Handy SpecThe Portable Tool for Spectral Analyses

Rugged Construction

HandySpec VIS is a high-quality handheld device for spectral analysis and performs exact colorimetric measurements
in liquids. It allows the direct analysis in
the production area without the time consuming procedure of taking samples followed by measurements in the lab.

The HandySpec is designed especially for measurement tasks in the field and is outstanding concerningcompactness and robustness. It delivers an absolute
wavelength read-out accuracy of 0.3 nm with a very
high reproducebility over a wide dynamic range.
The current version is designed for performing
absorption measurements in liquids in the visible andnear NIR range.
Coca Cola
OPL: 2mmHandyspec

CADAS

0,70,60,5E0,40,30,20,10,0
380

430

480

530

580

630

680

730

780-0,1ll [nm]Absorbance spectra of Coca-Cola, measured withHandySpecVis and laboratory spectrometer.

HandySpec

••
•
•
•
•
•

Portable
Compact
Rugged construction
Wide dynamic range
Immediate data evaluation
Memory for 50 spectra
Standard RS-232 interface

Easy HandlingAn integrated Hellma immersion probe is directly
immersed into the media to be analysed, theperformance of the measurement is activated by one
single press of a button. After a second theLCD display shows the colorimetric L*a*b values
graphical
well as the complete transmission spectrum.
asReliable Components
The HandySpec is based on the very reliable
spectrometer components from Carl Zeiss like theMMS 1 Spectral Sensor module (with range from 320
to 1100 nm) and the immersion probe from Hellma.
The light troughput and the extreme stability of thesecomponents together with smart electronics allow
precise measurements within parts of a second.
Optimised for low power consumption the HandySpec can perform up to 10,000 measurements intotal without recharging. Up to 50 spectra can be
stored on the internal memory.
The complete spectral data as well as thecolorimetric results can be forwarded to a computer
via serial interface.

株式会社 スペクトラ・コープ〒164‑0011 東京都中野区中央4‑4‑5第一小林ビル
Tel:03‑5328‑2858 Fax:03‑5328‑2859
URL http://www.spectra.co.jp

Technical data: HandySpec VIS
Immersion Probe

Spectral Sensor
Module manufacturer:
Spectral range:
Resolution(according Rayleigh):Pixel dispersion:
Number of pixels:
Wavelengh accuracy:

Carl Zeiss
360* to 1100 nm
10 nm
3.3 nm256
0.3 nmManufacturer:
Material:
Fibers:Light path:

HellmaQuartz
Suprasil, 600 µm dia
10 mm (standard),
2 to 20 mm available

Light Source* limited by light source

Type: Halogen
Power: 2 W

Electronics

Other measurement probes, light sources, Spectral
Sensors on request.

Microprocessor:
Dynamic range:
Integration time:
Measurement time:
Number ofmeasurements:
Display:
Memory:Siemens SAB80C537, 8 Bit
14 Bit14ms (min.)approx. 300ms

Housing

10,000
LCD graphical display
50 mm x 27 mm
50 spectra* without probe

Dimensions:
Weight: 1 kg

275* mm x 70 mm x 60 mm

Software
Interfaces
Serial interface in accordance with RS-232 for data
export, bidirectional (Sub-D connector)
Connector for charging the batteries
Charging voltage: 12 VDC (wall-outlet power supply)
Operating Elements
1 button to perform the measurement,
4 buttons for additional functions e.g selecting menu,
referencing, numerical indication of data etc.

In der Au 25
D- 61440 Oberursel
Phone: +49 6171/9758-0
Fax:+49 6171/9758-50
e-mail: Info@tec5.comInternet: www.tec5.com

Integrated software for measurement control,
data reduction, dark current subtraction,
evaluation of transmission and absorption,
calculation of colorimetric data,
data storage and data export

Customer Specific Measurement
Systems
For applications in process or production control, tec5offers the high-end MultiSpec OEM Spectrometer
System. In addition, we design solutions matched tocustomers’ demands.