Catalog No. E12045

ABSOLUTE Digimatic Calipers

Standard & Coolant Proof
300 mm measuring range, now with highly visible LCD!

Small Tool Instruments and Data Management

Digital caliper equipped with absolute scale

ABS Digimatic Caliper
Measuring range: 300 mm

500-153-30

500-714-20

FeaturesEasy-to-see large numbers!
The character height on the LCD is increased to 11.0 mm toenhance visibility (Standard model: Increased by approx.16%,
IP67 model: Increased by approx. 8%, in-house comparison).

Highly resistant to water,oil, and stains! *1
These calipers conform to the IP67 ingress protectionlevel and can be used in the same harsh environments
as purely mechanical calipers.

No zero-adjustment necessary!
Thanks to the ABS (absolute) scale, troublesome zeroadjustment is eliminated.

20 mm beam width aids stability!
Thanks to the 20 mm beam width, the straightness of thescale is excellent and so accuracy is maintained over thewhole measuring range.

Environment-friendly battery life!
The battery life is extended from 3.5 years to 5 years
for the Standard model and from 1 year to 5 years for
the IP67 model (in-house comparison). Reducing battery
replacements helps to protect our environment.

COOLANT PROOF *2 is the universal termfor Mitutoyo Digimatic Small Tool Instrumentsthat are not only resistant to dust and water
ingress but also to deterioration of materials
due to contact with the cutting oil or coolant
fluids in normal use.

Simple battery replacement! *1
There is no screw on the battery lid. Quick battery
replacement and airtightness are achieved with a clicklock lid.

New appearance!
These newly designed products continue the user-friendlinessof the previous models with 200 mm or less measuring range.

*1: This feature is only available on the IP67 model, an IP67 ABS coolant proof caliper.
*2: Some types of aggressive cutting oil or coolant may degrade the sealing materials over time.

2with Dust/Water Protection Conforming to IP67 Level

IP67 ABS Digimatic Coolant Proof Caliper
Measuring range: 300 mm

Centralized data management

By connecting to U-WAVE fit ... Measurement data can be
wirelessly transmitted, eliminating errors caused by handwriting notes.

500-153-30
500-173-30

U-WAVE fit can be directly installed on therear of the display of the measuring instrument
without impeding the operator because
U-WAVE fit has been downsized and thinned.

500-714-20
500-764-20

Transmission unit

U-WAVE-TC

264-621
Connecting unit 02AFZ300

USB input tool direct

USB-ITN-C

Transmission unit

for 500-153-30/500-173-30
06AFM380C

U-WAVE-TC

264-620
Connecting unit 02AFZ310

ReceiverU-WAVE-R

Connecting cable

02AZD810D

See the optional
accessories table at thebottom left of this page.

USB input tool direct

USB-ITN-A

for 500-714-20/500-764-20
06AFM380A

Digimatic mini-processor

DP-1VA LOGGER
264-505

USB cable
06AFZ050

The data logger function allows
storage of up to 1,000 items ofdata in memory.

MeasurLink is an IoT platform for quality managementthat realizes “Visualization of Quality” by enabling
real-time data collection from networked Digimatic
gages, and centralized control and analysis.
Wireless
Digimatic output, communication through wire
USB output, communication through wire

Note: All items except the digimatic calipers are optional.

Optional accessories
ProductFor 500-153-30
500-173-30
For 500-714-20
500-764-20

Connecting cable
Hold unit
Connecting cable

Order No.
959149 (1 m)
959150 (2 m)959143
05CZA624 (1 m)
05CZA625 (2 m)

By connecting to Input Tool Direct ... By connecting to DP-1VA LOGGER ...
Measurement data can be input to Microsoft Measurement data can be printed, and measureExcel, Microsoft Notepad and similar software. ment data and log data can be transmitted in a
Note: Microsoft Excel is a registered trademark ofbatch to a PC with a one-touch operation.
Microsoft Corporation.

3049 1905(1) A(CH)PR, Printed in Japan

Specifications
MetricOrder No.
Measuring rangeResolution
Accuracy*1
Repeatability
MassFine adjustment
Data output
IP code levelPower suppply
Battery life
Scale typeMax. response speed
LCD display
Function
Alarm
Operating temperature
Storage temperature

500-153-30

Inch/Metric500-714-20

500-173-30

0 - 300 mm
0.01 mm±0.03 mm0.01 mm
350 g

500-764-20
0 - 12 in / 0 - 300 mm
0.0005 in / 0.01 mm±0.0015 in / ±0.03 mm
0.0005 in / 0.01 mm
350 g

Thumb roller
Digimatic code format—IP67—
IP67
Silver oxide battery SR44 (1 pc.)
Approx. 5 years under normal use
Electromagnetic induction-type ABSOLUTE linear encoder
Unlimited
7-segment/6-digit, minus [-] sign, character height: 11.0 mm
Power on/off, origin-set, ZERO/ABS, Origin-set, automatic power on/off*4, Power on/off, origin-set, ZERO/ABS, Origin-set, automatic power on/off*4,
data output*2
data output*2, data hold*3
data output*2
data output*2, data hold*3
Low battery voltage, Scale surface contamination, Overflow
0 to 40 ˚C-10 to 60 ˚C*1: Excluding quantizing error*2: By using the optional connecting cable with data switch, measurement data can be output*3: By using the optional data hold unit, the displayed value can be held.
*4: Display will turn off after 20 minutes inactivity. Power is restored when the slider is moved.

Dimensions

Unit: mm40464

2019.627.522500-153-30

3.8404

64

20

19.7

27.5

22500-714-20

3.8

Find additional product literatureand our product catalogue
http://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law. Please consult us
in advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technical specifications,are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
http://www.mitutoyo.co.jp