2391E

Separate type digital torque driver

WDIS-RL series
The detector is small and lightweight. For managing small torque.
Since it is a built-in battery type, it can be used for a wider range ofpurposes.

It works with a personal computer or PLC to manage torque measurement
more highly.
A color display that can display various information.
Data output is standard equipment for both USB and RS232C.
Numerical values such as pass / fail conditions can be set up to 10 channels.
The pass / fail judgment is displayed in color for easy understanding.
Powered by direct power from the AC adapter or 4 AA batteries.

SUGISAKI METER CO.,LTD.

2391E●Color display that can display various information.
●Data output is standard equipment for both USB and RS232C.

●Setting various measurement parameters

Measurement example
Saving and managing screw
tightening data
Saving screw tightening data
Data transfer to personal computer

Various measurements of screw tightening
Screw break test, yield point estimation, sampling inspection,
Measurement of return torque, measurement of starting torqueof rotating body, etc.
Tightening test

Measurement of return torqueWorks with "PC" and "PLC"
Various commands can be input from PLC or personal computer. Finer torque management is possible with external control.

Command list
Clear signal: Clears the display and outputs the held numerical value as data and saves it inmemory.
Measurement mode: Change the measurement mode.
Measurement channel: Change individual channels for which conditions such as pass / failjudgment are set.
Peak hold upper limit : Change the pass / fail judgment upper limit of the current channel.
Peak hold lower limit: Change the pass / fail judgment lower limit of the current channel.
Peak down lower limit: Change the peak down judgment start lower limit of the current
channel.
Real-time output lower limit: Changed the output lower limit of real-time output.
Auto clear time: Changed the time to automatically clear after the measurement iscompleted.
Buzzer notification: Changed the buzzer notification method.

SUGISAKI METER CO.,LTD.

2391E
WDIS -RL seriesSpecification
Model

WDIS-RL005

WDIS-RL05

WDIS-RL6

WDIS-RL10

Applications
Power supply

It is a torque measuring equipment to measure the tightening torque and loosening torque of screws.
Powered directly from the AC adapter or 4 AA batteries

Rate input voltage

12V DC

Measurement
range

0.20 ~ 50.00[mN-m] (Note 1)

2.0 ~ 500.0[mN-m]

0.020 ~ 6.000[N-m]

Measurement
Unit

gf-cm / ozf-in / mN-m /cN-m

kgf-cm / lbf-in / mN-m /cN-m0.20 ~ 10.00[N-m]

kgf-cm / lbf-in / N-m / cN-m

±0.5% (If 499 digit or less, ±3 digit.)±2.0%±0.3mN-m
Accuracy*Maintain a constant temperature during measurement.
Sampling rate

1000 data / 1sec

Measurement modeMeasurement mode

Data output

Contents

Peak holdPeak downPPPD○○

Measure the peak torque.
Measure the firs peak torque.

Real time output

C○The data output cycle is approximately 180 data / 1
second.

TrackTR－
Mainly used for calibration.
If the measurement mode is TR, the data isn’t outputted.
Data output
Memory size

Wired (ASCII format)
800 data
Detector (Refer below)
Detector cable
Plus bit #1, #2

Plus bit#0, #1

Bit holder
One way bit holder

Accessories, Attachments

Auxiliary handle×2－AC adaptorRubber footNote 1
Detector
Model

Result of calibration, Certification on calibration, Traceability system figure
WDISR-RL005 measurements are in 0.05 mN-m increments.

RL005 / 05

RL6 / 10

Φ4 (HIOS shank)

6.35sq (Square drive)

Shapes(Unit: mm)

SocketSUGISAKI METER CO.,LTD.

2391E

Ordering number
The order number indicates the model of the indicator followed by the ‘model of the detector’, ‘detector cable length’, ‘Packing’,‘destination’ and ‘AC adapter plug’.
WDISRL05

L1CO

1PlugA plug (Japan, Taiwan, USA)A plug (China)
C plugF plugG plug123
45Destination
JJapan O

Outside Japan

Packing
C
CardboardA
Plastic case
Detector cable length
L11 meter
L22 meters
L44 meters
Mode of the detector
RL005
RL05
RL6
RL10

model of the indicator

About the AC adapter
The certification mark differs depending on the specified plug.
Plug
A plug (Japan, Taiwan, USA)A plug (China)
C plugF plugG plugcertification
PSE、FCC、CCC、BSMI、CE
PSE、CE、UKCA、KC

Packing
The specifications of the case are shown below.
Packing-C-A

Specifications

Cardboard

PP

Exterior /InteriorSUGISAKI METER CO.,LTD.
4-2-12 shirahane Ryugasaki-shi,Ibaraki,301-0901,Japan
URL https://cedar.co.jp E-mail sales@cedar.co.jp
The contents of a catalog may change specification and a design without a preliminary announcement.

SUGISAKI METER CO.,LTD.