Truly wirelesswireless
Truly
Ranger Pro
ProRanger

ATEX/IECExZone
Zone 0/CSA
0/CSA ClassClass 11 Div
Div 11
ATEX/IECEx

ISA100&&WirelessHART
WirelessHART protocol
protocolISA100User-friendly andand collaborative
collaborative
User-friendly

Connectedwith
with System
System 11
Connected

Tempandandtriaxial
triaxial capable
capableTemp

Specifications summary (See datasheet 125M5237 for full details)

Accelerometers

Staticoverallvariables

Waveforms
and spectra

Temperature sensor

Wireless

Battery and power
Operating conditions

PhysicalAxis

1 or 3 axes

Amplitude range±20 g peakMeasurement accuracy±5% (160 Hz) Z-axis, ±10% (160 Hz) X and Y axis

Acceleration frequency
range

Z axis: 5 Hz (±3dB) to 10 kHz (±3dB),
X and Y axis: 5 Hz (±3dB) to 4 kHz (±3dB) (Tri-axial sensor only)

Accel Fmin / Fmax (Hz)2, 5, 10, 100, 200 / 200, 500, 1000, 2000, 5000, (10000 Z-axis only)

Velocity range

5 – 2000 (Hz) / 0 – 50 mm/s (0 - 2 in/s)

Velocity Fmin / Fmax (Hz)5, 10 / 200, 500, 1000, 2000

Measurement interval

10, 20, 30, 60, 120, 180, 240, 360 (minutes)

Accel Fmin / Fmax (Hz)2, 5, 10 / 200, 500, 1000, 2000, 5000, 10000

Number of samples

1024, 2048, 4096, 8192

Velocity Fmin / Fmax (Hz)5, 10 / 200, 500, 1000, 2000

Number of lines

400, 800, 1600, 3200

Waveform Measurement
intervals

6, 8, 12, 24, 48, 168, 336, 672 (hours)

Measurement range-40 °C to 120 °C (-40 °F to 248 °F)

Network standard

ISA100.11a or WirelessHART®

Radio standard

IEEE 802.15.4, 2.45 GHz ISM band

Encryption/ security

128-bit AES encrypted packets

Wireless range

150 meters sensor to access point, 100 meters sensor to sensor, line of sight(actual range depends on obstacles present, gateway antenna type, andorientation of the sensor relative to the gateway antenna)

Type

Replaceable D size 3.6V lithium-thionyl chlorideLife

Up to five years depending on the operating mode and configuration

Operating temperature-40 °C to 85 °C (-40 °F to 185 °F)

Weight

230 grams (without battery; 300 grams with battery)Dimensions

Height: 88 mm; diameter: 40 mm

Case material

316 stainless steel body and glass-reinforced, impact-resistant

Mounting holeM6 x 1mm X 5mm deep internal thread

IP rating

IP67 dust and water resistant (pending)

EMC conformity

IEC 61326-1, ETSI EN 301 489-1, CISPR22, ETSI EN 301 489-17

Radio spectrum

ETSI EN 300 328

Regulatory compliance Safety

Compatible gateways

ETSI EN 61010-1, IEC 62479

Hazardous atmosphere

CSA Class 1 Division 1 Groups A, B, C, D T4 ATEX/IECEx Zone 0Conformity

Compliant with all CE and FCC/IC requirements

Bently Nevada

ISA100, up to 50 Ranger Pro sensors per gateway

EmersonWirelessHART®-Emerson 1410 or 1420 v4.7.84 or higher, up to 70 Ranger Prosensors per gateway

Honeywell

ISA100–Honeywell WDM R310.2-4 or higher, up to 40 Ranger Pro sensors peraccess point

Yokogawa

ISA100–Yokogawa YFGW 410/510, up to 40 Ranger Pro sensors per access point

Copyright 2021 Baker Hughes Company. All rights reserved. Baker Hughes provides this information on an “as is” basis
for general information purposes. Baker Hughes does not make any representation as to the accuracy or completeness of
the information and makes no warranties of any kind, specific, implied or oral, to the fullest extent permissible by law,
including those of merchantability and fitness for a particular purpose or use. Baker Hughes hereby disclaims any andall liability for any direct, indirect, consequential or special damages, claims for lost profits, or third party claims arisingfrom the use of the information, whether a claim is asserted in contract, tort, or otherwise. Baker Hughes reserves theright to make changes in specifications and features shown herein, or discontinue the product described at any time
without notice or obligation. Contact your Baker Hughes representative for the most current information. The BakerHughes logo, the Bently Nevada logo, and Ranger Pro are trademarks of Baker Hughes Company.
GEA34045A(06/2021)

bakerhughes.com