Q U I C K

S T A R T

G U I D E

D-TEK Stratus™
Refrigerant Leak Detector and Portable Monitor
English ·Español ·Deutsch ·Français ·Italiano ·中文 · 日本語·Pyccĸий· Polski · Svenska · Türkçe · Nederlands · 한국어Power button(not visible)
MODE button

Headphonejack

SENS/ZERObuttonVolumebuttonMicro USB
power input(not visible)

Eject button(not visible)
Batterycover
2EnglishProbe tip/filter

1. Turning D-TEK Stratus on and preparing for use
Englisha. Long-press the power button to turn D-TEK Stratus on or off.

NOTE: If the screen does not turn on, the battery is very low
and needs to be charged. D-TEK Stratus can be usedwhile charging.
b. D-TEK Stratus will begin a warm-up for 45 to 90 seconds. Whenwarm-up is complete, D-TEK Stratus is ready for use.
c. To switch modes, press the MODE button. This toggles between
Cloud Hunting, Pinpoint and Manual Zero modes.

NOTE: D-TEK Stratus always starts up in the last mode thatwas used.

2. Using Cloud Hunting (Portable Monitor) modea. Slowly move through the suspect areas and observe the ppm reading.
b. Follow the ppm reading to find areas of higher refrigerant
concentration. The higher the number, the higher the concentrationof refrigerant and the closer you are to the leak source.

NOTE: Lingering in an area with refrigerant for several minutes
may cause the ppm reading to decrease. If this occurs,
move to an area of lower concentration for a few seconds.
c. Press the SENS/ZERO button to enable/disable the MAX
feature. When enabled, the highest ppm level observed isshown below the main ppm display. To reset the max reading,long-press the SENS/ZERO button.

NOTE: There is no sensitivity setting in Cloud Hunting mode.
3. Using Pinpoint modea. Place the tip of D-TEK Stratus as close as possible to thesuspected leak. Do not block the air flow.
b. Slowly (2.5 to 5 cm [1 to 2 in.] per second) move the probe pasteach possible leak point. If a leak is detected, D-TEK Stratus will
alarm and the on-screen indicators will illuminate.

NOTE: In Pinpoint mode, D-TEK Stratus will automaticallyzero to the background refrigerant after a few seconds.

3

NOTE: For large leaks or for SAE applications, press the
SENS/ZERO button to switch the sensitivity mode. The
current sensitivity is displayed on the screen.

NOTE: Do not use the Super sensitivity setting for SAE applications.
4. Manual Zero mode
Manual Zero mode operates in a similar way to Pinpoint mode, butallows the user to manually zero the instrument to the background
refrigerant by pressing the SENS/ZERO button.

NOTE: There is no sensitivity setting in Manual Zero mode.
5. Removing and installing the lithium ion battery
EjectbuttonBattery
Releasea. Press the eject button to release the battery cover.
b. Move the battery release tab to the side until the battery begins
to eject.
c. Slide the battery out.
d. Align the new battery with the rails.
e. Gently push the battery along the rails until the battery releasetab engages.

NOTE: Do not force the battery. If it does not slide freely, checkalignment and try again.

4

EnglishWhen this occurs, either continue looking for a higher
concentration of refrigerant or move to an area of lower
concentration to reset the zero point.
c. When a leak is identified, pull the probe away from the leak for a few
seconds and return to the suspected leak site to verify the leak.

6. Replacing and installing the sensor
EnglishEject button

Sensor release taba. Press the eject button to release the battery cover.
b. Grasp the sensor release tab and gently pull the sensor out.
c. Align the new sensor with the rails.
d. Gently push the sensor along the rails until it clicks into place.

NOTE: Do not force the sensor. If it does not slide freely, checkalignment and try again.
e. Reinstall the battery cover.

7. Replacing filters
a. Unscrew the probe tip and remove the filter.
b. Insert the new filter into the probe tip.
c. Screw on the probe tip.

8. Charging and Battery Information
To charge D-TEK Stratus, plug the supplied wall or car charger into
an appropriate power source and plug the micro USB cable into the
D-TEK Stratus micro USB power input. The on-screen battery symbol shows the charge percentage and indicates that the battery is
charging.

NOTE: D-TEK Stratus can be operated while charging.
NOTE: A new lithium ion battery will have a charge of 30% or
less. It is recommended to fully charge the battery
before use. Allow approximately three hours for a fullcharge.

5

WARNING
This symbol is used to alert the user to the presence ofimportant operating and maintenance (servicing)
instructions in the literature accompanying this instrument.

6

EnglishCautions:• Only use a certified charger/cord with an output of 5V (dc), 1 A• Keep the device out of extremely high or low temperature locations• Temperature ranges and humidity:• Storage: -20–60°C (-4–140°F)• Operating: -20–50°C (-4–122°F)• Charging: 0–45°C (32–113°F)• Humidity: 95% RH NC max.
• Do not expose the battery to liquid• Do not use the device if you notice any damage to the battery• Do not disassemble or modify the battery• Handle and dispose of the battery per local regulations• If the recharging operation fails to complete, even when the specified recharging time has elapsed, immediately stop further
recharging• Do not leave the battery unattended while charging• Unplug the charger when the battery is fully charged• Improper use or disposal of lithium ion batteries can cause a fire
• Not intended for use in flammable environments

Table 1: Symbol Table
SymbolDescription

EnglishBattery charge 75–100%
Battery charge 50–74%
Battery charge 30–49%
Battery charge 10–29%
Battery charge <10%
Battery is charging
Volume is set to 100% (default)
Volume is set to 50%
Volume is muted
CO2 sensor is installed
Indicates Cloud Hunting mode
Indicates Pinpoint mode
Indicates Manual Zero mode
Sensitivity = Super
Sensitivity = HighSensitivity = Medium
Sensitivity = Low
Battery error
Battery is above or below ideal temperature range andmay not charge properly
Sensor error has occurred

7

Botón deencendido (novisible)Conexión paraauriculares

Botón MODE

Botón devolumen

Botón
SENS/ZERO

Entrada dealimentación
micro USB(no visible)

Botón deexpulsión(no visible)
Cubierta de labatería
8

Español

Filtro/punta
de la
sonda

Español

1. Encendido del D-TEK Stratus y preparación para su uso
a. Mantenga pulsado el botón de encendido para encender oapagar el detector D-TEK Stratus.
NOTA: Si la pantalla no se enciende,significa que el nivel de labatería es muy bajo y debe cargarse. El D-TEK Stratus
se puede utilizar mientras se carga.
b. El D-TEK Stratus iniciará una fase de calentamiento de entre 45
y 90 segundos.
c. Para cambiar de modo, pulse el botón MODE. Se alternará entre
los modos Cloud Hunting (Búsqueda en nubes), Pinpoint(Detección clásica) y Manual Zero (Ajuste punto cero).
NOTA: D-TEK Stratus siempre se inicia en el último modo
utilizado.

2. Uso del modo Cloud Hunting (Búsqueda en nubes)
monitor portátil
a. Desplácese lentamente a través de las áreas sospechosas yobserve la lectura de ppm.
b. Siga la lectura de ppm para detectar áreas con mayor
concentración de refrigerante. Cuanto mayor sea la cifra, mayor
será la concentración de refrigerante y más cerca estará de lafuente de la fuga.
NOTA: Si permanece en un área con refrigerante durantevarios minutos, la lectura de ppm puede disminuir. Si
esto ocurre, desplácese a un área con menor
concentración durante unos segundos.
c. Pulse el botón SENS/ZERO para activar/desactivar la función
MAX. Cuando esta función esté activada, debajo de la lectura
principal de ppm se mostrará el nivel más alto de ppm
observado. Para restablecer la lectura máxima, mantenga
pulsado el botón SENS/ZERO.
NOTA: No hay ninguna opción de sensibilidad en el modo
Cloud Hunting.

9a. Coloque la punta del D-TEK Stratus tan cerca como sea posibledel punto en el que se sospecha que hay una fuga. No bloqueeel flujo de aire.
b. Mueva la sonda lentamente (entre 1 y 5 cm por segundo) por
cada posible punto de fuga. Si se detecta una fuga, el D-TEK
Stratus emitirá una alarma y se encenderán los indicadores defuga en pantalla.
NOTA: En el modo Pinpoint, el D-TEK Stratus pondrá
automáticamente a cero el refrigerante de fondo
después de unos segundos. Cuando esto ocurra,
continúe buscando una mayor concentratión derefrigerante o desplácese a un área de menor
concentratión para restablecer el punto cero.
c. Si se identifica una fuga, aleje la sonda de la fuga durante unos
segundos y vuelva a colocarla en ese punto para verificar la fuga.
NOTA: En el caso de fugas grandes o con aplicaciones SAE,
pulse el botón SENS/ZERO para cambiar el modo desensibilidad. La sensibilidad actual se indica en la pantalla.
NOTA: No utilice la opción de sensibilidad Super con
aplicaciones SAE.

4. Modo Manual Zero (Ajuste punto cero)
El modo Manual Zero funciona de forma similar al modo Pinpoint,
pero permite al usuario ajustar a cero manualmente el refrigerante
de fondo en el instrumento pulsando el botón SENS/ZERO.
NOTA: No hay opción de sensibilidad en el modo Manual Zero.

10

Español

3. Uso del modo Pinpoint (Detección clásica)5. Extracción e instalación de la batería de iones de litio

Español

Botón deexpulsión

Lengüeta deliberacióna. Presione el botón de extracción para soltar la cubierta de la batería.
b. Mueva la lengüeta de liberación hacia el lado hasta que labatería comience a soltarse.
c. Deslice la batería para extraerla.
d. Alinee la nueva batería con los rieles.
e. Empuje suavemente la batería a lo largo de los rieles hasta que lalengüeta de liberación se enganche.
NOTA: No fuerce la batería. Si no se desliza libremente,compruebe que esté bien alineada e inténtelo de nuevo.

6. Sustitución e instalación del sensor
Botón de expulsión

Lengüeta de liberacióna. Presione el botón de extracción para soltar la cubierta de la batería.
b. Sujete la lengüeta de liberación del sensory tire suavemente de élpara extraerlo.
c. Alinee el nuevo sensor con los rieles.
d. Empuje suavemente el sensor por los rieles hasta que quede
encajado en su sitio.
NOTA: No fuerce el sensor. Si no se desliza libremente,compruebe que esté bien alineado e inténtelo de nuevo.

11

e. Vuelva a colocar la cubierta de la batería.
a. Desenrosque la punta de la sonda y retire el filtro.
b. Inserte el nuevo filtro en la punta de la sonda.
c. Enrosque la punta de la sonda.

8. Información sobre la carga de la batería
Para cargar el D-TEK Stratus, enchufe el cargador de pared o decoche incluido en una fuente de alimentación adecuada y conecte
el cable micro USB en el puerto de entrada micro USB delD-TEK Stratus. El símbolo de batería que aparece en pantalla indicará que la batería se está cargando.
NOTA: El D-TEK Stratus se puede utilizar mientras se estácargando.
NOTA: Las baterías de litio nuevas están cargadas a un 30 %como máximo. Se recomienda cargarlas por completoantes de su uso. Para cargar la bateríacompletamente, déjela cargándose durante 3 horas.
Precauciones:• Utilice solo cargadores y cables certificados con una salida de
5 V (CC) y 1 A• Mantenga el dispositivo alejado de lugares con temperaturas
extremadamente altas o bajas• Intervalos de temperature y humedad:• Almacenamiento: Entre -20–60°C (entre -4–140°F)• Funcionamiento: Entre -20–50°C (entre -4–122°F)• Carga: Entre 0–45°C (entre 32–113°F)• Humedad: 95% HR NC máx• La batería no se cargará en absoluto a temperaturas inferiores a
10°C o superiores a 40°C. No se cargará en absoluto a temperaturas inferiores a 0°C y superiores a 45°C
• Evite que la batería entre en contacto con líquidos• No utilice el dispositivo si observa que la batería está dañada• No desmonte ni modifique la batería• Manipule y deseche la batería de acuerdo con las regulaciones locales

12

Español

7. Sustitución de los filtros

Español

• Si la recarga no se completa una vez transcurrido el tiempo derecarga especificiado, no siga cargando la batería• Supervise en todo momento la batería mientras se carga• Desenchufe el cargador cuando la batería esté completamente
cargada• Si usa o desecha incorrectamente la batería de iones de litio,
podría provocar un incendio• Instrumento no diseñado para su uso en entornos inflamables
WARNING
Este símbolo advierte al usuario de la existencia de
instrucciones de uso y mantenimiento importantes en la
documentación incluida.

13

Tabla 1: Table de símbolos
Descripción
Carga de la batería al 75–100%
Carga de la batería al 50–74%
Carga de la batería al 30–49%
Carga de la batería al 10–29%
Carga de la batería inferior al 10%
La batería se está cargando
El volumen está al 100% (por defecto)
El volumen está al 50%
El volumen está silenciado
El sensor de CO2 está instalado
Indica el modo Cloud Hunting (Búsqueda en nubes)
Indica el modo Pinpoint (Detección clásica)
Indica el modo Manual Zero (Ajuste punto cero)Sensibilidad = Super
Sensibilidad = Alta
Sensibilidad = Media
Sensibilidad = Baja
Error de la batería
La batería está por encima o por debajo del rango de temperatura ideal y es posible que no se cargue correctamente
Se ha producido un error en el sensor

14

EspañolSímboloBetriebsschalter(nicht sichtbar)
Kopfhörer-anschluss
MODE-Taste
SENS/ZEROTaste

Lautstärkeknopf

Micro-USBEingang(nicht sichtbar)

Auswurftaste(nicht sichtbar)
Akkuabdeckung
15

DeutschSondenspitze/Filter

Deutsch

1. Einschalten des D-TEK Stratus und Vorbereitung zur
Verwendunga. Das D-TEK Stratus wird durch längeres Drücken des
Betriebsschalters ein-oder ausgeschaltet (ON/OFF).
HINWEIS: Wenn sich der Bildschirm nicht einschaltet, ist die
Batterie fast leer und muss aufgeladen werden.
Das D-TEK Stratus kann während des Aufladensverwendet werden.
b. Das D-TEK Stratus beginnt einen 45 bis 90 Sekunden langen
Aufwärmvorgang. Wenn der Aufwärmvorgang abgeschlossen
ist, ist das D-TEK Stratus einsatzbereit.
c. Zum Wechseln der Betriebsart die MODE Taste drücken. Die
verfügbaren Modi sind Cloud Hunting (Leckageortung),
Pinpoint (exakte Lokalisierung) und Manual Zero
(Nullpunktabgleich).
HINWEIS: Das D-TEK Stratus startet immer im zuletzt
benutzten Modus.

2. Verwendung des Modus Cloud Hunting (Portabler Monitor)
a. Tasten Sie den Bereich der vermuteten Undichtigkeit langsam
ab und beobachten Sie den ppm-Messwert.
b. Folgen Sie dem ppm-Messwert, um Bereiche höherer
Kältemittelkonzentration zu finden. Je höher der Messwert um sohöher ist die Kältemittelkonzentration und um so näher befinden
Sie sich an der Leckquelle.
HINWEIS: Wenn Sie mehrere Minuten in einem Bereich mit
Kältemittelverweilen, kann der ppm-Messwert
abfallen. In diesem Fall solten Sie für ein paar
Sekunden auf einen Bereich mit geringerer
Konzentration übergehen.
c. Drücken Sie die Taste SENS/ZERO, um die Funktion MAX zu
aktivieren/deaktivieren. Wenn diese Funktion aktiviert ist, wird
die höchste gemessene ppm-Konzentration unter der16

Hauptanzeige für ppm angezeigt. Um den höchsten Messwert
zurückzusetzen, die Taste SENS/ZERO länger drücken.
HINWEIS: Im Modus Cloud Hunting gibt es keine
Empfindlichkeitseinstellung.
a. Bringen Sie die Spitze des D-TEK Stratus möglichst nahe an die
vermutete Leckstelle. Luftstrom nicht blockieren.
b. Bewegen Sie die Sonde langsam (etwa 2,5 bis 5 cm pro
Sekunde) an den einzelnen potenziellen Leckstellen vorbei.
Wird ein Leck erkannt, signalisiert das D-TEK Stratus einen
Alarm und die Bildschirmanzeigen leuchten auf.
HINWEIS: Im Pinpoint-Modus stellt das D-TEK Stratus den
Nullpunkt nach ein paar Sekunden automatisch auf
die Hintergrundkonzentration des Kältemittels ein.
In diesem Fall sollten Sie entweder nach einer
höheren Konzentration des Kältemittels suchen oderin einen Bereich niedrigerer Konzentrationübergehen, um den Nullpunktzurückzusetzen.
c. Wenn ein Leck identifiziert wurde, entfernen Sie die Sonde für ein
paar Sekunden von der undichten Stelle und bringen Sie sie
dann wieder an die vermutete Leckstelle zurück, um das Leckzu verifizieren.
HINWEIS: Für große Lecks oder für SAE-Anwendungenkönnen Sie den Empfindlichkeitsmodus durch
Drücken der SENS/ZERO-Taste ändern. Die
aktuelle Empfindlichkeit wird auf dem Bildschirmangezeigt.
HINWEIS: Für SAE-Anwendungen nicht die
Empfindlichkeitseinstellung Super verwenden.

4. Manueller Nullpunktabgleich
Der Modus manueller Nullpunktabgleich arbeitet ähnlich wie derPinpoint-Modus, ermöglicht jedoch den manuellen Nullpunktabgleich

17

Deutsch

3. Verwendung des Pinpoint-Modus

des Geräts auf die Hintergrundkonzentration des Kältemittels durch
Drücken der Taste SENS/ZERO.
HINWEIS: Im Modus manueller Nullpunktabgleich gibt es
keine Empfindlichkeitseinstellung.

Deutsch

5. Austausch des Lithium-Ionen-Akkus
Auswurftaste

Akkuentriegelunga. Drücken Sie die Auswurftaste, um die Akkuabdeckung freizugeben.
b. Schieben Sie die Akku-Entriegelungslasche zur Seite, bis sichder Akku löst.
c. Ziehen Sie den Akku dann heraus.
d. Richten Sie den neuen Akku auf die Führungsschienen aus.
e. Schieben Sie den Akku dann vorsichtig entlang derFührungsschienen ein, bis die Akku-Entriegelungslasche
einrastet.
HINWEIS: Dabei bitte keine Gewalt anwenden. Wenn der
Akku sich nicht unbehindert einschieben lässt,überprüfen Sie seine Ausrichtung und versuchen
Sie es erneut.

6. Austausch und Installation des Sensors
Auswurftaste

Sensor-Entriegelungslaschea. Drücken Sie die Auswurftaste, um die Akkuabdeckung freizugeben.

18

7. Austausch der Filter
a. Schrauben Sie die Sondenspitze ab und nehmen Sie den Filter
heraus.
b. Setzen Sie den neuen Filter in die Sondenspitze ein.
c. Schrauben Sie die Sondenspitze wieder auf.

8. Informationen über das Aufladen und den Akku
Zum Aufladen des D-TEK Stratus das mitgelieferte Wand- oder
Fahrzeugladegerät an eine geeignete Stromquelle anschließen unddas Mikro-USB-Kabel in den Mikro-USB-Stromeingang desD-TE Stratus einstecken. Das Akkusymbol auf dem Display zeigt
die Akkuladung in Prozent an und gibt an, dass der Akku aufgeladen wird.
HINWEIS: Das D-TEK Stratus kann während des Aufladensverwendet werden.
HINWEIS: Ein neuer Lithium-Ionen-Akku hat eine Akkuladungvon 30 % oder weniger. Es wird empfohlen, den
Akku vor der Verwendung vollständig aufzuladen.
Die vollständige Aufladung sollte ca. 3 Stunden in
Anspruch nehmen.

19

Deutschb. Greifen Sie die Sensor-Entriegelungslasche und ziehen Sie den
Sensor vorsichtig heraus.
c. Richten Sie den neuen Sensor auf die Führungsschienen aus.
d. Schieben Sie den Sensor vorsichtig entlang derFührungsschienen ein, bis er einrastet.
HINWEIS: Dabei bitte keine Gewalt anwenden. Wenn der
Sensor sich nicht unbehindert einschieben lässt,überprüfen Sie seine Ausrichtung und versuchen
Sie es erneut.
e. Setzen Sie die Akkuabdeckung wieder auf.

VORSICHT:• Bitte nur ein zertifiziertes Ladegerät und Kabel mit einem Ladestrom
von 5 Volt DC/1 Ampere verwenden• Gerät bitte nicht an Orten mit extremen Temperaturen aufbewahren• Termperaturbereiche und Luftfeuchtigkeit:• Lagerung: -20–60°C (-4–140°F)
Deutsch• Bei Betrieb: -20–50°C (-4–122°F)• Während des Ladens: 0–45°C (32–113°F)• Luftfeuchtigkeit: Maximal 95% relative Luftfeuchtigkeit, nicht
kondensierend• Bei Temperaturen unter 10°C oder über 40°C kann der Akku nicht
schnell aufgeladen werden. Bei Temperaturen unter 0°C oder über45°C kann der Akku gar nicht aufgeladen werden• Akku nicht in Kontakt mit Flüssigkeiten bringen• Gerät nicht benutzen, wenn Schäden am Akku festgestellt werden• Akku nicht auseinandernehmen oder modifizieren• Akku in Übereinstimmung mit örtlich geltenden Vorschriften entsorgen• Wenn der Wiederaufladevorgang nicht beendet wird, obwohl die
angegebene Ladezeit abgelaufen ist, sollten Sie jedes weitere
Aufladen sofort anhalten• Akku beim Aufladen nicht unbeaufsichtigt lassen• Wenn der Akku voll aufgeladen ist, Ladegerät vom Netz trennen• Die unsachgemäße Verwendung oder Entsorgung vonLithium-Ionen-Akkus kann einen Brand verursachen• Nicht zur Verwendung in feuergefährlichen Umgebungen
WARNING
Dieses Symbol wird verwendet, um den Benutzer auf wichtige
Betriebs- und Wartungsanweisungen in der diesem Gerät
beiliegenden Dokumentation hinzuweisen.

20

Table 1: Tabelle de Symbole
Beschreibung
Akkuladung 75–100%
Akkuladung 50–74%
Akkuladung 30–49%
Akkuladung 10–29%
Akkuladung <10%
Akku wird aufgeladen
Lautstärke ist auf 100% eingestellt (Standardeinstellung)
Lautstärke ist auf 50% eingestellt
Lautstärke ist stumm geschaltet
CO2 Sensor ist installiert
Zeigt Cloud Hunting-Modus an (Leckageortung)Zeigt Pinpoint-Modus an (exakte Lokalisierung)
Zeigt manueller Nullpunktabgleich an
Empfindlichkeit = Super (Super)Empfindlichkeit = High (Hoch)
Empfindlichkeit = Medium (Mittel)
Empfindlichkeit = Low (Niedrig)
Akkufehler
Akku befindet sich unterhalb oder oberhalb des idealen Temperaturbereichs und wird möglicherweise
nicht korrekt aufgeladen
Es ist ein Sonsorfehler aufgetreten

21

DeutschSymbolFrançais
Français

Embout desonde/filtre

Bouton d’alimentation (nonvisible)Prise decasqued’écoute

Bouton MODE

Bouton duvolumeBoutonSENS/ZERO(Sonde/zéro)

Entrée d’alimentation
Micro USB(non visible)

Bouton d’éjection(non visible)
Couvercle
de labatterie
22

Français

1. Mise sous tension du D-TEK Stratus et préparation pourl’utilisation
a. Appuyez longuement sur le bouton d’alimentation pour mettre le
D-TEK Stratus sous/hors tension.
REMARQUE : Si l’écran ne s’allume pas, la batterie est très
faible et doit être chargée. Il est possible d’utiliser
le D-TEK Stratus pendant qu’il est en charge.
b. Le D-TEK Stratus commence un préchauffage de 45 à 90
secondes. Lorsque le préchauffage est terminé, l’appareil
D-TEK Stratus est prêt à être utilisé.
c. Pour changer de mode, appuyez sur le bouton MODE. Cela fait
passer l’appareil en mode Cloud Hunting (Recherche
Ambiance), Pinpoint (Recherche Point de Fuite) ou Manual
Zero (Zéro Manuel).
REMARQUE : Le D-TEK Stratus démarre toujours dans ledernier mode utilisé.

2. Utilisation du mode Cloud Hunting (Recherche Ambiance) (moniteur portable)
a. Déplacez-vous lentement dans les zones suspectes et observez
la mesure en ppm.
b. Suivez la mesure en ppm pour trouver les zones où laconcentration de fluide frigoirgène est plus élevée. Plus le chiffre
est élevé, plus la concentration de fluide frigorigène est élevéeet plus vous vous approchez à la source de la fuite.
REMARQUE : Le fait de rester dans une zone contenant dufluide frigorigène pendant plusieurs minutes,
peut entraîner une diminution de la mesure en
ppm. Si cela se produit, allez dans une zone
de concentration plus faible pendant quelques
secondes.
c. Appuyez sur le bouton SENS/ZERO (Sonde/zéro) pouractiver/désactiver la fonction MAX. Lorsque cette fonction est
activée, le niveau en ppm le plus élevé observé apparaît sous23

l’affichage principal des valeurs en ppm. Pour réinitialiser lalecture maximale, appuyez longuement sur le bouton
SENS/ZERO (Sonde/zéro).
REMARQUE : Le mode Cloud Hunting (Recherche Ambiance)
ne comporte pas de réglage de sensibilité.
a. Placez l’embout du D-TEK Stratus aussi près que possible de lafuite suspectée. Ne bloquez pas la circulation d’air.
b. Déplacez lentement la sonde (2,5 à 5 cm par seconde) en laplaçant devant chaque point de fuite possible. En cas dedétection d’une fuite, le D-TEK Stratus émet une alarme et lestémoins s’allument sur l’écran.
REMARQUE : En mode Pinpoint (Recherche Point de Fuite),le D-TEK Stratus se remet automatiquement à
zéro sur le fluide frigorigéne de fond après
quelques secondes. Lorsque cela se produit,
continuez à chercher une concentration plusélevée de fluide frigorigène ou allez dans unezone de concentration plus faible pourréinitialiser le point zéro.
c. Lorsqu’une fuite est identifiée, éloignez la sonde de la fuite
pendant quelques secondes et ramenez-la sur le site de la fuite
suspectée pour vérifier cette derniére.
REMARQUE : Pour les fuites importantes ou pour lesapplications SAE, appuyez sur le bouton
SENS/ZERO (Sonde/zéro) pour changer lemode de sensibilité. La sensibilité en cours est
affichée à l’écran.
REMARQUE : N’utilisez pas le réglage de sensibilité Super
pour les applications SAE.

4. Mode Manual Zero (Zéro Manuel)Le mode Manual Zero (Zéro Manuel) fonctionne de manière similaire
au mode Pinpoint (Recherche Point de Fuite), mais permet de mettre

24

Français
Français

3. Utilisation du mode Pinpoint (Recherche Point de Fuite)

l’instrument à zéro manuellement en appuyant sur le bouton
SENS/ZERO (Sonde/zéro).
REMARQUE : Le mode Manual Zero (Zéro Manuel) ne
comporte pas de réglage de sensibilité.

5. Retrait et installation de la batterie lithium-ion

Français

Boutond’ejection
Déverrouilllage de labatteriea. Appuyez sur le bouton d’éjection pour ouvrir le couvercle de labatterie.
b. Déplacez la languette de déverrouillage de la batterie sur le côtéjusqu’à ce que la batterie commence à sortir.
c. Sortez la batterie en la faisant coulisser.
d. Alignez la nouvelle batterie sur les rails.
e. Poussez doucement la batterie le long des rails jusqu’à ce que lalanguette de déverrouillage de la batterie s’enclenche.
REMARQUE : Ne forcez pas la batterie. Si elle ne coulisse
pas librement, vérifiez l’alignement et réessayez.

6. Remplacement et installation de la sonde
Bouton d’ejection

Languette de déverrouillage
de la sondea. Appuyez sur le bouton d’éjection pour ouvrir le couvercle de labatterie.

25

7. Remplacement du filtre
a. Dévissez l’embout de la sonde et retirez le filtre.
b. Introduisez le filtre neuf dans l’embout de la sonde.
c. Vissez sur l’embout de la sonde.

8. Informations sur la charge et la batterie
Pour charger le D-TEK Stratus, branchez le chargeur mural ou
automobile fourni dans une source d’alimentation appropriée etbranchez le câble micro USB dans l’entrée d’alimentation micro
USB du D-TEK Stratus. Le symbole de batterie à l’écran indique lepourcentage de charge et que la batterie est en cours de charge.
REMARQUE : Il est possible d’utiliser le D-TEK Stratus
pendant qu’il est en charge.
REMARQUE : Une batterie lithium-ion neuve aura une charge
de 30 % ou moins. Il est recommandé decharger complètement la batterie avant del’utiliser. Il faut compter environ 3 heures pourune charge complète.

26

Français
Françaisb. Saisissez la languette de déverrouillage de la sonde et extrayez
doucement la sonde.
c. Alignez la nouvelle sonde sur les rails.
d. Poussez doucement la sonde le long des rails jusqu’à ce qu’elle
s’enclenche en place.
REMARQUE : Ne forcez pas la sonde. Si elle ne coulisse pas
librement, vérifiez l’alignement et réessayez.
e. Remettez le couvercle de la batterie en place.

Précautions:• N’utilisez qu’un chargeur/cordon certifié avec une sortie de 5 V(cc),
1 A. Plages de température et humidité:• Entreposage: -20–60°C (-4–140°F)

Français• Fonctionnement: -20–50°C (-4–122°F)• Charge: 0–45°C (32–113°F)• Humidité: 95% HR max. sans condensation• Conservez l’appareil à l’écart des endroits à températures extrêmement élevées ou basses• La batterie ne se rechargera pas rapidement à des températures
inférieures 10°C (50°F) ou supérieures à 40°C (104°F). Ella ne serechargera pas du tout à des températures inférieures 0°C (32°F)
ou supérieures à 45°C (113°F)• N’exposez pas la batterie à un liquide• N’utilisez pas l’appareil si vous remarquez que la batterie est
endommagée• Ne démontez jamais ni modifiez la batterie• Manipulez et éliminez la batterie conformément aux réglementations
locales• Si la recharge ne s’achéve pas, même lorsque le temps spécifié s’estécoulé, arrêtez immédiatement la charge• Ne laissez pas la batterie sans surveillance pendant le charge• Débranchez le chargeur lorsque la batterie est complétement chargée• L’utilisation ou l’élimination incorrecte des batteries lithium-ion peut
provoquer un incendie• Ne doit pas êntre utilisé dans un environnement inflammable
WARNINGCe symbole est employé pour avertir l’utilisateur de la présenced’instructions d’utilisation et de maintenance (entretien) importantes
dans les documents liés à cet instrument.

27

Table 1: Tableau des symboles
Description
Charge de la batterie de 75–100%
Charge de batterie de 50–74%
Charge de la batterie de 30–49%
Charge de la batterie de 10–29%
Charge de la batterie <10%
La batterie est en cours de charge
Le volume est réglé à 100% (par défaut)
Le volume est réglé à 50%
Le volume est coupé
Le capteur CO2 est installéeIndique le mode Cloud Hunting (Recherche Ambiance)
Indique le mode Pinpoint (Recherche Point de Fuite)
Indique le mode Manual Zero (Zéro Manuel)Sensibilité = Super
Sensibilité = Élevée
Sensibilité = Moyenne
Sensibilité = Faible
Erreur de datterie
La betterie est au-dessus ou en dessous de la plagede température idéale et peut ne pas se charger correctement
Une erreur de sonde s’est produite

28

Français
Français

SymboleItaliano

Punta/filtro
sonda

Pulsante di alimentazione(non visibile)
Presa cuffie

Pulsante
MODALITÀ

Pulsante
volumePulsante
SENS/ZERO

Input Micro
USB (nonvisibile)

Pulsante diespulsione(non visibile)
Coperchiodel vano
batterie29

Italiano

1. Accensione di D-TEK Stratus e preparazione all’usoa. Premere a lungo il pulsante di alimentazione per accendere
D-TEK Stratus.
NOTA: Se lo schermo non si accende, la batteria è molto
scarica e deve essere caricata. D-TEK Stratus può
essere usato durante la carica.
b. Il tempo di riscaldamento di D-TEK Stratus è di 45–90 sec.
Quando il riscaldamento è completo, D-TEK Stratus è pronto perl’uso.
c. Per cambiare modalità, premere il pulsante MODALITÀ.
Quest’operazione permette di alternare tra le modalità Ricercadelle nubi, Localizzazione e Azzeramento manuale.
NOTA: D-TEK Stratus si accende sempre nell’ultima modalitàutilizzata.

2. Utilizzo della modalità ricerca delle nubi (monitor portatile)a. Attraversare lentamente le aree sospette e osservare la lettura ppm.
b. Seguire la lettura ppm per trovare le aree con una
concentrazione di refrigerante più elevata. Più alto è il numero,più alta è la presenza di refrigerante e più vicini sarete all’origine
della perdita.
NOTA: Trattenersi in un’area con refrigerante per diversi
minuti potrebbe causare la riduzione della lettura ppm.
In questo caso, spostarsi in un’area con una
concentrazione ridotta per qualche secondo.
c. Premere il pulsante SENS/ZERO per abilitare/disabilitare lafunzione MAX. Quando abilitata, il livello ppm più alto rilevato èmostrato sotto il display ppm principale. Per ripristinare la letturamax, premere a lungo il pulsante SENS/ZERO.
NOTA: Non vi è alcuna impostazione di sensibilità nella
modalità Ricerca delle nubi.

30

a. Posizionare la punta del D-TEK Stratus il più vicino possibile al
punto in cui si sospetta la perdita. Non bloccare il flusso d’aria.
b. Procedendo lentamente (2,5–5 cm al secondo), passare lasonda su tutti i possibili punti di perdita. Se viene rilevata una
perdita, il D-TEK Stratus emette un suono e gli indicatori sullo
schermo lampeggiano.
NOTA: In modallità Localizzazione, il D-TEK Stratus si azzera
automaticamente, dopo qualche secondo, al "rumore difondo." Quando ciò si verifica, continuare a cercare
una concentrazione più alta di refrigerante oppure
spostarsi in un’area con concentrazione ridotta per
tornare al punto zero.
c. Quando viene identificata una perdita, allontanare la sonda per
alcuni secondi e poi riavvicinarla al punto sospetto per verificare lapresenza della perdita.
NOTA: Per le perdite importanti o per le applicazioni SAE,
premere il pulsante SENS/ZERO per cambiare lamodalità di sensibilità. La sensibilità corrente èvisualizzata sullo schermo.
NOTA: Non usare l’impostazione di sensibilità Super per leapplicazioni SAE.

4. Modalità Azzeramento manuale
La modalità Azzeramento manuale opera analogamente alla modalità Localizzazione, ma consente all’utente di riportare manualmente
lo strumento al rumore di fondo premendo il pulsante SENS/ZERO.
NOTA: Non vi è alcuna impostazione di sensibilità nella
modalità Azzeramento manuale.

31

Italiano

3. Utilizzo della modalità Localizzazione

5. Rimozione e installazione della batteria agli ioni di litio
Pulsante diespulsione

ItalianoRimozione
batteria
a. Premere il pulsante di espulsione per togliere il coperchio delvano batterie.
b. Spostare la linguetta di rilascio della batteria sul lato finchè labatteria non inizia a uscire.
c. Far scorrere la batteria.
d. Allineare la nuova batteria alle guide.
e. Spingere delicatamente la batteria lungo le guide finchè lalinguetta di rilascio della batteria non si aggancia.
NOTA: Non forzare la batteria. Se non scorre liberamente,verificare il corretto allineamento e riprovare.

6. Sostituzione e installazione del sensore
Pulsante di espulsione

Linguetta per la rimozionedel sensore
a. Premere il pulsante di espulsione per togliere il coperchio delvano batterie.
b. Afferrare la linguetta per la rimozione del sensore ed estrarredelicatamente il sensore.
c. Allineare il nuovo sensore alle guide.

32

d. Spingere delicatamente il sensore lungo le guide finchè non
scatta in posizione.
NOTA: Non forzare il sensore. Se non scorre liberamente,verificare il corretto allineamento e riprovare.
e. Reinstallare il coperchio del vano batterie.

7. Sostituzione dei filtri

8. Informazioni sulla ricarica e sulla batteria
Per caricare il D-TEK Stratus, collegare il caricabatterie da parete oda auto fornito a una fonte di alimentazione adeguata e collegare ilcavo micro USB all'alimentazione micro USB dello D-TEK Stratus. Il
simbolo della batteria sullo schermo mostra la percentuale di ricarica e indica che la batteria è in fase di carica.
NOTA: il D-TEK Stratus può essere usato durante la ricarica.

NOTA: una batteria agli ioni di litio nuova avrà un livello di
carica pari al 30% o inferiore. Si consiglia di caricarecompletamente la batteria prima dell'uso. La ricarica
completa richiede circa 3 ore.

33

Italianoa. Svitare la punta della sonda e rimuovere il filtro.
b. Inserire il nuovo filtro nella punta della sonda.
c. Avvitare la punta della sonda.

Avvertenze:• Usare esclusivamente un caricatore/cavo certificato con una potenzapari a 5 V(cc), 1 A• Intervalli di temperature e umidità:• Conservazione: -20–60°C (-4–140°F)

Italiano• Funzionamento: -20–50°C (-4–122°F)• Carica: 0–45°C (32–113°F)• Umidità: 95% max di umidità rel. senza condensa• Tenere il dispositivo lontano da temperature estremamente alte obasse• La batteria non si ricaricherà rapidamente a temperature inferiori a
10°C (50°F) o superioria a 40°C (104°F). La batteria non si ricaricherà affatto a temperature inferiori a 0°C (32°F) e superiori a
45°C (113°F)• Non esporre la batteria ai liquidi• Non usare il dispositivo se si notano danni alla batteria• Non smontare né modificare la batteria• Maneggiare e smaltire la batteria secondo le normative locali• Se l’operazione di ricarica non viene completata, anche se il
tempo di ricarica specifico è trascorso, interrompere immediatamente la ricarica• Non lasciare la batteria incustodita durante la ricarica• Scollegare il caricatore quando la batteria è completamente carica• L’uso o lo smaltimento scorretti delle batterie agli ioni di litio possono causare incendi• Non destinato all’uso in ambienti infiammabili
WARNING
Questo simbolo richiama l’attenzione dell’utente sulla presenzadi importanti istruzioni per il funzionamento e la manutenzione
all’interno della documentazione allegata.

34

Table 1: Tabella dei simboli
Descrizione
Carica batteria 75–100%
Carica batteria 50–74%
Carica batteria 30–49%
Carica batteria 10–29%
Carica batteria <10%
La batteria sta caricando
Il volume è impostato su 100% (predefinito)
Il volume è impostato su 50%
Il volume è disattivato
Il sensore CO2 è installato
Indica la modalità Ricerca delle nubi
Indica la modalità Localizzazione
Indica la modalità Azzeramento manuale
Sensibilità = Super
Sensibilità = Alta
Sensibilità = Media
Sensibilità = Bassa
Errore batteria
La batteria è sopra o sotto l’intervallo di temperatura
ideale e potrebbe non caricarsi correttamente
Si è verificato un errore del sensore

35

Italiano

Simbolo电源按钮 （图
中不可见）
耳机插孔
模式按钮
音量按钮

灵敏度 / 归零按钮

Micro USB
电源输入
（图中不可
见）

弹出按钮 （图
中不可见）
电池盖

36

??

中文

探针头 / 过滤器

1.

打开 D-TEK Stratus，以备使用a. 长按电源按钮，以打开或关闭 D-TEK Stratus.

备注 : 如果屏幕未打开，则表示电池电量非常低，需要充电 .
D-TEK Stratus 充电时可以使用 .
b. D-TEK Stratus 将开始预热 45-90 秒. 预热完成后，D-TEK
Stratus 即可供使用.
c. 要切换模式，请按下模式按钮. 仪器将在云采集、定位和手动归
零模式之间切换.

备注 : D-TEK Stratus 启动时始终是上次使用的模式 .

2.

使用云采集（便携式监视器）模式

中文

a. 缓慢穿过可疑区域，观察 ppm 读数.
b. 参照 ppm 读数找到冷媒浓度较高的区域. 读数越高，冷媒浓度越
高，则您距离泄漏源越近.

??

备注 : 在冷媒区域停留数分钟 ppm 读数可能会变小 . 如出现这种情况，前往冷媒浓度较低的区域停留数秒 .
c. 按下灵敏度/归零按钮可启用/禁用最大值功能. 启用后，观察得
到的最高 ppm 水平将显示在主 ppm 显示屏下方. 要重置最大读数，长按灵敏度/归零按钮.

备注 : 云采集模式下没有灵敏度设置 .

3.

使用定位模式a. 将 D-TEK Stratus 的探针头尽可能靠近疑似泄漏点. 请勿堵塞气流.
b. 探针头以（2.5-5 cm[1-2 in.]/秒）的速度缓慢经过各个可能的泄漏点. 如果检测到泄漏，D-TEK Stratus 将发出警报，屏幕指
示灯将亮起.

备注 : 在定位模式下， D-TEK Stratus 将在数秒后根据背景冷媒
自动归零 . 如出现这种情况，请继续查找较高的冷媒浓
度，或前往冷媒浓度较低区域以重置零点 .
c. 确定泄漏点后，将探针从泄漏点移开数秒，然后再次回到疑似泄
漏点以验证泄漏点.

37备注 : 对于较大的泄漏点或 SAE 应用，按下灵敏度 / 归零按钮，切换为灵敏度模式 . 当前灵敏度显示在屏幕上 .
备注 : 超高灵敏度设置不适用于 SAE 应用 .

4.

手动归零模式
手动归零模式与定位模式的工作方式类似，但用户在前者模式下可通过按下灵敏度/归零按钮，根据背景冷媒手动将仪器归零.

备注 : 手动归零模式下没有灵敏度设置 .

5.

拆卸和安装锂离子电池电池释放片a. 按下弹出按钮，打开电池盖.
b. 将电池释放片移动到侧面，直至电池开始弹出.
c. 滑出电池.
d. 将新电池与导轨对齐.
e. 沿导轨轻轻按压电池，直至卡合住电池释放片.

备注 : 请勿对电池使用蛮力 . 如果无法自由滑动，检查对齐情况后重试 .

6.

更换和安装传感器
弹出按钮传感器释放片a. 按下弹出按钮，打开电池盖.

38

??

中文

弹出按钮

b. 抓住传感器释放片，然后轻轻地拉出传感器.
c. 将新传感器与导轨对齐.
d. 沿着导轨轻轻地推动传感器，直至嵌入到位.

备注 : 请勿对传感器使用蛮力 . 如果无法自由滑动，检查对齐情况后重试 .
e. 重新装上电池盖.

7.

更换过滤器
a. 旋松探头针并移除过滤器.
b. 将新过滤器插入探针头.

??

中文c. 旋上探针头.

8.

充电与电池信息为 D-TEK Stratus 充电时，请将自带的墙面或车载充电器插入适当的电源中，然后将 Micro USB 电缆插入 D-TEK Stratus Micro USB
电源输入端。屏幕上的电池符号显示电量百分比，表示电池正在充
电。
备注 : D-TEK Stratus 充电时可以使用。

备注 : 新的锂离子电池电量只有 30% 或更少。建议电池充满后再使用。电池充满后可使用 3 小时左右。
注意：• 仅使用输出为 5 V(dc)、1 A• 的认证充电器/充电线温度和湿度:• 存储温度：-20 至 +60°C（-4 至 +140°F）• 工作温度：-20 至 +50°C（-4 至 +122°F）• 充电温度：0 至 45°C（32 至 113°F）• 湿度：95% 相对湿度 NC 最大值• 让设备远离极高温或极低温场所.
• 电池在 10°C (50°F) 以下或 40°C (104°F) 以上无法快速充. 电池
在 0°C (32°F) 以下以及 45°C (113°F) 以上无法充电• 请勿将电池暴露在液体中.
• 如发现电池损坏，请勿使用设备.
• 请勿拆卸或改装电池.
• 根据当地法规操作和处理电池.

39

• 如果充电操作无法完成，甚至超过规定的充电时间仍未完成，请立
即停止充电.
• 电池充电时请小心看管.
• 电池充满电后，拔掉充电器.
• 不适用于易燃环境.

??

中文WARNING此标识用于提示用户本仪器随附的文献中出现的重要操作和维护（维修）说明。.

40

Table 1: 符号表符号说明电池充电 75–100%电池充电 50–74%电池充电 30–49%电池充电 10–29%
电池充电 <10%中文电池正在充电
音量设为 100%（默认值）音量设为 50%??

音量设为静音CO2 传感器已安装
指示云采集模式
指示定位模式
指示手动归零模式
灵敏度 = 超高灵敏度 = 高灵敏度 = 中等灵敏度 = 低电池故障电池高于或低于理想温度范围，可能无法正确充电传感器发生故障41

スニファチップ /日本語

フ ィ ルタ

電源ボタン(裏側)
ヘッドフォンジャック

MODE (モード)ボタン

Volume（ボ
リュー
ム）ボタ
ン

SENS/ZERO（感度/ゼロ点調整）ボタン

Micro USB 充
電口（裏側）

イジェクトボタン (裏側)
バッテリー
カバー

42

1. D-TEKをオンにし、使用の準備をする
a. D-TEK Stratusをオンまたはオフにするには、電源ボタンを長押しします. 注意

注意 :画面がオンにならない場合は、バッテリーが非常に低いため、充電が必要です .D-TEK Stratus は充電中も使用できます .
b. D-TEK Stratusは45～90秒間のウォームアップを行います.
ウォームアップ完了後、D-TEK Stratusの使用が可能になります.
c. モードを切り替えるには、MODE ボタンを押します. Cloud

日本語

Hunting (ポータブルモニター)、Pinpoint(特定)およびManual Zero (手動ゼロ点調整)モードの間で切り替えられます.

注意 :D-TEK Stratus は常に、前回使用したモードで起動します .

2. Cloud Hunting（ポータブルモニター）モードを使用する
a. 冷媒リークが疑われるエリアをゆっくりと移動しながら、ppm
読み値を確認します.
b. ppm 読み値に基づいて、冷媒濃度が高くなっている場所を見つけます. 数値が高いほど、冷媒濃度が高くなるため、リーク箇所に近づきます.

注意 :冷媒が充満している場所で数分間検知機を置くと、ppm 読み値が低下する可能性があります . この場合、冷媒濃度が低い場所へ移動し、数秒間置いてください .
c. SENS/ZERO(感知/ゼロ点)ボタンを押して、MAXピーク機能
を有効/無効にします. 有効にした場合、測定された最大ppm
読み値がメインppmディスプレイの下に表示されます. ピーク
読み値をリセットするためには、SENS/ZERO(感知/ゼロ点)
ボタンを長押しします.

43

注意 :

Cloud Hunting( ポータブルモニター ) モードでは、感度の設定がありません .

3. Pinpoint(特定)モードを使用する
a. D-TEK Stratusのスニファチップをリークが疑われる箇所にできるだけ近づけます. 空気の流れをブロックしないでください.
b. リークが疑われる箇所に対して、プローブをゆっくりと (2.5 ～D-TEK Stratusはアラーム音を鳴らし、インジケータが点灯します. （改行して）注意：Pinpoint(特定)モードでは、D-TEK
Stratusは数秒後、バックグラウンドの冷媒に対して自動的にゼロになります. この場合、より高い濃度の冷媒を探し続けるか、低い濃度の領域に移動してゼロ点をリセットします.
c. リーク箇所が特定されたら、プローブをリーク箇所から離し数秒間待ってから、再度リークが疑われる箇所に戻り、リークを確認します.

注意 :注意 :

大漏れまたは SAE アプリケーションでは、SENS ( 感知 )/ZERO( ゼロ点 ) ボタンを押して、感度モードを切り替えます . 現在の感度は画面に表示されます .
SAE アプリケーションでは、Super 感度を使用しな
いでください .

4. Manual Zero（手動ゼロ点調整）モード
Manual Zero（手動ゼロ点調整）モードは、Pinpoint(特定)モードに類似する方法で動作しますが、SENS/ZERO（感知/ゼロ点調整）ボタンを押すと、手動でバックグラウンドの冷媒をゼロにすることができます.

注意 :Manual Zero（手動ゼロ点調整）モードでは感度の設定がありません .

44

日本語

5 cm [1 ～2 in.] /秒）通過させます. リークが検出されると、

5.

リチウムイオンバッテリーの取り外しと取り付け
バッテリーリリース
パッテリー
リリース
（見にくいため、上記のように改行）

日本語

イジェクタ
ボタン
（見にくいため 上記のように改行）

a. イジェクトボタンを押して、バッテリーカバーを取り外します.
b. バッテリーが外へ動き始めるまで、バッテリーリリースタブを横に動かします.
c. バッテリーを外にスライドさせ、取り出します.
d. 新しいバッテリーをレールに合わせます.
e. バッテリーリリースタブがかみ合うまで、ゆっくりバッテリーをレールに沿って入れます.

注意 :

6.

バッテリーを無理に押 し 込ま ないで く だ さ い . スムー
ズにスライドしない場合は、位置合わせ具合を確認
し、もう一度試してください .

センサーを交換し、取り付ける
イジェクタボタン

センサーリリースタブ

45

a. イジェクタボタンを押して、バッテリーカバーを取り外します.
b. センサーリリースタブをつかみ、センサーを外側にゆっくり引き出します.
c. 新しいセンサーをレールに合わせます.
d. カチッとかみ合うまでレールに沿ってセンサーをゆっくり入れ
ます.

注意 :

センサーを無理に押し込まないで下さい . スムーズにスライドしない場合は、位置合わせ具合を確認し、もう一度試してください .
e. バッテリーカバーを取り付け直します.

7.

フィルタを交換する
b. 新しいフィルタをプローブチップに取付けます.
c. プローブチップのネジを締めます.

8.

充電とバッテリーに関する情報
D-TEK Stratusを充電するためには、指定された壁または車載充電器を適切な電源に接続し、micro USBケーブルをD-TEK Stratusのmicro USB電源に挿し込んでください。画面上のバッテリー記
号に充電率(%)が表示され、バッテリーが充電中であることが表示されます。

注意 :注意 :D-TEK Stratus は充電中に操作できます。
新しいリチウムイン電池は 30% 以下のみ充電されています。使用前に完全に充電することをお勧めします。完全に充電するには約 3 時間かかります。

46

日本語

a. スニファチップのネジを外し、フィルターを取り外します.

日本語

注意:
• 5 V（DC）の出力を持つ認定された充電器/コー• ドのみを使用, 1アンペア 温度範囲と湿度：• 保管: -20 ～ +60°C (-4 ～ +140°F)• 操作-20 ～ +50°C (-4 ～ +122°F)• 充電: 0 ～ 45°C (32 ～ 113°F)• 湿度: 95% RH NC最大
• ドットサイズを統一する極端な高温または低温の場所にデバイスを置かないでください• バッテリーは10°C (50°F) 以下または 40°C (104°F) 以上では高速充電されません. バッテリーは0°C (32°F) 以下または 45°C
(113°F) 以上では全く充電されません• バッテリーを液体に曝さないでください• バッテリーの損傷がある場合には、デバイスを使用しないでください• バッテリーを解体または改造しないでください• バッテリーの取り扱いや廃棄については、地域の規則に従ってください• 規定された充電時間が経過しても、充電できない場合、直ちに再充電を停止してください• 充電中にバッテリーを放置しないでください• バッテリーが完全に充電された時点で、充電器を外してください• リチウムイオンバッテリーの不適切な使用や廃棄により、火災が発生する場合があります• 引火性の強い場所で使用しないでください
WARNING
この記号はこの装置に同梱された資料に重要な動作と保守（サービス）手順が存在することを、ユーザーに警告するために使用されます。

47

Table 1: 記号表
説明
バッテリー充電 75～100%
バッテリー充電 50～74%
バッテリー充電 30～49%
バッテリー充電 10～29%
バッテリー充電 <10%
バッテリーが充電中です
ボリュームが100%に設定されます（デフォルト）
ボリュームが50%に設定されます
ボリュームがミュートされます
CO2 センサーが取り付けられます
Cloud Hunting (ポータブルモニター)モードを示
します
Pinpoint (特定)モードを示します
Manual Zero （手動ゼロ点調整）モードを示します
感度 = Super(スーパー)
感度 = High(高)感度 = Medium(中)感度 = Low(低)バッテリーエラー
バッテリーは使用温度範囲を超えた場合で充電できません
センサーエラーが発生しました

48

日本語

記号

Кнопка
питания (невиднаГнездо длянаушников

Кнопка MODE(РЕЖИМ)Кнопка громкости

КнопкаSENS/ZERO(ЧУВСТВ./НОЛЬ)Кнопка высвобождения
крышки (невидна)

Разъем
питания
Micro-USB(не виден)Крышка
аккумул
яторного
отсека
49Русский

Наконечник
зонда/фильтр

1. Настройка устройства D-TEK Stratus и подготовка егок использованию
a. Чтобы включить или выключить устройство D-TEK Stratus,нажмите и удерживайте кнопку питания.

заметка:Русский

Если экран не включается, значит, аккумулятор
почти разряжен, и его необходимо зарядить.
Устройство D-TEK Stratus можно использовать вовремя зарядки.
b. Начнется прогрев устройства D-TEK Stratus, который
продлится от 45 до 90 секунд. По завершении прогрева
устройство D-TEK будет готово к использованию.
c. Для переключения режима нажмите кнопку MODE(РЕЖИМ). Ее нажатие обеспечивает переключение между
режимами поиска облака, точного определения иобнуления вручную.

заметка:Устройство D-TEK Stratus всегда включается втом режиме, который использовался последним.

2. Использование режима поиска облака (портативный
монитор)a. Медленно исследуйте подозрительные участки, следя запоказателем ppm (частей на млн).
b. Отслеживайте показатель ppm (частей на млн), чтобы
отыскать участки с наибольшей концентрацией хладагента.
Чем выше показатель, тем выше концентрация хладагентаи ближе место утечки.

заметка:Нахождение устройства вблизи участка схладагентом в течение нескольких минут может
привести к снижению показателя. Если такое
произойдет, переместите устройство на несколько
секунд в зону с его меньшей концентрацией.
c. Нажмите кнопку SENS/ZERO (ЧУВСТВ./НОЛЬ), чтобы
включить/отключить функцию MAX (МАКС.). Когда
включена эта функция, под основным показателем

50

количества частей на миллион отображается самое высокоеиз полученных значений показателя. Для сброса
максимального значения нажмите и удерживайте кнопкуSENS/ZERO (ЧУВСТВ./НОЛЬ).

заметка:Для режима поиска облака настройка
чувствительности не предусмотрена.

3. Использование режима точного определения

заметка:В режиме точного определения
автоматическое обнуление показаний устройства
D-TEK Stratus до фонового уровня происходит
через несколько секунд. Когда такое произойдет,продолжите поиск участка с более высокой
концентрацией хладагента или переместите
устройство в зону с его менее высокой
концентрацией, чтобы выполнить сброс нулевой
точки.
c. Обнаружив утечку, отведите зонд от зоны утечки нанесколько секунд, а затем верните его в точку
предполагаемой утечки, чтобы подтвердить ее наличие.

заметка:Перед определением крупных утечек илипроведением работ по SAE нажмите кнопкуSENS/ZERO (ЧУВСТВ./НОЛЬ), чтобы переключить
режим чувствительности. Текущий уровень
чувствительности отображается на экране.

заметка:Не используйте при проведении работ по SAEсверхвысокую чувствительность.

51Русский

a. Поместите наконечник зонда устройства D-TEK Stratus какможно ближе к месту предполагаемой утечки. Незакрывайте поток воздуха.
b. Медленно (со скоростью 2,5–5 см/с [1–2 дюйма/сек])проведите зондом над каждой вероятной точкой утечки. Приобнаружении утечки устройство D-TEK Stratus выдаст
звуковой сигнал и загорятся индикаторы на экране.

4. Режим обнуления вручнуюВ режиме обнуления вручную устройство работает так же, каки в режиме точного определения, однако пользователь имеет
возможность вручную обнулить его до фонового уровня хладагента, нажав кнопку SENS/ZERO (ЧУВСТВ./НОЛЬ).

заметка:Для режима обнуления вручную настройка
чувствительности не предусмотрена.

5. Извлечение и установка литий-ионного аккумулятора

Русский

Кнопка
высвобождения
крышки
Фиксатор
аккумулятора
a. Нажмите кнопку высвобождения крышки аккумуляторного
отсека.
b. Отведите в сторону фиксатор, чтобы аккумулятор
выдвинулся.
c. Извлеките аккумулятор.
d. Выровняйте новый аккумулятор по направляющим.
e. Аккуратно вставьте аккумулятор по направляющим досрабатывания фиксатора.

заметка:Не давите на аккумулятор слишком сильно.
Если он не вставляется, выровняйте его иповторите попытку.

6. Замена и установка датчика
Кнопка высвобождения крышки

Фиксатор датчика

52

a. Нажмите кнопку высвобождения крышки аккумуляторного
отсека.
b. Возьмитесь за фиксатор датчика и аккуратно извлеките
датчик.
c. Выровняйте новый датчик по направляющим.
d. Аккуратно вставьте датчик по направляющим до щелчка.

заметка:Не давите на датчик слишком сильно. Если онне вставляется, выровняйте его и повторите
попытку.
e. Верните крышку батарейного отсека на место.

7. Замена фильтров

8. Зарядка и аккумулятор
Чтобы зарядить устройство D-TEK Stratus, подключите одинконец входящего в комплект поставки сетевого или автомобильного зарядного устройства к подходящему источнику питания, а разъем micro USB — к разъему питания micro USBустройства D-TEK Stratus. Индикатор зарядки на экране указывает, что производится зарядка аккумулятора, а также показывает уровень заряда в процентах.

заметка:Устройство D-TEK Stratus можно использоватьво время зарядки.

заметка:Уровень заряда нового литий-ионного
аккумулятора составляет 30% или меньше.
Рекомендуется полностью зарядить аккумулятор
перед использованием устройства. Полная зарядка
занимает около 3 часов.

53Русский

a. Открутите наконечник зонда и извлеките фильтр.
b. Вставьте в наконечник новый фильтр.
c. Прикрутите наконечник зонда.

Русский

Предостережения:• Используйте только сертифицированное зарядное устройство/кабель с выходом 5 В (пост. тока), 1 А• Температурные диапазоны и влажность:• Хранение: от -20 до +60 °C (от -4 до +140 °F)• Эксплуатация: от -20 до +50 °C (от -4 до +122 °F)• Зарядка: от 0 до 45 °C (от 32 до 113 °F)• Влажность: макс. 95% относительной влажности безконденсации
• Не используйте устройство в местах со сверхвысокой илисверхнизкой температурой
• При температурах ниже 10°C (50°F) или выше 40°C (104°F)быстрая зарядка аккумулятора невозможна.При температурах
ниже 0°C (32°F) и выше 45°C (113°F) он не заряжается вовсе.
• Не подвергайте аккумулятор воздействию влаги
• Не используйте устройство, если заметите, что аккумулятор
поврежден
• Не разбирайте аккумулятор и не пытайтесь изменить его конструкцию
• Транспортировка и утилизация аккумулятора должны осуществляться в соответствии с местными нормативными требованиями
• Если окажется, что зарядка не завершилась даже по истечении указанного времени, ее необходимо будет немедленно
прекратить
• Во время зарядки не оставляйте аккумулятор без присмотра
• После полной зарядки аккумулятора отключите зарядное
устройство
• Использование или утилизация литий-ионных аккумуляторов
ненадлежащим образом может стать причиной пожара
• Не предназначено для использования в огнеопасных средахWARNINGДанный символ используется для предупреждения
пользователя о наличии важных инструкций поэксплуатации и техническому обслуживанию (ремонту) вдокументации, прилагаемой к данному прибору.

54

Table 1: Таблица условных символов
Описание
Заряд аккумулятора 75–100%Заряд аккумулятора 50–74%Заряд аккумулятора 30–49%Заряд аккумулятора 10–29%
Заряд аккумулятора <10%Аккумулятор заряжается
Уровень громкости — 100 % (по умолчанию)Уровень громкости — 50 %Звук отключен
Установлен датчик CO2Указатель режима поиска облака
Указатель режима точного определения
Указатель режима обнуления вручную
Чувствительность — сверхвысокая
Чувствительность — высокая
Чувствительность — средняя
Чувствительность — низкая
Сбой аккумулятора
Температура аккумулятора выше или ниже оптимальной, и, возможно, он не заряжается должным
образом
Сбой датчика

55Русский

Символ

Końcówka
sondy/filtr

Przycisk zasilania(niewidoczny)

Przyciskregulacjigłośności

Przycisk
SENS/ZERO
Gniazdozasilania
mikro USB
(niewidoczne)

Przycisk zwolnienia(niewidoczny)
Pokrywabaterii
56

PolskiGniazdo słuchawkowe

Przyciskzmianytrybu
MODE

1. Włączanie i przygotowanie do użytku urządzenia D-TEK
Stratusa. Naciśnij i przytrzymaj przycisk zasilania, aby włączyć lubwyłączyć urządzenie D-TEK Stratus.

Uwaga: Jeśli ekran nie włącza się, oznacza to, że poziom
naładowania baterii jest bardzo niski i trzeba jąnaładować. Urządzenia D-TEK Stratus można
używać podczas ładowania.
b. Uruchamianie urządzenia D-TEK Stratus trwa od 45 do 90
sekund. Po zakończeniu procedury uruchamiania urządzenie
D-TEK Stratus jest gotowe do użytku.
c. Przełączanie pomiędzy trybami odbywa się za pomocą
przycisku MODE. Przycisk ten umożliwia zmianę między trybami
Cloud Hunting, Pinpoint i Manual Zero.

Uwaga: Urządzenie D-TEK Stratus uruchamia się w trybie, w
Polskiktórym było ostatnio używane.

2. Używanie trybu Cloud Hunting (przenośnego monitora)a. Powoli przesuń urządzenie po obszarach, co do których istnieje
podejrzenie wycieku, i obserwuj odczyt ppm.
b. Obserwuj odczyt ppm, aby znaleźć obszary wyższego stężenia
chłodziwa. Im wyższa liczba, tym wyższe stężenie chłodziwa itym bliżej jesteś źródła wycieku.

Uwaga: Utrzymywanie urządzenia w danym obszarze z
chłodziwem przez kilka minut może spowodować
spadek wartości odczytu ppm. Jeśli to nastąpi,przesuń urządzenie na kilka sekund w obszar
niższego stężenia.
c. Naciśnij przycisk SENS/ZERO, aby włączyć/wyłączyć funkcję
MAX. Kiedy jest włączona, najwyższy zaobserwowany poziom
ppm wyświetla się poniżej głównego wyświetlacza ppm. Aby
zresetować maksymalny odczyt, naciśnij i przytrzymaj przycisk
SENS/ZERO.

57

Uwaga: W trybie Cloud Hunting ustawienie czułości jest
niedostępne.

3. Używanie trybu Pinpointa. Umieść końcówkę D-TEK Stratus jak najbliżej miejsca
domniemanego wycieku. Nie tamuj przepływu powietrza.
b. Powoli (2,5 do 5 cm [1 do 2 cali] na sekundę) przesuwaj sondę
przez kolejne domniemane punkty wycieku. Jeśli zostanie
wykryty wyciek, urządzenie D-TEK Stratus uruchomi alarm izaświecą się wskaźniki na ekranie.

Uwaga: W trybie Pinpoint po upływie kilku sekund urządzenie

Uwaga: W przypadku dużych wycieków lub stosowaniaurządzenia w zastosowaniach SAE, wciśnij przycisk
SENS/ZERO, aby włączyć tryb czułości. Aktualna
czułość wyświetli się na ekranie.
Uwaga: W przypadku zastosowań SAE nie stosuj trybu
czułości Super.

4. Tryb Manual Zero
Tryb Manual Zero działa podobnie do trybu Pinpoint, pozwala jednak użytkownikowi ręcznie zerować urządzenie do wartości chłodziwa w tle, poprzez naciśnięcie przycisku SENS/ZERO.

Uwaga: W trybie Manual Zero ustawienie czułości jest
niedostępne.

58

Polski

D-TEK Stratus automatycznie wyzeruje się dopoziomu chłodziwa w tle. Jeśli to nastąpi, wyszukajwyższe stężenie chłodziwa lub przejdź do obszaruniższego stężenia, aby zresetować punkt zerowy.
c. Po wykryciu wycieku odsuń sondę na kilka sekund od wycieku,
a następnie umieść ją ponownie w miejscu domniemanego
wycieku, aby go zweryfikować.

5. Wyjmowanie i instalowanie baterii litowo-jonowej
Przyciskzwolnienia

Zwolnienie bateriia. Naciśnij przycisk zwolnienia, aby zdjąć pokrywę baterii.
b. Przesuń element zwalniający baterii na bok, aż bateria zaczniesię wysuwać.
c. Wysuń baterię.
d. Włóż nową baterię, dopasowując ją do prowadnic.
e. Delikatnie dociśnij baterię do prowadnic, aż zaskoczy element
zwalniający baterii.

PolskiUwaga: Nie wkładaj baterii na siłę. Jeśli bateria nie wsuwa
się swobodnie, sprawdź dopasowanie i spróbuj
ponownie.

6. Wymiana i instalacja czujnika
Przycisk zwolnienia

Element zwalniający czujnik
a. Naciśnij przycisk zwolnienia, aby zdjąć pokrywę baterii.
b. Chwyć za element zwalniający czujnika i delikatnie wyciągnij
czujnik.
c. Włóż nowy czujnik, dopasowując go do prowadnic.
d. Delikatnie dociśnij czujnik do prowadnic, aż wskoczy na miejsce.

59

Uwaga: Nie wkładaj czujnika na siłę. Jeśli czujnik nie wsuwa się
swobodnie, sprawdź dopasowanie i spróbuj ponownie.
e. Załóż ponownie pokrywę baterii.

7. Wymiana filtrów
a. Odkręć końcówkę sondy i wyjmij filtr.
b. Włóż nowy filtr do końcówki sondy.
c. Przykręć końcówkę sondy.

8. Informacje dotyczące ładowania i baterii
Aby naładować urządzenie D-TEK Stratus, podłącz dołączoną doniego ładowarkę ścienną lub samochodową do odpowiedniego źródła zasilania, a następnie podłącz kabel micro USB do wejścia zasilania micro USB w urządzeniu D-TEK Stratus. Symbol baterii na
ekranie pokazuje procent naładowania i oznacza, że bateria jest w
trakcie ładowania.

Uwaga: Urządzenie D-TEK Stratus może być używane

wynosi 30% lub mniej. Przed użyciem zaleca się
naładowanie baterii do pełna. W celu naładowania
baterii do pełna urządzenie należy ładować przez
około 3 godziny.

60

Polskipodczas ładowania.

Uwaga: Stopień naładowania nowej baterii litowo-jonowej

PolskiUwaga:• Używać wyłącznie certyfikowanej ładowarki/przewodu o parametrach wyjściowych 5V (DC), 1A• Zakres temperatur i wilgotność:• Przechowywanie: -20 do +60°C (-4 do +140°F)• Praca: -20 do +50°C (-4 do +122°F)• Ładowanie: 0 do 45°C (32 do 113°F)• Wilgotność: Maks. 95% RH w warunkach normalnych• Trzymaj urządzenie z dala od miejsc o bardzo wysokiej lub bardzo
niskiej temperaturze• Bateria nie będzie szybko się ładować w temperaturach poniżej
10°C (50°F) lub powyżej 40°C (104°F). Bateria nie będzie w ogólesię ładować w temperaturach poniżej 0°C (32°F) i powyżej 45°C
(113°F)• Nie wystawiaj baterii na kontakt z płynem
• Nie używaj urządzenia, jeśli zauważysz jakiekolwiek uszkodzeniebaterii• Nie rozmontowuj baterii ani nie modyfikuj jej• Używaj baterii i utylizuj je zgodnie z lokalnymi przepisami• Jeśli nie uda się dokończyć ładowania, nawet jeśli minął określony
czas ładowania, natychmiast zaprzestań dalszego ładowania• Nie zostawiaj baterii bez nadzoru podczas ładowania• Po całkowitym naładowaniu baterii, wyłącz ładowarkę z gniazdka• Nieprawidłowe użytkowanie lub utylizowanie baterii litowo-jonowych może prowadzić do powstania pożaru
• Produkt nie nadaje się do stosowania w środowisku łatwopalnym
WARNING
Ten symbol informuje użytkownika o obecności ważnych
wskazówek dotyczących działania i konserwacji (serwisowania),
znajdujących się w dokumentacji dołączonej do przyrządu.

61

Table 1: Tabela symboli
Opis
Naładowanie baterii 75–100%
Naładowanie baterii 50–74%
Naładowanie baterii 30–49%
Naładowanie baterii 10–29%
Naładowanie baterii <10%
Bateria jest w trakcie ładowaniaGłośność jest ustawiona na 100% (domyślnie)Głośność jest ustawiona na 50%Dźwięk jest wyciszony
Czujnik CO2 został zainstalowany
Wskazuje tryb Cloud Hunting
Wskazuje tryb Pinpoint
Wskazuje tryb Manual ZeroCzułość = Super
Czułość = Wysoka
Czułość = Średnia
Czułość = NiskaBłąd baterii
Bateria znajduje się w stanie powyżej lub poniżej idealnej temperatury i może nie ładować się prawidłowoWystąpił błąd czujnika

62

Polski

SymbolSondspets/filterStrömknapp(inte synlig)

Volymknapp

SENS/ZEROknapp

Micro
USB-ströminmatning(inte synlig)

Utmatningsknapp(inte synlig)
Batterilucka

63

Svenska

Uttag för
hörlurar

MODE-knapp

1. Sätta på och förbereda D-TEK Stratus för användninga. Håll strömknappen nedtryckt för att slå på eller av D-TEK Stratus.

NOTERA: Om skärmen inte aktiveras är batteriet nästan helt
urladdat och måste laddas. D-TEK Stratus kan
användas under laddning.
b. D-TEK Stratus påbörjar uppvärmning under 45 till 90 sekunder.
När uppvärmningen är klar är D-TEK Stratus redo att användas.
c. Byt läge genom att trycka på knappen MODE. Detta växlar
mellan lägena Cloud Hunting, Pinpoint och Manual Zero.

NOTERA: D-TEK Stratus startas alltid i det läge som senast
användes.

Svenska

2. Använda Cloud Hunting-läget (bärbar monitor)
a. Förflytta dig långsamt genom misstänkta områden och håll ett
öga på ppm-avläsningen.
b. Följ ppm-avläsningen för att hitta områden med högre
köldmedelskoncentration. Ju högre värde desto högre är
köldmedelskoncentrationen och ju närmare befinner du dig
läckagekällan.

NOTERA: När du stannar kvar i ett område med köldmedel
under flera minuter kan det leda till att
ppm-avläsningen sjunker. Om detta sker ska duförflytta dig till ett område med lägre koncentration
under ett par sekunder.
c. Tryck på SENS/ZERO-knappen för att aktivera/inaktivera
MAX-funktionen. När den är aktiverad visas högsta ppm-nivå
som iakttagits nedanför den huvudsakliga ppm-skärmen.
Återställ maxavläsningen genom att hålla SENS/ZERO-knappen
nedtryckt en stund.

NOTERA: Det finns ingen känslighetsinställning i Cloud
Hunting-läget.

64

3. Använda Pinpoint-lägeta. Placera spetsen på D-TEK Stratus så nära det misstänkta
läckaget som möjligt. Blockera inte luftflödet.
b. För långsamt sonden (2,5-5 cm (1-2 tum) per sekund) över varje
möjligt läckage. Om ett läckage upptäcks larmar D-TEK Stratus
och indikatorerna på skärmen tänds.

NOTERA: I Pinpoint-läget nollställs D-TEK Stratus
automatiskt till bakgrundsköldmedlet efter ett par
sekunder. När detta sker, fortsätt antingen att titta
efter en högre köldmedelskoncentration eller
förflytta dig till ett område med lägre koncentration
för att återställa nollpunkten.
c. När ett läckage identifieras, avlägsna sonden från läckaget
under ett par sekunder och återvänd till det misstänkta läckaget
för att verifiera läckaget.

NOTERA: För större läckage eller för SAE-applikationer tryck på

NOTERA: Använd inte känslighetsinställningen Super för
SAE-applikationer.

4. Läget Manual Zero
Manual Zero-läget fungerar ungefär som Pinpoint-läget men ger
användaren möjlighet att nollställa instrumentet till köldmedlet i bakgrunden genom att trycka på SENS/ZERO-knappen.

NOTERA: Det finns ingen känslighetsinställning i ManualZero-läget.

65

Svenska

SENS/ZERO-knappen för att växla känslighetsläge.
Nuvarande känslighet visas på skärmen.

5. Avlägsna och installera litiumjonbatteriet
Utmatningsknapp

Batterifrigöring
a. Tryck på utmatningsknappen för att frigöra batteriluckan.
b. Flytta batterifrigöringsfliken åt sidan tills batteriet börjar matas ut.
c. Dra ut batteriet.
d. Rikta in det nya batteriet med spåren.
e. Tryck försiktigt in batteriet längs spåren tills
batterifrigöringsfliken aktiveras.

NOTERA: Tvinga inte batteriet. Om det inte glider utan
hinder, kontrollera inpassningen och försök igen.

Svenska

6. Byta ut och installera sensorn
Utmatningsknapp

Sensorfrigöringsflik
a. Tryck på utmatningsknappen för att frigöra batteriluckan.
b. Ta tag i sensorfrigöringsknappen och dra försiktigt ut sensorn.
c. Rikta in den nya sensorn med spåren.
d. Tryck försiktigt in sensorn längs spåren tills den klickar på plats.

NOTERA: Tvinga inte sensorn. Om den inte glider utan
hinder, kontrollera inpassningen och försök igen.
e. Sätt tillbaka batteriluckan.

66

7. Byta ut filter
a. Skruva av sondspetsen och avlägsna filtret.
b. Sätt i det nya filtret i sondspetsen.
c. Skruva på sondspetsen.

8. Information om laddning och batteri
För att ladda D-TEK Stratus, anslut den medföljande vägg- eller bilkontakten i en lämplig strömkälla och koppla in mikro-USB-sladdentill ingången för mikro-USB på din D-TEK Stratus. Batterisymbolen på
skärmen visar laddningen i procent och indikerar att batteriet laddas.

NOTERA: D-TEK Stratus kan användas medan den laddas.
NOTERA: Ett nytt litiumjonbatteri kommer att vara laddat till

Var försiktig:• Använd endast en certifierad laddare/sladd med effekt 5 V(likström), 1A• Temperaturomfång och luftfuktighet:• Förvaring: -20 till +60 °C• I drift: -20 till +50 °C• Vid laddning: 0 till 45°C
• Luftfuktighet: Max 95% relativ luftfuktighet, icke-kondenserande• Håll enheten borta från platser med extremt höga eller låga temperaturer• Batteriet kommer inte att snabbladdas vid temperaturer
under10°C (50°F) eller över 40°C (104°F). Det kommer inte att
laddas alls vid temperaturer under 0°C (32°F) och över 45°C
(113°F)• Utsätt inte batteriet för vätska
• Använd inte enheten om du lägger märke till någon skada på batteriet• Ta inte isär och modifiera inte batteriet• Hantera och kassera batteriet enligt lokala förordningar• Om laddningsåtgärden inte slutförs, även när den angivna laddningstiden har förflutit, avbryt omedelbart ytterligare laddning• Lämna inte batteriet utan tillsyn vid laddning• Koppla från laddaren när batteriet är fulladdat

67

Svenska

30 % eller mindre. Det rekommenderas att ladda
upp batteriet till fullo innan det används. Tillåt cirkatre (3) timmar tills batteriet är fulladdat.

• Oriktig användning eller kassering av litiumjonbatterier kan leda till
att de fattar eld• Ej avsedd för användning i brandfarliga miljöer

SvenskaWARNING
Denna symbol används för att varna användaren om viktiga
drifts- och skötselanvisningar (underhåll) i litteraturen som
medföljer detta instrument.

68

Table 1: Symboltabell
Beskrivning
Batteriladdning 75-100%
Batteriladdning 50–74%
Batteriladdning 30–49%
Batteriladdning 10–29%
Batteriladdning <10%
Batteriet laddas
Volymen är inställd till 100% (standard)
Volymen är inställd till 50%
Volymen är avstängd
CO2-sensorn är installerad
Indikerar Cloud Hunting-läge
Indikerar Pinpoint-läge
Indikerar Manual Zero-läge
Känslighet = Super
Känslighet = High
Känslighet = Medium
Känslighet = Low
Batterifel
Batteriet är över eller under idealiskt temperaturintervall och kanske inte kan laddas korrekt
Sensorfel har uppstått

69

Svenska

SymbolProb ucu/filtre

Güç düğmesi
(görünmez)
Kulaklık jakı

Ses seviyesidüğmesi

SENS/ZEROdüğmesi

Mikro USBgüç girişi
(görünmez)Çıkarma düğmesi
(görünmez)
Pil kapağı

70

TürkçeMODE düğmesi

1. D-TEK Stratus'un açılması ve kullanıma hazırlanması
a. D-TEK Stratus'u açmak veya kapatmak için güç düğmesineuzun basın.

NOT: Ekran açılmazsa şarj az demektir. Bu durumda, cihazınşarj edilmesi gerekir. D-TEK Stratus'u şarj ederken
kullanabilirsiniz.
b. D-TEK Stratus, 45 ila 90 saniyelik bir ısıtma yapacaktır. Isıtma
tamamlandığında D-TEK Stratus kullanıma hazırdır.
c. Mod değiştirmek için MODE düğmesine basın. Böylelikle, Cloud
Hunting, Pinpoint ve Manual Zero modları arasında geçiş
yapabilirsiniz.

NOT: D-TEK Stratus, her zaman için son kullanıldığı moddan
açılır.

2. Cloud Hunting (Taşınabilir Ekran) modunun kullanımı

Türkçea. Şüpheli alanlarda yavaşça hareket ettirerek ppm değerine bakın.
b. Ppm değerini izleyerek soğutucu konsantrasyonunun yüksek
olduğu alanları bulun. The higher the number, the higher theconcentration of refrigerant and the closer you are to the leaksource.

NOT: Soğutucunun olduğu alanda birkaç dakika beklemek ppm
değerinin düşmesine neden olabilir. Bu durumda, birkaç
saniyeliğine daha düşük konsantrasyonlu bir alana geçin.
c. SENS/ZERO düğmesine basarak MAX özelliğini açın/kapatın.
Bu özellik açıkken, gözlemlenen en yüksek ppm değeri ana ppm
göstergesinin altında görüntülenir. Maximum değeri sıfırlamak
için SENS/ZERO düğmesine uzun basın.

NOT: Cloud Hunting modunda hassasiyet ayarı yoktur.
3. Pinpoint modunun kullanımı
a. D-TEK Stratus'un ucunu, kaçak olduğundan şüphelendiğiniz
yere olabildiğince yakın konumlandırın. Hava akışını
engellemeyin.

71

b. Probu, her bir olası kaçak noktasından yavaşça (saniyede 2,5 ila
5 cm [1 ila 2 inç]) geçirin. Kaçak tespit edilirse D-TEK Stratus
alarm verir ve ekran göstergeleri yanar.

NOT: Pinpoint modunda, D-TEK Stratus, birkaç saniye sonra
otomatik olarak arka plana sıfırlanır. Bu durumda, dahayüksek soğutucu konsantrasyonu aramaya devam edere
ya da daha düşük konsantrasyonlu bir alana geçerek sıfır
noktasını sıfırlayın.
c. Kaçak tespit edildiğinde, probu kaçaktan birkaç saniyeliğine
çekin ve kaçak olduğundan şüphelendiğiniz alana geri dönerek
kaçak olduğunu doğrulayın.

NOT: Büyük kaçaklar veya SAE uygulamaları için,
SENS/ZERO düğmesine basarak hassasiyet modunu
etkinleştirin. Güncel hassasiyet ekranda görüntülenir.

NOT: SAE uygulamaları için Super hassasiyet modunu
kullanmayın.

4. Manual Zero (Manuel Sıfır) modu

NOT: Manual Zero modunda hassasiyet ayarı yoktur.
5. Lityum iyon pilin çıkarılması ve takılması
ÇıkarmadüğmesiPilçıkarmaa. Çıkarma düğmesine basarak pil kapağını çıkarın.
b. Pil çıkarma tırnağını, pil çıkmaya başlayana kadar yana hareket
ettirin.

72

TürkçeManual Zero modu, Pinpoint moduna benzer şekilde çalışmakla
beraber kullanıcının SENS/ZERO düğmesine basarak aleti ara plansoğutucusuna otomatik olarak sıfırlamasına olanak tanır.

c. Pili kaydırarak dışarı çıkarın.
d. Yeni pili raylarla hizalayın.
e. Pili, pil çıkarma tırnağı yerine oturana kadar raylar boyunca
hafifçe itin.

NOT: Pili zorlamayın. Kolayca kaymıyorsa hizasını kontrol edip
tekrar deneyin.

6. Sensörün çıkarılması ve takılması
Çıkarma düğmesi

Sensör çıkarma tırnağı
a. Çıkarma düğmesine basarak pil kapağını çıkarın.
b. Sensör çıkarma tırnağını tutup sensörü yavaşça dışarı çekin.
c. Yeni sensörü raylarla hizalayın.
d. Sensörü, yerine oturana kadar raylar boyunca hafifçe itin.

Türkçe

NOT: Sensörü zorlamayın. Kolayca kaymıyorsa hizasını
kontrol edip tekrar deneyin.
e. Pil kapağını yerine takın.

7. Filtrelerin değiştirilmesi
a. Prob ucunun vidasını açıp filtreyi çıkarın.
b. Yeni filtreyi prob ucuna takın.
c. Prob ucunu vidalayın.

73

8. Şarj ve Batarya Bilgisi

D-TEK Stratus’u şarj etmek için duvar veya araç şarj cihazını uygun
güç kaynağına takın ve mikro USB kablosunu da D-TEK Stratus
mikro USB güç girişine takın. Ekrandaki batarya sembolü, şarj yüzdesini gösterir ve bataryanın şarj olduğunu belirtir.
NOT: D-TEK Stratus, şarj sırasında çalışabilir.

NOT: Yeni lityum iyon batarya %30 veya daha az şarja sahip
Uyarılar:• Yalnızca 5 V (dc), 1 A çıkışlı onaylı bir şarj cihazı/kablo kullanın• Sıcaklık aralıkları ve nem:
• Saklama: -20 ila +60°C (-4 ila +140°F)• Çalışma: -20 ila +50°C (-4 ila +122°F)• Şarj: 0 ila 45°C (32 ila 113°F)• Nem: Maksimum %95 yoğuşmasız bağıl nem• Cihazı aşırı yüksek ya da düşük sıcaklık koşullarından uzak tutun• Pil, 10°C (50°F) altı veya 40°C (104°F) üzeri sıcaklıklarda hızlı şarjolmaz. 0°C (32°F) altı ve 45°C (113°F) üzeri sıcaklıklarda ise hiçşarj olmaz• Pili sıvılara maruz bırakmayın• Pilde hasar varsa cihazı kullanmayın• Pili sökmeyin ve pil üzerinde değişiklik yapmayın• Pili, yerel düzenlemeler uyarınca kullanın ve elden çıkarın• Belirtilen şarj süresi dolmasına rağmen şarj işlemi tamamlamazsaşarj etmeyi hemen bırakın• Şarj etme sırasında pili gözetimsiz bırakmayın• Pil tamamen dolduğunda şarj aletinin fişini çekin
• Lityum iyon pillerin uygunsuz şekilde kullanılması veya elden çıkarılması yangına neden olabilir• Yanıcı ortamlarda kullanılmamalıdır
WARNING
Bu cihaza ilişkin basılı materyallerde yer alan bu sembol,
kullanıcıyı önemli çalışma ve bakım (servis) talimatlarınınmevcudiyeti konusunda uyarmak için kullanılır

74

Türkçeolacaktır. Kullanımdan önce bataryayı tamamen şarjetmeniz önerilir. Tam şarj için yaklaşık 3 saat bekleyin.

Table 1: Sembol Tablosu
SembolAçıklama
Pil şarjı %75–100
Pil şarjı %50–74
Pil şarjı %30–49
Pil şarjı %10–29
Pil şarjı <%10
Pil şarj oluyor
Ses seviyesi %100 (varsayılan)Ses seviyesi %50
Ses kapalı
CO2 sensörü takılı
Cloud Hunting modunu belirtir

TürkçePinpoint modunu belirtir
Manual Zero modunu belirtir
Hassasiyet = Süper
Hassasiyet = Yüksek
Hassasiyet = Orta
Hassasiyet = Düşük
Pil hatası
Pil, ideal sıcaklık seviyesinin üstünde veya altında vedoğru şekilde şarj olmayabilir
Sensör hatası meydana geldi

75Uiteinde vansonde/filter

Aan/uit-knop(niet zichtbaar)
Koptelefoonstekker

MODE-knop

Volumeknop
Nederlands

SENS/ZEROknop

microUSBvoedingsingang (nietzichtbaar)

Uitwerpknop (nietzichtbaar)
Batterijklepje76

1. De D-TEK Stratus inschakelen en gereedmaken voor
gebruika. Druk de aan/uit-knop lang in om de D-TEK Stratus aan of uit tezetten.

NOTITIE: Als het scherm niet wordt ingeschakeld, is de batterij
bijna leeg en moet deze worden opgeladen. De
D-TEK Stratus kan worden gebruikt tijdens het
opladen.
b. De D-TEK Stratus wordt gedurende 45 tot 90 seconden
opgewarmd. Als het opwarmen is voltooid, is de D-TEK Stratus
gereed voor gebruik.
c. Als u van modus wilt veranderen, drukt u op de MODE-knop.
Hiermee wordt omgeschakeld tussen de modi Cloud Hunting,Pinpoint en Manual Zero.

NOTITIE: De D-TEK Stratus start altijd in de als laatst gebruikte
modus.

Nederlands

2. De modus Cloud Hunting (draagbare monitor) gebruiken
a. Beweeg langzaam langs de verdachte plekken en observeer de
ppm-waarde.
b. Volg de ppm-waarde om gebieden met een hogere concentratie
koelmiddelen te vinden. Hoe hoger de waarde, des te hoger dekoelmiddelconcentratie en des te dichter u in de buurt bent van
de bron van het lek.

NOTITIE: Als u meerdere minuten doorbrengt in een gebied
met koelmiddelen kan de ppm-waarde afnemen. Als
dat gebeurt, gaat u gedurende enkele seconden naar
een gebied met een lagere concentratie.
c. Druk op de SENS/ZERO-knop om de functie MAX in of uit teschakelen. Als deze is ingeschakeld, wordt het hoogste
ppm-gehalte dat is waargenomen weergegeven onder het
primaire ppm-display. U kunt de maximale waarde resetten door
de SENS/ZERO-knop lang ingedrukt te houden.

77

NOTITIE: Er is geen gevoeligheidsinstelling beschikbaar in demodus Cloud Hunting.

3. De modus Pinpoint gebruiken
a. Plaats het uiteinde van de D-TEK Stratus zo dicht mogelijk bij
het verdachte lek. Blokkeer de luchtstroom niet.
b. Beweeg de sonde langzaam (ongeveer 2,5 tot 5 cm perseconde) langs elk mogelijk lekpunt. Als een lek wordt
gevonden, geeft de D-TEK Stratus een alarm af en lichten deschermindicatoren op.

NOTITIE: In de modus Pinpoint stelt de D-TEK Stratus na
enkele seconden automatisch in op nul voor dekoelmiddelen op de achtergrond. Als dit gebeurt,
kunt u blijven zoeken naar een hogere concentratie
koelmiddel of naar een gebied met een lagere
concentratie gaan om het nulpunt te resetten.
c. Als een lek wordt gevonden, haalt u de sonde enkele seconden
lang weg van het lek en keert u vervolgens terug naar deverdachte plek om het lek te bevestigen.
op de SENS/ZERO-knop om de gevoeligheidsmodus
in te schakelen. De huidige gevoeligheid wordt
weergegeven op het scherm.

NOTITIE: Gebruik de gevoeligheidsinstelling Super niet voor
SAE-toepassingen.

4. Modus Manual Zero
De modus Manual Zero werkt op vergelijkbare wijze als de modus
Pinpoint, maar stelt de gebruiker in staat het instrument handmatig
op nul in te stellen voor de koelmiddelen op de achtergrond door op
de SENS/ZERO-knop te drukken.

NOTITIE: Er is geen gevoeligheidsinstelling beschikbaar in demodus Manual Zero.

78

Nederlands

NOTITIE: Voor grote lekken of voor SAE-toepassingen drukt u5. De lithium-ionbatterij verwijderen en plaatsen
Uitwerpknop

Batterijvergrendelinga. Druk op de uitwerpknop om het batterijklepje te ontgrendelen.
b. Beweeg het lipje van de batterijvergrendeling opzij totdat debatterij wordt uitgeworpen.
c. Schuif de batterij naar buiten.
d. Lijn de nieuwe batterij uit met de geleiders.
e. Druk de batterij voorzichtig langs de geleiders totdat het lipje vande batterijvergrendeling wordt geactiveerd.

NOTITIE: Forceer de batterij niet. Als deze niet gemakkelijk
naar binnen glijdt, controleert u de uitlijning enprobeert u het opnieuw.

6. De sensor vervangen en installeren

Nederlands

Uitwerpknop

Ontgrendelingslipje
voor sensor
a. Druk op de uitwerpknop om het batterijklepje te ontgrendelen.
b. Pak het ontgrendellipje voor de sensor vast en trek voorzichtig
de sensor naar buiten.
c. Lijn de nieuwe sensor uit met de geleiders.
d. Druk voorzichtig de sensor langs de geleiders totdat de sensor
vastklikt.

79

NOTITIE: Forceer de sensor niet. Als deze niet gemakkelijk
naar binnen glijdt, controleert u de uitlijning enprobeert u het opnieuw.
e. Plaats het batterijklepje terug.

7. Filters vervangena. Schroef het uiteinde van de sonde los en verwijder het filter.
b. Plaats het nieuwe filter in het uiteinde van de sonde.
c. Schroef het uiteinde van de sonde weer vast.

8. Informatie over opladen en de accu
Voor het opladen van de D-TEK Stratus, steekt u de meegeleverde
stekker in het stopcontact of de autolader in de stroomvoorziening
en steekt u de micro-USB-kabel in de micro-USB-ingang van de
D-TEK Stratus. Het accusymbool op het scherm toont het oplaadpercentage en geeft aan dat de accu wordt opgeladen.

NOTITIE: D-TEK Stratus kan gedurende het opladen gewoon
gebruikt worden.

NOTITIE: Een nieuwe lithium-ion-accu heeft een lading van

Nederlands

30% of minder. Het wordt aanbevolen om de accu
alvorens gebruik eerst volledig op te laden. Hetvolledig opladen neemt ongeveer 3 uur in beslag.

80Nederlands

Waarschuwingen:• Gebruik uitsluitend een gecertificeerd(e) snoer/lader met een uitvoer van 5 V (gelijkspanning), 1 A• Temperatuurbereiken en luchtvochtigheid:• Opslag: -20 tot +60 °C (-4 tot +140 °F)• Gebruik: -20 tot +50 °C (-4 tot +122 °F)• Opladen: 0 tot 45 °C (32 tot 113 °F)• Luchtvochtigheid: Max. 95% RV, zonder condensvorming• Houd het apparaat uit de buurt van locaties met buitengewoon
hoge of lage temperaturen• De batterij wordt niet snel opgeladen bij temperaturen lager dan
10°C of hoger dan 40°C. De batterij wordt helemaal niet opgeladen bij temperaturen lager dan 0°C of hoger dan 45°C
• Stel de batterij niet bloot aan vloeistof• Gebruik het apparaat niet als u merkt dat de batterij beschadigd is• Demonteer of modificeer de batterij niet
• Hanteer de batterij en voer deze af volgens de lokale regelgeving• Als het oplaadproces niet wordt voltooid, zelfs nadat de aangegeven oplaadtijd is verstreken, stopt u onmiddellijk met verder opladen• Laat de batterij niet onbeheerd achter tijdens het opladen• Koppel de oplader los als de batterij volledig is opgeladen• Onjuist gebruik of onjuiste afvoer van lithium-ionbatterijen kan
brand veroorzaken• Niet bedoeld voor gebruik in ontvlambare omgevingen
WARNING
Dit symbool wordt gebruikt om gebruikers te waarschuwen voor
de aanwezigheid van belangrijke bedienings- en
onderhoudsinstructies in de literatuur die bij dit instrument
geleverd wordt.

81

Table 1: Symbolentabel
Beschrijving
Battterijlading 75–100%
Battterijlading 50–74%
Battterijlading 30–49%
Battterijlading 10–29%
Battterijlading <10%
Batterij wordt opgeladen
Volume ingesteld op 100% (standaard)
Volume ingesteld op 50%
Volume gedemptCO2-sensor is geïnstalleerd
Geeft modus Cloud Hunting aan
Geeft modus Pinpoint aan
Geeft modus Manual Zero aan
Gevoeligheid = Super
Gevoeligheid = Hoog
Gevoeligheid = Gemiddeld
Gevoeligheid = Laag
Batterijfout
De temperatuur van de batterij bevindt zich boven ofonder het ideale temperatuurbereik en de batterij wordt
mogelijk niet correct opgeladen
Er is een sensorfout opgetreden

82

Nederlands

Symbool프로브 팁/필터전원 버튼(보이지 않음)
헤드폰 잭
MODE 버튼SENS/ZERO버튼마이크로USB 전원 입력(보이지않음)꺼내기 버튼(보이지 않음)
배터리 덮개

83

한국어

볼륨 버튼

1. D-TEK Stratus 켜기 및 사용 준비a. 전원 버튼을 길게 눌러 D-TEK Stratus 을 켜거나 끕니다.

노트 :화면이 켜지지 않으면 배터리가 매우 부족하므로 충전해야 합니다 . D-TEK Stratus 는 충전 중에도 사용할 수있습니다 .
b. D-TEK Stratus 가 45~90초 동안 작동 준비를 시작합니다. 작동준비가 완료되면 D-TEK Stratus 를 사용할 수 있습니다.
c. 모드를 전환하려면 MODE 버튼을 누릅니다. 그러면 클라우드 헌팅, 핀포인트 및 수동 영점 모드 사이를 전환합니다.
注意 : D-TEK Stratus 는 항상 마지막으로 사용한 모드에서 시작됩니다 .

2. 클라우드 헌팅(휴대용 모니터) 모드 사용a. 의심되는 영역을 천천히 이동하면서 ppm 판독값을 관찰합니다.
b. ppm 판독값을 따라 냉매 농도가 더 높은 영역을 찾습니다. 숫자가 높을수록 냉매 농도가 더 높고 누설 지점에 더 가까워지는 것입니다.
노트 : 냉매 영역에 몇 분 동안 머물러 있으면 ppm 판독값이감소할 수 있습니다 . 이 경우에는 농도가 더 낮은 영역으로 몇 초 동안 이동하십시오 .
c. SENS/ZERO 버튼을 눌러 최대 기능을 활성화/비활성화합니다.
활성화되면, 관찰된 가장 높은 ppm 레벨이 주 ppm 디스플레이아래에 표시됩니다. 최대 판독값을 재설정하려면 SENS/ZERO한국어버튼을 길게 누릅니다.
노트 : 클라우드 헌팅 모드에는 민감도 설정이 없습니다 .

3. 핀포인트 모드 사용
a. D-TEK Stratus 의 팁을 의심되는 누설 지점에 최대한 가까이 놓
습니다. 공기 흐름이 차단되지 않게 하십시오.

84

b. 천천히(초당 2.5~5cm[1~2인치]) 이동하여 누설 가능성이 있는각 지점을 지나갑니다. 누설이 감지되면 D-TEK Stratus 가 경보를 울리고 화면의 표시기가 켜집니다.
핀포인트 모드에서 , D-TEK Stratus 는 몇 초가 지나면노트 :배경 냉매에 대해 자동으로 영점 설정됩니다 . 이 경우에는 더 높은 냉매 농도를 계속 찾거나 농도가 더 낮은영역으로 이동하여 영점을 재설정하십시오 .
c. 누설이 식별되면 누설 지점에서 프로브를 당겨 몇 초 동안 빼낸후 의심되는 누설 부위로 돌아가 누설을 확인합니다.

노트 :누설 부위가 큰 경우 또는 SAE 적용개소의 경우에는노트 :재 감도가 화면에 표시됩니다 .
SAE 적용개소에는 매우 높음 감도 설정을 사용하지 마

SENS/ZERO 버튼을 눌러 감도 모드를 전환합니다 . 현십시오 .

4. 수동 영점 모드수동 영점 모드는 핀포인트 모드와 비슷한 방식으로 작동하지만,이 모드에서 사용자는 SENS/ZERO 버튼을 눌러 배경 냉매에 대해기기를 수동으로 영점 설정할 수 있습니다.
노트 : 수동 영점 모드에는 민감도 설정이 없습니다 .

5. 리튬 이온 배터리 분리 및 장착배터리 분
해a. 꺼내기 버튼을 눌러 배터리 덮개를 엽니다.
b. 배터리가 나오기 시작할 때까지 배터리 분리 탭을 옆으로 움직
입니다.

85

한국어

꺼내기 버
튼

c. 배터리를 당겨서 빼냅니다.
d. 새 배터리를 레일에 맞춥니다.
e. 배터리 분리 탭이 맞물릴 때까지 레일을 따라 배터리를 조심스럽게 밀어 넣습니다.

노트 :배터리를 억지로 밀어 넣지 마십시오 . 자연스럽게 들어가지 않으면 정렬 상태를 확인하고 다시 시도하십시
오.

6. 센서 교체 및 장착꺼내기 버튼센서 분해 탭a. 꺼내기 버튼을 눌러 배터리 덮개를 엽니다.
b. 센서 분리 탭을 잡고 센서를 부드럽게 당겨서 빼냅니다.
c. 새 센서를 레일에 맞춥니다.
d. 딸깍 하면서 제자리에 고정될 때까지 레일을 따라 센서를 조심스럽게 밀어 넣습니다.

노트 :센서를 억지로 밀어 넣지 마십시오 . 자연스럽게 들어가지 않으면 정렬 상태를 확인하고 다시 시도하십시오 .
e. 배터리 덮개를 다시 장착합니다.

7. 필터 교체한국어a. 프로브 팁을 풀고 필터를 제거합니다.
b. 새 필터를 프로브 팁에 삽입합니다.
c. 프로브 팁을 조입니다.

8. 충전 및 배터리 정보
D-TEK Stratus를 충전하려면 제공된 벽면 또는 차량 충전기를 적절한 전원에 연결하고 마이크로 USB 케이블을 D-TEK Stratus 마이크86

WARNING이 기호는 이 기기와 함께 제공되는 자료에 중요한 작동 및 유지
보수(정비) 지침이 있음을 사용자에게 알리는 데 사용합니다.

87

한국어

로 USB 전원 입력부에 연결합니다. 화면상 배터리 기호에 충전율이 표시되고 배터리가 충전 중임을 나타냅니다.
노트 : 참고 : D-TEK Stratus 는 충전 중에도 작동할 수 있습니다.
노트 : 참고 : 새로운 리튬 이온 배터리는 30% 미만으로 충전되어 있습니다 . 사용하기 전에 배터리를 완전히 충전하는 것이 좋습니다 . 완전히 충전하는 데 약 3 시간이걸릴 수 있습니다 .
주의:• 5V(dc), 1A 출력의 인증된 충전기/코드만 사용하십시오
• 온도 범위 및 습도:• 보관: -20 ~ +60°C(-4 ~ +140°F)• 작동: -20 ~ +50°C(-4 ~ +122°F)• 충전: 0 ~ 45°C(32 ~ 113°F)• 습도: 95% RH NC 최대•온도가 극도로 높거나 낮은 위치에 장치를 두지 마십시오• 10°C (50°F) 미만 또는 40°C (104°F) 초과 온도에서는 배터리가 빠르게 충전되지 않습니다. 0°C (32°F) 미만 및 45°C (113°F) 초과온도에서는 충전이 전혀 이루어지지 않습니다•배터리를 액체에 노출하지 마십시오•배터리에서 손상이 발견되면 장치를 사용하지 마십시오•배터리를 분해하거나 개조하지 마십시오•현지 규정에 따라 배터리를 처리하고 폐기하십시오
•지정된 재충전 시간이 경과되었는데도 재충전 작업이 완료되지 않는 경우, 재충전을 즉시 중지하십시오
•충전하는 동안 배터리를 방치해 두지 마십시오•배터리가 완전히 충전되면 충전기의 플러그를 뽑습니다•리튬 이온 배터리를 잘못 사용하거나 폐기하면 화재가 발생할 수있습니다•가연성 환경에서는 사용할 수 없습니다

Table 1: 기호표기호설명배터리 충전 75~100%배터리 충전 50~74%배터리 충전 30~49%배터리 충전 10~29%
배터리 충전 10% 미만배터리 충전 중
볼륨이 100%로 설정됨(기본값)볼륨이 50%로 설정됨음소거됨
CO2 센서가 장착됨클라우드 헌팅 모드를 나타냄핀포인트 모드를 나타냄수동 영점 모드를 나타냄
감도 = 매우 높음
감도 = 높음
감도 = 중간

한국어감도 = 낮음배터리 오류
배터리가 적정 온도 범위보다 높거나 낮으며 충전이 올바르게 되지 않을 수 있음센서 오류가 발생함

88

Download the full operating manual from
Descargue el manual de operación completo en
Das komplette Betriebshandbuch können Sie herunterladen bei
Téléchargez le manuel d’instructions complet sur
Scaricare il manuale operativo completo da请浏览詳細な操作マニュアルは以下のURLからダウンロードして下さいЗагрузите полное руководство по эксплуатации с веб-сайта
Pobierz pełną instrukcję obsługi na
Ladda ner den fullständiga bruksanvisningen frånKullanım kılavuzunun tamamını indirmek için
Download de volledige gebruikshandleiding van다음 웹 사이트에서 전체 작동 설명서를 다운로드하십시오
www.inficonservicetools.com

Two Technology Place
East Syracuse, NY 13057-9714 USA
Phone: +1.800.344.3304
Fax: +315.434.2520
E-Mail: service.tools@inficon.com
www.inficonservicetools.com
Bonner Strasse 498
D-50968 Cologne, Germany
Phone: +49 221 56788-660
Fax: +49 221 56788-9660
E-Mail: servicetools.europe@inficon.com
www.inficonservicetools-europe.com
074-709-P10A © 2019 INFICON