Remote sensor system
Temperature sensor andWireless Analog signal transmission＆ Wireless Power Supply
Load cell type

Thermocouple type

Temperature sensor type

Applications
Just by facing the items, you can easily Supply!
Wireless, so no mechanical operate contact helps long-life.
Thermocouple
Thermocouple

While rotating, appropriate temperature management is possible

Thermocouple

Thermocouple

Temperature controlof the Mixing machine.

Instead of the connector
Remote system
Instead of using slip ring

Remote system

Introduction the products

Analog sensor type(1 signal)

Load cell type(1 signal)

Transmit output
signal of analog
sensor and output
it as 0…10V.

Transmitter
output sensor
OutputRNT-1803-VS10-PU-01
RNE-1803A-PU-02

Temperature managementof the die, forming die andthe casting type.

Transmit output signal ofcompression load cell andoutput as 4…20mA.

Transmitter
output sensor
OutputResistance thermometer type（1 signal )for Resistance thermometer
Pt100. Transmit thetemperature measuring signal
and output as 4…20mA.

RNT-1804-LC10-PU-01 RNT-1804-LC15-PU-01 RNT-1804-LC20-PU-01 Transmitter
RTT-1804-PT1B10-PU-01 RTT-1804-PT1B20-PU-01 RTT-1804-PT1B30-PU-01
RTE-1804E-PU-02
output sensor RTE-1804E-PU-02
4...20mA x1signal
0...10V x1signal
Output4...20mAｘ1signal
Transmission range/Center offset 1...4mm/ ± 2.5 ｍｍ
Transmission range/Center offset 1...4mm/ ± 2.5 ｍｍ
Transmission range/Center offset 0...2.5mm/ ± 2 ｍｍ
protection class IP67
protection classprotection class IP67
IP67
Response delay ≦ 0.5 sec.
Response delay≦ 0.2 sec.
Response delay ≦ 0.5 sec.
compression load cell 350 Ω± 10％
Sensor type
Analog sensor output 0...10V Sensor type
Sensor type
Standard Resistance Thermometer type Pt100 3-wire
Input sensitivity 1mV/V
1.5mV/V
2mV/V
Operating temperature 0...+60℃
Measuring
0...100℃
0...200℃
0...300℃
Operating temperature 0...+60℃
Operating temperature 0...+60℃
Size Transmitter
M18 X 76mm
Output sesnsor
M18 X 94mm
Size Transmitter
M18 X 61mm
Size Transmitter
M18 X 61mm
Output sensor
M18 X 94mm
Output sensor M18 X 94mm

Thermocouple type (K, J) 1 transmission M18, M30 type

Thermocouple type (K, J) 2 signals ring typeTransmit the temperature measuri signalof thermocouple K,J type and output it as4…20mA.

M18 type
Transmitter
output sensor
OutputTransmission range/Center offset
protection class
Response delay
Sensor typeTransmit the temperature measuring
signal of thermocouple K/J type andoutput it as 4…20mA. The central
inside diameter for medium size of φ50mm and large size of φ 100mm areavailable.
Diameter φ 50 types

M30 type

RS02T-018-K300
RTT-1804-K-100
RS02T-018-J300 RS02T-030-K300
RS02T-018-K1000
RTE-1804E-PU-02 RS02E-018E-PU-02 RS02E-018E-PU-02 RS02E-030E-PU-02

Transmitter

4...20mAｘ1signal 4...20mAｘ2signals
1...4mm / ± 2.5mm
2...6 ｍｍ / ± 4mm
IP67≦ 0.5 sec.
JIS thermocouple J JIS Thermocouple K
JIS Thermocouple K
0...300℃
Measuring range
0...1000℃
0...300℃
0...300℃
0...1000℃
Operating temperature 0...+60℃
0...+80℃

Size Transmitter
Output sensor

M18 X 76mm
M18 X 94mm

M18 X 61mm
M18 X 94mm

M30 X 74mm
M30 X 95mm

output sensor
OutputTransmission range/Center offset
protection class
Response delay
Sensor typeMeasuring range

Diameter φ 100 types

RS02T-R01-K300
RS02T-R01-K1000
RS02E-R01E-PU-02

RS02T-R01-J300

RS02T-R03-K300
RS02E-R03E-PU-02

4...20mA ｘ2 signals
0...8mm/ ± 8mm
IP67≦ 0.5 sec.
JIS Thermocouple K
0...300℃
0...1000℃

0...9mm/ ± 8mm *1

JIS Thermocouple J JIS Thermocouple K
0...300℃

Operating temperature 0...+50℃
Size (Transmitter･Output sensor） φ116ｘD 35mm

0...300℃

φ156ｘD 35mm

installation screw], please prepare for M8 screw x4 pcs.
*1 when with metal shaft

Temperature sensor Specification（Thermocouple 8signals/Resistance thermometer 2signals/ Thermistor 2signals)

Resistance thermometer typeTransmission amplifier

RS080T-233

Transmitting head ring type
RSH080T-R01-CN

Or

Transmitter

Thermocouple
Thermistor

Output sensor

Power supply

Output Head ring type
RSH080E-R01R-CN

Type

RS080T-233

Output HeadOr
RS232C

Current output

PLC

Current output amplifier

RS801E-234E
RS232C

Output head compact type
RSH080E-422R-CN

Transmission head compact type
RSH080T-422-CN

Transmission amplifier Transmitting head

Power supply

Current output amplifier ■ The setting of the temperature sensor is simple and easy. Connect PC to a transmission amplifier using the attached software.

RSH080T-422-CN（compact） RSH080E-422R-CN（compact）
RS801E-234E
RSH080T-R01-CN（ring）
RSH080T-R01-CN（ring）

Output
RS232C Max 8ch (When using output amplifier：Current output 4-20mA Max 8ch）
Transmission range/Center offset compact：0...3mm / ± 3mm
ring：0...6mm / ± 6mm
Protection class
No Protection class IP67
IP67
No Protection classCompatible sensor type and max. transmission signal Thermocouple:8signals Resistance thermometer:2signals,Thermocouple:2signals

Operating temperature
Compact：0...70℃
Ring：0...50℃
Connection cable between the head - amplifier 06F-M08S01-02 (Straight、2m) / 06F-M08R01-02 (Angle、2m)
Size (Transmitter･Output sensor） Compact：H55 Ｘ W55 Ｘ D 25mm
Ring：φ 116 Ｘ D 35mm
installation screw], please prepare for M8 screw x4, a compact type: M4 screw x2, an amplifier:M4 screw x4.

8 signals transmission of Thermocouple（J,K,E,N,R,S,T,B)
2 signals transmission of resistance thermometer type（RTD PT10,50,100,200,500,1000/ RTD1000/RTDNI-120)
2 signals transmission of Thermistor（44004/44033,44005/44030,44007/44034,44006 /44031, 44008/44032,YSI400,Spectrum1003K)■Please contact for initial setting of the transmission amplifier.
■A temperature signal output by an output head is a serial signalof RS-232C,
Possible to input to PLC or a PC of the serial interface.
In addition, using the current output amplifier, and convert into
4...20mA.

Introduction of the custom example

Please let us know your requests.

Custom development request is also available,please feel free to contact us.

Case 1

Integrating the sensors.
We’ve commercialized an integral type with the sensor specified by the customer, and reduced the labor of sensor wiringand improvement of waterproofness.

Case 2

Request for the shape that fits the equipment.

Case 3

Wants to use other than thermocouple J or K.

Various shapes such as ring and sector that are easy to install on the axis. Prepared exclusive board in accordance withthe customer’demands.

Customized 2 signals type of thermocouples to standard typeto 2 signals types B types. Depending on customers requests.

Wireless Power Supply byMail ：b-plus-usa@b-plus-kk.com
Web ：http://www.b-plus-kk.com* Info may change the mention contents such as specifications without a notice.
Thank you for understanding* Please refer to instruction manual or the user's guide. It can be download by HP.

BN1713Ae2018.02