TM700

HIGH-SPEED TORQUE WAVEFORM MONITORFOR UTM⋜/UTMVSpecifications
Torque input

Rotation
input

Display
External
I/O signal● High-speed sampling at 20kS/s!!

Interface● A torque curve monitor supporting thehigh response time of UTMϩ/UTMV (1kHz).
● Display the curves of torque, rotation speed, and power.
● Enable the cogging torque measurement with UTMϩ/UTMV .
● Upper/lower limits & hold function.

Power supply
Power consumption
Operatingconditions
Dimension
WeightAttachment

Operation manualҩҩҩ1, External input/output connectorҩҩҩ1,Cable for connecting UTMϩ/UTMV‫ق‬2m‫ك‬ҩҩҩ1, Operating tool㸦with D/A converter(3ch) option㸧ҩҩҩ1Accessories

DTC2‫؟‬CA372-I/O‫؟‬CA81-USB‫؟‬
CATM21-M‫؟‬
CATM51-M‫؟‬
CATM(R)21‫؟‬
CATM(R)21-M‫؟‬
CATM(R)51-M‫؟‬
CN50‫؟‬CN55‫؟‬CN73‫؟‬CN90‫؟‬CN91‫؟‬
GMP96x96‫؟‬
TSU03‫؟‬CE markingcertification

EMC directives EN61326-1● Operation power can be supplied to the UTMϩ/UTMV .ࠉ ࠉ
● Easy connection to UTMϩ/UTMV with the supplied cable.
● Data recording with USB.
Torque↓↑Speed‫؞‬Voltage input‫ؙ‬Signal input range -5 to +5V Input impedance‫؟‬1MΩ or more
Non-linearity‫؟‬within 0.02٫FS±1digit‫ؙ‬Accuracy
Zero drift‫؟‬within 0.2mV/Υ RTI
Gain drift ‫؟‬within 0.01٫/Υfilter
Lowpass filter‫ٕق‬6dB/oct.‫ك‬Select from 10, 30, 100, 300, 1k, 3k, 10k, 30kHz‫ؙ‬Analog‫ؙ‬A/D converter Rate‫؟‬20000times/secResolution‫؟‬24bit‫ق‬binary‫ؙؙك‬Approx. 1/30000 against 5V
Output level: 1V output per 1V input (load resistance: 2kΩ or above)‫ؙ‬Monitor output‫؞‬Pulse input‫ق‬open-collector‫ك‬■ high-speed‫ق‬for standard UTMϩ/UTMV‫ك‬
Accordance with UTMϩ/UTMV‫ؙ‬Max. frequency
Select from 15, 10, 5, 3, or 2 rpm (when pulse rate is 4 ppr)‫ؙ‬Min. frequency
60, 40, 20, 12, 8 rpm (when pulse rate is 1 ppr)‫ؙ‬Min. pulse width 50μs
‫ؙ‬Circuit configuration:No-voltage contact input (minus common) Open collector connectable‫ق‬IcٙApprox. 10mA‫ك‬■ low-speed‫ق‬for UTMϩwith rotary encoder option‫ك‬
Approx.1000rpm‫ؙ‬Max. frequency
0.1rpm‫ؙ‬Min. frequency‫ؙ‬Min. pulse width 5μs‫ؙ‬Circuit configuration:No-voltage contact input (minus common) Open collector connectable‫ق‬IcٙApprox. 10mA‫ك‬Display unit
3.5 inch TFT color LCD
External output‫ق‬8‫ك‬ex) high and low limit outputs
External input‫ق‬10‫ك‬ex) hold control input, digital zero input(DZ)
SIF‫؟‬2-wire serial interfaceUSB‫؟‬USB interface
D3V‫؟‬D/A converter voltage output‫ق‬3ch‫(ك‬Option)
DC24V‫ق‬±15٫‫ك‬
6W typTemperature‫؟‬Operation‫ؙٕ‬10Υ ‫ٔ ع‬40Υ‫ؙ‬Storage‫ؙٕ‬40Υ‫ٔ ع‬80Υ
Humidity‫؟‬85%RH or below (non-condensing)96(W)× 96(H)× 138(D)mm‫ق‬excluding projections‫ك‬Approx.1.0kg

Case for TM700 (with AC power supply)
Cable with FCN connector at one-end 3m
miniUSB-computer USB cable 1.8m
Cable for connecting UTMϩ/UTMV 2m
Cable for connecting UTMϩ/UTMV 5m
Cable for UTMϩwith rotary encoder connection (non-connector at one end)UTMϩ 2m
Cable for connecting UTMϩrotary encoder 2m
Cable for connecting UTMϩrotary encoder 5m
FCN series I/O connector (with cover)
FCN series I/O connector (with diagonal cover)'$FRQYHUWHU FK FRQQHFWRUWaterproof plastic connector for UTMϩ/UTMVWaterproof plastic connector for UTMϩrotary encoder
Rubber seal
DC lighting surge unit

Structure of product code

↑Monitor the fluctuationof torque, rotation speed, and power.

Power① Standard unit

TM700 ‫ڧ‬ձղ

② Interface

SignInterfac
Standard SI/F、USB↓One optional interface can be added in addition the standard interface.
D/A converter(Voltage output)（3ch）
D3VExternal dimension(Front)(Side)(Rear)

Panel cutout size
Panel thickness
1.6 to 3.2mm

Unit

DTC2：Case for TM7001201432m

20715°

168190At stand usemm