Measurement Data Management System

Catalog No. E12051

Small Tool Instruments
and Data Management

U-WAVE / USB Input Tool / DP-1VA LOGGER

“Visualization of Quality” Increases theCompetitiveness of Manufacturing.
Breaking away from handwritten records and embracing centralized
management of measurement data can lead to an increase in quality andefficiency in the future.

Greater possibilities using data output
Higher Efficiency
Human
error

Human
error

Conventionally...
MeasureData is displayed

When data
output is used..
MeasureHigherEfficiency

Troublesome

Data is displayed

Manually record data

Load data to PC
by pushbutton

Input by keyboard

A stage susceptible tohuman error is removed,saving time in the processcentralized management of measurement data
Centralized Data Management Achieves
Contributes to the prevention of defects through visualization of quality.

Operation in an Excel sheet

Digitalization enables easy data collection and analysis

The data can be directly written into an Excel sheet.

The measurement data from each process can be stored
and managed centrally.

Multiple gages can be connected at the same time

Centralized DataManagement

・
・
・
・

・
・
・
・

ReceiverNote: Receiver not necessary when usingwith Mitutoyo Bluetooth® U-WAVE ﬁt

Digimatic gage

Easy to install on existing Digimatic gage (*)

Cost Reduction Effect Allows you to build a system with lower initial costs and running costs.
* Some U-WAVE-TM/TC models are not compatible.

If a Digimatic gage is damaged, operation can becontinued using another gage of the same type
The transmitter can be reconnected to another Digimatic
gage having the same connection type as the original.

Connectable to any of your existing Digimatic gages
No need to buy a replacement if your tool is equipped with
the Digimatic function.

Transmitter
Digimatic gage

Cost Reduction
EffectApproximately 400,000 continuous data transmissions are possible
Just one CR2032 lithium battery provides power for about
400,000 data transmissions.
* When the battery is used with U-WAVE-TM/TC/T. The battery life is approx. one
year under normal use when used with U-WAVE-TMB/TMC.

*We cannot guarantee that this application/software and U-WAVE TMB/TCB will connect with all Bluetooth ® equipped devices.

2

3

Steps to Achieving “Visualization of Quality”
Select your desired data communication method.

When you need easy data import to your PC via
connection cable■ USB input tool series

Pages 6 - 9, 18 - 19, 26 - 29

When you need to import data via wirelessconnection■ U-WAVE

Pages 12 - 35

When you need on-the-spot printout■ DP-1VA LOGGER

4

Pages 10 - 11, 18 - 195

Measurement Data Input Unit

USB Input Tool
Simple and robust option to collectdata from a digimatic gage.

Simply connect it to the USB portof your PC for automatic recognition,
add your digimatic data cableand gage for immediate use
If you have a Digimatic gage with an output terminal and a PC,
you can import the data into whatever record format you normally use (Excel, Word, Notepad, etc.).
The ergonomic data button is isolated from the gage where it might otherwise introduce
unwanted movement. A 3.5mm jack socket allows attachment of an optional foot switch.
Most suitable tool for taking the first step in breaking away from handwritten records and digitizing quality records.

+
Pair with the appropriate digimatic data cable to suit your gage. (Refer to page 18)

Can be used with your own Digimatic
gage with an output terminal
Measureme data
Measurement
can be sent to a PC
with a sing
single switch.

You may already own a measuring tool that can be used. See the photographs
below to check whether it has a Digimatic output connector. The connector typemay be different, depending on the measuring tool model. See the list showing
measuring tool compatibility on pages 8 to 9 and 18 to 19.

Digimatic Indicator

6

Digimatic Micrometers

Coolant-proof Digimatic
Micrometer

Digimatic Height Gage

Remove the coverto access theoutput connector.

Hardness Testing
Machine (rear view)

Production Line Surface
Roughness Measuring
Instrument7

Input Tool System Configuration/Specifications
USB Input Tool Direct (USB-ITN)

Input tool series (IT-020U/IT-007R)● Converts measurement data to keyboard signals anddirectly inputs them to spreadsheet software such as Excel
and Notepad.
● When using with optional software (see page 26), you caneasily create Microsoft Excel worksheets, enabling further
improvement of work efficiency.

IT-020U

USB-ITN

IT-007R■ IT-020U/IT-007R System Configuration
Connector

C

Digimatic indicator

B

Refer to pages 18 and 19 for details
of Digimatic ports and connectors.

Digimatic micrometer

Digimatic caliper

A
Digimatic caliper

ConnectorAB1) Icon operation
2) Function key operation
3) Foot switch operation
One of the above operations

C

D
E

Type

Model No.

Code No.

Data switch

A

USB-ITN-A

06AFM380A

㾎

B

USB-ITN-B

06AFM380B

㾎㾎

C

USB-ITN-C

06AFM380C

D

USB-ITN-D

06AFM380D

G

E

USB-ITN-E

06AFM380E

F

USB-ITN-F

06AFM380F

SF

G

USB-ITN-G

06AFM380G

SF

USB-ITN-SF

06AGQ001F

F

Foot switch (optional)
937179T

Option for USB-ITN
USB Foot Switch Adapter USB-FSW
06ADV384

Refer to pages 18 and 19 fordetails of Digimatic ports
and connectors.
System Environment
Hardware: PC with USB port (Type A)
Supporting OS (USB-ITPAK V3.0):
Windows 10 64 bit
Windows 11
Supporting OS (USB-ITPAK V2.1):
Windows2000 SP4
Windows XP SP2 or later
Windows Vista
Windows7
Windows8/8.1
Windows10
Software: Programs supporting keyboard
input(Excel, Word, memo pad, etc.)

Options
Measurement Data Collection
Software
USB-ITPAK V3.0/V2.1

Cable length: 2 m
Note 1: When using a foot switch in combination with USB-ITN, the optional USB-FSW and USB-ITPAK V3.0/V2.1 are required.
Note 2: When the measuring gage and USB-ITN have no data pushbuttons, the optional USB-ITPAK V3.0/V2.1 is required.

(Refer to pages 26 to 28 for details)

Refer to pages 32 to 35 for details of the options.

■ Main Specifications of USB-ITN
Output specification: USB2.0 or USB1.0 Supported driver software: Changeable between two types 1) Stand-alone: HID keyboard device*2) Using USB-ITPAK V3.0/V2.1: Virtual COM port (VCP) Communication speed: 12 Mbps (Full Speed) Power source: USB bus power
Mass: 59 g USB2.0 certificate Conforms to EU EMC Directive* Since this device is compatible with Windows standard driver software, dedicated driver software is not required.
Note: Information regarding USB-ITPAK V3.0 can be downloaded from our website.

8

Connect the
Digimaticconnector
to the gage

Digimatic indicator
Connector is plugged
into a USB port on a PC

Cable length: 2 m

Refer to pages 18 and 19
for details.

D

Connector
A - G, SF (8 types)+
L-shape(4 types)(1)

Connect to PC
USB terminal

Model: IT-020U
Code No.264-020(2)

RS-232C
communication
conversion type
Model: IT-007R
Code No.264-007

G

(2)
(1)

SF

Data output is available bytransmitting data request
commands from the PC.

Foot switch (optional)

Gage selector 3 (optional)

937179T

939039Cable length: 2 m

USB keyboard
signalconverter type(1)

Cable length:
1 m or 2 m

E
F

USB Input Tool Direct USB-ITN

CDigimatic
micrometer(optional)

B

B

Data loading methods
Data switch on the input tool(Type A /B/ C only)
Data switch on the measuring gage
Command transmission from a PC** Only available when using USB-ITPAK V3.0/V2.1

connecting cable

(2)

DATA

C

GAGE

■ USB-ITN System Configuration

Connect the
Digimaticconnector to
the gage

IT-020U is an easy-to-use, expandable input tool with a large data switch and a footswitch connector.
It can be connected to various types of measuring instruments using optional
connection cables.
When using with optional software (see pages 28 to 29), you can easily create
Microsoft Excel worksheets, enabling further improvement of work efficiency.
IT-007R is an RS-232C communication type input tool that can be controlled by data
request commands from a PC.

Connect theunit to a serial
port of the PC

This selector can connect up
to 3 measuring gages andswitching is available with theslide switches without changing
connecting cables
Connecting cable (length: 1 m)
between two input tools is astandard accessory (936937).

System Environment
HardwarePC with USB port (Type A)
Supporting OS (USB-ITPAK V3.0):
Windows 10 64 bit
Windows 11
Supporting OS (USB-ITPAK V2.1):
Windows2000 SP4
Windows XP SP2 or later
Windows Vista, Windows7
Windows8/8.1, Windows10
Software:
Programs supporting keyboard input(Excel, Word, memo pad, etc.)
USB-ITPAK V3.0/V2.1 (optional)
Note: Please download the information
for USB-ITPAK V3.0 from our
website.

System Environment
HardwarePC or sequencer with a serial port (D-sub 9 pin)
Software:
A program compatible with the RS-232C
communication (such as a hyper terminal) is
required.

■ Main Specifications of IT-020U
Output specification: USB2.0 or USB1.0
Supported driver software: Changeable between two types 1) Stand-alone: HID keyboard device* 2) Using USB-ITPAK V3.0/V2.1: Virtual COM port (VCP)
Communication speed: 12 Mbps (Full Speed) Power source: USB bus power USB2.0 certificate Conforms to EMC Directive* This product is compatible with the standard driver software for Windows. No dedicated driver software is required.

■ Specifications of IT-007R RS-232C Communication
Output specification: RS-232C compliant
Communication method: Full duplex
Communication speed: 2400 bps (fixed)
Bit configuration: Start bit · · ·1/
Data length · · · 8 (Most significant bit, 0 (fixed))
Prity · · · · · ·None/ Stop bit · · · · · · 1
Flow control: None
Home position: DCE (modem definition)
Data request signal
Data can be output by transmitting a character from the PC.
Data format
1) When data output

2) Error code output

Output order —>Example of format
Display
Output data
0.123 —> 01A + 0000.123CR —>−0.1234 —> 01A − 000.1234CR

D1 D2 D3 D4 D5 D6 D7 D8 D9 D10 D11 D12 D13

D1 D2 D3 D4"0 1 A" (ﬁxed)
Data parts (Floating decimal point)
Signs, "+" or "–""9 1" (ﬁxed)
CR
Error code No.
1: No data input
2: Loaded data with format other than speciﬁed

CRConnector specification and power source from the PC

Pin No.
1
2
3
4
5
6
7
8
9

This product operates while accumulating the power supplied from thePC. A second or more input interval is required.
Symbolin / out
Description of functions(N.C.)−
No connection
RXD
OUT
Data output from this product to the PC
TXD
IN
Data input from the PC to this product
DTR
IN+12 V power source from the PC*
GND−
GroundDSR
OUT
Not used
RTS
IN+12 V power source from the PC*
CTS
OUT
Not used(N.C.)−
No connection* When connecting to a sequencer, a power source is required.
Input voltage: Supplied in the range 6 V to 16 V
Power source terminal: Supplied to pins 4 and 7
Note: "4" and "6", "7" and "8" are connected with each other inside
this product.

9

Mini-Printer Equipped with Data Logging Function

DP-1VA LOGGER

Portable for on-the-spot output
This small, portable printer can be used to print measurement data from Digimatic gages and perform statistical calculations. It has a 2-way power supply system, providing excellent mobility and flexibility. Being portable, it allows you to output
measurement results from a Digimatic gage on the spot. It is a powerful tool for streamlining quality control and improving
quality.

Highly effective for quality checking

When you need on-the-spot printout

The DP-1VA LOGGER has high functionality. Not only is it capable of printing measurement data from Digimatic gages, performing various statistical calculations, and producing histograms and D charts, but it is also capable of performing complex
calculations for X-R control charts.
Up to 1,000 data items can be saved in the device using the data logger function. By connecting it to a PC with a USB cable(optional), you can transfer all saved data to your own Excel forms with a single touch.

Example of printout

Code No.
Model No.

MODE1
Various statistical calculations are executedusing all input data. If the tolerance limits
have been set, GO/±NG judgment andhistogram creation are also enabled.

MODE2

Printing method
Character specification
Printing speed

I n a d d i t i o n t o t h e M O D E1 f u n c t i o n ,measurements within the tolerance limits are
printed out as a D chart*. This chart allows
you to identify the trend of variations inmeasurement data.

Measurement data can be
printed on the spot and also besent to a PC.

Data output* D chart stands for Displacement chart.

Printing paper*2

Power supply

MODE3
MODE2

MODE3

Only input of data automatically enables
calculation processing of complex control
limit values as well as calculation for creating
an Xbaar-R control chart.

Battery life*3

Data processing capacity
MODE1

Example of batch printing log data
In OUT LOG Setting 1
This setting allows printout of measurement
time, measurement value, and GO/±NG
judgment result.
In OUT LOG In OUT LOG
Setting 2
Setting 1

In OUT LOG Setting 2This setting allows printout of measurement
In OUT LOG time, measurement value, and GO/±NG
Setting 3 judgment result.

In OUT LOG Setting 3
This setting allows printout of data number,measurement date and time, measurement
value, and GO/±NG judgment result.

Options

See video from
here

10

"d2" is the generic name for Mitutoyo Digimatic
output compatible with up to 8 digits of I/O data.

USB cable (A-microB)

Foot switch(optional)(optional)

Code No. O06AFZ050

Thermal line printer
Total number of dots: 384 dots/line
Dot size: 8 dots/mm
0.8s per line (6.5mm/s)
High durability thermo-sensitive paper, Width 58mm × length
48m
2-way power supply system
1. 100-240V 50/60Hz AC adapter (6V, 2A)
2. AA alkaline battery (LR6) or nickel-metal-hydride battery(NiMH Size AA) 4 pieces(Manganese dioxide batteries are not usable.)
About 10,000 lines(if data is printed once every 5 seconds using 1,600mA NiMH
batteries at 20ºC)
MODE0: 100,000 pieces of data
MODE1, MODE2: 9,999 pieces of data
MODE3: Sample size 10 × 9999 subgroups = 99,990 pieces ofdata
GO/±NG judgment (five sets can be defined)
Five sets can be set.

Tolerance judgment
Measurement data logging
Up to 1,000 measurements(storage)
Input timer
0.25s, 1s, 5s, 30s, 1 min, 30min, 60min
USB output
Data output
RS-232C data output at TTL levels
GO/±NG judgment result output (−NG, GO, +NG)
Clock accuracy
Maximum time difference per month: ±2 minutes
0 to 45ºC (using AC adapter)
Operating temperature
10 to 45ºC (using battery)
Storage temperature-10 to 50ºC
Mass
390g (main unit)
External dimensions
94 (W) × 201 (D) × 75.2 (H) mm
Standard accessories

Optional accessories

For details, see pages 18 to 19.

264-505*1
DP-1VA LOGGER
Digimatic input, Digimatic 2 input(specific to Mitutoyo KA counter)

Consumable items

AC adapter : 06AEG180, printing paper (one roll), strap, user’s
manual
1. USB cable (A-microB): 06AFZ050 (1m)
2. RS-232C output cable: 09EAA084 (1m, D-SUB 9 pin)
3. GO ±NG judgment cable: 965516 (2m, 10 pin terminal/
separate)
4. Foot switch: 937179T (2m)Printing paper (10 rolls)*1: To denote your AC line voltage add the following suffixes. A for North America, D for Europe, E for
UK, K for Korea, DC for China, B for Oceania without AC adapter and no suffix is required for Japan.
*2: The printer paper has excellent environmental and chemical resistance, but it has limitations indurability due to thermosensitivity. If recorded paper is stored for more than 5 years, or used as apublic document, it is recommended to make a more durable copy.
*3: The battery life quoted is not a guaranteed value, but only a typical value.

Code No. 937179T

11

Measurement Data Wireless Communication System

U-WAVE

Record measurement data wirelessly
Achieve smart measurement
Measurement results can be sent wirelessly and saved on a PC, smartphone or tablet. We have created a smart system that
does not require handwriting or manual input from a keyboard.

When you need to import data via wireless connection

What is theSeries ?

“U-WAVE”, the measurement data wireless communication system, collects data in the inspection process swiftly and accurately, and increases a company's competitiveness based on detailed data analysis.

Acquire measurement datafrom various Digimatic gages
equipped with U-WAVE.

Instantly transfer data
wirelessly to your PC,
smartphone or tablet.

Measurement data canbe sent wirelessly to a PC,
smartphone or tablet.

Wireless transfer expands measurement possibilities
Stable communication is possible up to a maximum communication distance of 16-20m*.
Efficiency is greatly improved by being able to work without worrying about cable length or snags, etc.

Freedom of movement

Flexible layout of measurement site

Human error during manual input is eliminatedStable wireless communication

Approx.

Digimatic gage + Transmitter
See videofrom here

12

You can choose from a variety of products and applications.

*May be affected by the electromagnetic environment.

16 to 20 m
ReceiverNote: Receiver not necessary when using with
Mitutoyo Bluetooth® U-WAVE fit

13

Introduction to the U-WAVE Product Lineup

U-WAVE-TMB/TCB (U-WAVE fit Bluetooth®)

U-WAVE-T
No receiver necessary

Send measurement data to your PC
The system consists of a “U-WAVE-T” transmitter that attaches to a measuring instrument with digimatic output function
and a “U-WAVE-R” receiver that connects to a PC.
It is compatible with many types of measuring instruments with digimatic output function.
PC (for storing data)

Easy connection with Bluetooth®
No receiver is required, and up to 7 devices can be connected to one PC. It is also possible to connect not only to PCs but
also to Bluetooth® compatible devices such as smartphones and tablets.
You can send measurement data through wireless transmission more easily.
These models inherit the same functions and performance of the U-WAVE-TM/TC, which are smaller, thinner, and have
improved integration with Digimatic gages and improved operability.
Furthermore, we also have a lineup of dustproof and water-proof IP67 models to withstand on-site environments.

Import measurement
data to your PC
wirelessly

U-WAVE-T
U-WAVE-TMB(Transmitter) U-WAVE-T+ Digimatic gage

(Receiver) U-WAVE-R
PC/smartphone/tablet (for storing data)
U-WAVE-TCB(Transmitter) U-WAVE-TMB or WAVE-TCB+ Digimatic gage

U-WAVE-TM/TC (U-WAVE fit)

Fits onto Digimatic gages for improved
operability* Connectivity of the dedicated Mitutoyo Bluetooth® U-WAVE application and U-WAVE-TMB/TCB to every single Bluetooth® device is not guaranteed.
* If using U-WAVE/U-WAVE-fit, please note that it does not support bidirectional serial communication.
* Only some models are IP67 compatible. For details, see pages 19 to 22.

While inheriting the functions and performance of U-WAVE, these models have been made smaller and thinner, and have
improved operability, being designed specifically for use with small digimatic calipers and small digimatic micrometers.
PC (for storing data)

Function comparison table

U-WAVE-TM

Import measurement
data to your PC
wirelessly(Transmitter) U-WAVE-TMB or WAVE-TCB+ Digimatic gage

14

(Receiver) U-WAVE-R

U-WAVE fit Bluetooth® *1

Transmission method

Original <based on IEEE802.15.4 (2.4 GHz)>

Bluetooth® 4.2

Communication distanceApprox. 20m (line of sight)

Approx. 16m (line of sight)

Connectible model

U-WAVE-TC

U-WAVE fit*1

U-WAVE

Dedicated application/
softwareDigimatic gages

Digimatic calipers, micrometers,indicators and linear Height

U-WAVEPAK
USB-ITPAK*2

U-WAVEPAK-BW
U-WAVEPAK-BM
U-WAVE Navi
USB-ITPAK*2*1 Please check the list of compatible models since the unit may not be attachable to some models.
*2 Please note that USB-ITPAK may not be recognized if your computer's OS build is old.

15

Compatibility Table for Digimatic Gages and U-WAVE

Note1: U-WAVE fi t and Bluetooth ® U-WAVE fi t can only be connected to digimatic calipers, digimatic micrometers, and some digimatic indicators witha measurement range of 0 to 300 mm.
Refer to the lists of compatible models on pages 32 to 35 for details.
Note2: U-WAVE should only be used in countries where they have obtained wireless certifi cation.

Product combinations
Appearance when fitted together

Digimatic gage

Connecting unit/connecting cable

Transmitter
Buzzer type

Standard type

U-WAVE-TM
264-623

Digimatic
micrometers

Digimatic
micrometers

ReceiverHigh lead typeor
02AZF310

IP67 type

U-WAVE-TM
264-622
Standard type

Digimatic caliper

Digimatic caliper

Standard type

Coolant-proof type

Buzzer type+

02AZF300

U-WAVE-TC
264-621+

IP67 type

ID-CNX/ID-FNX
seriesAppearance when fitted together

Buzzer type

Connecting unit/connecting cable

U-WAVE-T
02AZD730G

Transmitter
Buzzer type

U-WAVE-TMB
264-627

Digimatic
micrometers

Digimatic
micrometers

Dust/water-proof typeNo receiver necessaryStandard typeHigh lead type

U-WAVE-TM
264-622

U-WAVE-T
02AZD880G

BracketDigimatic gageor

U-WAVE-TM
264-623

02AZF700

Other
Connecting
cable

Product combinations

IP67 type

Buzzer type(12.7mm type only)or

Digimatic indicator,
Other

Digimatic indicator,
Other

Other

(12.7mm type only)

U-WAVE-R
02AZD810D

Designregisteredin Japan

U-WAVE-TC
264-620

02AZF310
ID-CNX/ID-FNX series+orCoolant-proof
type

or
02AZF310

IP67 type

Standard type

Coolant-proof type

Standard type

Digimatic caliper

Digimatic caliper

U-WAVE-TMB
264-626+

Buzzer type
02AZF300

Coolant-proof
type

Digimatic indicator

Digimatic indicator

16(12.7mm type only)or
IP67 type

02AZF310
ID-CNX/ID-FNX series+

U-WAVE-TCB
264-625

U-WAVE-TCB
264-624
Buzzer type

U-WAVE-TMB
264-627

ID-CNX/ID-FNX
series(12.7mm type only)or
02AZF700

IP67 type

U-WAVE-TMB
264-626

17

[U-WAVE] [USB input tool] [DP-1VA LOGGER]

Compatibility Lists of Digimatic Gages and Connecting Cables(1)

U-WAVE-T(2)

L-shape type with

Connector type

Standard

For foot switch

type
Water-proof type
Straight type withbutton,A) Water-proofCR) output
D) Flat 10-pin typewith output button B) with output button C) output button
cable outlet on right
02AZD790A

02AZE140A

02AZD790B

02AZE140B

02AZD790C

02AZE140C(3)USB Input Tool Direct
USB-ITN

IT-020U/IT-007R/DP-1VA
LOGGER/MUX-10F/EC Counter Connector type

Connector (12 types, A to G and SF)
Type D on the other end for allmodels02AZD790D

02AZE140D

type
Water-proof type
Straight type withbutton,A) Water-proofCR) output
D) Flat 10-pin typewith output button B) with output button C) output button
cable outlet on right
USB-ITN-A
06AFM380A

USB-ITN-B
06AFM380B

USB-ITN-C
06AFM380C

No applicable models
USB-ITN-C can beused. See figure
below

02AZD790E

02AZD790F

02AZD790G

02AZG011

02AZE140G

02AZG021

No applicable models
Use 02AZD790F or 02AZD140F.

L-shape type with

Connector type

Model No.
Code No.

+No applicable models
Type C can be used,
but be careful of thecable when operating
the thimble. (See figure
below)

L-shape (cable
L-shape (cable
L-shape (cablestraight
Straight standardE) Round 6-pin type F) Flat straight type FB) Flat
FR) FlatFL) FlatG) Flatoutlet is back)outlet is right)
outlet is left)
water-proof type SF) typeUSB-ITN-D
06AFM380D

L-shape type withtype
Water-proof type
Straight type withbutton,A) Water-proofCR) output
D) Flat 10-pin typewith output button B) with output button C) output button
cable outlet on right

02AZE140E

02AZE140F

L-shape (cable
L-shape (cable
L-shape (cablestraight
Straight standardE) Round 6-pin type F) Flat straight type FB) Flat
FR) FlatFL) FlatG) Flatoutlet is back)outlet is right)
outlet is left)
water-proof type SF) typeUSB-ITN-E
06AFM380E

USB-ITN-F
06AFM380F

No applicable models
USB-ITN-F can be used.

USB-ITN-G
06AFM380G

USB-ITN-SF
06AGQ001F

L-shape (cable
L-shape (cable
L-shape (cablestraight
Straight standardE) Round 6-pin type F) Flat straight type FB) Flat
FR) FlatFL) FlatG) Flatoutlet is back)outlet is right)
outlet is left)
water-proof type SF) type1m

05CZA624

05CZA662

95914904AZB51293693793738790533890568990569190569321EAA194

06AGL011

2m

05CZA625

05CZA663

95915004AZB513

96501496501390540990569090569290569421EAA190

06AGL021

Code No.

For connector types (circled in broken lines), select the model (A to G, SF) that matches the connector shape of the Digimatic gage you are using.

Note 1: USB-ITN is required to use in the combination with USB-ITPAK, when connected to D) ID-F, D) digital Mu-checker, EB, EC-101D, and F) ID-U, ID-SS, ID-SX.
Note 2: USB-ITN, IT-020U, and U-WAVE-T cannot be used with D) EF/EH and D)VL-B/S-B.
Note 3: USB-ITN and U-WAVE-T cannot be used with D) SJ-500/SV-2100.
Note 4: IT-020U and USB-ITN-SF cannot be used with D) SJ-210 and D) EB.

L-shape type with

Connector type

type
Water-proof type
Straight type withbutton,A) Water-proofCR) output
D) Flat 10-pin typewith output button B) with output button C) output button
cable outlet on rightL-shape (cable
L-shape (cable
L-shape (cablestraight
Straight standardE) Round 6-pin type F) Flat straight type FB) Flat
FR) FlatFL) FlatG) Flatoutlet is back)outlet is back)outlet is back)water-proof type SF) type

Cable
Gage connectors on data cable Connector shapeon Digimatic
gage side
Data switch

Digimatic ports on gage

Please note that some highprecision Digimatic gages arecapable of displaying the measurement result to more than
6 digits. However, according to
the Digimatic output specification, the result may be output
in 6 digits only.
Digimatic gages whose display
may exceed 6 digits• Laser Scan Micrometers• Litematic• Linear gage counter (EH)

18

㾎

㾎

㾎

• Digimatic caliper
CD67-S_PM
CD-PMX
CD-PM
CDC-P_PMX
CDN-P_PMX
CFC-G
CFC-GLCFC-GC
CFC-GUNTD10B-P_C
NTD10PB-P_C
NTD-PMX• Digimatic depth gage
VDS-PMX• Digimatic scale unit
SD-G

• Digimatic micrometer
MDH-25M/25MC
MDE-MJ/MX
MDC-MJ/MX
OMC-MX
MDQ-MX
CLM1-QMX
CLM2-QMX
CLM1-DKX
CLM2-DKXOther models ending
with -MX• Digimatic
micrometer head
MHN-MX/MXN• Digimatic hole test
HTD-R
• Digimatic depth gage
DMC-MX• Digimatic caliper
CD-AX
CD-C
CD-SC/SCTNTD25-AX
NTD31-AX• Digimatic depth gage
VDS-AX• Digimatic
micrometer headMHD-50MB• Digimatic scale unit
SD-D
SDV-D

–

㾎

–

–

–

–

–

–

–

• Digimatic indicator
ID-N
ID-B

• Digimatic indicator
ID-CNX
ID-FNX• Digimatic depth gage
Digimatic type
(ID-CNX)• Digimatic thickness
gage
Digimatic type
(ID-CNX)

Picture of
Digimatic port

Applicablemodels• Digimatic
micrometer
MDC-MBOMC-MB
PMU300-MB

L-shape

Straight

Type C straightconnectors areavailable, but may
interfere with thimble
operation.

• Digimatic indicator
ID-H
ID-F (Note 1)• High-precision height gage
QMH-AX/BX• Mu-checker
Digital Mu Checker• Laser scan
micrometer
LSM-9506• Linear gage counter
EF/EH (Note 2)
EB (Note 1) and (Note 4),EC-101D (Note 1)• Litematic
VL-B/S-B (Note 2)• Surface roughness
measuring machine
SJ-210 (Note 4)/
310/410
SJ-500/SV-2100 (Note 3)• Hardness testing machines
HM-210/220
HV-110/120
HR-530HR-600[Hardness testingmachines]
HM-102
HM-103HM-320HM-430HM-411• Digimatic indicator
ID-CX, ID-C112AX, ID-C112RXB, ID-C112GXB, ID-U (Note 1), ID-SS (Note 1), ID-SX (Note 1)
HDM-AX, HD-AX, HDS-HC, HDS-C• ABS borematic• Digimatic cylinder gage
SBM-CXCG-D• Scale unit
• Hardness testing machines
SD-E, SDV-E, SD-F, SDV-F
HH-300• Digimatic depth gage
Digimatic type (ID-CX)[How to check the code No. of the connecting cable used for the measuring instrument]
For models currently being sold
Please check the optional accessories for the applicable Digimatic gage in the general catalog.
The general catalog can also be viewed on the Mitutoyo website. Web address: https://www.mitutoyo.co.jp/
Regarding discontinued models not listed in the general catalog(1) How to look for items from the general catalog
For a digimatic indicator, you can check the “New and Old Product Compatibility Table” for compatible connecting
cables.
(2) How to look for items from the Mitutoyo website
You can search the “Parts List”. Enter the code number of the Digimatic gage you want to search for.
Example: 293-421, 500-151, etc.

19

[U-WAVE] Specifications
Wireless Communication Specifications
Product nameCompatible standards

Wireless communication

Wireless communication distanceTransmission output

Modulation method

Communication frequency

Used band

U-WAVE-T/TM/TC

ARIB STD-T66 (Japan)

Original (based on IEEE802.15.4)

Approx. 20 m (line of sight)

U-WAVE-T: 1 mW (0 dBm) or less
U-WAVE-TC/TM: 2.5mW (4dBm) or less

DS-SS (Direct Sequence - Spread
Spectrum)Resistant to interfering signals
and noise

2.4 GHz band

15 channels(2.405 to 2.475 GHz at intervals of 5
MHz)

U-WAVE-TMB/TCB

ARIB STD-T66 (Japan)

Bluetooth® 4.2 Low Energy

Approx. 16 m (line of sight)
Approx. 16 m (on shop fl oor)

3.2 mW (5 dBm) or less (Class2)

FH-SS (Frequency-Hopping
Spread Spectrum)

2.4 GHz band

In accordance with FH-SS(2.400 to 2.483 GHz)

* Not compatible with the previous product Mu-Wave because the communication specifications are different.

* U-WAVE should only be used in countries where they have obtained wireless certification.
* “U-WAVE-R/T/U-WAVE fit” and the Bluetooth® type “U-WAVE-TCB/TMB” are not compatible because they have different communication specifications.

Transmitter (For combinations, refer to pages 16 to 17.)
Product name

Model

Code No.

Protection level

Data reception
indication

Power supply

Battery life

Mass

Appearance

External dimensions

Unit: mm
Buzzer type264-623–

18 g

LED/Buzzer

Dust/water-proof
type

264-622

IP67

LED

38.8

Approximately 400,000 transmissions
in continuous use

U-WAVE-TM

18 g

41.9

12.9

Unit: mm264-627–

LED/Buzzer

18 gApproximately 1 year under normal
conditions of use, but varies
according to usage.

U-WAVE-TMB
Dust/water-proof
type

264-626

IP67

LED

38.8

Buzzer type

18 g

41.9

12.9

Unit: mm–

18 g

LED/Buzzer
Lithium battery
CR2032 ×1

U-WAVE-TC
Dust/water-proof
type

264-620

IP67

56

Approximately 400,000 transmissions
in continuous use

LED

30.4

264-621

18 g

11.45

Buzzer type

Unit: mm264-625–

LED/Buzzer

264-624

IP67

LED

56
30.4

Approximately 1 year under normal
conditions of use, but varies
according to usage.

U-WAVE-TCB
Dust/water-proof
type

18 g

18 g

11.45

Buzzer type

Unit: mm–

Dust/water-proof
type

02AZD730G

IP67

LED

44
29.6

Approximately 400,000 transmissions
in continuous use

U-WAVE-T

20

23 g

LED/Buzzer

23 g
18.5

02AZD880G

16.6

Buzzer type21

[U-WAVE] Specifications/Options
Connecting unit (Refer to pages 32 to 35.)
Product name

Model

Code No.

Protection level

Mass

Appearance

Dimensions
Unit: mm

26.25

6g

(12.75)NA(0.9)

02AZF300

17.65

For standard type

21.8

(19.6)

26.25

Unit: mm

ID-CNX series
ID-FNX series
12.7 mm typeexclusive

21.8
27.15
(12.75)

6g

28.5

IP67

(19.3)

02AZF310

7

For dust/water-proof type

Unit: mm

02AZF700

NA

10 g

19.3

43.5
(12.1)

Connecting unit

(14.8)(19.6)

27.15

Receiver (Refer to pages 16 to 17.)
Model
No.

Product name

Code No.

Power
supplyU-WAVE-R
U-WAVE-T
Number of connected
Number ofdevices (per PC)
connected devices

Mass

AppearanceStandard accessory: U-WAVEPAK software

Dimensions

System Environment: Compatible OS Main specifi cations

Data format

Unit: mm Windows 2000 Professional (SP4 or higher)* ・Setup of dedicated

U-WAVE-R

U-WAVE-R 02AZD810D

USB buspower
systemUp to 15

Up to 100

130 g
80

140

Example of format when the Digimatic gage
Windows XP Home Edition (SP2 or higher)* driver software (USB displays 12.34
Windows XP Professional (SP2 or higher)*and virtual COM
DT1 01 02 +00000012.34 M
Windows Vista*
port)
HeaderDigimatic gage
Unit M: mm
DT1: Measurement data
・Initial setting of ID
Windows 7*
display value *1
ST1: Status code
number andWindows 8 / 8.1*
U-WAVE-TC/TM/T ID: 00 to 99
U-WAVE-R ID: 00 to 99
frequency selection
Windows 10*(required only once *1: Data interface function is switchable to “Measurement value only”
Windows 11 Ver.1022B or later
for the fi rst time)* 32-bit, 64-bit OS supported <Version
Example of status code format
confi rmed to work with Windows 10>
・Load data to
ST1 01 02 0999999073 99
Microsoft Excel or· U-WAVEPAK Ver.1.020 or later
HeaderNotepad through<Connectability confi rmed for tablet PC>
Status code
Device ID *2
Status code
99: Data cancellation
the data interface
31.6 · Microsoft Surface Pro 6 (the version whose01:
No response from
U-WAVE-R
ID
operation on Windows10 Professional isfunction
Digimatic gageconfi rmed)· Required environment: DVD drive (forinstallation), USB
ports x 2 or more

U-WAVE-TC/TM/T ID

03: Measurement data
missing, etc.

*2: Unique number assigned to U-WAVE shipment

Note: U-WAVEPAK software is packed together with the U-Wave receiver. If your computer has no DVD drive then please download and install U-WAVEPAK from our website.

Options

For details, see pages 18 to 19.

Standard

For foot switch

Details regarding connecting cables
are provided.

Connector type
A to G

22

Foot switch (optional)

U-WAVE-T

Foot switch
Digimatic gage

Connector type
A to G

Code No. 937179T

23

[U-WAVE] Options (Brackets/Attachment Fixture)
Installation bracket Kit for U-WAVE-T
Product nameApplication
exampleApplication examples of the mounting plate (02AZE200)
Type

Code No.

Attachable unit

Accessory

Digimatic indicator (ID-C0512NXB)
Front view

Bracket for
U-WAVE-T

Standard

02AZE200

U-WAVE-T

Image of fastener attachment to main
gage

Rear view

Side view

In the case of an indicator

Detachable fastener: 1 set
Mounting screw: 2 pcs.

ID-C112XB
A back plate without a center lug
is recommended. If a center lug ispresent, attach fastener after cutting
away material to clear the lug.

Dedicated for digimatic
indicators

02AZF670

ID-C
Front view

U-WAVE-TM/
TMB
BracketU-WAVE-TM/TMB

Mounting screw: 4 pcs.

Attachable to ID-C in the various positions

Dedicated for ABS coolantproof digimatic indicator 02AZF675
ID-N

24

Bracket for
QM-Height

Dedicated for QM-Hight

02AZE990

U-WAVE-T

Detachable fastener: 2pcs.
Mounting screw: 2 pcs.

Attachment
fi xture for LH

Linear Height LH-600F/FG 12AAY486

U-WAVE-T

Mounting screw: 4 pcs.

QM-Height

LH-600F/FG

25

Excel-specific Measurement Data Collection Software and Measurement Examples
USB-ITPAK V3.0 measurement examples

Simultaneous measurement

The following are three measurement methods that can be set with USB-ITPAK V3.0, illustrated with specific examples.

Example of measurement involving importing data from multiple Digimatic gages at once

<Reference> The user’s manual for USB-ITPAK V3.0 is posted on our website. You can find detailed information such as setting procedures.

(Measurement
example)Batch (simultaneous)measurement of 4
U-WAVE-R
heights (A to D) of
the workpieceshown below.

http://www.mitutoyo.co.jp

Sequential measurement
Example of measurement using input according to pre-registered procedures(Measurement example)
The outside diameters at X
and Y and length (H) of 5
workpieces are measured inorder, as shown on the right.
Finally, a visual inspection
is performed (for scratches,
uneven coloring, etc.) todetermine Go/±NG judgment.

USB-ITN-F ×2 pcs.

Y

U-WAVE-T ×2 pcs.

H

U-WAVE-R

USB hub(Commercial item)

2) Measure length H of
5 workpieces.

USB-FSW ×2 pcs.
+
Foot switch ×2 pcs.

X

U-WAVE-T,
U-WAVE-TM/TC

1) Measure outside
diameter at X and Y
of 5 workpieces witha micrometer.

USB hub (Commercial item)

USB-ITN-C

Height A

B

C

Data
Data
request cancellation

D

A

B
C
D
E
Height A Height B Height C Height D

1
2

1

5.02

8.03

9.96

6.03

3

2

4.98

8.02

10.01

5.99

4

3

4.97

8.04

10.07

5.96

5

4

6

5

First measurement(finished)
Second measurement(finished)
Third measurement(finished)
Fourth measurement(Wait for next input)Note) When using U-WAVE for batch measurement, it will operate in “Event Drive mode.”

USB-FSW ×4 pcs.
+
Foot switch ×4 pcs.

Individual measurement
Example of simultaneous work by multiple operators

3) Inspect external viewto check if there are
any scratches or color
shading and input“OK” or “NG”.

DatacancellationA

USB hub* No data request for U-WAVE.

When a measuring procedure is executed, a window is displayed. “Data request*”, “Data cancel*”, “Data skip*”,“Aborting”, “Complete” can be specified.

B

U-WAVE-R

Character Character
Data
strings strings Input request*
Input “OK”
“NG”

C(Commercial item)

Operator 1(Measurement example)
Two operators measure
6 workpieces
3 workpieces per operator

Operator 2

USB-ITN-C

U-WAVE-T

1

A

B

C

D

E

F

G

Setting

1

2

3

4

5

6

2 Dimension A 10.02 10.03* These operations can be allocated to the function key or foot switch (via USB-FSW).

3 Dimension B 9.98

Workpiece Nos. 1 to 3

Data
Data
request cancellation

Workpiece Nos. 4 to 4

The last data input may becanceled by pressing andholding the data switch

9.96

10.01 10.07

4 Dimension C 10.15 10.14

USB-FSW ×2 pcs. + Foot switch ×2 pcs.

10.15 10.23 10.04
9.99

9.78

9.96

10.27

Cell that will

Operator 1 receive next
input

Cell that will

Operator 2 receive next
input

Notes on using USB-ITPAK V3.0/V2.1:
Cell movement direction after inputting data
(down and right)
A

B

C

D

E

F

Setting

1

2

3

4

5

2 Dimension X

10.02510.0339.964

10.03110.046

3 Dimension Y

9.982

10.01710.0089.996

10.027

4 Dimension H

29.97

30.02

30.07

29.96

30.04

External
appearance

OK

OK

NG

1

5

Carriage return (Low, column)
Microsoft Excel sheet previously
specified
Input range of micrometer(B2 to F3)
Input range of caliper(B4 to F4)
Input range of visual judgment(B5 to F5)

Cell that will receive next input is highlighted in green.

*These are common options for IT-020U, USB-ITN, and U-WAVE. They cannot be used with IT-007R.

26

Do not merge the cells in the specified range as a measurement data input.
During measurement, the Microsoft Excel worksheet cannot be modified in any way apart from entering data.
If you need to modify the sheet, it is necessary to abort or finish the measurement.
If the OS build version is old, it may not be possible to use U-WAVE fit Bluetooth® and U-WAVE fit or U-WAVE-T together.

Option for USB-ITPAKUSB Foot Switch Adapter USB-FSW
This is the necessary USB adapter for connecting to a PC when performing data control (data request, data cancellation, etc.) using a footswitch (No.937179T) with USB-ITPAK. It must be used in combination with USB-ITPAK in order to function.
Foot switch (Code No. 937179T)

Main Specifications of USB-FSW
You can use USB-ITPAK to set different purposes for the foot switch according to intended use.
Data control “data request,” “data cancel,” “data skip”
Enter any particular character string, e.g., “Pass”, “Fail”, etc.
Note: As with USB-ITN, USB-FSW also uses a built-in VCP driver.

USB-FSW appearance
Total length 160 mm

USB-FSW

Code No. 06ADV384

27

Software
Useful functions of USB-ITPAKCan be combined with a bidirectional-communicationsupporting Digimatic gage for even greater work efficiency

This is optional software that allows you to create procedures for inputting measurement data sent from U-WAVE, input
tools, etc. into Excel. With this, you can reduce the time and effort required to create inspection sheets and further
streamline inspection work.

Equipped with an automatic sorting function for sorting input measurement data [Easy input mode]This function can be implemented even if the measuring instrument
does not support bidirectional serial communication. After setting,measurement values are automatically sorted into an Excel sheet as
needed.
Only the number of measurement items isWith normal input

With automatic sorting function(Entered into column
A only.)(Once entered into column A, similar data is
automatically classified.)

Control and setting of the main unit, collection of gage information, etc. can be performed as a batch operation from a PC.
This contributes to drastic improvement in work efficiency.
・An optional cable and measurement data input unit are required for bidirectional serial communication.
・USB-ITPAK V3.0 must be installed on the PC used for communication.

Example of ID-C/F bidirectional communication support

preset.
(Example: number of measurement items = 3)

D: 10 mm
H: 20 mm
W: 30 mm

0.000

0.000
0.000

Settings
Save

0.000
0.000

Details

Windows 2000 SP4, Windows XP SP2 or later, Windows Vista, Windows 7, Windows 8 / 8.1

Functions28

・ Batch zero setting and power ON/

・ Various functions of ID-C/ID-F units

V3.0
V2.1

Trial
Full
version(free) version

can be set from USB-ITPAK.

・ The contents of various function

settings can be saved on a PC, andthe same settings can be copied toother ID-C/ID-F units.
Reduced work time for setting

Old model(Old ID-C/ID-F + USB-ITPAK V2.1)

Old model(Old ID-C/ID-F + USB-ITPAK V2.1)

・ For older ID-C/ID-F units that do・ Since bidirectional serial

not support bidirectional serial
communication, zero setting mustbe performed individually on each
Digimatic gage.

Symbols: ✓1: Can be used only when connected to a device supporting USB-ITPAK V3.0 Digimatic S1,✓: Can be used, —: Cannot be used

d1: 1st generation, unidirectional communication, 6-digit communication / d2: 2nd generation,
Digimatic d1/d2
Supported
unidirectional communication, 8-digit communication
communication
standard
Digimatic S1
3rd generation, bidirectional serial communication, 8-digit communication

Compatible OS

New model(ID-C/ID-F + USB-ITPAK V3.0)

Zero set
Presetting
Resolution
Tolerance setting...etc.

ITPAKOperating environment andfunctions, etc.

New model(ID-C/ID-F + USB-ITPAK V3.0)

Deploy settings to other ID-C/ID-F units

0.000

USB-ITPAK V3.0/V2.1 Function comparison table

Function example (2)Measuring instrument setting

OFF operation, etc. of multiple ID-C/
ID-F units is made possible by use of
the dedicated software “USB-ITPAK.”

Simplifies measuring instrument setting
Batch setting of ID-C/ID-F units can be
performed from your PC. Moreover, thesettings can be saved on your PC and set toother ID-C/ID-F units. You can perform settings
without even touching the ID-C/ID-F units.

Function example (1)Control of ID-C/ID-F from PC

communication is not supported,function setting from a PC is notpossible.

Zero set Zero set Zero set

✓
—

✓

✓

—

・USB-ITPAK V3.0
Use this QR code to
download the software.

✓

Windows10

・USB-ITPAK V2.1

USB dongle

USB dongle

Can only be used with a PC
connected to a USB dongle

Can only be used with a PC
connected to a USB dongle✓

Windows11—

Sequential
measurement

With this method, when using one or several measuring instruments, the measurement data are input into
an Excel sheet from the measuring instrument(s) registered in advance.

✓

—

✓

Batch
measurement

With this method, measurement data are acquired in batch from several measuring instruments and input
into an Excel sheet.

✓

—

✓

Individual
measurement

The Excel sheets and cells for inputting measurement data are set individually for each measuring instrument.
With this method, measurements performed randomly by multiple operators can be input from each
instrument into their specified sheets and cells.

✓

—

✓

Simplemeasurement
function

This function makes it possible to start measuring without prior detailed settings and to sort data into Excelcolumns according to measurement location.

—✓

Measuring
instrument
setting

This function is used to change the various settings (zero setting, registration of preset values, setting of unit,
counting direction, and tolerance) of connected measuring instruments.

—✓1

Measurement
historyThis function saves information on the measurement operator and the measurement equipment used within
the measurement data.
(It records in the data who used what to measure the data.)—✓1

Deviceinformation

This function reads various information about connected measuring instruments (model, serial No.,calibration date) and displays it on the PC.

—✓1

Data input into
Microsoft Excel

This function is used to input values into user-specified cells in Excel.

✓

—

✓

Text data input
with virtual
keyboard

This function is used to input text (characters and values) into specified cells in Excel.

✓

—

✓

Code No. 06AGR543

Code No. 06AFM386

29

Measurement Navigation Applications
Mitutoyo U-WAVE Navi

Using the application, create ameasurement procedure, display
and navigate the measurement,
and manage the measurement
results

U-WAVEPAK-BM

Measure a workpiece to performa simple trend management

U-WAVEPAK-BW

Up to seven units of U-WAVE-TCB/
TMB can be set up. Measuring
results can be transferred to Excel
spread sheet.*

Introduction to the MeasurLink Software
What is?

See videofrom here.

Achieve “Visualization of Quality”
Collecting the measurement data

IoT of Quality Control

Measurement Data Wireless Communication System

Measurement Data Network System

U-WAVE / USB Input Tool

MeasurLink

MeasurLink is an IoT platform for quality management that realizes “Visualization of Quality” by enabling real-time datacollection from the networked Digimatic gages and global control and analysis. U-WAVE supports MeasurLink as an
infrastructure that collects and controls data.

PurposePreventing defectives
Collects data from the Digimatic gages on the network and performs
statistical process control (SPC) to warn of possible generation of defectives.

Diagnosis by data analysis
Checking measurement results by accessing the data base and performing
various analyses helps investigate and resolve process performance concerns.
・ Create / perform a measurement

Possible
actions(Functions)

Displaylanguageprocedure (including GO/NG
judgement)
・ Navigate a measurement procedure
・ Manage/ transfer a measurement
procedure・ Display a list of measurement results
・ Transfer a measurement result
Japanese / English (Depends on the OS
settings)

・ Judgment
・ Data logging
・ Graphical display of measurement

result・ Display the histogram of
measurement results
・ Transfer a measurement result (log
data)

Simply start achieving IoT
Identification of data origin
Upper application (connection with
USB-ITPAK, MeasurLink)

Integration of U-WAVE/USB input tool and MeasurLink
Japanese / English (Depends on the OS
settings)

EnglishIn addition to conventional data storage, the network can be
confi gured in steps to simply start IoT of Quality Control.

Building a MeasurLink measurement data network system based on U-WAVE and USB input tools
Data Collection Software: MeasurLink Real-Time

検査室room
Inspection

Android 7.0 or later(iOS not supported)

Android 7.0 or later / iOS 10.0 or later
U-WAVEPAK-BM(English version only)
can be downloaded
for free from eachapp store.

Available from
Google Play for freedownload.

Compatible
OS

U-WAVE NaviThis is SPC software that collects data from various measuring instruments
and displays statistical processing data such as control charts, histograms, andprocess capability indexes in real time.

Windows10 Pro 64bit

Process Management for Administrators: MeasurLink Process Manager

DL from the following URL
https://www2.mitutoyo.co.jp/eng/
contact/products/u-wave/index2.html

This administrative software enables centralized monitoring of information fromall MeasurLink data collection terminals networked together on the shop fl oor.

Shop floor

This is administrative software for accessing the database where measurement
data collected by MeasurLink Real-Time is stored, confirming measurement
results, and performing statistical analysis.

MeasurLink
DatabaseserverU-WAVEPAK-BM

Note1: We cannot guarantee the operation of this application and U-WAVE TMB/TCB with all Bluetooth ® equipped devices.

Process Analysis Module for Administrators: MeasurLink Process Analyzer

Network

Gage Management Software: MeasurLink Gage Management
This software plans and implements a complete calibration schedule andincorporates a powerful retrieval function in addition to recording and managing
the operational status of gages.

U-WAVE

Evaluation/Analysis Software for Measurement System
Analysis (MSA): MeasurLink Gage R&R

Note2: Google Play and the Google Play logo are trademarks of Google LLC. Apple and the Apple logo are trademarks of Apple Inc.
Inspection room

* By enabling the direct input feature within U-WAVEPAK-BW the results can be written directly to Excel. The results do not include the transmitter
device ID. Use of our optional USB-ITPAK or MeasurLink software allows for data sorting by individual transmitter IDs.

Measurement
data

30

Collect
dataThis is evaluation and analysis software compliant with the MSA* required by
IATF16949.
*Measurement System Analysis

Unify

Analyze

Visualize

31

Connection Unit
No.02AZF310

U-WAVE-TM/TMB compatible gages

U-WAVE-TM/TMB compatible gages

No.02AZF310
Product name

Code No.
Unit Current product293-140293-140-30
㾎
293-141
293-141-30㾎
mm
293-142293-142-30
㾎293-143
293-143-30
㾎QuantuMike
293-180
293-180-30
㾎293-181293-181-30
㾎inch/mm
293-182293-182-30
㾎293-183293-183-30
㾎293-230293-230-30
㾎
293-231293-231-30
㾎
293-232293-232-30
㾎
293-233
293-233-30
㾎
293-234
293-234-30
㾎
293-235
293-235-30
㾎
293-236293-236-30
㾎
293-237293-237-30
㾎
293-238293-250
293-250-10
293-250-30
㾎293-251mm
293-251-10
293-251-30
㾎
293-252
293-252-10
293-252-30
㾎293-253
293-253-10
Coolant- 293-253-30
㾎
proof
293-254
micrometer
293-254-10
293-254-30
㾎293-255
293-255-10
293-255-30
㾎
293-256
293-256-10
293-256-30
㾎
293-257
293-257-10
293-257-30
㾎293-330293-330-30
㾎
293-331293-331-30
㾎
293-332293-332-30
㾎
293-333
293-333-30
㾎inch/mm
293-334
293-334-30
㾎
293-335
293-335-30
㾎
293-336293-336-30
㾎
293-350
293-350-10
293-350-30
㾎32

Current productNo.02AZF310
Product name

Code No.
Unit Current product293-351
293-351-10
293-351-30
㾎293-352
293-352-10
293-352-30
㾎293-353
293-353-10
293-353-30
㾎
293-354293-354-10
293-354-30
㾎inch/mm
293-355
293-355-10
293-355-30
㾎Coolant- 293-356
proof
293-356-10
micrometer 293-356-30
㾎
293-357
293-357-10
293-357-30
㾎
293-960
293-960-30
㾎
293-961293-961-30
㾎
293-962
293-962-30
㾎
293-963
293-963-30㾎
mm
293-964
293-965
293-966
293-966-30
㾎
314-251314-251-10
314-251-30
㾎
314-252314-252-10
314-252-30
㾎
314-253314-253-10 mm
314-253-30
㾎
314-261314-261-10
314-261-30
㾎
314-262314-262-10
314-262-30
㾎V-anvilmicrometer 314-351
314-351-10
314-351-30
㾎
314-352314-352-10
314-352-30
㾎
314-353314-353-10 inch/mm
314-353-30
㾎
314-361314-361-10
314-361-30
㾎
314-362314-362-10
314-362-30
㾎
317-251
317-251-30㾎
mm
317-252
317-252-30
㾎
317-351
Uni317-351-30
㾎micrometerinch/mm317-352
317-352-30
㾎323-250
323-250-30 mm
㾎323-251

Product name

Code No.
Unit Current product323-251-30
㾎
323-252323-252-30 mm
㾎323-253
323-253-30
㾎323-350Uni㾎
micrometer 323-350-30
323-351
323-351-30
㾎inch/mm323-352323-352-30
㾎323-353
323-353-30
㾎
324-251324-251-10
324-251-30
㾎
324-252324-252-10
324-252-30
㾎
mm
324-253324-253-10
324-253-30
㾎
324-254324-254-10
㾎Gear tooth 324-254-30
micrometer 324-351
324-351-10
324-351-30
㾎
324-352324-352-10
324-352-30
㾎inch/mm324-353324-353-10
324-353-30
㾎
324-354324-354-10
324-354-30
㾎
326-251326-251-10
326-251-30
㾎
326-252326-252-10
326-252-30
㾎
mm
326-253
326-253-10
326-253-30
㾎
326-254326-254-10
㾎Screw thread 326-254-30
micrometer 326-351
326-351-10
326-351-30
㾎
326-352326-352-10
326-352-30
㾎inch/mm326-353326-353-10
326-353-30
㾎
326-354326-354-10
326-354-30
㾎
329-250
329-250-10
329-250-30㾎
mm
329-251329-251-10
329-251-30
㾎Depth
micrometer 329-350
329-350-10
329-350-30
㾎inch/mm329-351329-351-10
329-351-30
㾎Product name

Code No.
Unit Current product331-251331-251-30
㾎
331-252
331-252-30
㾎
331-253
331-253-30
㾎
331-254331-254-30㾎
mm
331-261331-261-30
㾎
331-262331-262-30
㾎
331-263331-263-30
㾎
331-264331-264-30
Spline㾎
micrometer 331-351
331-351-30
㾎
331-352
331-352-30
㾎
331-353
331-353-30
㾎
331-354331-354-30
㾎inch/mm331-361331-361-30
㾎
331-362331-362-30
㾎
331-363331-363-30
㾎
331-364331-364-30
㾎
340-251
340-251-10
340-251-30
㾎
mm
340-252
340-252-10
340-252-30
㾎Outsidemicrometer 340-351
340-351-10
340-351-30
㾎inch/mm
340-352
340-352-10
340-352-30
㾎
342-251342-251-30
㾎
342-252mm
342-252-30
㾎
342-253342-253-30
㾎
342-254342-254-30
㾎Point
micrometerr 342-261
342-261-30
㾎
mm
342-262342-262-30
㾎
342-263342-263-30
㾎
342-264342-264-30
㾎
Crimp height 342-271
mm
micrometer 342-271-30
㾎
342-351342-351-30
㾎
342-352342-352-30
㾎
342-353342-353-30
㾎Pointinch/mmmicrometer 342-354
342-354-30
㾎
342-361342-361-30
㾎
342-362342-362-30
㾎

Unit Current productCode No.
342-363342-363-30
Point㾎
inch/mmmicrometer 342-364
342-364-30
㾎
Crimp height 342-371
inch/mmmicrometer 342-371-30
㾎
343-250
343-250-30
㾎
343-251343-251-30
㾎
mm
343-252343-252-30
㾎
343-253
Caliper type 343-253-30
㾎
outsidemicrometer 343-350
343-350-30
㾎
343-351
343-351-30
㾎inch/mm343-352343-352-30
㾎
343-353343-353-30
㾎
345-250
345-250-10
345-250-30㾎
mm
345-251345-251-10
Caliper type 345-251-30
㾎
insidemicrometer 345-350
345-350-10
345-350-30
㾎inch/mm345-351345-351-10
345-351-30
㾎350-251
350-251-10
350-251-30
㾎350-252
350-252-10
350-252-30
㾎350-253
350-253-10
350-253-30
㾎350-254350-254-10
350-254-30
㾎350-261
350-261-10
350-261-30
㾎350-271350-271-10
350-271-30
㾎350-272
Digimatic 350-272-10
mm
micrometer
350-272-20
head
350-272-30
㾎350-273350-273-10
350-273-30
㾎350-274
350-274-10
350-274-20
350-274-30
㾎
350-281350-281-10
350-281-30
㾎
350-282
350-282-10
350-282-30
㾎
350-283
350-283-10
350-283-30
㾎350-284
350-284-10

Product nameProduct name

Code No.
Unit Current product350-284-30 mm
㾎350-351
inch/mm
350-351-10
350-351-30
㾎
350-352
350-352-10
350-352-30
㾎350-353
350-353-10
350-353-30
㾎
350-354350-354-10
350-354-30
㾎350-357-10
Digimatic 350-357-30
㾎
micrometer
350-361head
350-361-10
350-361-30
㾎350-381350-381-10
350-381-30
㾎
350-382
350-382-10
350-382-30
㾎
350-383
350-383-10
350-383-30
㾎350-384
350-384-10
350-384-30
㾎
369-250369-250-30
㾎
369-251369-251-30
㾎
mm
369-252369-252-30
㾎
369-253369-253-30
㾎Disk
micrometer 369-350
369-350-30
㾎
369-351369-351-30
㾎inch/mm369-352369-352-30
㾎
369-353369-353-30
㾎
389-251389-251-30
㾎
389-252389-252-30
㾎
389-261389-261-30
㾎
mm
389-262389-262-30
㾎
389-271389-271-30
㾎
389-272㾎
Sheet metal 389-272-30
micrometer 389-351
389-351-30
㾎
389-352
inch/mm389-352-30
㾎
389-361389-361-30
㾎
389-362389-362-30
㾎
389-371mm
389-371-30
㾎
389-372389-372-30
㾎33

Connection Unit
No.02AZF310
No.02AZF700

U-WAVE-TM/TMB compatible gages

U-WAVE-TC/TCB compatible gages

No.02AZF310
Product name

Code No.
Unit Current product395-251395-251-30
㾎
395-252395-252-30
㾎
395-253395-253-30
㾎
395-254395-254-30
㾎
395-261395-261-30
㾎
395-262395-262-30㾎
mm
395-263395-263-30
㾎
395-264395-264-30
㾎
395-271395-271-30
㾎
395-272395-272-30
㾎
395-273395-273-30
㾎
395-274
Tube
micrometer 395-274-30
㾎
395-351395-351-30
㾎
395-352395-352-30
㾎
395-353395-353-30
㾎
395-354395-354-30
㾎
395-362395-362-30
㾎
395-363
inch/mm
395-363-30
㾎
395-364395-364-30
㾎
395-371395-371-30
㾎
395-372395-372-30
㾎
395-373395-373-30
㾎
395-374395-374-30
㾎
406-250406-250-30
㾎
406-251406-251-30
㾎
mm
406-252406-252-30
㾎
406-253
Outside㾎
micrometer, 406-253-30
non-rotating 406-350
spindle406-350-30
㾎
406-351406-351-30
㾎inch/mm406-352406-352-30
㾎
406-353406-353-30
㾎
422-230422-230-30
㾎
422-231422-231-30
㾎
422-232422-232-30
㾎
422-233
Blade㾎
micrometer, 422-233-30
mm
non-rotating 422-260
spindle422-260-30
㾎
422-261422-261-30
㾎
422-270422-270-30
㾎
422-271422-271-30
㾎34

Connection Unit
No.02AZF300
No.02AZF310

No.02AZF310
Code No.
Unit Current product422-330422-330-30
㾎
422-331422-331-30
㾎
422-332422-332-30
㾎422-333
Blade㾎
micrometer, 422-333-30
inch/mm
non-rotating 422-360
spindle422-360-30
㾎
422-361422-361-30
㾎
422-370422-370-30
㾎
422-371422-371-30
㾎Product name

No.02AZF700
Product name

ID-CNX

ID-FNX

Code No.
Unit Current product543-700B
㾎
543-705B
㾎
mm
543-710B
㾎
543-715B
㾎
543-701B
㾎
543-702B
㾎
543-706B
㾎
543-707B㾎
inch/mm
543-711B
㾎
543-712B
㾎
543-716B
㾎
543-717B
㾎
543-855B
㾎
mm
543-856B
㾎

U-WAVE-TC/TCB compatible gages
No.02AZF310
Product name

Code No.
Unit Current product500-712-10
500-712-20
㾎
500-713-10
500-713-20
㾎
500-714-10
500-714-20
㾎
500-716-11mm
500-716-20
㾎
500-717-11
500-717-20
㾎
500-718-11
500-718-20
㾎
500-719-10
500-719-20
㾎
500-720-11
inch/mm500-720-20
㾎ABS coolant- 500-721-10
proof caliper 500-721-20
㾎
500-722-10
500-722-20㾎
mm
500-723-10
500-723-20
㾎
500-724-10
500-724-20
㾎
500-727-11
500-727-20㾎
mm
500-728-11
500-728-20
㾎
500-735-10
500-735-20
500-736-10
inch/mm500-736-20
㾎
500-737-10
500-737-20
㾎Code No.
Unit Current product500-738-10
500-738-20
㾎
500-762-10
500-762-20
㾎
500-763-10
ABS coolant- 500-763-20
㾎
proof caliper 500-764-10 inch/mm
500-764-20
㾎
500-769-10
500-769-20
㾎
550-301-10mm
550-301-20
㾎
550-311-10ABS
inch/mm
㾎
Digimatic 550-311-20
caliper,
550-331-10mm
nib style jaws 550-331-20
㾎
550-341-10
inch/mm550-341-20
㾎
551-301-10
mm
551-301-20
㾎ABS
Digimatic 551-311-10 inch/mm
551-311-20
㾎
caliper,nib style and 551-331-10
mmstandard 551-331-20
㾎jaws
551-341-10
inch/mm551-341-20
㾎Tire tread 571-100-20 mm
depth gage 571-200-20
㾎
571-251-10
571-251-20
㾎
571-252-10
571-252-20
㾎
571-253-10mm
571-253-20
㾎
571-254-10
571-254-20
㾎
571-255-10
571-255-20
㾎
571-261-10
571-261-20
㾎
571-262-10
inch/mm571-262-20
㾎Depth gage571-263-10
571-263-20
㾎
571-264-10
571-264-20
㾎inch/mm571-265-10
571-265-20
㾎
571-301-10
571-301-20
㾎
mm
571-302-10
571-302-20
㾎
571-311-10
571-311-20
inch/mm571-312-10
571-312-20
㾎
573-601
573-601-20
㾎
573-602573-602-20
㾎
573-604
573-604-20
㾎
573-605
573-605-20
㾎
573-606
573-606-20
㾎
Dedicated 573-608mm
caliper573-608-20
㾎
573-611573-611-20
㾎
573-612573-612-20
㾎
573-614573-614-20
㾎
573-615573-615-20
㾎
573-616
Product name

No.02AZF310
Product name

Dedicatedcaliper

Code No.
Unit Current product573-616-20
㾎
573-618
㾎
573-618-20
㾎
573-621573-621-20
㾎
573-622573-622-20
㾎
573-625573-625-20
㾎
573-626573-626-20
㾎
mm
573-634573-634-20
㾎
573-635
573-635-20
㾎
573-645573-645-20
㾎
573-646573-646-20
㾎
573-647573-647-20
㾎
573-648573-648-20
㾎
573-651573-651-20
㾎
573-652573-652-20
㾎
573-653573-653-20
㾎
573-654573-654-20
㾎
mm
573-661573-661-20
573-662573-662-20
㾎
573-676573-676-20
㾎
573-677573-677-20
㾎
573-679573-679-20
㾎
573-701
573-701-20
㾎
573-702573-702-20
㾎
573-704
573-704-20
㾎
573-705
573-705-20
㾎
573-706
573-706-20
㾎
573-708
573-708-20
㾎
573-721573-721-20
㾎inch/mm
573-725573-725-20
㾎
573-734573-734-20
㾎
573-745573-745-20
㾎
573-746573-746-20
㾎
573-751573-751-20
㾎
573-752573-752-20
㾎
573-761573-761-20
㾎Current productNo.02AZF300
Product nameABS
Digimatic
caliperr

Code No.
Unit Current product500-150-20
500-150-30㾎
500-151-20
500-151-30
㾎
500-152-20
500-152-30
㾎
500-153500-153-30
㾎
500-154-20
500-154-30
㾎
500-155-20mm
500-155-30
㾎
500-156-20
500-156-30
㾎
500-157-20
500-157-30
㾎
500-158-20
500-158-30
㾎
500-161-21
500-161-30
㾎
500-162-21
500-162-30
㾎
500-167500-167-30
㾎
500-168
500-168-30
㾎
500-170-20
500-170-30㾎
500-171-20
500-171-30
㾎
500-172-20
500-172-30
㾎
500-173
inch/mm500-173-30
㾎
500-174-20
500-174-30
㾎
500-175-20
500-175-30
㾎
500-176-20
500-176-30
㾎
500-177-20
500-177-30
㾎
500-178-20
500-178-30
㾎
500-201-21mm
500-201-30
㾎
500-202-21
inch/mm500-202-30
㾎
500-203-21mm
500-203-30
㾎
500-204-21
inch/mm500-204-30
㾎
500-205mm
500-205-30
㾎
500-233-21
500-233-30
㾎
500-234-21
500-234-30
㾎
500-235-21
500-235-30 mm
㾎500-236-21
500-236-30
㾎
500-443500-444500-445500-463500-464inch/mm500-465

Product name

Code No.
Unit Current product571-201-20
571-201-30
㾎
571-202-20mm
571-202-30
㾎
571-203-20
571-203-30
㾎Depth gage571-211-20
571-211-30
㾎
571-212-20
inch/mm571-212-30
㾎
571-213-10
571-213-30
㾎
573-191-20
ABS low 573-191-30 mm
㾎force
573-291-20
caliperinch/mm
573-291-30
㾎

352312(JP1), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Notes on Export Regulations
Do not commit an act which could directly or indirectly violate any law or regulation of Japan, your country or any other international treaty,
relating to the export or re-export of any commodities
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.
All product information contained in this brochure is current as of December 2023.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp