AvaSpec-Mini4096CL
Small and powerful OEM spectrometer
This first-to-the-market, 4096 pixel CMOS
array miniature spectrometer is theperfect combination of small size and highresolution (up to 0.09 nm)!

to-unit reproducibility and temperaturestability. These are key para-meters for OEM
customers for reliable integration into theirproducts.

It’s only the size of a deck of cards, yetdelivers a dynamic range better than 3000:1,
stray-light levels lower than 0.2% and weighs
only 175 grams. Easy to take anywhere youlike.

Many areas of research can becovered with this device, such as lightanalysis, chemical research and Raman
spectroscopy. The possibilities are endless.

The AvaSpec-Mini4096CL is producedwith the latest automated production
technology, providing excellent unit-

Of course, the AvaSpec-Mini works
seamlessly with Avantes spectroscopy
software and the Windows and Linuxlibraries.

Technical Data
Optical bench
Wavelength range

Symmetrical Czerny-Turner, 75 mm focal length, MK II
200 - 1100 nm

Stray light

0.2 - 1%

Sensitivity

261.000

Detector

HAM S13496 , CMOS linear array,4096 pixels (7x200μm)

Signal/noise

300:1

Dynamic range

3300

Dark noise
AD converter
Integration time
Interface

16 cnts
16-bit, 6 MHz
30 μs - 50 s
USB 2.0 (480 Mbps) / pigtailed (40 cm) USB-A

Sample speed with onboard averaging

6.5 ms/scan

Data transfer speed

8.9 ms/scan

I/O
Dimensions, weight
Power supply
Temperature range

5 bidirectional programmable I/O; 1 analog out; 1 analog in,
1x5V
95 x 68 x 20 mm, 175 grams
Default USB power, 500 mA
0-550C

info@avantes.com | www.avantes.com | 29

Spectrometers

Grating selection table for AvaSpec-MiniUse

Useable range(nm)Spectral range (nm)Lines/mm

Blaze (nm)Order code

4096CLUV

200-400

170

1800

250

MN 1800-0.25

VIS

330-900

535

600

500

MN 600-0.50

NIR

550-1100

525

600

1000

MN 600-1.00

UV/VIS/NIR

200-1100

900

300

300

MN 300-0.30

VIS/NIR360-1100

720

300

500

MN 300-0.50* Other gratings are available on request.

Resolution table (FWHM in nm) for AvaSpec-Mini*
Slit size (µm)
Grating (lines/mm)

102550

100

200

500

4096CL300

0.50-0.70

1.20-1.30

2.17

4.60

9.00

20.00

600

0.30-0.36

0.58-0.681.17

2.20

4.50

10.00

1800

0.09-0.11

0.18

0.36-0.40

0.78

1.50

3.70

* Typical values. Small deviations are possible.

Ordering Information
AvaSpec-Mini4096CL• Mini Fiber-optic Spectrometer, 75 mm focal length, 4096 pixel CMOS detector,
USB 2 powered interface, including DCL
Specify grating, wavelength range and options. Other gratings are available on request.

Options
SLIT-XXOSC• Slit size, please specify XX = 10, 25, 50, 100, 200 or 500 µm (5 µm possible on request)• Order sorting coating for grating MN 600-0.50, recommended with OSF-305

OSC-UA• Order sorting coating for grating MN 300-0.30

OSC-VA• Order sorting coating for grating MN 300-0.50, recommended with OSF-305

OSF-YYY• Order-sorting filter for reduction of 2nd order effects,
please specify YYY= 305, 395, 457, 515, 550 or 600 nm, depends on range
For non-OEM users a set of preconfigured models are available

www.phototechnica.co.jp〒336-0017 埼玉県さいたま市南区南浦和 1-2-17

TEL:048-871-0067

FAX:048-871-0068

e-mail:voc@phototechnica.co.jp

30 | info@avantes.com | www.avantes.com