AUTO SCAN
SPECTROPHOTOMETER
FD-9
E INSTRUCTION MANUAL
Before using this instrument, please read this manual.

Safety Symbols
The following symbols are used in this manual and on the FD-9 to prevent accidents which may occur as a result of incorrect use of the
instrument.
Denotes an instruction regarding a safety warning or note. Read the instruction carefully to ensure safe and correct use.
Denotes an instruction regarding the risk of electric shock. Read the instruction carefully to ensure safe and correct use.
Denotes an instruction regarding the risk of fire. Read the instruction carefully to ensure safe and correct use.
Denotes a prohibited action. This action must never be performed.
Denotes an instruction. This instruction must be strictly adhered to.
Denotes a prohibited action. Never disassemble the instrument.
Denotes an instruction. Be sure to disconnect the AC adapter from the AC outlet.
This symbol indicates A.C.
This symbol indicates D.C.

Notes on this Manual• Copying or reproduction of all or part of the contents of this manual without KONICA MINOLTA’s permission is strictly prohibited.
• The contents of this manual are subject to change without prior notice.
• Every effort has been made in the preparation of this manual to ensure the accuracy of its contents. However, should you have any
questions or find any errors, please contact your retailer or a KONICA MINOLTA-authorized service facility.
• KONICA MINOLTA will not accept any responsibility for consequences arising from the use of the instrument.

iSafety Precautions
To ensure correct use of this instrument, read the following points carefully and adhere to them. After you have read this manual, keep it
in a safe place where it can be referred to anytime a question arises.

WARNING

Failure to adhere to the following points may result in death or serious injury.

Do not use the instrument in places where flammable or
combustible gases (gasoline, etc.) are present. Doing so
may cause a fire.

Do not disassemble or modify the instrument or the AC
adapter. Doing so may cause a fire or electric shock.

Always use the AC adapter supplied as a standard
accessory or the optional AC adapter, and connect it toan AC outlet of the rated voltage and frequency. If an
AC adapter other than those specified by KONICAMINOLTA is used, it may result in damage to the unit,
fire or electric shock.

Take special care not to allow liquid or metal objects toenter the instrument. Doing so may cause a fire orelectric shock. Should liquid or metal objects enter the
instrument, turn the power off immediately, disconnect
the AC adapter plug from the AC outlet, and contact thenearest Konica Minolta-authorized service facility.

If the instrument will not be used for a long time,
disconnect the AC adapter plug from the AC outlet.
Accumulated dirt or water on the prongs of the AC
adapter’s plug may cause a fire and should be removed.

The instrument should not be operated if it is damaged
or the AC adapter is damaged, or if smoke or odd
smells occur. Doing so may cause a fire. In suchsituations, turn the power off immediately, disconnect
the AC adapter plug from the AC outlet and contact thenearest Konica Minolta-authorized service facility.

Do not forcibly bend, twist, or pull the AC adapter
power cable. Do not scratch or alter the power cable orplace heavy objects on it. Doing so may damage thepower cable and cause a fire or electric shock.

Always hold the plug itself when disconnecting the AC
adapter plug from the AC outlet. Pulling on the power
cable may damage it and cause a fire or electric shock.

Insert the power plug fully and securely. Incomplete
inserting may cause fire or electric shock.

Do not insert or disconnect the AC adapter plug froman AC outlet with wet hands. Doing so may cause
electric shock.

CAUTION

Failing to adhere to the following points may result in injury or damage to the
instrument or other property.

Do not place the instrument on an unstable or sloping
surface. Doing so may result in its falling or
overturning, causing injury. Be careful not to drop the
instrument when carrying it.

Make sure that the AC outlet is located near the
instrument and that the AC adapter plug can beconnected to and disconnected from the AC outlet
easily.

Do not open the cover during scanning. If the cover isopened during scanning, there is a risk that your handmight get caught in the instrument, resulting in injury.

When cleaning, disconnect the power plug. Cleaning
with the power plug connected may result in electric
shock.

1

Introduction
The Auto Scan Spectrophotometer FD-9 is designed for the rapid automatic scanning of printed materials.

Notes on Use
Be sure to use this instrument properly. Use of this instrument in ways other than those specified in this manual may result in risk of
injury, electric shock, instrument damage, or other problems.

❙❙ Operating Environment• Use the FD-9 at an ambient temperature of between 10°C and 35°C and relative humidity of 30 to 85% with no condensation. Do not
use the instrument in areas subject to rapid temperature changes.
• Do not leave the FD-9 in direct sunlight or near sources of heat, such as stoves, etc. The internal temperature of the instrument may
become much higher than the ambient temperature in such cases.
• Do not use the FD-9 in areas where dust, cigarette smoke or chemical gases are present. Doing so may cause deterioration in
performance or a breakdown.
• Do not use the FD-9 near equipment which produces a strong magnetic field (such as speakers, etc.).
• The FD-9 belongs to installation category I products (equipment which is powered by an AC adapter connected to commerciallyavailable power).
• The FD-9 belongs to pollution degree 2 products (equipment which may cause temporary electrical hazards due to contamination orcondensation or products which are used in such an environment).
• Do not use the FD-9 at altitudes higher than 2,000 m.
• The FD-9 and the AC adapter supplied as a standard accessory have been designed exclusively for indoor use. They should never beused outdoors because rain or other factors may damage the instrument.

❙❙ Measurement• Make sure that test charts are clean and not dusty.
• When bringing test charts from another location, avoid sudden changes in temperature and allow the test charts to become sufficiently
adapted to room temperature before taking measurements. If condensation occurs on the test chart, it may cause sluggish scanning or
paper jams.
• Be sure that there are no objects blocking the paper entry or paper exit slots of the FD-9.
• The FD-9 is only for measuring paper test charts or similar subjects. It cannot be used for measuring printed cloth or other subjects
with insufficient stiffness.

❙❙ Power Source• Make sure that the power switch is set to off (“  “) when the FD-9 is not in use.
• Always use the AC adapter supplied as a standard accessory and connect it to an AC outlet of the rated voltage and frequency.
• Use an AC power supply of the rated supply voltage (within ±10%).

❙❙ System••••

Do not subject the FD-9 to strong impact or vibration. Doing so may cause deterioration of performance or breakdown.
When the FD-9 is not in use, it should be covered to prevent entry of foreign matter.
The FD-9 may cause interference if used near a television, radio, etc.
Since the FD-9 uses a microcomputer, external magnetic noise may cause malfunction. In this case, turn the power off, and wait 30
minutes, and then turn it on again.

❙❙ Obtaining the latest instrument firmware• The latest FD-9 firmware can be downloaded from the following URL:
http://www.konicaminolta.com/instruments/download/software/instrument_firmware/index.html

2Notes on Use❙❙ Notes on Storage• The FD-9 should be stored at temperatures between 0°C and 40°C, and at a relative humidity of 80% or less (35°C) without
condensation. Do not store the instrument in areas subject to high temperatures, high humidity, sudden changes in temperature, orwhere freezing or condensation may occur, because these circumstances may cause a breakdown. It is recommended to store the FD-9
with a drying agent at a temperature around 20°C.
• Do not leave the FD-9 inside a car such as in the trunk. Otherwise, the temperature and/or humidity may exceed the allowable range
for storage during midsummer or midwinter, resulting in a breakdown.
• Keep the packing materials used for shipment and use them to transport the FD-9. This protects the instrument from sudden changes intemperature, vibration, and shock.
• Do not store the FD-9 in areas where dust, cigarette smoke or chemical gases are present. Doing so may cause deterioration in
performance or a breakdown.
• Entry of dust into the instrument will hinder accurate measurement. When the instrument is not in use, please cover the instrumentwith the supplied Dust Cover to prevent the entry of dust.
• Be sure to keep all packing materials (cardboard box, cushioning material, plastic bags, etc.). They can be used to protect the
instrument during storage or during transportation to the service facility for maintenance (recalibration, etc.).

❙❙ Notes on Cleaning• If the FD-9 becomes dirty, wipe it with a soft, clean dry cloth. Never use solvents such as thinner or benzene.
• If the internal White Calibration Plate becomes dirty, it can be cleaned according to the procedure on p. 35.
• If the FD-9 breaks down, do not try to disassemble and repair it by yourself. Contact a KONICA MINOLTA authorized service facility.

❙❙ Disposal Method• Make sure that the FD-9 and its accessories and packing materials are either disposed of or recycled correctly in accordance with local
laws and regulations.

3

Table of Contents
Safety Symbols��������������������������������������������������������������������������������������������������������������������������������������������������������������������������� i
Notes on this Manual����������������������������������������������������������������������������������������������������������������������������������������������������������������� i
Safety Precautions��������������������������������������������������������������������������������������������������������������������������������������������������������������������� 1
Introduction������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 2
Notes on Use������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 2
Operating Environment���������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 2
Measurement�������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 2
Power Source������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 2
System������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������ 2
Obtaining the latest instrument firmware������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 2
Notes on Storage�������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 3
Notes on Cleaning������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������ 3
Disposal Method�������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 3

Table of Contents���������������������������������������������������������������������������������������������������������������������������������������������������������������������� 4
Unpacking���������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 6
Unpacking the FD-9��������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 6
Packing Materials:���������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 6
Removing Tapes:������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������ 6
Unpacking the Optional Auto Sheet Feeder FD-A09������������������������������������������������������������������������������������������������������������������������������������������������������������ 7
Packing Materials����������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 7
Removing FD-A09 from Box����������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 7
Removing Tapes������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 7

Standard Accessories���������������������������������������������������������������������������������������������������������������������������������������������������������������� 9
Optional Accessories��������������������������������������������������������������������������������������������������������������������������������������������������������������� 10
System Diagram���������������������������������������������������������������������������������������������������������������������������������������������������������������������� 11
Names and Functions of Parts����������������������������������������������������������������������������������������������������������������������������������������������� 12
Preparations���������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 13
Setting up FD-9�������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 14
Necessary space������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������ 14
Attaching/Removing Paper Guide��������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 15
Attaching����������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 15
Removing���������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 15
Setting Up Auto Sheet Feeder FD-A09 (optional accessory)���������������������������������������������������������������������������������������������������������������������������������������������� 16
Necessary space for Auto Sheet Feeder FD-A09��������������������������������������������������������������������������������������������������������������������������������������������������������� 16
Attaching Output Tray to Auto Sheet Feeder FD-A09������������������������������������������������������������������������������������������������������������������������������������������������� 17
Mounting FD-9 on Auto Sheet Feeder FD-A09����������������������������������������������������������������������������������������������������������������������������������������������������������� 18
Removing FD-9 from Auto Sheet Feeder FD-A09������������������������������������������������������������������������������������������������������������������������������������������������������� 19
Connecting the AC adapter�������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 20

Switching Power On/Off��������������������������������������������������������������������������������������������������������������������������������������������������������� 21
Switching power on������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 21
Switching power off������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 21

4

Connecting FD-9 to Computer���������������������������������������������������������������������������������������������������������������������������������������������� 22
Connecting FD-9 to Computer using USB cable����������������������������������������������������������������������������������������������������������������������������������������������������������������� 22
Determining connected USB port number������������������������������������������������������������������������������������������������������������������������������������������������������������������� 23
Setting Network Settings for Connecting FD-9 to Computer via a LAN���������������������������������������������������������������������������������������������������������������������������� 24
Setting network settings directly���������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 24
Setting network settings via DHCP������������������������������������������������������������������������������������������������������������������������������������������������������������������������������ 26

Taking measurements������������������������������������������������������������������������������������������������������������������������������������������������������������� 27
Feeding Test Charts into FD-9��������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 27
Feeding test charts when using the FD-9 alone������������������������������������������������������������������������������������������������������������������������������������������������������������ 28
Feeding test charts when using optional Auto Sheet Feeder FD-A09�������������������������������������������������������������������������������������������������������������������������� 28

Settings screen������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 29
Opening/closing settings screen������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 29
Settings screen layout���������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 29
Setting items������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������ 30
Selecting and changing settings������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 30

Setting display language��������������������������������������������������������������������������������������������������������������������������������������������������������� 31
Troubleshooting����������������������������������������������������������������������������������������������������������������������������������������������������������������������� 32
Error messages��������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 32
Clearing paper jams������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 32
When using FD-9 without Auto Sheet Feeder�������������������������������������������������������������������������������������������������������������������������������������������������������������� 32
When using FD-9 with Auto Sheet Feeder������������������������������������������������������������������������������������������������������������������������������������������������������������������� 33

Cleaning����������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 35
Cleaning the white calibration plate������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������ 35
Cleaning inside the FD-9����������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 36

Scan Measurement Chart Conditions����������������������������������������������������������������������������������������������������������������������������������� 37
Dimensions������������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 38
Specifications��������������������������������������������������������������������������������������������������������������������������������������������������������������������������� 39

5

Unpacking❙❙ Unpacking the FD-9
Packing Materials:
Keep all packing materials (cardboard box, cushioning materials, plastic bags, etc.) in a safe place. The FD-9 is a precision measuring
instrument. In the event that the instrument needs to be shipped to a Konica Minolta authorized service facility, the packing materials canbe used to protect the instrument from impact and vibration during shipment. If the original packing materials are lost or damaged, pleasecontact the nearest Konica Minolta authorized service facility.

Removing Tapes:
The cover, color measurement sensor unit, and conductive pads of the FD-9 have been taped in position to avoid movement during
shipment. The tapes must be removed before use.
After unpacking the FD-9 and placing it in its final location, remove the tape strips ① and ② as indicated in the illustrations below.

Open the cover and remove tape ③ as shown at right. If power is switched on before tape
③ is removed, the FD-9 may malfunction. After removing tape ③, slide the color
measurement sensor unit slightly by hand to check that it moves smoothly. If it does not
move smoothly or if there is excessive noise, please contact your local Konica Minolta
dealer.

Carefully remove the tapes ④ as shown in the figure below. Be careful not to remove the conductive pads under the tapes. If tapes ④ are
not removed, repeatability may suffer.

6Unpacking❙❙ Unpacking the Optional Auto Sheet Feeder FD-A09
Packing Materials
Keep all packing materials (cardboard box, cushioning materials, plastic bags, etc.) in a safe place. In the event that the FD-A09 needs tobe shipped to a Konica Minolta authorized service facility, the packing materials can be used to protect the product from impact andvibration during shipment. If the original packing materials are lost or damaged, please contact the nearest Konica Minolta authorized
service facility.

Removing FD-A09 from Box
After opening the box of the FD-A09, remove the output tray as indicated by ① at rightbefore removing any of the other materials.

Remove the cardboard inserts and cushioning materials, and then carefully lift the FD-A09 out of the box and place it where it will beused.

Removing Tapes
To protect some parts of the FD-A09 and prevent their movement during shipment, spacers have been taped to the FD-A09 or parts havebeen taped in place. The tapes and spacers must be removed before use. After placing the FD-A09 where it will be used, remove the tapes
and spacers in the order specified below.
Remove the two spacers under the supply tray and the tapes holding them in place (② below) and then remove the tape ③ from the
supply tray.

7

Unpacking

Remove the two tapes ④ holding the supply tray closed against the FD-A09, remove the tape ⑤ holding the extension tray closed, andthen remove the two tapes ⑥ holding the cover closed.

Open the supply tray by tilting it down and remove the tape and cushion indicated by ⑦ below.

8Standard Accessories
AC Adapter AC-A324F
Used to supply power from an AC outlet to the instrument.
Input: 100 to 240 V
50/60Hz
Output (Typical): 24 V
6.25A
(STD-24050)

USB Cable (3m) IF-A18
Used to connect the instrument to a computer.

Paper Guide
Used to guide the test chart into the instrument when feeding test charts
manually.

Measurement Utility Software FD-S2w
Used to control the instrument from a computer.

9

Optional Accessories
Auto Sheet Feeder FD-A09
Used to automatically feed test charts into the instrument.
Recommended maximum number of loaded sheets:
Normal paper (80 g/m2): 100 sheets
Heavy paper (130 g/m2): 70 sheets
Paper size:
Paper width:
Paper length:
Paper thickness:

10

90 to 320 mm
170 to 450 mm
0.08 to 0.25 mm

System Diagram
Auto Scan
Spectrophotometer
FD-9

AC Adapter

Measurement
Utility Software
FD-S2w

USB Cable (3m)
IF-A18

Auto Sheet Feeder
FD-A09Computer(commercially available)

Standard accessories
Optional accessories

11Names and Functions of Parts

12

Front view

Rear view

Bottom view

Inside view

1

Cover

2

Paper guide mounting slots

For attaching the Paper Guide.

3

Display panel

Shows instrument settings, status, error messages, etc.

4

Up / down buttons

For selecting items from the setting screen or making numerical settings.

5

buttonFor confirming selection or settings.

6(Return) button

For entering the settings screen or returning from the setting screen.

7

Power button

For switching power on and off.

8

AC adapter input socket

For connecting to the AC adapter

9

LAN connection terminal

For connecting the instrument to a wired network.

10

USB connection terminal

For connecting the instrument to a computer via USB.

11

Auto Sheet Feeder alignment holes

12

Auto Sheet Feeder connector cover

13

Color measurement sensor unit

Moves back and forth during operation to scan the test chart.

14

Base mounting screws

These screws secure the instrument to its base. They can be loosened if necessary toclear jammed paper (see p. 32) or for cleaning (see p. 35).
• Be sure that these screws are firmly tightened. If they are loose whenmeasurements are taken, measurement accuracy may suffer.

15

White calibration plate

Used for performing white calibration.

Preparations

13

Preparations❙❙ Setting up FD-9
The FD-9 should be set up on a flat, level surface with sufficient space for the color chart in front of and behind the FD-9. There should beenough space in front of the FD-9 for the color chart to be measured to be inserted into and ejected from the FD-9 without any obstacles,
and enough space behind the FD-9 for the color chart to fully extend from the FD-9 without obstacles. (See below.)• If the FD-9 is set up on a rough surface or a surface which is not level, measurement accuracy may suffer.
• Obstacles may cause the color chart to bend or be shifted to either side, resulting in damage to the color chart or misalignment during
scanning.
• After scanning, the scanned color chart will be ejected from the front of the FD-9. If the FD-9 is placed at the edge of a table, theejected color chart might fall to the floor and be damaged.

Necessary space

Maximum chart
paper length

Maximum chart
paper length14

Preparations❙❙ Attaching/Removing Paper Guide
Paper Guide is used when test charts will be fed manually into the FD-9.
• If the Auto Sheet Feeder FD-A09 will be used, the Paper Guide should be removed.
• When the Paper Guide is not attached to the FD-9, it should be stored carefully where it will not be lost or broken.

Attaching1

While gripping the ends of the Paper Guide, align thehooks of the Paper Guide with the paper guide mounting
slots of the FD-9, and insert the hooks fully.

2

Slide the Paper Guide straight down until the Paper Guide
clicks in place.
• Be careful to slide the Paper Guide straight down. Twisting the Paper
Guide or tilting it away from the FD-9 may cause the hooks to be broken.

Removing

1

While gripping the ends of the Paper Guide, slide thePaper Guide straight up.
• Be careful to slide the Paper Guide straight up. Twisting the Paper Guideor tilting it away from the FD-9 may cause the hooks to be broken.

2

Remove the hooks from the paper guide mounting slots.

15

Preparations❙❙ Setting Up Auto Sheet Feeder FD-A09 (optional accessory)
The optional Auto Sheet Feeder FD-A09 can automatically feed a stack of test charts into the FD-9. Up to 100 test charts can be placed in
the Auto Sheet Feeder, and they will be automatically fed into the FD-9 one by one to be scanned.
The FD-A09 should be set up on a flat, level surface with sufficient space for both the supply tray and the output tray to be fully opened.
There should be enough space for the color charts to be placed in the supply tray in front of the FD-A09 before measurement and for thecharts to be ejected from the FD-9 and lay in the output tray after measurement. (See below.)

Necessary space for Auto Sheet Feeder FD-A09

Approx. 400mm

Supply tray

Output tray

Approx. 400mm

16Preparations

Attaching Output Tray to Auto Sheet Feeder FD-A09
To attach the output tray to the FD-A09, hold the tray tilted upward, and insert the two tabs at the base of the tray into the corresponding
slots on the back of the FD-A09 as shown below.

17Preparations

Mounting FD-9 on Auto Sheet Feeder FD-A09• If the Paper Guide is attached to the FD-9, remove it before proceeding. See p. 15.

1
2

Unplug all cords and cables (AC adapter power cord, USB cable, LAN cable) from the FD-9.

3

Push down on the tab of the Auto Sheet Feeder connector
cover, pull the cover open, and remove it.

While holding the cover of the FD-9 closed, tilt the FD-9
back so that it rests on its back side.

• Please store the cover carefully so that it will not be lost or broken.

18

4
5

Return the FD-9 to the normal upright orientation.

6

Lift the FD-9 up, align the Auto Sheet Feeder alignment
holes on the bottom of the FD-9 with the alignment pegs
on the Auto Sheet Feeder, and slowly lower the FD-9
straight down onto the Auto Sheet Feeder until it clicks inplace.

Place the Auto Sheet Feeder FD-A09 on a flat surface.

Preparations

Removing FD-9 from Auto Sheet Feeder FD-A09

1

Switch off the power of the FD-9.

2

Unplug all cords and cables (AC adapter power cord, USB
cable, LAN cable) from the FD-9.

3

Lift the FD-9 straight up off of the Auto Sheet Feeder.

4

Replace the Auto Sheet Feeder connector cover on the
FD-9.

19

Preparations❙❙ Connecting the AC adapter
Warning• Always use the AC adapter supplied as a standard accessory or a specified replacement AC adapter, and use the AC adapter only withan AC outlet of the rated voltage and frequency. Failure to do so may damage the FD-9 or AC adapter, or may cause electric shock orfire.
• If the FD-9 will not be used for a long period of time, disconnect the AC adapter from the AC outlet.
• When plugging the AC adapter into an AC outlet, be sure that the prongs of the plug are clean and dry. Accumulated dirt or moistureon the prongs of the plug may cause a fire and should be removed.
• When plugging in or unplugging a cord, be sure to hold the plug, not the cord. Do not pull on or forcibly bend the cord. Doing so may
result in broken wires.
• Do not plug in or unplug the AC adapter with wet hands. Doing so may cause electric shock.
• Be sure to insert plugs fully and securely. Incomplete insertion may cause fire or electric shock.
• Do not disassemble or modify the AC adapter or cords. Doing so may cause fire or electric shock.
• Do not unplug or plug in the AC adapter with the instrument’s power set to “ON” ( | ) Doing so may cause a malfunction.
Procedure

1

Make sure that the power switch of the FD-9 is set to off().

2

Insert the output plug of the AC adapter power cord into
the power cord inlet of the AC adapter as shown.

3

Insert the output plug of the AC adapter into the AC
adapter inlet of the FD-9 as shown. Be sure the plug isoriented correctly.

4

Insert the input plug of the AC adapter power cord into an
AC wall outlet.
• When disconnecting the AC adapter, make sure that the power switch is set
to off ().

20

Switching Power On/Off❙❙ Switching power on
1

Press the on ( | ) side of the power switch. The FD-9 will
start up.

The Konica Minolta logo will be displayed for a few seconds, followed by the initialization display. When initialization has been
completed (after about 25 seconds), the “Please connect” display will be shown.

❙❙ Switching power off
1

Press the off () side of the power switch to switch off thepower.

21

Connecting FD-9 to Computer
The FD-9 can be connected to the computer directly using a USB cable or via a LAN.
When connecting via a LAN, either DHCP or a fixed IP address can be used.

❙❙ Connecting FD-9 to Computer using USB cable• Use the included USB Cable IF-A18 (3m) to connect the FD-9 and computer.
• It is recommended that the software to connect to and control the instrument (such as the included Measurement Utility Software
FD-S2w) be installed prior to connecting the FD-9.
• The USB communication port of the instrument conforms to USB 2.0.
• When connecting the instrument and computer, the USB driver for the instrument needs to be installed. If it is not installed with thesoftware, you will be prompted to install the driver when the instrument is first connected to the computer and switched on.
• The instrument is not designed to be powered via the USB cable. The AC adapter must be used.
• Make sure that the USB connector plugs are properly oriented and connected securely. Check the shapes of the receptacle and plug,and insert the plug fully into the receptacle.
• When connecting/disconnecting the USB cable, always hold onto the connector plug. Do not pull on or forcibly bend the cable. Doingso may result in wire breakage.
• Make sure that the cable has sufficient length. Putting tension on the cable may cause connection failure or wire breakage.
In general, a USB cable can be connected/disconnected while the instrument and computer are switched on, but in the procedure below
the FD-9 and computer are switched off.

1

Plug the USB Type B plug of the USB Cable into the USB
connection terminal on the back of the FD-9.

2

Plug the USB Type A plug of the USB Cable into the USB
port of the computer.

3

Connect the AC adapter and switch on the FD-9 andcomputer.
• When you are prompted to install the USB driver, specify the USB driver
included with the software or on the Measurement Utility Software
FD-S2w disk.
• When using the included Measurement Utility Software FD-S2w, refer tothe FD-S2w Installation Guide.

22

Connecting FD-9 to Computer

Determining connected USB port number
The number of the USB port assigned to the FD-9 is required when connecting to the FD-9 via USB.
To see which port has been assigned, follow the steps below.
On Windows 7:

1

Open Control Panel.

2

Select System and Security.

3

In System, click Device Manager.

4

Click on the  next to Ports (COM & LPT). The list of connected devices will appear.

5“Measuring Instruments” will appear in the list, followed by the assigned COM port in parentheses.

On Windows 8, Windows 8.1, or Windows 10:

1

Point to the lower-left corner of the screen and then right-click in the window that appears.

2

In the menu that appears, click “Device Manager” to open the Device Manager.

3

Click “Ports (COM & LPT)” to expand it, and check the COM port assigned to “Measuring Instruments”.

On Mac OS X:

1

Press and hold the Option key on your keyboard, and click the Apple menu.

2

Choose “System Information” or “System Profiler”.

3

In the window that appears, select “USB” from the left column and check the COM port assigned to “Measuring
Instruments”.

23

Connecting FD-9 to Computer❙❙ Setting Network Settings for Connecting FD-9 to Computer via a LAN
The FD-9 can be connected to a LAN and shared by multiple computers. When connecting the FD-9 to a LAN, network settings can be
performed directly or DHCP (Dynamic Host Configuration Protocol) can be used.
• Network settings can also be performed from Measurement Utility Software FD-S2w when the FD-9 is connected via USB.

Setting network settings directly
The IP address, subnet mask, and default gateway can be set directly in the settings screen.
• It is not possible to set the network name directly using the FD-9 buttons.

1

Press

2

Press
or
repeatedly to select “IP address” and press
The first number of the IP address will be highlighted.

to open the settings screen.
.

• “IP address” will be disabled if “DHCP” is set to “ON”. To change the setting to “OFF”, seep. 26.

3

Press
number.

or

repeatedly to set the desired value for the first• Holdingorpressed will cause the value to change continually.
• Values between 0 and 255 can be set. If
oris pressed again when the lowest or
highest value has been reached, the number will change to the highest or lowest value.

4

Press

5

Repeat steps 3 and 4 until all numbers have been set for the IP
address.

6

Press
to select “Subnet mask” and press
the subnet mask will be highlighted.

7

Press
number.

to move to the next number.

or. The first number of

repeatedly to set the desired value for the first• Holdingorpressed will cause the value to change continually.
• Values between 0 and 255 can be set. If
oris pressed again when the lowest or
highest value has been reached, the number will change to the highest or lowest value.

24

8

Press

9

Repeat steps 3 and 4 until all numbers have been set for the subnet
mask.

to move to the next number.

Connecting FD-9 to Computer

10 Pressto select “Default gateway” and press
highlighted.

11 Pressor

12 Press

to move to the next number.

. The first number of the default gateway will be

repeatedly to set the desired value for the first number.

• Holdingorpressed will cause the value to change continually.
• Values between 0 and 255 can be set. If
oris pressed again when the lowest or highest value has been reached, the number will change
to the highest or lowest value.

13 Repeat steps 3 and 4 until all numbers have been set for the default
gateway.

14 Pressto select “SAVE DATA” and press. The message“Caution: It reboots after saving setting data” will be shown.

15 Pressto save the settings and reboot the FD-9.

Once these values have been set, connection to the FD-9 from within the same subnet canbe performed by specifying the IP address during connection.

25

Connecting FD-9 to Computer

Setting network settings via DHCP
If your network has DHCP enabled, you can set DHCP on the FD-9 to “ON” and allow your system to automatically allot the IP address,
subnet mask, and default gateway.
Changing the DHCP setting

26

1

Press

to open the settings screen.

2

Press

or

repeatedly to select “DHCP” and press

3

Press

or

to change the setting to the desired setting.

4

Press
to select “SAVE DATA” and press. The message“Caution: It reboots after saving setting data” will be shown.

5

Press

.

ON: Network settings are performed automatically via DHCP.
OFF: Network settings are not performed automatically.

to save the settings and reboot the FD-9.

Taking measurements
Measurements with the FD-9 are controlled from software, such as the included Measurement Utility Software FD-S2w. For details on
taking measurements, please refer to the instruction manual for the software that you are using.

❙❙ Feeding Test Charts into FD-9
• If the FD-9 base mounting screws inside the cover are loose when measurements are taken, measurement accuracy may suffer. Be surethat the screws are firmly tightened.
• When bringing test charts from another location, avoid sudden changes in temperature and allow the test charts to become sufficiently
adapted to room temperature before taking measurements. If condensation occurs on the test chart, it may cause sluggish scanning or
paper jams.
• The following test charts should not be fed into the FD-9 or loaded in the optional Auto Sheet Feeder FD-A09. Doing so may cause
jams or may damage the FD-9 and/or the FD-A09.
• Wrinkled, folded, or torn charts• Charts with staples or paper clips• Charts bound together in booklet form
• Charts that are glued together• Charts with cutouts removed or which are cutouts• Charts which are adhesive labels• Permissible curl amount:Downward curl
Upward curl
tion

direc

R

Curl in feed direction

Feed

Feed

tion

direc

Lift ≤
5 mm≥ R70

Curl in scan direction

Feed

tion

direc

≥ R60

Curl radius whenpaper is stood on
edge

Feed

Lift
Lift when paper islaid flat on table.

tion

direc

Lift ≤
1 mm• When using test charts which are easily scratched, such as test charts on glossy paper, it is recommended that they be fed into the FD-9
by hand instead of using the optional Auto Sheet Feeder FD-A09. If the FD-A09 is used, the chart may become scratched.

27

Taking measurements

Feeding test charts when using the FD-9 alone• For information on feeding test charts when using the optional Auto Sheet Feeder FD-A09, please see below.

1

Place the end of the test chart on the Paper Guide with theprinted surface facing up, and slide the paper guides sothat they are positioned touching the sides of the test
chart as shown.
• For details on chart conditions, see p. 37.

2

Slide the test chart gently between the guides and into the FD-9.
When the test chart enters the FD-9 by a certain amount, it will
be automatically pulled into the FD-9 for measurements. When
the FD-9 starts pulling the test chart in, let go of the test chart.
• Be sure to slide the test chart in straight. If the paper guide positions are
too narrow or if the test chart goes in at an angle, the test chart may ride up
onto the paper guides, which may cause measurement accuracy to suffer.

Feeding test charts when using optional Auto Sheet Feeder FD-A09

1

When using the Auto Sheet Feeder, place the test charts
in the supply tray with the printed surface facing up andslide the paper guides so that they are positioned
touching the sides of the test charts as shown.
Recommended maximum number of loaded sheets:
Normal paper (80 g/m2): 100 sheets
Heavy paper (130 g/m2): 70 sheets
Paper size:
Paper width:
90 to 320 mm
Paper length:
170 to 450 mm
Paper thickness:
0.08 to 0.25 mm• For more details on chart conditions, see p. 37.

Feeding of the test charts from the supply tray into the FD-9 is controlled by the software. For details, please refer to the software
instruction manual.

28

Settings screen
The settings screen shows instrument information, error messages, etc. and enables setting of some instrument settings.

❙❙ Opening/closing settings screen
If the settings screen is not shown, press
When the settings screen is shown, press

to open the settings screen.
to close the settings screen and cancel any incomplete settings.

❙❙ Settings screen layout
The settings screen is divided into 4 main parts.
Selection area for selecting the item to view or set.
Value/setting area for displaying the value and changing the setting of the selected item.
Message area for showing messages related to operation of the display panel.
Key function description area describing the functions of the keys in the current operation.

29

Settings screen❙❙ Setting items
Serial No.

The serial number of the instrument. Not changeable.

VersionThe version of the FD-9 firmware. Not changeable.

MAC address

The MAC address of the FD-9’s network interface board. Not changeable.

Network name

The network name assigned to the FD-9. It can be set or changed from within the software. For details, refer tothe software instruction manual.

Display language(See p. 31.)English:English日本語:
Japanese
中文:
Chinese (Simplified)Deutsch:GermanFrançais:
FrenchEspañol:
SpanishItaliano:
Italianpolski:
PolishРусский:Russian
Português: Portuguese• Change is applied immediately after

DHCP(See p. 26.)

ON: The network settings are automatically assigned using DHCP
OFF: Use of DHCP is disabled

IP address(See p. 24.)

IP address (xxx.xxx.xxx.xxx) set on FD-9. It can be set using the,, andsoftware when FD-9 is connected.
• Disabled if DHCP is set to “ON”
• Changes made using the buttons are not applied until SAVE DATA is performed.

Subnet mask
(See p. 24.)Subnet mask (xxx.xxx.xxx.xxx) set on FD-9. It can be set using the,, andsoftware when FD-9 is connected.
• Disabled if DHCP is set to “ON”
• Changes made using the buttons are not applied until SAVE DATA is performed.

Default gateway(See p. 24.)

Default gateway (xxx.xxx.xxx.xxx) set on FD-9. It can be set using the,, andsoftware when FD-9 is connected.
• Disabled if DHCP is set to “ON”
• Changes made using the buttons are not applied until SAVE DATA is performed.

SAVE DATA

Saves changes to network settings (DHCP, IP address, subnet mask, and default gateway), and reboots theFD-9 to apply the new settings.

is pressed.

buttons or frombuttons or frombuttons or from❙❙ Selecting and changing settingsPressorto move the cursor to the desired item. The value or current setting of the selected item will appear in the value/settingarea.
To change the selected item, refer to the page specified for that item in the table above.
• Setting items which are shown in gray in the settings screen cannot be changed.

30

Setting display language
The language shown on the display panel can be selected according to the following procedure.

1

Press

2

Press
or
repeatedly to select “Display language” and press. A list of the language selections will appear, with the current
setting highlighted.

3

Press

4

Press
to confirm the selection. The display screen will change
immediately to the selected language.

to open the settings screen.

or

repeatedly to select the desired language.

31

Troubleshooting❙❙ Error messages
The messages below may be displayed when using the instrument. When one of these messages appears, please take the action indicatedbelow. If the message continues to appear even after taking the recommended action, contact the nearest KONICA MINOLTA authorized
service facility.
Error message

Possible cause/problem

ActionPaper jammed

Paper has jammed in the FD-9.

Follow the procedure below to clear the paper jam.

Paper has jammed in the Auto Sheet Feeder.

Follow the procedure on p. 32 to clear the paper jam.

Cover opened

Cover of FD-9 is open.

Close the cover and wait until FD-9 initialization is complete.

Fatal Error
Error code: ********
Warning code: ********

A fatal error has occurred.

Switch off the FD-9, wait 30 seconds, and then switch it backon. If the error continues to occur, note the Error code andWarning code numbers, and contact the nearest KONICAMINOLTA authorized service facility❙❙ Clearing paper jams
If the paper becomes jammed during measurement, follow the procedures below to remove the jammed paper.

When using FD-9 without Auto Sheet Feeder
When using the FD-9 without the Auto Sheet Feeder, the location of paper jams is usually where the paper is fed between the base and thescan mechanism.

32

1

Press
to eject the paper. If the paper cannot beejected by the FD-9, continue with the steps below tomanually clear the paper jam.

2

Switch off the FD-9.

3

Disconnect all cables (USB cable, LAN cable, AC adapter
cable).

4

Open the cover.

5

Loosen the two screws as shown completely so that they
are free of the base.

6

Close the cover.

7

Grasp the ends of the FD-9 and tilt the FD-9 bodybackward as shown. The base should stay flat.

8

Remove the jammed paper.

Troubleshooting

9

Grasp the ends of the FD-9 and tilt the FD-9 body onto thebase again.

10 Open the cover
11 Tighten the two screws as shown until snug.

• If the screws are loose when measurements are taken, measurement
accuracy may suffer.
• Do not overtighten.

12 Close the cover.
13 Reconnect the cables and switch the FD-9 back on.
When using FD-9 with Auto Sheet Feeder
When using the FD-9 with the Auto Sheet Feeder, the location of paper jams is usually in one of the following locations: Where the paper
feeds from the supply tray into the Auto Sheet Feeder and then into the FD-9, within the FD-9, or where the paper feeds from the FD-9
through the Auto Sheet Feeder into the exit tray. To clear paper jams, check each section.

1

Switch off the FD-9.

2

Disconnect all cables (USB cable, LAN cable, AC adapter
cable).

3

Remove the FD-9 from the Auto Sheet Feeder.

4

If paper is still jammed, lift the lever on the left side of the
supply tray and swing the supply tray mechanism out tothe right.

5

Remove any paper from inside the Auto Sheet Feeder.

• If the paper jam is inside the FD-9, follow the procedure in the preceding
section “When using FD-9 without Auto Sheet Feeder” to remove thejammed paper.

33

Troubleshooting

6

Pull the handle on the inside cover to open the inside
panel and check that there is no paper inside the panel. If
there is paper there, remove it.

7

Pull up on the handle to open the top cover and checkthat there is no paper inside the panel. If there is paper
there, remove it.

8

Close the top cover and inner cover, and then swing the
supply tray mechanism closed again.

9

If there is paper jammed on the exit tray side, remove it.

10 Replace the FD-9 in the Auto Sheet Feeder, connect all
cables again, and switch the FD-9 back on.

34

Cleaning❙❙ Cleaning the white calibration plate
If the white calibration plate becomes dirty, it will affect measurement accuracy. If the white calibration plate becomes dirty, it should becleaned as described below.

1

Make sure that the FD-9 is switched off.

2

Disconnect all cables (USB cable, LAN cable, AC adapter cable).

3
4Open the cover.

5

Close the cover.

6

Grasp the ends of the FD-9 and tilt the FD-9 bodybackward as shown. The base should stay flat.

7

Wipe the surface of the white calibration plate with aclean, dry, soft, lint-free cloth or lens-cleaning tissue.

Loosen the two screws as shown completely so that they
are free of the base.

• If the white calibration plate is stained, lens-cleaning fluid may be used totry to remove the stain.
• Never use benzene or other solvents to clean the white calibration plate.
Doing so may cause discoloration or damage to the plate.
• If a stain cannot be removed, contact the nearest Konica Minolta service
facility.

8

Grasp the ends of the FD-9 and tilt the FD-9 body onto thebase again.

9

Open the cover

10 Tighten the two screws as shown until snug.

• If the screws are loose when measurements are taken, measurement
accuracy may suffer.
• Do not overtighten.

11 Close the cover and reconnect cables.

35

Cleaning❙❙ Cleaning inside the FD-9
If paper dust, etc. accumulates inside the FD-9, it may cause paper jams or problems with patch recognition. If paper jams or patch
recognition problems start to occur more frequently, the inside of the FD-9 should be cleaned according to the procedure below.

1

Make sure that the FD-9 is switched off.

2

Disconnect all cables (USB cable, LAN cable, AC adapter cable).

3
4Open the cover.

5

Close the cover.

6

Grasp the ends of the FD-9 and tilt the FD-9 bodybackward as shown. The base should stay flat.

7

Clean the cover glass of the line sensor and the whitebacking and other plastic surfaces with a clean, dry, soft,
lint-free cloth.

Loosen the two screws as shown completely so that they
are free of the base.

• If the white backing is extremely dirty, clean the white backing with a soft
cloth moistened with alcohol.

8

Wipe off the rubber rollers with a soft cloth moistenedwith alcohol to clean and condition them.

9

Grasp the ends of the FD-9 and tilt the FD-9 body onto thebase again.

10 Open the cover
11 Tighten the two screws as shown until snug.

• If the screws are loose when measurements are taken, measurement
accuracy may suffer.
• Do not overtighten.

12 Close the cover and reconnect cables.

36

Scan Measurement Chart Conditions
The FD-9 can scan charts conforming to the following conditions:• Even if paper conditions and chart layout conditions meet the conditions below, some charts may not be measurable.

When feeding test charts using FD-9 alone

When using optional
Auto Sheet Feeder FD-A09

Paper width

45 to 330 mm

90 to 320 mm

Paper length

170 to 1500 mm

170 to 450 mm

Paper thickness

0.05 to 0.45 mm

0.08 to 0.25 mm *1

Minimum patch size

6 × 6 mm

Maximum number of patches per pageA4-size paper: 1394
A3-size paper: 2928

• Positioning markers are not necessary.
*1 When using optional Auto Sheet Feeder FD-A09, the maximum recommended number of loaded sheets is:		 Normal paper (80 g/m2): 100 sheets
		 Heavy paper (130 g/m2): 70 sheets
The layout of the chart on the paper must conform to the following layout. (See below.)
Minimum leading margin

23 mm

Minimum tailing margin

33 mm

Minimum side margins

4 mmHorizontal shift

Within 150 mm of paper center line

Into FD-9

Paper center line

Leading margin

Chart area
Chart width

Horizontal shift

Side margin

Side margin

Trailing margin\37

Dimensions
Auto Scan Spectrophotometer FD-9

(mm)

199

716

457

39

159

251

337

175

343
205

601

Auto Sheet Feeder FD-A09

1158
339

461

235

38

245

200

173

121
108

156

225

Specifications
Model

FD-9

Illumination/ viewing system

45°a:0° (annular illumination) *1
Conforms to CIE No. 15, ISO 7724/1, DIN5033 Teil 7, ASTM E 1164, and JIS Z 8722 Condition afor reflectance measurements.

Spectral separation device

Concave grating

Wavelength range

380 to 730 nm

Wavelength pitch

10 nm

Half bandwidth

Approx. 10 nm

Measurement area

Approx. Ø3 mm

Measurement light source

LED

Measurement range

Reflectance: 0 to 150%

Short-term repeatability

Colorimetric: Within σΔE00 0.05
(Under Konica Minolta test conditions where a white calibration plate is measured 30 times at
10-second intervals after white calibration has been performed.)Inter-instrument agreement

Within ΔE00 0.3(Average of 12 BCRA Series II color tiles compared to values measured with a master body under
Konica Minolta standard conditions.)

Measurement time

Approx. 4 min. for 1500 patches(According to Konica Minolta standard conditions) *3

Measurable charts *4(When manually feeding charts)

Paper width

45 to 330 mm

Paper length

170 to 1500 mm

Paper thickness

0.05 to 0.45 mm

Minimum patch size

6 × 6 mm

Maximum number of patches per
sheet of paper

A4-sized paper: 1394
A3-sized paper: 2928

Margins (minimum)

Leading: 23 mm; Trailing: 33 mm; Left/right sides: 4 mm each

Output item

Spectral reflectance

Illumination conditions *2

M0 (A), M1 (D50), M2 (A+UV filter), C, ID50, D65, ID65, F2, F6, F7, F8, F9, F10, F11, F12,
User-defined illuminant

Backing condition

White backing, compliant with ISO 13655

InterfaceUSB2.0; 100Base-TX

Power

Dedicated AC adapter: Input: AC 100 to 240 V 50/60 Hz
Output: DC 24 V 5 A

Dimensions(W × D × H)

Main body only: 716 × 251 × 159 mm
Auto Sheet Feeder FD-A09 (optional accessory) only: 601 × 1158 (with trays extended) × 245 mm

WeightMain body only: Approx. 10.5 kg
Auto Sheet Feeder FD-A09 (optional accessory) only: Approx. 18.0 kg

Operating temperature/ humidity range

10 to 35°C, 30 to 85% relative humidity with no condensation

Storage temperature/ humidity range

0 to 45°C, 0 to 85% relative humidity with no condensation

Standard accessories

AC adapter, USB cable, Measurement Utility Software FD-S2w, Paper Guide

Optional accessories

Auto Sheet Feeder FD-A09*1 Illumination for wavelengths below 400 nm is unidirectional.
*2 M0, M1, and M2 illumination conditions conform to the illumination conditions in ISO 13655 Section 4.2.2 Illumination
requirements and measurement conditions.
*3 Paper size: A3; Patch size: 6 × 6 mm; Patch distribution: 32 rows × 47 columns*4 Even if paper size and margins are within the ranges specified, some charts may not be measurable.

39

< CAUTION >KONICA MINOLTA WILL NOT BE LIABLE FOR ANY DAMAGES RESULTING FROM THE MISUSE, MISHANDLING,UNAUTHORIZED MODIFICATION, ETC. OF THIS PRODUCT, OR FOR ANY INDIRECT OR INCIDENTAL DAMAGES(INCLUDING BUT NOT LIMITED TO LOSS OF BUSINESS PROFITS, INTERRUPTION OF BUSINESS, ETC.) DUE TO
THE USE OF OR INABILITY TO USE THIS PRODUCT.

E 9222-A8AN-13©2015-2016 KONICA MINOLTA, INC.

BGGBGX

Printed in Japan