Coordinate Measuring Machines

5-axis CNC CMM

CRYSTA-Apex EX 1200R Series

Catalog No. E16015(4)• CRYSTA-Apex EX 1200R series products areadvanced CNC CMMs equipped with the REVO-25-axis scanning probe head.

Helical scan

Gasket scan

• REVO-2 can be positioned at any angle, making it easy
to measure even complex form workpieces.
In addition, the synchronized 5-axis movement
considerably reduces measurement times.

• The RSP2 for 5-axis scanning, the RSP3 that allows
use of a cranked stylus, and the SFP2 for surface
roughness measurement are all probes that can beused to record various shape measurements.
Automatic changeover of these probes is possible
with an auto probe changer, enabling fully automatic
measurement of parts with diverse shapes.

• REVO has laser sensing technology that ensures highaccuracy measurement. This accuracy is maintained
when using various lengths of styli (up to 500 mm*).

• After initial calibration, the infinite positioning of theRSP-2, means that any probe angle is available formeasurement.

* Distance from probe rotation center to stylus tip.

SPECIFICATIONSCRYSTA-Apex EX 121210R

Model No.
X axisMeasuring range Y axis
Z axis
CNC MODE(Key selectorswitch: AUTO)
Drive speed

1200 mmJ/S MODE
Drive acceleration
Resolution
Guide method

CRYSTA-Apex EX 122010R
1200 mm2000 mm
960 mmDrive speed each axis: Max. 300 mm/s

Sweep scan

Measuring Speed 1 - 5 mm/s
Max. 80 mm/s (J/S Mode: High Speed)
Max. 3 mm/s (J/S Mode: Low Speed)
Max. 3 mm/s (J/S Mode: Touch Speed)
375 mm/s²
0.0001 mm
Air bearings on each axis
1200 mm2500 kg

CRYSTA-Apex EX 121210R/122010R/123010R Series Accuracy (Unit: μm)
Temperature Maximum permissible error of length measurement
ISO 10360-2:2009 (JIS B 7440-2:2013)environment
1E 0, MPE = 2.9+4L/1000 E 250, MPE = 2.9+4L/1000
REVO+RSP2+RSH250
2E 0, MPE = 2.9+5L/1000 E 250, MPE = 2.9+5L/1000
1
E 0, MPE = 2.5+3L/1000 E 250, MPE = 2.5+3L/1000
REVO+RSP3-3+ RSH32
E 0, MPE = 2.5+4L/1000 E 250, MPE = 2.5+4L/1000* L=Measuring length (unit: mm)* Table below defines temperature environments 1 and 2
Probe used

Airfoil section scan

CRYSTA-Apex EX 121210R/122010R/123010R Series Installation Temperature
Temperature environment 1 Temperature environment 218 - 22 °C
16 - 26 °C
Guaranteed accuracy Temperature Rangetemperature
Rate of change 2 °C or less per day
5 °C or less per day
environment
Gradient
1 °C or less per meter 1 °C or less per meter

Specification of REVO-2 Scanning Probe
CRYSTA-Apex EX 123010R

CRYSTA-Apex EX 1200R Series DIMENSIONSREVO-2 Probes
Automatic changeover of these probes is possible with an auto probechanger.

• RSP2Optimal for 5-axis scanning and touch-trigger measurement
Stylus holder supports 175 mm, 250 mm, 350 mm, 450 mm,
and 500 mm effective stylus length• RSP3-3

SP25 scanning probe, allowing the use of a cranked stylus• SFP2

Surface roughness probe

Rotation angle Vertical (A-axis) -5° to +120° (0.08 sec)
(Pitch angle) Horizontal (B-axis) ∞ (0.08 sec)
Maximum stylus length

500 mm (Distance from probe rotation center to stylus tip)

(Unit: mm)

2200

MAFIS Express [Blade measurement/Evaluation program]
SFP2

SFH-1

RSP2

RSP3-1 RSP3-2 RSP3-3 RSP3-4

160

1200XZ100

RSP3-6SFH-23645

This software program enables creation of measurement programs and measurement
and analysis of blades and blisks. A part program for measurement can be
automatically created just by selecting required contents and evaluation conditions.
The measurement results are displayed in a report including 2D graphics.

570

3000 mm

Maximum height
Maximum mass
2000 kg
3000 kg
Mass (including the control device
4050 kg
6150 kg
9110 kgand installation platform)
Pressure
CMM: 0.4 MPa REVO: 0.5 MPa
Air supply
Consumption
150 L/min under normal conditions (air source: 230 L/min or more), 0.6 MPa or more
Note: While the appearance of the natural stone measuring table varies according to the source, the high stability for which this material is known can always be relied upon.
Table loading

760

CRYSTA-Apex EX 123010R

RSH175

RSH3-1
RSH3-2
RSH3-3240 278

RSH250Y

RSH3-4

1345
2140

600

74712

RSH350

200

F.L.
C

B
A

Dashed line: Measuring range
Model No.
CRYSTA-Apex EX 121210R
CRYSTA-Apex EX 122010R
CRYSTA-Apex EX 123010R

2X
1200
1200
1200

Y
1200
2000
3000

Z
960
960
960

A2595
3395
4395B1700
1890
2500

C470
775
970

SFM-A1 SFM-A2 SFM-B1 SFM-D1

RSH450
RSH500

RSH3-6c-500
RSH3-6c-600
RSH3-6-600
RSH3-6-700
RSH3-6-800

32402(JP9), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo CorporationNotes on Export Regulations:
Do not commit an act, which could directly or indirectly, violate any law or regulation of Japan,
your country or any other international treaty, relating to the export or re-export of any commodities.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan

Note: Product illustrations are without obligation. Product descriptions, in particular any and all technical specifications,are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.
All product information contained in this brochure is current as of February 2024.

T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp