Product Catalog
2019
Tally Counter
Electronic Counter
Electromagnetic Counter
Mechanical Counter
Length Measuring Counter
Length Sensor
Rotary Encoder
Tachometer
Thermometer
Measuring InstrumentsCompany Information
Company Overview"As long as humans live counting will be a necessity." ~ Line Seiki
Human progress and development over the years has been measured bynumbers. Statistics are as vital to ordinary students and hobbyists as they
are to experienced professionals. For 60 years now, Line Seiki has served
this everyday need to count, measure, track, and monitor various activities
with its wide variety of industrial counters, measuring instruments,and sensors.
From its humble beginnings as a counter maker in post-World War II Japan, Line Seiki has grown to become
a global leader in industrial, professional, and even personal counter-serving the needs of markets across the
Asia Pacific, Europe, and the Americas. As a pioneer in counting technology, it has perfected the science andart of measurements while keeping up with the latest counting technology. Wherever you may see humans andtheir needs to count and measure, you will see Line Seiki.
Now with five locations throughout Asia Pacific and a vast global network of distributors, Line Seiki
manufactures and supplies product models for just about every field or industry. Our products include the mostadvanced electronic, electro-mechanical and mechanical counters, tachometers, digital thermometers, tallycounters, sensors, and encoders as well as other customized products. Applications range from factory
automation to speed and time measurements for hobbyists, to everyday household uses. Moreover, manyof our products are commonly used by original equipment manufacturers (OEM), including those found in vending
machines, gaming machines, textile machines, plastic injection machines, metal stamping machines, and so on.
Our core business, above all, is product development. We fuse decades of experience to the latest advancesin technology to engage in technological cooperation and joint product development with other world leaders in
the field of counting. Thus, as long as humans live, so will Line Seiki.

Line Seiki GroupLine Seiki Co., Ltd. (Japan)
Line Seiki Philippines, Inc. (Philippines)
Line Seiki Asia Pacific, Inc. (Philippines)
LS Instrumentation Sales & Services (Philippines)
Taiwan Line Seiki Co., Ltd. (Taiwan)
Korea Line Seiki Co., Ltd. (Korea)

History
History
Feb. 1949

A trading company handling measuring tools was inaugurated in Tokyo, Japan.

Apr. 1954

Line Seiki Co., Ltd. was incorporated in Tokyo, Japan.

Jun. 1985

Shirakawa factory was established in Fukushima Prefecture, Japan.

Jun. 1988

Korea Line Seiki was established in Seoul, Korea.

Jun. 1988

Taiwan Line Seiki was established in Taipei, Taiwan.

Aug. 1995

Line Seiki Philippines, Inc. was established in Cavite, the Philippines.

Jun. 1999

Line Seiki Asia Pacific, Inc. was established in the Philippines.

Jul. 2008

LS Instrumentation Sales & Services was established in Quezon City, the Philippines.

2

DK-5000 Series

Digital Tally Counter with Timestamp

Models : DK-5000 Series

DK - 50
Input
Features05 : 5 inputs A : Counters+ timestamp
10 : 10 inputs
B : Timestamp Recorder
30 : 30 inputs
C : Counters

DK-5005 (5-inputs)+ automatic recording
D : Status Recorder
E : Counters+ timestamp+ Input Key Operation Tone
+ Preset Alarm (Total Count)

DK-5010 (10-inputs)

DK-5030 (30-inputs)

ModelsRecording Method

DK-50

A / DK-50

E

Press # Key or F + 5 Key

DK-50

B

Every 1 to 30 Key Press

Specifications

DK-50

C

DK-50

Automatic recording
at preset intervals

D

Every 1 to 30 Key Press

SoftwareRubber Key
7-Segment LCD
Counter / Status (x 5) : 7 mm (H), Total Count : 5 mm (H)Record Number / Time Display : 3.5 mm (H)Input Display
0 to 9999 (Internal : 0 to 99999) / D Type : HI or LO
Total Count Display *0 to 999999
Clock Function
Year / Month / Day / Hour (24 hour) / Minute / Second
MemoryBuilt-in flash memory
Data Communication
USB micro-B (USB 2.0)
Power Source
AAA Battery x 4 pcs. / USB Cable / AC/DC Adapter
Battery Life (Approx.)
200 hour min. **Weight
DK-5005 : Approx. 130 g / DK-5010 : Approx. 185 g /(excl. Battery)
DK-5030 : Approx. 340 g
Ambient Temp. : Operating
0 to 50 °C (Should not be frozen)
Ambient Humid. : Operating
35 to 85 %RH (Non-condensing)Ambient Temp. : Storage-10 to +60 °C (Should not be frozen)
Accessories
Instruction Manual (Battery & USB Cable Not included)

Input Type
Display Type* Except D Type (DK-5005D / DK-5010D / DK-5030D)

The companion software enables the user
to use the following functions when thedevice is connected to the computer witha general-purpose USB cable.
- Real-time display of count values
displayed on the device- Edit name of each counter- Download the data from the device- Export the downloaded data to a file with.XLS or .CSV extension

Software Specifications
CPU
Pentium2 266 MHz
OS
Windows 7, 8, 8.1, 10
Disk
Application: 50MBSpace
Java Runtime Environment: 160MB
Memory1 GB
Download Downloadable fromhttps://www.lineseiki.com** Under normal operating temperature using Alkaline type battery

Dimensions in mm

25
TIMEM

ROW

SUB

DK-500570

DK-5000 Series
F
POWER

SUB

1

2

3

4

5

#

DK-501096

DK-5030

186

Clock /
Memory No. Display
Row Display *1
Input Display
Date/Time Icon
Subtract Icon
Battery Warning Icon
Software Link Icon
DC Jack*1 DK-5010
DK-5030 only
*2 Except D TypePower Key[#] Key
Row Key *1
Input Keys
Subtract Key
Function Key
USB 2.0 Port
Total Count Display *2
Strap Hole *3*3 DK-5005 only

OptionUSB-02 (USB Cable)

170

18

* A general-purpose micro B-to-A USB
cable can be used.

AC/DC Adapter (DK-5)* Please use Line Seiki AC/DC Power Adapter3

DK-5000 Series Accessories

Digital Tally Counter with Timestamp

Models

M16-600

M16-301

M16-302

M16-601

Model
FeaturesM16-602

M16-600

M16-601 *M16-301*

M16-303

M16-602 **M16-302 **M16-303 ***Sensor Input Device Sensor Input Device Wireless LAN Sensor Input Device Wireless End Wireless Coordinator+ Wireless LAN
Adapter+ Wireless (802.15.4) Device (802.15.4)(802.15.4)
Transmission ofThis device
OOO
establishes
Function
External Input
wireless
Wireless access fromO
O
O
O
network (802.15.4)
PC to DK-5000
Wireless LAN
Wireless LAN Wireless (802.15.4) Wireless (802.15.4) Wireless (802.15.4)
Wireless Network* Windows PC + Wireless LAN Router** Windows PC + M16-303*** Windows PC + M16-302 / M16-602

Wireless System
Wireless LAN System(M16-601 / M16-301)

Wireless (802.15.4) System(M16-602 / M16-302 + M16-303)

Windows PC

Windows PC

AC/DC Power Adapter

AC/DC Power Adapter

M16-303

M16-601 / M16-301

Wireless LAN Router

M16-602 / M16-302

DK-5000 Series

DK-5000 SeriesSpecifications
Model
M16-601 / M16-301
M16-602 / M16-302 / M16-303 Model
Wireless Communication IEEE 802.11 b/g/n
IEEE 802.15.4
Input Mode
Input
Wireless Range Up to 30 m Line-of-Sight (LOS)
Operating Frequency 2.4 GHz
Operating Channel 13 (Automatic)
16
DS-SS
Modulation
DS-SS (IEEE802.11b)
MethodOFDM (IEEE802.11g/n)
Input Speed

M16-600 / M16-601 / M16-602
Open Collector / Contact Input
Non-contact Input : Open Collector
Sink Current approx. 1.5 mA L : < 1.4 V
Contact Input :Relay, Microswitch, Others

Sink Current approx. 1.5 mA
Count Input : 10 Hz (Duty 1:1)
Status Input : pulse width of ****
1 second min.
Sensor Power Supply 12 VDC (100 mA)

Sensor Input Terminal Terminal Screw
Power Source AC/DC Power Adapter (except M16-303) / USB 5 V (Consumption Current Approx. 55 mA)
Power Supply USB Power Supply : Vbus 5 V (100 mA) except M16-303
DK-5000 Connection USB 2.0 (USB Micro-B) except M16-303
Ambient Temp. : Operating 0 to 50 °C (Should not be frozen)
Ambient Humid. : Operating 35 to 85 %RH (Non-condensing)Ambient Temp. : Storage -10 to +60 °C (Should not be frozen)
Dimensions
M16-600 M16-601 M16-602 : 150 x 100 x 32 mm (excluding antenna) /M16-301 M16-302 : 75 x 50 x 32 mm (excluding antenna) / M16-303 : 97 x 48 x 16 mm

WeightAccessories

M16-600 : Approx. 240 g / M16-601 M16-602 : Approx. 250 g / M16-301 M16-302 : Approx. 60 g / M16-303 : Approx. 35 g
Instruction Manual x 1, USB On-the-Go (OTG) Host Cable x 1, Micro B-to-A USB Cable x 1, AC/DC Power Adapter x 1 (except M16-303)**** DK-5005D / DK-5010D / DK-5030D only ***** These wireless devices are compliant with radio law in Japan, the United States, Canada and EU.

4

DK-5000A / DK-5000B Series

Digital Tally Counter with Timestamp

DK-5000A Series
Type

Count Values with Date & Time

5-30 combined tallycounters with timestamp

A DK-5000A series

Electronic tally device with 5 to 30 independent counters.
TIME

:

TOTAL

1

2

3

4

5

This device can record the individual count values and the total count with a corresponding timestamp.
It can save up to 250 records in its memory.
Memory Recall Mode Display 1 (Count values)
MEM

Memory Recall Mode Display 2 (Date & Time)
MEM

YYYY

MM

DD

hh

mm

Recorded data is displayed and saved on the computer when the device is connected to the computer with USB cable.
The saved data is editable with spreadsheet software.
Count values can be displayed on the computer on a real-time basis.
Auto-AcquireAcquire All Records

Timestamp
1
2
3
4
5

Total Input1 Input2 Input32015 - 04 - 01 11:00:00 37
2015 - 04 - 01 11:05:30 162015 - 04 - 01 11:18:25 4802015 - 04 - 01 11:45:50
1
2015 - 04 - 01 11:58:03 93

5
2
50
1
50

7
0
100
0
15

9
5
15
0
3

Input4 Input5
11
0
305
0
8

5
9
10
0
17

DK-5000B Series

Timestamp on every pushType

Timestamp Recorderwith 5-30 input keys

B DK-5000B series

Date & Time is recorded whenever the keys are pressed.
The number of times the keys are pressed is displayed.
Start time and End time are recorded.
You can analyze the saved data on the computer.

No need totake notes.

Total Count3 Counts

DK-5005B

Record Start# (11:00)
Record Stop
# (12:00)

1st Count
Key 1 (11:15)
2nd Count
Key 3 (11:30)

3rd Count
Key 5 (11:45)

Auto-Acquire2015 - 04 - 01 12:00:00

Timestamp
1
2
3
4
5

2015 - 04 - 01 11:00:00
2015 - 04 - 01 11:15:03
2015 - 04 - 01 11:30:07
2015 - 04 - 01 11:45:15
2015 - 04 - 01 12:00:20

Total Input1 Input2 Input3
0
1
2
3
3

0
1
1
1
1

0
0
0
0
0

0
0
1
1
1

Input4 Input5
0
0
0
0
0

5

0
0
0
1
1

Timestamp andthe numberof inputs arerecorded.

DK-5000C / DK-5000D Series

Digital Tally Counter with Timestamp

DK-5000C Series

Record values
at preset intervals

Type5-30 combined tally counterswith automatic recording function

C DK-5000C series

The individual count values and the total count are recorded automaticallyat preset intervals with a corresponding timestamp.
You can select whether or not to reset the count values after a data log is saved.

DK-5005C

Auto-Acquire2015 - 04 - 01 23:00:00

Timestamp
1
2
3
4
5

Total Input1 Input2 Input32015 - 04 - 01 11:00:00
2015 - 04 - 01 12:00:00
2015 - 04 - 01 13:00:00
2015 - 04 - 01 14:00:00
2015 - 04 - 01 15:00:00

0
123
245
272
302

0
8
12
12
20

0
10
18
35
38

0
25
52
55
60

Input4 Input5
0
30
85
92
99

0
50
78
78
85

At preset
intervals(1 minute to 99
hours 59 minutes)

DK-5000D Series

Status Recorder (HI & LO)

Type

Status Recorder with
5-30 input keys

D DK-5000D series

The input status is toggled between HI and LO and recorded with timestamp whenever pressing input keys.
HI and LO time of each input key can be summed up.

1

2

Key1 : HI Key1 : LO Key2 : HI

Key1
HI TIMELO TIME3

Key2 : LO

Example 1
Example 2

5

Key3 : HI Key3 : LO Key4 : HI Key4 : LO Key5 : LO Key5 : LO

Key2
HI TIMELO TIME

4

Key3
HI TIME

Key4
HI TIMELO TIMEKey5
HI TIMELO TIME

Measurement for working hours of each working process.
Recording downtime of equipment by cause.
DK-5005D

Auto-Acquire2015 - 04 - 01 12:00:00

Timestamp
1
2
3
4
5

2015 - 04 - 01 11:00:00
2015 - 04 - 01 11:00:03
2015 - 04 - 01 11:00:07
2015 - 04 - 01 11:00:15
2015 - 04 - 01 11:00:20

Total Input1 Input2 Input3
-

H
-

H
-

-

Input4 Input5-6-

DK-5000E Series / Sensor Input Device

Digital Tally Counter with Timestamp

DK-5000E Series

Count up with key soundBeep at goal you set

5-30 combined tally counterswith timestampType (Keypad
Tone + Preset Alarm)

E DK-5000E series

Electronic tally device with 5 to 30 independent counters.
TIME

:

1

TOTAL

2

3

4

* The preset value is changeable.

5

When the total value reaches a preset value,* the device makes an electronic alarm sound.
You can input while looking into the microscope since an electronic sound is generated when inputting.
Key Operation Sound Setting

Auto-AcquireAcquire All Records

Timestamp
1
2
3
4
5

Preset Alarm Setting

Example :Preset Alarm Setting = Every 50 counts

Total Input1 Input2 Input32015 - 04 - 01 11:05:15
2015 - 04 - 01 11:10:30
2015 - 04 - 01 11:15:25
2015 - 04 - 01 11:20:50
2015 - 04 - 01 11:25:03

50
100
150
200
250

10
20
30
40
50

10
20
30
40
50

Microscope

Input4 Input5

10
20
30
40
50

10
20
30
40
50

10
20
30
40
50

DK-5005E

Sensor Input Device | M16-600 / M16-601 / M16-602
Sensor Input Device

M16-600+

5 combined tallycounters with timestamp

DK-5005A

Auto-AcquireAcquire All Records

Timestamp
M16-600 can transfer inputs from exteral
sensors to DK-5005A, instead of pushing
input keys.

M16-600

1
2
3
4
5

Total Input1 Input2 Input32015 - 04 - 01 11:00:00 37
2015 - 04 - 01 11:05:30 162015 - 04 - 01 11:18:25 4802015 - 04 - 01 11:45:50
1
2015 - 04 - 01 11:58:03 93

5
2
50
1
50

7
0
100
0
15

9
5
15
0
3

11
0
305
0
8

5
9
10
0
17

Place sensors wherever you want
Wire them up to M16-600
No more to stand by DK-5005A

Count Input

DK-5005A

Andon sensor

Input4 Input5

Push switch

Proximity sensor

7

DK-5000 Series Wireless Accessories

Digital Tally Counter with Timestamp

M16-301 / M16-601 (Using Customer's Wireless LAN Network)

The change of status isdisplayed by getting
the signal fromequipment directly.

Sensor Input Device+ Wireless LAN

M16-601

Status Recorder with
5 input keys

DK-5005D+

Customer's
Wireless LAN Router

Andon Sensor(M13-215)

Se
n

so

Se
n

r In

pu

so

r In

t (F

las

h)

pu

t (L

igh

Windows PC

tin

g)

Auto-Acquire2015 - 04 - 01 12:00:00

Timestamp
1
2
3
4
5

2015 - 04 - 01 11:00:00
2015 - 04 - 01 11:00:03
2015 - 04 - 01 11:00:07
2015 - 04 - 01 11:00:15
2015 - 04 - 01 11:00:20

Total Input1 Input2 Input3
-

H
-

H
-

DK-5005D

Input4 Input5--

M16-601-

M16-302 / M16-602 + M16-303 (Using Wireless Network built by M16-303)[ 8 0 2 . 1 5 . 4 ]The change of status is M16-602
displayed by getting
the signal fromM16-303equipment directly.

Sensor Input Device+ Wireless (802.15.4)+

Status Recorder with
5 input keys

DK-5005D

Wireless Coordinator(802.15.4)

Andon Sensor(M13-215)

Windows PC

Se

ns

Se

ns

or

or

Inp

ut

Inp

ut

(Fl

as
h)

(Li

gh
tin

g)

M16-303

Auto-Acquire2015 - 04 - 01 12:00:00

Timestamp
1
2
3
4
5

2015 - 04 - 01 11:00:00
2015 - 04 - 01 11:00:03
2015 - 04 - 01 11:00:07
2015 - 04 - 01 11:00:15
2015 - 04 - 01 11:00:20

Total Input1 Input2 Input3
-

H
-

H
-

-

Input4 Input5-8-

M16-602

DK-5005DDK-100 Series / DK-2300 / H-102 Series

Tally Counter

Models : DK-100 Series / DK-2300 (Digital Tally Counter)
DK-100H /DK-100D / DK-100M

Model
Display
Digit Size (H)DK-2300

7-Segment LCD
10 mm

Main : 10 mm
Sub : 5 mm
4Main : 4.5
No. of Digits
Sub : 40 to 9999
Main : -9999 to 19999
Count Range
Sub : -9999 to 9999
DK-100H : CR2032
DK-100M / 100D : CR1632
CR2032
Power Source
DK-100M / 100D : 1 Year
3 Years
Battery Life (Approx.) DK-100H : 2.5 Years
DK-100H : 42 x 65.5 x 18.4 mm DK-100M / 100D : 42.1 x 60 x 25.5 mm 50.8 x 77.8 x 18.5 mm
Dimensions
Approx. 40 g
Weight (incl. Battery) DK-100H : Approx. 30 g DK-100M / 100D : Approx. 25 g
5
°C
to
40
°C
(Should
notbefrozen)Ambient Temp. : Operating
85 %RH max. (Non-condensing)Ambient Humid. : Operating-10 to +60 °C (Should not be frozen)
Ambient Temp. : Storage
530 mm length lanyard (DK-100H / DK-2300 only), Battery, Instruction Manual
Accessories
4-Function Tally Counter
DK-100H : Hand-Held Tally / DK-100D : Desk Tally /Others
DK-100M : Desk Tally (Master Unit)

DK-100H

DK-100M

DK-100D

DK-2300

DK-100M Function

DK-2300 Function

Power Supply (to All Connected DK-100D)
AC/DC Adapter
1. Can be connected to DK-100D 9pcs maximum.
(Sold separately) 2. Can be supply power to DK-100D.
(AC / DC Adapter is necessary)Type

DK-2300

RS

M

DK-100M

DK-100D

DK-100D

Type 1
Up/Down Counter
Type 2 (Up only)
Tally of 2 Counters
Type 33 Up / Down Counters
Type 4 (Up only)
Counter with Memory

Switch1

Switch
2

ON

ON

ON

OFF

OFF

ON

OFF

OFF

To switch types, remove the battery first
before changing DIP Switch settings.

DK-100D

Connected to DK-100D and Power Supply

4-Function Tally Counter

Models : H-102 / H-102P Series (Mechanical Tally Counter)
Model

Type

No. of Units Weight (Approx.)

H-102
Hand
H-102B
Mountable
H-102M-2
H-102M-3
H-102M-4
Bank
H-102M-5
H-102M-6
H-102M-7
H-102M-8
H-102M-9
H-102M-10
Hand
H-102P
H-102PB Mountable
Bank
H-102PM-5
H-102PM-6

1
1
2
3
4
5
6
7
8
9
10
1
1
5
6

61 g
68 g
157 g
236 g
314 g
392 g
472 g
550 g
628 g
707 g
785 g
32 g
42 g
220 g
260 g

Count Method
No. of Digits
Digit Size
Count Speed
Count Range
Reset
Ambient Temp. : Operating
Ambient Humid. : Operating
Enclsoure

Hand-operated (by pushing count lever)4
H-102 : 4.8 x 2.4 mm (White on Black)H-102P : 4.0 x 2.3 mm (White on Black)200 counts/minute0 to 9999
Knob-5 °C to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)H-102 : Metal, H-102P : Plastic

H-102P
H-102PB

H-102M-5

H-102PM-5

H-102M-10
H-102

H-102B

H-102PM-6

9

Electronic Totalizer/

Hourmeter/

Minutemeter/GR2 Series

Secondmeter

Models

GR2 -

0

Type
C : Totalizer
H : Hourmeter
M : Minutemeter
S : Secondmeter

Wiring 0 : Terminal Screw
0 : Dry Contact / Open Collector /Voltage (NPN Transistor) Input
1 : Wide Range Voltage Input

Input

Reset 0 : Remote

1 : Remote & Front

Specifications
GR2-C : Totalizer

Type

GR2-H : Hourmeter

GR2-M : Minutemeter GR2-S : Secondmeter7-segment LCD 7 digits 8 mm (H)Display
0 to 9999999

Count Range

0.0 to 999999.9

Max. Count Speed
GR2-C 00 30 Hz
GR2-C 10 20 Hz

Spec. of Input

Resolution
1 count per 1/10 hour

Resolution
1 count per 1/10 minute

Resolution
1 count per 1/10 second

Mechanical / Environmental Specifications
Ambient Temp. : Operating-10 to +55 °C (Should not be frozen)

Ambient Temp. : Storage-20 to +65 °C (Should not be frozen)

Ambient Humid. : Operating

25 to 85 %RH (Non-condensing)Vibration /Malfunction : Single Amplitude 0.35 mm Frequency 10 to 55 Hz X.Y.Z. Axis 100 m/s²

Shock Resistance

Destruction : Single Amplitude 0.75 mm Frequency 10 to 55 Hz X.Y.Z. Axis 300 m/s²

Front Panel Protection

IP54

Weight (incl. Battery)Approx. 50 g

WiringM3 Terminal Screw

Electrical Specifications
Dry Contact / Open Collector / Voltage (NPN Transistor) L : <0.8 VDC H : 2 to 30 VDC Sink Current : 8 A
Wide Range Voltage
L : 0 to 5 VDC / AC
Input Impedance : 300 k
H : 20 to 250 VDC / AC
* 50 / 60 Hz Convertible
Front Reset
Remote Reset
Dry Contact / Open Collector / Voltage (NPN Transistor) L : <0.8 VDC H : 2 to 30 VDC Sink Current : 8 A
Reset
Wide Range Voltage
L : 0 to 5 VDC / AC
Input Impedance : 300 k
H : 20 to 250 VDC / AC
* 50 / 60 Hz Convertible
6 ms min.
Reset Pulse Width

Input

Power Source

3.6 V 1/2AA Lithium Battery (replaceable), 7 years min. with continuous input at 25 °C *The battery life is not a guaranteed value.
CE, RoHS, UL

ComplianceDimensions in mm
4836.9

45 +1.0
-0

39.4

10
24

22.2 -0

+1.0

3

Front

Panel Cut Size

View

44.8

View

22

Rear

4 1.1
8

5
Digit Size

10

Electronic Totalizer / Hourmeter / Minutemeter / Secondmeter GC2/GH2/GM2/GS2 Series
Models

G 2-

0

Type
C : Totalizer
H : Hourmeter
M : Minutemeter
S : Secondmeter

Wiring 0 : Terminal Block
Input

0 : Dry Contact / Open Collector /Voltage (NPN Transistor) Input
1 : Wide Range Voltage Input

Reset

0 : Remote
1 : Remote & Front

Display 6 : 6 Digits

8 : 8 Digits (GC2 only)

Specifications
Type

GC2 : Totalizer

GH2 : Hourmeter

Spec. of Input

GS2 : Secondmeter7-segment LCD 6 / 8 (GC2 only) digits 10.6 mm (H)DisplayCount Range

GM2 : Minutemeter

0 to 999999 / 0 to 99999999
Max. Count Speed
GC200 30 Hz
GC210 20 Hz

0.0 to 99999.9
Resolution
1 count per 1/10 hour

Resolution
1 count per 1/10 minute

Resolution
1 count per 1/10 second

Mechanical / Environmental Specifications
Ambient Temp. : Operating-10 to +55 °C (Should not be frozen)

Ambient Temp. : Storage-20 to +65 °C (Should not be frozen)

Ambient Humid. : Operating

25 to 85 %RH (Non-condensing)Vibration /Malfunction : Single Amplitude 0.35 mm Frequency 10 to 55 Hz X.Y.Z. Axis 100 m/s²

Shock Resistance

Destruction : Single Amplitude 0.75 mm Frequency 10 to 55 Hz X.Y.Z. Axis 300 m/s²

Front Panel Protection

IP54

Weight (incl. Battery)Approx. 60 g

WiringTerminal type : 14-22 AWG lead wires to be connected by screws

Electrical Specifications
Input

Reset

Dry Contact / Open Collector / Voltage (NPN Transistor) L : <0.8 VDC H : 2 to 30 VDC Sink Current : 8 A
Wide Range Voltage
L : 0 to 5 VDC / AC
Input Impedance : 300 k
H : 20 to 250 VDC / AC
* 50 / 60 Hz Convertible
Front Reset
Remote Reset
Dry Contact / Open Collector / Voltage (NPN Transistor) L : <0.8 VDC H : 2 to 30 VDC Sink Current : 8 A
Wide Range Voltage
L : 0 to 5 VDC / AC
Input Impedance : 300 k
H : 20 to 250 VDC / AC
* 50 / 60 Hz Convertible

Reset Pulse Width

6 ms min.

Power Source

3.6 V 1/2AA Lithium Battery (replaceable), 7 years min. with continuous input at 25 °C *The battery life is not a guaranteed value.
CE, RoHS, UL

ComplianceDimensions in mm
Dry Contact / Open Collector / Voltage (NPN Transistor) Input Type

49.8

6.5

50 -0

+0.7

Terminal No.

24.8

10.2

+0.7

12345

25 -0

38.3

12.6
29

3.5

Panel Cut Size

Wide Range Voltage Input Type

49.8

11

4.8

Terminal No.

1 2 3 4 5

Digit
Size

3.8

6 Digits

11

10.6

38.3

10.6

3.5

24.8

54
43.2

8 Digits

E48 Series

Electronic Preset Counter

Models

E48 Preset

Power Source

1 : 1 Level Preset
2 : 2 Level Preset
3 : 1 Level Preset+ Prewarn

1 : 100 to 240 VAC
2 : 12 to 24 VDC (E48-102 only)

Output0 : Relay Output
1 : Open Collector Output

Specifications

E48-101 / E48-111
E48-102 / E48-112
E48-201 / E48-211
E48-301 / E48-311
ModelBlue STN LCD with white backlight ; Digit Size : 10 x 5 mm
Display6
No. of Digits
6
No. of Digit Setting Keys
1 Level1 Level2 Levels
1 Level Preset + Prewarn
Preset Level-99999 to 999999
Setting Range
0 to 999999
Prewarn Range
Contact / Open Collector / Voltage (Selectable)
Input Mode
Open Collector (Sink Current Approx. 11 mA) L : 0 to 4 V / Voltage (Input Impedance 7 k ) L : 0 to 4 V H: 6 to 30 V (Available to 2-wire DC sensor.)
Input Signal
30 Hz, 1 kHz, 5 kHz, 10 kHz (Selectable)
Count Speed
Add / Subtract (Add / Subtract individual input, 90° quadrature input)
Count Mode
-99999 to 999999
Count Range
0.001 to 99.999 (excl. 0)
Prescale0.0, 0.00, 0.000, No Decimal Point
Decimal Point Position-99999 to 999999
Write
Front Reset, Remote Reset , Auto-reset
Reset
2 ms or 20 ms (Selectable)
Remote Reset Time
Selectable
from
A
to
G
; fixed to Mode A if output mode is not "Standard"
Operation Mode
E²PROM (10 years, rewritable 100,000 times)
MemoryRelay Output (Main : 1c, Sub : 1a) : max. resistance load of 250 VAC, 5 A / 30 VDC, 5 A
E48 - 0
OutputE48 -NPN Open Collector Output : 45 VDC, 100 mA MAX.

1

Relay Output at 30 Hz : 17 ms max. / 1 kHz, 5 kHz,10 kHz : 12 ms max. Open Collector Output at 30 Hz : 7 ms MAX. / 1 kHz, 5 kHz,10 kHz : 2 ms MAX.
1-Level Preset :Standard, Equal, Lower Limit, Upper Limit
2-Level Preset :Standard, Equal, Upper-Lower Limit, Upper-Upper Limit
1-Level Preset + Prewarn : Standard, Equal
One Shot (10 to 9990 ms)* or HOLD
* Standard Output only

Output Delay
Output Mode
Output Time
Error Display
Power Supply

In 90° quadrature mode, error message will be displayed on LCD if count value exceeds upper / lower limit of count range.(overflow error :100 to 240 VAC -15 %, +10 %12 to 24 VDC -15 %, +10 %50 / 60 Hz Convertible

Sensor Power Source
Power Consumption
Ambient Temp./Humid. : Operating
AltitudeInstallation Environment
Front Panel Protection
Compliance
Weightunderflow error :)100 to 240 VAC -15 %, +10 %50 / 60 Hz Convertible

12 VDC 100 mA-

12 VDC 100 mA

Approx. 7 VA for 240 VAC

Approx. 2.5 W for 24 VDC

Approx. 7 VA for 240 VAC-10 to +50 °C (Should not be frozen) 45 to 85 %RH (Non-condensing)
2,000 m max.
Over-voltage Category II, Pollution Degree 2 (IEC61010-1)
IP54 (Panel Surface)
CE, RoHS, UL
Approx. 160 g

Approx. 120 g

Approx. 160 g

Dimensions in mm
Digit Size
5
44.5

48

Standard Size
(AC Models)

10

RESET

99.8
63.8

Short Body Type(DC Model)

12

Accessories(Gasket, Clip, & Terminal Cover)

45 -0

The dimension of the clip is47.5 x 60.5.
Install the clip into therear of the counter, push forward
and clamp it into place while
holding the counter firmly
against the direction of the clip.

Panel
Cut Size+0.6

45 +0.6-0

Front Panel(common to
AC & DC Models)

6.2

44.5

48

G48 Series

Electronic Preset Counter

ModelsG48 - 3PresetPower Source

0 : 1 Level Preset
1 : 2 Level Preset
2 : 1 Level Preset+ Prewarn

5 : 100 to 240 VAC
6 : 12 to 24 VDC (G48-306 only)

Specifications
G48-305
G48-306
G48-315
G48-325
Model
LCD with backlight ; Digit Size : 10 x 5 mm
Display6
No. of Digits
6
No. of Digit Setting Keys
1 Level1 Level2 Levels
1 Level Preset + Prewarn
Preset Level-99999 to 999999
Setting Range
0 to 999999
Prewarn Range
Contact / Open Collector / Voltage (Selectable)
Input Mode
Open Collector (Sink current 10 mA, DC power model : power supply volt / 1.2K ) L : 0 to 4 V / Voltage (Input impedance 7 K ) L : 0 to 4 V H : 6 to 30 V (Available to 2- wire DC sensor.)
Input Signal
30 Hz, 1 kHz, 5 kHz (Selectable)
Count Speed
Add / Subtract (Add / Subtract individual input, 90° quadrature input)
Count Mode
-99999 to 999999
Count Range
0.001 to 99.999 (excl. 0)
Prescale0.0, 0.00, 0.000, No Decimal Point
Decimal Point Position-99999 to 999999
Write
Front Reset, Remote Reset , Auto-reset
Reset
2 ms or 20 ms (Selectable)
Remote Reset Time
Selectable from A to G ; fixed to Mode A if output mode is not "Standard"
Operation Mode
E²PROM (10 years, rewritable 100,000 times)
MemoryRelay Output (1a) : max. resistance load of 250 VAC, 5 A / 30 VDC, 5 A
OutputRelay Output at 30 Hz : 20 ms max. / 1 kHz, 5 kHz : 7 ms max.
Output Delay
1-LevelPreset:Standard, Equal, Lower Limit, Upper Limit
Output Mode
2-Level Preset:
Standard, Equal, Upper-Lower Limit, Upper-Upper Limit
1-Level Preset + Prewarn: Standard, Equal
One Shot (10 to 9990 ms)* or HOLD
* Standard Output only

Output Time
Error Display
Key Protection
Power Supply

In Add / Subtract mode, error message will be displayed on LCD if count value exceeds upper / lower limit of count range.(overflow error :underflow error :)Key operation can be disabled at 4 protection levels (L1, L2, L3, L4) by selecting in Program Mode and shorting key protection terminals.
100 to 240 VAC -15 %, +10 %12 to 24 VDC -15 %, +10 %100 to 240 VAC -15 %, +10 %50 / 60 Hz Convertible

50 / 60 Hz Convertible

12 VDC 100 mA
12 VDC 100 mA
Sensor Power Source
Approx. 7 VA for 240 VAC
Approx. 1.2 W for 24 VDC
Approx. 7 VA for 240 VAC
Power Consumption-10 to +50 °C (Should not be frozen) 45 to 85 %RH (Non-condensing)Ambient Temp./Humid. : Operating
2,000 m max.
AltitudeOver-voltage Category II, Pollution Degree 2 (IEC61010-1)
Installation Environment
IP54 (Panel Surface)
Front Panel Protection
CE, RoHS, UL
ComplianceApprox. 170 g
Approx. 110 g
Approx. 170 g
WeightDimensions in mm
Digit Size

10

99.8
45 -0

The dimension of the clip is47.5 x 60.5.
Install the clip into therear of the counter, push forward
and clamp it into place while
holding the counter firmly
against the direction of the clip.

Short Body Type(DC Model)

13

Panel
Cut Size

45 -0

Accessories (Clip)

+0.6

+0.6

63.8
44.5

Front Panel(common to
AC & DC Models)

6.2

44.5

48
48

5

Standard Size
(AC Models)Counter / Display / Tachometer E21/G62/E60/E10/MD/MDR/G90/G95/F90 Series
Models

E21 Series

G62 Series

F90 Series

G95 Series
G90 Series
MDR Series

Model
Type
DisplayNo. of Digits
Digit Size H (Main)
Count Mode
Count Range
Count Speed
Preset Level
Preset Range
Reset
Input Mode
OutputMemoryPower Source
Sensor Power Source
Power Consumption
Panel Cut Size
Dimensions
WeightAmbient Temp. : Operating
Ambient Humid. : Operating
Function

Model
Type
DisplayNo. of Digits
Digit Size H (Main)
Count Mode
Count Range
Count Speed
Preset Level
Preset Range
Reset
Input Mode
OutputMemoryPower Source
Sensor Power Source
Power Consumption
Panel Cut Size
Dimensions
WeightAmbient Temp. : Operating
Ambient Humid. : Operating
Function

E21 Series
Preset Counter / Tachometer
Red and Green LED
6
10 mm

MD Series

G62 Series

E60 Series

Red LED
4

Preset Counter
LCD (Backlight)
6

E10 Series

E10 Series
Red LED6
10.16 mm

Add / Subtract, 90° Quadrature-99999 to 999999
0 to 999999
0 to 9999
30 Hz / 1 kHz / 5 kHz / 10 kHz
30 Hz / 10 kHz
25 Hz / 10 kHz
1 to 20 to 9999-99999 to 999999
0 to 999999
Front / Remote / Auto
Contact
Open Collector
Voltage
Relay (1c) / Open Collector
Relay (1a) / Open Collector Relay (1c / 1a) / Open Collector
Relay (1c)
E²PROM
100 to 240 VAC
100 / 110 / 200 / 220 VAC
12 VDC 100 mA
12 VDC 60 mA
7 VA
5 VA
7 VA
4.3 VA
68 x 68 mm
54 x 54 mm
54 x 54 mm
68 x 68 mm
72 x 72 x 77 mm
60 x 60 x 106 mm
60 x 60 x 106 mm
72 x 72 x 125 mm
Approx. 280 g
Approx. 200 g
Approx. 240 g
Approx. 550 g-10 to +50 °C (Should not be frozen)
35 to 85 %RH (Non-condensing)
45 to 85 %RH (Non-condensing)
35 to 85 %RH (Non-condensing)Write / Prescale / Key Protection /Key Protection /Write / Prescale /Optional :RS-485 Serial Communication
Dust-proof Cover
Input Inhibition
Key Protection /-99999 to 999999
30 Hz / 1 kHz / 5 kHz / 8 kHz
1 to 4-99999 to 999999

MD Series

MDR Series

Totalizer / Preset Counter

G90 Series
Totalizer / Display
Red LEDG95 Series

F90 Series

Totalizer / Display

Tachometer
Red (and Green) LED
6
14.22 / 10 mm
0.0001 to 999999
0.11 Hz to 20 kHz-/20.0001 to 999999

46
614.22 mm14.22 / 10.16 mm
56.8 mm101.6 mmAdd
Add / Subtract, 90° Quadrature
0 to 999999
0 to 9999-99999 to +999999
20 Hz / 5 kHz
20 Hz / 1 kHz
20 Hz / 1 kHz- / 1 or 2-/1
- / 0 to 999999- / 0 to 9999
RemoteFront / Remote / Auto
Contact
Contact
Open Collector
Voltage
Voltage- / Relay (1c) / Open Collector
E²PROM
None or E²PROM
100 to 240 VAC
100 / 110 / 200 / 220 VAC
12 VDC 30 mA
12 VDC 50 mA / 100 mA
12 VDC 100 mA
6 VA
2.5 / 4 VA
3.5 / 5.5 VA
68 x 68 mm
68 (140) x 68 mm
401 x 149 x 47 mm
626 x 195 x 47 mm
72 x 72 x 125 mm
72 (144) x 72 x 125 mm
Approx. 1.5 kg
Approx. 3 kg
Approx. 500 to 560 g Approx. 530 / 950 to 1050 g
0 to 40 °C / -5 to +40 °C (Should not be frozen)-10 to +50 °C (Should not be frozen)
45 to 85 %RH (Non-condensing)
Optional : Dust-proof Cover
Decimal Point Position Setting / with Bracket /* Switchable between Totalizer and Display(*G90/95-303)

Contact
Open Collector
Voltage
Magnetic Sensor
Relay (1c)
100 to 240 VAC
12 VDC 100 mA / 50 mA
92 x 45 mm
96 x 48 x 106 mm
Approx. 280 g-5 to +50 °C (Should not be frozen)
Rate Indicator / Controller

Please check the details on our website. (https://www.lineseiki.com)

14

MZ Series

Electromagnetic Counter

Models

MZ -

-

Digit

Voltage6 : 6 digits
7 : 7 digits

DC12V : 12 VDC
DC24V : 24 VDC

Mounting & Connection

00 : Rear (Mounting Pitch : 15 mm) / Lead wire
10 : Rear (Mounting Pitch : 20 mm) / Lead wire
20 : Base (Horizontal) / Lead wire
30 : Behind Panel (Screw) / Lead wire
40 : Base (Vertical) / Lead wire
70 : Panel (Snap-In) / Lead wire
02 : Rear (Mounting Pitch : 15 mm) / Terminal Pins (Pitch = 12.7 mm)
12 : Rear (Mounting Pitch : 20 mm) / Terminal Pins (Pitch = 10 mm)

Specifications
6 digits : 3.0 x 2.0 mm / 7 digits : 3.0 x 1.6 mm (white on black)
12 VDC, 24 VDC-15 % to +10 %1.2 W
Permissible
One Pulse - One Count (Energizing - ½ count, De-energized - ½ count )20 Hz max.
25 ms min.
1:1
6 digits : 3,000,000 counts minimum / 7 digits : 10,000,000 counts minimum
None
Lead wire 265 mm / Terminal : Pin-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)
1,500 VAC for 1 minute
50 M minimum (500 VDC Megger)
Destruction : 16.7 Hz : Double Amplitude 4 mm
Malfunction : 10 to 55 Hz : Double Amplitude 0.5 mm
Destruction : 30 G
Malfunction : 5 G
Cover : Plastic (Black) / Base: SteelApprox. 30 g
CE, RoHS, (UL: depends on models)Digit Size
Power Source
Allowable Voltage Fluctuation
Power Consumption
Continuous Energizing
Count Method
Count Speed
Pulse Width
Make (Duty)
Count Life
Reset
Connection
Ambient Temp. : Operating
Ambient Humid. : Operating
Test Voltage
Insulation ResistanceVibration
Shock
Structure
WeightComplianceDimensions in mm
29.6

1

5.15

15 ±0.2

2.0

M3p=0.5

2.9

Lead wires can exitout of the side
1.6

5.5

MZ-670•770- DC V

1
39.5

30.5 ±0.2

30

29.6
27.8

41.5

18

27

Panel
Cut Size

17.6

23.5

5.15

MZ-612•712- DC V

15 ±0.2

7.1 ±1.0

30

1
2-

0.70

17.6
M3p=0.5

2.0

10

18

12.7

41.5

1

15

2-

0.70

29.6
5.15

20 ±0.2

17.6

29.6

7.1 ±1.0
41.5

18

11.2

2

33.5
23
10

10

MZ-602•702- DC V
30

3.5
Mounting HoleM3p=0.52.9

24.4± 0.2

3.5

46.5

19

MZ-630•730- DC V
25.4 ±0.2

1.6

2-R1.75

5.3

2- 2.6

46.5

28.6 ±0.2
2-R1.75

Mounting Hole
32.4

30

20 ±0.2

M3p=0.5

MZ-620•720- DC V
57
46.4 ±0.2
30

41.4

17.6

18

17.6
8.8

5.15

MZ-640•740- DC V

7.0

41.5

MZ-610•710

29.6

7

30

MZ-600•700

1847.2 ±0.2
60.5

MZ-600•700- DC V
MZ-610•710- DC V

ME Series

Electromagnetic Counter

ModelsME - 0 0 Digit

6 : 6 digits
7 : 7 digits

Voltage

AC100V : 100 VAC
AC200V : 200 VAC
AC220V : 220 VAC
Mounting
DC12V : 12 VDC
0 : Rear
DC24V : 24 VDC
2 : Base (Horizontal)
3 : Behind Panel (Screw)
4 : Base (Vertical)
7 : Panel (Snap-In)

Specifications
6 digits : 3.5 x 1.8 mm / 7 digits : 3.5 x 1.4 mm (white on black)100 / 200 / 220 VAC 50 / 60 Hz Convertible | 12 / 24 VDC-15 % to +10 %AC : 1.1 to 2.1 VA / DC : 1.2 W
Permissible
One Pulse - One Count (Energizing - ½ count, De-energized - ½ count )10 Hz max.
50 ms min.
1:1
3,000,000 counts minimum
None
Lead wire 250 mm-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)
1,500 VAC for 1 minute
50 M minimum (500 VDC Megger)
Destruction : 16.7 Hz : Double Amplitude 4 mm
Malfunction : 10 to 55 Hz : Double Amplitude 0.5 mm
Destruction : 30 G / Malfunction : 5 G
Cover : Plastic (Black) / Base: SteelME- 000-AC / DC V : Approx. 50 g, ME- 020-AC / DC V : Approx. 60 g, ME- 030-AC / DC
Approx. 52 g, ME- 040-AC / DC V : Approx. 60 g, ME- 070-AC / DC V : Approx. 55 g,
CE, RoHS, (UL : depends on models)Digit Size
Power Source
Allowable Voltage Fluctuation
Power Consumption
Continuous Energizing
Count Method
Count Speed
Pulse Width
Make (Duty)
Count Life
Reset
Connection
Ambient Temp. : Operating
Ambient Humid. : Operating
Test Voltage
Insulation ResistanceVibration
Shock
Structure
WeightComplianceV:

Dimensions in mm
ME- 000-AC / DC34.4
26 ±0.2

V

55.4

4.5

1

ME- 070-AC / DCV

37.5 ±0.2
35 +0.2-0

ME- 030-AC / DC

M3

V

55.4

1

41.4
34.6

58.8

1.6

ME- 020-AC / DC

7

5.5

4

16

58.8

19.7

3.5

60.5

47.2 ±0.2

1

V57
46.4 ±0.2
34.6

19.7

52.9

29

ME- 040-AC / DC

28.6 ±0.2

2.5

39

10

22.8

19.7

11.2±0.2

2- 2.6
deep 710.3 ±0.2
26 +0.2-03
34.6
25.4 ±0.2

t=1-2

Panel
Cut Size

19.5

19.7

3

34.6

V
1.6

MCU Series

Electromagnetic Counter

Models

MCU -

-

Digit

Mounting

Voltage6 : 6 digits S : Rear
AC100V : 100 VAC
7 : 7 digits P : Panel (Screw)
AC200V : 200 VAC
C : Base (Horizontal) AC220V : 220 VAC
CY : Base (Vertical) DC24V : 24 VDC

Specifications
Digit Size
Power Source
Allowable Voltage Fluctuation
Power Consumption
Continuous Energizing
Count Method
Count Speed
Pulse Width
Make (Duty)
Count Life
Reset
Connection
Ambient Temp. : Operating
Ambient Humid. : Operating
Test Voltage
Insulation ResistanceVibration
Shock
Structure
WeightCompliance

6 digits : 4.3 x 2.6 mm / 7 digits : 4.3 x 2.0 mm (white on black)100 / 200 / 220 VAC 50 / 60 Hz Convertible | 24 VDC±10 %AC : 3.4 to 4.3 VA / DC : 1.7 W
AC : Less than 5 minutes / DC : Permissible
One Pulse - One Count (Energizing - ½ count, De-energized - ½ count )10 Hz max.
50 ms min.
1:1
10,000,000 counts minimum
None
Lead wire 250 mm-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)
1,500 VAC for 1 minute
50 M minimum (500 VDC Megger)
Destruction : 16.7 Hz : Double Amplitude 4 mm
Malfunction : 10 to 55 Hz : Double Amplitude 0.5 mm
Destruction : 30 G / Malfunction : 5 G
Cover : Plastic (Black) / Base : Plastic
MCU- S-AC / DC V : Approx. 85 g, MCU- P-AC / DC V : Approx. 100 g,
MCU- C-AC / DC V : Approx. 100 g, MCU- CY-AC / DC V : Approx. 105 g
CE, RoHS, (UL : depends on models)Dimensions in mm
V

5.3

9

P-AC / DC

V

+0.5

1.6

Panel
Cut Size46

35.6 ±0.2

22.2

53.2

3.6

2-

7.5

CY-AC / DC

V
56.3

5

7

11.5

60
47.2 ±0.2

41.4
4128.6 ±0.2

24.5

2-

MCU-

4

17

55.8
47.8 ±0.2
41

2-M3

44 -0

46
3.3

V

11.5
24.5

1.6

11.5
24.5

56.4

C-AC / DC

18.1
8.2

25.6
17.6 ±0.2

41

MCU-

MCU-

41.4

16 ±0.2
18.5
24.8

S-AC / DC

27 +0.5-035.6 ±0.2

MCU-

4- 3.456.3

5

MCF Series

Electromagnetic Counter

Models

MCF -

-

Digit

VoltageMounting

4 : 4 digits
6 : 6 digits

X : Panel (Clip)
P : Panel (Screw)

Specifications
Digit Size
Power Source
Allowable Voltage Fluctuation
Power Consumption
Continuous Energizing
Count Method
Count Speed
Pulse Width
Make (Duty)
Count Life
Reset
Connection
Ambient Temp. : Operating
Ambient Humid. : Operating
Test Voltage
Insulation ResistanceVibration
Shock
Structure
WeightCompliance

AC100V : 100 VAC
AC110V : 110 VAC
AC200V : 200 VAC
AC220V : 220 VAC
DC12V : 12 VDC
DC24V : 24 VDC

4.0 x 2.4 mm (white on black)100 / 110 / 200 / 220 VAC 50 / 60 Hz Convertible | 12 / 24 VDC-15 % to +10 %AC : 2.8 to 3.1 VA / DC : 2.0 to 2.1 W
Permissible
One Pulse - One Count (Energizing - ½ count, De-energized - ½ count )10 Hz max.
50 ms min.
1:1
10,000,000 counts minimum
Manual Reset (Push Button)
Lead wire 250 mm-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)
1,500 VAC for 1 minute
50 M minimum (500 VDC Megger)
Destruction : 16.7 Hz : Double Amplitude 4 mm
Malfunction : 10 to 55 Hz : Double Amplitude 0.5 mm
Destruction : 30 G / Malfunction : 5 G
Cover : Plastic (Gray)
MCF-4X / P -AC / DC V: Approx. 100 g, MCF-6X / P -AC / DC V : Approx. 120 g
CE, RoHS, (UL : depends on models)Dimensions in mm
9.5 2

64.7

4.3

2.4

14.8
29

MCF-4X-AC / DCV

42

X-Type(Panel mount with clip)

13
63.5

4.5

9.5 2

64.7

4.3

Panel
Cut Size0 0 0 013
2- 6.0P-Type(Panel mount with screw)

2.4

50.5 +0.2
-0

3.8

14.8
29

MCF-6X-AC / DC

V

56

13
9.5 3

4.5

2- 3.4

2.4

13

23.2
45.7

MCF-6P-AC / DC
35.6

V

55.9

63.5

2- 6.018

Panel
Cut Size

25.5 +0.2-02- 3.4

2.4

23.2
45.7

35.6

MCF-4P-AC / DC

V

45.7

9.5 3

25.5 +0.2-0

3.8

37 +0.2-00 0 0 0

MCR Series

Electromagnetic Counter

Models

MCR -

-

Digit

AC100V : 100 VAC
AC200V : 200 VAC
DC24V : 24 VDC
Mounting PN : Panel (Screw)
CN : Base (Screw)

4 : 4 digits
5 : 5 digits
6 : 6 digits

VoltageSpecifications
Digit Size
Power Source
Allowable Voltage Fluctuation
Power Consumption
Continuous Energizing
Count Method
Count Speed
Pulse Width
Make (Duty)
Count Life
Reset
Connection
Ambient Temp. : Operating
Ambient Humid. : Operating
Test Voltage
Insulation ResistanceVibration
Shock
Structure
WeightCompliance4 & 5 digits : 5.4 x 3.8 mm / 6 digits : 5.4 x 2.8 mm (white on black)100 / 200 VAC 50 / 60 Hz Convertible | 24 VDC±10 %AC : 3.7 to 4.2 VA DC : 3.3 W
Permissible
One Pulse - One Count (Energizing - ½ count, De-energized - ½ count )15 Hz max.
33 ms min.
1:1
10,000,000 counts minimum
Manual Reset (Push Button)
PN Type: Soldering Terminal (Lead wires with terminal (260mm) are included) CN Type: Lead wire (AWG22)-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)
1,500 VAC for 1 minute
50 M minimum (500 VDC Megger)
Destruction : 16.7 Hz : Double Amplitude 4 mm
Malfunction : 10 to 55 Hz : Double Amplitude 0.5 mm
Destruction : 30 G / Malfunction : 5 G
Cover : Plastic (Black)
MCR- PN-AC / DC V : Approx. 260 to 265 g, MCR- CN-AC / DC V : Approx. 275 to 280 g
CE, RoHS, (UL : depends on models)Dimensions in mm
V
75.5

7.5

4- 3.4holes spot-faced in 6.4 deep 1.5
V
6.5

86

8

CN-AC / DC100
86 ±0.2
69

10.5
30 ±0.245.2

MCR-80±0.268.4 +2-040±0.244.6+2
-0

6.5 4

8

PN-AC / DC90
80 ±0.2

15.5
40 ±0.2
50

MCR-

4- 4.519

6

MB / HK-30 / HK-46 Series

Electromagnetic Counter / Hourmeter

Models (Preset Counter)
Model

Digit

Reset

Weight (Approx.)

MB-4111
MB-5111
MB-4211
MB-5211

4
5
4
5

Front

260 g
260 g
290 g
290 g

Front & Remote

Specifications
Model
No. of Digits
Digit Size
Count Mode
Count Range
Count Speed
Preset Level
Preset Range
Reset
OutputPower Source

Power Consumption
Panel Cut Size
Ambient Temp. : Operating
Ambient Humid. : Operating
Function
Mounting

MB Series
4/5
3.8 x 2.6 mm
Add
0 to 9999 / 0 to 99999
10 Hz
1
0 to 9999 / 0 to 99999
Front / Remote
Relay (1c)100 / 200 VAC
50 / 60 Hz Convertible |24 VDCCount : AC 5.0 VA / DC 3.6 W
Reset : AC 25 VA / DC 16 W
54 x 54 mm-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)
Dust-Proof
Panel mount with clip

MB Series

Models (Hourmeter)
Model

Power Source

HK-3001
HK-3011HK-3021HK-3031
HK-3041
HK-3051
HK-4601
HK-4611
HK-4621
HK-4631
HK-4661

AC

DC
AC

DC

110 V
110 V
230 V
230 V
6 to 12 V
12 to 36 V
110 V
110 V
230 V
230 V
6 to 30 V

Frequency
50 Hz
60 Hz
50 Hz
60 Hz
50 Hz
60 Hz
50 Hz
60 Hz
-

HK-30 Series

HK-46 SeriesSpecifications
Model
HK-30 Series (AC) HK-46 Series (AC) HK-30 Series (DC) HK-46 Series (DC)
HourmeterType
5 integers 2 decimal places
No. of Digits
6 integers 1 decimal place
4 mm
Digit Size
Add
Count Mode
0.00 to 99999.99 hours
Count Range
0.0 to 999999.9 hours
1/100 hours (36 seconds)
Reading Resolution
1/10 hours (6 minutes)
Integers : white on black / Decimal place : red on black
Color of Digit
Synchronous motor
Motor
Step motor
IP65
Front Panel Protection
Fork clamp
Mounting
Flat pins (6.3 x 0.8 mm) with screw terminals
Connection
Ambient Temp. : Operating-25 to +70 °C *-25 to +80 °C *-10 to +55 °C *-20 to +70 °C *±10 %Allowable Voltage Fluctuation
Panel Cut Size
33 x 22 mm
46 x 46 mm
33 x 22 mm
46 x 46 mm
WeightApprox. 32 g
Approx. 50 gApprox. 32 g
Approx. 50 g
OptionHK-30P :HK-46A :HK-30P :HK-46A :DIN 48 x 24 mm panel

DIN rail adapter* Should not be frozen

20

DIN 48 x 24 mm panel

DIN rail adapter

KZ Series

Mechanical Counter

Models

Revolution Type
Direct Drive TypeCount Type
Revolution
Direct Drive
RatchetType (I)Type (II)Type (III)Type (IV)

KZ-105 (I)KZ-205 (I)KZ-305 (I)KZ-105 (II)KZ-205 (II)KZ-305 (II)KZ-105 (III)KZ-205 (III)KZ-305 (III)KZ-105 (IV)
KZ-205 (IV)
KZ-305 (IV)I

Rotating
Shaft

Ratchet Type(with Lever)

II

Right
Top-Going

III

Right
Top-Coming

IV

Left
Top-Going

Left
Top-Coming

Specifications
Revolution (KZ-105) : Adds 1 count for every complete revolution of the shaft
Direct Drive (KZ-205) : Adds 10 counts for every complete revolution(1 count / 1/10 shaft revolution) of the shaft
Ratchet (KZ-305) : Adds 1 count for each 40 - 45° actuation of lever
5 digits
4.2 mm
KZ-105 : 1000 rotation/minute | KZ-205 : 100 rotation/minute | KZ-305 : 500 counts/minute0 to 99999
None
Approx. 10 to 14 g
-5 to +40 °C (Should not be frozen)
35 - 85 %RH max. (Non-condensing)
RoHS

Count Method

No. of Digits
Digit Size (H)Maximum Count Speed
Count Range
Reset
WeightAmbient Temp. : Operating
Ambient Humid. : Operating
ComplianceDimensions in mm
KZ-105 (I) / KZ-105 (II)KZ-105 (III) / KZ-105 (IV)

KZ-205 (I) / KZ-205 (II)KZ-205 (III) / KZ-205 (IV)

KZ-305 (I) / KZ-305 (II)KZ-305 (III) / KZ-305 (IV)

22.5
2.3

21

1.9

9

12.3

15

1.9

3

32

9 20

24

24

21.5

10.2±0.15

2x2M3x4 Cross Recessed Pan Head Screw
25.4

M3x0.54
3

8

3

M3x0.5 THRU

28.7
31.7

21

2.3

21

32

21.5

4 +0.1- 0 THRU

Lever Included in KZ-305
5.6

22.5
15

10.2±0.15
2 x 4 +0.1
- 0 THRU

9 20

RS Series

Mechanical Counter

ModelsModel

Type (I)Type (II)Type (III)Type (IV)

RS-204-5 (I)RS-4 (I)
RS-5 (I)

RS-204-5 (II)RSL-204-5 (III)

RSL-204-5 (IV)

RS-5 (II)I

Rotating
Shaft

II

Right
Top-Going

III

Right
Top-Coming

IV

Left
Top-Going

Left
Top-Coming

Specifications
Ratchet : Add 1 count per 40° to 45° actuation of lever
5 digits / RS-4 : 4 digits
RS-204 : 4.8 x 2.4 mm / RS-4, RS-5 : 4.4 x 2.8 mm
200 to 500 counts/minute0 to 99999 / RS-4 : 0 to 9999
Knob
Approx. 50 to 105 g-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)
RoHS (RS(L)-204)

Count Method
No. of Digits
Digit Size
Maximum Count Speed
Count Range
Reset
WeightAmbient Temp. : Operating
Ambient Humid. : Operating
ComplianceDimensions in mm
RS-204-5 (I) / RS-204-5 (II)
12

38.7

7 7

RSL-204-5 (III) / RSL-204-5 (IV)
32.1
20.6

30

32.1
20.6

7 7

20.5 ±0.2

12

12
24.5
30.5 ±0.2
36.6

25.4
4
3.2

3.2

12

4
24.5
30.5 ±0.2
36.6
25.4

20.5 ±0.2

1.6

6.2

1.6

6.2

50

RS-5 (I)

5 10

29.7
19

10
4

58
30 ±0.2

4
16

4- 4.24- 4.2

RS-5 (II)Mechanical Counter Operation

11
32.5
19

Add 1 count per 40° to 45°
actuation of lever56
30 ±0.2

9 10
13

4
14.8

29.5
21

4- 4.2
RSL-204-5 (IV)221637 ±0.2
45

1.6

RSL-204-5 (III)
RS-204-5 (I)RS-4 (I)
RS-5 (I)

RS-204-5 (II)
RS-5 (II)

5 10
14

22

29.2

16

16
38 ±0.246

10

4

30 ±0.2

16
38 ±0.2
46

RS-4 (I)
10.5

38.7
30

RS / LB / RL / SR / SP1-101 / SP1-102 Series

Mechanical Counter

Models (RS / LB / RL / SR Series)
Count Type Model
Ratchet RS-6 (I)DigitsDigit Size

6
5
4
5
4
5
5
4

5.0 x 3.8 mm
6.0 x 4.5 mm
5.0 x 3.8 mm
5.0 x 3.8 mm
5.0 x 3.8 mm
5.0 x 3.8 mm
6.0 x 4.5 mm
4.3 x 2.8 mm

RS-303-5 (I)Revolution LB-4 (I)
LB-5 (I)RL-4 (I)
Direct Drive RL-5 (I)RL-219-5 (I)SR-4 (I)Type
Rotating
Shaft

Weight (Approx.) Maximum Count Speed
115 g
120 g
90 g
100 g
90 g
100 g
110 g
60 g

200 to 500 counts/minute600 rotations/minute100 rotations/minute
RS-303-5 (I)200 rotations/minuteRatchetType (I)

Add 1 count per 40° to 45° actuation of lever

RevolutionI

Add 1 count per complete revolution of the shaft

Right
Top-Going

Direct Drive

Add 10 counts per complete revolution(1 count per 1/10 revolutions) of the shaft

Specifications
Count Range
Reset
Ambient Temp. : Operating
Ambient Humid. : Operating

LB-4 (I)5-digit models : 0 to 99999 / 4-digit models : 0 to 9999 / 6-digit model : 0 to 999999
Knob / SR-4 : Push Button-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)

Models (SP1-101 / SP1-102 Series)
Count Type Type (I)Type (II)Digits
Revolution SP1-101-4 (I) SP1-101-4 (II)4
SP1-101-5 (I) SP1-101-5 (II)5
4
5

Direct Drive SP1-102-4 (I) SP1-102-4 (II)SP1-102-5 (I) SP1-102-5 (II)SP1-101-4 (I)TypeType (I)Type (II)Revolution

RotationI

II

Direct Drive

Top-Going

Top-Coming

SP1-101-5 (I)Add 1 count per complete revolution of the shaft
Add 10 counts per complete revolution(1 count per 1/10 revolutions) of the shaft

Specifications
Maximum Count Speed
Count Range
Digit Size
WeightReset
Ambient Temp. : Operating
Ambient Humid. : OperatingSP1-101 : 3000 rotations/minute SP1-102 : 300 rotations/minuteSP1-101-4, SP1-102-4 : 0 to 9999 / SP1-101-5, SP1-102-5 : 0 to 99999
6.0 x 4.0 mm
Approx. 215 to 265 g
Push Button-5 to +40 °C (Should not be frozen)
85 %RH max. (Non-condensing)

Dimensions in mm

81

48

23

52±0.223

2

4- 4.37.5
18

9.5 18648
53
4- 4.51689
102

18 9.5

47

16

7 17

6

75

7

17 7

SP1-101-5 (I) / SP1-101-5 (II) /SP1-102-5 (I) / SP1-102-5 (II)

49.5
53

SP1-101-4 (I) / SP1-101-4 (II) /SP1-102-4 (I) / SP1-102-4 (II)1761±0.2
71

RY1 / MWC Series

Length Measuring Counter

ModelsRY1 Series
RY1-3:10-5 (I)RY1-3:100-5 (I)(Top-Going)

RY1-3:10-5 (II)RY1-3:100-5 (II)(Top-Coming)
M
W
C

L
IN
E
I

IK

E

S

MWC Series
MWC-3:1-5 (I)MWC-3:10-5 (I)(Clockwise)

W

M
C

IN

L
E
E

S
I

IK

MWC-3:1-5 (II)MWC-3:10-5 (II)(Anti-clockwise)

Specifications
Model

No. ofDigitsRotations for
AddingResolution

Maximum
Count Speed

Starting Torque

Weight(Approx.)

5Top-Going
0.1 m
300 m/min
15mN.m570 g
5Top-Coming
0.1 m
300 m/min
15mN.m570 g
25mN.m5
Top-Going
0.01 m
30 m/min570 g
25mN.m5
Top-Coming
0.01 m
30 m/min570 g
15mN.m5
Clockwise
1m
300 m/min
545 g
15mN.m5
Anti-clockwise
1m
300 m/min
545 g
15mN.m5
Clockwise
0.1 m
30 m/min545 g
15mN.m5
Anti-clockwise
0.1 m
30 m/min545 g
2 wheels are furnished as standard accessories. Please pay attention to copy products in the market.
Color of Figure
RY1-3:100-5 (I)MWC-3:1-5 (I)RY1-3:10-5 (I)0000m0000m000000 0
RY1-3:100-5 (II)RY1-3:10-5 (II)MWC-3:10-5 (I)Decimal place : yellow on black
Decimal place : yellow on black0 0 0 0 . 00
Integer : white on black
Integer : white on black
All figures are white on black.
Ambient Temp. : Operating -5 to +40 °C (Should not be frozen) Ambient Humid. : Operating 85 %RH max. (Non-condensing)RY1-3:10-5 (I)RY1-3:10-5 (II)RY1-3:100-5 (I)RY1-3:100-5 (II)MWC-3:1-5 (I)MWC-3:1-5 (II)MWC-3:10-5 (I)MWC-3:10-5 (II)Dimensions in mm
RY1-3 : 10-5 / RY1-3 : 100-5

Wheel
Unit
Meter Yard Meter Meter Wheel Yard Wheel
26
Model
MRM096 MRM097 MRM110
26
18
18
Type of Surface
Plain
Material of Surface Thermoplastic Elastomer Urethane6

MWC-3 : 1-5 / MWC-3 : 10-525
68

Digit Size
(2nd to 5th
wheels)
8

4.1

6
7.5

Digit Size
(1st wheel of
MWC-3:10-5)

110

134

24

7.5

Digit Size
(1st wheel of
MWC-3:1-5)
3

20

32

48

47
38

4.5

3- 5.5+0.3

39

Reset Knob

4

59

4

52 ±0.2

ATTENTION !!!
Measuring wheels are subject to slip
due to wear after prolonged usage.
They may cause measurement errors
if not replaced on time.
Please periodically check the wheels
for signs of deterioration.

97 -0

4

7.5

Digit Size

17
61 ±0.2
71

+0.3

16

4- 4.3

2

7.5

49.5
53

48

106.1 -0

157
81

CR Series

Length Sensor

ModelsModel

Sensor

CR-3:1AC
CR-3:10AC
CR-3:100AC
CR-3:1RC
CR-3:10RC
CR-3:100RC

Photoelectronic Sensor,
1 Non-contact Output(Open Collector Output)
Photoelectronic Sensor,
2 Non-contact Outputs(Open Collector Output,
90° Quadrature)

Resolution

Maximum
WeightCount Speed (Approx.)

1m
0.1 m
0.01 m1m
0.1 m
0.01 m

700 m/min
700 m/min
250 m/min
700 m/min
700 m/min
250 m/minRotation
TorqueProper Counter
Electronic Counter

605 g
20 mN.m
Bidirectional
Electronic Counter

610 gSpecifications
OutputModel

Power Source

Current
Consumption

1 Output
CR-3:1AC / CR-3:10AC
5 to 24 VDC ±10 %CR-3:100AC
CR-3:1RC / CR-3:10RC 2 Outputs (90° Quadrature)CR-3:100RC

Maximum
Load Current

Cable
Type

100 mA

2-meters,
Shielded Cable

35 mA
44 mA

* 2 wheels are furnished as standard accessories.
When you order, please specify type of wheels : meter or yard.
Please pay attention to copy products in the market.

Ambient Temp. : Operating -5 to +40 °C (Should not be frozen)
Ambient Humid. : Operating 35 to 85 %RH (Non-condensing)CE, RoHS
ComplianceDimensions in mm
71

Meter Wheel

17

Yard Wheel

26
48

26

18

157

2000 ±1018+0.3

Source: 5-24VDC

61 ±0.2

16

2

14.6

4- 4.3
Mounting Holes 52 ±0.211Count Method
Wheel

CS-3:1CT1-3:1 / CT1-3:10
CT1-3:1A / CT1-3:10A
CT1-3:1R / CT1-3:10R
CT1-3:100A / CT1-3:1000A CT1-3:100R / CT1-3:1000R
CR-3:1AC / CR-3:10AC
CR-3:100AC

CR-3:1RC / CR-3:10RC
CR-3:100RC

Top Going = Add
Top Coming = Add

Top Going = Add
Top Coming = Subtract

Unit
Meter Yard Meter
Model
MRM096 MRM097 MRM110
Type of Surface
Plain
Material of Surface Thermoplastic Elastomer Urethane
ATTENTION !!!
Measuring wheels are subject to slip
due to wear after prolonged usage.
They may cause measurement errors
if not replaced on time.
Please periodically check the wheels
for signs of deterioration.

Note : If wiring for output A & B was reversed unintentionally,
counting method will also be reversed.

25

7.5

PYYMM

97 -0

LINE SEIKI
CR-3:XXXXC

7.5

49.5

106.1 -0

+0.3

81.7

CT1 Series

Length Sensor

ModelsSensorModel

Resolution

Maximum
WeightRotation
Count Speed (Approx.) Torque

Proper Counter

Microswitch

1m
240 m/min
700 g
Electronic Counter
0.1 m
24 m/min
710 g
Electromagnetic Counter
Photoelectronic Sensor,
1m
700 m/min
730 g
Electronic Counter
1 Non-contact Output
0.1 m
500 m/min
730 g
15 mN.m(Voltage Output)
0.01 m
250 m/min
740 g
0.001 m
100 m/min
790 g
Photoelectronic Sensor,
Bidirectional
1m
700 m/min
740 g
2 Non-contact Outputs
Electronic Counter
0.1 m
500 m/min
740 g(Voltage Output /0.01 m
250 m/min
750 g
90° Quadrature)
0.001 m
100 m/min
780 g* Factory-Installed Option : Open collector output is available for models with 1 or 2 non-contact outputs.
"C" should be added to the model number. (Ex.) CT1-3:10AC

CT1-3:1
CT1-3:10
CT1-3:1A
CT1-3:10A
CT1-3:100A
CT1-3:1000A
CT1-3:1R
CT1-3:10R
CT1-3:100R
CT1-3:1000R

Specifications
OutputModel

Power Source

Current
Consumption

OutputImpedance

Maximum
Load Current

Microswitch Contact Output, Switching Capacity : 250 VAC 5 A , 125 VDC 0.5 A (Resistance Load)
1 Output
CT1-3:1 / CT1-3:10
1 Output
5 to 24 VDC ±10 %
35 mA
3.9 k
100 mA
CT1-3:1A / CT1-3:10A /CT1-3:100A
1 Output
5 to 12 VDC ±10 %
20 mA
20 mA
2k
CT1-3:1000A
70 mA
100 mA
3.9 k
CT1-3:1R / CT1-3:10R / 2 Outputs (90° Quadrature) 5 to 24 VDC ±10 %CT1-3:100R
2 Outputs (90° Quadrature) 5 to 12 VDC ±10 %
40 mA
20 mA
2k
CT1-3:1000R* 2 wheels are furnished as standard accessories.
When you order, please specify type of wheels : meter or yard.
Please pay attention to copy products in the market.

Ambient Temp. : Operating -10 to +50 °C (Should not be frozen)
Ambient Humid. : Operating 35 to 85 %RH (Non-condensing)CE, RoHS
ComplianceDimensions in mm
CT1-3:1 / CT1-3:10

Wheel
85.5

Unit
Meter Yard Meter
Model
MRM096 MRM097 MRM110
Type of Surface
Plain
Material of Surface Thermoplastic Elastomer Urethane

7

18

ATTENTION !!!

7.5

Measuring wheels are subject to slip
due to wear after prolonged usage.
They may cause measurement errors
if not replaced on time.
Please periodically check the wheels
for signs of deterioration.

5

19

30

61

32

100±1047±0.54- 5
Mounting Holes

57
130

25

95±0.5105

CT1-3:1A / CT1-3:10A / CT1-3:100A / CT1-3:1000A
CT1-3:1R / CT1-3:10R / CT1-3:100R / CT1-3:1000R

Meter Wheel

26

18

85.5
18

Yard Wheel

26

18

4- 5
Mounting Holes25

95±0.5
105

26

7.5

97 -0

+0.3

+0.3

7.5

5
47±0.557130106.1 -0

7.5

30

19

61

32

7

CS / CB / CE Series

Length Sensor / Rotary Encoder

Model (CS Series : Length Sensor)
Model

SensorResolution

CS-3:1

Reed Switch

1m

Maximum
Rotation
Count Speed Torque

WeightProper Counter

15 mN.m

Approx.
740 g

Electronic Counter
Electromagnetic Counter

500 m/minSpecifications
Ambient Temp. : Operating -5 to +40 °C (Should not be frozen) Ambient Humid. : Operating 85 %RH max. (Non-condensing)

Dimensions in mm
38
320+20

68

39

8

48

20

7.5

47±0.5

3- 6
Mounting Hole
59

59
110

25

134

Models (CB / CE Series : Rotary Encoder)

CB -

CE -

OutputC : Open Collector Output
V : Voltage Output
D : Line Driver Output (5 VDC only)

Supply Voltage

H : 12-24 VDC (Open Collector Output only)
L : 5-12 VDCResolution

100 : 100 PPR
200 : 200 PPR
250 : 250 PPR
300 : 300 PPR
360 : 360 PPR
400 : 400 PPR

500 : 500 PPR
600 : 600 PPR
800 : 800 PPR1000 : 1000 PPR
1024 : 1024 PPR
1200 : 1200 PPR

1500 : 1500 PPR
1800 : 1800 PPR
2000 : 2000 PPR
2048 : 2048 PPR2500 : 2500 PPRResolution

60 : 60 PPR
100 : 100 PPR
200 : 200 PPR
250 : 250 PPR
400 : 400 PPR
500 : 500 PPR

CB Series

CE Series

Specifications
Model
CB- HC
CB- LV
CB- LDPower Source
12 to 24 VDC ±10 %5 to 12 VDC ±10 %5 VDC ±10 %(Ripple 3 % max.)
Current Consumption
50 mA max.
OutputOpen Collector
Line Driver
Voltage
Output Signal
90° Quadrature x 2 + Zero Marking
Output Phase
A,B,Z
A, B, Z, A, B, Z
Output Capacity
Sink Current : 35 mA max.
Sink Current :Residual Voltage : 0.5 V max.
20 mA max.
±20 mA
30 V max.
H : Supply Voltage -1 V min.
H : 2.5 V min.
L : 0.5 V max.
L : 0.5 V max.
Max. Freq. Response
100 kHz
Signal Rise / Fall Time
1 s max.
0.1 s max.
Output Impedance
2.2 k
500 +10mm
Cable Length0Max. Applied Voltage
Output Voltage

Cable Outside Diameter
Weight
Shaft
Ambient Temp. : Operating
Ambient Humid. : Operating
Compliance

5.1 mmApprox. 120 g (incl. 500mm cable)
6 mm-10 to +70 °C (Should not be frozen)
35 to 95 %RH (Non-condensing)CE

27

CE5 to 12 VDC ±5 %40 mA max. (20 mA for S Type)
Voltage
90° Quadrature x 2
A, B
Sink Current :20 mA max.
Logic1 = 4 to 11 VDC
Logic0 = 0.5 VDC or less
60 kHz
1 s max.
2.0 k
4 mm
Approx. 100 g
4 mm-

TM-5 Series

Digital Tachometer

ModelsModel
TM-5000

Main Unit only - Non-contact

Type

TM-5010E

Main Unit only - Non-contact

TM-5000EK Kit Type

Measuring Method Scope of Supply- Non-contact- Contact

TM-5010EK Kit Type- Non-contact- Contact

TM-5013--

TM-5014--

Accessories- TM-5000 (Main Unit) - ZTM051 Reflective Tape x 10 pcs- AAA Alkaline Battery x 4 pcs- Instruction Manual- Accessories- TM-5010E (Main Unit) - ZTM051 Reflective Tape x 10 pcs- AAA Alkaline Battery x 4 pcs- Instruction Manual- Accessories- TM-5000 (Main Unit) - ZTM051 Reflective Tape x 10 pcs- ZTM004-SET Rubber Tip x 3 pcs- Contact Adapter- AAA Alkaline Battery x 4 pcs - ZM4052 Surface Speed Wheel 0.1 m x 1 pc
- ZM4053 Surface Speed Wheel 0.1 y x 1 pc
- Instruction Manual- C-4000 Carrying Case
- Accessories- TM-5010E (Main Unit) - ZTM051 Reflective Tape x 10 pcs- Contact Adapter- ZTM004-SET Rubber Tip x 3 pcs- AAA Alkaline Battery x 4 pcs - ZM4052 Surface Speed Wheel 0.1 m x 1 pc
- Instruction Manual- C-4000 Carrying Case
- Accessories- Contact Adapter- ZTM004-SET Rubber Tip x 3 pcs- Accessories- ZM4052 Surface Speed Wheel 0.1 m x 1 pc
- Contact Adapter- ZTM004-SET Rubber Tip x 3 pcs- Accessories- ZM4053 Surface Speed Wheel 0.1 y x 1 pc

TM-5010EK

TM-5000EK

TM-5013 / TM-5014

Specifications (Main Unit)
Model
TM-5000
TM-5010E
Measuring Range 6.0 to 99999.9 r/min 6.0 to 99999.9 r/min

Model
TM-5000
TM-5010E
Auto Power-off After 3 minutes from last measurement or key operation
Measurement Data : until the next data is defined
Data Hold Time

0.10 to 1666.66 r/s
0.60 to 9999.99 m/min

Memory Data : same as battery life-

1.00 to 16666.6 cm/s

Measuring Method Non-contact Measurement : Main Unit / Extended Sensor

0.65 to 10936.1 y/min+ Reflective Tape

0.39 to 6561.65 inch/sContact Measurement : Rubber Tip / Surface Speed Wheel

1.9 to 32808.3 f/min+ Contact Adapter

0.1 r/min

Resolution
Accuracy

50 to 300 mm, ±30°

0.1 r/min f/min, others 0.01±(0.01 % + 1 digit) ±(0.01 % + 1 digit) (r/min, m/min)±(0.05 % + 1 digit) incl. conversion
accuracy for other units

Measuring Distanceand Angle
Power Supply

1.0 to 10.0 seconds
Measurement Display : 6 digits, 7-segment LCD

Battery Life underContinuous
Measurement

Approx. 20 hours

Low-battery mark :Ambient Temp. : Operating
Ambient Humid. : Operating
Ambient Temp. : Storage
Dimensions / Weight(Battery Included)
Compliance

5 to 40 °C (Should not be frozen)

Sampling Time

Reflective light mark :Unit : r/min

Display

Unit : r/min, r/s, m/min, cm/s,y/min, inch/s, f/min
Memory : MEM1-8, MAX, MIN

AAA Alkaline Battery, 4 pcs.

35 to 85 %RH (Non-condensing)-10 to +60 °C (Should not be frozen)
122 mm (H) x 58 mm (W) x 28 mm (D)Approx. 130 g
CE, RoHS

Functions
The extended sensor can be used byturning the slide switch inside the battery
box located at the back of main unit.

1
2a
2b
r/min

2

2c

ZTM004 ZM4052 / 4053

4

cm/s m/min r/s r/min

MAX MEM
MIN

OptionZTM051
Reflective Tape, 10 pcs.

ZTM004-SET
Rubber Tip, 3 pcs.

2d
6a

3

6b

ZM4052
Surface Speed Wheel(1/10 m), 1 pc.

4a

5
5a

ZM4053MEAS

TM-5000

5cLINE SEIKI

TM-5000
1 Emitter / Receiver Part
2 Displays and Indicators

2a Photoelectric Detection Indicator
2b Unit Display
2c Memory Number Display

5e

MAX MIN.../min

5b

6

.../s

(1/10 y), 1 pc.

5d

TM-4015
Extended sensor

CALL
CLR

TM-5010(21 mm diameter,

LINE SEIKI

TM-5010E

cable length 1 m max)

TM-40152d Measurement Data Display

3 Measurement Button
4 Extended Sensor

Surface Speed Wheel

Contact Adapter
with Rotating Shaft

5 Keys / Switches

5a Measurement Mode Key
5b 8-Point Memory Data Input Key
5c 8-Point Memory Data Read/Clear Key

4a Connector for Extended

5d Rotational Speed Key

Sensor(Unit Selecting Key)28

5e Linear/Surface Speed Key(Unit Selecting Key)

6 Contact Adapter
6a Rubber Tip

6b Surface Speed Wheel

TM-7 Series

Digital Laser Tachometer

ModelsModel
TM-7000K

Type

Measuring Method Scope of Supply- Non-contact- TM-7000 (Main Unit)- Contact- Contact Adapter

Kit Type

Accessories- ZTM051 Reflective Tape x 10 pcs- ZTM004-SET Rubber Tip x 3 pcs- AAA Alkaline Battery x 4 pcs- ZM4052 Surface Speed Wheel 0.1 m x 1 pc

- Instruction Manual- C-7000 Carrying Case

- Accessories

TM-7010EK Kit Type- Non-contact- Contact

Kit Type

TM-7020K- Contact

TM-7000K- TM-7010E (Main Unit)- Contact Adapter- AAA Alkaline Battery x 4 pcs- Instruction Manual- Accessories- TM-7020 (Main Unit)- ZTM051 Reflective Tape x 10 pcs- AAA Alkaline Battery x 4 pcs- ZM4052 Surface Speed Wheel 0.1 m x 1 pc

- Instruction Manual- C-7020 Carrying Case

- ZTM004-SET Rubber Tip x 3 pcs- ZM4052 Surface Speed Wheel 0.1 m x 1 pc
- C-7000 Carrying Case
- ZTM004-SET Rubber Tip x 3 pcs
TM-7010EK- Accessories

TM-7020K

Specifications (Main Unit)
TM-7000

Model

TM-7010E

Model

TM-7020

TM-7000

TM-7010E

TM-7020

Measuring Range 6.0 to 99999.9 r/min 6.0 to 99999.9 r/min Prescale
Auto Power-off
After 3 minutes from last measurement or key operation(Non-contact)
Data Hold
Measurement Data : until the next data is defined
0.10 to 1666.66 r/sec Mode
6.0 to 15000.0 r/min
Measuring Range
1.0 to 6000.0 r/min
Time
Memory Data : until the battery runs down
0.10 to 250.00 r/sec
Contact(Contact)
Non-contact Measurement : Main Unit
Measuring0.1
to0.60 to 1500.00 m/min
Measurement :6.0 to 15000.0 r/min 1.00 to 2500.00 cm/sec 99999.9 /minwith Reflective Tape
MethodMain Unit with
0.65 to 1640.41 y/min
Contact
Measurement:
RubberTip/
SurfaceSpeed
0.01 toRubber Tip /0.39 to 984.25 in/sec
Wheel with Contact Adapter Surface Speed Wheel
1.9 to 4921.25 f/min 9999.99 /sec
: Laser, Class 2 compliant to IEC 60825-1
Resolution
0.1 r/min
0.1 r/min f/min, others 0.01
0.1 r/min
Optical System Light Source
Accuracy±(0.01 % + 1 digit) ±(0.01 % + 1 digit) (r/min, m/min)±(0.01 % + 1 digit) (Except TM-7020) Measuring Range : 50 to 1000 mm, ±40°±(0.05 % + 1 digit) incl. conversion
Wavelength: 650 nm, CW
accuracy for other units(1.0 to 2.0 seconds, with Contact Adapter)
Measurement Display : 6 digits, 7-segment LCD Laser On mark :Display

Contact Measurement mark :On Target mark :(Except TM-7020)

Unit : /min, /sec, m/min, cm/sec,

Battery Life under Approx. 30 hours
Approx.15 hours
Approx. 30 hours
Continuous
Measurement
Backlight
Blue (Auto-Backlight Off after 10 seconds of inactivity)
Ambient Temp. / Humid. : 5 to 40 °C (Should not be frozen) 35 to 85 %RH (Non-condensing)Operating-Ambient Temp. : Storage-10 to +60 °C (Should not be frozen)
Dimensions / Weight TM-7000 / TM-7010E : 125 mm (H) x 58 mm (W) x 27.4 mm (D) / Approx. 140 g(Battery Included) TM-7020 : 144 mm (H) x 58 mm (W) x 36 mm (D) / Approx. 170 g
Compliance
CE, RoHSy/min, in/sec, f/min
Prescale Mode Indicator : PS-

1 mW max.

Unit : /min

Memory : MEM1-25, MAX, MIN
Prescale Range:

AAA Alkaline Battery, 4 pcs.

Power Supply

Low-battery mark :Unit : /minPower Output

1.0 to 2.0 seconds

1.0 to 10.0 seconds

Sampling Time0.001 to 999.999

Functions
2d
2e
2g
2h

2h
2g
2f
2e

1
2

MEM

Option
2h

2a
/ min

LINE SEIKI

2b
2c

2a
PS

f/

min

in /

sec

y/

min

cm/

sec

m/

/ min

min

LINE SEIKI

LINE SEIKI

Rubber Tip, 3 pcs.

ZM4052
Surface Speed Wheel

3

4a

ENT

4b

MEM
CALL

4c

CLR

MAX MIN/min/sec

ZM4053
Surface Speed Wheel

5

MEAS

1 Emitter / Receiver Part
2 Displays and Indicators

5b

2c

4

TM-7000(Laser)

ZTM004-SET
5a(1/10 m), 1 pc.

3

TM- 7000

ZTM051
Reflective Tape, 10 pcs.

2

2d

MAX
MIN

ZTM004 ZM4052 / 4053

2d

4f

(1/10 y), 1 pc.

4e

Image for illustrative
purposes only

4d

TM- 7010

TM-7010E(Laser)2d Low-battery Indicator
2e Contact Measurement Indicator

2a Measurement Data Display 2f Memory Number & MAX-MIN Display

TM - 7020

TM-7020(Rotating Shaft)
3 Measurement Button
4 Keys / Switches
4a Enter Key

2b Prescale Setting Indicator

2g Laser On Indicator

4b Memory Call Key

2c Unit Display

2h On Target Indicator

4c Clear Key

29

Contact Adapter
with Rotating Shaft
4d Linear/Surface Speed Key(Unit Selecting Key)
4e Rotational Speed Key(Unit Selecting Key)
4f Measurement Mode Key

5 Contact Adapter
5a Rubber Tip

5b Surface Speed Wheel

TC-400A / TC-330AWP / TC-6000

Digital Thermometer

Models : TC-400A / TC-330AWP
TC-400A (Advanced type)TC-330AWP (Water Proof type)

AUTO POWER OFF
ON

1.°C / °F (Selectable : TC-400A only)
2.Auto Power-off
3.Data Hold
4.Data Record(MAX / MIN / Memory)
5.Relative Measurement

OFF

HO LD

REC

DATA
CALL

MAX-MIN

o

C

o

F

REL

TC-400A
AUTO POWER OFF
ON

OFF

HO LD

REC

DATA
CALL

MAX-MIN

REL

TC-330AWP

Specifications
Model
TC-400A
TC-330AWPType K (TC-330AWP is Water Proof Probe only)
Sensor Type-160 °C (-256 °F) to +1,372 °C (+2,502 °F)
Measuring Range+199.9 °C (+391.8 °F) : 0.1 °C (0.2 °F), +200 °C (+392 °F) : 1 °C (2 °F)
Resolution
0 °C (32 °F) to 199.9 °C (391.8 °F) : ±(0.1 % of reading +0.7 °C (1.3 °F))
Accuracyat Ambient Temp.
25 °C (77 °F) ±5 °C (9 °F)

Temperature Coefficient
Sampling TimeOperation Function-0.1 °C (31.8 °F), +200 °C (+392 °F) : ±(0.2 % of reading +1 °C (1.8 °F))±(0.015 % of reading + 0.06 °C (0.108 °F)) / °C (°F)
Approx. 0.5 second

Ambient Temp. : Operating
Ambient Humid. : Operating
Power Supply
Battery Life
Dimensions
Weight (incl. Battery)Sensor Connector
Accessories
Compliance°C / °F (Selectable)*, Auto Power-Off,
Data Hold, Data Record (MAX / MIN / Memory),
Relative Measurement * TC-400A only
0 to 50 °C (Should not be frozen)
90 %RH max. at 0 to 35°C (Non-condensing)Battery 006P (9 V)
Approx. 500 hours
166 mm (H) x 68 mm (W) x 35 mm (D)TC-400A : Approx. 200 g / TC-330AWP : Approx. 210 g
Thermocouple Connector
Water-Proof Connector
Instruction Manual, Battery 006P, 1 pc.
CE, RoHS

Probe for TC-400A
Model Max. Temp. Application / TypeKF-01A
500 °C
Surfaces(932 °F)
Surfaces /500 °C(932 °F) Hard-to-reach Surfaces
800 °CInterior of Semi-solids(1,472 °F) with High Temperature
Sheath Type : SUS316

KF-02A
KS-21A

Probe for TC-330AWP
Model Max. Temp. Tip Shape / Material
KB-101WP 300 °C (572 °F) Slanted / SUS304
KB-102WP 300 °C (572 °F) Slanted / SUS304
KS-103WP 600 °C (1,112 °F) Pointed / SUS316
KS-104WP 700 °C (1,292 °F) Spherical / SUS316
KS-105WP 800 °C (1,472 °F) Pointed / SUS316
Cable : 1,000 mm

Dimensions in mm
KF-01A90

KF-02A15

KB-101WP

100

12
3.2

12

150

KS-21A
150

40

16

15

150

25

KB-102WP
100

100

3.2

KS-104WP
300

100

100

1.6

KS-103WP
100

2.3

KS-105WP

40

12

150

100
3.2

1

Model : TC-6000

Specifications

16

Diameter of Measuring Area (mm)
100
80
25
250
800

1000

Distance to Object (mm)-30 °C to +550 °C
0.5 °C / 1 °C Automatic Change±2 % of reading or ±3 °C whichever is higher
4 times/second
0.95 fixed
100 at 1000 mm distanceAmbient Temp./Humid. : Operating 0 to 50 °C / 70 %RH max.*
Ambient Temp./Humid. : Storage -20 to +60 °C / 80 %RH max.*
Display
3.5-Digit LCD
Power Supply
9 V 006P
Dimensions / Weight (incl. Battery) 148 (H) x 105 (W) x 42 (D) mm / Approx. 160 g
Accessories
One Battery 006P, Instruction Manual,
Plastic CoverCompliance
CE, RoHS

Measuring Range
Resolution
Accuracy
Sampling Time
Emissivity
Measuring Area Size

Simple & Easy Operation
Infrared Thermometer
Handy gun type withlaser pointing function
EL back light function
In Conformance to CE(EMC)

Laser
Pointer* Should not be frozen, Non-condensing

30

EM-1000 / EH-2000 / EL-2000 / TC-3200 / TH-4000

Measuring Instruments

Model : EM-1000
AnemometerAir flow meter

Specifications

Humidity / Temp. meter<built-in sensor>

Thermometer<Probe sSold<Probes
Sold
Separately>

Light meter

Measurement
Unit
ft/minAir velocity
m/s
Air flow
Remarks

km/h
MPH
knots
CMM
CFM

Range

Digit Size (H) on LCD 8 mm
Anemometer / Air flow (CFM / CMM), Light,
Measurement
Humidity / Temp. meter, Thermometer
Indication of "- - - - "Over Input Display
006P 9 VDC battery (Heavy duty type)
Power Supply
156 mm (H) x 60 mm (W) x 33 mm (D)Dimensions
Weight (incl. Battery) Approx. 160 g
Instruction Manual, 006P Battery, Hand Strap
Accessories
Ambient Temp. : Operating 0 to 50 °C (Should not be frozen)
Ambient Humid. : Operating 80 %RH max. (Non-condensing)
Thermocouple Probes Sold Separately
Remarks
CE, RoHS
Compliance

Resolution Measurement

80 to 5,910 ft/min 1.0 ft/min
Humidity
0.4 to 30.0 m/s
0.1 m/s
1.4 to 108.0 km/h 0.1 km/h
0.9 to 67.0 MPH
0.1 MPHLight
0.8 to 58.3 knots 0.1 knots
54,000 CMM
0.001 to 1 CMM * auto range
1,908,400 CFM
0.001 to 100 CFM

ft/min : feet per minutem/s : meters per second
km/h : kilometers per hour
CMM : cubic meters/min

MPH : miles per hourknots : nautical miles per hour
ft-cd : feet candle
CFM : cubic feet/min

Temperature(Type K)

Unit

Range

%RH
10 to 95 %RH
Temperature 32 to 122 °F(thermistor) 0 to 50 °C
0 to 2,200 lux
lux
1,800 to 20,000 lux
ft-cd
0 to 204.0 ft-cd
170 to 1,860 ft-cd-148 to +2,372 °F-100 to +1,300 °C

Resolution
0.1 %RH
0.1 °F
0.1 °C
1 lux
10 lux
0.1 ft-cd
1 ft-cd
0.1 °F
0.1 °C

Models : EH-2000 / EL-2000 / TC-3200 / TH-4000

Sold
Separately
Sold Separately

Specifications
Model
Digit Size (H) on LCD
Measurement
Sensor

EH-2000

EL-2000

TC-3200

TH-4000

15 mm
pH MeterGlass Electrode Method

14 mm
Main : 14 mm / Sub : 7 mm Main : 14 mm / Sub :7 mm
Light Meter
Thermometer
Humidity / Temp. Meter
Precision Thin Film
Thermocouple Probe
Silicon Photo Diode
Humidity : HighCapacitve Sensor
Thermocouple Probe Type K / J
Temp. : Thermistor
Type K / J
Measurement Range
pH 1.00 to pH 13.00
Light : 0 to 20,000 lux
Humidity : 10 to 95 %RH-199.9 to +1,300.0 °C
Temp. : -199.9 to +1,300.0 °C
Temp. : 0 to 80 °C
Resolution
pH 0.01
Light : 1 lux
0.1 °C
Humidity : 0.01 %RH
Temp. : 0.1 °C
Temp. : 0.01 °C
Accuracy
pH 5 to pH 9 : ±0.07
Light : ±(5 % +8)lux±(0.4 %+0.8 or 1 °C)
Humidity : ±(3 % +1) %RH
Other : ±0.1 or 0.2
Temp. : ±(0.4 % +0.8 or 1 °C)
Temp. : ±0.8 °C
Sampling Time
Approx. 0.4 second
Approx. 1 second
Approx. 1 second
Approx. 0.8 second
006P 9 VDC battery (Heavy duty type)
Power Supply
Battery Life
Approx. 240 hours
Light : Approx. 48 hours
Approx. 87 hours
Approx. 104 hours
Temp. : Approx. 80 hours
Main Unit : 135 x 60 x 33 mm
Dimensions
Sensor : 9.5 x 120 mm Light Sensor : 82 x 55 x 7 mm
Sensor : 26 x 200 mm
Weight (incl. Battery)Approx. 278 g
Approx. 265 g
Approx.196 g
Approx. 264 g
Instruction Manual, 006P Battery
Accessories
0 to 50 °C (Should not be frozen)
Ambient Temp. : Operating
80 %RH max. (Non-condensing)Ambient Humid. : Operating
Thermocouple Probes Sold Separately
Remarks
CE, RoHS
Compliance

Accessories (Sold Separately)
Model
EH-2000 Model
TC-3200 TH-4000
Buffer pH 4.00 solution (40 cc)
EH-804 Real-time Measurement Software U801
U801
Buffer pH 7.00 solution (40 cc)
EH-807 Logger Software
DL2005
Replacement pH ElectrodeEH-801 USB Cable
USB-01 USB-01
pH Electrode Storage Solution (100 cc) EH-800 AC/DC Adapter
ZTC9021 ZTC9021
Thermocouple Probes (Please refer to the catalog of Thermocouple Probes.)31

www.lineseiki.com

Line Seiki Co., Ltd.
Line Seiki Co., Ltd. - Head Office37-7 Chuo-cho, 2-Chome
Meguro-ku, Tokyo JAPAN 152-0001
Please visit our company website for contact details
Fax: +81-3-37104552
E-mail: webtrade@line.co.jp