HAND TACHOMETER MANUAL

TM-5000/5010/E

HAND TACHOMETER INSTRUCTION MANUAL

Please note that misuse of this device
may lead to injury to the user ordamage to the device.

A t t e n t io n

In-contact Adaptor
Remote Sensor

Customer Service

TM-5000/5010/E TM-5013 TM-4015
Thank you for purchasing our hand tachometer.
Please read this instruction manual carefully before using toensure the correct usage of this device.
Please keep this instruction manual for future reference.

Safety P reca u tio n s

Address:
Contact :E-mail :Web:

LINE SEIKI CO., LTD.
37-7 Chuo-cho, 2-Chome
Meguro-ku, Tokyo 152-0001 Japan
Please visit our company website for contact details
webtrade@line.co.jp
https://www.lineseiki.com

1. Preparations

For safe usage of this device, please observe all statements regarding
precautions and warnings in this instruction manual.

Attention

1. Installing the batteries
1) Press and slide down the battery cover at the back side of the main unit.
2) Install the four batteries (AAA 1.5V) noting the polarity as indicated inside.
3) After installing the batteries, replace and slide back the cover.
2. Using the reflective tape (for non-contact type)Cut off a length of 1 to 3 cm. of the reflective tape. Remove the kraft paper
backing and stick the reflective tape on the object whose rotational speed is to bemeasured.
Wipe off oils or stains from the surface where the reflective tape will be attached.
Stick the reflective tape as near as possible to the outer edge of the object whoserotational speed is to be measured.

Do not use this device near machines that emit strong
electromagnetic fields or objects that store static electricity.
Do not drop this device or submit it to strong impact.
Do not use or store the device where it can be exposed todirect sunlight, dust, high temperature and high humidity.
Inspect the device before conducting measurements.
Use the battery case to ensure that the battery is properly
installed.
Remove the battery when the device will not be used for along period of time.
Do not attempt to disassemble or modify this device.

3. Installing the In-contact Adaptor (for contact type)
Loosen the lock screw at the back of the unit, insert the in-contact adaptor to theupper side of the main unit and tighten the lock screw using a screw driver. Use acoin if a screw driver is not available. The lock screw must be tightened securely
to ensure that the in-contact adaptor does not fly off while in use.
4. Installing the remote sensor
Set the switch in the battery compartment to off position. Insert the connector of
the remote sensor to the modular telephone style connector on the left hand side.

2.Components and Functions
Surface speed wheel (circumference 1/10 m or 1/10 y)

Rubber Tip

Combination of adaptor with TM-5000 or TM-5010 is available.
For TM-5000, display is 1/10 m/min.
For TM-5010/E, calculations are done internally to display
the units m/min, cm/sec, f/min, inch/sec and y/min,
which can be read directly.

Remote Sensor

In-contact adaptor
In-contact measurement or surface speed measurement is possible using therubber tip or the surface speed wheel in combination with TM-5000 or TM-5010.
Attention!

When using at high speed, be very careful that the portion being measured don't get dragged on.

Photoelectric part
Unit display

rotational speed mode(r/min r/s)

Memory number display
r/minPhotoelectric detector indicator

f/min*inch/s* y/min*cm/s m/min r/s r/min count
MAX MEMMIN

rotation count and length
measurement mode
(count m y* inch* f *)Rotational / Linear speed
Units Group selection key

Measurement display
Measurement button

While it is pressed, measurement is performed.

Connector for remote sensor probe
Measurement mode switch
Connector for remote sensor probe
The remote sensor can be used byturning the slide switch inside the battery
box located at the back of this unit.

MEASMAX MIN

By pressing this button, display changes as follows:Measurement value > MAX value > MIN value
TM-5000

LINE SEIKI

TM-5000

8-point memory data
input key
8-point memory data
read/clear key

linear speed mode
(m/min cm/s y/min* inch/s*f/min* ).../min.../sCALL
CLR

TM-5010LINE SEIKI

TM-5010 / TM-5010E

Unit selection key
r/minr/s
m/min cm/s y/min*
f/min*
countf*inch/s*
my*inch** TM-5010E Only

HAND TACHOMETER MANUAL
3. Measurement

1) Press the measurement button to turn the unit on, the LCD will then light up.
Only while the measurement button is pressed, measurement is performed.
Press the measurement button again to reset the previous value on thedisplay and perform the measurement again when measuring the numberof revolutions (COUNT) and length(m).
(Applicable for both TM-5010 and TM-5010E).
2) To measure, press the measurement button for at least 1 second.
3) When measuring using the reflective tape, the " " mark will appear confirmingthat the photosensor is working.
4) If the measurement is out of range, the "----" mark will appear.
1.Non-contact type measurement
1) The distance between reflective tape and tachometer main unit (or remote sensor)
is 5 to 30 cm. and the angle of incidence of the light is within +/- 30 degrees.
2) Press the measurement button, confirm that the "" mark is displayed on thescreen, then read the rotational speed.
3) When the measurement button is released, the display is held for about 3 mins.
The unit automatically turns off after 3 mins. of inactivity.
2.Contact type measurement
1) Use the rubber tip correctly by touching it perpendicular to the center of the axleof the rotating object under test. To measure the peripheral speed of the items
such as belts, replace the rubber tip with the surface speed wheel.
2) To attach the surface speed wheel, pull out the rubber tip with your hand and press
the surface speed wheel axle so that the boss of the wheel axle is inserted into
the slit of the adaptor axle. Pull the surface speed wheel gently to confirm thatit does not come off.
3) When measuring the circumferential velocity, the surface speed wheel should touch
the measured object so that the surface speed wheel turns in parallel with it.
4) When measuring length, use the surface speed wheel. Length is calculated basedon the circumference of the wheel (1/10 m) multiplied by the rotation count.
3. Auto-power off
After the measurement button is released, the display is held for about 3 minutes
then the unit automatically turns OFF if no further measurements are made duringthat time. (Applicable for both TM-5000 and TM-5010/E)
Note
Make sure that lights coming from a stroboscope and/or fluorescent lights do notreach the photoelectric part and reflective tape attached to the rotating object toprevent measurement errors. Before starting rotation of the object under test,
please confirm that the measurement on the reflective tape is 0.0 rpm. If themeasurement is affected by external light, change the position of the unit.

*4. Functions

1. Changing of Measurement Units (TM-5010/E only)
Changing of "rotational speed mode" to "linear speed mode" to "rotation countand length measurement mode" can be done by pressing the Rotational/Linear
Speed Units Group selection key.
Pressing the Unit Selection key changes the units of rotational speed or linear
speed or count/length into other units of measurement.
After the installation of batteries, the default unit is r/min which will be displayed on
the screen.
2. Maximum / Minimum Values (TM-5010/E onlyrotational speed mode / linear speed mode only)
Pressing the Measurement Mode switch will change the data displayed on thescreen. "MAX" display represents the maximum value and "MIN" represents theminimum value.
The maximum and minimum values are values measured from the start ofmeasurement until the measurement button is released.
Pressing the measurement button clears the previous maximum and minimummeasurement values.
3. Memory (TM-5010/E only)
By pressing the 8 point memory data input key, the measured values and unit canbe stored in the 8 point memory. At each press of the 8 point memory data input
key, data are stored accordingly into No. 1 to No. 8. MEM and the current memory
No. is displayed on the screen. After memory No. 8 is displayed, no additional
data can be inputted.
Pressing the 8 point memory data read/clear key will recall the stored data. Whilethe data are being called, MEM will blink and the memory no. is displayed on thescreen. When the data from a memory no. is called out and the 8 point memory
read/clear key is pressed, ALLCLR will be displayed on the screen. WhenALLCLR is displayed on the screen, pressing the 8 point memory data input key
will clear the memory and the display will return to the measured value.
When the memory is being recalled and ALLCLR is displayed, not pressing any
key for 20 seconds will return the display to the measured value.

5. Battery Replacement

When themark at the left portion of the screen starts to blink, please replace
the batteries.
Please refer to '1.1 Installing the Batteries' for the battery replacement procedure.
Please replace all batteries with new ones.
Be careful not to mix different types of batteries - i.e. manganese batteries andalkaline batteries together.
* The instrument cannot measure accurately after low battery indicator isdisplayed. Please replace the battery immediately.

TM-5000/5010/E
6.Storage
When storing this device, please avoid areas which are very humid, exposed
to direct sunlight, high temperature, subject to vibration, dust, dirt, saline,or organic gases.

SPECIFICATIONS (MAIN UNIT)
TM-5000

TM-5010/E

MeasuringRange

6.0 - 99999.9 rpm

6.0 - 99999.9 rpm 0.65 - 10936.1 y/min*
0.10 - 1666.66 rps 0 - 999999 count0.60 - 9999.99 m/min 0.0 - 99999.9 m
1.00 -16666.6 cm/s 0.00 - 99999.9 y *0.39 - 6561.65 inch/s* 0.00 - 99996.1 inch *1.9 - 32808.3 f/min * 0.0 - 99999.7 f *Resolution

0.1 rpm

1 count, 0.1 rpm fpm m f,others 0.01

Model+_ 0.01% +_ 1 digit rpm m/min(for other units, the conversion accuracy is +_ 0.05% +_ 1 digit)

Accuracy
Sampling Time

1.0 - 10.0 seconds
Display: 6 digits, 7 segment LCD
Battery alarm :mark
Reflective light :markDisplay

Display unit : r/min

Display unit : r/min, r/s, m/min, cm/sy/min,* inch/s,* f/min,* count, m, y,* inch,* f*Memory display : MAX, MIN, MEM1~8

After 3 mins. from last measurement or key operation

Auto Power-off

Measurement data: until the next data is defined

Data Hold Time

Memory data: same as battery life

Measuring
MethodMeasuringDistanceNon-contact measurement using the main unit or with remote sensor (use with reflective tape)Contact measurement using the in-contact adaptor (use with rubber tip, surface speed wheel)

Power Supply

4 pcs. of AAA alkaline battery(continuous measurement of 20 hrs.)

OperatingTemperature
Storage
Temperature
Storage
Humidity
Dimension/
Weight50 - 300mm(using reflective tape)5 - 40 °C Non-Freezing-10 - +60 °C Non-Freezing
35 - 85%RH Non-Condensing

122(H) x 58(46)(W) x 28(D)mm / approx. 130g (including batteries)* TM-5010E Only

MODELS
TM-5000TM-5000 main unit
AAA alkaline battery, 4 pcs.
Reflective tape, 10 pcs.
Instruction manual

TM-5010/E

TM-5010/E main unit
AAA alkaline battery, 4 pcs.
Reflective tape, 10 pcs.
Instruction manual

TM-5013

In-contact adaptor
Rubber tip, 3 pcs.
Surface speed wheel (1/10 m)

TM-5014

In-contact adaptor
Rubber tip, 3 pcs.
Surface speed wheel (1/10 y)

TM-4015

Remote sensor(21mm diameter, maximum cable length 1m)

RoHS
This device conforms to the following EMC regulations:
EN 61326-1 Section 7
EN 61326-1 Section 6
Test was performed without using the remote sensor.
This manual was last revised Aug. 2, 2021.
*Subject to change without prior notice.
All Rights Reserved,Copyright © 2021,LINE SEIKI CO.,LTD.