Ultrasonic Thickness Gauges
ISO9001

JQA0950

Options
Attachment for measuring pipes（Equipment for standard probe）＊ Picture :Attachment with standard probe

TI-P01A Attachment

Probe type 10Z6NDT-A for TI-45CA
Probe for narrow and small
measuring points
Probe type 10Z6NDT-A can work with only TI-45CA

The new Ultrasonic Wall Thickness Gauges TI-45NA, TI-45BA & TI-45CA
accurately measure wall thickness of all metals, ceramics, glass andmost rigid plastics

Features

The attractive, ergonomic housing de-Resolution of 0.01mm
Extremely compact & lightweight
Switch-selectable units - mm or inches

sign is extremely compact, lightweightand rugged. It fits comfortably in thepalm of your hand and incorporates aback light that illuminates the LCD inpoorly lit areas.

Built-in calibration test plate
Quick display update with last reading retainedon the display
Display symbols alert user to poor coupling andlow battery conditions

Made in Japan byformerly Kawatetsu Advantech Co.,Ltd.

UltrasonicThickness Gauges
TI-45NA/TI-45BA/TI-45CA
Applicable to measure wall thickness and state of corrosion of tanks,
pipings, hull plate of ships, and structures.
Also applicable to product inspection of various materials.
Metals : steel, cast steel, aluminium, copper, brass, titanium,etc.
Non-metallic materials : glass, hard plastics（except acrylic）, etc.

Storage Tanks

Piping Systems

Ocean Vessels

Parts Inspection

BottleVelocity table

Front Panel
Mode select switch

Materials

CL（m/s）

Back light switch

Steel

5920

Zero adjustment switch

SUS 304

5790

Turning on-off switch

Glass

5570

Test plate（5mm）

Copper

4700

Zinc

4170

TI-45BA

TI-45NA

Specifications
Measuring range
in steel & pipe dia
Resolution accuracy*1Probe Diameter(Contact surface Diameter)
Display
Display resolution
Velocity range
Cable
Battery
Calibration plate
Size
WeightTemparature
Warranty

Resin

TI-45 series Complete Kit
The TI-45 series is supplied as a complete kit withthe gauge and wrist strap, probe and cable assembly, 1 bottle of coupling fluid, battery and instruction
manual - all supplied in a fitted, hard plastic carryingcase.

TI-45CA

0.40−15.00mm
0.50−19.00mm
1.00−199.99mm
φ10.5 × thickness 1.50mm or more
φ10.5 × thickness 0.8mm or more
φ27.2 × thickness 1.50mm or more
±0.03mm±0.03mm±0.05mm or ±0.2%rdg *2
5Z10NDT-M（5MHz）
10Z10NDT-B（10MHz）
10Z6NDT-A（10MHz）
φ13.0(φ11.5)mm
φ13.0(φ11.5)mm
φ7.5(φ6.5)mm
4 1/2 - digit LCD display, light with push button illumination
0.01mm
1 000m/s to 12 000m/s
1m long cable with locking, quick removable connections
One AA battery
Built-in test piece
69（W）
× 144
（H）
× 30（D）（mm）
Gauge:150g ± 10g, Probe type 5Z10NDT-M:50g ± 10g, Probe type 10Z10NDT-B:50g ± 10g, Probe type 10Z6NDT-A:35g ± 10g-5℃ to 50℃（Ambient・Materials）
Gauge：1 year, Probe：6 months*1 in case of steel measurement*2 "rdg" means abbreviation of "reading"

OpitonAttachment for measuring pipes and tubes
Couplant(coupling fluid)

TI-P01A
TI-C01

PRECAUTIONS
The probe surface is fabricated from acrylic resin and care should be placed down for measurements and lifted vertically when complete. Do not slide on surfaces.
Do not use these gauges where material temperatures exceed 60℃（140°F）
as the probe will be damaged. Use the High Temperature Thickness Gauge for these applications.
Keep the gauge free of dust
（especially metal powders, carbon, etc.）
as they will damage the PC Board. Use a damp cloth to clean the gauge after use. DO NOT USE
CHEMICAL SOLVENTS OF ANY KIND.

＊Specifications may be changed without notice due to proceeding improvements.

formerly Kawatetsu Advantech Co.,Ltd.

ＵＲＬ : http://www.jfe-advantech.co.jp/
Tokyo Head Oﬃce (Overseas Sales)
JFE Kuramae Bldg.,2-17-4, Kuramae Taito-Ku, Tokyo
111-0051, Japane-mail: tokyo@jfe-advantech.co.jp Tel. ＋81-3-5825-5577

Fax. ＋81-3-5825-5591

Head Oﬃce and Main Plant
3-48 Takahata-cho, Nishinomiya-shi, Hyogo Pref.
663-8202 Japane-mail: honsha@jfe-advantech.co.jp
Tel. ＋81-798-66-1508
EC-TI-45-01A

Fax. ＋81-798-65-7025
15.05.data K