APPENDIX B:

SAR DIPOLE VERIFICATION PLOTSELEMENT
DUT: Dipole 750.0 MHz; Type: D750V3 - SN1054
Communication System: UID: 0, CW; Frequency: 750.0 MHz
Medium: 750 Body; Medium parameters used:
f = 750.0 MHz; cond = 0.968 S/m; perm = 54.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 04/13/2022; Ambient Temp: 22.9°C; Tissue Temp: 21.4°C
Probe: EX3DV4 - SN7552; ConvF:(10.12,10.12,10.12); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

750 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 2.84 W/kg
SAR(1 g) = 1.80 W/kg
Deviation (1 g) = 4.29%

B1

ELEMENT
DUT: Dipole 750.0 MHz; Type: D750V3 - SN1054
Communication System: UID: 0, CW; Frequency: 750.0 MHz
Medium: 750 Body; Medium parameters used:
f = 750.0 MHz; cond = 0.963 S/m; perm = 54.3; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 04/15/2022; Ambient Temp: 23.1°C; Tissue Temp: 22.3°C
Probe: EX3DV4 - SN7552; ConvF:(10.12,10.12,10.12); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

750 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 2.82 W/kg
SAR(1 g) = 1.79 W/kg
Deviation (1 g) = 3.71%

B2

ELEMENT
DUT: Dipole 750.0 MHz; Type: D750V3 - SN1057
Communication System: UID: 0, CW; Frequency: 750.0 MHz
Medium: 750 Body; Medium parameters used:
f = 750.0 MHz; cond = 0.929 S/m; perm = 55.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 04/27/2022; Ambient Temp: 22.5°C; Tissue Temp: 23.4°C
Probe: EX3DV4 - SN7416; ConvF:(9.89,9.89,9.89); Calibrated: 2021-05-18
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn701; Calibrated: 2021-05-11
Phantom: Twin-SAM V8.0; Serial: 1357
Measurement SW: DASY Module SAR V16.0.2.136

750 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 2.55 W/kg
SAR(1 g) = 1.69 W/kg
Deviation (1 g) = -2.20%

B3

ELEMENT
DUT: Dipole 750.0 MHz; Type: D750V3 - SN1057
Communication System: UID: 0, CW; Frequency: 750.0 MHz
Medium: 750 Body; Medium parameters used:
f = 750.0 MHz; cond = 0.954 S/m; perm = 54.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/02/2022; Ambient Temp: 22.9°C; Tissue Temp: 21.7°C
Probe: EX3DV4 - SN7416; ConvF:(9.89,9.89,9.89); Calibrated: 2021-05-18
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn701; Calibrated: 2021-05-11
Phantom: Twin-SAM V8.0; Serial: 1357
Measurement SW: DASY Module SAR V16.0.2.136

750 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 2.47 W/kg
SAR(1 g) = 1.63 W/kg
Deviation (1 g) = -5.67%

B4

ELEMENT
DUT: Dipole 750.0 MHz; Type: D750V3 - SN1057
Communication System: UID: 0, CW; Frequency: 750.0 MHz
Medium: 750 Body; Medium parameters used:
f = 750.0 MHz; cond = 0.949 S/m; perm = 53.0; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/03/2022; Ambient Temp: 22.4°C; Tissue Temp: 21.8°C
Probe: EX3DV4 - SN3837; ConvF:(8.85,8.85,8.85); Calibrated: 2022-01-19
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn793; Calibrated: 2022-01-13
Phantom: Twin-SAM V8.0; Serial: 2034
Measurement SW: DASY Module SAR V16.0.2.136

750 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 2.81 W/kg
SAR(1 g) = 1.84 W/kg
Deviation (1 g) = 6.48%

B5

ELEMENT
DUT: Dipole 750.0 MHz; Type: D750V3 - SN1097
Communication System: UID: 0, CW; Frequency: 750.0 MHz
Medium: 750 Body; Medium parameters used:
f = 750.0 MHz; cond = 0.937 S/m; perm = 55.4; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/16/2022; Ambient Temp: 25.0°C; Tissue Temp: 21.3°C
Probe: EX3DV4 - SN3837; ConvF:(8.85,8.85,8.85); Calibrated: 2022-01-19
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn793; Calibrated: 2022-01-13
Phantom: Twin-SAM V8.0; Serial: 2034
Measurement SW: DASY Module SAR V16.0.2.136

750 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 2.63 W/kg
SAR(1 g) = 1.75 W/kg
Deviation (1 g) = 4.04%

B6

ELEMENT
DUT: Dipole 750.0 MHz; Type: D750V3 - SN1097
Communication System: UID: 0, CW; Frequency: 750.0 MHz
Medium: 750 Body; Medium parameters used:
f = 750.0 MHz; cond = 0.954 S/m; perm = 54.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/18/2022; Ambient Temp: 23.3°C; Tissue Temp: 21.5°C
Probe: EX3DV4 - SN3837; ConvF:(8.85,8.85,8.85); Calibrated: 2022-01-19
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn793; Calibrated: 2022-01-13
Phantom: Twin-SAM V8.0; Serial: 2034
Measurement SW: DASY Module SAR V16.0.2.136

750 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 2.72 W/kg
SAR(1 g) = 1.73 W/kg
Deviation (1 g) = 2.85%

B7

ELEMENT
DUT: Dipole 835.0 MHz; Type: D835V2 - SN4d132
Communication System: UID: 0, CW; Frequency: 835.0 MHz
Medium: 835 Body; Medium parameters used:
f = 835.0 MHz; cond = 1.01 S/m; perm = 54.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 04/27/2022; Ambient Temp: 22.2°C; Tissue Temp: 21.6°C
Probe: EX3DV4 - SN7552; ConvF:(9.86,9.86,9.86); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

835 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 3.31 W/kg
SAR(1 g) = 2.07 W/kg
Deviation (1 g) = 5.50%

B8

ELEMENT
DUT: Dipole 835.0 MHz; Type: D835V2 - SN4d040
Communication System: UID: 0, CW; Frequency: 835.0 MHz
Medium: 835 Body; Medium parameters used:
f = 835.0 MHz; cond = 0.970 S/m; perm = 52.7; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 04/27/2022; Ambient Temp: 21.5°C; Tissue Temp: 21.9°C
Probe: EX3DV4 - SN7639; ConvF:(10.45,10.45,10.45); Calibrated: 2021-11-16
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1646; Calibrated: 2021-11-11
Phantom: Twin-SAM V8.0; Serial: 1736
Measurement SW: DASY Module SAR V16.0.2.136

835 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm;
Peak SAR (extrapolated) = 3.14 W/kg
SAR(1 g) = 2.00 W/kg
Deviation (1 g) = 4.93%

B9

ELEMENT
DUT: Dipole 835.0 MHz; Type: D835V2 - SN4d040
Communication System: UID: 0, CW; Frequency: 835.0 MHz
Medium: 835 Body; Medium parameters used:
f = 835.0 MHz; cond = 0.968 S/m; perm = 52.8; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/04/2022; Ambient Temp: 22.5°C; Tissue Temp: 21.0°C
Probe: EX3DV4 - SN7639; ConvF:(10.45,10.45,10.45); Calibrated: 2021-11-16
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1646; Calibrated: 2021-11-11
Phantom: Twin-SAM V8.0; Serial: 1736
Measurement SW: DASY Module SAR V16.0.2.136

835 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 3.17 W/kg
SAR(1 g) = 2.01 W/kg
Deviation (1 g) = 5.46%

B10

ELEMENT
DUT: Dipole 835.0 MHz; Type: D835V2 - SN4d133
Communication System: UID: 0, CW; Frequency: 835.0 MHz
Medium: 835 Body; Medium parameters used:
f = 835.0 MHz; cond = 0.942 S/m; perm = 55.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/23/2022; Ambient Temp: 22.1°C; Tissue Temp: 21.5°C
Probe: EX3DV4 - SN7571; ConvF:(9.87,9.87,9.87); Calibrated: 2021-12-10
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn859; Calibrated: 2021-12-08
Phantom: Twin-SAM V5.0; Serial: 1646
Measurement SW: DASY Module SAR V16.0.2.136

835 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 3.20 W/kg
SAR(1 g) = 2.04 W/kg
Deviation (1 g) = 5.26%

B11

ELEMENT
DUT: Dipole 850.0 MHz; Type: D850V2 - SN1010
Communication System: UID: 0, CW; Frequency: 850.0 MHz
Medium: 835 Body; Medium parameters used:
f = 850.0 MHz; cond = 0.980 S/m; perm = 52.7; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/06/2022; Ambient Temp: 22.7°C; Tissue Temp: 22.0°C
Probe: EX3DV4 - SN7639; ConvF:(10.45,10.45,10.45); Calibrated: 2021-11-16
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1646; Calibrated: 2021-11-11
Phantom: Twin-SAM V8.0; Serial: 1736
Measurement SW: DASY Module SAR V16.0.2.136

850 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 3.28 W/kg
SAR(1 g) = 2.07 W/kg
Deviation (1 g) = 3.81%

B12

ELEMENT
DUT: Dipole 850.0 MHz; Type: D850V2 - SN1009
Communication System: UID: 0, CW; Frequency: 850.0 MHz
Medium: 835 Body; Medium parameters used:
f = 850.0 MHz; cond = 1.01 S/m; perm = 57.3; density = 1000 kg/m3
Phantom Section: Flat; Space: 15 mm
Test Date: 05/24/2022; Ambient Temp: 20.7°C; Tissue Temp: 20.1°C
Probe: EX3DV4 - SN7674; ConvF:(10.12,10.12,10.12); Calibrated: 2021-09-06
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1683; Calibrated: 2021-08-06
Phantom: Twin-SAM V8.0; Serial: 2071
Measurement SW: DASY Module SAR V16.0.2.83

850 MHz System Verification at 23 dBm (200 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 3.14 W/kg
SAR(1 g) = 2.08 W/kg
Deviation (1 g) = 5.05%

B13

ELEMENT
DUT: Dipole 1750.0 MHz; Type: D1750V2 - SN1104
Communication System: UID: 0, CW; Frequency: 1750.0 MHz
Medium: 1750 Body; Medium parameters used:
f = 1750.0 MHz; cond = 1.47 S/m; perm = 52.5; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/09/2022; Ambient Temp: 22.3°C; Tissue Temp: 20.5°C
Probe: EX3DV4 - SN7532; ConvF:(8.51,8.51,8.51); Calibrated: 2022-04-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn501; Calibrated: 2022-04-13
Phantom: Twin-SAM V8.0; Serial: 1357
Measurement SW: DASY Module SAR V16.0.2.136

1750 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 6.80 W/kg
SAR(1 g) = 3.73 W/kg
Deviation (1 g) = 2.75%

B14

ELEMENT
DUT: Dipole 1750.0 MHz; Type: D1750V2 - SN1104
Communication System: UID: 0, CW; Frequency: 1750.0 MHz
Medium: 1750 Body; Medium parameters used:
f = 1750.0 MHz; cond = 1.43 S/m; perm = 52.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/16/2022; Ambient Temp: 20.2°C; Tissue Temp: 20.4°C
Probe: EX3DV4 - SN7532; ConvF:(8.51,8.51,8.51); Calibrated: 2022-04-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn501; Calibrated: 2022-04-13
Phantom: Twin-SAM V8.0; Serial: 1357
Measurement SW: DASY Module SAR V16.0.2.136

1750 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 6.85 W/kg
SAR(1 g) = 3.80 W/kg
Deviation (1 g) = 4.68%

B15

ELEMENT
DUT: Dipole 1750.0 MHz; Type: D1765V2 - SN1008
Communication System: UID: 0, CW; Frequency: 1750.0 MHz
Medium: 1750 Body; Medium parameters used:
f = 1750.0 MHz; cond = 1.51 S/m; perm = 51.8; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/24/2022; Ambient Temp: 23.2°C; Tissue Temp: 21.3°C
Probe: EX3DV4 - SN7670; ConvF:(8.36,8.36,8.36); Calibrated: 2021-08-05
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1681; Calibrated: 2021-08-03
Phantom: Twin-SAM V8.0; Serial: 1966
Measurement SW: DASY Module SAR V16.0.2.136

1750 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 7.59 W/kg
SAR(1 g) = 3.93 W/kg
Deviation (1 g) = 3.97%

B16

ELEMENT
DUT: Dipole 1750.0 MHz; Type: D1750V2 - SN1104
Communication System: UID: 0, CW; Frequency: 1750.0 MHz
Medium: 1750 Body; Medium parameters used:
f = 1750.0 MHz; cond = 1.51 S/m; perm = 51.1; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/25/2022; Ambient Temp: 20.3°C; Tissue Temp: 19.0°C
Probe: EX3DV4 - SN7532; ConvF:(8.51,8.51,8.51); Calibrated: 2022-04-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn501; Calibrated: 2022-04-13
Phantom: Twin-SAM V8.0; Serial: 1357
Measurement SW: DASY Module SAR V16.0.2.136

1750 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 6.31 W/kg
SAR(1 g) = 3.43 W/kg
Deviation (1 g) = -5.51%

B17

ELEMENT
DUT: Dipole 1900.0 MHz; Type: D1900V2 - SN5d180
Communication System: UID: 0, CW; Frequency: 1900.0 MHz
Medium: 1900 Body; Medium parameters used:
f = 1900.0 MHz; cond = 1.57 S/m; perm = 51.0; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/03/2022; Ambient Temp: 21.5°C; Tissue Temp: 23°C
Probe: EX3DV4 - SN7308; ConvF:(8.58,8.58,8.58); Calibrated: 2022-02-21
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn467; Calibrated: 2022-02-24
Phantom: Twin-SAM V8.0; Serial: 2029
Measurement SW: DASY Module SAR V16.0.2.136

1900 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 7.59 W/kg
SAR(1 g) = 4.03 W/kg
Deviation (1 g) = 3.33%

B18

ELEMENT
DUT: Dipole 1900.0 MHz; Type: D1900V2 - SN5d180
Communication System: UID: 0, CW; Frequency: 1900.0 MHz
Medium: 1900 Body; Medium parameters used:
f = 1900.0 MHz; cond = 1.54 S/m; perm = 52.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/05/2022; Ambient Temp: 21.3°C; Tissue Temp: 22.6°C
Probe: EX3DV4 - SN7308; ConvF:(8.58,8.58,8.58); Calibrated: 2022-02-21
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn467; Calibrated: 2022-02-24
Phantom: Twin-SAM V8.0; Serial: 2029
Measurement SW: DASY Module SAR V16.0.2.136

1900 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 7.40 W/kg
SAR(1 g) = 3.83 W/kg
Deviation (1 g) = -1.79%

B19

ELEMENT
DUT: Dipole 1900.0 MHz; Type: D1900V2 - SN5d181
Communication System: UID: 0, CW; Frequency: 1900.0 MHz
Medium: 1900 Body; Medium parameters used:
f = 1900.0 MHz; cond = 1.57 S/m; perm = 52.1; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/10/2022; Ambient Temp: 21.9°C; Tissue Temp: 23.8°C
Probe: EX3DV4 - SN7546; ConvF:(7.89,7.89,7.89); Calibrated: 2022-04-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1402; Calibrated: 2022-04-14
Phantom: Twin-SAM V8.0; Serial: 2070
Measurement SW: DASY Module SAR V16.0.0.116

1900 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 7.68 W/kg
SAR(1 g) = 4.16 W/kg
Deviation (1 g) = 4.79%

B20

ELEMENT
DUT: Dipole 1900.0 MHz; Type: D1900V2 - SN5d181
Communication System: UID: 0, CW; Frequency: 1900.0 MHz
Medium: 1900 Body; Medium parameters used:
f = 1900.0 MHz; cond = 1.51 S/m; perm = 51.8; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/10/2022; Ambient Temp: 24.2°C; Tissue Temp: 20.7°CProbe: EX3DV4 - SN7532; ConvF:(8.19,8.19,8.19); Calibrated: 2022-04-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn501; Calibrated: 2022-04-13
Phantom: Twin-SAM V8.0; Serial: 1357
Measurement SW: DASY Module SAR V16.0.2.136

1900 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 6.73 W/kg
SAR(1 g) = 3.68 W/kg
Deviation (1 g) = -7.30%

B21

ELEMENT
DUT: Dipole 1900.0 MHz; Type: D1900V2 - SN5d181
Communication System: UID: 0, CW; Frequency: 1900.0 MHz
Medium: 1900 Body; Medium parameters used:
f = 1900.0 MHz; cond = 1.58 S/m; perm = 52.1; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/12/2022; Ambient Temp: 21.9°C; Tissue Temp: 22.1°C
Probe: EX3DV4 - SN7546; ConvF:(7.89,7.89,7.89); Calibrated: 2022-04-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1402; Calibrated: 2022-04-14
Phantom: Twin-SAM V8.0; Serial: 2070
Measurement SW: DASY Module SAR V16.0.2.83

1900 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 90.0): Measurement grid: dx=10.0 mm, dy=15.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=6.0 mm, dy=6.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 7.66 W/kg
SAR(1 g) = 4.13 W/kg
Deviation (1 g) = 4.03%

B22

ELEMENT
DUT: Dipole 2300.0 MHz; Type: D2300V2 - SN1064
Communication System: UID: 0, CW; Frequency: 2300.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2300.0 MHz; cond = 1.80 S/m; perm = 53.8; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/05/2022; Ambient Temp: 21.9°C; Tissue Temp: 20.6°C
Probe: EX3DV4 - SN7421; ConvF:(7.5,7.5,7.5); Calibrated: 2022-03-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn604; Calibrated: 2022-03-22
Phantom: Twin-SAM V4.0; Serial: 1275
Measurement SW: DASY Module SAR V16.0.2.1362300 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 9.31 W/kg
SAR(1 g) = 4.93 W/kg
Deviation (1 g) = 1.86%

B23

ELEMENT
DUT: Dipole 2300.0 MHz; Type: D2300V2 - SN1064
Communication System: UID: 0, CW; Frequency: 2300.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2300.0 MHz; cond = 1.79 S/m; perm = 52.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/11/2022; Ambient Temp: 23.5°C; Tissue Temp: 21.2°C
Probe: EX3DV4 - SN7427; ConvF:(7.28,7.28,7.28); Calibrated: 2022-02-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1403; Calibrated: 2022-02-22
Phantom: Twin-SAM V8.0; Serial: 1944
Measurement SW: DASY Module SAR V16.0.2.1362300 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 9.59 W/kg
SAR(1 g) = 4.72 W/kg
Deviation (1 g) = -2.48%

B24

ELEMENT
DUT: Dipole 2450.0 MHz; Type: D2450V2 - SN719
Communication System: UID: 0, CW; Frequency: 2450.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2450.0 MHz; cond = 2.03 S/m; perm = 52.5; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 03/17/2022; Ambient Temp: 24.0°C; Tissue Temp: 22.9°C
Probe: EX3DV4 - SN7406; ConvF:(7.43,7.43,7.43); Calibrated: 2021-07-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1676; Calibrated: 2021-06-21
Phantom: Twin-SAM V8.0; Serial: 2058
Measurement SW: DASY Module SAR V16.0.0.65

2450 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 9.62 W/kg
SAR(1 g) = 4.81 W/kg
Deviation (1 g) = -7.50%

B25

ELEMENT
DUT: Dipole 2450.0 MHz; Type: D2450V2 - SN719
Communication System: UID: 0, CW; Frequency: 2450.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2450.0 MHz; cond = 1.90 S/m; perm = 51.1; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/06/2022; Ambient Temp: 22.0°C; Tissue Temp: 23.6°C
Probe: EX3DV4 - SN7552; ConvF:(7.44,7.44,7.44); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.0.65

2450 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 10.2 W/kg
SAR(1 g) = 4.90 W/kg
Deviation (1 g) = -5.77%

B26

ELEMENT
DUT: Dipole 2450.0 MHz; Type: D2450V2 - SN797
Communication System: UID: 0, CW; Frequency: 2450.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2450.0 MHz; cond = 2.04 S/m; perm = 51.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/28/2022; Ambient Temp: 20.8°C; Tissue Temp: 20.8°C
Probe: EX3DV4 - SN7552; ConvF:(7.44,7.44,7.44); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2450 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 10.8 W/kg
SAR(1 g) = 5.15 W/kg
Deviation (1 g) =4.25%

B27

ELEMENT
DUT: Dipole 2450.0 MHz; Type: D2450V2 - SN797
Communication System: UID: 0, CW; Frequency: 2450.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2450.0 MHz; cond = 1.99 S/m; perm = 50.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/01/2022; Ambient Temp: 23.0°C; Tissue Temp: 22.5°C
Probe: EX3DV4 - SN7552; ConvF:(7.44,7.44,7.44); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2450 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 10.7 W/kg
SAR(1 g) = 5.17 W/kg
Deviation (1 g) = 4.66%

B28

ELEMENT
DUT: Dipole 2450.0 MHz; Type: D2450V2 - SN797
Communication System: UID: 0, CW; Frequency: 2450.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2450.0 MHz; cond = 1.96 S/m; perm = 51.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/07/2022; Ambient Temp: 20.9°C; Tissue Temp: 24.0°C
Probe: EX3DV4 - SN7552; ConvF:(7.44,7.44,7.44); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2450 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 10.7 W/kg
SAR(1 g) = 5.30 W/kg
Deviation (1 g) = 7.29%

B29

ELEMENT
DUT: Dipole 2450.0 MHz; Type: D2450V2 - SN797
Communication System: UID: 0, CW; Frequency: 2450.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2450.0 MHz; cond = 1.93 S/m; perm = 51.3; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/10/2022; Ambient Temp: 23.4°C; Tissue Temp: 22.0°C
Probe: EX3DV4 - SN7552; ConvF:(7.44,7.44,7.44); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2450 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 9.99 W/kg
SAR(1 g) = 4.74 W/kg
Deviation (1 g) = -4.05%

B30

ELEMENT
DUT: Dipole 2600.0 MHz; Type: D2600V2 - SN1064
Communication System: UID: 0, CW; Frequency: 2600.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2600.0 MHz; cond = 2.18 S/m; perm = 51.0; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/28/2022; Ambient Temp: 20.8°C; Tissue Temp: 20.8°C
Probe: EX3DV4 - SN7552; ConvF:(7.28,7.28,7.28); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2600 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 11.4 W/kg
SAR(1 g) = 5.27 W/kg
Deviation (1 g) = -5.22%

B31

ELEMENT
DUT: Dipole 2600.0 MHz; Type: D2600V2 - SN1064
Communication System: UID: 0, CW; Frequency: 2600.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2600.0 MHz; cond = 2.11 S/m; perm = 49.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/01/2022; Ambient Temp: 23.0°C; Tissue Temp: 22.5°C
Probe: EX3DV4 - SN7552; ConvF:(7.28,7.28,7.28); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2600 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 11.4 W/kg
SAR(1 g) = 5.31 W/kg
Deviation (1 g) = -4.50%

B32

ELEMENT
DUT: Dipole 2600.0 MHz; Type: D2600V2 - SN1064
Communication System: UID: 0, CW; Frequency: 2600.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2600.0 MHz; cond = 2.16 S/m; perm = 51.5; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/04/2022; Ambient Temp: 21.9°C; Tissue Temp: 23.9°C
Probe: EX3DV4 - SN7552; ConvF:(7.28,7.28,7.28); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2600 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 11.5 W/kg
SAR(1 g) = 5.16 W/kg
Deviation (1 g) = -7.19%

B33

ELEMENT
DUT: Dipole 2600.0 MHz; Type: D2600V2 - SN1064
Communication System: UID: 0, CW; Frequency: 2600.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2600.0 MHz; cond = 2.16 S/m; perm = 50.6; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/07/2022; Ambient Temp: 20.9°C; Tissue Temp: 24.0°C
Probe: EX3DV4 - SN7552; ConvF:(7.28,7.28,7.28); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2600 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 11.5 W/kg
SAR(1 g) = 5.27 W/kg
Deviation (1 g) = -5.22%

B34

ELEMENT
DUT: Dipole 2600.0 MHz; Type: D2600V2 - SN1064
Communication System: UID: 0, CW; Frequency: 2600.0 MHz
Medium: 2450 Body; Medium parameters used:
f = 2600.0 MHz; cond = 2.14 S/m; perm = 50.7; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/10/2022; Ambient Temp: 23.4°C; Tissue Temp: 22.0°C
Probe: EX3DV4 - SN7552; ConvF:(7.28,7.28,7.28); Calibrated: 2021-09-20
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1680; Calibrated: 2021-08-04
Phantom: Twin-SAM V8.0; Serial: 2065
Measurement SW: DASY Module SAR V16.0.2.83

2600 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (30.0 x 30.0 x 30.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.5 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 12.6 W/kg
SAR(1 g) = 5.48 W/kg
Deviation (1 g) = -1.44%

B35

ELEMENT
DUT: Dipole 3500.0 MHz; Type: D3500V2 - SN1097
Communication System: UID: 0, CW; Frequency: 3500.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3500.0 MHz; cond = 3.17 S/m; perm = 49.5; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/27/2022; Ambient Temp: 20.5°C; Tissue Temp: 20.5°C
Probe: EX3DV4 - SN7661; ConvF:(6.7,6.7,6.7); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3500 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 16.9 W/kg
SAR(1 g) = 6.58 W/kg
Deviation (1 g) = 2.49%

B36

ELEMENT
DUT: Dipole 3500.0 MHz; Type: D3500V2 - SN1097
Communication System: UID: 0, CW; Frequency: 3500.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3500.0 MHz; cond = 3.20 S/m; perm = 51.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/02/2022; Ambient Temp: 21.3°C; Tissue Temp: 21.0°C
Probe: EX3DV4 - SN7661; ConvF:(6.7,6.7,6.7); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3500 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 16.5 W/kg
SAR(1 g) = 6.58 W/kg
Deviation (1 g) = 2.49%

B37

ELEMENT
DUT: Dipole 3700.0 MHz; Type: D3700V2 - SN1018
Communication System: UID: 0, CW; Frequency: 3700.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3700.0 MHz; cond = 3.39 S/m; perm = 49.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/27/2022; Ambient Temp: 20.5°C; Tissue Temp: 20.5°C
Probe: EX3DV4 - SN7661; ConvF:(6.66,6.66,6.66); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3700 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 17.0 W/kg
SAR(1 g) = 6.39 W/kg
Deviation (1 g) = 0.63%

B38

ELEMENT
DUT: Dipole 3700.0 MHz; Type: D3700V2 - SN1018
Communication System: UID: 0, CW; Frequency: 3700.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3700.0 MHz; cond = 3.43 S/m; perm = 50.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/02/2022; Ambient Temp: 21.3°C; Tissue Temp: 21.0°C
Probe: EX3DV4 - SN7661; ConvF:(6.66,6.66,6.66); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3700 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 15.9 W/kg
SAR(1 g) = 6.22 W/kg
Deviation (1 g) = -2.05%

B39

ELEMENT
DUT: Dipole 3700.0 MHz; Type: D3700V2 - SN1018
Communication System: UID: 0, CW; Frequency: 3700.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3700.0 MHz; cond = 3.42 S/m; perm = 50.5; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/05/2022; Ambient Temp: 21.0°C; Tissue Temp: 21.0°C
Probe: EX3DV4 - SN7661; ConvF:(6.66,6.66,6.66); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3700 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 17.4 W/kg
SAR(1 g) = 6.65 W/kg
Deviation (1 g) = 4.72%

B40

ELEMENT
DUT: Dipole 3700.0 MHz; Type: D3700V2 - SN1018
Communication System: UID: 0, CW; Frequency: 3700.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3700.0 MHz; cond = 3.44 S/m; perm = 49.3; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/30/2022; Ambient Temp: 22.0°C; Tissue Temp: 20.9°C
Probe: EX3DV4 - SN7661; ConvF:(6.66,6.66,6.66); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3700 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 17.2 W/kg
SAR(1 g) = 6.49 W/kg
Deviation (1 g) = 2.20%

B41

ELEMENT
DUT: Dipole 3900.0 MHz; Type: D3900V2 - SN1073
Communication System: UID: 0, CW; Frequency: 3900.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3900.0 MHz; cond = 3.67 S/m; perm = 50.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/05/2022; Ambient Temp: 21.0°C; Tissue Temp: 21.0°C
Probe: EX3DV4 - SN7661; ConvF:(6.51,6.51,6.51); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3900 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 18.9 W/kg
SAR(1 g) = 6.68 W/kg
Deviation (1 g) = 3.89%

B42

ELEMENT
DUT: Dipole 3900.0 MHz; Type: D3900V2 - SN1073
Communication System: UID: 0, CW; Frequency: 3900.0 MHz
Medium: 3600 Body; Medium parameters used:
f = 3900.0 MHz; cond = 3.68 S/m; perm = 48.9; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 05/30/2022; Ambient Temp: 22.0°C; Tissue Temp: 20.9°C
Probe: EX3DV4 - SN7661; ConvF:(6.51,6.51,6.51); Calibrated: 2021-06-28
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn1450; Calibrated: 2021-08-16
Phantom: Twin-SAM V5.0; Serial: 1692rightback
Measurement SW: DASY Module SAR V16.0.2.136

3900 MHz System Verification at 20 dBm (100 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (28.0 x 28.0 x 28.0): Measurement grid: dx=5.0 mm, dy=5.0 mm, dz=1.4 mm; Graded Ratio: 1.5
Peak SAR (extrapolated) = 18.6 W/kg
SAR(1 g) = 6.53 W/kg
Deviation (1 g) = 1.56%

B43

ELEMENT
DUT: Dipole 5250.0 MHz; Type: D5GHzV2 - SN1191
Communication System: UID: 0, CW; Frequency: 5250.0 MHz
Medium: 5200-5800 Body; Medium parameters used:
f = 5250.0 MHz; cond = 5.46 S/m; perm = 47.7; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/04/2022; Ambient Temp: 22.3°C; Tissue Temp: 21.5°C
Probe: EX3DV4 - SN7417; ConvF:(4.91,4.91,4.91); Calibrated: 2022-02-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn665; Calibrated: 2022-02-22
Phantom: Twin-SAM V8.0; Serial: 2060
Measurement SW: DASY Module SAR V16.0.2.83

5250 MHz System Verification at 17 dBm (50 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (22.0 x 22.0 x 22.0): Measurement grid: dx=4.0 mm, dy=4.0 mm, dz=1.4 mm; Graded Ratio: 1.4
Peak SAR (extrapolated) = 13.4 W/kg
SAR(1 g) = 3.56 W/kg
Deviation (1 g) = -3.91%

B44

ELEMENT
DUT: Dipole 5250.0 MHz; Type: D5GHzV2 - SN1057
Communication System: UID: 0, CW; Frequency: 5250.0 MHz
Medium: 5200-5800 Body; Medium parameters used:
f = 5250.0 MHz; cond = 5.27 S/m; perm = 47.2; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 06/28/2022; Ambient Temp: 21.6°C; Tissue Temp: 20.9°C
Probe: EX3DV4 - SN7417; ConvF:(4.91,4.91,4.91); Calibrated: 2022-02-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn665; Calibrated: 2022-02-22
Phantom: Twin-SAM V8.0; Serial: 2060
Measurement SW: DASY Module SAR V16.0.2.83

5250 MHz System Verification at 17 dBm (50 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (22.0 x 22.0 x 22.0): Measurement grid: dx=4.0 mm, dy=4.0 mm, dz=1.4 mm; Graded Ratio: 1.4
Peak SAR (extrapolated) = 13.6 W/kg
SAR(1 g) = 3.71 W/kg
Deviation (1 g) = 0.00%

B45

ELEMENT
DUT: Dipole 5600.0 MHz; Type: D5GHzV2 - SN1191
Communication System: UID: 0, CW; Frequency: 5600.0 MHz
Medium: 5200-5800 Body; Medium parameters used:
f = 5600.0 MHz; cond = 5.96 S/m; perm = 47.0; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/04/2022; Ambient Temp: 22.3°C; Tissue Temp: 21.5°C
Probe: EX3DV4 - SN7417; ConvF:(4.35,4.35,4.35); Calibrated: 2022-02-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn665; Calibrated: 2022-02-22
Phantom: Twin-SAM V8.0; Serial: 2060
Measurement SW: DASY Module SAR V16.0.2.83

5600 MHz System Verification at 17 dBm (50 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (22.0 x 22.0 x 22.0): Measurement grid: dx=4.0 mm, dy=4.0 mm, dz=1.4 mm; Graded Ratio: 1.4
Peak SAR (extrapolated) = 15.0 W/kg
SAR(1 g) = 3.71 W/kg
Deviation (1 g) = -3.51%

B46

ELEMENT
DUT: Dipole 5600.0 MHz; Type: D5GHzV2 - SN1057
Communication System: UID: 0, CW; Frequency: 5600.0 MHz
Medium: 5200-5800 Body; Medium parameters used:
f = 5600.0 MHz; cond = 5.78 S/m; perm = 46.6; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 06/28/2022; Ambient Temp: 21.6°C; Tissue Temp: 20.9°C
Probe: EX3DV4 - SN7417; ConvF:(4.35,4.35,4.35); Calibrated: 2022-02-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn665; Calibrated: 2022-02-22
Phantom: Twin-SAM V8.0; Serial: 2060
Measurement SW: DASY Module SAR V16.0.2.83

5600 MHz System Verification at 17 dBm (50 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (22.0 x 22.0 x 22.0): Measurement grid: dx=4.0 mm, dy=4.0 mm, dz=1.4 mm; Graded Ratio: 1.4
Peak SAR (extrapolated) = 14.9 W/kg
SAR(1 g) = 3.75 W/kg
Deviation (1 g) = -2.60%

B47

ELEMENT
DUT: Dipole 5750.0 MHz; Type: D5GHzV2 - SN1191
Communication System: UID: 0, CW; Frequency: 5750.0 MHz
Medium: 5200-5800 Body; Medium parameters used:
f = 5750.0 MHz; cond = 6.18 S/m; perm = 46.7; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 04/04/2022; Ambient Temp: 22.3°C; Tissue Temp: 21.5°C
Probe: EX3DV4 - SN7417; ConvF:(4.43,4.43,4.43); Calibrated: 2022-02-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn665; Calibrated: 2022-02-22
Phantom: Twin-SAM V8.0; Serial: 2060
Measurement SW: DASY Module SAR V16.0.2.83

5750 MHz System Verification at 17 dBm (50 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (22.0 x 22.0 x 22.0): Measurement grid: dx=4.0 mm, dy=4.0 mm, dz=1.4 mm; Graded Ratio: 1.4
Peak SAR (extrapolated) = 14.1 W/kg
SAR(1 g) = 3.36 W/kg
Deviation (1 g) = -9.68%

B48

ELEMENT
DUT: Dipole 5750.0 MHz; Type: D5GHzV2 - SN1057
Communication System: UID: 0, CW; Frequency: 5750.0 MHz
Medium: 5200-5800 Body; Medium parameters used:
f = 5750.0 MHz; cond = 6.00 S/m; perm = 46.3; density = 1000 kg/m3
Phantom Section: Flat; Space: 10 mm
Test Date: 06/28/2022; Ambient Temp: 21.6°C; Tissue Temp: 20.9°C
Probe: EX3DV4 - SN7417; ConvF:(4.43,4.43,4.43); Calibrated: 2022-02-22
Sensor-Surface: 1.4mm (VMS + 6p)
Electronics: DAE4 Sn665; Calibrated: 2022-02-22
Phantom: Twin-SAM V8.0; Serial: 2060
Measurement SW: DASY Module SAR V16.0.2.83

5750 MHz System Verification at 17 dBm (50 mW)
Area Scan (40.0 x 80.0): Measurement grid: dx=10.0 mm, dy=10.0 mm
Zoom Scan (22.0 x 22.0 x 22.0): Measurement grid: dx=4.0 mm, dy=4.0 mm, dz=1.4 mm; Graded Ratio: 1.4
Peak SAR (extrapolated) = 14.5 W/kg
SAR(1 g) = 3.56 W/kg
Deviation (1 g) = -4.94%

B49