Diode Pumped Sub-Nanosecond
Actively Q-Switched Laser

MPL15100
FEATURESAPPLICATIONSዧ More than 0.5 mJ pulse energy at 1064 nm
ዧ Short pulse duration < 700 ps

ዧ 1000 Hz repetition rate
ዧ 532 nm, 355nm, 266 nm wavelengths asstandard option
ዧ Actively Q-Switchedዧ High Peak Power 0.7 MWዧ Other wavelengths (e.g. 1053 nm, 1342 nm,671 nm, 447 nm) are available

ዧ OLED repair
ዧ Markingዧ Nonlinear Optics
ዧ Seeding laser amplifiersዧ Pollution Monitoringዧ Remote sensing

MPL15100 series robust DPSS actively

Q-switched sub-nanosecond lasers deliver
multi-kW peak powers, less than 1 ns pulse
duration at 1000 Hz repetition rate. Shortinnovative laser cavity with is fixed on thermostabilized baseplate which gives extremely
stable output parameters performance. Small
footprint is welcome point for integration into
OEM lasers. Sub-nanosecond pulse duration of< 700 ps with near transform limited spectral
linewidth at repetition rates up to 1 kHz with
low timing jitter of <200 ps and energies
more than 500 µJ covers broad spectrum of
applications starting from LIBS, laser inducedfluorescence to many others. Standard
optional harmonics generator to green(532 nm) and ultraviolet (355 nm, 266 nm) isalso available.

+370 620 42058
sales@qslasers.com
Savanoriu Ave. 231, Vilnius LT-02300, Lithuania

www.qslasers.com

Specifications 1)
MODEL

MPL15100

1)

MPL15100-1K

Pulse energy:
at 1064 nm

0.5 mJ

at 532 nm

0.25 mJ

at 355 nm

0.15 mJ

at 266 nm

0.05 mJ

Typical pulse duration

2)

3)
4)

< 700 ps

Pulse to pulse energy stability (RMS):5)

at 1064 nm< 0.5 % 2)

at 532 nm< 1.0 % 2)

at 355 nm< 1.5 % 2)

at 266 nm< 2.0 %

Typical pulse duration≤ 700 ps 3)

Power drift± 3.0 % 4)

Pulse repetition rate 5)

100 HzBeam spatial profile6)

2)
7)

8)

1000 Hz9)Close to Gaussian< 4 mradBeam divergence 6)
Polarization

Linear, horizontal at 1064 nm

Spectral linewidth

DANGER

SLM< 50 µrad

Beam pointing stability 7)
Typical beam diameter

VISIBLE AND/OR INVISIBLE LASER RADIATIONAVOID EYE OR SKIN EXPOSURE TO DIRECTREFLECTED OR SCATTERED RADIATION

1.2 mm8)

Optical jitter

Due to continuous improvements
all specifications are subject tochange. Unless stated otherwise
all specifications are measured at1064 nm.
Averaged from 60 seconds timeinterval.
FWHM level at 1064 nm.
Over 8-hour period after max
5 minutes of warm-up when
ambient temperature variation isless than ±2 °C.
Factory-set pulse repetition
rate is fixed at max repetition
rate. Higher repetition rates
are available, please inquire fordetails.
Full angle measured at the1/e² level.
RMS value measured from 1000shots.
Beam diameter is measured
20 cm from laser output at the1/e² level.
In respect to Q-switch triggeringrising edge pulse.

Nd:YAG 1064 nm, 532 nm, 355 nm, 266 nm
Max. 0.5 mJ, pulse < 700 ps
LD 800 nm, max 60 W
CLASS IV LASER PRODUCT< 0.3 ns 9)

DIMENSIONS
Laser head (W×L×H)

163 × 295 × 53.5 mm

Controller unit (W×L×H)

257 × 271 × 153 mm

Cable cord length

1m

OPERATING REQUIREMENTS
Cooling requirements

air cooled

Ambient temperature

15 – 30 ºC

Relative humidity

10 – 80 % (non-condensing)

Mains voltage

100 – 240 VAC, single phase, 50 – 60 Hz

Power consumption< 10 W< 100 W

3053.51355280295

Ø615065163

MPL15100

Typical beam intensity profile(20 cm from laser output) ofMPL15100 series lasers
25125

MPL15100 laser head dimensions with attached harmonics unit (in mm)+370 620 42058
sales@qslasers.com
Savanoriu Ave. 231, Vilnius LT-02300, Lithuania

www.qslasers.com

www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067

FAX:048-871-0068

e-mail:voc@phototechnica.co.jp