GP01197T/09/EN/01.22-00
71596800
2022-08-30

Products

Solutions

Description of Device Parameters
iTEMP TMT182B
Temperature transmitter

Services

iTEMP TMT182B

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

1.1
1.2
1.3
1.4

Document function . . . . . . . . . . . . . . . . . . . . .
Target group . . . . . . . . . . . . . . . . . . . . . . . . . .
Using this document . . . . . . . . . . . . . . . . . . . .
Documentation . . . . . . . . . . . . . . . . . . . . . . . .

2

Overview of the operating menu . . . . . 7

3"Guidance" menu . . . . . . . . . . . . . . . . . . 14

3.1
3.2

"Commissioning" wizard . . . . . . . . . . . . . . . . . 14
"Import / Export" submenu . . . . . . . . . . . . . . . 21

4

"Diagnostics" menu . . . . . . . . . . . . . . . . 22

4.1
4.2
4.3
4.4
4.5
4.6

"Active diagnostics" submenu . . . . . . . . . . . . . 23
"Diagnostic list" submenu . . . . . . . . . . . . . . . . 24
"Event logbook" submenu . . . . . . . . . . . . . . . . 25
"Simulation" submenu . . . . . . . . . . . . . . . . . . . 26
"Diagnostic settings" submenu . . . . . . . . . . . . . 27
"Minimum/maximum values" submenu . . . . . . 28

5"Application" menu . . . . . . . . . . . . . . . . 30

5.1
5.2
5.3
5.4

"Measured values" submenu . . . . . . . . . . . . . .
"Sensor" submenu . . . . . . . . . . . . . . . . . . . . . .
"Current output " submenu . . . . . . . . . . . . . . .
"HART configuration" submenu . . . . . . . . . . . .

6"System" menu . . . . . . . . . . . . . . . . . . . . 43

6.1
6.2
6.3

"Device management" submenu . . . . . . . . . . . . 45
"User management" submenu . . . . . . . . . . . . . 47
"Information" submenu . . . . . . . . . . . . . . . . . . 52

Endress+Hauser

4
4
4
6

31
34
40
42

3

About this document

iTEMP TMT182B

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters, providing a detailed explanation of each individual parameter of the operating
menus.
It is used to perform tasks that require detailed knowledge of the function of the device:• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Preferred
Procedures, processes or actions that are preferred.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.
Reference to documentation
Reference to pageA

Reference to graphic
Notice or individual step to be observed
1. , 2. , 3. …

Series of steps
Result of a step
Help in the event of a problem
Visual inspection
Operation via local display
A0028662

Operation via operating tool
A0028663

Write-protected parameterA0028665

4

Endress+Hauser

iTEMP TMT182B

About this document

SymbolMeaning

SymbolPermitted
Procedures, processes or actions thatare permitted.

Preferred
Procedures, processes or actions thatare preferred.

Forbidden
Procedures, processes or actions thatare forbidden.

Tip
Indicates additional information.

Reference to documentation
Reference to graphicA

1. , 2. , 3. …

Result of a step

1.3.2

Meaning

Reference to page
Series of steps
Visual inspection

Information on the document structure

The parameters of all the operating menus and the commissioning wizard are described inthis document.
• Guidance menu with the Commissioning wizard (→  14), which guides the user
automatically through all the device parameters that are needed for commissioning• Diagnostics menu (→  22)• Application menu (→  30)• System menu (→  43)

1.3.3

Structure of a parameter description

The individual parts of a parameter description are described in the following section:
Complete parameter name

Navigation

Write-protected parameter =Navigation path to the parameter via the operating tool
The names of the menus, submenus and parameters are abbreviated to the form in which they appear on
the display and in the operating tool.

Prerequisite

The parameter is only available under these specific conditions

Description

Description of the parameter function

Selection

List of the individual options for the parameter• Option 1• Option 2

User entry

Input range for the parameter

User interface

Display value/data for the parameter

Factory setting

Default setting ex works (if not explicitly selected)

Additional information

Additional explanations (e.g. in examples):• For individual options• For display values/data• For the input range• For the factory setting• For the parameter function

Endress+Hauser

5About this document

iTEMP TMT182B

1.4

Documentation

The Description of Device Parameters is part of the following documentation:
Temperature transmitter iTEMP TMT182B: BA02260T

6

Endress+Hauser

iTEMP TMT182B

Overview of the operating menu

2

Overview of the operating menu

Navigation



Operating tool→  14

Guidance‣ Commissioning→  14‣ Device management
Device tag→  14

Device name→  15

Serial number→  15Extended order code 1 to 3

→  15‣ Sensor→  16Unit→  16

Sensor type→  16

Connection type→  17

2-wire compensation→  17

Reference junction→  18

RJ preset value→  18‣ Current output→  18Lower range value output→  18Upper range value output→  19Failure mode→  19‣ User management

Endress+Hauser→  14→  19

Access status→  20

New password→  207Overview of the operating menu

iTEMP TMT182B

Confirm new password→  20Status password entry→  20‣ Import / Export→  21Create configuration report→  21

Save / Restore→  21→  22

Diagnostics‣ Active diagnostics
Actual diagnostics 1→  23Operating time→  23‣ Diagnostic list

→  24
Actual diagnostics 1 to 3

→  24

Actual diag 1 to 3 channel→  24

Timestamp 1 to 3

→  24‣ Event logbook→  25Previous diagnostics 1 to 10→  25Previous diag 1 to 10 channel→  25

Timestamp 1 to 10→  25‣ Simulation8→  23→  26
Diagnostic event simulation→  26

Current output simulation→  26

Current output value→  26

Sensor simulation→  27

Sensor simulation value→  27

Endress+Hauser

iTEMP TMT182B

Overview of the operating menu

‣ Diagnostic settings→  27

Alarm delay→  27

Thermocouple diagnostic→  28‣ Minimum/maximum values→  28Sensor min value→  28Sensor max value→  28Reset sensor min/max values→  29Device temperature min value→  29Device temperature max value→  29Reset device temp. min/max values→  29→  30Application‣ Measured values→  31
Sensor value→  31

Sensor raw value→  32

Output current→  32

Percent of range→  32

Device temperature→  32PV→  32SV→  33TV→  33QV→  33‣ Sensor→  34‣ Sensor

Endress+Hauser→  34Unit→  34

Sensor type→  349

Overview of the operating menu

iTEMP TMT182B

Connection type→  352-wire compensation→  35

Reference junction→  36

RJ preset value→  36

Sensor offset→  36‣ Linearization→  37Call./v. Dusen coeff. R0→  37Call./v. Dusen coeff. A→  37Call./v. Dusen coeff. B→  37Call./v. Dusen coeff. C→  38Polynomial coeff. R0→  38Polynomial coeff. A→  38Polynomial coeff. B→  39Sensor lower limit→  39Sensor upper limit→  39‣ Current output

10→  40Lower range value output→  40Upper range value output→  40

Failure mode→  40Current trimming 4 mA

→  41

Endress+Hauser

iTEMP TMT182B

Overview of the operating menu

Current trimming 20 mA→  41

Damping→  41‣ HART configuration→  42

HART address→  42

No. of preambles→  42→  43

System‣ Device management→  45

Device tag→  45

HART short tag→  45

Device reset→  45

Configuration counter→  46

Configuration changed→  46

Reset configuration changed flag

→  46‣ User management→  47‣ User management
Access status‣ Delete password→  47→  48

Status password entry‣ Enter password→  48→  48Password→  48

Status password entry→  48‣ Define password

Endress+Hauser→  47→  49New password→  49Confirm new password→  49Status password entry→  5011Overview of the operating menu

iTEMP TMT182B‣ Recover password
Reset password→  50Status password entry→  50‣ Change password→  51

Old password→  51

New password→  51

Confirm new password→  51

Status password entry→  52‣ Information→  52‣ Device→  52
Serial number→  52

Order code→  53

Firmware version→  53

Hardware revision→  53Extended order code 1→  53

Device name→  54

Manufacturer→  54‣ HART info12

→  50→  54
Device type→  54

Device revision→  55

HART revision→  55

HART descriptor→  55

HART message→  55

Hardware revision→  56

Software revision→  56Endress+Hauser

iTEMP TMT182B

Endress+Hauser

Overview of the operating menu

HART date code→  56

Manufacturer ID→  56

Device ID→  56

13

"Guidance" menu

iTEMP TMT182B

3"Guidance" menu

Main functions for use – from fast and safe commissioning to guided support during
operation.
Navigation

Guidance

Guidance‣ Commissioning→  14‣ Device management→  14‣ Sensor→  16‣ Current output→  18‣ User management→  19‣ Import / Export

3.1→  21Create configuration report→  21

Save / Restore→  21"Commissioning" wizard

Navigation

3.1.1



Guidance → Commissioning"Device management" wizard

Navigation

Guidance → Commissioning → Device management

Device tagNavigationDescription

Displays the name for the measuring point.

User entry

Character string comprising numbers, letters and special characters (32)14

Guidance → Commissioning → Device management → Device tag

Endress+Hauser

iTEMP TMT182B"Guidance" menu

Device name
Navigation

Guidance → Commissioning → Device management → Device name

Description

Displays the name of the transmitter.
Additional information:
The name can also be found on the transmitter's nameplate.

User interface

Character string comprising numbers, letters and special characters

Factory setting

iTEMP TMT182B

Serial number
Navigation

Guidance → Commissioning → Device management → Serial number

Description

Displays the serial number of the measuring device. The serial number can be used to
identify the measuring device and to retrieve further information via the Device Viewer or
Operations app, such as the related documentation.
Additional information:
The serial number can also be found on the nameplate of the sensor and transmitter.

User interface

Character string comprising numbers, letters and special characters

Extended order code 1 to 3
Navigation

Guidance → Commissioning → Device management → Extended order code 1 to 3

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Endress+Hauser

Character string comprising numbers, letters and special characters

15

"Guidance" menu

iTEMP TMT182B

3.1.2"Sensor" wizard

Navigation

Guidance → Commissioning → Sensor

Unit



Navigation

Guidance → Commissioning → Sensor → Unit

Description

Selection of the unit for all measured values.

Selection

SI units
• °C
•K
• Ohm
Custom-specific units
• °F
• mV

Factory setting°C

Sensor type

Navigation

Guidance → Commissioning → Sensor → Sensor type

Description

Use this function to select the sensor type for the sensor input.
Info:
Please observe the terminal assignment when connecting the sensor.

Selection

16• Pt100 IEC60751, a=0.00385 (1)• Pt200 IEC60751, a=0.00385 (2)• Pt500 IEC60751, a=0.00385 (3)• Pt1000 IEC60751, a=0.00385 (4)• Pt100 JIS C1604, a=0.003916 (5)• Ni100 DIN 43760, a=0.00618 (6)• Ni120 DIN 43760, a=0.00618 (7)• Ni100 OIML/GOST 6651-09, a=0.00617 (12)• Ni120 OIML/GOST 6651-09, a=0.00617 (13)• Type A (W5Re-W20Re) IEC60584 (30)
• Type B (PtRh30-PtRh6) IEC60584 (31)
• Type C (W5Re-W26Re) IEC60584 (32)
• Type D (W3Re-W25Re) ASTM E988-96 (33)
• Type E (NiCr-CuNi) IEC60584 (34)
• Type J (Fe-CuNi) IEC60584 (35)
• Type K (NiCr-Ni) IEC60584 (36)
• Type N (NiCrSi-NiSi) IEC60584 (37)
• Type R (PtRh13-Pt) IEC60584 (38)
• Type S (PtRh10-Pt) IEC60584 (39)
• Type T (Cu-CuNi) IEC60584 (40)
• Type L (Fe-CuNi) DIN43710 (41)Endress+Hauser

iTEMP TMT182B"Guidance" menu

• Type L (NiCr-CuNi) GOST R8.585-01 (43)
• Type U (Cu-CuNi) DIN43710 (42)
• Pt50 GOST 6651-94, a=0.00391 (8)• Pt100 GOST 6651-94, a=0.00391 (9)• Cu50 GOST 6651-09, a=0.00428 (10)• Cu100 OIML/GOST 6651-09, a=0.00428 (11)• Cu50 OIML R84:2003, a=0.00428 (10)• Cu50 OIML/GOST 6651-94, a=0,00426 (14)• RTD Platinium (Callendar/van Dusen)• RTD Poly Nickel (OIML R84, GOST 6651-94)• RTD Polynomial Copper (OIML R84:2003)• 10...400 Ohm• 10...2000 Ohm• -20...100 mV
Factory setting

Pt100 IEC60751, a=0.00385 (1)Connection type

Navigation

Guidance → Commissioning → Sensor → Connection type

Prerequisite

An RTD sensor or a resistance transmitter must be specified as the sensor type.

Description

Use this function to select the connection type for the sensor.

Selection• 2- wire• 3- wire• 4- wire

Factory setting

4-wire2-wire compensation



Navigation

Prerequisite

An RTD sensor or a resistance transmitter with a 2-wire connection type must be specified
as the sensor type.

Description

Use this function to specify the resistance value for two-wire compensation in RTDs.

User entry

0.0 to 30.0 Ohm

Factory setting

0.0 Ohm

Endress+Hauser

Guidance → Commissioning → Sensor → 2-wire compensation

17

"Guidance" menu

iTEMP TMT182B

Reference junction



Navigation

Guidance → Commissioning → Sensor → Reference junction

Prerequisite

A thermocouple (TC) sensor must be selected as the sensor type.

Description

Use this function to select reference junction measurement for temperature compensationof thermocouples (TC).
Additional information:
If "Fixed value" is selected, the compensation value is specified via the RJ preset value
parameter.

Selection• Internal measurement• Fixed Value

Factory setting

Internal measurement

RJ preset value

Navigation

Guidance → Commissioning → Sensor → RJ preset value

Prerequisite

The Fixed value parameter must be set if the Reference junction option is selected.

Description

The Fixed value parameter must be set if the Reference junction n option is selected.

User entry–58.0 to 360.0

Factory setting

0.0

3.1.3

"Current output " wizard

Navigation

Guidance → Commissioning → Current output → Current output

Lower range value output

Navigation

Guidance → Commissioning → Current output → Lower range value output

Description

Use this function to assign a measured value to the current value 4 mA.
Additional information:
The set point that can be set depends on the sensor type used in the Sensor typeparameter.

18

Endress+Hauser

iTEMP TMT182B"Guidance" menu

User entry–50 000.0 to 50 000.0

Factory setting

0.0Upper range value output

Navigation

Guidance → Commissioning → Current output → Upper range value output

Description

Use this function to assign a measured value to the current value 20 mA.
Additional information:
The set point that can be set depends on the sensor type used in the Sensor typeparameter.

User entry–50 000.0 to 50 000.0

Factory setting

100.0

Failure modeNavigation

Guidance → Commissioning → Current output → Failure mode

Description

Use this function to select the signal on alarm level of the current output in the event of anerror.

Selection• Max.
• Min.

Factory setting

Min.

3.1.4

"User management" wizard

The initial password can be assigned via user management during commissioning.
Each additional change to user management (delete password, change password, etc.)
is implemented in the menu: System → User management

Endress+Hauser

19

"Guidance" menu

iTEMP TMT182B

Navigation

Guidance → Commissioning → User management

Access status
Navigation

Guidance → Commissioning → User management → Access status

Description

If additional write protection is active, this restricts the current access authorization evenfurther.

User interface• Operator• MaintenanceFactory setting

MaintenanceNew passwordNavigation

Guidance → Commissioning → User management → New password

Description

If the factory setting is not changed, the device works without write-protection, using
userrole 'Maintenance'. The configuration data of the device can always be modified.
Once the password has been defined, write-protected devices can only be set tomaintenance mode if a correct password is entered in the parameter 'Password'.
A new password is valid, after it has been confirmed within the parameter 'Confirm new
password'.
Any new password must consist of at least 4 and a maximum of 16 characters and cancontain letters and numbers.

User entry

Character string comprising numbers, letters and special characters (16)

Confirm new passwordNavigation

Guidance → Commissioning → User management → Confirm new password

Description

Enter the new password again to confirm.

User entry

Character string comprising numbers, letters and special characters (16)

Status password entry
NavigationDescription

Use this function to display the status of the password verification.

20

Guidance → Commissioning → User management → Status password entry

Endress+Hauser

iTEMP TMT182B"Guidance" menu

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

Factory setting--------

3.2"Import / Export" submenu

Navigation

Guidance → Import / Export

Create configuration report
Navigation

Guidance → Import / Export → Create configuration report

Description

Generates the configuration report in the PDF format. This report documents the device
configuration.

User entry

Clicking the Create configuration report button enables a configuration report to begenerated. This report can be printed out or saved in PDF format.

Save / Restore

Navigation

Guidance → Import / Export → Save / Restore

Description• The device settings can be saved with 'Save'.
• Saved device settings can be written to the device with 'Restore'.
In addition, the user can choose whether to upload the data from the device before saving
or to download the data to the device after restoring the settings.

User entry

Endress+Hauser

Clicking the Save/restore button activates the Save and Restore functions.

21

"Diagnostics" menu

iTEMP TMT182B

4"Diagnostics" menu

Troubleshooting and preventive maintenance – settings for device behavior during process
and device events as well as assistance and measures for diagnostic purposes.
Navigation

Diagnostics

Diagnostics‣ Active diagnostics
Actual diagnostics 1→  23Operating time→  23‣ Diagnostic list

→  24
Actual diagnostics 1→  24

Actual diag 1 channel→  24

Timestamp 1→  24‣ Event logbook→  25Previous diagnostics 1→  25Previous diag 1 channel→  25

Timestamp 1→  25‣ Simulation22→  23→  26
Diagnostic event simulation→  26

Current output simulation→  26

Current output value→  26

Sensor simulation→  27

Sensor simulation value→  27

Endress+Hauser

iTEMP TMT182B"Diagnostics" menu

‣ Diagnostic settings
Alarm delay→  27

Thermocouple diagnostic→  28‣ Minimum/maximum values

4.1→  27→  28Sensor min value→  28Sensor max value→  28Reset sensor min/max values→  29Device temperature min value→  29Device temperature max value→  29Reset device temp. min/max values→  29"Active diagnostics" submenu

Navigation

Diagnostics → Active diagnostics

Actual diagnostics 1
Navigation

Diagnostics → Active diagnostics → Actual diagnostics 1

Description

Use this function to display the current diagnostics message. If several messages arepresent at the same time, the message with the highest priority is displayed.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Operating time
NavigationDescription

Use this function to display the length of time the device has been in operation up to now.

User interface

Hours (h)Endress+Hauser

Diagnostics → Active diagnostics → Operating time

23

"Diagnostics" menu

iTEMP TMT182B

4.2"Diagnostic list" submenu

Navigation

Diagnostics → Diagnostic list

Actual diagnostics 1 to 3
Navigation

Diagnostics → Diagnostic list → Actual diagnostics 1 to 3

Description

Use this function to display the current diagnostics message. If several messages arepresent at the same time, the message with the highest priority is displayed.

User interface

Example: F041 Sensor interrupted

Actual diag 1 to 3 channel
Navigation

Diagnostics → Diagnostic list → Actual diag 1 to 3 channel

Description

Use this function to display the source (HW- or SW module) of the corresponding
diagnostics message.

User interface• -------• Sensor• Device temperature• Current output• Sensor RJ

Factory setting--------

Timestamp 1 to 3
NavigationDescription

Displays the time stamp of the previous diagnostic message in relation to the operating
time.

User interface

Hours (h)24Diagnostics → Diagnostic list → Timestamp 1 to 3

Endress+Hauser

iTEMP TMT182B"Diagnostics" menu

4.3

"Event logbook" submenu

Navigation

Diagnostics → Event logbook

Previous diagnostics 1 to 10
Navigation

Diagnostics → Event logbook → Previous diagnostics 1 to 10

Description

Use this function to display historic diagnostic messages.

User interface

Example: F041 Sensor interrupted

Previous diag 1 to 10 channel
Navigation

Diagnostics → Event logbook → Previous diag 1 to 10 channel

Description

Use this function to display the channel of a historic diagnostic message.

User interface• -------• Sensor• Device temperature• Current output• Sensor RJ

Factory setting--------

Timestamp 1 to 10
NavigationDescription

Displays the time stamp of the previous diagnostic message in relation to the operating
time.

User interface

Hours (h)Endress+Hauser

Diagnostics → Event logbook → Timestamp 1 to 10

25

"Diagnostics" menu

iTEMP TMT182B

4.4"Simulation" submenu

Navigation

Diagnostics → Simulation

Diagnostic event simulation



Navigation

Diagnostics → Simulation → Diagnostic event simulation

Description

Select the diagnostic event to be simulated.
Note:
To terminate the simulation, select "Off".

User entry

Enter one of the diagnostic events using the drop-down menu. The assigned status signals
and diagnostic behaviors are used in the simulation mode.
Example: x043 Short circuit

Factory setting

Off

Current output simulation



Navigation

Diagnostics → Simulation → Current output simulation

Description

Use this function to switch simulation of the current output on and off. The display
alternates between the measured value and a diagnostics message of the "function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Current output value

NavigationDescription

Use this function to set a current value for the simulation. In this way, users can verify thecorrect adjustment of the current output and the correct function of downstream switchingunits.

User entry

3.58 to 23.0 mA

Factory setting

3.58 mA

26

Diagnostics → Simulation → Current output value

Endress+Hauser

iTEMP TMT182B"Diagnostics" menu

Sensor simulation



Navigation

Diagnostics → Simulation → Sensor simulation

Description

Use this function to activate the simulation of the process variable. The simulated value
can be set with parameter "Sensor simulation".

Selection

• Off
• On

Factory setting

Off

Sensor simulation value

Navigation

Diagnostics → Simulation → Sensor simulation value

Description

Use this function to enter a simulation value of the process variable. Subsequent measured
value processing and the signal output use this simulation value. In this way, users canverify whether the measuring device has been configured correctly.

User entry

Signed floating-point number

Factory setting

0.0

4.5

"Diagnostic settings" submenu

Navigation

Diagnostics → Diagnostic settings

Alarm delay

NavigationDescription

Use this function to set the delay time during which a diagnostics signal is suppressed
before it is output.

User entry

0 to 5 s

Factory setting

2sEndress+Hauser

Diagnostics → Diagnostic settings → Alarm delay

27

"Diagnostics" menu

iTEMP TMT182B

Thermocouple diagnostic

Navigation

Diagnostics → Diagnostic settings → Thermocouple diagnostic

Description

Use this function to turn off the 'Sensor corrosion' and 'Sensor interrupted' diagnostic
functions of the thermocouple measurement. This may be necessary to allow theconnection of electronic simulators (e.g. calibrators) during thermocouple measurement.
The measurement accuracy of the transmitter is not affected when the thermocouple
diagnostics are turned on or off.

Selection

• Off
• On

Factory setting

On

4.6

"Minimum/maximum values" submenu

Navigation

Diagnostics → Minimum/maximum values

Sensor min value
Navigation

Diagnostics → Minimum/maximum values → Sensor min value

Description

Use this function to display the minimum temperature measured in the past at sensor
input (peakhold indicator).

User interface

Signed floating-point number

Factory setting

Positive floating-point number

Sensor max value
NavigationDescription

Use this function to display the maximum temperature measured in the past at sensor
input (maximum indicator).

User interface

Signed floating-point number

Factory setting

Negative floating-point number

28

Diagnostics → Minimum/maximum values → Sensor max value

Endress+Hauser

iTEMP TMT182B"Diagnostics" menu

Reset sensor min/max values



Navigation

Diagnostics → Minimum/maximum values → Reset sensor min/max values

Description

Reset the maximum indicators for the minimum and maximum temperatures measured at
the sensor inputs.

User entry

Clicking the Reset sensor min/max values button activates the reset function. As a resultof this action, the min/max values of the sensor only display the reset, temporary values.

Device temperature min value
Navigation

Diagnostics → Minimum/maximum values → Device temperature min value

Description

Displays the minimum electronics temperature measured in the past (minimum indicator).

User interface

Signed floating-point number

Factory setting

Positive floating-point number

Device temperature max value
Navigation

Diagnostics → Minimum/maximum values → Device temperature max value

Description

Use this function to display the maximum electronics temperature measured in the past(maximum indicator).

User interface

Signed floating-point number

Factory setting

Negative floating-point number

Reset device temp. min/max values



NavigationDescription

Resets the maximum indicators for the minimum and maximum electronic temperatures
measured.

User entry

Clicking the Reset device temperature min/max values button activates the resetfunction. As a result of this action, the min/max values of the device temperature only
display the reset, temporary values.

Endress+Hauser

Diagnostics → Minimum/maximum values → Reset device temp. min/max values

29

"Application" menu

iTEMP TMT182B

5"Application" menu

Targeted optimization to the application – comprehensive device settings from sensor
technology to system integration for optimum application adaptation.
Navigation

Application

Application‣ Measured values→  31
Sensor value→  31

Sensor raw value→  32

Output current→  32

Percent of range→  32

Device temperature→  32PV→  32SV→  33TV→  33QV→  33‣ Sensor→  34‣ Sensor→  34Unit→  34

Sensor type→  34

Connection type→  352-wire compensation→  35

Reference junction→  36

RJ preset value→  36

Sensor offset→  36‣ Linearization→  37Call./v. Dusen coeff. R030→  37

Endress+Hauser

iTEMP TMT182B"Application" menu

Call./v. Dusen coeff. A→  37Call./v. Dusen coeff. B→  37Call./v. Dusen coeff. C→  38Polynomial coeff. R0→  38Polynomial coeff. A→  38Polynomial coeff. B→  39Sensor lower limit→  39Sensor upper limit→  39‣ Current output→  40Lower range value output→  40Upper range value output→  40

Failure mode→  40Current trimming 4 mA

→  41

Current trimming 20 mA→  41

Damping→  41‣ HART configuration→  42

HART address→  42

No. of preambles→  42

5.1"Measured values" submenu

Navigation

Application → Measured values

Sensor value
NavigationDescription

Use this function to display the current measured value at the sensor input.

Endress+Hauser

Application → Measured values → Sensor value

31

"Application" menu

User interface

iTEMP TMT182B

Signed floating-point number

Sensor raw value
Navigation

Application → Measured values → Sensor raw value

Description

Use this function to display the non-linearized mV/Ohm value at the specific sensor input.

User interface

Signed floating-point number

Output current
Navigation

Application → Measured values → Output current

Description

Use this function to view the calculated output current in mA.

User interface

3.58 to 23.0 mA

Percent of range
Navigation

Application → Measured values → Percent of range

Description

Use this function to display the measured value in % of the span.

User interface

Signed floating-point number

Device temperature
Navigation

Application → Measured values → Device temperature

Description

Use this function to display the current electronics temperature.

User interface

Signed floating-point number

PV
NavigationDescription

Use this function to display the primary HART value.

32

Application → Measured values → PV

Endress+Hauser

iTEMP TMT182B

User interface"Application" menu

Signed floating-point number

SV
Navigation

Application → Measured values → SV

Description

Use this function to display the secondary HART value.

User interface

Signed floating-point number

TV
Navigation

Application → Measured values → TV

Description

Use this function to display the tertiary HART value.

User interface

Signed floating-point number

QV
NavigationDescription

Use this function to display the quaternary (fourth) HART value.

User interface

Signed floating-point number

Endress+Hauser

Application → Measured values → QV

33

"Application" menu

iTEMP TMT182B

5.2"Sensor" submenu

Navigation

5.2.1



Application → Sensor"Sensor" submenu

Navigation

Application → Sensor → Sensor

Unit



Navigation

Application → Sensor → Sensor → Unit

Description

Selection of the unit for all measured values.

Selection

SI units
• °C
•K
• Ohm
Custom-specific units
• °F
• mV

Factory setting°C

Additional information

Please note: If another unit has been selected instead of the factory setting (°C), all the set
temperature values are converted to correspond to the configured temperature unit.
Example: 150 °C is set as the upper range value. Following the selection of °F as theengineering unit, the new (converted) upper range value = 302 °F.

Sensor type

Navigation

Application → Sensor → Sensor → Sensor type

Description

Use this function to select the sensor type for the sensor input.
Info:
Please observe the terminal assignment when connecting the sensor.

Selection

34• Pt100 IEC60751, a=0.00385 (1)• Pt200 IEC60751, a=0.00385 (2)• Pt500 IEC60751, a=0.00385 (3)• Pt1000 IEC60751, a=0.00385 (4)• Pt100 JIS C1604, a=0.003916 (5)• Ni100 DIN 43760, a=0.00618 (6)• Ni120 DIN 43760, a=0.00618 (7)• Ni100 OIML/GOST 6651-09, a=0.00617 (12)• Ni120 OIML/GOST 6651-09, a=0.00617 (13)Endress+Hauser

iTEMP TMT182B"Application" menu

• Type A (W5Re-W20Re) IEC60584 (30)
• Type B (PtRh30-PtRh6) IEC60584 (31)
• Type C (W5Re-W26Re) IEC60584 (32)
• Type D (W3Re-W25Re) ASTM E988-96 (33)
• Type E (NiCr-CuNi) IEC60584 (34)
• Type J (Fe-CuNi) IEC60584 (35)
• Type K (NiCr-Ni) IEC60584 (36)
• Type N (NiCrSi-NiSi) IEC60584 (37)
• Type R (PtRh13-Pt) IEC60584 (38)
• Type S (PtRh10-Pt) IEC60584 (39)
• Type T (Cu-CuNi) IEC60584 (40)
• Type L (Fe-CuNi) DIN43710 (41)
• Type L (NiCr-CuNi) GOST R8.585-01 (43)
• Type U (Cu-CuNi) DIN43710 (42)
• Pt50 GOST 6651-94, a=0.00391 (8)• Pt100 GOST 6651-94, a=0.00391 (9)• Cu50 GOST 6651-09, a=0.00428 (10)• Cu100 OIML/GOST 6651-09, a=0.00428 (11)• Cu50 OIML R84:2003, a=0.00428 (10)• Cu50 OIML/GOST 6651-94, a=0,00426 (14)• RTD Platinium (Callendar/van Dusen)• RTD Poly Nickel (OIML R84, GOST 6651-94)• RTD Polynomial Copper (OIML R84:2003)• 10...400 Ohm• 10...2000 Ohm• -20...100 mV
Factory setting

Pt100 IEC60751, a=0.00385 (1)Connection type

Navigation

Application → Sensor → Sensor → Connection type

Prerequisite

An RTD sensor or a resistance transmitter must be specified as the sensor type.

Description

Use this function to select the connection type for the sensor.

Selection• 2- wire• 3- wire• 4- wire

Factory setting

4- wire2-wire compensation



Navigation

Prerequisite

An RTD sensor or a resistance transmitter with a 2-wire connection type must be specified
as the sensor type.

Description

Use this function to specify the resistance value for two-wire compensation in RTDs.

Endress+Hauser

Application → Sensor → Sensor → 2-wire compensation

35

"Application" menu

iTEMP TMT182B

User entry

0.0 to 30.0 Ohm

Factory setting

0.0 Ohm

Reference junction



Navigation

Application → Sensor → Sensor → Reference junction

Prerequisite

A thermocouple (TC) sensor must be selected as the sensor type.

Description

Use this function to select reference junction measurement for temperature compensationof thermocouples (TC).
Additional information:
If "Fixed value" is selected, the compensation value is specified via the RJ preset value
parameter.

Selection• Internal measurement• Fixed Value

Factory setting

Internal measurement

RJ preset value

Navigation

Application → Sensor → Sensor → RJ preset value

Prerequisite

The Fixed value parameter must be set if the Reference junction option is selected.

Description

The Fixed value parameter must be set if the Reference junction n option is selected.

User entry–58.0 to 360.0

Factory setting

0.0

Sensor offset

NavigationDescription

Use this function to set the zero point correction (offset) of the sensor measured value.
The value indicated is added to the measured value.

User entry–18.0 to 18.0

Factory setting

0.0

36

Application → Sensor → Sensor → Sensor offset

Endress+Hauser

iTEMP TMT182B"Application" menu

5.2.2

"Linearization" submenu

Navigation

Application → Sensor → Linearization

Call./v. Dusen coeff. R0

Navigation

Application → Sensor → Linearization → Call./v. Dusen coeff. R0

Prerequisite

The RTD platinum (Callendar/Van Dusen) option is enabled in the Sensor type parameter.

Description

Use this function to set the R0 value for sensor linearization with the Callendar/Van Dusen
polynomial.

User entry

10.0 to 2 000.0 Ohm

Factory setting

100.0 Ohm

Call./v. Dusen coeff. ANavigation

Application → Sensor → Linearization → Call./v. Dusen coeff. A

Prerequisite

The RTD platinum (Callendar/Van Dusen) option is enabled in the Sensor type parameter.

Description

Use this function to set the coefficients for sensor linearization with the Callendar/Van
Dusen polynomial.

User entry

0.003 to 0.004

Factory setting

0.0039083

Call./v. Dusen coeff. B

Navigation

Prerequisite

The RTD platinum (Callendar/Van Dusen) option is enabled in the Sensor type parameter.

Description

Use this function to set the coefficients for sensor linearization with the Callendar/Van
Dusen polynomial.

User entry–2.0 · 10-06 to 2.0 · 10-06

Factory setting–5.775E-07

Endress+Hauser

Application → Sensor → Linearization → Call./v. Dusen coeff. B

37

"Application" menu

iTEMP TMT182B

Call./v. Dusen coeff. CNavigation

Application → Sensor → Linearization → Call./v. Dusen coeff. C

Prerequisite

The RTD platinum (Callendar/Van Dusen) option is enabled in the Sensor type parameter.

Description

Use this function to set the coefficients for sensor linearization with the Callendar/Van
Dusen polynomial.

User entry–1.0 · 10-09 to 1.0 · 10-09

Factory setting–4.183E-12

Polynomial coeff. R0

Navigation

Application → Sensor → Linearization → Polynomial coeff. R0

Prerequisite

The RTD poly nickel or RTD copper polynomial option is enabled in the Sensor typeparameter.

Description

Use this function to set the R0 value for linearization of nickel/copper sensors.

User entry

10.0 to 2 000.0 Ohm

Factory setting

100.0 Ohm

Polynomial coeff. ANavigation

Prerequisite

The RTD poly nickel or RTD copper polynomial option is enabled in the Sensor typeparameter.

Description

Use this function to set the coefficients for sensor linearization of copper/nickel resistancethermometers.

User entry

0.004 to 0.006

Factory setting

0.0054963

38

Application → Sensor → Linearization → Polynomial coeff. A

Endress+Hauser

iTEMP TMT182B"Application" menu

Polynomial coeff. B

Navigation

Application → Sensor → Linearization → Polynomial coeff. B

Prerequisite

The RTD poly nickel or RTD copper polynomial option is enabled in the Sensor typeparameter.

Description

Use this function to set the coefficients for sensor linearization of copper/nickel resistancethermometers.

User entry–2.0 · 10-05 to 2.0 · 10-05

Factory setting

6.7556E-06

Sensor lower limitNavigation

Application → Sensor → Linearization → Sensor lower limit

Prerequisite

The RTD platinum, RTD poly nickel or RTD copper polynomial option is enabled in the
Sensor type parameter.

Description

Use this function to set the lower calculation limit for special sensor linearization.

User entry–10 000.0 to 10 000.0

Factory setting

Depends on the sensor type selected.

Sensor upper limitNavigation

Prerequisite

The RTD platinum, RTD poly nickel or RTD copper polynomial option is enabled in the
Sensor type parameter.

Description

Use this function to set the upper calculation limit for special sensor linearization.

User entry–10 000.0 to 10 000.0

Factory setting

Depends on the sensor type selected.

Endress+Hauser

Application → Sensor → Linearization → Sensor upper limit

39

"Application" menu

iTEMP TMT182B

5.3"Current output " submenu

Navigation

Application → Current output

Lower range value output

Navigation

Application → Current output → Lower range value output

Description

Use this function to assign a measured value to the current value 4 mA.
Additional information:
The set point that can be set depends on the sensor type used in the Sensor typeparameter.

User entry–50 000.0 to 50 000.0

Factory setting

0.0Upper range value output

Navigation

Application → Current output → Upper range value output

Description

Use this function to assign a measured value to the current value 20 mA.
Additional information:
The set point that can be set depends on the sensor type used in the Sensor typeparameter.

User entry–50 000.0 to 50 000.0

Factory setting

100.0

Failure modeNavigationDescription

Use this function to select the signal on alarm level of the current output in the event of anerror.

Selection• Max.
• Min.

Factory setting

Min.

40

Application → Current output → Failure modeEndress+Hauser

iTEMP TMT182B"Application" menu

Current trimming 4 mA



Navigation

Application → Current output → Current trimming 4 mA

Description

Use this function to set the correction value for the current output at the start of themeasuring range at 4 mA.

User entry

3.85 to 4.15 mA

Factory setting

4.0 mA

Additional information

The trimming only affects the current loop values from 3.8 to 20.5 mA. A failure modewith min. and max. current values is not subject to trimming.

Current trimming 20 mA



Navigation

Application → Current output → Current trimming 20 mA

Description

Use this function to set the correction value for the current output at the end of themeasuring range at 20 mA.

User entry

19.85 to 20.15 mA

Factory setting

20.0 mA

Additional information

The trimming only affects the current loop values from 3.8 to 20.5 mA. A failure modewith min. and max. current values is not subject to trimming.

DampingNavigationDescription

Use this function to set the time constant for the damping of the measured value.

User entry

0 to 120 s

Factory setting

0s

Additional information

The current output responds to fluctuations in the measured value with an exponential
delay. The time constant of this delay is defined by this parameter. If a low time constant isentered, the current output responds quickly to the measured value. On the other hand,
the response of the current output is delayed significantly if a high time constant isentered.

Endress+Hauser

Application → Current output → Damping

41

"Application" menu

iTEMP TMT182B

5.4"HART configuration" submenu

Navigation

Application → HART configuration

HART addressNavigation

Application → HART configuration → HART address

Description

Enter the address via which the data exchange takes place via HART protocol.
Additional information:
In FDT/DTM based operating tools, e.g. FieldCare, DeviceCare from Endress+Hauser the
HART address can be set via the communication DTM.
Only with address "0" a measured value transmission via the current value is possible. For
all other addresses, the current is fixed to 4.0 mA (multidrop mode).

User entry

0 to 63

Factory setting

0

No. of preambles



NavigationDescription

Use this function to define the number of preambles in the HART telegram

User entry

5 to 20

Factory setting

542

Application → HART configuration → No. of preambles

Endress+Hauser

iTEMP TMT182B"System" menu

6

"System" menu

Overall device management and security settings – management of system settings andadaption to operational requirements.
Navigation



SystemSystem‣ Device management→  45

Device tag→  45

HART short tag→  45

Device reset→  45

Configuration counter→  46

Configuration changed→  46

Reset configuration changed flag

→  46‣ User management→  47‣ User management
Access status‣ Delete password→  47→  48

Status password entry‣ Enter password→  48→  48Password→  48

Status password entry→  48‣ Define password

Endress+Hauser→  47→  49New password→  49Confirm new password→  49Status password entry→  5043

"System" menu

iTEMP TMT182B‣ Recover password
Reset password→  50Status password entry→  50‣ Change password→  51

Old password→  51

New password→  51

Confirm new password→  51

Status password entry→  52‣ Information→  52‣ Device→  52
Serial number→  52

Order code→  53

Firmware version→  53

Hardware revision→  53Extended order code 1→  53

Device name→  54

Manufacturer→  54‣ HART info44

→  50→  54
Device type→  54

Device revision→  55

HART revision→  55

HART descriptor→  55

HART message→  55

Hardware revision→  56

Software revision→  56Endress+Hauser

iTEMP TMT182B"System" menu

6.1

HART date code→  56

Manufacturer ID→  56

Device ID→  56"Device management" submenu

Navigation



System → Device management

Device tagNavigation



System → Device management → Device tag

Description

Displays the name for the measuring point.

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

Depends on the product root and serial number
EH_TMT182B_serial number

HART short tag

Navigation



System → Device management → HART short tag

Description

Defines the short tag for the measuring point.
Maximum length: 8 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Character string comprising numbers, letters and special characters (8)Factory setting????????

Device reset

NavigationDescription

Use this function to reset the device configuration - either entirely or in part - to a definedstate.

Endress+Hauser

System → Device management → Device reset

45

"System" menu

iTEMP TMT182B

Selection• Not active• Restart device• To delivery settings• To factory defaults

Factory setting

Not active

Configuration counter
Navigation



System → Device management → Configuration counter

Description

Use this function to display the counter reading for changes to device parameters.
Info:
Static parameters, whose values change during optimization or configuration, cause thisparameter to increment by 1. This supports parameter version management. If several
parameters change, e. g. due to loading of parameters from FieldCare, etc. in the device,
the counter can show a higher value. The counter can never be reset and is not reset to adefault value after a device reset. If the counter overflows, (16 bit), it starts again at 1.

User interface

0 to 65 535

Factory setting0

Configuration changed
Navigation



System → Device management → Configuration changed

Description

Indicates whether the configuration of the device has been changed by a master (primaryor secondary).

User interface• Primary master• Secondary master

Reset configuration changed flag
NavigationDescription

The Configuration changed information is reset by a master (primary or secondary)

User entry

Clicking the button resets the information.

46

System → Device management → Reset configuration changed flag

Endress+Hauser

iTEMP TMT182B"System" menu

6.2

"User management" submenu

Logout →Maintenance

Switch to 'Operator' access authorization

Change user role →Operator

Password
Status password entry

Forgot password? →Operator

Reset password
Status password entry

Change password →MaintenanceOld password
New password
Confirm new password
Status password entry

Delete password →MaintenanceStatus password entry

Define password →MaintenanceNew password
Confirm new password
Status password entry

Navigation in the submenu is supported by the following operating elements:• BackReturn to the previous page
• Cancel
If Cancel is selected, the status before the submenu was started is restored
Navigation

6.2.1



System → User management"User management" submenu

Navigation



System → User management → User management

Access status
NavigationDescription

If additional write protection is active, this restricts the current access authorization evenfurther.

User interface• Operator• MaintenanceFactory setting

MaintenanceEndress+Hauser

System → User management → User management → Access status

47

"System" menu

iTEMP TMT182B

6.2.2"Delete password" submenu

Navigation



System → User management → Delete password

Status password entry
Navigation



System → User management → Delete password → Status password entry

Description

Use this function to display the status of the password verification.

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

Factory setting--------

6.2.3"Enter password" submenu

Navigation



System → User management → Enter password

Password
Navigation



System → User management → Enter password → Password

Description

Enter the password for the 'Maintenance' user role to get access to the functionality of thisrole.

User entry

Character string comprising numbers, letters and special characters (16)

Status password entry
NavigationDescription

Use this function to display the status of the password verification.

48

System → User management → Enter password → Status password entry

Endress+Hauser

iTEMP TMT182B"System" menu

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

Factory setting--------

6.2.4"Define password" submenu

Navigation



System → User management → Define password

New passwordNavigation



System → User management → Define password → New password

Description

If the factory setting is not changed, the device works without write-protection, using
userrole 'Maintenance'. The configuration data of the device can always be modified.
Once the password has been defined, write-protected devices can only be set tomaintenance mode if a correct password is entered in the parameter 'Password'.
A new password is valid, after it has been confirmed within the parameter 'Confirm new
password'.
Any new password must consist of at least 4 and a maximum of 16 characters and cancontain letters and numbers.

User entry

Character string comprising numbers, letters and special characters (16)

Confirm new passwordNavigationDescription

Enter the new password again to confirm.

User entry

Character string comprising numbers, letters and special characters (16)Endress+Hauser

System → User management → Define password → Confirm new password

49

"System" menu

iTEMP TMT182B

Status password entry
Navigation



System → User management → Define password → Status password entry

Description

Use this function to display the status of the password verification.

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

Factory setting--------

6.2.5"Recover password" submenu

Navigation



System → User management → Recover password

Reset password
Navigation



System → User management → Recover password → Reset password

Description

Enter a code to reset the current password.
CAUTION: Use this function only if the current password is lost. Contact your Endress+Hauser Sales Center.

User entry

Character string comprising numbers, letters and special characters (16)

Status password entry
NavigationDescription

Use this function to display the status of the password verification.

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch

50

System → User management → Recover password → Status password entry

Endress+Hauser

iTEMP TMT182B"System" menu

• Reset password accepted• Invalid user role
• Wrong sequence of entry
Factory setting--------

6.2.6"Change password" submenu

Navigation



System → User management → Change password

Old passwordNavigation



System → User management → Change password → Old password

Description

Enter the current password, to subsequently change the existing password.

User entry

Character string comprising numbers, letters and special characters (16)

New passwordNavigation



System → User management → Change password → New password

Description

If the factory setting is not changed, the device works without write-protection, using
userrole 'Maintenance'. The configuration data of the device can always be modified.
Once the password has been defined, write-protected devices can only be set tomaintenance mode if a correct password is entered in the parameter 'Password'.
A new password is valid, after it has been confirmed within the parameter 'Confirm new
password'.
Any new password must consist of at least 4 and a maximum of 16 characters and cancontain letters and numbers.

User entry

Character string comprising numbers, letters and special characters (16)

Confirm new passwordNavigationDescription

Enter the new password again to confirm.

User entry

Character string comprising numbers, letters and special characters (16)Endress+Hauser

System → User management → Change password → Confirm new password

51

"System" menu

iTEMP TMT182B

Status password entry
Navigation



System → User management → Change password → Status password entry

Description

Use this function to display the status of the password verification.

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

Factory setting--------

6.3"Information" submenu

Navigation

6.3.1



System → Information"Device" submenu

Navigation



System → Information → Device

Serial number
Navigation



System → Information → Device → Serial number

Description

Displays the serial number of the measuring device. The serial number can be used to
identify the measuring device and to retrieve further information via the Device Viewer or
Operations app, such as the related documentation.
Additional information:
The serial number can also be found on the nameplate of the sensor and transmitter.

User interface

52

Character string comprising numbers, letters and special characters

Endress+Hauser

iTEMP TMT182B"System" menu

Order code
Navigation



System → Information → Device → Order code

Description

Displays the device order code.
Additional information:
The order code can be used for instance to order a replacement or spare device or to verify
that the device features specified on the order form match the shipping note.

User interface

Character string comprising numbers, letters and special characters

Firmware version
Navigation



System → Information → Device → Firmware version

Description

Use this function to view the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

Hardware revision
Navigation



System → Information → Device → Hardware revision

Description

Use this function to display the hardware revision of the device.

User interface

Character string comprising numbers, letters and special characters

Extended order code 1 to 3
Navigation



System → Information → Device → Extended order code 1 to 3

Description

Displays the first, second and/or third part of the extended order code. Due to character
length restrictions, the extended order code is split into a maximum of 3 parameters. The
extended order code indicates for each feature in the product structure the selected option,
thereby uniquely identifying the device model.
Additional information:
The extended order code can also be found on the nameplate.

User interface

Endress+Hauser

Character string comprising numbers, letters and special characters

53

"System" menu

iTEMP TMT182B

Device name
Navigation



System → Information → Device → Device name

Description

Displays the name of the transmitter.
Additional information:
The name can also be found on the transmitter's nameplate.

User interface

Character string comprising numbers, letters and special characters

Factory setting

iTEMP TMT182B

Manufacturer
Navigation



System → Information → Device → Manufacturer

Description

Displays the manufacturer.

User interface

Character string comprising numbers, letters and special characters

Factory setting

Endress+Hauser

6.3.2"HART info" submenu

Navigation



System → Information → HART info

Device type
NavigationDescription

Use this function to view the device type with which the device is registered with the HART
FieldComm Group. The device type is specified by the manufacturer. It is needed to assign
the appropriate device description file (DD) to the device.

User interface

0x11D2

54

System → Information → HART info → Device type

Endress+Hauser

iTEMP TMT182B"System" menu

Device revision
Navigation



System → Information → HART info → Device revision

Description

Use this function to view the device revision with which the device is registered with the
HART® FieldComm Group. It is needed to assign the appropriate device description file(DD) to the device.

User interface

0x01

HART revision
Navigation



System → Information → HART info → HART revision

Description

Display the HART protocol revision of the measuring device

User interface

7

HART descriptor

Navigation



System → Information → HART info → HART descriptor

Description

Maximum length: 16 characters; permitted characters: A-Z, 0-9, certain special characters.

User entry

Character string comprising numbers, letters and special characters (16)

Factory setting????????????????

HART message

Navigation



System → Information → HART info → HART message

Description

Use this function to define a HART message which is sent via the HART protocol whenrequested by the master.
Maximum length: 32 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting????????????????????????????????

Endress+Hauser

55

"System" menu

iTEMP TMT182B

Hardware revision
Navigation



System → Information → HART info → Hardware revision

Description

Use this function to display the hardware revision of the device.

User interface

0x01

Software revision
Navigation



System → Information → HART info → Software revision

Description

Displays the software revision of the measuring device

User interface

0x01

HART date code

Navigation



System → Information → HART info → HART date code

Description

Use this function to define date information for individual use. Date in the format yearmonth-day (YYYY-MM-DD)

User entry

The date can be saved here in the predefined format.

Manufacturer ID
Navigation



System → Information → HART info → Manufacturer ID

Description

Use this function to view the manufacturer ID with which the device is registered with theFieldComm Group.

User interface

0x0011

Device ID
NavigationDescription

Shows the device ID for identifying the device in a HART network. In addition to the device
type and manufacturer ID, the device ID is part of the unique ID. Each HART device isuniquely identified by the unique device ID.

56

System → Information → HART info → Device ID

Endress+Hauser

iTEMP TMT182B

User interface

Endress+Hauser"System" menu

Positive integer57

*71596800*
71596800

www.addresses.endress.com