CERTIFICATE OF CALIBRATION
Customer: DIGIPAS

200 SPECTRUM CENTER DRIVESUITE 300
IRVINE, CA 92618
ANAB AC-2489

PO Number: ED26C

Certificate/SO Number: 1-B5X9X-40-1 Revision 1
Manufacturer: DigiPas USA

As-Found: In Tolerance

Model Number: DWL1500XY

As-Left: In Tolerance

Description: Digital Level
Serial Number: 13A00426

Calibration Date: Feb 09, 2018

ID: NONECalibrated To: Manufacturer Specification
Calibration Procedure: 1-AC80074-2

Transcat Calibration Laboratories have been audited and found in compliance with ISO/IEC 17025:2005. Accredited calibrations performed within the Lab’s Scope ofAccreditation are indicated by the presence of the Accrediting Body’s Logo and Certificate Number. Any measurements on an accredited calibration not covered by thatLab’s Scope of Accreditation are listed in the notes section of the certificate. SCC, NRC, CLAS or ANAB do not guarantee the accuracy of an individual calibration byaccredited laboratories.
Transcat calibrations, as applicable, are performed in compliance with the requirements of the Transcat Quality Manual QAC-P01-000 Revision 1.0, the customer’sPurchase Order and/or Quality Agreement requirements, ISO 9001:2008, ANSI/NCSL Z540.1-1994 (R2002). Complete records of work performed are maintained byTranscat and are available for inspection. Laboratory standards used in the performance of this calibration are listed below.
Transcat documents the traceability of measurements to the SI units through the National Institute of Standards and Technology (NIST), or the National Research Councilof Canada (NRC), or other national measurement institutes (NMI) that are signatories to the CIPM Mutual Recognition Arrangement, or accepted fundamental and/or
natural physical constants, or by the use of specified methods, consensus standards or ratio type measurements. Documentation supporting traceability information isavailable for review upon written request at a Transcat facility. The measured quantity and the measurement uncertainty are required for further dissemination oftraceability.
Uncertainties are reported with a coverage factor k=2, providing a level of confidence of approximately 95%. All calibrations have been performed using processes
having a TUR of 4:1 or better (3:1 for mass calibrations), unless otherwise noted. The Test Uncertainty Ratio (TUR) is calculated in accordance with NCSL International
RP-18. For mass calibrations: Conventional mass referenced to 8.0 g/cm³.
The results in this report relate only to the item calibrated or tested. Recorded calibration data is valid at the time of calibration within the stated uncertainties at theenvironmental conditions noted. The determination of compliance to the specification is specific to the model/serial no./ID no. referenced above based on the tolerances
shown; these tolerances are either the original equipment manufacturers (OEM’s) warranted specifications or the client’s requested specifications. This certificate may
not be reproduced except in full, without the written approval of Transcat. Additional information, if applicable may be included on separate report(s).

Notes:

One or more test points are close to the tolerance limit, however this instrument is not adjustable.

Date Received: February 05, 2018
Service Level : R6
Revised On: Revised on February 20, 2018

Certificate - Page 1 of 3
Reprinted on February 20, 2018

Customer Number:

1-636816-000

OPS-F20-013R1 01/23/2017 FP001R1 10/12/2017

CERTIFICATE OF CALIBRATION
Customer: DIGIPAS

200 SPECTRUM CENTER DRIVESUITE 300
IRVINE, CA 92618
ANAB AC-2489

PO Number: ED26C

Certificate/SO Number: 1-B5X9X-40-1 Revision 1
As Found/As Left Data

DescriptionOOT

TUR

Setpoints

Accuracy

Low Limit

High Limit

As Found / As Left

0.000°±( 0.002 °)-0.0020.002-0.002 °1.7 : 1-0.229°±( 0.002 °)-0.231-0.227-0.230 °1.7 : 1-0.451°±( 0.002 °)-0.453-0.449-0.453 °1.7 : 1

-1.146°±( 0.004 °)-1.150-1.142-1.150 °3.3 : 1-1.500°±( 0.004 °)-1.504-1.496-1.504 °3.3 : 10.000°±( 0.002 °)-0.0020.002

0.002 °

1.7 : 10.229°±( 0.002 °)

0.227

0.231

0.230 °

1.7 : 10.451°±( 0.002 °)

0.449

0.453

0.453 °

1.7 : 1

1.146°±( 0.004 °)

1.142

1.150

1.150 °

3.3 : 11.500°±( 0.004 °)

1.496

1.504

1.503 °

3.3 : 10.000°±( 0.002 °)-0.0020.002-0.002 °1.7 : 1-0.229°±( 0.002 °)-0.231-0.227-0.230 °1.7 : 1-0.451°±( 0.002 °)-0.453-0.449-0.452 °1.7 : 1

-1.146°±( 0.004 °)-1.150-1.142-1.149 °3.3 : 1-1.500°±( 0.004 °)-1.504-1.496-1.504 °3.3 : 10.000°±( 0.002 °)-0.0020.002

0.002 °

1.7 : 10.229°±( 0.002 °)

0.227

0.231

0.231 °

1.7 : 10.451°±( 0.002 °)

0.449

0.453

0.452 °

1.7 : 1

1.146°±( 0.004 °)

1.142

1.150

1.149 °

3.3 : 11.500°±( 0.004 °)

1.496

1.504

1.503 °

3.3 : 1

Angle Measure: X-Axis

Angle MeasureAngle Measure: Y-Axis

Angle Measure

Date Received: February 05, 2018
Service Level : R6
Revised On: Revised on February 20, 2018

Certificate - Page 2 of 3
Reprinted on February 20, 2018

Customer Number:

1-636816-000

OPS-F20-013R1 01/23/2017 FP001R1 10/12/2017

CERTIFICATE OF CALIBRATION
Customer: DIGIPAS

200 SPECTRUM CENTER DRIVESUITE 300
IRVINE, CA 92618
ANAB AC-2489

PO Number: ED26C

Certificate/SO Number: 1-B5X9X-40-1 Revision 1

Traceable Standards
AssetManufacturer

Model Number

Description

Cal Date

Due Date

Traceability Number

Use101131

Regal BeloitSM1-112

Gage Block Set, 112 pcs.

10-Apr-17

30-Apr-18

15-&101131-2-1

AF/AL

18719

Pratt & Whitney

15 inchPrecision Level, 15 in.

3-Nov-17

31-May-18

1-&18719-2021-1

AF/AL

20914

Brown & Sharpe

701-818

Master Granite Square, 9x18x3
Grade AA

7-Aug-17

31-Aug-18

37340

AF/AL

3037

Starrett

AG11C

Angle Block Set, 11 pcs.

21-Oct-17

31-Oct-19

17-38283-A

AF/AL

3045

Starrett/Webber

Unknown (RD0307)

Gage Block, English, 0.01", 0.02",
0.03", Grade 028-Dec-17

31-Dec-18

15-&3045-17-1

AF/AL

3148

Tru-Stone

24 in. x 36 in.

Surface Plate

6-Dec-1730-Dec-18

1-171207103135

AF/AL

The use of the standard is defined as: AF - used for as-found readings, AL - used for as-left readings.

Environmental Data
Temperature

Relative Humidity

68.11°F /20.06°C

3025

26.80%

Calibrated At:
35 Vantage Point Dr

Facility Responsible:
35 Vantage Point Dr
Rochester, NY 14624
800-828-1470

Rochester, NY 14624

Unit Barcode:

Temp / RH Asset

0900B220024

Calibrated By:

Reviewed By:

Electronically Signed By:

Electronically Signed By:

Tyler Melnyk

Frederick Tank

Tyler Melnyk

Feb 09, 2018

Frederick Tank

Feb 09, 2018

Calibration Technician

10:44:26 -05:00

Lab Manager

11:31:25 -05:00

0900B220024

Date Received: February 05, 2018
Service Level : R6
Revised On: Revised on February 20, 2018

Certificate - Page 3 of 3
Reprinted on February 20, 2018

Customer Number:

1-636816-000

OPS-F20-013R1 01/23/2017 FP001R1 10/12/2017