MultiSpec

OEM Detectorarray Spectrometer Systems

MultiSpec is a modular instrument family of fastand process capable simultaneous readout
spectrometer systems for the UV/VIS/NIR, basedon flexible 19“- chassis technology. Various
spectral ranges, resolutions and PC-interfaces
are available. The integrated spectrometers are

high-quality optical modules without moving
parts and with high reliability and long term
stability. An electronic multiplexer makes multichannel sampling possible. The standardized
SMA-connectors at the front side allow theconnection of fibres and various probes.

Plug-In-cassette designMultiSpec systems follow a modular concept. Eachcomponent, such as spectrometer or lamp module,
are integrated in cassettes, which can be changedeasily. So you keep the necessary flexibility for future
measuring demands.

Operating electronics
Different interfaces guarantee easy transfer fromlaboratory to process:
Desktop

MultiSpec: The Desktop version with PCI-Bus
plug-in board is especially suitable for laboratory
use.
StandAlone

MultiSpec

••
•
•••

Remotepossible spectral range
from 200 – 1700 nm ( - 2200*)
fast, precise, robust
modern detector array technology
wide dynamic range
various Operating Electronics
versatile software packages

Spectrometer module

MultiSpec: The Remote version offers a RS232/RS-485 interface. With RS-485 an external
control by host PC from up to 6 units over a distanceof more than 1 km is possible.
StandAlone

MultiSpec: The Stand-Alone line withintegrated embedded PC offers a powerful andcompact unit. For process communication it provides
Ethernet interface, analog 4...20mA outputs anddigital I/O´s.

Electronic multiplexer

MultiSpec systems are based on the monolithic
spectrometers from Carl Zeiss. The high lightsensitivity and the extreme stability of thiscomponent allow together with the tec5 15/16-bit
electronics very accurate measurements with highdynamic. The spectrometers are available withvarious wavelength ranges and resolutions. In
addition you can control two different types ofspectrometers in parallel.

Electronic spectral sensor multiplexers provide
important advantages in case sample and reference
data have to be acquired at the same time (2channel-operation)orspectraatdifferent
measurement locations have to be taken
sequentially. The tec5 multiplexer has no moving
parts and is very fast. Variations of the light sourceor the temperature can be compensated.

株式会社 スペクトラ・コープ〒164‑0011 東京都中野区中央4‑4‑5第一小林ビル
Tel:03‑5328‑2858 Fax:03‑5328‑2859
URL http://www.spectra.co.jp

Light Sources

Accessories•

Equipped with appropriate fiberoptics and probes
the system can be adapted for your measurementtask. We assist you in finding the optimized solution.

••

Usable wavelength range:
Halogen lamp:
360nm – 2400nm
Deuterium lamp:
200nm – 600nm
Shine through lamp: 200nm – 2400nm
Xenon flash lamp:
200nm – 1100nm
Lamp and shutter control
2 internal filter ports

Software modules••••MultiSpecView and MultiSpecCalc
Modern and powerful 32 Bit software packages
TMbased on Spectacle from CreonLabControl
MultiSpecPro process software with various dataprocessing algorithms (e.g. color evaluation /TMchemometrics) based on LabVIEW
TM32 Bit function library for LabVIEW
andprogrammer interface for C++/ VB/ Delphi for thedevelopment of application specific software
Additional modules on request

Technical Data:
Spectral sensor(Specifications depend on applied module)
Spectral range:
Resolution(Rayleigh):

200 – 1700 nm(*- 2200 nm on request)
2.5 - 18 nmPixel dispersion: 0.8 - 6 nm
Wavelengthprecision:
0.1 - 0.6 nm
Number of pixels: 128 / 256 / 512 / 1024
Noise:< 0.00005 abs. rms for Si-array< 0.0002 abs. rms for InGaAs-array

Operating Electronics(Specifications depend on applied module)
Resolution:

15 / 16 Bit

Measurement time:

variable from 1.5 ms – 6 s

Optical Interface
Standard-SMA connectors
Other

Spectroscopy software MultiSpecVIEWPower supply:

110/220V, 50/60Hz

Dimensions:(Std.-enclosure)

18 x 427 x 411 [mm](3/4 HE / 84TE)

Weight:≈ 12 kg

Operating temperature: 0 °C – 45 °C

tec5 AGIn der Au 25
61440 Oberursel
Tel: 06171-9758-0
Fax:06171-9758-50
e-mail: info@tec5.com
Internet: www.tec5.com