MULTI GAUGE CONTROLMG15

MULTI GAUGE CONTROL
MG15 ION MULTI GAUGE CONTROLLER
MG15ION MULTIGAUGECONTROLLER
ION MULTIGAUGECONTROLLER

Real time pressure charts (data export possibility).

DESCRIPTIONTECHNICALReal
DATA
time pressure charts (data export possibility).

CONTROL APPLICATION

Supply voltage

TECHNICAL DATA

Measurement channels
Supply voltage

100 - 240 VAC, 50 - 60 Hz(power consumption max 200 W)
XSWR DFWLYHFKDQQHOVSDVsivechannels:
to Hz
3 - number
100- 240
VAC, 50up- 60
GHÀQHGDWWKHWLPHRIRUGHU(powerconsumption max 200 W)

Supported activegauges XSWRCTR90,DFWLYHFKDQQHOVSDVTTR91, TTR211, PTR225,
Measurement
channels
PKR251/360/361,
PTR90,
sivechannels: up toPCR280,
3 - number
ITR90, ITR100, Baratron, ANAGHÀQHGDWWKHWLPHRIRUGHU
LOG IN, PG105, ATMION, User
Supported active gauges
CTR90,
TTR91, TTR211, PTR225,
GHÀQHG
PKR251/360/361, PCR280, PTR90,
Supported passive gauges ITR90,
IE414,
IE514,Baratron,
UHV-24, UHV-24p,
ITR100,
ANA0.66HULHV18'(8+9$
LOG
IN, PG105, ATMION, User
NR-F-UHV, G8130, BARION basicGHÀQHG,,8VHUGHÀQHG
Supported passive gauges IE414, IE514, UHV-24, UHV-24p,
Measurement range:
0.66HULHV18'(8+9$
Passive channel
2×12-12 mbar
NR-F-UHV,
G8130, BARION basic
Active channel
5×10-10 mbar,,8VHUGHÀQHG
Display range
Measurement
range:
Passivechannel
Setpoints
relay outputs
Active channel
5HOD\RXWSXWVZLWKFRQÀDisplayrange
gurableactivation signal
Setpoints
relay outputs
Digital inputs

2×10-12 - 2×105 mbar
-122×12mbar
10 (4xNO/NC,6xNO)
5×10-10 mbar
4 (Emission,Degas)
2×10-12 - 2×105 mbar

ELECTRONICS
ELECTRONICS

The MG15 is able to support four active gauges and up to
three passive gauges, extending the measurement range toDESCRIPTION
2×10-12 mbar. The unit is fully software controlled. The unit
can be remotely controlled via one of available interfaces.
The MG15 is able to support four active gauges and up to
The unit is equipped with autosave function (the device savethree passive gauges, extending the measurement range toyour parameters, preset and apply them automatically after
2×10-12 mbar. The unit is fully software controlled. The unitrestart).
can be remotely controlled via one of available interfaces.
The unit is equipped with autosave function (the device saveyour parameters,
FEATURES preset and apply them automatically afterrestart).
ƒ Measurement range from atmospheric to 2×10-12 mbar
ƒ Pressure plots
ƒ FEATURES
Pressure trend graphs
ƒ 0HDVXUHPHQWÀOWHULQJ ORZPHGKLƒ Measurement range from atmospheric to 2×10-12 mbar
ƒ Controls almost all commercially available Bayard-Alpert
ƒ Pressure plotsgauges
ƒ Pressure trend graphs
ƒ Degassing of Bayard-Alpert with power and time limit
ƒ 0HDVXUHPHQWÀOWHULQJ ORZPHGKLƒ 6XSSRUWV,U:7KRULDÀODPHQWV
ƒ Controls almost all commercially available Bayard-Alpert
ƒ Bayard-Alpert overpressure protectiongauges
ƒ Selectable measurement units: mbar, Torr, Pascal, psia
ƒ Degassing of Bayard-Alpert with power and time limit
ƒ *DVVSHFLÀFFRUUHFWLRQZLWKRQHFXVWRPL]HGVHWWLQJƒ 6XSSRUWV,U:7KRULDÀODPHQWVƒ Unit over-temperature protection
ƒ Bayard-Alpert overpressure protection
ƒ 10 individually programmable set-points with thresholdƒ Selectable measurement units: mbar, Torr, Pascal, psia
and hysteresis functions
ƒ *DVVSHFLÀFFRUUHFWLRQZLWKRQHFXVWRPL]HGVHWWLQJ
ƒ 8VHUGHÀQHGFKDQQHOQDPHV
ƒ Unit over-temperature protection
ƒ 3RVVLELOLW\RIDFWLYHJDXJHVVHOILGHQWLÀFDWLRQ
ƒ 10 individually programmable set-points with thresholdƒ %HDPÁX[PRQLWRU
and hysteresis functions
ƒ %ODFN%R[YHUVLRQDYDLODEOH RSWLRQ ZLWKRXW7)7GLVSOD\ƒ 8VHUGHÀQHGFKDQQHOQDPHV
ƒ 3RVVLELOLW\RIDFWLYHJDXJHVVHOILGHQWLÀFDWLRQƒ %HDPÁX[PRQLWRUCONTROL APPLICATION
ƒ %ODFN%R[YHUVLRQDYDLODEOH RSWLRQ ZLWKRXW7)7GLVSOD\104(4xNO/NC, 6xNO)
4 (Emission,4 (0 - 10V)Degas)

5HOD\RXWSXWVZLWKFRQÀAnalog outputs
gurable activation signal
Communication interface RS232/485, Ethernet
Digital inputs
4 PROFINET - option (specify at order)

Analogoutputs- 10V)
Communication
protocol 4 (0MODBUS-TCP
Communication
Ethernet
User interface interface RS232/485,µ7)7GLVSOD\ZLWKWRXFKVFUHHQPROFINET - option (specify at order)
Dimensions
212.6 x 128.4 x 260.3 mm
Communication protocol
MODBUS-TCP:[+[' RI+(µUserinterfaceµ7)7GLVSOD\ZLWKWRXFKVFUHHQ
Weight (approx.)
3.8 kgDimensions
212.6x 128.4x 260.3 mm* prepared to support many othercustomgauges:[+[' RI+(µ
Weight(approx.)
APPLICATIONS

3.8 kg* preparedsupport
manyothercustomsystems
gaugesƒ Anytohighand ultra
highvacuumƒPREVAC sp. z o.o.

sales@prevac.euƒƒPL44362 Rogów+48 32 459 20 01

ƒ+48 32 459 21 30
Raciborska
Str. 61 Technology
Precisionand Vacuum

Precision and Vacuum TechnologyMG15

ƒMeasurement of ultimate pressure (from atmosphere to UHVrange)
APPLICATIONS

Bakeable and non bakeable systems
Any high and ultra high vacuum systems
Measurement of ultimate pressure (from atmosphere to UHVrange)
INSTRUMENTS WITH ELECTRONICS
Bakeable and non bakeable systems

39

www.prevac.eu
39

INSTRUMENTS WITH ELECTRONICS