LED (ON when output is actuated)
0.1 to 1 second per shot (adjustable via front panel knob) orPower Source for Sensor 12 VDC, 60mA max.
Operating Humidity

Front Panel Reset(simultaneous input of add and subtract is not possible)

Pulse Width

Output TimeABC

Contact Input: Relay, Microswitch, etc. (sink current 7mA max.)
90° Quadrature Input / Individual Add/Subtract Input
Open Collector Input: 10 kHz / Contact Input: 30 Hz

Output IndicatorCount Display

Open Collector Input L : 0 to 2 V (sink current 7mA max.)

Count Speed

Output Display
OUT100/110 VAC or 200/220 VAC, 50/60 Hz

Power Consumption○FRONT PANEL(E10-166MR)

6 Digits 7-Segment Red LED 10.16 mm (H) x 5.54 mm (W)Noise Immunity
Vibration Immunity
Shock Immunity
Weight35 to 85 % RH (non-condensing)1500 VAC (1 minute)
100 MΩ min. (500 VDC Megger)(on Power leads and between non-charged metal parts)
Square wave noise from Noise Simulator±2.0 kV (PowerTerminals), ±500 V(Input Terminals)
Malfunction: 10 to 55 Hz, double amplitude 0.5mm
Destruction: 16.7 Hz, double amplitude 4mm
Malfunction: 100 m/s 2 (10G) Destruction: 300 m/s2 (30G)
550g※ When set to 0, counter will count down from displayed value to 0, when output

signal occurs.

WIRINGOPERATION MODE SELECTIONABCSENSOR
External Reset
Input A Input B

TERMINAL
WIRING+12V

GND

342

1
7

89COM.

N.O.

N.C.

10121314

Auto-reset, one shot output
Overrun, one shot output
Overrun, latch output

Prior to operation, please select both display mode and output mode by slide switch
on front panel.

611GND

ADJUSTMENT OF OUTPUT TIME

Input Selection

Relay Output

GNDGND

5A
BC

1516200/220 100/110

17When operation mode A or B is selected, please adjust output
time from 0.1 to 1 second by turning the knob below the slideswitch.

0V
0.1

Speed Selection

1SecAC Source(1) 90° Quadrature Input (from rotary encoder, etc.)

OPERATION EXAMPLESENSOR
Input A Input B

1Auto-reset, One shot output

+12V

GND

3

4

2

Overrun, One shot output

PresetValue (6)10

PresetValue (4)11
disconnected(2) Individual Add/Subtract Input (Open Collector Input)

Add

Sub

1

INPUTGND

2CountValue
OutputCountValue
Output

Reset

Reset

Overrun, Latch output

4101154Preset
Value (4)shorted54344332(3) Individual Add/Subtract Input(Contact Input(relay, microswitch, push button switch, etc.))

2

11Count
Value

3200Output
Reset

Add

Sub12

GND

CAUTION

41011• The counter will not accept count or reset signals during the first 100 ms after
power-up.
• In case Individual Add/Subtract Input is selected, additive and subtractive
signals cannot be input simultaneously.
• Shielded wires should be used for input/output leads.
• Input/output leads should be separated from power leads as much as possible.
• Wires for short circuit (jumpers) should be as short as possible.
• When noise is observed on input or power leads, noise suppressor or powersource noise filter should be used.

shorted(1) Contact Output (Relay Output)

OUTPUT
9N.C.

GND

556

126

DIMENSIONSCan be reset when& areshorted(via microswitch, open collectortransistor, etc.)

68 +0.7072 ±0.1

OUT

13disconnected 10 kHz (Open Collector Input)
shorted

30 Hz (Contact Input)

A
B
C

68 +0.7
0

COUNT SPEED
SELECTION

8N.O.

72 0.1

EXTERNAL
RESET

7COM.

Panel Cutout

RESET

100/110 VAC 0 VAC

16200/220 VAC

0 VAC

15171 2 3 4 5 6

In case of 100/110 VAC,
should be used.

&

In case of 200/220 VAC,
should be used.

&

80.1

1S

107.5 ±0.1

9.5 ±0.2

Digit Size
5.54
10.16

POWER
SOURCE

17DUSTPROOF COVER (OPTION)
In case the counter will be used in a bad environmental conditions (e.g. dust / splash),a dustproof cover should be used.
If the cover is to be used, the panel cutout should be 70 x 70 mm

LINE SEIKI CO., LTD.
Head OfficeThis manual was last revised June 15, 2017.
Contents are subject to change even without prior notice.
All Rights Reserved. Copyright © 2017, LINE SEIKI CO., LTD.

Contact
E-mai
URL

37-7 Chuo-cho, 2-Chome Meguro-ku,
Tokyo JAPAN 152-0001
TEL : 03-3716-5151 FAX : 03-3710-4552
webtrade@line.co.jp
http://www.lineseiki.com