Rotors & Accessories Selection Guide
Angle Rotors for LCX-200 / 1002ml

Click
Here

CA-1

CA-102ml×1810,000rpm：7,940G

Click
Here

2ml×2410,000rpm：9,730G

CA-15

Click
Here

[Outer line] 2ml×1810,000rpm：8,500G

CA-5

Click
Here

CA-16

Click
Here

2ml×2410,000rpm：9,390G[Inner line] 0.5ml×1210,000rpm：6,600G[Middle line] 0.2ml×1210,000rpm：5,810G

2ml

5ml

50ml
Rotor for Conical Tube

Rotor for 5ml Conical Tube

CA-6

Click
Here

2ml×36[Outer line] 2ml×1810,000rpm：9,390G[Inner line] 2ml×1810,000rpm：8,270G

CA-12

Click
Here5ml×1210,000rpm：9,170G＊Recommend Tube : TOMY 5ml tube,PT0050-11 for CA-12.

CA-8

ClickHere

50ml×410,000rpm：9,390G
or 15ml×410,000rpm：9,060G

50ml×810,000rpm：11,740G

50ml
Rotor for Conical Tube and5ml BDF (FACS tube)

CA-14HS

ClickHere

50ml×410,000rpm：9,390G
or 5ml×410,000rpm：7,490G＊The above speciﬁcations show Max. speed, Max. RCF, and Max. capacity. These values may change according to combination of centrifuge, rotor, bucket, adapter, and tube.
has been tested and certiﬁed by the Health Protection Agency (HPA, UK) to be compliant with the International Standard (Annex AA of the IEC 61010-2-020), meets requirements
for biohazard safety.
adopted a hermetically sealed structure equivalent toexcellent sealing performance.

rotors designed and tested inaccordance with the International Standard (Annex AA of the IEC 6101 0-2-020) to oﬀer

CA-1
Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

7,940 G

Tube angle

45 degree

Max. radius

71 mm

Min. radius

40 mm

Rotor capacity

2ml×18Rmin
Rmax

Applicable Tube to CA-1
Tube (φ×L [mm])AdapterMax.Number−−

Radius[mm]Max. RCF[G]

BART
CodeMaxMin

1871407,940

1186940

7,720

1MA006-01

1862406,930

1A-004

1870407,830

1A-004

1862406,930

1MA002-01PC

1854406,040

12 ml（11×42）

1.5 ml（11×41）

0.5 ml（7.9×31）

0.4 ml（5.7×50）

0.25 ml（5.7×38）

0.2 ml（6.3×24）

CA-10Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

9,730 G

Tube angle

45 degree

Max. radius

87 mm

Min. radius

55 mm

Rotor capacity

2ml×24

Rmin
Rmax

Applicable Tube to CA-10
Tube (φ×L [mm])AdapterMax.Number−−

Radius[mm]Max. RCF[G]

BART
CodeMaxMin

248755

9,730

1248655

9,620

1MA006-01

2478558,720

1A-004248655

9,620

1A-0042477558,610

1MA002-01PC

246955

7,720

12 ml（11×42）

1.5 ml（11×41）

0.5 ml（7.9×31）

0.4 ml（5.7×50）

0.25 ml（5.7×38）

0.2 ml（6.3×24）

CA-15
Rmax1

Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

8,500 G

Tube angle

45 degree

Max. radius 1 (Rmax1)

76 mm

Max. radius 2 (Rmax2)

61 mm

Max. radius 3 (Rmax3)

52 mm

Min. radius 1 (Rmin1)

44 mm

Min. radius 2 (Rmin2)

35 mm

Min. radius 3 (Rmin3)

37 mm

Rmin1

Rmin3
Rmax3

Rmin2
Rmax2

2ml×18 (outer line)
Rotor capacity

0.5ml×12 (inner line)
0.2ml PCR tube×12 (middle line)

Applicable Tube to CA-15
Tube (φ×L [mm])AdapterMax.Number−

Radius[mm]Max. RCF[G]

BART
CodeMaxMin

18 (outer line)

76

44

8,500

1−

18 (outer line)

74

44

8,270

1−

12 (inner line)

59

356,600

1MA006-01

18 (outer line)

67

44

7,490

1MA004-01

18 (outer line)

76

44

8,500

1MA004-01

18 (outer line)

67

44

7,490

112 (middle line)

52

37

5,810

118 (outer line)

58

44

6,490

12 ml（11×42）

1.5 ml（11×41）

0.5 ml（7.9×31）

0.4 ml（5.7×50）

0.25 ml（5.7×38）−
0.2 ml（6.3×24）
MA002-01PC

CA-5
Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

9,390 G

Tube angle

44 degree

Max. radius

84 mm

Min. radius

52 mm

Rotor capacity

2ml×24

Rmin
Rmax

Applicable Tube to CA-5
Tube (φ×L [mm])AdapterMax.Number−−

Radius[mm]Max. RCF[G]

BART
CodeMax

Min

2484529,390

12482

52

9,170

1MA006-01

2474528,270

1MA004-01

2484529,390

1MA004-01

2475

528,390

1MA002-01PC

2466

52

7,380

12 ml（11×42）

1.5 ml（11×41）

0.5 ml（7.9×31）

0.4 ml（5.7×50）

0.25 ml（5.7×38）

0.2 ml（6.3×24）

CA-6
Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

9,390 G

Tube angle

44 degree

Max. radius 1 (Rmax1)

84 mm

Max. radius 2 (Rmax2)

74 mm

Min. radius 1 (Rmin1)

52 mm

Min. radius 2 (Rmin2)

42 mm

Rotor capacity

2ml×36

Rmin1

Rmin2
Rmax2

Rmax1

Applicable Tube to CA-6
Tube (φ×L [mm])Adapter−2 ml（11×42）−
1.5 ml（11×41）

MA006-01
0.5 ml（7.9×31）

0.4 ml（5.7×50）

MA004-01

MA004-01
0.25 ml（5.7×38）

MA002-01PC
0.2 ml（6.3×24）

Max.Number

Radius[mm]Max. RCF[G]

BART
CodeMaxMin

18 (outer line)

84529,390

118 (inner line)

74428,270

118 (outer line)

82

529,170

118 (inner line)

72

428,050

118 (outer line)74

52

8,270

118 (inner line)

64

42

7,160

118 (outer line)

83

52

9,280

118 (inner line)

73

42

8,160

118 (outer line)

74

52

8,270

118 (inner line)

64

42

7,160

118 (outer line)

66

52

7,380

118 (inner line)

56

42

6,260

1

CA-12
Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

9,170 G

Tube angle

34 degree

Max. radius

82 mm

Min. radius

44 mm

Rmin
Rmax

5ml conical tube×12

Rotor capacity

Applicable Tube to CA-12PT0050-11(PP)

C5ml＊ Recommend Tube : TOMY 5ml tube, PT0050-11 for CA-12.

ActualTube Dimension Allowable capacity[ml](φ×L [mm])
RCF [G]

φ15×59ー3.5Max.
Number12

Max.RCF
Allowable[G]speed(Radius[mm])[rpm]

10,0009,170(82)1Remarks

TOM

BottomShape

Mfr
abbr. Model Name(Material)

BART
Code

Tube
Nominalcapacity[ml]CA-14HS
Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

9,390 G

Tube angle

25 degree

Max. radius 1 (Rmax1)

84 mm

Max. radius 2 (Rmax2)

67 mm

Min. radius 1 (Rmin1)

22 mm

Min. radius 2 (Rmin2)

33 mm

Rotor capacity

Rmin2

50ml conical tube×4＊1

5ml×4＊1

Rmin1
Rmax2

Rmax1

Allowablespeed[rpm]

Max.RCF[G](Radius[mm])＊1 Diﬀerent types of tubes cannot be loaded together.

Applicable Tube to CA-14HS

φ29×116

15,50010,0009,390(84)430304(PET)C

φ29×116

3,600

6,100

3,500(84)430829(PP)C

φ29×116

15,500FLC

352070(PP)C

φ30×1159,400IWA

2343-050(PP)C

φ29×1179,400

339652(PP)C

φ30×115

17,00010,0009,390(84)362696(PP)C

φ26.6×113.7

9,500

MS-56500(PP)C

φ30×1159,400

3182-345(PP)C

φ29×115

15,0003181-345(PP)C

φ29×115

15,000430053(PET)C

φ16×120

3,600

6,300

3,550(80)430766(PP)C

φ16×120

12,00010,0008,950(80)430791(PP)C

φ16×120

12,00010,0008,950(80)FLC352095(PS)C

φ17×120

1,800

4,400

1,730(80)NUC

339650(PP)C

φ17×120

10,50010,0008,950(80)3132-345(PP)C

φ17×118

15,00010,0008,950(80)3131-345(PP)C

φ17×118

15,00010,0008,950(80)352002(PP)R

φ12×75

3,000

6,300

2,970(67)352063(PP)R

φ12×75

3,000

6,300

2,970(67)352008(PS)Rφ12×75

1,400

4,300

1,390(67)352054(PS)R

φ12×75

1,400

4,300

1,390(67)352235(PS)R

φ12×75

1,400

4,300

1,390(67)

NUC

INA

Adapter

CA1500-01

CORINA

FLC
5ml＊Diﬀerent types of tubes cannot be loaded together.

Tube Dimension Allowable(φ×L [mm])
RCF [G]

3510.5

3.5

4441

11Remarks

Bottom
Shape

C

SUM

15ml

Max.
Number430291(PP)
COR

50ml

Actualcapacity[ml]

Mfr
abbr. Model Name(Material)

BART
CodeTube
Nominalcapacity
[ml]

CA-8
Applicable model

LCX-200 / LCX-100

Max. speed10,000 rpm

Max. RCF

9,390 G

Tube angle

25 degree

Max. radius 1 (Rmax1)

84 mm

Max. radius 2 (Rmax2)

81 mm

Min. radius 1 (Rmin1)

27 mm

Min. radius 2 (Rmin2)

34 mm

Rotor capacity

Rmin1

Rmin2

Rmax1

Rmax2

50ml conical tube×4＊1

15ml conical tube×4＊1＊1 Diﬀerent types of tubes cannot be loaded together.

Applicable Tube to CA-8Max.
NumberAllowable
speed
[rpm]

Max.RCF[G](Radius[mm])15,50035410,000

9,390(84)1

φ29×116

15,50035410,000

9,390(84)1C

φ29×1163,600

35

46,1003,500(84)1352070(PP)C

φ30×1159,400

35410,0009,390(84)1NAL3148-0050(PPCO)C

φ28.6×113.8 50,000

24.5

410,0009,390(84)1COR430766(PP)C

φ16×120

12,00010.5

410,0009,060(81)1COR430791(PP)C

φ16×120

12,00010.5

410,0009,060(81)1FLC352196(PP)C

φ17×120

6,000

10.5

48,1005,940(81)1COR430053(PET)C

φ16×120

3,600

10.5

46,3003,600(81)1FLC352095(PS)C

φ17×120

1,800

10.5

44,4001,750(81)1430291(PP)C

φ29×116430829(PP)C

430304(PET)

FLC

COR

50ml

Tube Dimension Allowable(φ×L [mm])
RCF [G]

35ml

15ml

＊Diﬀerent types of tubes cannot be loaded together.

Remarks

Actualcapacity
[ml]

Mfr
abbr. Model Name(Material)

BottomShape

Nominalcapacity[ml]

BART
CodeTubeCA-16
Applicable model

LCX-200 / LCX-100

Max. speed

10,000 rpm

Max. RCF

11,740 G

Tube angle

25 degree

Max. radius

105 mmMin. radius

42 mm

Rotor capacity

50ml conical tube×8

Rmin
Rmax

Applicable Tube to CA-16

15,50010,00011,740(105)430304(PET)C

φ29×1163,600

5,5003,550(105)430829(PP)C

φ29×116

15,500FLC

352070(PP)C

φ30×115

9,400

IWA

2343-050(PP)C

φ29×1179,400

339652(PP)C

φ30×115

17,000362696(PP)C

φ26.6×113.7

9,500

10,00011,740(105)MS-56500(PP)C

φ30×1159,400

3182-345(PP)C

φ29×115

15,0003181-345(PP)C

φ29×115

15,000430053(PET)C

φ16×120

3,600

5,600

3,510(100)430766(PP)C

φ16×120

12,00010,00011,180(100)430791(PP)C

φ16×120

12,00010,00011,180(100)FLC352095(PS)C

φ17×120

1,800

4,000

1,790(100)NUC

339650(PP)C

φ17×120

10,5009,600

10,310(100)3132-345(PP)C

φ17×118

15,00010,00011,180(100)3131-345(PP)C

φ17×118

15,00010,00011,180(100)

NUC

CA1500-01

CORINA

Tube Dimension Allowable(φ×L [mm])
RCF [G]

35

10.5

88

1

1

Remarks

BottomShape

φ29×116

INA

Adapter

Allowable
Max.RCF
speed[G][rpm](Radius[mm])

C

SUM

15ml

Max.
Number430291(PP)
COR

50ml

Actualcapacity[ml]

Mfr
abbr. Model Name(Material)

BART
CodeTube
Nominalcapacity[ml]Symbols in the Table
Manufacturer Abbreviation
BDC ： Becton, Dickinson and Company.
BEC ： BECKMAN COULTER ,INC.
COR ： CORNING INTERNATIONAL CORP.
EIK ： EIKEN CHEMICAL CO., LTD.
EPP ： EPPENDORF AG
FLC ： Falcon/CORNING INTERNATIONAL CORP.
HER ： Herolab GmbH Laborgeraete
HIT ： Eppendorf Himac Technologies CO., LTD.
IED ： IEDA TRADING CORPORATIONINA ： Ina-optika corporation.
IWA ： Iwaki/AGC TECHNO GLASS CO., LTD.
NAL ： NALGENE/ Thermo Fisher Scientific K.K.
NEG ： NICHIDEN-RIKA GLASS CO., LTD.
NIP ： NIPRO CORPORATION
NUC ： NUNC / Thermo Fisher Scientific K.K.
SEK ： SEKISUI MEDICAL CO., LTD.
SIO ： SHIONOGI & CO., LTD.
SUM ： SUMITOMO BAKELITE COMPANY LIMITEDTER ： TERUMO CORPORATION
TOM ： TOMY SEIKO CO., LTD.
TRE ： TOHO KK.
WAT ： WATSON CO., LTD.

Materials

HDPE ：High-density Polyethylene

FEP ：Teflon FEP

G ：Glass

PA：Polyallomer

PC ：Polycarbonate

PET ：Polyethylene Terephthalate

PP：Polypropylene

PS ：Polystyrene

PPCO ：Polypropylene Copolymer

PSF ：Polysulfone

SS：Stainless Steel

Bottom shape

C ：Conical

F：FlatR ：Round＊ The speciﬁcations of the tube listed in the applicable tube table indicate the nominal value of the manufacturer.
＊ If the speciﬁcations of the tube have been changed by the manufacturer, it may not be able to fulﬁll all the conditions stated in the tables. For the latest
speciﬁcations of the tube, please ask the manufacturer.