Battery BA19 Lead Acid
Data sheet

LINAK.COM/MEDLINE-CARELINEBA19 Lead Acid
The BA19 lead acid backup battery has been developed
specifically for use with the CA/CO control box platform. It
is a compact and cost-efficient battery with built-in charger
and cable management.
A further advantage of the BA19 lead acid battery is thefunctional attachment of the battery by means of the LINAK
mounting bracket and a securing screw on the application.
The battery provides charging status information to thecontrol box which makes it possible to receive an indication
on OpenBus™ system hand controls.

Features and options:• Lead acid technology: Lead acid
• Output voltage: 24 V DC
• Max. discharge current: 6 A• Charging current: 300 mA, controlled by BA19 internal charger• Capacity: 1.2 Ah/28.8 Wh• Weight: 1.6 kg• Housing colour: Light grey (RAL 7035)• Protection class: IPX6, IPX6 Washable DURATM• Packaging: Multibox• PCP version: PCP 2.0

LINAK.COM/MEDLINE-CARELINE

Usage:• Compatibility: Battery backup for CA/CO control box platform• Duty cycle: 10%, 2 minutes continuous use followed by18 minutes not in use• Charging: Via integrated charger• Charging time: Approx. 6 hours• Recharging during storage: Battery recharging no later than
6 months after production date stated on the label• Operating temperature: +5 °C to +40 °C• Storage temperature: -5 °C to +40 °C
The batteries must be stored in an applicable storage room
to avoid direct sunlight• Relative humidity: 20% to 80% - non-condensing• Atmospheric pressure: 700 to 1060 hPa• Height above sea level: Max. 3000 meters• Service: Battery cells cannot be replaced as the battery covercannot be closed properly afterwards• Approvals (pending): IEC60601-1:2005 3rd edition,ANSI/AAMI ES60601-1: 2005, 3rd edition
CAN/CSA-22.2 No. 60601-1:2008

LED functionality:
LED

Indication of operation

Solid yellow

Charging (battery not ready)No LED light

Fully charged (battery ready)Flashing yellow

Error during charging

Buzzer functionality:
The buzzer will make a warning when a button on the hand control is pressed and the battery capacity is low.
The buzzer can also be activated by an intelligent control box to signal other conditions. This must be specified in the control box software.

BA19Ordering example:

BA1916111100
Not usedNot used
Enclosure

1 = Standard Enclosure (PP)Battery type

1 = Lead Acid

PCBA

1 = Analogue PCBA (PCP 1.0)
2 = Digital PCBA (PCP 2.0)

Label

1 = Standard BA19 label
X = Custom label

IP degree

6 = IPX6
W = IPX6 Washable DURA™

Colour

1 = RAL7035

BA19 Article

Dimensions:

Drawing no. 1020w4001

BA19 mounting possibilities:
Battery mounted on platewith control box

Cable management

Mounting instructions:

BA19 must be mounted with attachment screw and mounting
bracket (see below) due to the battery weight.
Screw M4 countersunk
torque 1.1Nm +/- 0.1Nm.

45°dkdkdM4dk8.4

k2.7
According to ISO 2009

Mounting bracket (frame flat) - article no. 1015W1001:

Copyright ® LINAK 2020.02 . MA M9-02-657-D . Chapter 10.43
Terms of use
The user is responsible for determining the suitability of LINAK products for specific application. LINAK takes great care in providing
accurate and up-to-date information on its products.
However, due to continuous development in order to improve its products, LINAK products are subject to frequent modifications andchanges without prior notice. Therefore, LINAK cannot guarantee the correct and actual status of said information on its products.
While LINAK uses its best efforts to fulfil orders, LINAK cannot, for the same reasons as mentioned above, guarantee the availability ofany particular product. Therefore, LINAK reserves the right to discontinue the sale of any product displayed on its website or listed in its
catalogues or other written material drawn up by LINAK.
All sales are subject to the Standard Terms of Sale and Delivery for LINAK. For a copy hereof, please contact LINAK.

FOR MOUNTING INSTRUCTIONS AND GUIDANCE IN USAGE, PLEASE SEE THE RELEVANT USER MANUAL