Simplifying Progress
PFAS — Analysis Arium® CellPlus Ultrafilter
SampleDetection threshold

Detected Concentration

UnitMethodPFBA

50

Under detection thresholdng/L (ppt)

MS-0047387 1; APFPeA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFHxA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFHpA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; APFNA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; APFDA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFUnDA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFDoDA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFTrDA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFTeDA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFHxDA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFBS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFPeS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFHxS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; A

PFHpS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; APFNS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; APFDS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFUnDS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFDoDS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFTrDS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

4:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

6:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

8:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

10:2 FTS

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

N-MeFOSAA

0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

N-EtFOSAA

0.5

Under detection thresholdng/L (ppt)

MS-00473871; PV

8:2diPAP

0.5Under detection thresholdng/L (ppt)

MS-00473871; PVPFECHS

0.5

Under detection thresholdng/L (ppt)

MS-00473871; PV

Execution and Analysis Procedure
The water analysis was executed by TÜV Rheinland Energy & Enviroment GmbH, an internationally recognized testing laboratory for special analytics,
based on following measurement method: MS-0047387 Rev. 0, in accordance with DIN 38407-42, 2011-03. Relative expanded measurement
uncertainty (k=2): 50 %.
A= The method has been accredited.
PV= The method has been partially validated.
The tests were performed with the Arium® CellPlus Ultrafilter attached to an Arium® Mini Essential, fed with tap water. The CellPlus Ultrafilter was flushedwith 20 liters of ultrapure water before the sample was collected.

SampleDetection threshold

Detected Concentration

UnitMethod

PFOA (lin)0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFOA (br)0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFOA (sum)0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFOS (lin)0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; APFOS (br)0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

PFOS (sum)0.5

Under detection thresholdng/L (ppt)

MS-0047387 1; A

Execution and Analysis Procedure
The water analysis was executed by TÜV Rheinland Energy & Enviroment GmbH, an internationally recognized testing laboratory for special analytics,
based on following measurement method: MS-0047387 Rev. 0, in accordance with DIN 38407-42, 2011-03. Relative expanded measurement
uncertainty (k=2): 50 %.
A= The method has been accredited.
PV= The method has been partially validated.
The tests were performed with the Arium® CellPlus Ultrafilter attached to an Arium® Mini Essential, fed with tap water. The CellPlus Ultrafilter was flushedwith 20 liters of ultrapure water before the sample was collected.

Germany
Sartorius Lab Instruments
GmbH & Co. KG
Otto-Brenner-Strasse 2037079 Goettingen
Phone +49 551 308 0
For further information, visit	
sartorius.com

Specifications subject to change without notice.
© Sartorius Lab Instruments GmbH & Co. KG.
Status: 06 | 2024USASartorius Corporation
565 Johnson Avenue
Bohemia, NY 11716
Phone +1 631 254 4249
Toll-free +1 800 635 2906