Catalog No. E16024(2)Operable Adjacent to a Machine Tool on the Production Line

Coordinate Measuring Machines

Highly Environment-resistant Coordinate Measuring Machine,
Allowing On-site Inspection outside the Measuring Room
• Has achieved an accuracy guaranteed temperature range of 10 to 40ºC thanks to the combination of technologies such as a symmetricstructure, uniform material and temperature compensation.
• Has achieved a contamination resistance more than 2 times better than Mitutoyo's conventional machine by adopting a newly
developed absolute scale* that is highly resistant to the challenging production-line environment.
*：A scale scale that provides an absolute value for each measurement point. This eliminates the need for an initialization operation.

Scale signal magnitude• Adopts the horizontal-arm structure and installs the CMM controller and PC under the measuring table to save installation space. This
has reduced the footprint to about 70% compared with that of the conventional moving bridge model.

MiSTAR scaleThis scale will not deteriorate evenwith more than twice thecontamination as on a regular scale.

MiSTAR scale
Regular scaleError occurrence on a regular scale
Threshold value of error detection

Contamination value

Note: These results were obtained using
Mitutoyo's in-house testing methods.

User-friendly Design Reduces Operator's Workload• The adoption of the horizontal-arm structure provides a three-sided open architecture, significantly easing the task of moving
workpieces on and off the measuring table.
• Equipped as standard with Quick Launcher that enables simple and intuitive operation.
• The use of clamping tools and pallet receiver enables quick measurement of similar
workpieces.
Note: Clamping tools and the pallet receiver are optional accessories.

Clamping tools (option)

Quick Launcher

2Pallet receiver (option)

New Functionality has the "Smart Factory" in View
• From status control to preventive maintenance. Smart Factory starts with "visualization".
• MiSTAR 555 is compatible with the following three applications that consolidate information management of the manufacturing process
through a network.
Status Monitor:		 Allows remote monitoring of the operational statusof measuring instruments.
1
Condition Monitor* : Allows remote monitoring of the current conditionof measuring instruments.
MeasurLink*1:		 Reduces the production of defective parts through"Visualizing Quality".		 *1: Optional

StatusStatus Monitor
MonitorOperationalstatus of measuringmachine

稼働率の向上Improve Operating Rate

MeasurLink

Real-timeacquisition/analysis ofmeasurement
dataData

Data visualization

SMS (Smart Measuring System)
This system allows on-line monitoring of theoperational status of a measuring machineand visualization of measurement data
produced during the manufacturing process
to enable product quality improvement.

Improvement inaccuracy/quality

Condition Monitor

Condition
MonitorCondition ofmeasuringmachineSPECIFICATIONSMiSTAR 555
XY
Measuring range
ZMax. Permissible Length Measurement Error (ISO 10360-2: 2009)
Drive speed
Drive acceleration
Maximum workpiece height
Measuring table capacity
Maximum loading
Accuracy guaranteed temperature
Mass (including the control device and installation platform)*2: Specifications vary by configuration and thermal environment.

570 mm
500 mm
500 mm
(2.2+3L/1000) µm*2 L: Measured length (unit: mm)
5 - 350 mm/s (max. combined speed 606 mm/s) in CNC MODE
1556 mm/s2 (max. combined acceleration 2695 mm/s2)
660 mm120 kg
10 ºC - 40 ºC
655 kgDIMENSIONS (unit: mm)
1188.31122

76

Z:5006841025

110

Y:500248

169

X:570 95.5

Equipped with the PH6M probe head
800885Equipped with the PH10MQ probe head

102.5

12589.5

Measuring
range

2350

1121

720F.L.

3

052 1906(1)e-(CH)PR, Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed upby comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment
and Seismometers

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver bespoke measuringsolutions and even, if deemed cost-effective, takeyour critical measurement challenges in-house on a
sub-contract basis.

Find additional product literatureand our product catalogue
http://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law. Please consult usin advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
http://www.mitutoyo.co.jp