Pressure
Sensors
CONTENTS
Application and Caution

2

SD Series / Semiconductor Pressure Sensor Chip

3

SP Series / Semiconductor Pressure Sensor

4

SM Series / Semiconductor Pressure Sensor

5

SSP Series / Semiconductor Pressure Sensor

6

SPAN Series / Pressure Sensor Non-influenced by Noise7

TFP12 Series/ Metallic Thin Film Pressure Sensor(Pressure Transmitter)

8-9

TCP10 Series
10-11/ Double Diaphragm Semiconductor Pressure Sensor(Pressure Transmitter)

MKT-TAISEI Co.,Ltd.

Pressure Sensor Application
Micro-Pressure Range

Gas pressure control, washing machine water level control, filter clogging detection.
FA equipment such as vacuum cleaner clogging detectors, electronic blood pressure meter,

Low-Pressure Range

weather forecast equipment, altimeter, water pressure detector, water depth measurement
device, vehicle engine controller, compressor pressure controller, vacuum absorber and noncontact switches.

Low-Pressure Range

Vehicle suspension, vehicle break controller, air conditioner refrigerant pressure controllers
and hydraulic controllers for building and construction equipment.

CAUTION

1) Use within the specified conditions. No warranty is made over specified conditions of use.
2) Avoid measuring corrosive pressure medium or in catalysis, unless otherwise, it would be thecause of defects.
3) Avoid measuring pressure medium in high temperature, unless otherwise, it would be the causeof defects.
4) Avoid measuring flammable pressure medium, unless otherwise, it would be the cause of
explosion or fire.
5) Avoid disassembling, taking off a part or changing it.

Semiconductor Pressure Sensor Chip / SD Series
FEATURES
Corresponds to customization from design to mass production.
Small chip size. ( 1 mm ~ 3 mm )Two model types are available. ( absolute pressure and gauge pressure )Wide pressure specification. ( 10k ~ 50M Pa or 0.1 ~ 500 kg/cm2 )Low cost, high accuracy, high reliability, high breaking pressure resistance.

MODEL

SD 20 0C A 103
Rated Pressure
Type of Pressure A: Absolute Pressure
G: Gauge Pressure
Chip Size

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT

SD200C

SD250N

5
4

6

3
4

5

1.25

2
3

1.25

0.75

Pressure

1
2

6

1

6
5
4

1
2
3

3.1

2.6

2.1

1.1

SD300C

Pressure

Pressure

1

6

2

5

3

4

1.25

SD101K

Unit : mm

Pressure

0.8

0.8
Output
5Input2

Input43

Input1

Output
1
6

Output3

2
6
Output SubstrateOutput
15
InputInput2
6Input3

4
Substrate

Input2
Output
1
6

Output3

5
4
Output Substrate5Input4
Substrate

SPECIFICATIONS
SD101K-A

Model No.
Rated Pressure[ Pa ]

Pressure Type

SD200C-A or G

SD300C-G

SD250N-A
103

303

503

104

504

101

201

501

1M 3.5M 100k 200k 500k 1M 1.5M 1M
Absolute or Gauge
Absolute

3M

5M

10M 20M 35M 50M

10k

20k

50k

103

353

102

202

502

103

Maximum Overload Pressure

153

204

354

Absolute

Gauge

Rated Pressure X 3

Temperature Range[ C ]

20 ~ 100

20 ~ 100

20 ~ 100

Excitation Current[ mA ]

1.0

1.0

1.0

Bridge Resistance[ k ]

41

51

4.51

Offset Voltage[ mV ]

025

025

025

Output Span Voltage[ mV ]

60 ~ 140

60 ~ 140

50 ~ 80

Linearity[ %FS ]

0.3

20 ~ 100
1.5

1.0
51

030

025

60 ~ 140

0.3

0.3

Offset Voltage Temperature Characteristics * 1 [ %FS/C ]

0.1

0.1

0.1

0.5

0.1

Span Voltage Temperature Characteristics * 1[ %FS/C ]

0.05

0.05

0.05

0.2

0.05

Chip Size[ mm ]

1.10.20.1

2.10.20.1

2.6 0.20.1

3.1 0.2
0.1

Chip Height[ mm ]

0.750.2

1.25

1.25 0.2

1.25 0.2

Stand Hole Diameter (Gauge Pressure)[ mm ]



0.8 0.1

0.2

0.80.1

0.5

NOTE *1 The temperature range is 0 to 60C. If there is no specification, the temperature range is 255C.
These specifications are examples. Please contact us for other specifications.

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

0.3

1.5

e-sd-p01-11

Semiconductor Pressure Sensor / SP Series
FEATURESThe SP series is an inexpensive plastic package type ( DIP type ) .
These units correspond to regular equipment,
measurement equipment, home electronic equipment,
medical equipment and others.

MODEL

SP2 0 C G 102 U
Terminal Direction
Rated Pressure
Type of Pressure G : Gauge

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENTSP20CG

12

4

8

12

0.8

4

0.8

8

6 5 4
1

15.24

R

13.5

13.5

15.24

R

1

3

0.25

3

6 5 4

U

0.25

SP20CG

Unit : mm

5

0.

C

15
x.

6.8

ma

5



5
Input2.54

Input
2

15

2.54

2.54

x.

Output
1
6

Output3maInput
2

0.



1 2 3
2.54

C

1 2 3

Output
1
6

Output
3

4
Substrate5Input4
Substrate

SPECIFICATIONSSP20CG

Model No.
Rated Pressure[ kPa ]

101

201

501

102

202

10

20

50

100

100 ~ 200

Pressure Type
Temperature range

20 ~ 100
Non-Corrosive Gases[ mA ]

1.5

1.0

Maximum Overload Pressure

Rated Pressure X 3

Output Span Voltage[ mV ]

Brige Resistance[k]Offset Voltage[ mV ]

X 1.5

60 ~ 140
51
025

020

Offset Voltage Temperature Characteristics *1 [ %FS/C ]

0.5

0.1

Span Voltage Temperature Characteristics *1 [ %FS/C]

0.2

0.05

1.5

0.3

Linearity

103

Gauge
[ C ]

Pressure Medium
Excitation Current

502

100 ~ 500 100 ~ 1000

[ %FS ]NOTE *1 The temperature range is 0 to 60C. If there is no specification, the temperature range is 255C.
These specifications are examples. Please contact us for other specifications.

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-sp-p01-12

Semiconductor Pressure Sensor / SM Series
FEATURESThe SM series are high precision and high durability metal
can package type ( TO-5 CAN equivalent package ) .

MODEL

SM20C G 102
Rated Pressure
Type of Pressure A : Absolute Pressure
G : Gauge Pressure

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT
13

7.9

Unit : mm

5.6

10

SM10C
3

13

SM20C

0.45

7.9

4
6
5
Bottom View

Input5
1 6
Output

5.6

15
10

1

2

Output
Substrate343
Input2

3

Low Pressure High Pressure
7.9
15
10

SM30C

4
6
5
Bottom View

1 6
Output5.6
12

Output
Substrate
3
4

3

2.5

4
6
5
Bottom View

Input
5



0.45

8.4

Input245

10.6

Input
5



2.5

45

0.45

8.4

3



High Pressure

10.6

Output
Substrate34
Input
2

45

10.6
8.4

1

2

High Pressure

1 6
Output

SPECIFICATIONS
SM10C

Model No.
Rated Pressure[ kPa ]

501

102

202

10

20

50

100

100 ~ 200

502

20 ~ 100
Non-Corrosive Gases[ mA ]

1.5

1.0

Maximum Overload Pressure

Rated Pressure X 3

Output Span Voltage[ mV ]

Bridge Resistance[k]Offset Voltage[ mV ]

60 ~ 140
51
025

020

Offset Voltage Temperature Characteristics *1 [ %FS/C ]

0.5

0.1

Span Voltage Temperature Characteristics *1 [ %FS/C ]

0.2

0.05

[ %FS ]
1.5
Linearity
NOTE *1 The temperature range is 0 to 60C. If there is no specification, the temperature range is 255C.
These specifications are examples. Please contact us for other specifications.

0.3

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/103

100 ~ 500 100 ~ 1000

Absolute or Gauge[ C ]

Pressure Medium
Excitation Current

SM30CA or G

201

Pressure Type
Temperature Range

SM20C

101

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-sm-p01-13

Semiconductor Pressure Sensor / SSP Series
FEATURESThe SSP11 series is a small plastic package type.
Built-in amplification and temperature compensation circuits,
eliminate the need for circuit design and characteristic adjustment.
Terminal types are DIP and SMD.
It can be used for various devices such as general-purpose devices,
measuring instruments, household devices, and medical devices.

MODEL

S S P 11  P G 1 0 2DTerminal Type D : DIP
S : SMD

Rated Pressure

Pressure Type PG : Gauge ()VG : Gauge ()CG : Gauge ()EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT

1

2

SMD Type

3

Unit : mm Tolerance ± 0.3mm
12

3
No. Signal Name
Name
Vcc
Power Input
1
2
NC
Unconnected
Output Voltage
Vout3
UnconnectedNC
4
Unconnected
NC
5
GND
GND
6

2.2

2.2

Vent Hole

Vent Hole
5

4

3.0

6

5

4

5.0

3.0

2.0

8.5

2.0

8.5

5.0

6

Recommended Circuit(9.5)

7.0 max.
2.5 2.5

2.5 2.5
7.2 max.

7.5

7.2 max.

6-0.90.5

2.5 2.5

Vcc

Vout

C1

C2

9.5

7.0 max.
2.5 2.5

0.5

10.0

0.25

0.15

(5.0)

0.25

1.9

DIP Type

1.1
0.5

Recommended hole drillingfor printed circuit boards

GND
C1=0.1μF
C2=1000pF

Recommended padfor printed circuit board

SPECIFICATIONSCG102Model No.
Terminal Type○○D (DIP)
S (SMD)

Rated Pressure
Maximum Overload Pressure
Pressure Type
Pressure Medium
Operating Temperature Range
Compensation Temperature Range
Supply Voltage
Output Voltage *1,2
Total Accuracy *1,2
Maximum Current Consumption *1[ kPa ]

100

VG102

○
○

100

[ C ][ C ][V][V][ %FS ][ mA ]

PG501

○

PG102

○
○

PG202

○
○

No Description
50
100
200
Rated Pressure  2
Gauge
Non-Corrosive Gases
20 ~ 85 (No Freezing and Condensation)
0 ~ 60 (No Freezing and Condensation)
50.25

PG502

○
○

500

PG103

○
○

1000
Rated Pressure  1.5

0.5 ~ 4.5
1.25
10

NOTE *1 Shows the value when the supply voltage is 5V. The value fluctuates due to fluctuations in voltage, but the error is not included.
*2 The total accuracy is the factory value. Please design so that it can be corrected when the offset voltage deviates.

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-ssp_series-03

Pressure Sensor Non-influenced by Noise / SPAN Series
FEATURES
Can be used in environments where high-frequency noise occurs.
● Insulation separation structure is adopted in the sensor case.
● High output and high reliability achieved through combination of
metal diaphragm and thin film gauge.
● High insulation.
● Wide operating temperature range.
● Small size.

EXTERNAL DIMENSIONS AND TERMINAL ARRANGEMENT

Signal Name

OutputPower Supply

Red

GND

Black

OutputWhite

Dimensions:A
A
58±2
68±2

M terminal
7114-1260
M housing
Dedicated Housing
F terminal
7116-1261
F housing7123-1835
Waterproof Stopper 7157-3596-30

SPECIFICATIONS

Case

Model No.

SPAN-

18.6 0.1
0
15
±1

°

2

58 connector V Type Part No.

13 or more

G3/8

HEX24

28

2.5 0.4
0GND

Specifications
Voltage Output
Free Power Supply /Current Output

Wire Color

18 or more

15.3±0.5

A

Power Supply

6 .3 zWire Color

Terminal Arrangement

O RingUnit : mm

Recommended Dimensions
for Sensor Mounting Holes

G3/8-Class B

9Structure to Isolate Insulation

Element Part-5  10  20  35  50 MPa

Rated Pressure

150%FS

Overload
Operating Temperature range±1.0%FS or less
40 ~ 120 C (at DC 5V)

Insulation Ring

20 ~ 85 °C

Compensation Temperature range
Power Supply

DC 5V  12V  24V  8 ~ 35V

Output

0.5 ~ 4.5V  1 ~ 5V  4 ~ 20mA

Zero temperature Characteristics

Within ±0.025%FS/°C

Span temperature Characteristics

Within ±0.025%FS/°C

Sensor Output Fluctuation Rate(% FS)

Total Accuracy

2 ms or less

Response Time

About 60 g

WeightIP67equivalent

IP Protection Rating

Total Accuracy = √ (Linearity) 2 + (Hysteresis) 2 + (Reproducibility) 2

MODEL NUMBER CONFIGURATION

1

SPAN
Rated
Pressure

05

5 MPa

10

10 MPa2020 MPa3535 MPa5050 MPa

2

Chalk
200

Sensor Output Fluctuation Due to Noise

150

Conventional Product
Noise Resistance Product

100
50
0
50

0.1k

1k

10k

100k

*

3
Output Format

Input Voltage
Voltage Output Specifications
Current Output Specifications

5

5V

6

12 V

0

8 ~ 35 V(Free Power Supply)

7

24 V

Ex.1) 10 MPa Input Voltage 12 V Voltage Output (0.5 to 4.5 V)
Ex.2) 35 MPa Free Power Supply 24 V Voltage Output (0.5 to 4.5 V)
Ex.3) 50 MPa I nput Voltage 24 V Current Output
Ex.4) 5 MPa Free Power Supply and Voltage Output (1 to 5 V)

h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >H

0.5 ~ 4.5 V

G

1~5V

I

4 ~ 20 mA

Voltage Output
Current Output→ SPAN-10-6H→ SPAN-35-0H→ SPAN-50-7I→ SPAN-05-0G

Specifications of products are subject to change without notice.

<

1M

Applied Noise Frequency (Hz) 10VP-P Applied

3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

e-span-p01-03

Metallic Thin Film Pressure Sensors(Pressure Transmitters) / TFP12 Series
FEATURESHigh Accuracy and High Liability by the combinationof Metal Diaphragm and Metallic Thin Film Gauge.
Medium and high pressure transmitters with stainlesssteel diaphragm.
Comply with a wide range of pressure media like Oil,
Water, Gas which dose not spoil stainless steel.
Comply with wide operating temperature range.
Built-in amplifier.
Economical, Low cost.

MODEL

TFP12
Voltage OutputElectric
Specification

Structural
Specification

EXTERNAL DIMENSIONS
Unit : mm(33)

141000  50

S/N

5

R e d : Power ()Black : Common ()White : Output
Shield Wire

Name Plate

6

HEX.2465  10

27

R1/4

24

Q 70001M50.8

S/N : Q 7 0001
Serial Number
Date of Manufacture
Pressure Range Code (Q : 20MPa)

SPECIFICATIONSTFP12

Model No.

E
0~0.5 MPa

Pressure Range

G
0~1 MPaJ
0~2 MPa

Pressure TypeL
0~5 MPa20 C~80 C

Maximum Over Load Pressure

Rated Pressure X 1.5

Power Source  Rated Output

Power Source: Vs12~24 V DC  Rated Output: 1~5 V

Load ResistancePower Source: Vs5.0 V DC  Rated Output: 0.5~4.5 V

More than 10 k

Response TimeLess than 3 msec

Total AccuracyLess than 3 %FS

Linearity*1

Hysteresis*1

Temperature Characteristic*1

Less than 0.3 %FS

Less than 0.25 %FS
Less than 0.2 %FS
Less than 0.03 %FS C
100V AC 1 min
Correspondence IP67

Connection

R1/4, other specication consult.

Structure

Shield Cable Type , other specication consult.

Cable

R
0~35 MPa

Non-corrosive gases or liquids compatible with SUS630 or SUS304

Operating Temperature Range

Protection

Q
0~20 MPa

Gauge Pressure

Pressure Medium

Withstand VoltageN0~10 MPa

1,00050 mm , Specify cable length.

WeightApprox. 100 g
Environmental
Shock Proof
490 m/s2 (X,Y,Z direction)
Performance
Vibration Proof
19.6 m/s2 (Oscillating Freguency:10~150 Hz, Total Sweep Time:75 min. , Sweep Times :10 times Each X,Y,Z direction)(255 C)
e-tfp12-p01-04_ai15*1 : Reference Value. It provides for accuracy by "Total Accuracy".

TFP12 Series
MODEL NUMBER CONFIGURATION

TFP12





Voltage Output
Built-in Amplifier
Pressure Sensors (Pressure Transmitters)

Pressure Range [MPa]

E
G

0~0.5
0~1

J

0~2
0~5

L
N
Q
Power Source
Rated Output
Accuracy
Connection
Wetted Parts Material*2
Cable Type
Additional Spec.

R
1
2
4
6
3
1
2
6
1
F
0
F

0~10
0~20
0~35
5.0 VDC (ratiometric)
12~24 VDC
0.5~4.5 V (ratiometric)
1~5 V
Total Accuracy Less than3%FS (20~80 C)
R1/4
Diaphragm:HASTELLOY C276 , Fitting:SUS316L
Diaphragm:SUS630 , Fitting:SUS304
Cable Type Length:1mSpecify your requirement.
NilSpecify your requirement.

*2 : "2" is less than pressure range 1 MPa, "6" is more than pressure range 2 MPa.

e-tfp12-p02-05

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

Double Diaphragm Semiconductor Pressure Sensors(Pressure Transmitters) / TCP10 Series

FEATURESHigh Accuracy and High Liability by the combination ofMetal Diaphragm and Semiconductor Pressure Sensor.
Low and Medium Pressure Transmitters with stainlesssteel duplex diaphragm.
High accuracy and excellent durability supported bywide pressure range available from very low pressure.
Comply with a wide range of pressure media like Oil,
Water, Gas which dose not spoil stainless steel.
Built-in amplifier, Output signal: 1~5 VDC.
Economical, Low cost.

MODEL

TCP10
Voltage OutputElectric
Specification

Structural
Specification

EXTERNAL DIMENSIONS
Unit : mm
(28)

14
6

11

S/N

1000  50
Name Plate

R e d : Power ()Black : Common ()White : Output
Shield Wire

A 80001M50.8
R1/4

30

27

65  10

HEX.27

S/N : A 8 0001
Serial Number
Date of Manufacture
Pressure Range Code (A : 100kPa)

SPECIFICATIONSTCP10

Model No.
Pressure Range



2

5

A

B

E

0~20 kPa

0~50 kPa

0~100 kPa

0~200 kPa

0~500 kPa

W

Gauge

Pressure Type

Non-corrosive gases or liquids compatible with SUS316

Pressure Medium
Operating Temperature Range10 C ~ 60 C

Maximum Over Load Pressure

Rated Pressure X 5Power Source  Rated Output

Power Souce:Vs12~24 V DC / Rated Output:1~5 V

Load ResistanceMore than 10 k

Response TimeLess than 3 msec
Less than 3 %FS

Total Accuracy
Linearity*1 Less than 1.5 %FS

Hysteresis*1

Less than 0.3 %FS
Less than 0.2 %FS
Less than 0.03 %FS/C

Temperature Characteristic *1

500V AC 1min

Withstand Voltage
Connection

R1/4, other specication consult.

Structure

Shield Cable Type, other specication consult.
1,00050 mm, Specify cable length.

Cable
WeightEnvironmental
Performance(255 C)V

100~0 kPa 100~100 kPa

Approx. 150 g
Shock Proof

490 m/s2 (X,Y,Z direction)

Vibration Proof

19.6 m/s2 (Oscillating Freguency:10~150 Hz, Total Sweep Time:75 min. , Sweep Times:10 times Each X,Y,Z direction)*1 : Reference Value. It provides for accuracy by "Total Accuracy".
e-tcp10-p01-05

TCP10 Series
MODEL NUMBER CONFIGURATIONTCP10 

Voltage Output
Built-in Amplifier
Pressure Sensors (Pressure Transmitters)

Pressure Range [kPa]

2
5
A
B
E

Power Source
Rated Output
Accuracy
Connection
Wetted Parts Material
Cable Type
Additional Spec.

V
W
2
6
3
1
3
1
F
0
F

0~20
0~50
0~100
0~200
0~500
100~0100~100
12~24 VDC
1~5 V
Total Accuracy Less than3%FS (10~60 C)
R1/4
Diaphragm:SUS316L Fitting:SUS316L
Cable Type Length:1mSpecify your requirement.
NilSpecify your requirement.

e-tcp10-p02-04

Specifications of products are subject to change without notice.

<h ttp ://www.m kt- ta is ei . c o. j p/

Head Office >< Kansai Branch >3-10-8, Nohagi 1st Bldg, Minami-Ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.
PHONE : 81-3-3982-0029 FAX : 81-3-3986-1537 E-mail : info@mkt-taisei.com
3-18-9, Nishi-Nakajima, Yodogawa-ku, Osaka-City, 532-0011, Japan.
PHONE : 81-6-6390-7264 FAX : 81-6-6390-6945

Pressure Sensors

TAISEI Co.,Ltd.
HEAD OFFICE 6972, Shimo-yoshida, Chichibu-shi, Saitama-ken, 369-1593, Japan.

TEL:81-494-77-1211 FAX:81-494-77-1415( BUSINESS )MKT-TAISEI Co.,Ltd.
HEAD OFFICE 3-10-8, Nohagi 1st Bldg, Minami-ikebukuro, Toshima-ku, Tokyo, 171-0022, Japan.

TEL:81-3-3982-0029 FAX:81-3-3986-1537
E-mail : info@mkt-taisei.com
KANSAI BRANCH 3-18-9, Nishi-nakajima, Yodogawa-ku, Osaka-shi, 532-0011, Japan.
TEL:81-6-6390-7264 FAX:81-6-6390-6945

URL : http://www.mkt-taisei.co.jp/
2024.0485