Form Measurement

Surface Roughness Measuring System

SURFTEST SJ-500/SV-2100

Catalog No. E15006(5)Surface roughness testers offer benchtop or portable
operation and the choice of data analysis by PC or aneasy-to-use dedicated processor

Dedicated data processor type

Surftest SJ-500/ SV-2100
Improved operability

Various types of analysis

High-durability

7.5" Color TFT LCD
The dedicated data processor has a high-visibility 7.5" color TFT LCD. Icon display
and touch panel operation provide user-friendly display and easy operation.
Positioning by joystick and manual control knobs on the processor
Easy-to-operate joystick. Fine positioning of stylus required for small-holemeasurements can be easily performed using the manual fine-adjustment knobs.
Multiple trace function
A machine can be programmed to take up to three traces, one after the other.

Easy operation, high-accuracy analysis of surface
roughness and fine contours!

High-visibility color display panel

Efficient positioning by joystick and adjustment knobs

A high-visibility 7.5" color TFT LCD, color icon display and touchoperated panel provide user-friendly, easy operation.
Built-in thermal printer. Fine contour analysis provided as standard.

Both a fast-traverse joystick (X-axis: 20 mm/s for SJ-500, 40 mm/s
for SV-2100, Z2-axis: 20 mm/s for SV-2100S4/H4/W4) andmanual fine-adjustment knobs, essential for positioning in small holemeasurement, are standard features.
Positioning in small hole measurement
Positioning in Y/Z-directions
with column fine-adjustment
knob (or detector elevationknob) and optional cross-travel
table.

Capable of fine-contour analysis
Supports 43 types of analysis parameters, complying with surface roughness standards
such as ISO 1997 and JIS 2001. Also capable of various fine-contour analysis.
* Contour analyses: Area, circle, angle, coordinate difference, step, inclination

Ceramic guideway
A ceramic guideway, inherently free from wear and deterioration with age, isemployed to maintain the traversing straightness of the drive unit (X axis) indefinitely.
Maintenance-free design, since anti-corrosion treatment is not required for ceramic.

SJ-500Traverse: 50 mm
Compact, high-performance type

PC data-processing system
Next-level, advanced processing

Positioning at the trace startpoint with X-axis fineadjustment knob.

Supports 16 languages
Japanese, English, German, French, Italian, Spanish, Portuguese,
Korean, Simplified Chinese, Traditional Chinese, Czech, Polish,
Hungarian, Turkish, Swedish, Dutch

Multiple trace programming function

Navigation function aids leveling

A machine can be programmed to take up to three consecutive tracesby one-key operation, as shown in the figure below.

Powerful support for leveling adjustments• SJ-500/SV-2100M4
Consecutive tracing in
X-axis direction only

When using an optional 3-axis adjustment table or leveling table, anavigation screen is available to help the operator level the surface tobe tested.

• SV-2100S4/H4/W4
X-axis tracing with programmed Z-axis
shifts possible

Example of 3-axis adjustable table

SV-2100M4

Traverse: 100 mm
Manual column type

Dedicated data processor

Advanced processing and easy operation

SV-2100S4/H4/W4

Measurement

Example: SV-2100S4 input screen

Traverse

Traverse: 100 mm
Power column typeThe user is guided through theleveling procedure to determine theamount of adjustment needed.

2

3

A portable tester also boasting high performancein desktop applications

Surftest SJ-500
High accuracy, high performance, userfriendly display and easy operation
Ceramic guide

Vertical adjustment knob
Essential for positioning the stylus close to theworkpiece!

A desktop tester that's easy to use for portableapplications

Surftest SV-2100

By setting the origin point at start-up, the Absolute scale system allows
accurate positioning for repeated or multiple measurements.

Detector elevation knob

Z-axis fine-adjustment
knob (0.67 mm/rev)

Z-axis fine-adjustment
knob (1 mm/rev)

Leveling knob

HandleX-axis fineadjustment knob
X-axis fine-adjustment
knob (2 mm/rev)

X-axis fine-adjustment
knob (2 mm/rev)
Detector

Detector

Support for testing problematic features
Supports measurement in the axial direction for shrouded
features, such as found on crankshafts, by simply swiveling thedetector through 90 degrees.

Skid/skidless measurement selector knob

Detector

Class-leading traverse straightness: 0.2 μm/50 mm Normal tracing
High-speed traverse at up to 20 mm/s under
joystick control
Smooth positioning using the verticaladjustment knob

SV-2100M4

Off

SV-2100S4

High-speed traverse at up to 40 mm/s (X-axis) under joystick control
Smooth positioning, using the Z-axis fine-adjustment knobs Stable,
high-accuracy measurement with a traverse straightness of
0.15 μm/100 mm

Digital Adjustment Tilting (DAT) function is supplied as
standard for efficient leveling of workpieces: ±1.5°On

Connector unit

Tracing a shrouded feature

Drive unit inclination adjustment mechanism

DAT function

Connector unit

1. Capable of a series of automatic measurements, plus auto leveling(optional) and stylus retraction. Accurate positioning for repeated or
multiple measurements possible.

2. SV-2100S4 / H4 / W4 models are equipped with an emergency
stop button.
Emergency stop
buttonPowerful support for manual leveling!!

Preparatory
measurement

Adjustment
value display

Adjust according todisplayed value usingleveling knob

Preparatory
measurement

Adjust usingleveling knob

Verification
measurement

Checking
counter

Actualmeasurement
OK

3. Base sizes and vertical travel range on column

Reduction ofsetting time
Accurate leveling
Easy operation

4

Vertical travel
range

SV-2100S4

350 mm

SV-2100H4

Measurement setup screen

SV-2100W4NG

RepeatModel No.

SV-2100M4

5

550 mm
350 mm

Vertical traverse
methodPower and manual

Base size
600×450 mm
1000×450 mm

Manual only

600×450 mm

Dedicated data processor

Data processing unit
• Data saving (internal memory)• High-speed printing• Expansion slot for external
memory (CF card)• High-visibility 7.5" color LCD• Touch panel with color
icon display

Analysis to international standards

Easy, icon-based input of setup conditions

Evaluates surface roughness using up to 43 parameters complying
with international standards such as ISO 1997 and JIS 2001.
Bearing Area Curve (BAC), Amplitude Distribution Curve (ADC), andpower spectrum (wavelength display) are readily available in graphform.

Setups are aided by icons representing ISO / JIS roughness standard
parameters with appropriate values selected from recommended lists.
Typical surface roughness symbol on drawing

Note: Patent registered in Japan, U.S.A., China,
Germany, U.K., and France.

Grinding
Ra 1.5⊥ 0.0025-0.8/ Rz8max 3.3• Display supports 16
languages• Joystick• Key panel

Typical result of icon-based setup
U”X”0.0025—0.8/ Rz8max 3.3

Customizable menu screen
A large variety of optional accessories
The menu customization
function allows display of
frequently used menu icons

One-touch display of various
screens

Home screen

Options supporting measurement
including an auto leveling table,
a 3-axis adjustment table, and aleveling table. Furthermore, these canbe easily operated via a navigation
function. (Supported accessories differ
depending on the model.)Clicking a parameter icon
displays the recommended
cut-off value, etc.

Built-in thermal printerFine-contour analysis

Measurement data is printed by the high-definition,
high-speed thermal printer.
In addition to calculation results and evaluation results,
BAC, ADC and other curves can also be printed.

Various contour analyses (area, circle, angle, coordinate difference, step,
inclination) are supplied as standard.
Evaluation setup screen

Measurement setup screen

Calibration screen

Contour analysis screen

Statistical processing

Saving and recalling measurement setups

Statistical data processing possible (up to 300 data samples)
Statistical processing items: MAX., MIN., average, standard deviation,
histogram, probability of acceptance.

Up to 10 measurement setups can be saved to and recalled frominternal memory.
One-touch recall of stored setups
Select desired analysis icon and then specify the range.

Simplified communication program for SURFTEST SJ-500/SV-2100

Statistical data input

Click the desired measurement
setup file

Statistical results

Measurement screen opens

The Surftest SJ-500 / SV-2100 Series has a USB interface, enablingdata to be transferred to a spreadsheet or other software. We also
provide a program that lets you create inspection record tables using a
Microsoft Excel* macro.

• OS: Windows XP-SP3
Windows Vista
Windows 7
Windows 8
Windows 10• Spreadsheet software: Microsoft Excel 2000
Microsoft Excel 2002
Microsoft Excel 2003
Microsoft Excel 2007
Microsoft Excel 2010
Microsoft Excel 2013
* Windows OS and Microsoft Excel are products ofMicrosoft Excel 2016
Microsoft Corporation.

This program can be downloaded free of charge from the Mitutoyo
website.
https://www.mitutoyo.co.jp

The optional USB cable is also required.
USB cable for SJ-500 / SV-2100 Series 12AAH490

67

PC data processing type

Surftest SJ-500P/SV-2100M4

Specifications
Type of data processing

Dedicated data processor
PC system
SV-2100S4
SV-2100H4
SV-2100W4
SJ-500P
SV-2100M4
178-680-01
178-682-01
178-684-01
178-534-11
178-638-11
with 0.75 mN mm
detectorinch / mm
178-681-01
178-683-01
178-685-01
178-534-13
178-638-13
Order No.*mm
178-680-02
178-682-02
178-684-02——
with 4 mN
detectorinch / mm
178-681-02
178-683-02
178-685-02——
X axis
100 mm (4 in)
50 mm (2 in)
100 mm (4 in)Measuring range
Z axis (detector)
800 μm, 80 μm, 8 μm (32000 μin, 3200 μin, 320 μin)Detection method
Differential inductance
Resolution
0.01 μm (800 μm range), 0.001 μm (80 μm range), 0.0001 μm (8 μm range) (0.4 μin <32000 μin range>, 0.04 μin <3200 μin range>, 0.004 μin <320 μin range>)
Detector
Stylus tip shape (Angle /60°/ 2 μm, 0.75 mN
Radius) Measuring force
90°/ 5 μm, 4 mN
Measuring speed
0.02 - 5 mm/s
0 - 20 mm/s
Drive
speed
0
40
mm/s
or
manual
0 - 20 mm/s or manual
0 - 40 mm/s or manual
Drive unit
or manual(X axis)*1*1
0.15 μm / 100 mm
0.2 μm / 50 mm
0.15 μm / 100 mm
Straightness
0.2 μm / 50 mm
Resolution
0.05 μm (1.97 μin)
Travel—350 mm (H4, W4: 550 mm)—
350 mmDrive unit
Drive speed— *2
Manual
0 - 20 mm/s or manual— *2
Manual(Z2 axis)
Resolution—
1 μm (39.4 μin)—
Applicable standards
JIS1982 / JIS1994 / JIS2001/ ISO1997/ANSI / VDA
Pa, Pq, Pz, Pp, Pv, Pt, Psk, Pku, Pc, PSm, PΔq, Pmr, Pmr(c), Pδc, Rk, Rpk, Rvk, Mr1,
Pa, Pq, Psk, Pku, Pp, Pv, Pz, Pt, Pc, PSm, PΔq, Pmr(c), Pmr, Pδc,
Mr2, A1, A2, Rmax, Rz, Pmax, PzJIS
Ppq, Pvq, Pmq, PPc, PzJIS, PPI, Ra, Rq, Rsk, Rku, Rp, Rv, Rz, Rt, Rc,
Ra, Rq, Rz, Rp, Rv, Rt, Rsk, Rku, Rc, RSm, RΔq, Rmr, Rmr(c), Rδc, Ry, R3z, RPc, Pc, Sm,
RSm, RΔq, Rmr(c), Rmr, Rδc, Rpq, Rvq, Rmq, Rmax, RzJIS, Wa,
Parameter
S, HSC, Rmax, RzJIS, Rppi, RΔa, Rλa, Rλq, Rlo, Rlr, tp, Htp, Vo
Wq, Wsk, Wku, Wp, Wv, Wz, Wt, Wc, WSm, WΔq, Wmr(c), Wmr,
Wa, Wq, Wz, Wp, Wv, Wt, Wsk, Wku, Wc, WSm, WΔq, Wmr, Wmr(c), Wδc, Wy,
Wδc, Wst, Rk, Rpk, Rvk, Mr1, Mr2, A1, A2, Rx, AR, R, Wx, AW, W,
W3z, WPc, Wmax, WΔa, Wλa, Wλq, Wlo, Wlr
Wte, Ry, RyDIN, RzDIN, R3y, R3z, S, HSC, Lo, lr, Δa, λa, λq, Vo,
R, Rx, AR, W, Wx, AW, Wte
Htp, NR, NCRX, CPM, SR, SAR, NW, SW, SAW
Primary profile, Roughness profile, Waviness profile, Filteredwaviness profile, Band waviness profile, Rolling circle wavinessPrimary profile, Roughness profile, Waviness profile, DF profile,
Filtered profile
profile, Rolling circle center line waviness profile, envelope
Roughness motif profile, Waviness motif profile
residual profile, DIN4776 profile, Roughness motif profile,
Waviness motif profile
ADC, BAC Graph, power spectrum graph, auto-correlationgraph, Walsh power spectrum graph, Walsh auto-correlation
Analysis graph
ADC, BAC, power spectrum graph
graph, local peak distribution graph, slope distribution graph,parameter distribution graph
Inclination (Entire, Arbitrary), Circular compensation, Elliptical compensation,
Inclination (Entire / start / end /Arbitrary), R-surface correction,
Data compensation functions
Parabolic compensation, Hyperbolic compensation, Conic compensation,
Elliptical compensation, Parabolic compensation, Hyperbolic
Stylus circular-arc compensationcompensation, Conic compensation, Polynomial correction
Gaussian, 2CR75, 2CR50, 2CRPC75, 2CRPC50, Robust-Spline,
FilterGaussian, 2CR75, PC75, Robust-Spline
Robust-Gaussian
λs: 0.25 / 0.8 / 2.5 / 8 / 25 / 80 / 250 μm / Noneλs: 0.0008 / 0.0025 / 0.008 / 0.025 / 0.08 / 0.25 / 0.8*3
λc: 0.025 / 0.08 / 0.25 / 0.8 / 2.5 / 8 / 25 / 80 mm
λc: 0.025 / 0.08 / 0.25 / 0.8 / 2.5 / 8.0 / 25 /Arbitrary
Cutoff value
λf: 0.08 / 0.25 / 0.8 / 2.5 / 8 / 25 / 80 mm / None
λf: 0.08 / 0.25 / 0.8 / 2.5 / 8.0 / 25 /Arbitrary
Sampling length0.025 / 0.08 / 0.25 / 0.8 / 2.5 / 8 / 25 / 80 mm*30.025 / 0.08 / 0.25 / 0.8 / 2.5 / 8.0 / 25.0 / 80.0
Number of intervals
1 - 20- 700 (max.)
0.02 - 50 mm
0.02 - 100 mm
0.025 - 50 mm
0.025 - 100 mm
Arbitrary length(0.01 mm interval)(0.01 mm interval)(0.001 mm interval)(0.001 mm interval)
18 languages (Japanese, English, German, French, Italian,
16 languages (Japanese, English, German, French, Italian, Spanish, Portuguese, Korean,
Display language
Spanish, Portuguese, Korean, Chinese (traditional / simplified),
Chinese (traditional / simplified), Czech, Polish, Hungarian,Turkish, Swedish, Dutch)
Czech, Polish, Hungarian,Turkish, Swedish, Dutch, Russian, Thai)
S4: 766×450×966 mm
Main unit
H4: 766×450×1166 mm
425×94×160 mm
716×450×863 mm
425×94×160 mm 716×450×863 mm
External
W4: 1166×450×1176 mm
dimensions
Display unit
— *4
330×270×124 mm— *4(W×D×H)Electronic unit—
372×245×71.8 mm——
PC I / F unit—
350×263×86 mm
S4: 140 kg
Main unit
2.7 kg
140 kg
H4: 150 kg
2.7 kg
140 kg
W4: 220 kg
Mass
Display unit
4.0 kg— *4— *4
Electronic unit—
3.0 kg——PC I / F unit—
3.8 kgModel No.

A superior data processing tester with PC data analysis
for higher efficiency.
Note 1: If a power column type with PC data-processing is required, consider the
FORMTRACER Avant S3000 Series (Refer to the Brochure <E15030>
for specifications.)
Note 2: Printer is optional.

SJ-500P
SV-2100M4 with PC

FORMTRACEPAK: Best-seller for surface roughness and contour analysis• Surface Roughness analysis function• Microscopic contour analysis function

FORMTRACEPAK can perform surface roughness analyses that conformto various standards such as ISO, JIS, ANSI, and VDA. For comparing themeasurement values with the tolerance limits, you can use the 16% rule or the
maximum value rule.
Furthermore, since FORMTRACEPAK comes with parameter calculation functions
as well as a rich set of graphic analysis functions, it can be widely utilized foreverything from routine quality control to R&D applications. It also includes
many other functions, such as the function for eliminating (compensating)shapes, such as slopes and R-surface, and a data deletion function.

This function can calculate steps and surface areas from the roughness data.
Furthermore, as with the contour analysis function, a rich set of calculationcommands that combine various elements, such as points, lines, and circles,
to calculate angles, pitches, and distances are provided as standard features.

• Reference length dialog box
When setting up thereference length in ameasurement condition,
you can display thestandard values definedby the ISO/JIS standards
by selecting the applicablestandard.

• Analysis condition modification with a preview function• Simple input usingdrawing symbols

You can easily set up cumbersome
measurement conditions by simply entering
data according to the drawing symbols of
the ISO/JIS roughness standard.

You can easily modify various types of analysis conditions such as thestandard to be used, curve type, and filter. Furthermore, before eliminating(compensating) shapes such as slopes, R-surfaces, and parabolas, the preview
function allows you to check the impact on the spot.

Example of drawing symbol
Grinding
Ra 1.5⊥ − 2.5 / Rzmax 6.7

Before compensation

Example of entering data
U”X”0.08—0.8/Rz8max 3.3

After compensation

8

SJ-500178-532-01
178-533-01
178-532-02
178-533-02
50 mm (2 in)

SV-2100M4
178-636-01
178-637-01
178-636-02
178-637-02*1 When using the simplified stand (optional): 0.4 μm/ 50 mm
When using the manual column stand (optional): 0.3 μm/ 50 mm*2 The simplified stand or manual column stand is available as an optional accessory. (Refer to page 11 for details.)*3 80 mm only for the SV-2100 Series.
*4 Varies depending on the PC system.

9External Dimensions

Optional Accessories

Unit: mmElectronic unit

Dedicated data processor

SJ-500

124

.7

.8

330
-

+

178-085 Except for measuring unit
Vertical adjustment range: 300 mm
Dimension (W×D×H): 600×450×710 mm
Mass: 110 kg

11.8
7

245

270

160

15
300

.5

BACK

DOWN

50

598

94

17.5

UP

198.7

710

.6

203

0
.1

.4

10 - 60

8

.9

.3

62.5

Dimensions of SJ-500 with manual column stand Unit: mm

Suitable for desktop use in inspection rooms, etc.

320

.2

81.8

Manual column stand: 178-085 (for SJ-500)

FORWARD

50

T-groove

Only for SV-2100S4 / H4/ W4

12

69.5

71.8

100

372

Measuring range

600

Note: While the appearance of the natural stone stand varies according to the source, the high stability
for which this material is known can always be relied upon.

Simple column stand: 178-089 (for SJ-500)

80

130
450

4

40
60

35

100

175

600

12

116
716

116

175

716

Note: While the appearance of the natural stone stand varies according to the source, the high stability
for which this material is known can always be relied upon.

12

350

Measuring range

400

130
450

DAT leveling table: 178-048( ): SV-2100H4

SV-2100W4

XY leveling tables 178-043-1/178-042-1

スThis table can be used by itself or in conjunction
with other leveling tables.

14

8

658

7

100

80

15

178-042-1
120

Inclination adjustment angle
Maximum load
Table dimensions

550

1054

T-groove dimensions(common to SV-2100 Series)

178-043-1
1176

±1.5°
15 kg
130×100 mm

Order No.

178-043-1 (mm)178-042-1 (mm)178-049 (mm)178-053-1 (in)178-052-1 (in)178-058 (in/mm)*with analog heads *with digital heads *with digital heads

Table dimensions

130×100 mm

Maximum load

Measuring range

35

110

85

12

100

11

T-groove

85

35

600

15
250

100

60.7 50350（550）751

258

854（1054）80966（1166）

100

Unit: mm8.5

178-089 Except for measuring unit
Vertical adjustment range: 250 mm
Dimension (W×D×H): 400×250×578.6 mm
Mass: 20 kg

85

12

863

258

Dimension of SJ-500 with simple column standA portable simple column stand.

506.6

SV-2100S4/SV-2100H4

578.6

SV-2100M4±1.5°

Swiveling angle±3°

X/Y-axis travel range
1000

116
1116

10

15 kg

Inclination adjustment angle

175

Resolution

130

Dimensions (W×D×H)

450

Mass

11

—
—
±12.5 mm0.01 mm0.001 mm

220×189×83 mm

262×233×83 mm

262×233×55 mm

6 kg

6.3 kg5 kg

250

Optional Accessories

Optional Accessories

System configuration including optional accessories (for SJ-500 with optional manual column stand)3-axis adjustment table: 178-047
This table helps make the alignment adjustments required when measuring cylindrical surfaces. The corrections for the pitch angle and the swivel
angle are determined from a preliminary measurement and the Digimatic micrometers are adjusted accordingly. A flat-surfaced workpiece can also
be leveled with this table.
Traverse
Aligned

Inclination adjustment angle ±1.5°
Swiveling angle±2°

Y-axis range±12.5 mm

Resolution of heads

0.001 mmTable dimensions

130×100 mm

Maximum load

Traverse
direction

Not aligned

12AAG202: Extension rod (50 mm)

direction

End point

End point

Axial line

Axial line

Start point

Start point

178-397-2
Detector (4 mN)
178-396-2
Detector (0.75 mN)

Printer

12AAJ112: SPC cable

12AAG203: Extension rod (100 mm)
Data processing *3
system (PC type)

12AAA841: Memory card
*2

12AAJ111: RS-232C cable

15 kg

178-047:
3-axis adjustment table

Recorded profiles

218-001 (mm), 218-011 (in):Cross-travel table

SJ-500 with manualcolumn stand218-002:
Table*1

12AAG202: Extension rod (50 mm)

12AAJ112: SPC cable*3
Data processing
system (PC type)178-023:
Vibration isolator(Pump supplied)

178-025:
Vibration isolator(Air supplied)

Table

Printer

12AAG203: Extension rod (100 mm)

Dedicated data processor

12AAA876:
Printer paper (5 rolls)

System configuration including optional accessories (for SV-2100M4 / S4 / H4 / W4)

178-397-2
Detector (4 mN)
178-396-2
Detector (0.75 mN)

264-505:*4
DP-1VA

264-505:*4
DP-1VA

178-016:
Leveling
table

178-043-1:

Leveling table(with analog heads)

178-042-1:

Leveling table(with digital heads)

178-048:
DAT levelingtable

12AAG175:
Calibration stand

12AAQ587:
900×750×740 mm(Max. load 300 kg)

12AAQ583:
1500×900×740 mm(Max. load 800 kg)

12AAA841: Memory card
*2

12AAJ111: RS-232C cable
Dedicated data processor

178-047:
3-axis adjustment table218-001 (mm), 218-011 (in):Cross-travel table

998291:
V-block

218-002:
Table

178-019:
Precision vise

218-003:
Rotary vise172-144:
Rotary vise

12AAA876:
Printer paper (5 rolls)*1178-016:
Leveling
table998291:
V-block

178-043-1:
Leveling table(with analog heads)

178-019:
Precision vise

218-003:
Rotary vise

178-042-1:
Leveling table(with digital heads)

172-144:
Rotary vise

178-048:
DAT levelingtable

172-142:Center support

172-143:
Center support riser

178-023:
Vibration isolator(Pump supplied)

178-025:
178-115:Vibration isolator Vibration isolator for W4(Air supplied)(Desktop air supplied)

176-107:
Holderwith clamp

176-107:
Holderwith clamp

172-234:
V-blockwith clamp

172-378:
V-blockwith clamp

172-196:
Rotary table*1 7 0 mm stage to be mounted onto the base required for calibration of roughness specimen.
Not required when using cross-travel table, plain stage, cross-travel stage, or 3-axis adjustment
table.
*2 PC for managing the analysis result externally output from the dedicated data processor.
*3 Only SJ-500P can be connected. Use a USB cable when connecting the SJ-500P main unit anda PC. A USB cable is a standard accessory of the SJ-500P.
*4 Since print of unit does not support 'μm', use DP-1VA without print set of unit.

172-143:
Center support riser

178-119:
Vibration isolator for W4(Air supplied)

12AAQ587:
900×750×740 mm(Max. load 300 kg)

172-234:
V-blockwith clamp

172-378:
V-blockwith clamp

12AAQ583:
1500×900×740 mm(Max. load 800 kg)

Roughness specimen (standard accessory): 178-601

Reference step specimen: 178-611 (mm), 178-612 (in)Display

Ra=about 3 μm

For sensitivity calibration of detector

Material

Ni (Cr surface coating)

Nominal value of step

172-196:
Rotary table*1 7 0 mm stage to be mounted onto the base required for calibration of roughness specimen.
Not required when using cross-travel table, plain stage, cross-travel stage, or 3-axis adjustment table.
*2 PC for managing the analysis result externally output from the dedicated data processor.
*3 Only SV-2100M4 can be connected.
*4 Since print of unit does not support 'μm', use DP-1VA without print set of unit.

12172-197:
Swivel center
supportTable

12AAG175:
Calibration stand172-197:
Swivel center
support172-142:Center support

Roughness specimen: 178-604
Display

Ra=about 3 μm, about 0.4 μm

Note: R a=Approx. 0.4 μm should only be used for stylus tip
checking.

13

2 μm (79 μin), 10 μm (394 μin)

Optional Detectors, Extensions and Styli

φ8

0.9

ø1.2
60°

*1 Tip angle 90°*2 For downward-facing measurement only.

*3

Tip radius
1 µm
2 µm
5 µm
10 µm
250 µmColor coding
White
Black
No color
Yellow
No notch or color*4 Used for calibration, a "Reference step specimen" (178-611 or 178-612, optional) is also required

14

Color
coding*3

ø1.2
60°

43.8
42.6
5.2

ø3
ø3

7

94.7
87.7

36.5
35

7.6
6.4
5.2

ø1.2

For corner hole/Double-length for deep hole*2
93.8
87.7

12AAM601 (2 µm)
12AAM603 (5 µm)( ): Tip radius( ): Tip radius

12AAC739 (2 µm)
12AAB412 (5 µm)*1
12AAB424 (10 µm)*1

12AAE886 (250 µm)

ø2.4

0.9

ø2.4

60°

ø2.4

12AAC738 (2 µm)
12AAB411 (5 µm)*1
12AAB423 (10 µm)*1

45.2
37.7

0.9
A

For rolling circle waviness/Double-length for deep hole*2*4

ø1.

2

For hole bottom
44.3
37.7

0.5

12AAE899 (2 µm)
12AAE915 (5 µm)*1( ): Tip radius( ): Tip radius

5

10

7.6

ø2.4

ø1.2

For eccentric arm*2

12AAB338 (φ1.588)

7.6
6.4

ø0.3

12AAE938 (2 µm)
12AAE940 (5 µm)*1

細長穴用( ): Tip radius

36.5
35

ø1.2
60°

Detail-A
A

5.2

33
31.8
5.2
A

44.7
37.7

0.9

( ): Tip radius

ø1.2

ø1

( ): Tip radius

ø0.6

Detail-A

3X stylus
12AAC741 (2 µm)
12AAB414 (5 µm)*1
12AAB426 (10 µm)*1

Color
coding*3

94.4
87.7

0.6

1.6

7.6
5.2
Detail-A

2X stylus
12AAC740 (2 µm)
12AAB413 (5 µm)*1
12AAB425 (10 µm)*1

60°

6.4

For knife-edge

45
0.4

60°

ø1.2

0.6

ø2.4

ø1.2

Color 0.9
coding*3

ø2.4

7.6
5.2

A

Ball ø1.588

12AAE896 (2 µm)
12AAE912 (5 µm)

60°Color 0.9
coding*3

144.7
137.7
A

7.6

ø2.4

37.7
43.8

For small slotted hole*2
94.7
87.7

0.9

ø1.2

93.8
87.7

6.7

ø2.4

2.4

ø1.2

8.5
A

Color
coding*3

For deep hole*2

Detail-A

24.2
23
Detail-A

For gear tooth/Double-length for deep hole*2

7.4
6.2

A

ø0.6

0.8

60°
Detail-A（S＝5/1）( ): Tip radius

12AAJ662 (φ0.5 mm)
細穴形状用

7.6
5.2

2.5

ø0.3

Ball ø0.5

44.7
37.7

7.6

8.9

12AAC734 (2 µm)
12AAB406 (5 µm)*1
12AAB418 (10 µm)*1( ): Tip radius( ): Tip radiusA
60° ø1.2

ø2.4

0.4

12AAB339 (2 µm)
12AAB410 (5 µm)
12AAB422 (10 µm)

43.8
37.7

Color
coding*3

12AAE894 (2 µm)
12AAE910 (5 µm)*1

ø1.2

For rolling circle waviness surface*4

For ultra-small hole*4

93.8
87.7

( ): Tip radius

歯面用

ø2.4

44.2
37.7

Ball ø1.6

For deep groove (30 mm)/Double-length for deep hole*2

ø1.2

Detail-A

ø1.2

2.4

For extra-minute hole
0.8

ø2.4

2.5

1.2
A

A

12AAC737 (2 µm)
12AAB407 (5 µm)*1
12AAB419 (10 µm)*1

60°

12AAE884 (φ1.6 mm)

93.8
87.7
41

( ): Tip radius

Detail-A

37.7
45.2

For gear tooth

For small hole*2*4

Color
coding*3

ø0.6

1.2

60°
Detail-A（S＝5/1）( ): Tip radius

ø2.4
ø1.2

For deep groove*2 (30 mm)

Color
coding*3

12AAE895 (2 µm)
12AAE911 (5 µm)*1( ): Tip radius

60°

8.9

ø0.3

Detail-A

12AAC733 (2 µm)
12AAB405 (5 µm)*1
12AAB417 (10 µm)*1

44.2
37.7

0.4

30

A

A
ø1.2

60°

For extra-small hole

ø2.4

A

12AAE908 (5 µm)*

87.7

0.6

ø0.6

( ): Tip radiusColor
coding*3

ø1.2

Detail-A（S＝5/1）

ø2.4

15
3.4
1.6

1.6

60°

12AAC732 (2 µm)
12AAB404 (5 µm)*1
12AAB416 (10 µm)*1

44.4
37.7

0.6

3.4
1.6

0.4

0.6

45.2
For deep groove*2
37.7
深溝用2倍（20mm）(40 mm)( ): Tip radius

ø2.4

For small hole/Double-length for deep hole*2
0.6
12AAE892 (2 µm)
94.4深穴2倍用
112AAC736 (2 µm)
12AAB408 (5 µm)*1
12AAB420 (10 µm)*1

60°

Detail-Aø2.4( ): Tip radius( ): Tip radiusColor
coding*3

3.1

60°

For small hole
ø0.6

ø2.4

A

Detail-A

12AAE898 (2 µm)
12AAE914 (5 µm)*1

94.7
87.7

0.9

ø0.3

5.2
7.6

6.4

ø1.2

Double-length for deep hole*2

3
1.8
0.6

44.7

37.7
ø3.2

0.9
Color
coding*3

12AAE882 (1 µm)
12AAE924 (1 µm)*1
12AAC731 (2 µm)
12AAB403 (5 µm)*1
12AAB415 (10 µm)*1
12AAE883 (250 µm)*4

44.7
37.7

ø2.4

For deep groove*2 (20 mm)

Styli
Standard stylus

ø1.2

ø2.4

Note: No more than one extension rod can be connected.

ø2.4

ø2.4

Detail-A

Detectors that comply with previous standards,
for general use, etc.

4 mN( ): Tip radius( ): Tip radius

ø2.4

178-397-2

12AAE893 (2 µm)
12AAE909 (5 µm)*1

87.7
5.2

• 12AAG203 Extension rod 100 mm

Order No. Measuring force
ʼ 97ISO and ʼ 01JIS compliant detectors
178-396-2
0.75 mN

95.2

23
21.8

Color
coding*3

For deep groove*2 (20 mm)

ø2.4

Detector

Skidless nosepiece（12AAB355）

4

14.2
13

φ7

12AAC735 (2 µm)
12AAB409 (5 µm)*1
12AAB421 (10 µm)*1

44.7
37.7
A

ø2.4

φ14

14

φ

1.3
3.6

For deep groove (10 mm)• 12AAG202 Extension rod 50 mm

14

60

3.1

Styli

ø2.4

11.5 10Extension rods

ø2.4

Detectors

Detail-A*1 Tip angle 90°*2 For downward-facing measurement only.
Note: Customized special interchageable styli are available on request,
Please contact any Mitutoyo office for more information.

*3

Tip radius
2 µm
5 µm
10 µm
Color codingBlack
No color
Yellow*4 Used for calibration, a "Reference step specimen" (178-611 or 178-612, optional) is also required15

155 2202(5)g-(CH)NE-(JP8),Printed in Japan

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law.
Please consult us in advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

Mitutoyo Corporation

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp