Coordinate Measuring Machines

High Accuracy CNC Coordinate Measuring Machine

STRATO-Apex Series

Catalog No.E16001(9)Improved body rigidity and scanning measurement technology

STRATO-Apex

High-performance scanning measurement has been achieved through the improved structural
rigidity and incorporation of a newly developed compensation technology.
Scanning mode form error on sphere: P Form.Sph.Scan:SS: Tact.MPE = 1.3 μm (574 Series)

SERIES

Scanning mode time: τ Sph.Scan: PP: Tact.MPL = 40 s(574 Series)(cf. previous FALCIO Series: MPETHP = 2.2 μm)
		MPTtHP = 110 s)

Probe used: SP25M

STRATO-Apex Series: A state of-the-art CNC coordinate
measuring machine with high-speed operation.
The high drive speed and acceleration guarantee top scanning performance and high-accuracy
measurement, down to the 1 μm class.

Equipped with high-performance controller• Uses a digital servo system that processes all control loops for position, speed, and current as digital signals.
• The digital servo system offers the following benefits:
1) Little drift or deterioration with time2) Wide dynamic range
3) Easy implementation of various types of control algorithm

Ultra-high precision glass scales• An ultra-high precision crystallized glass scale which has practically nothermal expansion (coefficient of linear expansion 0.01 × 10-6/ °C) is combinedwith a high-performance reflective linear encoder with resolution of 2/100 μm
to create the ultra-high accuracy measurement unit installed on each axis of
STRATO-Apex.
• A unique securing method used for the scales minimizes the hysteresis errorthat can result from the difference in the coefficients of linear expansion
between the installation plane and scale. In addition, the glass scale is resistantto rust and corrosion, guaranteeing accuracy long term.

Vibration-damping unit included as a standard accessory
ISO accuracy guaranteed with long stylus• Vibration of the floor where the unit is installed shows up as
measurement value variations. The STRATO-Apex Series comes equipped

Compensation technology using the laser tracking interferometer newly developed in-house guarantees accuracy of E280, MPE.

with a vibration-damping unit that uses auto-leveling air springs. The

Workpieces requiring measurement of deep holes can also be measured with high accuracy.

vibration-damping unit not only prevents floor vibrations from reaching

STRATO-Apex 700/900 Series

Maximum permissible error of length measurement		
E 200, MPE = 0.7 + 2.5L/1000 µm
		E 280, MPE = 0.9 + 2.5L/1000 µm

the main unit, but also has a function that uses a sensor to detect load
Spatial compensation system developed in-house

Omnidirectional
measurement at high densitychanges caused by movements of the individual axes and placement of aworkpiece and quickly restores the main unit to horizontal orientation.

CMM measurement space

The use of this vibration-damping platform eliminates variation in

Laser tracking
interferometer

Measurement image from the laser tracking interferometer developed
in-house(omnidirectional measurement at high density)

measured values due to floor vibration.

Use of a wide-angle field angle reflector
enables an omnidirectional measurement
range

Vibration-damping unit with auto-leveling air springs

1

Vibration-damping unit positioning

2

STRATO-Apex 574

External dimensions

unit: mm

Maximum permissible error of length measurement
E 0, MPE=0.7 + 2.5L/1000 (μm)
Measuring range

140
X=500

570

160 175 Y:700

712

750

47

760

88

Z:400
573

2410

Measuring range

405

470

635
1270

800
1700

STRATO-Apex 574
Specifications

Guide method
CNC MODE
Drive speed
J/S MODE
Drive acceleration
Measuring method
Resolution
Material
Work table*
Size (table surface)
Tapped inserts
Maximum height
Workpiece
Maximum mass
Machine mass (includes the vibration-damping platform andcontroller, but not workpiece)
Power supply specifications
Pressure
Air supply
Consumption
Temperature range
Guaranteed accuracy
Per hourtemperature
Temperature change
Per 24 hours
environment
Temperature gradient vertical/horizontal

STRATO-Apex 574
500 mm
700 mm
400 mmAir bearings on all axes
Moving speed: Max.300 mm/s for each axis (maximum combined speed: 519 mm/s)
Measuring speed 1 – 3 mm/s
Moving speed 0 – 80 mm/s
Measuring speed 0 – 3 mm/s
Fine speed 0 – 0.05 mm/s
1333 mm/s2 for each axis (maximum combined acceleration: 2309 mm/s2)
Linear encoder
0.00002 mm
Granite676 × 1420 mm
M8 × 1.25
560 mm180 kgTapped insert positions in the table surface

unit: mm

20

140

*676

**120

1620 kg

13-M8×1.25

Power supply voltage: AC100-120/200-240 V±10 %; power supply capacity: 700 W
0.4 MPa
60 L/min under normal conditions (air source: At least 120 L/min)
18 to 22 °C
1.0 °C
2.0 °C
1.0 °C/m

700
525
350

* While the appearance of the natural stone measuring table varies according to the source, the high stability for which this material is known can always be relied upon.

175

Accuracy0

The following values are guaranteed when using the standard stylus.
Standard stylus (TP200: ø4 × 10, MPP-310Q: ø4 × 18, SP25M: ø4 × 50)

Maximum permissible limit of the repeatability
range
Title
Maximum permissible scanning mode form
error on a sphere

E 150,MPE
R 0,MPL
SymbolP Form.Sph.Scan:SS: Tact,MPE

Maximum permissible scanning mode timeτ Sph.Scan: PP: Tact,MPL

Title

SymbolMaximum permissible single-stylus form error

P Form.Sph.1 × 25:SS: Tact,MPE

0.7 + 2.5L/1000

0.7 + 2.5 L/1000

1.9 + 2.5 L/1000

0.7 + 2.5L/1000

0.7 + 2.5 L/1000

1.2

0.7

0.7

Standard
ISO 10360-5: 2020
JIS B 7440-5: 2022
ISO 10360-5: 2020
JIS B 7440-5: 2022

TP200

MPP-310Q*

SP25M

1.3

1.3

70 s

40 s

Standard
ISO 10360-5: 2020
JIS B 7440-5: 2022

TP200

MPP-310Q*

SP25M

1.8

0.7

0.7

500

1.4 + 2.5 L/1000

175.5
375

SP25M

250

MPP-310Q*0

E 0,MPE

unit: μm
TP200Standard
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013

125

Maximum permissible error of length
measurement

Symbol

88.5

Title

1420

Measuring range

Model
X axis
Y axisZ axis836*1 Workpiece loading area
*2 Y-axis guiding surface

Note: All models incorporate a main unit Startup system (relocation detection system), which disables operation when an unexpected vibration has occurred or themachine has been relocated. Be sure to contact your nearest Mitutoyo Sales Office prior to relocating your machine after initial installation.

20

1062

**120

* Available by custom order only.

3

140

20

1062

140
**120

18×M8

4

STRATO-Apex 700/900 Series

Accuracy
The following values are guaranteed when using the standard stylus.
Standard stylus (TP200: ø4 × 10, MPP-310Q: ø4 × 18, SP25M: ø4 × 50)
Title

ISO accuracy guaranteed with long stylus
Maximum speed and accuracy among moving bridge types

Symbol

E 0,MPEE 150,MPE

Maximum permissible
error of length
measurement

E 200,MPE
E 280,MPE

Maximum permissible
limit of the repeatability
range

unit: μm

Standard

TP200

ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013

1.4 + 2.5 L/1000(700Series)
1.5 + 2.5 L/1000(900Series)
1.9 + 2.5 L/1000(700Series)
2.0 + 2.5 L/1000(900Series)

SP25M+ SM25-3+ SH25-3

0.7 + 2.5L/1000

0.7 + 2.5L/1000

0.7 + 2.5L/1000

0.7 + 2.5L/1000

0.7 + 2.5L/1000

SP80*
0.7 + 2.5L/1000
0.7 + 2.5L/1000

0.9 + 2.5L/1000
1.2

0.7

Title

SymbolMaximum permissible scanning mode form
error on a sphere

P Form.Sph.Scan:SS: Tact,MPE

Maximum permissible scanning mode timeτ Sph.Scan: PP: Tact,MPL

ISO 10360-5: 2020
JIS B 7440-5: 2022
ISO 10360-5: 2020
JIS B 7440-5: 2022

Title

SymbolStandard

P Form.Sph.1 × 25:SS: Tact,MPE

ISO 10360-5: 2020
JIS B 7440-5: 2022

Maximum permissible single-stylus form error

SP25M+ SM25-2+ SH25-2

0.7 + 2.5L/1000

ISO 10360-2: 2009
JIS B 7440-2: 2013

R 0,MPL

MPP-310Q*

SP25M+ SM25-1+ SH25-1

Standard

0.7

0.7

0.7

0.7

MPP-310Q*

SP25M+ SM25-1+ SH25-1

SP25M+ SM25-2+ SH25-2

SP25M+ SM25-3+ SH25-3

SP80*

1.6

1.8

1.8

1.8

1.8

70 s

45 s

45 s

45 s

45 s

TP200

MPP-310Q*

SP25M+ SM25-1+ SH25-1

SP25M+ SM25-2+ SH25-2

SP25M+ SM25-3+ SH25-3

SP80*

1.8

0.9

0.9

0.9

0.9

0.9

TP200

* Available by custom order only.

External dimensions

unit: mmSTRATO-Apex 9106

2830

Measuring range

112

230

175

Y

780

X

Z

88

Measuring range

Item

STRATO-Apex 776

STRATO-Apex
776

43

A

A

C

1000

1410

590

D

B
D

STRATO-Apex
9166
940

700

C

W

STRATO-Apex
9106

740

B

700

STRATO-Apex
7106

1910

682.5
2210

W

2810

1460

1660

Specifications* While the appearance of the natural stone measuring table varies according to the source, the high stability for which this material is known can always be relied upon.

*1062

**120

140

18-M8X1.25

**120

1000

1022

STRATO-Apex 7106

533

250

1222

STRATO-Apex 9106

0
168

168
900

168

266

450

0

675

250

168

800

500

500

0

1067

88.5
0

467
350
233

750

225

750

1720

1000
700

STRATO-Apex 776

1334

13-M8X1.25

13-M8×1.25

1022

1600

13-M8X1.25

2320

140
**120

900

*862

1720

20

140
**120

140

675

3085 kg

*862

20

*1062

450

2410 kg

20

20

225

2180 kgPower supply voltage: AC100-120/200-240 V±10 %; power supply capacity: 700 W
0.4 MPa
60 L/min under normal conditions (air source: At least 120 L/min)
19 to 21 °C
1.0 °C
2.0 °C
1.0 °C/m

unit: mm88.5
01895 kg

Tapped insert positions in the table surface

700

Material
Work table*
Size (table surface)
Tapped inserts
Maximum height
Workpiece
Maximum mass
Machine mass (includes the vibration-damping platform
and controller, but not workpiece)
Power supply specifications
Pressure
Air supply
Consumption
Temperature range
Guaranteed accuracy
Per hourtemperature
Temperature change
Per 24 hours
environment
Temperature gradient vertical/horizontal

1000 mm1600 mm
600 mmAir bearings on all axes
Moving speed: Max.300 mm/s for each axis (maximum combined speed: 519 mm/s)
Measuring speed 1 – 3 mm/s
Moving speed 0 – 80 mm/s
Measuring speed 0 – 3 mm/s
Fine speed 0 – 0.05 mm/s
1500 mm/s2 for each axis (maximum combined acceleration: 2598 mm/s2)
Linear encoder
0.00002 mm
Granite862 × 1420 mm
862 × 1720 mm
1062 × 1720 mm
1062 × 2320 mm
M8 × 1.25
770 mm
500 kg
800 kg1200 kg

525

Drive acceleration
Measuring method
Resolution

700 mm

350

J/S mode

STRATO-Apex 9166
900 mm88.5
0175

Drive speed

STRATO-Apex 9106

1420

CNC mode

STRATO-Apex 7106
700 mm700

Guide method

STRATO-Apex 776

88.5
0
175
350
525

Measuring range

Model
X axis
Y axisZ axis1222

STRATO-Apex 9166
*1 Workpiece loading area
*2 Y-axis guiding surface

Note: All models incorporate a main unit Startup system (relocation detection system), which disables operation when an unexpected vibration has occurred or themachine has been relocated. Be sure to contact your nearest Mitutoyo Sales Office prior to relocating your machine after initial installation.

5

6

STRATO-Apex 1600 Series

STRATO-Apex 162016/163016

unit: μm

Title

Symbol

E 0,MPE

Maximum permissible error oflength measurement

High accuracy combined with wide measuring range
Best suited for highly accurate measurement of large workpieces.

E 150,MPE

Maximum permissible limit of therepeatability range

R 0,MPL

Title
Maximum permissible scanningmode form error on a sphere
Maximum permissible scanningmode time
Title
Maximum permissible singlestylus form error* Available by custom order only.

TP200

SP25M

SP80*

4.0 + 4.0 L/1000

3.0 + 4.0 L/1000

3.0 + 4.0 L/1000

4.0 + 4.0 L/1000

3.0 + 4.0 L/1000

3.0 + 4.0 L/1000

4.0

2.5

2.5

Standard
ISO 10360-5: 2020
JIS B 7440-5: 2022
ISO 10360-5: 2020
JIS B 7440-5: 2022

TP200

SP25M

SP80*

3.0

3.0

60 s

60 s

Standard
ISO 10360-5: 2020
JIS B 7440-5: 2022

TP200

SP25M

SP80*

4.0

2.8

2.8

Standard
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013

SymbolP Form.Sph.Scan:SS: Tact,MPE
τ Sph.Scan: PP: Tact,MPL
SymbolP Form.Sph.1 × 25:SS: Tact,MPE

External dimensions

unit: mm2805

D

STRATO-Apex
162012

Item
A
B
C
D
H

Measuring range

STRATO-Apex 1600 Series

Measuring range

STRATO-Apex 162012

1815
1000
3740

4340

5140

STRATO-Apex
163012STRATO-Apex
163016

B

Z

A

45

C

C

700
1415

1815
1350
4740

4390

5190

Tapped insert positions in the table surface

unit: mm
20

20

1850*

1850*

190
170**

190
170**
45-M8X1.25
3000

30-M8X1.25

2400

1600

2000

1200

15350 kg

1600

4280

2700

2000

1200

800

800

400

400

Power supply voltage: AC100-120/200-240 V±10 %; power supply capacity: 1500 W
0.4 MPa
100 L/min under normal conditions (air source: At least 250 L/min)
18 to 22 °C
1.0 °C
2.0 °C
1.0 °C/m

0

2060

386.5

0

STRATO-Apex
163012/163016

135
1600

135

1200

STRATO-Apex
162012/162016

015300 kg

1760

800

11200 kg

702

A
B
C
D
H

400

11150 kg

160

(893.5)3280

Material
Work table
Size (table surface)
Tapped inserts
Maximum height
Workpiece
Maximum mass
Machine mass (includes the vibration-damping platform andcontroller, but not workpiece)
Power supply specifications
Pressure
Air supply
Consumption
Temperature range
Guaranteed accuracy
Per hourtemperature
Temperature change
Per 24 hours
environment
Temperature gradient vertical/horizontal

X:1600Y

386.5

Drive acceleration
Resolution
Measuring method

135

250 386.5

1600

J/S MODE

570

800

Drive speed

760

1200

CNC MODE

Item

0

Guide method

STRATO-Apex 162016
STRATO-Apex 163012
STRATO-Apex 163016
1600 mm2000 mm3000 mm
1200 mm1600 mm
1200 mm1600 mm
Air bearings on all axes
Moving speed: Max.350 mm/s for each axis
(maximum combined speed: 606 mm/s)
Measuring speed 1 – 3 mm/s
Moving speed 0 – 80 mm/s
Measuring speed 0 – 3 mm/s
Fine speed 0 – 0.05 mm/s
780 mm/s2 for each axis (maximum combined acceleration: 1350 mm/s2)
0.00005 mm
Linear encoder
Granite1850 × 3280 mm
1850 × 4280 mm
M8 × 1.25
1350 mm
1750 mm1350 mm
1750 mm3500 kg
4000 kg

400

Model
X axis
Y axisZ axis712

Measuring range

650
1415

H

Specifications

STRATO-Apex
162016*1 Workpiece loading area
*2 Y-axis guiding surface

2060* While the appearance of the natural stone measuring table varies according to the source, the high stability for which this material is known can always be relied upon.

Accuracy
STRATO-Apex 162012/163012
The following values are guaranteed when using the standard stylus.
Standard stylus (TP200: ø4 × 10, MPP-310Q: ø4 × 18, SP25M: ø4 × 50)
Title
Maximum permissible error oflength measurement
Maximum permissible limit of therepeatability range
Title
Maximum permissible scanningmode form error on a sphere
Maximum permissible scanningmode time
Title
Maximum permissible single-stylus
form error* Available by custom order only.

7

Symbol

E 0,MPEE 150,MPE
R 0,MPL
SymbolP Form.Sph.Scan:SS: Tact,MPE
τ Sph.Scan: PP: Tact,MPL
SymbolP Form.Sph.1 × 25:SS: Tact,MPE

CNC coordinate measuring machine with a structure suited to high-accuracy
measurement of large and/or heavy workpieces

TP200

SP25M

SP80*

3.5 + 4.0 L/1000

2.5 + 4.0 L/1000

2.5 + 4.0 L/1000

3.5 + 4.0 L/1000

2.5 + 4.0 L/1000

2.5 + 4.0 L/1000

STRATO-Apex G

3.5

2.5

2.5

This CNC CMM has been developed with an original Mitutoyo structure (moving

Standard
ISO 10360-5: 2020
JIS B 7440-5: 2022
ISO 10360-5: 2020
JIS B 7440-5: 2022

TP200

SP25M

SP80*

2.5

2.5

60 s

60 s

Standard
ISO 10360-5: 2020
JIS B 7440-5: 2022

TP200

SP25M

SP80*

3.5

2.3

2.3

unit: μm
Standard
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013
ISO 10360-2: 2009
JIS B 7440-2: 2013bridge/foundation installed type) .

Note: All models incorporate a main unit Startup system (relocation detection system), which disables operation when an unexpected vibration has occurred or themachine has been relocated. Be sure to contact your nearest Mitutoyo Sales Office prior to relocating your machine after initial installation.

8

SOFTWARE

Optional Accessories

Application software with high performance and operability

For versatile measurements

MCOSMOS

MiCAT Plannercompact high-accuracy scanning probe SP25M

non-contact laser probe SurfaceMeasure<Data processing device for 3D measuring instruments><Automatic measurement program generation software for CMMs>

New processing program family for 3D measuring instruments, running on Windows. The
wide-ranging lineup of optional software supports various probes, enabling fully automatedmeasurement of all kinds of workpieces.

This software package dramatically reduces part-programming creation time byautomatically generating the part program. Tolerance information from a 3D CAD modelis read to determine which features of the part should be measured to verify conformanceto specification. Compared to conventional methods (teaching), this method creates moreefficient measurement programs as well as saving time.

This is a compact, highaccuracy, multi-function scanning probe with a 25mm outside diameter that makes scanning measurements, high-accuracy
point measurements, and centripetal point measurements (optional function).
The SP25M is used with the PH10MQ/10M auto probe head to provide a highdegree of measurement freedom.

The SurfaceMeasure probe moves while irradiating the workpiece with the laser, collecting
coordinate values from the workpiece surface. It is capable of obtaining large quantities of
surface 3D data at high speed.

See video here

Rotary table

9

GEARPAK Express

MSURF<Gear Measurement and Evaluation Software for CNC Coordinate Measuring Machines><non-contact laser measurement and evaluation program>

A 3D model created from the provided gear specifications enables you to visually and easily
check whether measurement will be performed as intended.
Furthermore, automatic program creation and on-screen measurement guidance help quick
and easy setting of the coordinate system.

MSURF-S is used for obtaining measured point cloud data with the SurfaceMeasure (noncontact laser probe), while MSURF-I is used for comparing this data with the master model
data, and for making dimensional measurements.
Furthermore, MSURF-G for offline teaching allows the user to create a measurement macroeven without the actual workpiece, improving the measuring machine's uptime.

MRT320

MRT240

These optional rotary tables allow highly-accurate
and efficient measurements of workpieces likerotationally symmetrical parts (gears, impellers
and cylindrical cams). In addition, the use of 4-Axis
nominal scanning enables top-class high-speed
measurement without impacting accuracy.

102406(JP14), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Notes on Export Regulations:
Do not commit an act, which could directly or indirectly, violate any law or regulation of Japan,
your country or any other international treaty, relating to the export or re-export of any commodities.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.
All product information contained in this brochure is current as of June 2024.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp