STRATO-Active Series

Catalog No. E16035(2)Coordinate Measuring Machines

High Accuracy CNC Coordinate Measuring Machine• The bridge type CNC CMM offers both high performance and cost-effective measurements.
• Achieves high accuracy of (1.2+3L/1000) μm*.
• High speed measurement at 3 mm/s.
• Suitable for measuring medium-sized workpieces accurately.
• Thermal compensation is included as standard.
Cost-effectiveSpecifications
Model

Measuring range

STRATO-Active 776

700 mm

700 mm

900 mm

900 mm

Y axis

700 mm1000 mm1000 mm1600 mm

Z axis

600 mmGuide system

HighHighuracy
performance Acc

STRATO-Active 7106 STRATO-Active 9106 STRATO-Active 9166

X axisDrive speed

Air bearing on each axis
CNC MODE(Mode selector switch:AUTO)
CNC MODE(Mode selector switch:MANUAL)
J/S MODE

Moving speed

Max. 300 mm/s (Max. combined speed 519 mm/s)

Measuring speed

1 to 3 mm/s

Moving speed

Max. 138 mm/s (Max. combined speed 239 mm/s)

Measuring speed

1 to 3 mm/s

High speed

0 to 80 mm/s

Low speed

0 to 3 mm/s

Fine speed

0 to 0.05 mm/s
Each axis 1333 mm/s2 (Max. combined acceleration 2309)

Drive acceleration
Length measuring method

Linear encoder

Resolution

0.00005 mm
Material

Measuring table

GraniteSize (stage)

876 × 1420 mm

876 × 1720 mm

Measured object fixing method
Workpiece

Maximum height

1076 × 2320 mm

800 mmMaximum mass

800 kgMain unit dimensions (H × W × D)1000 kg

1200 kg

1500 kg2730 × 1465 × 1700 mm 2730 × 1465 × 2000 mm 2730 × 1665 × 2000 mm 2730 × 1665 × 2740 mm

Mass (including the installation platform and control device)
Power source

1076 × 1720 mm

Screw (M8 × 1.25)

1900 kg

Voltage, Frequency

2150 kg

2350 kg

3050 kg100 to 120 / 200 to 240 VAC, 50/60 Hz

Maximum power consumption

1.5 kW

Accuracy*

Unit: µmTitle

E drive units for EV and FCV

E150,MPE

Semiconductor industry

Semiconductor manufacturing equipment

Medical industry

Surgical robotic system

1.2+3L/1000
ISO 10360-2: 2009
( JIS B 7440-2: 2013)

R0,MPL

Repeatability range of E0

Capable of measuring high-precision parts effectively.

Standard

E0,MPE

Length measurement error

Applications
Automotive industry

SymbolScanning mode form error on a sphere

P Form.Sph.Scan:PP:Tact,MPE

Scanning mode timeτSph.Scan:PP:Tact,MPL

Single-stylus form error

P Form.Sph.1×25:SS:Tact,MPE

1.2+3L/1000
1.22.0

ISO 10360-5: 2020
( JIS B 7440-5: 2022)

45 s
1.2

*guaranteed when using the SP25M probe system with a ø4 × 50 mm stylus [SM25-1+SH25-1+Stylus (ø4 × 50 mm)]. L: measuring length (mm).

Guaranteed accuracy temperature
Temperature range
Temperature change
Temperature gradient

218 to 22 °C
Per hour

1.0 °CPer day

2.0 °C

Vertical

1.0 °C/mHorizontal

1.0 °C/m

32309(2) -(JP0), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law.
Please consult us in advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.
All product information contained in this brochure is current as of September 2023.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp