DIGITAL PANEL METERDISPLAY FOR 15mm MOSAIC - LPDSA SERIESAPPLICATIONThis product is digital display for system which is most suitable fordisplay of various data of main instrument in mouting of graphic panel
and for centralized monitoring control and mosaic panel etc. along withthe development of automatic measuring recently. Easy-to-see highbrightness 7-segment LED, various types 1-digit to 6-digit and BCD
input (data bus system) is adopted. Also installation of masoic panel ispossible. Please use this product for display of electrical quantities andLPD04SA-15B-R

physical quantities according to purpose of usage.

(29.9×59.9×172mm)

FEATURES► Installation of one touch 15 mm mosaic panels is possible.
► Installation iron panel also possible.
► High quality and high reliability noise resistance design.
► Wiring simplification can be realized by adopting (wiring MAX50m)► High brightness LED element is adopted.
► 2 types of LED: Large (character height: 15mm),Small (character height: 10mm)► Changeover to static lighting system is possible.

1

DISPLAY FOR 15mm MOSAIC - LPDSA SERIESTYPE DESIGNATION(1)(2) (3)(4)

(5)

(6)

LPD 0 4 SA – 15 B – R – 61) Name of the Series4) Size of Number

SymbolSeries Name

SymbolSize of Number (h)LPD

Digital Indicator

1010.16mm

1514.22mm2) Polarity

5) Display color

SymbolDisplay of Polarity

SymbolDisplay color

0None

RRed

1Polarity display

3) Number of Digit
SymbolNumber of digit

11-digit

22-digit

33-digit

44-digit

55-digit

66-digit

6) Size of Box(Digit No: Treated as 1-digit, when with polarity.)
SymbolNone
6Size of Box
Standard: 4-digit use (1~4-digit)
6-digit use (5-digit, 6-digit)
4-digit or less is used for 6-digit BOX.

Item To Specify When Purchase
1. Type name
2. Aux. power
3. Option
4. Quantity

2

DISPLAY FOR 15mm MOSAIC - LPDSA SERIESCOMMON STANDARD SPECIFICATIONSITEMSPECIFICATION

Operation system

Data bus system

Display

LED (LED diode: Red)

Display diving signal

Open collector signal (DC24V, 1mA / signal line)

Auxiliary power

DC24V, ±10% (refer to specification list for power consumption)

Length of signal cable

50m or less: Flat cable, twist pair line etc.

Insulation resistance

DC500V, 50MΩ or more: Between electric circuit and outer case.

Withstand voltage
Operating temperature &humidity range
Storage temperature range

AC1500V, 50/60Hz 1 min: Between electric circuit and outer case.

External color

Munsell N1.5 (Black)

FilterWith axel antireflection treatment (Red: N-030)

0℃~+55℃, 30~85%RH-20℃~+70℃

OPTION
Specify below item according as necessary* One side connector with cable2 types: Flat cable (for insulation displacement contact use), Twist cable (for pressure bonding use)
Please specify the length as necessary by unit 0.5m.
* Meter with name, unit sculpture can be manufactured

PRODUCT SELL SEPARATELY
Specify below item according as necessary:* LPDSA - Install spring (2pcs for 1-unit meter use)
KCC&Co. and SEIWA Electirc Co is use 15mm Mosaic Panel, Installation of one touch is possible.
* LPDSA - Spacer (2pcs for 1-unit meter use)
KCC&Co. is used when installation of 15mm mosaic.
* LPDSA - 4 Install metal (Box for 4-digit: 1pcs for 1-unit meter use)
* LPDSA - 6 Install metal (Box for 6-digit: 1pcs for 1-unit meter use)
This product is necessary when shock strength exceeds 294m/s2 or mosaic tile install in half-tile slide at 15mmmosaic panel installation.

3

DISPLAY FOR 15mm MOSAIC - LPDSA SERIESTERMINOLOGY AND FUNCTIONS* BCD Input
Input signal to configure decimal 0~9 of indicator is with binary 4 bits as unit.
* Strobe Signal (for BCD input use)
Signal to read and store date in internal indicator.
Please make sure the short circuit from GND terminal when used in static lighting system.
* Latch Signal
Signal to read and store data in internal indicator.
* Lamp Test
All digit “8” and “－” (priority for all lamp test) are displayed by short circuit between lamp test terminal of indicatorand DC24V power GND.
* Input Signal Logic
Logic
“0”

“1”

Negative logic

HL

Signal levelDC10~24V

DC0~3V

Input Signal Driver

BCD INPUT* Data Bus System* Changeover to Static Lighting Display is possible

4

DISPLAY FOR 15mm MOSAIC - LPDSA SERIESTYPE AND OUTLINE DRAWING FOR BCD INPUTTypeNo Polarity
Digit No.

With PolarityNo. size: 14.22mm

No. size: 10.16mm

Digit No.

No. size: 14.22mm

No. size: 10.16mm

OutlineDrawing

1

LPD01SA-15B-R

LPD01SA-10B-R

－－－

A-1

1

LPD01SA-15B-R-6

LPD01SA-10B-R-6－－－

A-22

LPD02SA-15B-R

LPD02SA-10B-R±1

LPD11SA-15B-R

LPD11SA-10B-R

A-1

2

LPD02SA-15B-R-6

LPD02SA-10B-R-6±1

LPD11SA-15B-R-6

LPD11SA-10B-R-6A-2

3

LPD03SA-15BR

LPD03SA-10B-R±2

LPD12SA-15B-R

LPD12SA-10B-R

A-1

3

LPD03SA-15BR-6

LPD03SA-10B-R-6±2

LPD12SA-15B-R-6

LPD12SA-10B-R-6A-2

4

LPD04SA-15B-R

LPD04SA-10B-R±3

LPD13SA-15B-R

LPD13SA-10B-R

A-1

4

LPD04SA-15B-R-6

LPD04SA-10B-R-6±3

LPD13SA-15B-R-6

LPD13SA-10B-R-6A-2

5

LPD05SA-15B-R

LPD05SA-10B-R±4

LPD14SA-15B-R

LPD14SA-10B-R

A-2

6

LPD06SA-15B-R

LPD06SA-10B-R±5

LPD15SA-15B-R

LPD15SA-10B-R

A-2

POWER CONSUMPTION
Rise and fall at max. power supply voltage (W)No. size
Red LED12, ±1-digit

3, ±2-digit

4, ±3-digit

5, ±4-digit6, ±5-digit

14.22mm

1.0

1.5

2.0

2.5

3.5

4.0

10.16mm

1.0

1.3

1.7

2.0

2.5

3.0

SPECIFICATION FOR BCD INPUT USE
Signal line
1~4-digit

5-digit, 6-digit

11BCD: 1~4-digit, Total:16

BCD: 1~6-digit, Total: 24

D/P: 1~4-digit, Total: 4

D/P: 1~6-digit, Total: 6Error signal

11GND

33

Strobe signal
Data signal

5DISPLAY FOR 15mm MOSAIC - LPDSA SERIESDISPLAY* Error Display
Error (E) is displayed in lower digit by error date. Blank display in all upper digit.

* Blank Display
Blank is displayed when A~F (hexadecimal) is input to BCD data input line (display is OFF light)* Polarity (+ polarity will not displayed)+: when input “1” to BCD data, positive polarity will not display only blank display.
(When input 4, 7, 9, also is blank display)－: Display when “2” is input to BCD data.
(when 0, 3, 5, 6, 8 is input, “－” is displayed also)

DATA TRANSMISSION TIMING CHART* Decimal point allocation (when D.P input signal is not used)Setting switch for decimal point allocation is behind the BOX side.

* When adjusted to the digit number where decimal point is necessary, decimal point is ON.

6DISPLAY FOR 15mm MOSAIC - LPDSA SERIESRELATIONSHIP BETWEEN INPUT SIGNAL AND NUMERAL DISPLAY

80000000011111111※※※

BCD Input Code4200000101101011110000010110101111※
※
※
※
※
※

Truth TableD/P10101010101010101※※※

00000000000000001※※Input
LT0000000000000000010ADD.
11111111111111111※
0Display
0~9 Mark
0123456789
Light: OFF
Light: OFF
Light: OFF
Light: OFF
Light: OFF
Light: OFF
8Latch* Changeover strobe terminal from latch system to static system can be used with GND terminal by short circuit.

BLOCK DIAGRAM

7DISPLAY FOR 15mm MOSAIC - LPDSA SERIESBUS COMPOSITION AT LATCH SYSTEMCONNECTOR AND TERMINAL CONFIGURATION* 1~4-digit* Manufacturer: Hirose* Type: HIF3B-26PA-2.54DS* Compatible sokect: HIF3BA-26D-2.54R (for ribbon cable, flat cable crimping use)
HIF3BA-26D-2.54C (for solderless use)
No. Terminal
1

Description
GND

No. Terminal
2

Description
1×1003579111315
17192123
25

2×100
8×100
2×101
8×101
2×102
8×102
2×103
8×103
D.P. (101)
D.P.(103)
GND
NC

468
101214161820222426

4×100
1×101
4×101
1×102
4×102
1×103
4×103
D.P.(100)
D.P.(102)ERROR
GNDNC
Terminal assembly: 4-terminal (M3 screw)

No. terminal:1(+): +24V power
2 (-): GND
3: Lamp Test

4: Strobe

8DISPLAY FOR 15mm MOSAIC - LPDSA SERIESCONNECTOR AND TERMINAL CONFIGURATION* 5-digit, 6-digit* Manufacturer: Hirose* Type: HIF3B-34D-2.54DR* Compatible sokect: HIF3BA-34D-2.54R (for ribbon cable, flat cable crimping use)
HIF3BA-34D-2.54C (for solderless use)
No. Terminal

Description

No. Terminal

Description

1GND

2

1×100

32×10044×10058×1006

1×101

72×10184×101

98×101

101×102112×102124×102138×102

141×103152×103

164×103178×10318

1×104

192×10420

4×104

218×104

221×105

232×105

24

4×105

25

8×105

26

D.P.(100)

27

D.P.(101)

28

D.P.(102)

29

D.P.(103)30

D.P.(104)

31

D.P.(105)

32

ERROR

33

GND

34

GND

Terminal assembly: 4-terminal (M3 screw)

No. terminal:1(+): +24V power
2 (-): GND
3: Lamp Test

4: Strobe

9DISPLAY FOR 15mm MOSAIC - LPDSA SERIESINSTALLATION* Tighten Screw Type (Panel)* Installation One Touch Type (Mosaic Panel)* A-1 Type

* A-2 Type10