Small Tool Instruments andData Management

High-Accuracy Height Gage

Linear Height LH-600F/FG

CatalogCatalog
No.E12012(5)No.00000

Easy operation andHigh-accuracy
In pursuit of intuitive operation and outstanding ease of use
Achieves best-in-class accuracy of ± (1.1+0.6L/600) μm

High-Accuracy Height Gage

Linear Height LH-600F/FG
Easy operation using keypad and touch screen navigation,even suitable for beginners
Conduct various measurements such as 2D measurement
and perpendicularity measurement with just one tool
Versatile measurements through optional probes
Enhanced data output functions make it easier to manage
your measurement data23

Easy operation using keypad and touch screen navigation, even suitable for beginners
Contextual guidance on the large-screen touch panel supports your measurements
Simple, straightforward keys with icons

Icons allow the user to find the required operation at a glance.

Home screen
With the intuitive menu, even beginners can easily access various operations and settings.

Basic measurements

Height (top)measurement

Height (bottom)
measurement

Width (inside)
measurement

Width (outside)
measurement

Shaft (top)measurement

Shaft (bottom)measurement

Circle (shaft)measurement

Max.-Min. (top)measurement

Hole (bottom)
measurement

Hole (top)measurement

Circle (hole)measurement

Max.-Min. (bottom)
measurement

Measurements· Basic measurements (ABS)· Basic measurements (ABS,INC×5)· Angle measurements· Hole position measurements· Perpendicularity andstraightness measurements· 2D measurements· Part program measurements(1D, 2D)

Touch panel

orSheet key

Advanced measurement functions

Part programmeasurement

Touch panel with guidance
Measurement guidance is displayed on the large
touch panel of the 8.4-inch color LCD, enabling
intuitive operation.

Power2D measurement

Measurement results

PowerMeasurement settings

Measurement origin
setting

General settings

Guidance and measurement navigation

Probe selection

Measurement origin
switching

System settings

Data output

Guidance is available in 21 languages. The display shows each measurement step andit’s very easy to use even for beginner.

OtherHome screen
Guidance screen

Select basic measurements

Operable with gloves

Select a measurement method

Conduct the measurement

View the measurement results

Fantastic features for shop-floor use
Automatic scale check

Calibration reminder
On start up the user
is led through themenu to set thescale origin andrun the automatedprocedure tocheck the scale forcontamination.

4Operation log
A notification will
be displayed before
the calibration due
date that the user
has set.

Repeat measurement function
Operation log data
is retained for up to
2 months, and canbe output to a USB
memory device.

To enable efficientmeasurements theuser can repeat thelast measurementwith the optionalfoot switch or
on-screen button.

5

Various measurements with just one unit
Improved usability and accessibility, including advanced measurement functions
2D measurement - Pre-placement -Part program measurement

This function allows the user to register the hole position of the workpiece before measurement.

You can easily access and use the functions of Create, Run, Edit, and View results of part programs.

②③①Notes

Touch panelor
Part program contents

Sheet key
Part program list

ExecuteCreate a new part program

Save part programs

Edit part programs

Display measurement results from a part program

Example of performing part program measurement

Perpendicular/straightness
measurement- Graph creation You can check themeasurement results ofperpendicularity andstraightness in real time
during measurement. Aftermeasurement, you caneasily see the trends of themeasurement results in agraph.

Set the number of times to repeat
measurements and run the program.

Run the program repeatedly (when the number of executions is ≠1)

Run the registered contents
one by one.

Display the result for one
execution of the program.

Display all results for one
execution of the program.

Display all results of N times
of executing the program.

Copy part programs to other devices

By using a USB memory device, you can copy part programs to other devices.

Expanded measurement areaShow results in real time during measurement.

Show results after measurement.

With the new optional probes, you can now measure areas that were beyond the capacity of conventional probes.
Example of deep hole measurement

Maximum reach
300 mmHole depth
300 mmFull stroke
600 mm
You can monitor the graphs of past measurement results.

6Show measurement results in graph form.

7

Versatile measurements through optional probes
An extension holder and a depth stylus extend the measuring range both horizontally and verticallyThree types of optional probes:
1 The holder and the stylus can be freely combined according to the purpose ofmeasurement, and the measurement area can be changed.
2 This type is used for single-use measurements such as measuring a tapered holeor a knife edge.
3 This type is used to measure straightness and perpendicularity.
☆ You can adjust the balance of the slider by adjusting the number of auxiliaryweights.
(Magnetic auxiliary weights are easy to add and remove.)①Probes

Styli for position B/C

HolderStylusMounting example②③

Φ5 ball stylus L130 in position B

HolderDigimaticindicator

Slider☆

Optional probe mounting positions

Φ5 ball stylus L130 in position CAuxiliaryweights

DCAB

Holders/styli for position A
Mounting example
For extension holder 100

Part No.

Product name

Number ofweights

12AAY343

ø5 stepped probe (standard accessory)2

12AAA792

Holder for Dial Indicator (millimeter)

012AAA793

Holder (long)*1

For depth measurement probes

12AAB136

12AAY595

ø10 cylindrical universal probe2*1: Please refer to page 8 for the number of weights to be used.

Product name

Number ofweights*3

12AAF666

ø1 ball stylus (coaxial type)2957261

ø2 ball stylus (coaxial type)

212AAF667

ø2 ball stylus (coaxial type), ruby ball

2957262

ø3 ball stylus (coaxial type)2957263

ø4 ball stylus (coaxial type)

212AAB552

ø10 ball stylus (coaxial type), L=50

212AAF668

ø10 ball stylus (coaxial type), L=82

112AAF669

ø10 ball stylus (coaxial type), L=120

112AAF670

ø5 disk stylus

212AAF671

ø10 disk stylus

2957264ø14 disk stylus

2957265

ø20 disk stylus

212AAF672

ø1 ball stylus (eccentric type)

212AAF673

ø2 ball stylus (eccentric type)

212AAA788

ø4 ball stylus (eccentric type)

212AAA789

ø6 ball stylus (eccentric type)1226117

Shank with M2 thread*22226118

Shank with M3 thread*2

212AAY597

ø5 ball stylus L130

112AAY598

ø25 disk stylus

1*2: Stylus for coordinate measuring machine can be mounted. *3: When using an extension holder Note: Where the material is not described, the tip of the stylus is carbide.

Extension holder 100*1

For taper stylus (ø20)

Styli for position D
12AAY596

Part No.

Extension holder 200*112AAC072

Depth probe

212AAC073

Tapered stylus (ø20)

2Part No.

Product name

Number ofweights*4

Mounting example

12AAY599

Depth stylus 702

Depth stylus 70 is mounted in position D

12AAY600

Depth stylus 150

112AAY601

Depth stylus 300

012AAY602

Depth stylus 70 ø2 ball

212AAY603

Depth stylus 150 ø2 ball

112AAY604

Depth stylus 300 ø2 ball

012AAY605

Depth stylus 70 ø4 ball

212AAY606

Depth stylus 150 ø4 ball

112AAY607

Depth stylus 300 ø4 ball0

Depth stylus 150 ø4 ball is mounted in position D*4: Can only be attached to the extension holder. * Where the material is not described, the tip of the stylus is carbide.

89

Specifications
ModelOrder No.
Power grip

U-WAVE
Navi*2

Transmitting unit
U-WAVE-T
Digimatic
outputDigimatic
outputReceiving unit
U-WAVE-R

SerialoutputSmartphone/tablet(when usingBluetooth® U-WAVE)

Input tools

USB cable(A-B)
12BAF812*4

Excel*3

USB-ITPAK V3.0

*5

LH-Communication Tool V1.0

*1: Available at Apple Store and Google Play for free download. *2: Available at Google Play for free download.
*3: Excel is a registered trademark of Microsoft Corporation. *4: It is also possible to connect with a Bluetooth compatible PC.
*5: MeasurLink® is a registered trademark of Mitutoyo Corporation (Japan) and Mitutoyo America Corporation (USA).

Optional products for outputting measurement data
Part No.
12AAY481
12AAY482
12AAY483
12AAY484
12AAN052
12AAY485
12AAN146—
12BAF812
543-700B
543-701B
519-521519-561

Product name
Receipt printer (for Japan)*6
Receipt printer (for North America)*6
Receipt printer (for EU countries, excluding the UK)*6
Receipt printer (for the UK)*6
Printer paper for receipt printer (set of 10)
Printer mounting attachment
Connection cable for printer(USB memory device)*7
USB cable (type A - type B) (2 m)
Digimatic indicator (ID-C0512NXB)
Digimatic indicator (ID-C0512MNXB)
Lever head probe MLH-521
Mu-checker M-561*6: A small printer (optionally battery-powered) that can be mounted on the main unit.
It includes a printer cable and mounting bracket.
*7: USB memory devices should be formatted with FAT16/32.
NTFS and exFAT are not supported.

Part No.
936937965014264-505
06AGQ001F
264-02006AGL011
06AGL021
12AAJ088
02AZD810D
264-626264-627
02AZD730G
02AZD880G
12AAY486
02AZG011* Available at Mitutoyo website for freedownload.
* T o connect to a PC, use a USB cable (typeA-B).

Power supply
Battery operation time*10
Battery charging time*11
Dimensions (W×D×H)
Mass
Operating temperature / humidity ranges
Data output

4 µm (mechanical accuracy)
Motor-driven (5, 10, 15, 20, 25, 30, 40 mm/s: 7 steps) / Manual
Photoelectric incrermental encoder STVC-20Z
1 N (automatic constant-force function)
Full-floating (moving) / Semi-floating (measuring) Air bearing (built-in compressor)
8.4 inch touch-screen, LCD
Stepless tilt adjustment: 0 to 40°
Stepless swivel adustment: -30 to 180°
Scale status notification, calibration schedule notification· Semi-automatic compensation using the probe diameter calibration block (standard accessory)· Compensation by inputting the probe diameeter
AC adapter 100-240V±10% 50/60Hz/ Battery (NiMH)
Battery powered(standard): 4 hours, Powered by 2 batteries: 8 hours
Approx. 3.5 hours (can be used while charging)
238 (W) × 492 (D) × 996 (H) mm
26.1 kg
26.6 kg
5 to 40 °C/ 20 to 80% RH (non-condensing)
Digimatic d2/ S1 (bi-directional communication)*8: Specification determined at in-house ambient temperature*10: In-house standard(floating and motor-driven vertical movement, operated at 25%)

Dimensions*9: Guaranteed when using the Lever Head (519-521) and Mu-Checker (519-561).
*11: When ambient temperature is 30 °C or higher, the battery may not be charged sufficiently.

492

Product name
Digimatic cable (1 m)
Digimatic cable (2 m)
Digimatic mini processor (DP-1VA)
Input tool (USB-ITN-SF)
Input tool (IT-020U)
Bidirectional digimatic S1 cable, Flat and straight (1 m)
Bidirectional digimatic S1 cable, Flat and straight (2 m)Foot switch
U-WAVE-R
U-WAVE-TMB (IP67 type)
U-WAVE-TMB (Buzzer type)
U-WAVE-T (IP67 type)
U-WAVE-T (Buzzer type)
U-WAVE T mounting bracket
Bidirectional Diaimatic S1 cable (160 m)

LH-Communication Tool V1.0 software for creating inspection reports and configuring system settings
You can easily create andsave inspection reports andconfigure device parameters.

Preventive maintenance
Probe diameter compensation

Receipt printer

PCPC

Adjustment of display unit

5 µm (after compensation)

Unit: mm492

996

*1

Indication accuracy*8
Repeatability*8
Accuracy
Perpendicularity(at 20 °C) (forward and backward)*9
Straightness(forward and backward)*9
Driving method(speed)
Scale unit
Measuring force
Main unit moving mode
Display unit

377

U-WAVE
PAK-BM

Resolution

LH-600FG (with grip)
518-361-11/518-361-13

377

Digimatic
outputTransmitting unit
U-WAVE-TMB

0 to 977 mm (600 mm)
0 to 38 in (24 in)
0.0001/ 0.001/ 0.01/ 0.1 mm (selectable)
0.000001/ 0.00001/ 0.0001/ 0.001 in (selectable)± (1.1 + 0.6L / 600) μm, L= Measured length (mm)Plane: 0.4 μm (2 σ), Hole: 0.9 μm (2 σ)Measuring range (Stroke)

LH-600F (without grip)
518-360-11/518-360-13

977

Software977

Wireless connection

600

Wired connection (USB)

LH-600FG
518-361-11
518-361-13
with power grip

996

Data output improves work efficiency and the reliability of recorded data

LH-600F
518-360-11
518-360-13
without power grip

mminch/mm

600

Enhanced data output functions make it easier to manage your measurement data

238

238

LH-600F (without grip)

LH-600FG (with grip)

Standard accessories
ø5 stepped probe, ball-diameter compensation block (with cover and base), auxiliary weight (2 pcs. pre-mounted), battery pack (1 pc)*12, AC adapter,
power cable for AC adapter (optional), clear cover, conveying handle, cap, hex wrench, manual set, inspection certificate, Touch pen, protective sheet,Phillips screwdriver*12: One piece included as standard. Optional additional battery (using total of two batteries) for longer battery-powered operation.

Special accessoriesAdditional battery pack (Part No.: 12AAF712), model workpiece (Part No.: 12AAA879)

1011

2303(5)Ae-(TS)NE-(JP6), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contractbasis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law.
Please consult us in advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp