Spect r a l Sens or s by C a r l Zei s s

M C S C C D
M u lt i C h a n n e l- S p e c t r o m e t e r〒164‑0011
東京都中野区中央4‑4‑5第一小林ビル
Tel: 03‑5328‑2858 Fax: 03‑5328‑2859

P roduct

In f o r m a t io n

Construction
The module consist of a spectrometer bodymade of patented ceramic structure with anaberration corrected concave grating, a fibercross section converter or a mechanical slit
as optical entrance and a thermoelectricallycooled CCD detector of Hamamatsu. Allcomponents are cemented to thespectrometer body. The CCD detector has
an integrated Peltier element which must be
externally controlled. The heat dissipatedfrom the Peltier element passes through acopper block to the cooling body with
fan. An operating and coolingelectronic made by tec5 is available
from Carl Zeiss. One option is to equip
the spectrometer only with thepreamplifier for the tec5 operating
electronic. A mounting of thepreamplifier is recommended becauseof the light sealing of the module atthe detector.

Entrance Slit
(177)
(75)

86 ± 0.2

Peltier cooling

83 ± 0.321M4 x 8 (3 x)

26CCD ArrayBenefits••••

Very high sensitivity
Detection of low light levels
Use for diverse measuring tasks; e.g fluorescence measurement
Robust and thermally stable

18.5 18.5(128)7(12)

SMA connector

GratingSpecifications
Optical Entrance:
CSC version:

Entrance slit:

Cross section converter:
diameter: 0.5
NA = 0.22
mounted in SMA-coupling,
dismountable
70 µm x 1400 µm (optical entrance)

Grating:

Flat-field correction
248 l/mm (center)blazed for approx. 250 nm

Spectral range:

200 nm ... 980 nm(specifications for the range
220 nm... 980 nm depending on theposition and type of CCD array areused)

Wavelength accuracy absolute:< 0.5 nm

Reproducibility:< 0,1nmTemperature - induced drift:< 0.01 nm/K

Spectral distance of pixel:∆λ Pixel ≈ 0,8 nm

Resolution( FWHM: Full width half maximum):∆λ ≤ 3 nm (UV-Version)∆λ = 3...4 nm (UV-NIR Version)

Straylight:

0.1% measured at 340 nm with Deuterium lamp (transmission of NaNO2
solution, 50g/l, 1cm)

Dimensions:

177 x 178x 75 mm3(see drawing)

Options:

MCS CCD UV-NIR 200-980
MCS CCD UV
200-600

CCD Array
Producer:Type:

Hamamatsu
S 7031-906
S 7031-1006

Number of pixels:

532 x 64 oder 1044 x 64
24 x 24 µm²
2 MHz

Dimensions of pixels:
Maximum clock – rate:

Blocking filter for the second order is directly coated on the CCD array.

Interface
Symbols

Function

1RD

Reset Drain

Remarks

PinNo.

Symbols Function

13

TG

Transfer Gate

2os

Output Transistor
Source14

P 2V

CCD Vertical Register
Clock-2

3OD

Output Transistor Drain

15P 1V

CCD Vertical Register
Clock-1

4OG

Output Gate

16Thl

Thermistor

5SGSumming Gate

17Th2

Thermistor

6-18

P-

Peltier-78
9same pulse as P211

P 2H

19
CCD Horizontal.
Register Clock-2

Systemdaten
CCD Horizontal
PIHSystemdaten
Register Clock-1

10

IG2H

Test Point(Horizontal
Input Gate-2)

conn ect. to OV

11

1GIH

Test Point(Ilorizontal
Input Gate-1)

connect to OV

12

ISH

Test Point(1-lorizontal
Input Source)connect to RDP`

Remarks
same pulse as 1'2V

Peltier'

20

SS

Substrate(GND)

21

ISV

Test Point (Vertical lnput connect to RDSource)

22

IG2V

Test Point (Vertical Input connect to OVGate-2)

23

IGIV

Test Point (Vertical Input connect to OVGate-1)

24

RG

Reset Gate

113TG

OS

214P2VRDOD

315P1VOG

416Th1SG

517Th2618P-

719P+P2H

820

SS

P1H

921

ISV
IG2V

1022

IG1H

1123

IG1V

ISH

1224

RG

IG2HSystem data
Realised with:

16-Bit-AD-conversion,tec 5 operating electronic,
integration time 10 ms,
100 KHz and 50 cycles averaging≥ 14 bit*Dynamic range:*= log 2 (FS/sdd), FS = full scale = max. signal, sdd= standard deviation(rms) of dark signal, 100 single
readouts at 10 ms. The dynamic range can be enlarged by averaging.

Noise:

2...4 count standard deviation

Carl Zeiss Jena GmbH
Spektralsensorik

D-07740 Jena
Phone:++49 3641 64 2838
Fax:++49 3641 64 2485
E-Mail:info.spektralsensorik@zeiss.deInternet:
http:// www.zeiss.de/spectral

Subject to technical alteration. 12/99

PinNo.