ID-C/ID-F

CatalogCatalog
No.E12049(4)No.00000

Small Tool Instruments
and Data Management

High-performance ABS Digimatic Indicators

High-performance
ABS Digimatic
Indicator

ID-C/ID-F

The first Mitutoyo measuring tools to support bidirectional serial communication.
Dramatically improve work efficiency by connecting and linking with a PC.

New-generation ID series making
measurement operations smoother andenhancing production quality
Bidirectional serial communication that helps increase work efficiency
Meeting the need for more precise
measurements

The ID-C/ID-F units are Mitutoyo’s first measuring tools to support bidirectional serial communication.* They can be easily
connected and linked with a PC via a USB input tool, etc., and in addition to conventional measurement data collection, theyalso enable control and setting of the ID-C/ID-F units, collection of gauge information, and other operations to be performed
in batch from the PC. This contributes to drastic improvement in work efficiency.
*Achieved through I/F compatible with an original bidirectional serial communication specification (Digimatic S1). See P.6 for details.
An optional cable and measurement data input unit are required for bidirectional serial communication.
USB-ITPAK V3.0 must be installed on the PC used for communication.

0.000

0.000
0.000

for smoother measurement work

Zero set

ID-F Series

0.000

Zero set

0.000
0.000

Zero set

When tolerance judgment
result indicates failure

0.5 μm/0.00002 inresolution

The ID-C and ID-F ranges now include
models with 0.0005 mm/0.00002 inresolution. The units are also capableof resolution switching.**Except for the ID-C 0.01 mm/0.0005 inresolution model

2

The icon starts blinking at a set time before
calibration is due (e.g. 1 week before the calibrationdate). If the due date is exceeded, the entire screen
starts blinking to notify the user.

Old model(Old ID-C/ID-F + USB-ITPAK V2.1)

Old model(Old ID-C/ID-F + USB-ITPAK V2.1)

For older ID-C/ID-F units that do
not support bidirectional serial
communication, individual button
operation is necessary for zerosetting.

Since bidirectional serial
communication is not supported,function setting from a PC is notpossible.

The units have large screens that can display
various information in an easy-to-read manner.
They also have an analog bar, convenient for
observing subtle movements such as the approachto tolerance.

Ease of use is greatly
improved by three large
buttons. You can freely
set any frequently usedfunctions to these buttons.

Previous model

Calibration schedulewarning
Notification icon

➡ Ease of use greatly improved

Various functions of ID-C/ID-F units
can be set from USB-ITPAK.
The contents of various function
settings can be saved on a PC, andyou can copy them to other ID-C/ID-F
units.
➡ Reduced work time for setting

Setting of frequently used functions for easy operation

Avoid missing a pending calibration

The operator can set a
calibration validity and reminder
date. This function can support
better management of thegages.

New model(ID-C/ID-F + USB-ITPAK V3.0)

Improved work efficiency thanks to excellent readability

Addition of
analog bar

ID-C

Enabling more precise measurement

New model(ID-C/ID-F + USB-ITPAK V3.0)

Zero set

Large screen and analog bar*The ID-C series does not have illuminated backlighting.

Function example (2)Measuring instrument setting

Batch zero setting and power ON/OFF
operation, etc. of multiple ID-C/ID-F
units is made possible by use of thededicated software “USB-ITPAK.”

A wide range of support functions

ID-C Series

Function example (1)Control of ID-C/ID-F from PC

ID-F

Three large buttons

New model

ID-C

Parameter setting mode
• Counting direction switching• Tolerance judgment function setting • Resolution switching• Calculation function setting • Function lock setting

Improved analog bar

Switching between ABS (presetting) and INC(zero setting)Character size
enlarged about
1.5 times

Power ON/OFF• Data output (when connected to an external device)• Data hold (when not connected to an external device)

Watch the video for details.

ID-F

Improved measurement
work efficiency

Simple calculation
function
The result of the spindle
movement value multipliedby the calculation coefficient
can be displayed in realtime. This reduces the workof measuring with a jig or
similar tool.

f(x) = Ax
f(x): Displayed value
x: Spindle movement value
A: Selected value

Product information

3

ID-C Series

SPECIFICATIONSMetric ISO/JIS Type
Code No.
w/ lugFlat back543-700
543-705*2——
543-710543-715*2——

543-700B
543-705B*2
543-720B
543-730B
543-710B
543-715B*2
543-725B
543-735B

Range (mm)

12.7
25.4
50.8
12.7
25.4
50.8

Maximum permissible error (MPE)*1
Maximum permissible
Net mass (g)(mm)
limit (MPL)
Resolution (mm)Partial
Total
Hysteresis Repeatability Measuring force
Flat
measuring measuring
w/ lug
H MPE
R MPE
(N)
back
range P MPE range E MPE
1.5 or less
175
165
0.0005/
0.4 to 0.7
170
160
0.003
0.003
0.001/0.01
0.002
0.002
1.8 or less—195
(selectable)
0.005
0.005
2.3 or less—
260
0.9 or less
170
160
0.2 to 0.5
165
155
0.02
0.02
0.01
0.02
0.01
1.8 or less
—
190
0.04
0.04
2.3 or less—245

*1 These values apply to normal measurements at 20 °C (Resolution: 0.0005 mm, Allowable value: A=1)

*2 Low measuring force

Inch/ Metric ISO/JIS Type
Code No.

543-700

543-702B

543-710B

543-717543-720B

ID-C0512NX

ID-C0512ENXB

ID-C1012NXB

ID-C1012CENX

ID-C0525NXB■ Example of ID-CNX installedon optional bore gage

w/ lugFlat back543-701543-706*2——
543-711543-716*2——

543-701B
543-706B*2
543-721B
543-731B
543-711B
543-716B*2
543-726B
543-736B

Range

0.5 in/ 12.7 mm
1 in/ 25.4 mm
2 in/ 50.8 mm
0.5 in/ 12.7 mm
1 in/ 25.4 mm
2 in/50.8 mm

Maximum permissible error (MPE)*1
Maximum permissible
Net mass (g)(mm)
limit (MPL)
Resolution
Partial
Total
Hysteresis Repeatability Measuring force
Flat
measuring measuring
w/ lug
H MPE
R MPE
(N)
back
range P MPE range E MPE
1.5 or less
175
165
0.00002/0.00005/
0.0001/0.0005 in,
0.003
0.003
0.4 to 0.7
170
160
0.002
0.002
1.8 or less—
195
0.0005/0.001/
0.005
0.005
2.3 or less—
260
0.01 mm (selectable)
0.9 or less
170
160
0.02
0.02
0.2 to 0.5
165
155
0.02
0.01
0.0005 in/ 0.01 mm
1.8 or less
—
190
0.04
0.04
2.3 or less—245

*1 These values apply to normal measurements at 20 °C (Resolution: 0.0005 mm, Allowable value: A=1)

*2 Low measuring force

Inch/ Metric ASME/ANSI/AGD Type
Code No.
w/ lugFlat back543-702543-707*2——
543-712543-717*2——

543-702B
543-707B*2
543-722B
543-732B
543-712B
543-717B*2
543-727B
543-737B

Range
0.5 in/ 12.7 mm
1 in/ 25.4 mm
2 in/ 50.8 mm

Resolution
0.00002/0.00005/
0.0001/0.0005 in,
0.0005/0.001/
0.01 mm (selectable)

0.5 in/ 12.7 mm
1 in/ 25.4 mm
2 in/50.8 mm

0.0005 in/ 0.01 mm

Maximum permissible error (MPE)*1 Maximum permissible limit
(in)
(MPL)

Net mass (g)Overall*3

Measuring force (N)w/ lug

1.5 or less
0.4 to 0.7
1.8 or less
2.3 or less
0.9 or less
0.2 to 0.5

195
190
—
—
190
185
—
—

Hysteresis Repeatability±0.00012

0.00008

0.000080.00100.0005±0.00020±0.0010

1.8 or less±0.0015*1 These values apply to normal measurements at 20 °C (Resolution: 0.0005 mm, Allowable value: A=1)

2.3 or less
*2 Low measuring force

Flat
back
165
160
195
260
160
155
190
245

*3 Overall magnification and linearity

See page 7 for external dimensions
Direction setting, etc. is necessary when
ID-CNX is used with a bore gage. Referto the operation manual for details.

Common Specifications
12.7 mm/0.5 in models
Display
Display rotation
Protection level*2
Posssible plunger direction

543-722B

ID-C0525ENXB

543-730B

543-737B

ID-C0550NXB

ID-C1050ENXB

Power source
Battery life*3
Detection method
Response speed
Errors, Alarms
Operating temperature
Storage temperature

All directions

Low measuring force models*1
25.4 mm/1 in, 50.8 mm/2 in models
7 segements height: 11.0 mm, Analog bar (±20 scale)
330 °Equivalent to IP-42
0.0005 mm models: Plunger downward only
Up to direction in which plunger is horizontal
0.01 mm models: Up to direction in which plunger is horizontal
Lithium metal battery CR2032 (1pc.)
Approx. 2.5 years (normal use), Approx. 2,700 hours(continuous use)
Electrostatic capacitance type absolute linear encoder
No limit
Various setting errors, Sensor error, Display overflow, etc.
0 to 40 °C-10 to 60 °C*1: The item whose Code No. with an asterisk *2 is Low measuring force model like 543-706*2. See the specification table above.
*2: Protection level (IP=International Protection) is based on IEC 60529 (JIS C 0920). The levels shown are valid for factory conditions only.
*3: When the data processor is not connected. Battery life depends on use of the indicator. Use the above value as a reference.
Note: Allows high accuracy measurements of MAX/MIN and TIR (MAX-MIN). The peak detection speed is 50 times /s.Various contact points are available as optional accessories.

45

ID-F SeriesComparison of functions
ID-C Series
✓
✓
✓
✓
✓
✓*2
✓
✓

PresetZero set
Peak detection (Max, Min, TIR)
Unit system switching*1
Counting direction switching
Resolution selecting
Tolerance judgment
Simple calculation

ID-F Series
✓
✓
✓
✓
✓
✓
✓
✓

ID-C Series
✓
✓
✓
✓
✓
✓
✓

Analog bar display ON/OFF
Analog bar scale selecting
Key customize
Function lockCalibration schedule warning function
Auto OFF
Reset all settings

ID-F Series
✓
✓
✓
✓
✓
—
✓

*1: in/mm models only *2: Except 0.01 mm/0.0005 in models

Dimensions (ID-C Series)
12.7 mm range models

0.002
0.004
0.003

1.8 or less

Unlimited

ACadapter(5.9 V)

2.3 or less

ø59

170
240
330

0
ø8 -0.009

ø4.8

Ball contact (ø3)
901312(thread: M2.5×0.45)*1 These values apply to normal measurements at 20 °C (Resolution: 0.0005 mm, Allowable value: A=1)

Inch/Metric ASME/ANSI/AGD TypeNet mass(g)200
1.5 or less±0.00010
0.00008±0.00016±0.00012

0.00008

1.8 or less25.4 mm range models

170
ACadapter
Unlimited(5.9 V)

ø19.2

Unit: mm

2.3 or less33.2

See page 8 for external dimensions

ø0.76

Unit: inch

1.31

84.8
70
19.5

0
ø8 -0.009

7.3

40.8

26

Response speed
No limit
Errors, Alarms
Various setting errors, Sensor error, Diplay overflow, etc.
Outputd1, d2
I/O
S1Operating temperature
0 to 40 °C
Storage temperature-10 to 60 °C*1: Protection level (IP=International Protection) is based on IEC 60529 (JIS C0920). The levels shown are valid for factory conditions only.
Note: Allows high accuracy measurements of MAX/MIN and TIR (MAX-MIN). The peak detection speed is 50 times /s for resolution of 0.0005 mm and 500 times/s otherwise.
Various contact points are available as optional accessories.

0.43

3.30
2.67

ø59

Common Specifications
7 segements height: 11.0 mm, Analog bar (±20 scale)
330 °Equivalent to IP-40 (No protection against ingress of water)
Up to direction in which plunger is horizontal
AC adapter (DC 5.9 V)
Electrostatic capacitance type absolute linear encoder

11

330

*1 These values apply to normal measurements at 20 °C (Resolution: 0.0005 mm, Allowable value: A=1) *2 Overall magnification and linearity
Note: To denote your AC power cable add the following suffixes to the order No.: -A for UL/CSA, -D for CEE, -DC for CCC, -E for BS, -K for KC, No suffix is required for JIS/100 V.

Display
Display rotation
Protection level*1
Possible plunger direction
Power source
Detection method

ASME/ANSI/AGD Type

ISO/JIS Type

240

ø2.32

Power
sourceBall contact (ø3)
901312(thread: M2.5×0.45)

0ø0.375-0.0012

ø4.8Ball contact (ø0.12)
1/4

543-856
0.5 in/0.00002/543-856B 12.7 mm
0.00005/(flat back)
0.0001/ 0.0005/
1 in/
0.001 in,
543-852
25.4 mm
0.0005/ 0.001/
543-854
2 in/
0.01 mm
543-858 50.8 mm

0.30

Note: Products with a code No. suffixed "B" have a flat back, and other models have a center-lug back.

Maximum permissible limit (MPL) Response
Maximum permissible error (MPE)*1 (in)speedOverall*2
Hysteresis
Repeatability
Measuring force (N)0.68

Resolution

Ball contact (ø0.12)
21BZB005(thread: No. 4-48UNF)

2.27

Range

ø0.19

7.6

57.7

Code No.

0ø0.375 -0.0012

1.66

0.004
0.003

0.002

1/4

2.00
1.74
0.79 0.58
1/4

0.0025170.0025

1.02

25.4
50.8
50.8

0.0005/0.001/
0.01

1.5 or less

54.4
46.5

180
12.7

5

Net mass(g)/4

Hysteresis
H MPE

Power
source3/4

ø1Partial measuring Total measuring
range P MPE
range E MPE

Maximum permissible
limit (MPL)
Response
speed
Repeatability Measuring force
R MPE
(N)

13.3

Maximum permissible error (MPE)*1
(mm)

1.20

.5

543-855
543-855B(flat back)543-851543-853
543-857

Resolution
(mm)

ø0.44

Unit: inch20

ø6

w/ lug

Range
(mm)

30.5

50.1

Metric ISO/JIS Type
Code No.

ø11.2

Unit: mm

ø2.13

ID-F0550ENX

ø2.32

ID-F0525NX

ASME/ANSI/AGD Type1.97

543-854

0.52

SPECIFIACTIONS

ISO/JIS Type543-851ø54

543-855B

ID-F0512NXB

21.2
7.3

543-855

ID-F0512NX

ø0.19

21BZB005(thread: No. 4-48UNF)

Note: All products have a flat back.

6

7

25.4 mm range models

50.8 mm range models
ISO/JIS Type
ø19.2

Unit: mm

ISO/JIS Type

ASME/ANSI/AGD Type
33.2

11

ø0.76

Unit: inch

1.31

0.43

ASME/ANSI/AGD Type

Unit: mm

Unit: inch

38.1

11

1.50

ø 0.76

ø 2.32

ø59

0.68

1.38

0
ø0.375 -0.0012

ø4.8

901312(thread: M2.5×0.45)

Ball contact (ø0.12)

1.66

1.02

Ball contact (ø3)

ø 0.1921BZB005(thread: No. 4-48UNF)

1/4

7.3

ø4.8
Ball contact (ø3)
901312(thread: M2.5×0.45)

2.62

2.05

65.3

52

1/4

7.3

26

0.68

ø0.3750-0.0012

2.67

3.30
0
ø 8 -0.009

40.8

4.32
3.74
19.5

35

19.5

84.8

70

ø2.32

ø59
110.6
97.3

ø8

0
-0.009

0.43

2.27

57.7

3.35

85.2

ø19.2

50.8 mm range models

Ball contact (ø0.12) ø0.19
21BZB005(thread: No. 4-48UNF)

ISO/JIS Type

Note: All products have a flat back.

ASME/ANSI/AGD Type

Unit: mm

Unit: inch

ø19.2

38.1

ø0.76

11

1.50

0.43

85.2

12.7 mm range models

0.68

19.5
0.30

Ball contact (ø3)
901312(thread: M2.5×0.45)

0ø 0.375 -0.0012

2.62

ø4.8
2.05

Ball contact (ø0.12)
21BZB005(thread: No. 4-48UNF)

3.74

4.32

97.3

110.6

ø0.19

65.3

1/4

ø 8 -0.009

7.3

7.6

ø0.3750-0.0012

52

0.58
0.52

21.2
7.3

Ball contact (ø3)
901312(thread: M2.5×0.45)

0.79

17

54.4
46.5

0
-0.009

ø4.8

13.3

0

2.00
1.74

.5

ø8

1.38

/4

ø1

ø6

ø2.13

1/4

ø2.32

ø59

3/4

35

ø54

5

ø 2.32

1.39

1/4

ø0.44

Unit: inch

20

50.1

35.3

1.97

ø11.2

ASME/ANSI/AGD Typeø59

ISO/JIS Type
Unit: mm

3.35

Dimensions (ID-F Series)

Ball contact (ø0.12)

ø 0.1921BZB005(thread: No. 4-48UNF)

Note: Products with an Order No. suffixed "B" have a flat back, and other models have a center-lug back.

8

9

Optional Accessories

Software Reduces the time and effort needed for inspection work

Options
Cable Can be used for both ID-C/ID-F with SPC cable.
Measurement Data Input Unit
USB Input Tool Direct (2 m)
USB-ITN-SFSPC Connection Cable
SPC Cable for connecting ID-C/ID-F to an external device
such as IT-020U. Can be used for both ID-C/ID-F.

SoftwareMeasurement Data Collection Software

Measurement Data Collection Software
USB-ITPAK V3.0

USB-ITPAK is useful software to create procedures when inputting measurement data into Excel sheets. The latest version allows theuser to perform batch power-on for ID-C/ID-F units, batch power-off at the end of measurement, batch zero setting and presetting, dataacquisition instruction from a PC, etc.

USB-ITPAK V3.0

SPC Connection Cable (1 m)

Equipped with an automatic sorting function for sorting input measurement data [Easy input mode]This function can be implemented even if the measuring instrument
does not support bidirectional serial communication. After setting,measurement values ​​are automatically sorted into an Excel sheet as
needed.

No.06AGQ001F
Measurement Data Input Unit
USB Input Tool
IT-020U

USB-ITPAK V3.0 full version dongle

No.264-020

No.06AGL021

No.06AGR543

Connection Unit (for 12.7 mm range models only)

U-WAVE Mounting Plate(Once entered into column A, similar data is automatically classified.)

Simplifies measuring instrument setting
Batch setting of ID-C/ID-F units can be
performed from your PC. Moreover, thesettings can be saved on your PC and copy toother ID-C/ID-F units. Without even touching
the ID-C/ID-F units.

U-WAVE If using U-WAVE, please note that it does not support bidirectional serial communication.
Transmission Unit (U-WAVE-TM)

With automatic sorting function

D: 10 mm
H: 20 mm
W: 30 mm

SPC Connection Cable (2 m)

Measurement data collection software USB-ITPAK
V3.0 can be downloaded from our website. The above
dongle is required to use the full functions.

(Entered into column A only.)

Just preset the number of measurement
items.(Example: number of measurement
items = 3)

No.06AGL011

USB dongle

With normal input

Deploy settings to other ID-C/ID-F units• Presetting• Resolution• Tolerance setting...etc.

USB-ITPAK V2.1/V3.0 Function comparison table

Settings
Save

Symbols: ✓ 1 : Can be used only when connected with USB-ITPAK V3.0 and ID-C/ID-F;✓ : Can be used; — : Cannot be used

ITPAK
Operating environment andfunctions

Attachment example

No.264-623
No.264-622

Optional items such as alifting lever can be attachedwhile the U-WAVE-TM/TMB
is in place.

Transmission Unit (U-WAVE-TMB)*Cannot be attached to themodels with center-lug
back.

Buzzer type
Waterproof/dustproof type

Supported
communication
standard

Measurement Data Wireles Reception Unit
(U-WAVE-R)

No.02AZG011

Please note that a conventional
ID-C/ID-F cable cannot be
used because the shape of theconnector is different.

No.02AZG021

3rd generation, bidirectional serial communication, 8-digit communication—✓

Windows 2000 SP4, Windows XP SP2 or later, Windows Vista, Windows 7, Windows 8 / 8.1✓—

Windows 10✓

With this method, when using one or several measuring instruments, the measurement
data are input into an Excel sheet from the measuring instrument(s) registered in advance.

✓

—

✓

Batch measurement

With this method, measurement data are acquired in batch from several measuring
instruments and input into an Excel sheet.

✓

—

✓

No.02AZF670

Individual
measurement

The Excel sheets and cells for inputting measurement data are set individually for eachmeasuring instrument. With this method, measurements performed randomly by multiple
operators can be input from each instrument into their specified sheets and cells.

✓

—

✓

Foot Switch

Simple measurement
function

This function makes it possible to start measuring without prior detailed settings and tosort data into Excel columns according to measurement location.

—✓

Measuring instrument Function to change the various settings (zero setting, registration of preset values, setting ofsetting
unit, counting direction, and tolerance) of connected measuring instruments.

—✓1

FunctionsNo.937179T✓

Measurement history

This function saves information on the measurement operator and the measurement
equipment used within the measurement data.
(It records in the data who used what to measure the data.)—✓1

Device information

This function reads various information about connected measuring instruments (model,
serial No., calibration date) and displays it on the PC.

—✓1

Data input into
Microsoft Excel

This function is used to input values into user-specified cells in Excel.

✓

—

✓

Text data input with
virtual keyboard

This function is used to input text (characters and values) into specified cells in Excel.

✓

—

✓

to U-WAVE-T unit

Many other options are also available. For details, please visit our website. https://www.mitutoyo.co.jp

10✓

Sequential
measurement

to measuring tool

No.02AZD810D

d1: 1st generation, unidirectional communication, 6-digit communication /d2: 2nd generation, unidirectional communication, 8-digit communication—

U-WAVE-T Dedicated Connection Cable
for Foot SwitchFootswitch

V3.0
V2.1 Trial
Fullversion version(free)

Windows 11
Standard U-WAVE-T DedicatedConnection Cable

No.264-627
No.264-626

Digimatic S1

Supported operating systems

No.02AZF700

Buzzer type
Waterproof/dustproof type

Digimatic d1/d2

Details11

24 07 (JP4), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Notes on Export Regulations:
Do not commit an act, which could directly or indirectly, violate any law or regulation of Japan,
your country or any other international treaty, relating to the export or re-export of any commodities.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan

Note: Product illustrations are without obligation. Product descriptions, in particular any and all technical specifications,are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.
All product information contained in this brochure is current as of July 2024.

T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp