TGF11 Tracer Gas Filler

Leak testing with tracer gas requires filling your test object. The TGF11Tracer Gas Filler is a stand-alone instrument which allows for easy andreliable tracer gas filling and evacuating. It ensures that the entire test
object is filled with tracer gas at the right time and at the right pressure.
It also evacuates the tracer gas after the test avoiding contamination ofthe work area.
The new user interface allows easy, intuitive control of the instrument
reducing operating errors. The filler has no moving parts, thus
minimizing maintenance expense, and is well-suited for use in industrial
serial production.
The TGF11 is an accessory to INFICON hydrogen and helium leakdetectors, such as Sensistor Sentrac Hydrogen Leak Detector.

ADVANTAGESn

Ensured tracer gas filling in the entire test objectn

Avoids background contaminationnMade for demanding industrial environmentsn

Dual port for filling and evacuating for short cycle timesnMinimal maintenancenEasy set-up

1

TGF11 Tracer Gas Filler

FEATURESn

Evacuates, fills and purges the test objectn

Available in low pressure version (0.05 - 2 barg/ 0.7 - 29 PSIG) and standard version (0.3 - 10 barg /4.4 - 145PSIG)

nGross leak abort function prior to tracer gas filling

nProp valve for optimized fillingn

Uses a cost-effective venturi pump for gas evacuation - minimazing maintenance costs

ORDER INFORMATIONTGF11 Tracer Gas Filler, lowpressure

TypeTracer Gas Filler TGF11, low pressure version (0.05 - 2 barg/0.7 - 29 PSIG)
Tracer Gas Filler TGF11, standard pressure version (0.3 - 10barg /4.4 - 145 PSIG)

2TGF11 Tracer Gas FillerTGF11 Tracer Gas Filler,
standard pressure590-559590-558

SPECIFICATIONSTypeTGF11 Tracer Gas Filler, lowpressure

TGF11 Tracer Gas Filler,
standard pressure

Pressure

0.05-2 barg (0.7-29 PSIG)

0.3-10 barg (4.4-145 PSIG)

Tracer Gas pressure supply

100 - 400 kPa (1 - 4 barg,14,5 -58 PSIG)

100 - 1100 kPa (1 - 11 barg,14.5 -159.5 PSIG)

Compressed Air

300 - 1000 kPa (3-10 barg
/43.5-145 PSIG)

300 - 1000 kPa (3-10 barg
/43.5-145 PSIG)

Connections

Push in fittings, OD 6,10,12mm

Push in fittings, OD 6,10,12mm

Power Supply

100 – 240 VAC, 50/60 Hz, 2A

100 – 240 VAC, 50/60 Hz, 2A

Tracer Gas Pressure Supply

100 - 1100 kPa (1-11 barg /14.5-159.5 PSIG)

100 - 1100 kPa (1-11 barg /14.5-159.5 PSIG)

Evacuation Time

0.8 s/l to -0.5 barg (-7.2PSIG); 1.6 s/l to -0.7 barg (10.1 PSIG); 2.5 s/l to -0.8barg (-11.6 PSIG)

0.8 s/l to -0.5 barg (-7.2PSIG); 1.6 s/l to -0.7 barg (10.1 PSIG); 2.5 s/l to -0.8barg (-11.6 PSIG)

Maximum vacuum-0.85 barg (-11.6 PSIG) (85% -0.85 barg (-11.6 PSIG) (85%vacuum)
vacuum)Supplies

Capacity*

Communication Interface

USB device
RS232
PLC Input/Output
Operator interface

USB device
RS232
PLC Input/Output
Operator interface

Ambient temperature range

5°-45°C (41°-113°F)

5°-45°C (41°-113°F)

Dimensions (W x H x D)

305 x 160 x 284 mm(12 x 6.2 x 11.1 in.)

305 x 160 x 284 mm(12 x 6.2 x 11.1 in.)9.5 kg (19.8 lb.)

9.5 kg (19.8 lb.)

Weightwww.inficon.com

reachus@inficon.com

Due to our continuing program of product improvements, specifications are subject to change without notice.
RateWatcher is a trademark of INFICON. All other trademarks are the property of their respective owners.
(2022-02)© 2022 INFICON