Time Setting
Range

Display

Safety Devices

Chamber

Main Body

FLS-1000

SX-700SX-500

SX-300

SX-700BH

SX-500BH

105 to 135℃ *1

120V model
230V model(220 to 240V)

105 to 135℃ *1

45 to 104℃

230V model(220 to 240V)

105 to 123℃ *1
45 to 104℃

120V model
230V model (220 to 240V)

Sterilizing-

45 to 95℃ *1

230V model(220 to 240V)

Heating

45 to 95℃ *1

240V model

ASX-500

Heating

Sterilizing--

0:01 to 9:59 / 1 to 99 hours / 1 to 999 minutes /0:01 to 9:59, 10 to 99 hours (Selectable)

1 to 99 hours

0:01 to 9:59 / 1 to 99 hours / 1 to 999 minutes /0:01 to 9:59, 10 to 99 hours (Selectable)

1 to 99 hours

0:01 to 9:59 / 1 to 99 hours /1 to 999 minutes /0:01 to 9:59, 10 to 99 hours (Selectable)

Digital0 to 0.4 MPa

Warming

Digital0 to 0.4 MPa

DigitalPressure Gauge

DigitalΦ 370 × 774

58 LΦ 325 × 733

44 L

Φ 325 × 553

79 L

Φ 370 × 774

58 LΦ 325 × 733

TimeΦ 450 × 678

79 L

Safety Valve
Lid Interlock
Leakage Breaker
No Water Heating Prevention
Over-temperature Prevention
Over-pressure Prevention
Temperature Sensor Malfunction Detection
Pressure Sensor Malfunction Detection
Exhaust Valve Malfunction Detection
Low Sterilizing Water Level Detection
Chamber Lid Open/Close Detection
Exhaust Bottle Detection

100 L

Safety Valve
Lid Interlock
Leakage Breaker
No Water Heating Prevention
Over-temperature Prevention
Over-pressure Prevention
Temperature Sensor Malfunction Detection
Pressure Sensor Malfunction Detection
Exhaust Valve Malfunction Detection
Low Sterilizing Water Level Detection
Chamber Lid Open/Close Detection
Exhaust Bottle Detection

Dimensions(mm)

SUS304 (Stainless Steel)

SUS304 (Stainless Steel)

Capacity

72 kg

AC120V 50/60Hz
20A 1P
AC230V 50/60Hz
15A 1P

60 kg

AC120V 50/60Hz
15A 1P
AC230V 50/60Hz
10A 1P

50 kg

AC230V 50/60Hz
15A 1P

72 kg(Bio Box: 5 kg)

Stainless Steel Basket( Φ 300 × 181 mm)
2 pcs

AC120V 50/60Hz
20A 1P
AC230V 50/60Hz
15A 1P

60 kg(Bio Box: 5 kg)

470W × 528D × 1,003H 410W × 477D × 970H
410W × 477D × 790H (BioBox:Φ215 × 525 H) (Bio Box:Φ215 × 525 H)

AC230V 50/60Hz
15A 1P

Stainless Steel Basket( Φ 345 × 181 mm)
2 pcs-

Stainless Steel Basket( Φ 300 × 181 mm)
1 pc

Manufacturer:-

Stainless Steel Basket( Φ 300 × 181 mm)
2 pcs

Sales Office:3-14-17 Tagara, Nerima-ku, Tokyo 179-0073, Japan

Digital0 to 0.4 MPa

AnalogDigital--0:01 to 9:59

0.263 Mpa--

105 to 132℃

220 to 240V model

ASX-500

Secure Sterilization & Time-saving
Φ 325 × 733
58 L
SUS304 (Stainless Steel)

410W × 477D × 970H

60 kg
AC220 to 240V 50/60Hz
15A 1P-

0.6K05024②R

All TOMY products have a limited one-year warranty.Specifications are subject to change according to product advancement.
Tomy and Digital Biology is registered trademark of Tomy Seiko Co.,Ltd. and Tomy Digital Biology Co., Ltd.
Copyright 2021,Tomy Seiko and its subsidiaries. Printed in Japan.

Stainless Steel Basket( Φ 345 × 181 mm)
2 pcs

470W × 528D × 1,003H 410W × 477D × 970H

SUS304 (Stainless Steel)Material

AC240V 50/60Hz
15A 1P

600W × 600D × 940H

AC230V 50/60Hz
15A 1P

Dimensions(mm)(Excluding
protruding)

AC220V 50/60Hz
15A 1P

108 kgSafety Valve
Lid Interlock
Leakage Breaker
No Water Heating Prevention
Over-temperature Prevention
Over-pressure Prevention
Temperature Sensor Malfunction Detection
Pressure Sensor Malfunction Detection
Exhaust Valve Malfunction Detection
Low Sterilizing Water Level Detection
Chamber Lid Open/Close Detection
Drain Access Cover Open Detection
Exhaust Tank Full Level Detection

0 to 0.4 MPa

Analog0.263 MPa-

230V model

SX-700BH/500BH

Digital0.250 MPa

Warming

220V model

SX-700/500/300

Analog0.263 MPa

FLS-1000

Digital0.250 MPa

Easy Fast
Autoclave
Analog0.193 MPa

Biohazardous Waste
Autoclave
Safety Valve
Lid Interlock
Leakage Breaker
No Water Heating Prevention
Over-temperature Prevention
Over-pressure Prevention
Temperature Sensor Malfunction Detection
Pressure Switch Malfunction Detection
Exhaust Valve Malfunction Detection
Low Sterilizing Water Level Detection
Chamber Lid Open/Close Detection

Multiple Shared Use
Autoclave
Net Weight

Power Requirements

Stainless Steel Basket( Φ 415 × 233 mm)
2 pcs
IEC 61010-2-040, IEC61326-1

3-14-17 Tagara, Nerima-ku, Tokyo 179-0073, Japane-mail: info@digital-biology.co.jp
URL: https://www.digital-biology.co.jp/manufactured/
phone: +81-3-5971-8160 fax: +81-3-3970-6036*1. Max. operating temperature will be changed for the high altitude sterilization program and it depends on the altitude level.

Compliance of Standard

Standard Basket

High-throughput
Autoclave
Temperature

Maximum Operating Pressure

OperatingTemperature
Range

Model

Specifications

AUTOCLAVESSecure Sterilization & Time-saving
Autoclave is essential for almost any type of laboratory, andSecure Sterilization and Time-saving are the high priority towith innovative features as standard.

it is crucial to choose the best autoclave that suits your needs.
consider for researchers, and TOMY Autoclaves are designed

Secure Sterilization

Time-saving

STD Fine Pressure Adjustment

STD Cooling Fan System

Secure Sterilization condition is created by saturated steam that temperature andpressure keep the balance for a certain time.
TOMY Autoclaves heat water in the chamber and create steam. The generated steam isdischarged with air from the chamber to approach the saturated steam condition.
When air remains in the chamber, the balance of temperature and pressure breaks and it
makes pressure rise.
Fine Pressure Adjustment monitors temperature and pressure, works automatically toremove residual air from the chamber, and adjusts pressure to approach ideal
condition for more effective sterilization.

TOMY Autoclaves are standard-equipped with a dual-fan cooling system, shortening
overall operation time. The Cooling Fan System reduces cooling time by 60 min.*1
compared to non-fan operation.It will save your time and give you more to focus onother things.

Standard

Residual air may breakthe ideal balance.

Standard

timeefficient

Fine Pressure Adjustment
to approach ideal condition

Dual-fanRemoveresidual air*1 In the case of FLS-1000 under the test condition of no load operation, room temperature 25°C, sterilizing water 6 liters and power input AC200V 18A.

1

2

To Choose the Best Autoclave
Choosing the best autoclave largely depends on several factors,
your throughput needs (how large your articles to be sterilized are andas well as convenience and personal preference. TOMY Autoclave line-

High-throughput Autoclave

including articles to be sterilized (what you are autoclaving),
how much you need to autoclave) and available laboratory space,up will help you choose the right autoclave for your sterilizing needs.

Biohazardous Waste Autoclave

FLS-1000

SX-BH series⃝ Large volume of liquid sterilization
like water, media or solvent⃝ Bioreactor and fermentersterilization⃝ Animal cage sterilization
Page ➡ 5⃝ Next to biosafety cabinet⃝ Immediate sterilization after bench-work⃝ Decontamination before
disposal of biowasteHEPA

Bio Safety Cabinet

Page ➡ 9

6

Multiple Shared Use Autoclave

10

Easy Fast Autoclave

SX series

ASX-500
Share⃝ For central autoclave room and washing room
⃝ For shared use among some laboratories⃝ Both purpose for preparation of labwares anddecontamination of labwastes
Page ➡ 7

3⃝ Entry model⃝ Simple operation⃝ Dedicated to sterilize solid articles like labwares orlabwastes
8

Page ➡ 11 124More Secure Sterilizationconsidering Heat Transfer Time Lag

High-throughput Autoclave

FLS-1000

With an improved capacity and user-friendly design, the TOMY Autoclave FLS-1000 makes large
volume sterilization easier and accommodates unique shaped articles like bioreactors, fermenters andanimal cages. The FLS-1000 also supports you for secure sterilization in compliance with your facilities’
guidelines and global standards like GLP/GMP.

There is Heat Transfer Time Lag when using the steam
autoclave. The display shows the ambient temperatureinside the chamber, but the temperature of articles tobe sterilized arrives later than the display temperature.
This is called “Heat Transfer Time Lag“.
For more secure sterilization, it is recommended to add
the Heat Transfer Time Lag to the sterilization time.

Heat Transfer Time Lag
121℃

Article Temperature
Chamber Temperature

Required sterilization time + Heat transfer time lag= Setting timee.g. 500ml × 2 bottles at 121°C and 20min.
➡ 20min. + 12min. = 32 min.

OP Automatically timer countdown by sample temperature

Large Capacity

100ℓ

Optionw/External Temperature Sensor

The TOMY Autoclave FLS-1000 can check the temperature of articles to be sterilized with an optional External
Temperature Sensor. The detected temperature by the External Temperature Sensor can be used to count down
the sterilization time (e.g. 20 min.) . That means sterilization will be done with the temperature of articles inside
the chamber automatically without calculating Heat Transfer Time Lag manually by yourself.
The detected temperature also works to open the lid
safely after sterilization. The lid can be opened after the
121℃
temperature of articles drops down*1 and it prevents the20
121℃
sudden boiling of liquid sample.
minutes
These features help you to save your time to decide the
START!
20
sterilization time and to reduce troubles of spillage forminutes
media or solvent.

Wide Chamber

45㎝
45cm Wide Chamber

START!

*1 The temperature to release the lid interlock can be changed.

High-throughput Liquid Sterilization

Specifications

Do you have any experience that flasks and bottles can not be accommodated as desired when you sterilize alarge amount of liquid such as culture media and solvents? The FLS-1000 has a large chamber dimension of
45cm so that it can accommodate 5L × 4 flasks easier with a option flask stand and 500ml × 17 bottles with astandard stainless basket.

Model
Operating Temperature Range
Sterilization Course
Setting Time Range
Chamber Capacity and Size
Size and WeightStandard Accessory

FLS-1000
105 to 123℃ *2
Solid Sterilizing (Sterilize/Normal)
Liquid Sterilizing (Sterilize/Liquid)
0:01 to 99 hours
100 L
Φ 450 × 678 mm
600 W × 600 D × 940 H mm
108 kgStainless Steel Basket ACA-1000B × 2 pcs*2 Operating temperature range will be changed after selected the high altitude program.

5L×4 flasks

Full Package for High-throughput Autoclave FLS-1000
High-throughput Autoclave

FLS-1000

OP
Optionflask stand5

OP
OptionFlask StandOP

for 5L × 4 flasks

OptionExternal Temperature
Sensor

500ml×17 bottles/ Basket×1 unit×1 set×1 set6

OP PrinterMultiple Shared Use Autoclave

OptionSX series

The TOMY Multiple Shared Use Autoclaves SX series are designed to meet the sterilization needs fora wide variety of sizes and shapes of articles to be sterilized especially in the washing room or central
autoclave room arranged for shared use among a few or several laboratories.

supporting to keep Sterilization Log

Do you have any experience that you were worried if the sterilization was performed as set
when you use a shared autoclave in a remote wash room or an autoclave room?
The Optional Printer can print out the operating parameters, chamber temperature,chamber pressure, temperature of articles to be sterilized*2 and sterilization cycle process.
*2 In using the optional external temperature sensor

OP High Altitude Sterilization Program
OptionAutoclave operates by generating steam underpressure and temperature. Therefore, its sterilization
process can be greatly affected by the atmosphericpressure at the location where the autoclave isinstalled.
The SX series Autoclaves allow users to operate eventhough at high altitude place*3.

4,000m*3*3 Available atmospheric pressure is until 616 hPa or higher (equivalent to an altitude of 4,000m).
Operating temperature range will be changed after installed the high altitude program.

Specifications

Ergonomic Height Design
The SX series Autoclaves are designed to be theperfect height to easily load and unload articles into
the chamber.
Operatorworking
table

82.7cm*1*1 For SX-700

7

155cm

Model
Operating Temperature Range

Attentive Liquid
Sterilization Course

SX-700

Sterilization Course

When autoclaving liquid samples, opening the lid
while the liquid temperature is high may cause asudden boiling. It is necessary to allow the pressure
and temperature to drop naturally without forced
exhaust after sterilization.
The SX series Autoclaves have a LIQUID course that
automatically sterilizes like this process with a single
operation. It reduces spillage of media or solvent
inside the chamber and supports smooth shared use
in your facilities.
Prevent
suddenboiling

Setting Time Range for Sterilizing
Chamber Capacity and Size
Size and WeightStandard Accessory

SX-500
SX-3004
105 to 135℃ *Solid Sterilizing (Sterilize/Normal)
Liquid Sterilizing (Sterilize/Liquid)
Sterilizing and Warming (Sterilize - Warm)
Heating and Warming (Heat - Warm)
0:01 to 99 hours
79 L
58 L
44 L
Φ 370 × 774 mm
Φ 325 × 733 mm
Φ 325 × 553 mm
470 W × 528 D ×
410 W × 477 D ×
410 W × 477 D ×
1,003 H mm
970 H mm
790 H mm
72 kg
60 kg
50 kg
Stainless Steel Basket
Stainless Steel Basket
Stainless Steel Basket
ACA-700B × 2 pcs
ACA-315B × 2 pcs
ACA-315B × 1 pc

*4 Operating temperature range will be changed after installed the high altitude program.

Full Package for Multiple Shared Use Autoclave SX-700
Multiple Shared Use

Autoclave SX-700

×1 unit

OP Optional PrinterOption(Thermal Paper Type)×1 setOP External Temperature
OptionSensor

×1 setProtector Kit for

OP External Temperature
OptionSensor

×1 set8

Biohazardous Waste Autoclave

SX-BH series

Numerous biosafety guidelines endorse the use of autoclaves to inactivate potentially infectious
laboratory waste prior to disposal, as part of creating secondary barriers that prevent infectious
disease from spreading to laboratory workers, waste management personnel and the general public.
The TOMY SX-BH series are designed to exhaust air and steam coming from the chamber through
HEPA filter.

OP Biowaste Sterilization Container ST-ZERO
OptionHow do you handle disposables generating from the bench work of the biosafety cabinet including PCR test
kits, gloves, pipette tips and petri dishes containing specimens?
Biowaste Sterilization Container ST-ZERO can sit on the workbench of the biosafety cabinet without disturbing
air-flow and accommodate disposables inside the work area of the biosafety cabinet.
ST-ZERO can be vertically loaded to the Autoclave without a liquid drop falling. A several ST-ZERO can beaccommodated in the chamber of Autoclave SX-700BH.
Put disposals and close the lid of ST-ZERO

Load to Autoclave vertically.

inside the Bio Safety Cabinet.

HEPA
ST-ZERO

Bio Safety Cabinet

Specifications
Model
Operating Temperature Range

SX-700BH

SX-500BH
105 to 135℃ *2
Solid Sterilizing (Sterilize/Normal)
Liquid Sterilizing (Sterilize/Liquid)
Sterilizing and Warming (Sterilize - Warm)
Heating and Warming (Heat - Warm)
0:01 to 99 hours
79 L
58 L
Φ 370 × 774 mm
Φ 325 × 733 mm
470 W × 528 D × 1,003 H mm
410 W × 477 D × 970 H mm
72 kg
60 kg
Φ 215 × 525 H mm
5 kg
HEPA Filter 0.3 μm DOP 99.97 %Stainless Steel Basket
Stainless Steel Basket
ACA-700B × 2 pcs
ACA-315B × 2 pcs

Sterilization Course
Setting Time Range for Sterilizing
Chamber Capacity and Size

Air and steam exhaust through
HEPA Filter (DOP 99.97%)*1
Air and steam inside the chamber go to the coolingwater tank.
After cooling down, they are exhausted to thelaboratory work area through HEPA filter unit that is
equipped with a HEPA filter (DOP 99.97%).

Size andWeight

Autoclave Mainbody
Bio Box

Bio Box Exhaust Filter

HEPA

Standard Accessory*2 Operating temperature range will be changed after installed the high altitude program.

Full Package for Biohazardous Waste Autoclave SX-700BH
Autoclave SX-700BH(including Bio Box)

*1 It does not guarantee to remove contaminants completely.

Option×1 setFollow the biosafety guideline in your laboratory.

OP

Biowaste Sterilization Container

ST-ZERO×1 setCooling Water Tank

9

10

Power Failure Alert Function

Easy Fast Autoclave

Pure water, DI water andRO water available

Operation can be continued from the process when
the power failure occurs. It saves waiting time andre-sterilization work.

ASX-500

Pure water, DI water and RO water are available assterilization water to prevent contamination for lab
wares preparation.

OK!

The TOMY Easy Fast Autoclave ASX-500 is a dedicated model for a quick operation to sterilize solid
articles like flasks, bottles, petri dishes, pipet tips, lab glass & plastic wares and so on.
Easy operation just by setting temperature and time, provides fast sterilization to save your time.
The ASX-500 is a entry model but also has a wide variety of optional accessories to reduce troublesome
work and to shorten overall operation time. Each laboratory can choose the suitable accessories forthe articles to be sterilized.

Specifications
Model
Operating Temperature Range
Sterilization Course
Setting Time Range

ASX-500
105 to 132℃
Solid Sterilizing (Sterilize/Normal)
0:01 to 9:59 hours
58 L
325
× 733 mm
Φ410 W × 477 D × 970 H mm
60 kg

Chamber Capacity and Size
Size and Weight

Full Package for Easy Fast Autoclave ASX-500
A : Separator Kit Package
Easy Fast Autoclave

ASX-500

OP Separator Kit
OptionOption×1 unit

ASX-500

×1 set

B : Waste-bin Kit Package

Melted petri dishes remainin the long basket after
the sterilization.

Easy Fast Autoclave

ASX-500

OP Long Basket
OptionLong Basket
ACA-325TP

The ager melts andfalls down into thebucket.

ACA-325TP

×1 unitNaturally separated

MeltedPetri DishPut petri dishes with the medium remained directly
into the long basket of Separator Kit. They arenaturally separated into petri dishes and liquid aftersterilization.
11

for Disposal Petri Dish

OptionOP Waste-bin Kit

for Disposal Petri Dish

Melted Ager

Separator Kit

OP

Sterilization Bag
DAB-321

×1 set×1 setC : Stainless Steel Basket Package
Easy Fast Autoclave

ASX-500

Use the Long basket as a waste-bin with thesterilization bag.
Put a Sterilization Bag into the Long Basket anddirectly it into the chamber. Close the bag anddispose it after sterilization.

Waste-bin Kit
Lid for Long Basket OP Sterilization Bag
OP
OptionOption300TL
DAB-321

OP Stainless Steel Basket
Option×1 unit

ACA-315B

×2 set12

Model Selection Guide
FLS-1000

SX-700SX-500

SX-300

SX-700BH

SX-500BH

ASX-500

Fine Pressure Adjustment

●

●

●

●

●

●

●

Cooling Fan

●

●

●

●

●

●

●

●
(78.5 cm)●
(82.7 cm)●
(79.4 cm)●
(61.4 cm)●
(82.7 cm)●
(79.4 cm)●
(79.4 cm)

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

●

Model

Ergonomic Height Design(Height to working table)

Sterilization Course

OP High Altitude Sterilization Program*1

Sterilization - Normal(Labware and Waste)
Sterilization - Liquid(Media, Solvent and Water)
Sterilization & Warming(Sterilizing and keep warming media)
Warming(Melting and keep warming media)
until 616 hPa or higher(equivalent to an altitude of 4,000m)
until 750 hPa or higher(equivalent to an altitude of 2,000m)●(Standard)

Exhaust through HEPA filter (Bio Box)OP External Temperature Sensor

●

●

●

●

●

●
(Dot Matrix)●(Thermal)●(Thermal)●(Thermal)●(Thermal)●(Thermal)

●

●

●

●

●

●

OP Protector Kit for External Temperature Sensor
OP Printer●Biowaste Sterilization Container

OP ST-ZERO

OP Separator Kit for Disposal Petri Dish

Separator Kit 500

OP Flask Stand for 5L x 4 flasks

ACA-1100B

ACA-700B

ACA-315B

ACA-315B

ACA-700B

ACA-315B

ACA-315B

Standard equipped number per unit

2

2

2

1

2

2

0

Max. loaded number per unit

2

3

3

2

3

3

3

Φ 415 × 233 H mm

Φ 345 × 181 H mm

Φ 300 × 181 H mm

Φ 300 × 181 H mm

Φ 345 × 181 H mm

Φ 300 × 181 H mm

Φ 300 × 181 H mm

17

11

8

8

10

8

8

10

7

5

5

7

5

5

Size of basket
Max. loaded number of 500ml bottle perbasket*2
Max. loaded number of 500ml flask perbasket*2

OP Long Basket

Model

ACA-1000TP

ACA-710TP

ACA-325TP

ACA-310TP

ACA-710TP

ACA-325TP

ACA-325TP

OP Lid for Long Basket

Model

1000TL

700TL

300TL

300TL

700TL

300TL

300TL

OP Sterilization Bag

Model

DAB-700

DAB-321

DAB-321

DAB-700

DAB-321

DAB-321

OP Bucket

Model

ACA-700KS

ACA-315KS

ACA-315KS

ACA-700KS

ACA-315KS

ACA-315KS*1 Operating temperature range will be changed after installed/selected the high altitude program.
*2 Visit our website for more detailed information.

13

Separator Kit 500

●
Model

Basket●

ACA-1000KS

Please visit for more details

TOMY AUTOCLAVESSEARCH

14