LCX■Speciﬁcations
Model

LCX-200

LCX-100
10,000rpm

Maximum Speed
Maximum RCF

11,740G

Maximum Capacity

1,000ml (250ml x 4)

Speed Control System

Microprocessor feedback control

Drive Motor

Induction motor

Drive System

Flex-Spin drive system (Direct drive with an automatic alignment function)

Low Speed Bench-top Centrifuge

Data Entry

Jog Dial equipped with an <ENTER> button

Speed Setting Range

100rpm to 10,000 rpm（100rpm increments）

RCF Setting Range

10 G to 11,740 G(Less than 300G : 10G increments, Over 300G : 100G increments）

Time Setting Range

From 10 sec to 50 sec, in 10 sec increments,from 1 min to 99 min, in 1 min increments or <F-> for free

Safety Devices• Imbalance detector • Lid interlock • Lid open/close detector • Abnormal speed detector• Over-current circuit breaker • Rotor identiﬁcation system • Motor over-current detector

Additional Functions• Three-step acceleration characteristics selection • Three-step deceleration• characteristics selection • Three memory function • FLASH (momentary spin) function• BART Code selection • Memory function for the last run parameters

Rated Current
Power Requirements−220-240V : 4A

1-Phase AC120V 60Hz 15A−

1-Phase AC220-240V 50/60Hz 8A

1-Phase AC220-240V 50/60Hz 8A

660W (568kcal/h)430W (370kcal/h)Rated breaking current 120V : 15A−

Power Consumption(Heat output)
Breaker

120V : 10A
220-240V : 6A

Rated breaking current 220-240V : 7.5A

Rated breaking current 220-240V : 7.5A
Class I

Protection Against Electric ShockDimensions(excluding hinge)

460W x 584D x 355H (mm)430W x 554D x 355H (mm)

Net Weight

41kg

39kg

Environmental Requirements

Ambient temperature range：6 to 35℃, Relative humidity：not exceeding 75％RH,Altitude: up to 2000m, Intended exclusively for use indoors,
overvoltage category：II, pollution degree：2, inclination：not exceeding 2 degrees

Accessories Included

Operator's manual x1copy, BART Code table x1pc,
Clear storage case for operator's manual x1pc, Rotor locking knob x 1pc,
Power cord x1pc, Digital Warranty Information x1copy, Inspection sheet x1copy.

LCX-200

Sales Office:3-14-17 Tagara, Nerima-ku, Tokyo 179-0073, Japane-mail: info@digital-biology.co.jp
URL: https://www.digital-biology.co.jp/manufactured/
phone: +81-3-5971-8160 fax: +81-3-3970-6036

LCX-100

Manufacturer:
3-14-17 Tagara, Nerima-ku, Tokyo 179-0073, JapanAll TOMY products have a limited one-year warranty.
Specifications are subject to change according to product advancement.
Tomy and Digital Biology is registered trademark of Tomy Seiko Co.,Ltd.
and Tomy Digital Biology Co., Ltd. Copyright 2016,
Tomy Seiko and its subsidiaries. Printed in Japan.

Series05024⑤CLCX-SeriesMulti-purpose & User Friendly Features

High Speed Fixed Angle Rotor Available (MAX. 10,000 rpm)

Easy Rotor Mount with One-push Rotor Locking Knob

Rotor Type & ModelBucket

TS-4CAcceleration and Deceleration Characteristics
Max. speed

Various Tubes & Micro/Deep-well Plates Available(From Microtubes to 250ml Bottles, Blood Collection Tubes, etc.)

Swing - Out Rotors

Speed can be set
in 100 rpm
increments
500rpmTube Rack

S4096-02
7015-06
7215-06
7015-087115-08

1,800
3,600
3,600
3,500
3,500

7M5015-1 ＊93,500

3 Deceleration Steps& Natural Deceleration

3 Acceleration Steps

7050-01
7150-017050-02

4,000
5,000
3,500
3,100
3,100
1,600

0705-10P0705-FA10PB407Each Three (3) Acceleration and Deceleration Steps with Natural Deceleration

SC-23305-07P
3307-07P

Easy to See and Operate Control Panel with Jog Dial and Dual Color LED Display
Swing - Out Rotors

Lid Open & Sample Load Available during Power Oﬀ
Compatible Rotor Line-up with TOMY CAX/AX/NIX-Series Centrifuges
Dual Color LED Display

B433＊10TS-33C

Bucket cap kit B433

<Red: Rotor is spinning>
CS-1（Multi Rack for BloodCollection Tubes）
5 to 10ml×56
Multi Rackfor TS-38C Rotor

38M-TC0204P（Mix Rack for Conical Tubes）
50ml×8 or 15ml×16 for TS-38C Rotor

B438-96
B438-1507BH
B438-5002BH
TS-41C

Fixed Angle Rotors

Rotor Type & Model

Max. Speed (rpm)Max. RCF (G)

CA-1
CA-5

10,00010,000CA-6

10,000

7,940
9,390
9,390
8,270
9,390

CA-10
CA-12
CA-14HS

10,00010,000＊1

10,000Note＊5 Plate size up to 128.5L×86D×32Hmm
Microplate×4＊815 (R)×24, 15 (C)×1615 (R)(C)×24
Glass, Conical tube
＊815 (R)×32, 15 (C)×1615 (R)(C)×32
Glass
50 (R)×4＊7
Conical tube
50 (C)×4＊6
Glass
15 (R)×16
Conical tube
15 (C)×1650 (R)(C)×4, 15 (R)×4 ＊6 ＊7
Glass, Conical tube
50 (R)(C)×4, 15 (R)×4 ＊6 ＊750 (R)(C)×8, 15 (R)×8 ＊6 ＊7
Veno-Ject II tube 5ml (TER)
5 (R)×40
FLC＊3
Collection bucket of ﬂoating cells
8×4
Veno-Ject II tube 5ml (TER)
5 (R)×287 (R)×28
Veno-Ject II tube 7ml (TER)

FLC
Glass, PAX gene blood collection tube

14 (R)×1615 (R)×2815 (C)×1650 (C)×450 (R)×4250 (F)×4225 (C)×4175 (C)×43,030
3,140
3,1403,030
3,1803,1803,180

Conical tube
GlassNALFLCFLC＊11 ＊12＊113815-16P
3810M-14P
3806-EK20P
3810-N20P
3805-FA16P
3802-EP24P

3,500
3,500
3,500
3,500
3,500
3,500

4,200
4,000

2,370
2,380
2,330
2,270
2,080
1,780
3,650
3,590
3,650
3,650
3,100
3,310

4,000

3,310

4,500

3,150

＊14

2×16
1.5×16
0.5×200.4×36
0.25×3650 (C)×16
50 (R)×16
50 (C)×8 or 15 (C)×1650 (R)×8
15 (C)×40

Conical tube
NAL
Conical tube
Glass
Conical tube

Glass
15 (R)×64
5〜10 (R)×56
Blood collection tube
6 (R)×80
EIK
10 (R)×80
NAL
5 (R)×64, 4 (R)×64
FLC or SIO
2 (C)×96, 1.5 (C)×96
250 (F)×4
NAL
250 (R)×4
225 (C)×4＊11FLC175 (C)×4＊5 Plate size up to 128.5L×86D×33Hmm
Microplate×415 (C)×28
50 (C)×8
Conical tube
15 (C)×8
Deepwell plate×4 ＊5＊15 Plate size up to 128.5L×86D×90Hmm ＊15＊5Plate size up to 128.5L×86D×58Hmm
Microplate×8

Various series of Blood-Collection tubes available.

Fixed Angle Rotors

10,000

2,990

8,160
7,940
7,160
8,830
7,380
2,380
2,380
2,380
2,380
2,380

4,200

2ml×2410,000rpm : 9,390G

＊1

4,500
4,670
4,670
4,500
4,720
4,720
4,720

10,000

Bucket, Adapter and Cover are included.

CA-5

CA-8

4,100
4,100
4,100
4,100
4,100
4,100
4,100

2,760
2,990

3,500
3,500
3,500
3,500
3,500

＊1 ＊14(for LCX-200)

2,900

3850-04P
3850-N04P
38M-TC0204P ＊13
3850-02P
3815-10P

B438-29

3810M-14P（Mix Bucket）
50ml×4＋15ml×16 for TS-7C Rotor

4,100

Rotor Capacity
ml (bottom shape) × number(F) : Flat, (R) : Round, (C) : Conical

Sealed cap for Bucket B433
CAP433-1 with gasket

10,000TS-38C

LCX-100＊10M0406-05(for LCX-200)

Mix Bucket&Mix Rack

Max. RCF (G)LCX-200

510
2,320
2,430
2,190
2,380
2,150
2,160
2,340
2,370
2,810
4,670
2,150
1,670
1,670
420

4,000
5,000
5,000
5,000
5,000
5,000
5,000
5,000

10,000

B438

Blood Collection Tubes

4,100

M0415-04

Popular user-friendly Jog Dial with an〈ENTER〉 button
enables setting to be changed more quickly and surely.

7M5015-1

3314-04P
3315-G07P
3315-TC04P
3350-TC01P
3350-G01P

4,200

M0404-09<Green: Rotor is not spinning>

15ml & 50ml

LCX-100

TS-7C

Flex-Spin Drive allows excellent tolerance for minor diﬀerences in sample volumes.

Easy to See and Operate Control Panel

Max. Speed (rpm)
LCX-200

Rotor Capacity
ml (bottom shape) × number(R) : Round, (C) : Conical

9,060
9,730
9,170
9,390
7,490

2×18
2×24
2×18 (Outer line)
2×18 (Inner line)
50 (C)×4
15 (C)×42×24
5 (C)×12
50 (C)×4
5 (R)×4
2×18 (Outer line)
0.5×12 (Inner line)
0.2×12 (Middle line)
50 (C)×8
15 (C)×8

CA-15

10,0008,500
6,6005,810

CA-16

10,00011,74011,180CA-14HS

50ml×410,000rpm : 9,390G
or 5ml × 410,000rpm : 7,490G

CA-15

CA-16

Outer line 2ml×1810,000rpm : 8,500G
Inner line 0.5ml×1210,000rpm : 6,600G
Middle line 0.2ml×1210,000rpm : 5,810G

50ml×810,000rpm : 11,740G

Note

TS-7C

15ml×325,000rpm : 4,670G

Conical tube

250ml×45,000rpm : 4,720G ＊16

TS-38C

250ml×44,200rpm : 3,650G

TS-4C

Microplate×41,800rpm : 510G

TS-41C

Deepwell plate×44,500rpm : 3,150G ＊15■Manufacturer Abbreviation

TOM
Conical tube
FLC

＊2Conical tube

＊4

● The above speciﬁcations show the maximum value of speed, RCF(G) and sample capacity. These values may change according to the combination of rotor, bucket, adapter and tubes.

TS-33C＊3

TOM: TOMY SEIKO CO., LTD. FLC: Falcon/CORNING INTERNATIONAL CORP. COR: CORNING INTERNATIONAL CORP. EIK: EIKEN CHEMICAL CO., LTD.
NAL: NALGENE/Thermo Fisher Scientiﬁc K.K. SIO: SHIONOGI & CO., LTD. TER: TERUMO CORPORATION＊1 : Diﬀerent types of tubes cannot be loaded together. ＊2 : TOMY 5ml conical tube, PT0050-11. ＊3 : Falcon 5ml FACS tubes.
＊4 : For use with 15ml conical tubes, an optional Adapter CA1500-01 is required. ＊5 : Prior to centrifugation, check the durability of plates in use. ＊6 : For 15ml tubes, an adapter A1500-04 is required.
＊7 : For conical tubes, an adapter A3010-01P is required. ＊8 : For conical tubes, only four holes of the center of the bucket can be used. ＊9 : Can be loaded 50ml tube and 15ml tube mixed together.
＊10 : B433 bucket can be used as a sealed bucket by using an optional [Bucket cap kit B433]. ＊11 : An optional conical Tube Adapter (Falcon Tube Cushion, Cat. No.352090) is required.
＊12 : Falcon 225ml tube cannot be used with Bucket cap kit B433. ＊13 : Can be loaded up to 50ml ×4 and 15ml × 8 tubes mixed together. ＊14 : Sealed bucket＊15 : 2 deepwell plates (up to 128.5L×86D×80Hmm) are available with cover. ＊16 : Values indicate those of the rotor being mounted in the LCX-200.
has been tested and certiﬁed by the Health Protection Agency (HPA, UK) to be compliant with the International Standard (Annex AA of the IEC 61010-2-020), meets requirements for biohazard safety.
adopted a hermetically sealed structure equivalent torotors designed and tested in accordance with the International Standard (Annex AA of the IEC 61010-2-020) to oﬀer excellent sealing performance.