DIGITAL THERMOMETER

TC-400A
TC-330AWPFeatures

MULTI FUNCTION
MULTI FUNCTION+WATER PROOF

Thermocouple Type K / Auto-Power Off / Data Hold /Burn-out Display / Battery Alarm / MAX. MIN. Record /Selectable Measuring Unit °C or °F(TC-400A Only) /Data Memory / Relative Measurement /In Conformance to CE(EMC) / RoHS Compliant

TC-400A

TC-330AWPTHERMOCOUPLE PROBERoHSDIGITAL THERMOMETER

TC-400A/TC-330AWPSpecifications

FunctionsTC-400A

ModelSensor Type
Measuring RangeResolution

Type K

TC-330AWP

Input connector
MAX data indicator
MIN data indicator-160°C (-256°F) - +1,372 °C(+ 2,502°F)
199.9°C (391.8°F) and below: 0.1°C (0.2°F)

Hold status indicator
Auto-Power Offindicator

200°C (392°F) and above: 1°C (2°F)

Accuracy(at 25°C(77°F) ±5°C(9°F))

0~199.9°C (32°F~391.8°F): ±0.1% of reading + 0.7°C(1.3°F)

Thermocouple
Type indicator-0.1°C(31.8°F) or less: ±0.2% of reading + 1°C(1.8°F)
200°C(392°F) above:±0.2% of reading + 1°C(1.8°F)

Temperature Coefficient±0.015% of reading + 0.108°F / °F

Relative function
indicator
Stored data indicator

Data display
Hold key

Approx. 0.5 sec.
Digital Linearize

PowerON / OFF key°C/ °F (Selectable：TC-400A Only) , Auto Power-Off , Data Hold ,Data Record(MAX / MIN / Memory) , Relative Measurement

Operating Temp./ Humidity

Unit display

Data polarity
indicator
Recordingindicator±0.015% of reading + 0.06°C / °C

Sampling TimeLinearize Method
Operation Function

Low Battery indicator

Unit selection key(TC-400A Only)

Record key

0 - 50°C ( 32 - 122oF )Relative display
function key

REL0 - 90 % RH at 0 - 35°C (Non-condensing)Power Supply
Battery life
Size
WeightSensor Connector
Accessories
IP Protection
Compliance

Data recall key

Battery 006P ( 9V )Approx. 500 Hours
166(H) x 68(W) x 35(D)mm
200g

210g

Thermocouple Connector

Waterproof Connector

One Battery 006P, Instruction Manual
None

IP65
CE and RoHS

K Type Thermocouple Probes (Sold Separately)
Model No.

ThermometerMax.
Temp.

Dimensions(mm)Cable:L=1000

16

TC-400A

500°C

Type / Material ofprotection pipe

90

Surfaces

φ15

KF-01A

Applications

φ12φ15500°C

25TC-400A

400°C

φ61000°C

Small surfaces / Hard-to-reach surfaces

370

120

Surfaces (with high temperature)

φ28

TC-400A

φ1215023.5

KF-16A

Surfaces / Hard-to-reach surfaces

150

33

KF-08A

φ1240

TC-400A

7KF-02A

φ12150KS-21A

TC-400A

φ20100

800°C
φ3.2
150

KB-101WP

TC-330AWP

300°C

KS-104WP

TC-330AWP

700°C

φ12100

φ3.2
300

40

φ1KS-105WP

150

TC-330AWP

800°C

100
φ3.2

Liquids /Interiors of
Semi-solids(with high temperature)

Sheath Type / SUS316

Liquids /Interiors of
Semi-solids

Pipe Type / SUS304

Liquids(with hightemperature)

Sheath Type / SUS316

Liquids /Interiors ofSemi-solids(with high temperature)

Sheath Type / SUS316

More models of thermocouple probe are available.
Please refer to our website to choose the best model for your application.

Authorized Distributor :Address : 37-7 Chuo-cho, 2-Chome Meguro-ku,Tokyo 152-0001 Japan
Please visit our company website for contact details
E-mail : webtrade@line.co.jp Web : http: //www.line.co.jp
This catalog was last revised Oct. 8, 2021.
*Subject to change without prior notice.
All Rights Reserved,Copyright © 2021,LINE SEIKI CO.,LTD.