2351E

For managing various manual torque drivers and wrenches

WDISR-IP series
Manual torque tool management / built-in battery type
The "DIS series" has become easier to use and has higher performance.

It works with a personal computer or PLC to manage torque measurement
more highly. For managing manual torque drivers and wrenches.
Color display that can display various information.
Data output is standard equipment for both USB and RS232C.
A series of diverse lineups from up to 500mN・m (0.5N・m) to 1500N・m.
Numerical values such as pass / fail conditions can be set up to 10 channels.
Pass / fail judgment is clearly displayed in color.
Built-in battery drive.

SUGISAKI METER CO.,LTD.

2351E●Color display that can display various information.
●Data output is standard equipment for both USB and RS232C.

●Setting various measurement parameters

Measurement example
Manual torque drivermanagement

Manual torque wrench management
Click operation of torque wrench
Measurement is PD (peak downmode)
For tool check at the start of work

The actual torque applied is measured
PP (peak mode)
For worker experience training and tightening
operation check

Works with "PC" and "PLC"
Various commands can be input from PLC or personal computer. Finer torque management is possible with external control.

Command list
Clear signal: Clears the display and outputs the held numerical value as data and saves it inmemory.
Measurement mode: Change the measurement mode.
Measurement channel: Change individual channels for which conditions such as pass / failjudgment are set.
Peak hold upper limit : Change the pass / fail judgment upper limit of the current channel.
Peak hold lower limit: Change the pass / fail judgment lower limit of the current channel.
Peak down lower limit: Change the peak down judgment start lower limit of the current
channel.
Real-time output lower limit: Changed the output lower limit of real-time output.
Auto clear time: Changed the time to automatically clear after the measurement iscompleted.
Buzzer notification: Changed the buzzer notification method.

SUGISAKI METER CO.,LTD.

2351E

Specification
Model

WDISR-IP05

Applications

It is a torque measuring equipment to measure the tightening torque of tightening tools (torque screwdriver, torquewrench, etc.) in screw tightening work.

Power supplyInternal battery (12 hours continuous operation time, 3 hours charging time)

Rate input voltage

12V DC

Measurementrange

2.0 ~ 500.0[mN-m]

Measurement
Unit

kgf-cmlbf-inmN-mcN-m

WDISR-IP5

WDISR-IP50

0.020 ~ 5.000[N-m]0.20 ~ 50.00[N-m]

2.0 ~ 200.0[N-m]

kgf-cmlbf-inN-m
cN-m±0.5% (If 499 digit or less, ±3 digit.)*Maintain a constant temperature during measurement.

Sampling rate

1000 data / 1sec

Measurement mode

Data output

2.0 ~ 500.0[N-m]
kgf-cmlbf-inN-m

AccuracyMeasurement mode

WDISR-IP200 WDISR-IP500 WDISR-IP1500

20 ~ 1500
[N-m]
kgf-m
lbf-ftN-mContents

Peak holdPP○Measure the peak torque.

Peak downPD○Measure the firs peak torque.

Real time outputC○

The data output cycle is approximately 180 data / 1
second.

TrackTR－

Mainly used for calibration.

If the measurement mode is TR, the data isn’t outputted.
Data output

Wired (ASCII format)

Memory size

800 data

Auto power off

Power off after 10 minutes of non-use
Detector (Refer below)

Accessories,AttachmentsDetector cableAC adaptor
Rubber foot－Result of calibration, Certification on calibration, Traceability system figure

T-knob×2

Detector
Model

IP05 / 5

IP50

IP200

IP500

IP1500ShapesA

Detector shape

Φ40mm

Φ50mm

Φ60mm

Φ80mm

Φ100mmB

Height41.2mm

63mm

79mm

98mm

102.5mm

CThickness

5mm

10mm

15mm

15.5mm33mm

DFlange size

Φ60mm

Φ90mm

Φ108mm

Φ140mmΦ170mmEHole position

Φ50mm

Φ74mm

Φ90mm

Φ116mm

Φ140mm

Φ6.3mm(Every 120°)

Φ8.3mm(Every 120°)

Φ10.5mm(Every 120°)Φ13mm(Every 60°)

FHole size

Φ4.2mm(Every 120°)

GSocket6.35sq9.5sq

12.7sq19sq

25.4sqWeightAround 200g

Around 600g

Around 1kg

Around 2.5kg

Around 7.2kg

SUGISAKI METER CO.,LTD.

2351E

Ordering number
The order number indicates the model of the indicator followed by the ‘model of the detector’, ‘detector cable length’, ‘Packing’,‘destination’ and ‘AC adapter plug’.
WDISR

IP05

L1CO

1Plug
A plug (Japan, Taiwan, USA)A plug (China)
C plugF plugG plug123
4
5

Destination
JJapan O

Outside Japan

PackingC
CardboardA
Plastic case

※1Detector cable length
L1
1 meterL22 meters
L44 meters
Mode of the detector
IP05
IP5IP50
IP200
IP500
IP1500model of the indicator※1 If you select the detection unit IP200, IP500, or IP1500, the attache case (-A) cannot be used.

About the AC adapter
The certification mark differs depending on the specified plug.
Plug
A plug (Japan, Taiwan, USA)A plug (China)
C plugF plugG plugcertification
PSE、FCC、CCC、BSMI、CE

PSE、CE、UKCA、KC

Packing
The specifications of the case are shown below.
Packing-C-A

Specifications

Cardboard

PP

Exterior /Interior

SUGISAKI METER CO.,LTD.
4-2-12 shirahane Ryugasaki-shi,Ibaraki,301-0901,Japan
URL https://cedar.co.jp E-mail sales@cedar.co.jp
The contents of a catalog may change specification and a design without a preliminary announcement.
SUGISAKI METER CO.,LTD.