Diode Pumped Picosecond
Passively Q-Switched Laser

MPL2210

NEWFEATURES
ዧ More than 2 mJ pulse energy at 1064 nm
ዧ Short pulse duration < 250-270 ps
ዧ 1 – 100 Hz repetition rateዧ Ultra-compact
ዧ Passively Q-switchedዧ Average power 200 mW

ዧ High peak power > 7 MW
ዧ Guaranteed > 3 Gshot lifetime

ዧ Other wavelengths (e.g. 532 nm, 355 nm,266 nm) are available

APPLICATIONSዧ Laser-induced breakdown spectroscopy(LIBS)
ዧ Time resolved fluorescence measurements
ዧ DNA analysis
ዧ Pollution monitoringዧ Remote sensingዧ Supercontinuum generation
ዧ Ignition of gas mixtures

MPL2210 series DPSS passively

Q-switched picosecond laser deliver high peakpowers > 7 MW at 100 Hz repetition rate. Shortlaser cavity is fixed on thermo-stabilized andcontrolled baseplate which gives extremely
stable output parameters performance. Small
footprint is welcome point for integration into
OEM lasers. Sub-nanosecond pulse durationof < 250-270 ps, high pulse energy more than
2 mJ, variable repetition rate from 1 Hz to 100
Hz covers many applications like pollution
monitoring, DNA analysis, supercontinuum
generation and many others.
Due to short pulse duration and high pulse
energy laser delivers high peak power which
is up to 7 MW. Optional conversion to green(532 nm) and ultraviolet (355 nm, 266 nm) isalso available.

+370 620 65424
sales@qslasers.com
www.qslasers.com
Naujoji Riovoniu str. 25B, Vilnius LT-03153, Lithuania

Stand alone with harmonics option

OEMSpecifications 1)
MODEL

Due to continuous improvements
all specifications are subject tochange. Unless stated otherwise
all specifications are measured at1064 nm.
2)
FWHM level at 1064 nm.
3)
Averaged from 60 seconds timeinterval in 5 series.
4)
Over 8-hour period after max
5 minutes of warm-up when
ambient temperature variation isless than ±2 °C.
5)Factory-set pulse repetition
rate is fixed at 10 Hz repetition
rate. Higher repetition rates areavailable, please inquire for moredetails.
6)
Full angle measured at the 1/e2
level. Lower beam divergence isavailable upon request, pleaseinquire for more details.
7)
RMS value measured from1000 shots.
8)
Beam diameter is measured
20 cm from laser output at the1/e2 level.
9)
In respect to Q-switch triggeringrising edge pulse.
10)
Laser can be powered fromappropriate 12 VDC powersource. Inquire for details.
1)

MPL2210

Pulse energy
at 1064 nm

2 mJ

at 532 nm

1 mJ

at 355 nm

0.5 mJ

at 266 nm

0.25 mJ

Typical pulse duration< 250 ps - 270 ps 2)

Pulse to pulse energy stability (RMS)
at 1064 nm< 1.0 % 3)

at 532 nm< 2.0 % 3)

at 355 nm< 3.0 % 3)

at 266 nm< 4.0 % 3)

Power drift± 3.0 % 4)

Pulse repetition rate 5)1 – 100 Hz

Beam profile

close to Gaussian< 6 mradBeam divergence 6)
Polarization

linear, horizontal at 1064 nm

Spectral linewidth

SLM< 10 µrad

Beam pointing stability 7)

1.5 mmTypical beam diameter 8)
Optical jitter~ 2 µs RMS 9)DANGERDIMENSIONS
125 × 295 × 76 mm (with harmonics)

Laser head (W×L×H)

VISIBLE AND/OR INVISIBLE LASER RADIATIONAVOID EYE OR SKIN EXPOSURE TO DIRECT
REFLECTED OR SCATTERED RADIATION

99 × 174 × 45.5 mm (OEM version)

Nd:YAG 1064 nm, 532 nm, 355 nm, 266 nm
Max. 2 mJ, pulse < 500 ps
LD 800 nm, max 60 W
CLASS IV LASER PRODUCT

257 × 271 × 153 mm

Controller unit (W×L×H)

75 × 200 × 70 mm (OEM version)

OPERATING REQUIREMENTS
Cooling requirements

TEC

Ambient temperature

20 – 25 ºC

Relative humidity

10 – 80 % (non-condensing)

Mains voltage

100 – 230 VAC, single phase, 50 – 60 Hz 10)

Power consumption< 20 W
Typical beam intensity profile(20 cm from laser output) ofMPL2210 series lasers
122

150

52,74953280114

Laser Head

244,5 (x4)

75110

150

8740138

6,4(x4)491749922,7Radiator

1251

125

3076

20 2045,587

Radiator

16214(x4)66

MPL2210 series laser head dimensions OEM version (in mm)295

MPL2210 series laser head dimensions (in mm)+370 620 65424
sales@qslasers.com
www.qslasers.com
Naujoji Riovoniu str. 25B, Vilnius LT-03153, Lithuania

www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067

FAX:048-871-0068

e-mail:voc@phototechnica.co.jp