SPECIAL PRODUCTS
Ultrafast Joulemeter

MACH 6200 kHz Energy Meter
MONITORSKEY FEATURES
1. UP TO 200 kHz PULSE-TO-PULSE

ENERGY DETECTORS

Measure EVERY pulse, with no sampling, at high rep
rates, up to 200 kHz

200kHzMach 6 Module

Store 40 seconds of data at 100 kHz

3. TRACK MISSING PULSES AND PULSESBELOW THRESHOLD

POWER DETECTORS

Know how many pulses were missed or that didn’t
make the energy threshold with this unique pulsefeature4. SEVERAL HEADS TO CHOOSE FROMSilicon, InGaAs and Pyroelectric heads for a broad
wavelength and energy range

HIGH POWER SOLUTIONSPHOTO DETECTORSTHZ DETECTORS

2. CAPTURE AND STORE UP TO 4 MILLIONPULSES

5. ANALOG MODULE AVAILABLE

Use our fast M6 Detectors with the APM and anoscilloscope for fast analog energy measurements

6. FULL-SPEED USB 2.0 CONNECTION

Ensures high data rate transfer and fast operation

7. USER-FRIENDLY SOFTWARE WITH MANYDIAGNOSTIC FEATURESAVAILABLE MODELSs M6-6-Si
s M6-6-Si-L
s M6-6-In
s M6-6-In-L
s M6-6-PY
s M6-12.5-PY

6 mm Ø, Silicon Sensor (0.35-1.1 µm)
6 mm Ø, Silicon Sensor (0.35-1.1 µm), Low Noise Level
6 mm Ø, InGaAs Sensor (1.0-1.6 µm)
6 mm Ø, InGaAs Sensor (1.0-1.6 µm), Low Noise Level
6 mm Ø, Pyroelectric Sensor with Metallic Coating (0.35-2.5 µm)
12.5 mm Ø, Pyroelectric Sensor with Metallic Coating (0.35-2.5 µm)

Í Live Mode, Strip Chart, Histogram and Statistics
displays
Í FFT display of pulse energy data for temporal
diagnostics
Í Life Test Mode to automate laser testing

BEAM DIAGNOSTICS

SPECIAL PRODUCTS

OEM DETECTORSAll M6 heads need to be used with the Mach 6 Energy Meter

ACCESSORIESStand with Delrin Post
(Model Number: 200428)

APM Analog Power Supply(Model Number: 201848)

SEE ALSO

Additional 9V Power Supply(Model Number: 200960)

M5-UV-QED
Relative Measurements in UV

USB Cable(Model Number: 202373)

Pelican Carrying CaseTECHNICAL DRAWINGSCOMPATIBLE MODULES
MACH 6APM
LIST OF ALL ACCESSORIESAPPLICATION NOTES:PULSE-TO PULSE MEASUREMENTS
AT 130 KHZ

201923

MEASUREMENT LIMITS USINGJOULEMETERS

201932

170

186

Watch the Demo video available on our website atwww.gentec-eo.com

U LT R A F A S T J 0 U L E M E T E R

MACH 6

CONFORME

SPECIFICATIONS
M6-6-Si-L

M6-6-In

M6-6-In-L

M6-6-PY

M6-12.5-PY

MAX ENERGY

200 µJ

2 µJ

200 µJ

2 µJ

20 mJ

200 mJ

MAX AVERAGE POWER

5W

5W

5W

5W

5W

25 W

MAX REP RATE

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

EFFECTIVE APERTURE

6 mm Ø

6 mm Ø

6 mm Ø

6 mm Ø

6 mm Ø

12.5 mm Ø

MEASUREMENT CAPABILITY

With Mach 6

With Mach 6

With Mach 6

With Mach 6

With Mach 6

With Mach 60.35 - 1.1 µm

0.35 - 1.1 µm

0.9 - 1.6 µm

0.9 - 1.6 µm

0.35 - 2.5 µm

0.35 - 2.5 µm

Max Measurable Energy a

200 µJ

2 µJ

200 µJ

2 µJ

20 mJ

200 mJ

Noise Equivalent Energy

2 nJ

20 pJ

2 nJ

20 pJ

0.2 µJ

2 µJ

150 ns

150 ns

150 ns

150 ns

150 ns

150 ns

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

200 000 Hz

Max Pulse Width

100 nsec

100 nsec

100 nsec

100 nsec

100 nsec

100 nsec

Calibration Uncertainty

± 4%

± 4%

± 4%

± 4%

± 4%

± 4%Repeatability

± 1%

± 1%

± 1%

± 1%

± 1%± 1%

Max Avg Power (60 seconds)

5W

5W

5W

5W

5W

25 W

Max Probe Energy (@ 1064 nm)

200 µJ

2 µJ

200 µJ

2 µJ

20 mJ

200 mJ

Effective Aperture

6 mm Ø6 mm Ø

6 mm Ø

6 mm Ø

6 mm Ø

12.5 mm ØSensor

SiliconSiliconInGaAsInGaAsPyroelectric

Pyroelectric

Dimensions

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

38.1 Ø x 58.4D mm

76H x 111W x 76D

Weight (Head only)

150 g

150 g150 g

150 g

150 g

N/A

M6-6-Si

M6-6-Si-L

M6-6-In

M6-6-In-L

M6-6-PY

M6-12.5-PY

202115202116202117202118202119202120HIGH POWER SOLUTIONSRise Time (0-100%)
Max Repetition Rate

POWER DETECTORS

Spectral Range

ENERGY DETECTORS

M6-6-SiMONITORS

MACH 6DAMAGE THRESHOLDSPHYSICAL CHARACTERISTICS

PHOTO DETECTORSORDERING INFORMATION
Product NameTHZ DETECTORS

Product NumberModule

Mach 6: 202090
Specifications are subject to change without noticea. Maximum pulse energy reading will vary with wavelength and probe voltage responsivity (Rv). For more details, please read Application Note 121D-201932 and contact us at info@gentec-eo.com
OEM DETECTORSSPECIAL PRODUCTS
BEAM DIAGNOSTICS

Catalogue 2016_V1.0

T 418.651.80 03 | 1888 5G E N TE C | F 418.651.1174 | i nfo@gent ec -eo.c om.com

SPECIAL PRODUCTS

MACH 6MONITORS

MACH 6 JOULEMETERSPECIFICATIONS & FEATURES
MACH 6(Front View)

MACH 6(Rear View)

MACH 6Compatible Detector Heads

M6Maximum Repetition Rate

200 000 HzAnalog Output

0-3 V

External Trigger (TTL)

Optically CoupledInternal Trigger

2-20 %

Trigger Delay (User-Selectable)

38-3825 ns

Computer Input Connector

USB2.0

Product Number

202090INSTRUMENT CONTROL SCREEN
Our powerful LabView Application Software includes many unique control and diagnostic features. The
Instrument Control Screen, shown on the left, is used to set up the operation of the MACH 6, including
range, trigger, wavelength, and more. In addition, it is used to set a pulse batch size and to ARM theinstrument, which starts the data collection. It also gives you access to features like “Autoset”, “Call LiveMode”, “Run Life Test”, “Save Instrument Setup” and the like. These features can be accessed by clicking
directly on the feature or pushing the associated function key.

OEM DETECTORSTHZ DETECTORSPHOTO DETECTORSHIGH POWER SOLUTIONSPOWER DETECTORSENERGY DETECTORS

Measure every pulse at up to 200 kHz with MACH 6. Measure with 12-bit digital accuracy and capture up
to 4 million pulses in real time. Our MACH 6 Joulemeter is the only instrument in the world that performs
at this speed, and with this precision. It is designed to support our full complement of fast energy probesthat include Silicon, InGaAs and Pyroelectric Detectors. Measure from pJ to mJ and from 0.35 to 2.5 µm.
Using the M6-Si detector and the M5-UC-QED accessory, you can make relative measurements at 266 nm.

BEAM DIAGNOSTICS

SPECIAL PRODUCTS

LIVE MODE DISPLAY
The Live mode can be accessed from the Instrument Control Screen, or by simply pressing the “F4” function key on
your PC. This mode of operation is intended to act like your typical slower Digital Joulemeter, as it samples the laserpulses at a 10 Hz rate. It provides you with an energy strip chart, live energy reading, statistics and repetition rate. It isa very useful mode when setting up the Mach 6 with your laser. You can select “Auto Set”, where the instrument runs
through the ranges and trigger levels until it finds the correct range, or set them manually. When setup is complete,
you will exit this screen and return to the “Control” screen where you will select a Batch Size, Arm the instrument andstart taking pulse energy data.

U LT R A F A S T J 0 U L E M E T E R

MACH 6MONITORSSTRIP CHART

ENERGY DETECTORSThe Strip Chart Display provides a quick graphical look at the pulse data batch just collected. The data canbe displayed in full scale or in auto scale mode. You can also zoom-in on a portion of the data, like shownin the screen on the left. An ND0.3 filter has been dropped through the beam and you can see the effects
on the pulse data collected. You can fit trend and min/max lines to the data. Just below the chart, you willfind a complete set of statistics for the batch. At the top of this screen you will see tabs that will take you
to the Histogram, Statistics, and FFT (Fast Fourier Transform) displays.

POWER DETECTORSHISTOGRAMHIGH POWER SOLUTIONSInterested in viewing the statistical distribution of the pulse energy data set? The Histogram screen does
this for you and fits a “best Gaussian curve” to the data. It displays complete statistical calculations along
the bottom of the graph, along with pulse frequency. Just below the statistics, you will find instrument
controls, like range and trigger. You are also given the ability to adjust trigger delay and hold off asneeded.

PHOTO DETECTORSSTATISTICS

THZ DETECTORSThe Statistics Display offers a very complete set of useful energy readings and calculated statistics. These
include: minimum, maximum, average (mean), standard deviation, spread and variation. Some other very
handy features include: windows displaying, Average Frequency, Pulse Jitter, Pulses Below Trigger and Pulses
Below Threshold (a level set by you). In the screen on the left, you can see that there were 12 pulses below
an energy threshold of 3 µW, and 147 missing pulses (or pulses below trigger).

OEM DETECTORSLIFE TEST MODE
SPECIAL PRODUCTS

Catalogue 2016_V1.0〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067

FAX:048-871-0068,

http://www.phototechnica.co.jp
e-mail:voc@phototechnica.co.jp

BEAM DIAGNOSTICS

Would you like to run a life test on your high repetition rate pulsed laser? How about a periodic test vs. anenvironmental change like temperature? We have included a great feature to accomplish this. In the Life Test
screen, we give you the ability to select the statistics you want, a start time and date, a stop time and date,
the number of pulses and a test interval. You simply identify a file, a place to put the data, and then click on
start and walk away! When you come back, you have a data set that tracked the performance of your laser
over time, temperature, shock, vibration or anything you chose.

.com