2341

Measurement of friction torque of rotating body

Rotational Friction Torque Tester

NTS7-RT05 / NTS7-RT2
Rotation speed and rotation angle are adjustable.

Measure the rotation starting torque and friction
torque during rotation of the rotating body(which does not rotate by itself)
An example of measurement
・ Management of tightening torque for screws, etc.
・ Inspection of idling torque screwdriver・ Inspection of opening torque of caps such as pet bottles
・ Inspection of rotational torque of gears, bearings, etc.

SUGISAKI METER CO.,LTD.

2341

feature● Easy-to-understand color LCD● Prepare various commands for external control● Rotation speed can be selected from 5 to 50RPM*In increments of 5RPM
● Can be tightened with constant torque● Automatically turns on/off in conjunction with ACpower● 2 types of communication ports (USB/RS232C)

Measurement example (2 measurement modes are prepared)
Torque mode for motor control by torqueAngle mode for motor control based on
rotation angle*Usage example Screw tightening*Usage example Torque driver inspection

Linked with personal computer or PLC
Prepare various commands for external control.

Command listRotation start・・・Start rotation of the motor
Rotation stop・・・Stop the rotation of the motor
Measurement mode・・・Set the measurement mode
Measurement channel・・・Set the measurement channel
Pass/fail judgment upper limit value・・・Sets the pass/fail judgment
upper limit value for the current channel.
Pass/fail judgment lower limit value・・・Sets the pass/fail judgment
lower limit value for the current channel.
Buzzer notification・・・Set the buzzer notification method.
Rotation direction・・・Rotation direction setting
Rotation speed・・・Rotation speed setting
Stop command・・・Set conditions to stop
Data output・・・Set the type of output data
External port output・・・Set the contents to be output to the externalport
Angle command:・・・Set the angle to rotate

SUGISAKI METER CO.,LTD.

2341

Product Specifications
Model
Rated input voltage
Measurement range
unit of measureaccuracy
sampling period
Rotational speed
Direction of CW direction
rotation
CCW direction
control angle
rotation angle
communication port
external port
Attached document

NTS7-RT05
NTS7-RT224VDC
2.0 ～ 500.0 [mN・m]
0.020 ～ 2.000 [N・m]mN・m / cN・m
N・m / cN・m
±1.0% (±5 digits for 499 digits or less)1000 data / 1 second
5 to 50 [rpm] *Adjustable every 5 rpm
right-hand thread tightening direction
Right-hand thread loosening direction
About 0.968°
Can be set from 1° to 9999°
USB、RS232C
1ch input, 3ch output
Inspection report, calibration certificate, traceability system diagram*All listed on one sheet

Measurement mode
Measurement mode
Torque mode

PPangle mode

PAtruckTR

detector shapeModelContinuous
dataoutput
during motor rotation

content
Based on the measured torque, the motor stop judgment is performed.
Displays the measured torque while the motor is rotating.
Displays the maximum torque when stopped.
angle mode
Judgment to stop the motor is made based on the rotation angle.
Displays the measured torque while the motor is rotating.
Displays the maximum torque when stopped.

○○－

Mainly used for calibration.

RT05

RT2

Φ 10.3 mm
38.0 mm135.5 mm
Φ 37.0 mm

Φ 12.0 mm
55.5 mm
168.5mm
Φ 42.0 mm

External shape

A tip outer diameterB tip length
C body length
D body outer diameter
Applicable bit(Hios Shank H4)
Allowable radial load(tip of bit holder)
Allowable thrust load(B type)

20 [N]

30 [N]

40 [N]

80 [N]SUGISAKI METER CO.,LTD.

2341

Accessories
All products come with an AC adapter. Measurement bits etc. are as follows.
order number
NTS7-RT05-L2-A-O1
NTS7-RT2-L2-A-O1

Detector
RT05
RT2

AC adapter
model
Rated voltage, current
Frequency rating
Output voltage, current
certification

measuring bit
Phillips bits (#0, #1)
Phillips bits (#1, #2)

cable・Detector cable
・External port cableothersrubber footSMI36-24-1AC100V to 240V, 1.0A or less
50Hz～60Hz
24.0V, 1.5A or less
PSE、FCC、CCC、CE、UKCA

detector cablemodellengthRT-12P-2M
2m

Cable for external portmodel
length

EXT-5P-1M
1m

About order number
The order number indicates the "detector model", "detector cable length", "case", "destination", and "AC adapter plug"
after the display model.
NTS7

RT05

L2AO

1plug
1
A plug (Japan, Taiwan, USA)2
A plug (China)
3
C plug5
G plugDestinationJJapanO

Outside JapanPackingA
Plastic case

Detector cable length
L22 metersMode of the detector

RT05
RT2

model of the indicator

SUGISAKI METER CO.,LTD.
4-2-12 shirahane Ryugasaki-shi,Ibaraki,301-0901,Japan
URL https://cedar.co.jp E-mail sales@cedar.co.jp
The contents of a catalog may change specification and a design without a preliminary announcement.

SUGISAKI METER CO.,LTD.