PRODUCT OVERVIEWPR E S S UR E

„Successful medium-sized

AT TH E

because they are active in many

HIG H E ST L E V E L

concentrate on one area and do itcompanies are not successful
areas, but rather because they
better than anyone else“This is our philosophy. That´s why BDSENSORS hasconcentrated on electronic pressure measurement
technology from the beginning.
With our unremitting product and and quality strategy
we have been successful in becoming a major playeron the world market for electronic pressure sensing
devices within a few years.

With 260 employees at 4 locations in Germany, the Czech

INDEX

Republic, Russia and China BD|SENSORS has solutions from
0.1 mbar to 6000 bar:

PRESSURE

6-22

LEVEL

23-32

>> hydrostatic level probes

SWITCH

33-38

Two pressure transmitters and a submersible probe, based

EVALUATION

39-43

HART® communication or field bus interface.

SPECIAL VERSIONS

44-45

In addition we have developed hundreds of customer-spe-

APPLICATIONS

48-49

4 ARGUMENTS

50>> pressure sensors, pressure transducers		pressure transmitters>> electronic pressure switches>> pressure measuring devices with display andswitching outputs

on a stainless steel silicon sensor were the beginning.
Today the range extends to more than 70 standard products,from economical OEM devices to high-end products withcific applications, underlining the competence and flexibilityof BD|SENSORS. The excellent price/performance ratio of
our products is proof of the fact that we are able to meet the
toughest demand: Being a problem-solver for our customers.

For large production batches as well as for small production
numbers, no matter for what medium or external factors,
with almost any mechanical or electrical connection - wesolve your problem
flexibly, quickly and cost-efficiently.

4

SENSOR TECHNOLOGIESThe requirements on pressure and level measuring devices are various and, in addition to
intelligent design solutions, require most of all appropriate sensor technology.

12

stainless steel silicon sensor

stainless steel sensor

type DSP 210 without media isolation

type microfused(Ø = 18 mm)pressure ranges: 0 … 3.5 bar to 0 … 700 bar

pressure ranges:

0 … 20 mbar to 0 … 7 bar

application:gases, compressed air,liquids and non-aggressive media

type DSP 410 / DSP 411 / DSP 414 (Ø = 18 mm)with stainless steel diaphragmpressure ranges:

0 … 100 mbar to 0 … 600 bar

application:

gaseous and liquid media compatible
with stainless steel

application:

hydraulic and oxygen applications

type thinfilm-sensor
pressure ranges: 0 … 60 bar to 0 … 2200 bar
application:

hydraulic applications

type strain gaugepressure ranges: 0 … 1000 bar to 0 … 6000 bar

type DSP 413 / DSP 415 (Ø = 15 mm)with stainless steel diaphragmpressure ranges:

0 … 400 mbar to 0 … 600 bar

application:

for submersible probes Ø = 17 mm
as well as for pressure transmitterswith G 1/2“ flush and hydraulicapplications

application:

hydraulic applications,high dynamic pressure

SENSOR TECHNOLOGIES

5

BD|SENSORS is one of the few companies worldwide using four elements of modern pressure
measurement, offering sensors produced by BD|SENSORS itself or by partner companies with
BD|SENSORS know-how.

3

4

thickfilm ceramic sensor

capacitive ceramic sensor

type DSK 511 with flush diaphragmtype DSK 705 M(Ø = 18 mm)(Ø = 32.4 mm)pressure ranges:

0 ... 0.5 bar to 0 ... 600 bar

pressure ranges: 0 ... 60 mbar to 0 ... 20 bar

application:

for aggressive media and oxygen;with flush diaphragm preferred
for highly viscuous or contaminated media

type DSK 720 M(Ø = 18 mm)

type DSK 516 with flush diaphragmpressure ranges: 0 ... 100 mbar to 0 ... 50 bar(Ø = 15 mm)pressure ranges:

0 ... 0.5 bar to 0 ... 50 bar

application:

for submersible probes Ø 17 mm
and pressure transmitter with
G 1/2“ flush

type DSK 611 in monolithic design(Ø = 18 mm)pressure ranges:

0 ... 2 bar to 0 ... 400 bar

application:

OEM products with excellent price /performance ratio

application:

preferred for hydrostatic levelmeasurement as screw-in transmitteror submersible probes, a.o. for
aggressive media (acids, lyes, etc.)

6

APPLICATIONSPRESSURE>> hydraulics

7 - 17>> pneumatics

DIFFERENTIAL PRESSURE TRANSMITTER

18 - 20>> process monitoring and processengineering

DIGITAL PRESSURE GAUGE

21 - 22

PRESSURE TRANSMITTER>> control systems>> tool construction / presses /injection moulding machines>> power supply and distribution

PRESSURE TRANSMITTER7

For differential pressure measurement
Pressure ranges: 0 ... 10 mbar to 0 ... 6000 bar
Based on many years of experience and 4 different sensor technologies,
BD|SENSORS has developed a comprehensive range of standardized pressuretransducers. A virtually unlimited combination of mechanical and electricalconnections will allow implementation of a large range of device types. Different
types of output signals are available in addition. Whether analog or digital –current and voltage signals are as standard with us as the conventional digitalinterfaces found in industry.

DIGITAL
stainless steel sensor
nominal pressure

0 ... 100 mbar to 0 ... 400 bar

accuracy

0.25 / 0.35 % FSO

plant and machine engineering

DCT 531(according to IEC 60770)

output signal

RS485 with Modbus RTU protocol

characteristics

perfect thermal behaviour,
excellent long term stability

optionpressure port G 1/2“ flush,pressure sensor welded

RS485 Modbus RTU

application

stainless steel sensor
nominal pressure

0 ... 100 mbar to 0 ... 400 bar

accuracy

0.25 / 0.35 % FSO

plant and machine engineering(according to IEC 60770)

output signal

i2C, bus frequency max. 400 kHz, configurationof data format, interrupt signal

characteristics

perfect thermal behaviour,
excellent long term stability

optionpressure port G 1/2“ flush,pressure sensor welded

application

i²C interface

DCT 532

8

PRESSURE TRANSMITTER

DIGITAL
stainless steel sensor
nominal pressure

0 ... 100 mbar to 0 ... 400 bar

accuracy

0.25 / 0.35 % FSO

plant and machine engineering

DCT 533(according to IEC 60770)

output signal

IO-Link according to specification V 1.1,
data transfer 38.4 kBaud,smart sensor profile

characteristics

perfect thermal behaviour,
excellent long term stability

optionpressure port G 1/2“ flush,pressure sensor welded

IO-Link Interface

application

ceramic sensor

plant and machine engineering

nominal pressure

0 ... 600 mbar to 0 ... 600 bar

accuracy

0.5 % FSO

DCT 561(according to IEC 60770)

output signal

RS485 with Modbus RTU protocol

characteristics

good thermal behaviour,
good long term stability

optionpressure port G 1/2“ open port PVDF
for aggressive media,
oxygen application

RS485 Modbus RTU

application

ceramic sensor

plant and machine engineering

nominal pressure

0 ... 600 mbar to 0 ... 600 bar

accuracy

0.5 % FSO

(according to IEC 60770)

output signal

IO-Link according to specification V 1.1,
data transfer 38.4 kbit/s,
smart sensor profile

characteristics

good thermal behaviour,
good long term stability

optionpressure port G 1/2“ flush,pressure port G 1/2“ open port PVDF
for aggressive media,
oxygen application

application

IO-Link-Interface

DCT 563

PRESSURE TRANSMITTER9PRECISION
stainless steel sensor / ceramic sensor

process, oil and gas industry

nominal pressure

0 ... 400 mbar to 0 ... 600 bar (XMP i)(turn-down 1:10 adjustable)
0 ... 160 mbar to 0 ... 20 bar (XMP ci)(turn-down 1:5 adjustableaccuracy(according to IEC 60770)

0.1 % FSO (XMP i)
0.1 / 0.2 % FSO (XMP ci)process connection

inch and NPT threads, DRD, flange

housing

two chamber aluminium die cast case,stainless steel field housing

optiondisplay and operating module,
flameproof enclosure,cooling element up to 300 °C (XMP i),
diaphragm 99.9 % Al2O3 (XMP ci)

XMP i

XMP ci

HART ®

application

stainless steel sensor / ceramic sensor
nominal pressure

0 ... 400 mbar to 0 ... 40 bar (x|act i)(turn-down 1:10 adjustable)
0 ... 160 mbar to 0 ... 20 bar (x|act ci)(turn-down 1:5 adjustable)

accuracy

0.1 % FSO (x|act i)
0.1 / 0.2 % FSO (x|act ci)(according to IEC 60770)

process connection

G1“ cone, G1 1/2“ flush, clamp, dairy pipe,
varivent®, DRD, flange

characteristics

hygienic version,
display and operating module,
cooling element up to 300 °C (x|act i),diaphragm 99.9 % Al2O3 (x|act ci)

optionapplication

IS-version

hygienic applications

Hygienic Designx|act ix|act ci

HART ®

10

PRESSURE TRANSMITTERPRECISION
stainless steel sensor
nominal pressure

0 ... 400 mbar to 0 ... 40 bar

accuracy

0.1 % FSO

laboratory techniques, food and beverage

DMP 331 Pi(according to IEC 60770)

characteristics

excellent temperature response 0.04 % FSO / 10 K,
process connections suitable for hygienic application,
vacuum resistantoptionIS-version,communication interface for adjustment of offset,
span and damping

flush

application

stainless steel sensor

DMP 331 i

DMP 333 i

plant and machine engineering

DMP 320

laboratory, environmental industry

nominal pressure

0 ... 400 mbar to 0 ... 40 bar (DMP 331i)
0 ... 60 bar to 0 ... 600 bar (DMP 333 i)

accuracy

0.1 % FSO

(according to IEC 60770)

characteristics

digital electronics for linearisation andactive temperature compensation(temperature error 0.02% / 10 K),
with communication interface for offset
and span adjustment

optionIS-version,
digital output RS 232 with adapter

application

stainless steel sensor
nominal pressure

0 ... 100 mbar to 0 ... 600 bar

accuracy

0.1 % FSO

(according to IEC 60770)

characteristics

extremely fast response time ≤ 0.5 ms,
internal sample rate 10 kHz,
excellent thermal behaviour,
outstanding long term stability

application

fast response time

PRESSURE TRANSMITTER11PRECISION
stainless steel sensor
nominal pressure

0 ... 100 mbar to 0 ... 600 bar

accuracy

0.1 / 0.25 % FSO

plant and machine engineering

DMP 321HVAC

DMP 343(according to IEC 60770)

characteristics

compact design,
perfect thermal behaviour,
excellent long-term stability

optionIS-version,pressure sensor welded

application

INDUSTRY
stainless steel sensor without media isolation
nominal pressure

0 … 10 mbar to 0 … 1000 mbar

accuracy

0.35 % FSO(according to IEC 60770)

optionIS-version,compact field housing

application

extreme low pressure

stainless steel sensor
nominal pressure

0 ... 60 bar to 0 ... 600 bar

accuracy

0.35 % FSO

hydraulics(nach IEC 60770)

characteristics

G1/4“ flush

optionIS-version

applicationflush

DMP 339

12

PRESSURE TRANSMITTER

INDUSTRY
stainless steel sensor, welded
nominal pressure

0 … 6 bar to 0 … 600 bar

accuracy

0.5 % FSO

medical technology, hydraulics

DMP 335(according to IEC 60770)

characteristics

suitable for oxygen application,
resistant against pressure peaks

optionIS-version

applicationhigh overpressure
capability

thinfilm sensor, welded
nominal pressure

0 … 600 bar to 0 … 2200 bar

accurcay

0.35 % FSO

plant and mechanical engineering

DMP 334(according to IEC 60770)

optionIS-version,compact field housing,adjustability of span and offset

applicationhighpressure
Sauerstoffapplikation

strain gauge

oil and gas industry

nominal pressure

0 … 2000 bar to 0 … 6000 bar

accuracy

0.25 / 0.5 % FSO(according to IEC 60770)

characteristics

adjustability of offset and span
via external potentiometers,pressure port 9/16 UNF

optionIS-version

application

ultra high pressure

DMP 304

PRESSURE TRANSMITTER

13

INDUSTRY
ceramic sensor

universal applications

nominal pressure

0 … 400 mbar to 0 … 600 bar

accuracy

0.5 % FSODMK 331(according to IEC 60770)

optionIS-version,compact field housing,pressure port PVDF,
oxygen application,pressure port G 1/2“ flush

application

aggressive media

ceramic sensor

laboratory techniques, biogas plants

nominal pressure

0 … 40 mbar to 0 … 20 bar

accuracy

0.25 / 0.35 % FSO

DMK 351(according to IEC 60770)

optionIS-version,compact field housing,diaphragm 99.9 % Al2O3

applicationhigh overpressure
capability

ceramic sensor

plant and mechanical engineering / laboratory

nominal pressure

0 … 100 mbar to 0 … 60 bar

accuracy

0.25 / 0.35 % FSO(according to IEC 60770)

optionapplication

IS-version,
field housing,diaphragm of 99.9 % Al2O3

DMK 387

14

PRESSURE TRANSMITTER

INDUSTRY
ceramic sensor / stainless steel sensor
nominal pressure

0 ... 400 mbar to 0 ... 600 bar (DMK 457)
0 ... 100 mbar to 0 ... 600 bar (DMP 457)

accuracy

0.5 % FSO (DMK 457)
0.25 / 0.35 % FSO (DMP 457)

(according to IEC 60770)

optionDMK 457

DMP 457

marine / shipbuilding / offshore

DMK 456

marine / shipbuilding / offshore

IS-version,compact field housing,submersible version

application

ceramic sensor
nominal pressure

0 … 40 mbar to 0 … 20 bar

accuracy

0.1 / 0.25 % FSO(according to IEC 60770)

special feature

IS-version (temperature class T6),
stainless steel field housing

optiondiaphragm 99.9 % Al2O3,
thread or flange version

application

robust housing

ceramic sensor

marine / shipbuilding / offshore

nominal pressure

0 … 40 mbar to 0 … 20 bar

accuracy

0.1 / 0.25 % FSO(according to IEC 60770)

special feature

stainless steel field housing

optionIS-version (temperature class T4),
diaphragm 99.9 % Al2O3,
pressure port of CuNiFe

application

DMK 458

PRESSURE TRANSMITTER

15

INDUSTRY
stainless steel

hygienic applications

nominal pressure

0 … 100 mbar to 0 … 40 bar

accuracy

0.25 / 0.35 % FSO

DMP 331 P(according to IEC 60770)

process connection

G½“, G¾“, G1“ flush, dairy pipe,
clamp, varivent®

optionIS-version,compact field housing,
FDA conforming filling fluid,cooling element up to 300 °C

flush

application

ceramic sensor

hygienic applications

nominal pressure

0 … 60 bar to 0 … 400 bar

accuracy

0.5 % FSODMK 331 P(according to IEC 60770)

process connection

G½“ / G¾“ / G1“ flush

optioncompact field housing,
FDA conforming filling fluid,cooling element up to 300 °C

application

flush

ceramic sensor

hygienic applications

nominal pressure

0 … 40 mbar to 0 … 20 bar

accuracy

0.25 / 0.35 % FSO(according to IEC 60770)

process connection

G1 ½“ flush, clamp, dairy pipe,
varivent®, flange, flange DRD

optionIS-version,compact field housing,diaphragm 99.9 % Al2O3

applicationhighoverpressure capability

DMK 351 P

16

PRESSURE TRANSMITTER

OEM
stainless steel sensor, welded
nominal pressure

0 ... 6 bar to 0 ... 600 bar (17.600 G)
0 ... 6 bar to 0 ... 60 bar (17.609 G)

accuracy

0.5 % FSO

mobile hydraulics

17.600 G

17.609 G(according to IEC 60770)

output signal

4 … 20 mA / 2L
0 … 10 V / 3L
10 ... 90 % of VS / 3L rat.

pressure port

G ¼“, ¼“ NPT, G ½“, 7/16 UNF

heavy duty / refrigeration

application

stainless steel sensor without media isolation
nominal pressure

0 … 100 mbar to 0 … 6 bar

accuracy

0.5 % FSO

plant and mechanical engineering

18.600 G(according to IEC 60770)

output signal

4 … 20 mA / 2-wire
0 … 10 V / 3-wire
10 ... 90 % of VS / 3-wire rat.

pressure port

G ¼“, ¼“ NPT, G ½“

application

pneumatics

stainless steel sensor
nominal pressure

0 … 100 mbar to 0 … 6 bar

accuracy

0.5 % FSO

general industrial applications(according to IEC 60770)

output signal

4 … 20 mA / 2-wire
0 … 10 V / 3-wire
10 ... 90 % of VS / 3-wire rat.

pressure port

G ¼“, ¼“ NPT, G ½“

application

low pressure

18.601 GPRESSURE TRANSMITTER

17

OEM
stainless steel sensor
nominal pressure

0 … 1 mH2O to 0 … 10 mH2O

accuracy

0.5 % FSO

general industrial applications

18.605 G(according to IEC 60770)

output signal

4 … 20 mA / 2-wire
0 … 10 V / 2-wire
10 ... 90 % of VS / 3-wire rat.

pressure port

G ¼“ with PVC cableapplication

submersible

ceramic sensor

general industrial applications

nominal pressure

0 … 1 bar to 0 … 400 bar

accuracy

0.5 % FSO

26.600 G(according to IEC 60770)

output signal

4 … 20 mA / 2-wire
0 … 10 V / 3-wire
10 ... 90 % of VS / 3-wire rat.

pressure port

G ¼“, ¼“ NPT, G ½“

optionoil and grease free version

standard

application

ceramic sensor

general industrial applications

nominal pressure

0 … 1.6 bar to 0 … 250 bar

accuracy

1 % FSO(according to IEC 60770)

output signal

4 … 20 mA / 2-wire
0 … 10 V / 3-wire
10 ... 90 % of VS / 3-wire rat.

pressure port

G ¼“, ¼“ NPT

application

low cost

30.600 G

18

DIFFERENTIAL PRESSURE TRANSMITTERFor differential pressure measurement
Pressure ranges: 0 ... 1 mbar to 0 ... 70 bar
Thanks to different sensor technologies combined with compact aluminium
die-cast cases or plastic housings, our differential pressure transmitters may be
used for numerous fluids and gases, e. g. for monitoring ventilation ducts, filtersand fans in HVAC areas as well as for level measurement in closed pressurizedtanks.

DIGITAL
stainless steel sensor
differential pressure

0 … 10 mbar to 0 … 20 bar

accuracy

0.1 % FSO

plant and mechanical engineering

DPT 100(according to IEC 60770)

characteristics

fast response time,aluminium die cast case,zero adjustment via switch

process connection

internal thread 1/4“ - 18 NPT

optionseveral process connections

RS485 Modbus RTU

application

PRECISION
stainless steel sensor
differential pressure

0 … 1 mbar to 0 … 20 bar

accuracy

0,075 % FSO

oil and gas industry(according to IEC 60770)

characteristics

static over pressure 400 bar,rangeability max. 100:1,
aluminium die cast case,
HART®-communication

optionIS-version,
LC display,stainless steel housing

application

HART ®

DPT 200

DIFFERENTIAL PRESSURE TRANSMITTER19PRECISION
stainless steel sensor
nominal pressure

0 … 75 mbar to 0 … 20 bar

accuracy

0.1 % FSO

XMD

process, oil and gas industry(according to IEC 60770)

characteristics

IS-version,
turn-down 1:10,
aluminium die-cast caseprocess connection

internal thread ¼“ – 18 NPT

optiondisplay and operating module, flameproof
enclosure, chemical seals assembly

fluids + gases

HART ®

application

INDUSTRY
stainless steel sensor
nominal pressure

0 … 0.2 bar to 0 … 16 bardifferential pressure

0 … 20 mbar to 0 … 16 bar

accuracy

0.5 % FSO

plant and mechanical engineering

DMD 331(according to IEC 60770)

characteristics

compact design,
mechanically robust and reliable at dynamic
pressure as well as with shock and vibration

optionIS-version

fluids + gases

application

stainless steel sensor
differential pressure

0 … 1 bar to 0 … 70 bar

accuracy

1 % BFSL

plant and mechanical engineering(according to IEC 60770)

characteristics

display and pressure port rotatable,
up to 2 contacts,
turn-down 1:10

application

fluids + gases

DMD 831

20

DIFFERENTIAL PRESSURE TRANSMITTER

INDUSTRY
silicon sensor

plant and mechanical engineering

nominal pressure

0 … 6 mbar to 0 … 1000 mbar

accuracy

0.35 / 1 / 2 % FSO

DMD 341(according to IEC 60770)

optiondisplay and switching module withup to 2 contacts

application

gases and compressed air

silicon sensor

HVAC

differential pressure

0 ... 6 mbar to 0 ... 1000 mbar (DPS 200)
0 ... 1.6 mbar to 0 ... 1000 mbar (DPS 300)

accuracy

1 % FSO BFSLDPS 200(according to IEC 60770)

characteristics

adjustable ranges (DPS 300)

optioncontacts automatic zero adjustment,
square root extraction(DPS 300)

application

gases and compressed air

DPS 300

DIGITAL PRESSURE GAUGE

21

With a great variety of mechanical and electrical connections, BD|SENSORS
offers a new generation of digital pressure gauges for different applications.
Due to the two sensor technologies in use (stainless steel sensor or ceramic
sensor), our digital pressure gauges are suitable for nearly all fluids, pasty
media and gases.
The display module is rotatable, so that a clear readability is guaranteed
even in unusual installation positions.

PRECISION
stainless steel sensor

test and calibration equipment / leak detection

nominal pressure

0 … 100 mbar to 0 … 400 bar

accuracy

0.05 % FSO BFSL

DM 01

(according to IEC 60770)

characteristics

stainless steel housing Ø = 100 mm,
data logger function,
modular sensor concept

LC- display

graphic LC-display 128 x 64 pixel
with background illuminationpressure portinch, NPT threads

optionIS-version,accredited calibration certificate,service case with accessories,
USB-interface

application

modular concept

DL 01

22

DIGITAL PRESSURE GAUGEINDUSTRY
BAROLI 02

stainless steel sensor / ceramic sensor
nominal pressure

0 … 100 mbar to 0 … 600 bar (BAROLI 02)
0 … 400 mbar to 0 … 600 bar (BAROLI 05)

accuracy

0.125 % FSO BFSL (BAROLI 02)
0.25 % FSO BFSL (BAROLI 05)(according to IEC 60770)

LC- display

4.5-digit 7-segment display,6-digit 14-segment additional display

process connection

inch, NPT threads

BAROLI 05

application

housing rotatable

BAROLI 02 P

stainless steel sensor / ceramic sensor
nominal pressure

0 … 100 mbar to 0 … 40 bar (BAROLI 02 P)
0 ... 60 bar to 0 ... 400 bar (BAROLI 05 P)

accuracy

0.125 / 0.25 % FSO BFSL (BAROLI 02 P)
0.25 % FSO BFSL (BAROLI 05 P)(according to IEC 60770)

LC-display

4.5-digit 7-segment display,6-digit 14-segment additional display

process connection

G ½“ flush, G 1“ flush, dairy pipe, clamp

BAROLI
0505
P
BAROLI

application

flush

OEM
ceramic sensor / stainless steel sensor

plant and mechanical engineering

nominal pressure

0 ... 1.6 bar to 0 ... 250 bar (DM 10)
0 ... 6 bar to 0 ... 600 bar (DM 17)

accuracy

0.5 % FSO BFSL

(according to IEC 60770)

characteristic

adjustable housing

LC-display

4,5-digit 7-segment display

pressure port

G 1/2“ EN 837, G 1/4“ EN 837, 1/4“ NPT

functionmin-/max-function with reset-function

application

low-cost

DM 10

DM 17

23

LEVEL

APPLICATIONS>> ground water monitoring

HYDROSTATIC LEVEL PROBESSCREW-IN TRANSMITTERS

24 - 3132>> depth and level measurement
in wells>> drinking water systems>> level monitoring in open andclosed tanks>> storm water systemes>> pump and booster stations>> water treatment plants>> tank farms / fuel storage>> recycling of process water

24

HYDROSTATIC LEVEL PROBES

The hydrostatic level probes made by BD|SENSORS are suitable for measuring
the level of liquid and pasty media of all kind.
The separable submersible probes LMP 308 / LMP 808 / LMK 358 / LMK 858 area speciality; the cable part can be separated from the sensor head effortlessly
and without tools. This is an enormous advantage for many of ourcustomers during assembly as well as when performing service andmaintenance.
Special versions, such as integrated overvoltage protection, temperature
sensor or data logger are just as much a part of our standard program as thecommunication version with RS-485 interface, HART®-protocol or Modbus.

DIGITAL
stainless steel sensor
level

0 … 1 mH2O to 0 … 250 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.25 / 0.35 % FSO

Fuel / Oil

DCL 531(accuracy to IEC 60770)

special feature

RS485 with Modbus RTU

optioncable protection via corrugated pipe,
different kinds of cables,
different kinds of seal materials,
drinking water certificate acc. to DVGW and KTW

ø 26,5 mmstainless steel probewith RS485 Modbus RTU

recommended for

ceramic sensor

Fuel / Oillevel

0 … 60 cmH2O to 0 … 200 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.25 / 0.35 % FSO

DCL 551(accuracy to IEC 60770)

special feature

RS485 with Modbus RTU

optiondiaphragm ceramics Al2O3 99,9%

recommended forø 39,5 mmstainless steel probewith RS485 Modbus RTU

HYDROSTATIC LEVEL PROBE

25

DIGITAL
ceramic sensor

DCL 571

Fuel / Oillevel

0 … 1 mH2O to 0 … 100 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.25 / 0.35 % FSO(accuracy to IEC 60770)

special feature

RS485 with Modbus RTU

optiondrinking water certificate acc. to DVGW and KTW

recommended forstainless steel probewith RS485 Modbus RTU

PRECISION
ceramic sensor

environmental industrylevel

0 … 60 cmH2O to 0 … 100 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.1 / 0.2 % FSO

LMK 358 H(accuracy to IEC 60770)

special feature

HART® communication

optionIS-version,
cable protection via corrugated pipe,
diaphragm 99.9 % Al2O3

recommended for

ceramic sensor

ø 39,5 mm
detachable stainlesssteel probe

chemical industry, environmental industrylevel

0 … 60 cmH2O to 0 … 200 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.1 / 0.2 % FSO

HART ®

LMK 382 H(according to IEC 60770)

special feature

HART® communicaton

optionIS-version,flange version,
diaphragm 99.9 % Al2O3

recommended forø 39.5 mmstainless steel
probe

HART ®

26

HYDROSTATIC LEVEL PROBESPRECISION
ceramic sensor

marine / shipbuilding / offshorelevel

0 … 60 cmH2O to 0 … 200 mH2O

housing matieralstainless steel 1.4404 (316 L), CuNiFe

accuracy

0.1 / 0.2 % FSO

LMK 458 H(according to IEC 60770)

special feature

HART® communication

optionIS-version,
diaphragm 99.9 % Al2O3,
screw-in or flange version

ø 39.5 mm

HART ®

recommended forstainless steel sensor
level

0 … 4 mH2O to 0 … 200 mH2O

housing matieralstainless steel 1.4404 (316 L)

accuracy

0.1 % FSO

Fuel / Oil

LMP 307 i(according to IEC 60770)

characteristic

excellent long term stability

optionIS-protection,
cable protection via corrugated pipe,
drinking water certificate acc. to DVGW and KTW

recommended forstainless steel sensor
level

0 … 4 mH2O to 0 … 200 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.1 % FSO

ø 26.5 mmstainlesssteel probe

energy industry, environmental industry

LMP 308 i(according to IEC 60770)

special feature

cable part and sensor head detachable

optionIS-version,
cable protection via corrugated pipe

recommended forø 35 mmdetachable stainlesssteel probeHYDROSTATIC LEVEL PROBE

27

STANDARD
stainless steel sensor

energy industry, environmental industry

level

0 … 1 mH2O to 0 … 250 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.25 / 0.35 % FSO

LMP 305(according to IEC 60770)

special feature

suitable for level measurement in 1“ pipes

recommended forø 19 mmslimline probe

stainless steel sensor

energy industry, enviromental industry

level

0 ... 1 mH2O to 0 ... 250 mH2O

temperature

0 ... 30 °C to 0 ... 70 °C (LMP 307 T)

housing material

stainless steel 1.4404 (316 L)

accuracy(according to IEC 60770)

0.1 / 0.25 / 0.35 % FSO (LMP 307)0.25 / 0.35 / 0.5 % FSO (LMP 307 T)
1° C (LMP 307 T)

special feature(LMP 307T)

two galvanic seperated signal circuit for pressure
and temperature

option (LMP 307)IS-version,
cable protection via corrugated pipe,
drinking water certificate acc. to DVGW and KTW

LMP 307

LMP 307 T

ø 27 mmstainlesssteel probe

recommended forstainless steel sensor

energy industry, enviromental industry

level

0 ... 1 mH2O to 0 ... 250 mH2O (LMP 308)
0 ... 1 mH2O to 0 ... 100 mH2O (LMP 808)

housing material

stainless steel 1.4404 (316 L)

accuracy

0.1 / 0.25 / 0.35 % FSO (LMP 308)
0.25 / 0.35 % FSO (LMP 808)(according to IEC 60770)

optionIS-version (LMP 308),
cable protection via corrugated pipe (LMP 308),
cable protection via pipe (LMP 808)

special feature

cable part and sensor head detachable

recommended for

LMP 308

LMP 808

ø 35 mmdetachable probe

28

HYDROSTATIC LEVEL PROBESSTANDARD
ceramic sensor

energy industry, environmental industrylevel

0 … 6 mH2O to 0 … 200 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.5 % FSOLMK 306(according to IEC 60770)

special feature

for level measurement in ¾“ pipes

recommended forø 17 mmstainless steel probe

ceramic sensor		

energy industry, environmental industrylevel

0 … 4 mH2O to 0 … 250 mH2O

temperature

0 ... 30 °C to 0 ... 70 °C (LMK 307 T)

housing material

stainless steel 1.4404 (316 L)

accuracy(according to IEC 60770)

0.5 % FSO
1°C (LMK 307 T)

special feature(LMK 307T)

two galvanic seperated signal circuit for pressure
and temperature

optionIS-version

LMK 307

LMK 307 T

ø 27 mmstainless steel
probe

recommended for

ceramic sensor

energy industry, enviromental industry

level

0 ... 40 cm H2O to 0 ... 100 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.25 / 0.35 % FSO

LMK 358(according to IEC 60770)

special feature

cable protection via corrugated pipe,
cable part and sensor head detachable

optionIS-version,
diaphragm 99.9 % Al2O3

recommended forø 39.5 mm
detachable stainlesssteel probeHYDROSTATIC LEVEL PROBE

29

STANDARD
ceramic sensor

energy industry, environmental industrylevel

0 … 40 cmH2O to 0 … 200 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.25 / 0.35 % FSO

LMK 382(according toi IEC 60770)

optionIS-version,flange version,
diaphragm 99.9 % Al2O3,
mounting with stainless steel pipe

ø 39.5 mmstainless steel probe

recommended for

ceramic sensor

energy industry, environmental industry

level

0 … 1 mH2O to 0 … 100 mH2O

housing material

stainless steel 1.4404 (316 L)

accuracy

0.25 / 0.35 % FSO

LMK 387(according to IEC 60770)

optionIS-version,
mounting with stainless steel pipe,
drinking water certificate acc. to DVGW and KTW

ø 22 mmrecommended forstainless steel probe

ceramic sensor

marine / shipbuilding / offshorelevel

0 … 40 cmH2O to 0 … 200 mH2O

housing material

stainless steel 1.4404 (316 L), CuNiFe

accuracy

0.1 / 0.25 % FSO

LMK 458(according to IEC 60770)

special feature

optionrecommended for

permissible temperature up to 125 °C,
chemical resistance against seawaterand HFOIS-version,
diaphragm 99.9 % Al2O3,
screw-in and flange version

ø 39.539,5 mm

30

HYDROSTATIC LEVEL PROBESSTANDARD
ceramic sensor

Shipbuilding / Offshorelevel

0 ... 1 mH2O to 0 ... 100 mH2O

housing materialtitanium, stainless steel 1.4404 (316 L)

accuracy

0.25 % FSO

LMK 487(according to IEC 60770)

special feature

diaphragm ceramics 99.9% Al2O3

optionIS-version,different kinds of elastomer

ø 22mmrecommended fortitanium, stainless steel

ceramic sensor

chemical and eviromental industry

level

0 ... 6 mH2O to 0 ... 200 mH2O

housing material

PP-HT

accuracy

0.5 % FSOLMK 806(according to IEC 60770)

special feature

suitable for level measurement in 3/4“ pipes

recommended forø 21 mmplastic probe

ceramic sensor

chemical industry, enviromental industrylevel

0 … 4 mH2O to 0 … 100 mH2O

housing material

PP-HT

accuracy

0.5 % FSOLMK 807(according to IEC 60770)

optionFKM, EPDM, FFKM seals

recommended forø 35 mmplastic probeHYDROSTATIC LEVEL PROBE

31

STANDARD
ceramic sensor

water / sewagelevel

0 … 1 mH2O to 0 … 100 mH2O

housing material

PP-HT

accuracy

0.25 / 0.35 % FSO

LMK 808(according to IEC 60770)

special feature

diaphragm ceramics 99.9 % Al2O3,
cable and sensor section detachable

optionmounting accessories as screw fitting
and terminal clamp of stainless steelø 35 mmdetachable plastic probe

recommended for

ceramic sensor

chemical industry, environmental industrylevel

0 … 0.4 mH2O to 0 … 100 mH2O

housing material

PP or PVDF

accuracy

0.25 / 0.35 % FSO

LMK 809(according to IEC 60770)

special feature

diaphragm 99.9 % Al2O3,
chemical resistance

optionprepared for mounting with pipe extension

recommended forø 45 mmplastic probe

ceramic sensor

chemical industry, environmental industrylevel

0 … 0.4 mH2O to 0 … 100 mH2O

housing material

PP-HT

accuracy

0.25 / 0.35 % FSO

LMK 858(according to IEC 60770)

special feature

cable part and sensor head detachable

optioncable protection via pipe,
diaphragm 99.9 % Al2O3

recommended forø 45 mmdetachable
plastic probe

32

SCREW-IN TRANSMITTERSTANDARD
stainless steel sensor
nominal pressure

0 … 100 mbar to 0 … 40 bar

level

0 … 1 mH2O to 0 … 400 mH2O

accuracy

0.1 / 0.25 / 0.35 % FSO

plant and mechanical engineering

LMP 331

plant and mechanical engineering

LMK 331(according to IEC 60770)

pressure port

G 3/4“ flush

optionIS-version,compact field housing

recommended for

ceramic sensor
nominal pressure

0 … 400 mbar to 0 … 60 bar

level

0 … 4 mH2O to 0 … 600 mH2O

accuracy

0.5 % FSO

(according to IEC 60770)

pressure port

G ¾“ flush
for pasty and contaminated media

optionIS-version,pressure port PVDF,
compact field housing

aggressive media

reommended for

ceramic sensor

environmental industry, renewable energy

nominal pressure

0 … 40 mbar to 0 … 20 bar

level

0 … 0.4 mH2O to 0 … 200 mH2O

accuracy

0.25 / 0.35 % FSO(according to IEC 60770)

pressure port

G1 ½“ flush

optionIS-version,pressure port PVDF or stainless steel,diaphragm 99.9 % Al2O3,
compact field housing

recommended for

aggressive media

LMK 351

33

APPLICATIONSSWITCHPRESSURE SWITCH WITH DISPLAY>> mobile hydraulics

34 - 37>> dry running protection>> flow monitoring

PRESSURE SWITCH WITHOUT DISPLAY

38>> grease monitoring>> gas compressors>> test and construction engineering

34

PRESSURE SWITCH
with display

Due to the simple handling as well as the variety of software features

Depending on the requirements, the universal(switching points and hysteresis freely configurable, delay function, min/pressure measuring devices with display and swit-

max-value data storage, display and analogue output signal scalable, etc.)

ching contacts can be used as :the DS 200 / DS 400 series is especially suitable as an intelligent pressure switch for general plant and machine construction and the processing
industry.

>> pressure transmitter

output signal:>> electronic pressure switch

2-wire (4 … 20 mA) or 3-wire (0 … 10 V),>> digital pressure gauge

up to 4 contacts
electrical connection:
various plugs(e. g. DIN or circular plug) or cable outlet

stainless steel sensor / ceramic sensor
nominal pressure

0 ... 100 mbar to 0 ... 600 bar (DS 400)
0 ... 400 mbar to 0 ... 600 bar (DS 401)

accuracy

0.25 / 0.35 % FSO (DS 400)
0.5 % FSO (DS 401)

(according to IEC 60770)

characteristics

up to 2 contacts,
4-digit LED-display in ball housing,rotatable and configurable display modulepressure portinch and NPT threads

optionIS-version, pressure port PVDF (DS 401)

DS 400DS 401

plant and mechanical engineering

DS 300universal application

robust version

application

stainless steel sensor
nominal pressure

0 ... 100 mbar to 0 ... 600 bar

accuracy

0.5 % FSO

(according to IEC 60770)

characteristics

indication of measured values on a 4-digit LED display,rotatable and configurable display module,
parameter settings via IO-Link or menu
(VDMA-conform)

special feature

analogue output 4 ... 20 mA / 0 ... 10 V switchable,
contacts PNP / NPN switchable

pressure portinch and NPT-theads

application

IO-link
interfacePRESSURE SWITCH

35

with display

stainless steel sensor / ceramic sensor
nominal pressure

0 ... 100 mbar to 0 ... 600 bar (DS 200)
0 ... 400 mbar to 0 ... 600 bar (DS 201)

accuracy

0.25 / 0.35 % FSO (DS 200)
0.5 % FSO (DS 201)

(according to IEC 60770)

characteristics

up to 4 contacts,
4-digit LED-display,rotatable and configurable display modulepressure portinch and NPT threads

optionIS-version

DS 200

DS 201

medical technology, oxygen application

DS 202

universal applications

application

stainless steel sensor, welded
nominal pressure

0 … 6 bar to 0 … 600 bar

accuracy

0.5 % FSO

(according to IEC 60770)

characteristics

up to 4 contacts,
4-digit LED-display,rotatable and configurable display modulepressure portinch and NPT threads

optionIS-version,oxygen version

high overpressure capability

application

stainless steel sensor without media isolation
nominal pressure

0 ... 10 mbar to 0 ... 1000 mbar

accuracy

0.35 % FSO

HVAC(according to IEC 60770)

characteristics

up to 4 contacts,
4-digit LED-display,rotatable and configurable display modulepressure portinch and NPT threads

optionIS-version

application

low pressure

DS 21036

PRESSURE SWITCH
with display

thinfilm sensor

plant and machine engineering / mobile hydraulics

nominal pressure

0 ... 600 bar to 0 ... 2200 bar

accuracy

0.35 % FSO

DS 214(according to IEC 60770)

characteristics

up to 4 contacts,
indication of measured values on a 4-digit LED display,rotatable and configurable display modulepressure port

G 1/2“ EN 837, M20x1,5

very highpressureapplication

stainless steel sensor, welded
nominal pressure

0 … 6 bar to 0 … 600 bar

accuracy

0.5 % FSO

hydraulics, oxygen application

DS 217(according to IEC 60770)

characteristics

up to 2 contacts,
4-digit LED-display,rotatable and configurable display modulepressure port

G ½“, G ¼“, ¼“ NPT

applicationOEM

ceramic sensor

pneumatics, hydraulics

nominal pressure

0 … 2 bar to 0 … 400 bar

accuracy

1.5 % FSO

(according to IEC 60770)

characteristics

up to 2 contacts,
4-digit LED-display,rotatable and configurable display modulepressure port

G ¼“, ¼“ NPT

applicationOEM

DS 230PRESSURE SWITCH

37

with display

HVAC

DS 233

DS 200 P

DS 201 P

hygienic applications

DS 400 P

silicon sensor
nominal pressure

0 ... 6 mbar to 0 ... 1000 mbar

accuracy

0.35 % FSO(according to IEC 60770)

characteristics

aluminium housing,
LED display,rotatable and configurable display modulepressure port

G 1/8“ internal

gases andcompressed air

application

stainless steel sensor / ceramic sensor
nominal pressure

0 ... 100 mbar to 0 ... 40 bar (DS 200 P)
0 ... 60 bar to 0 ... 400 bar (DS 201 P)

accuracy

0.25 / 0.35 % FSO (DS 200 P)
0.5 % FSO (DS 201 P)(according to IEC 60770)

characteristics

up to 4 contacts,
4-digit LED-display,rotatable and configurable display modulepressure portinch thread (flush), dairy pipie, clamp,
varivent® (DS 200 P)

optioncooling element up to 300 °C (DS 201 P)

hygienic version

application

stainless steel sensor
nominal pressure

0 … 100 mbar to 0 … 40 bar

accuracy

0.25 / 0.35 % FSO(according to IEC 60770)

characteristics

up to 2 contacts,
4-digit LED-display,rotatable and configurable display modulepressure portinch thread (flush), dairy pipe, clamp,
varivent®

optionIS-version,cooling element up to 300 °C

application

hygienic design

38

PRESSURE SWICH
without display

The pressure switches DS 4 and DS 6 were designed also for OEMs (original equipment manufacturers). These electronic pressure switches areused in hydraulic and pneumatic applications for monitoring and controlling the pressure via switching outputs.
The 1 or 2 freely programmable contacts whose status is indicated bydifferently coloured LED’s can be configured quickly and comfortably either by the optionally available tools P-Set (PC software and programming
adapter) or via the programming device P6.

silicon sensor

pneumatics

nominal pressure

0 … 1 bar up to 0 … 10 bar

contacts

1 or 2

pressure port

G 1/8“ internal thread,M5 internal thread

characteristics

configurable via PC or programming device

DS 4applicationcompact version

ceramic sensor

hydraulics

nominal pressure

0 … 2 bar to 0 … 400 bar

contacts

1 or 2

pressure port

G ¼“

optionoil and grease free,oxygen version

characteristics

configurable via PC or programming device

applicationhigh pressure

DS 6

39

EVALUATIONDISPLAYSDATA LOGGERAPPLICATIONS

40 - 42>> in situ display for pressure,temperature and level>> in situ display at pumping stations

43>> display panel for silo battery>> combined level and temperature
measurement in heated container>> pressure regulation of hydrauliccircuits>> filter monitoring>> pressure and level measurementin biogas plants>> pressure regulation / monitoringof test stands

40

DISPLAYS

In order to correctly interpret analogue signals, display and evaluation
devices are indispensable. Besides the classic version with display andanalogue outputs (PA 430, ASM 430), BD|SENSORS offers with the processdisplays of CIT-series an evaluation device that can be combined with ourpressure measuring devices and hydrostatic submersible probes and isfurthermore also suitable for acquiring for example temperature and potentiometer signals.
The multifunctional process transmitter CIT 400 has been exclusively
developed for supplying 2- and 3-wire sensors with current signal and for
acquiring measuring results. Two different types of housing and a combination of independent limit contacts and a freely configurable analogue output
are available. We are therefore able to offer you solutions for nearly every
measurement task.

PA 430plug-on display
display

4-digit LED display,
display and housing rotatablesignal input

4 … 20 mA / 2-wire
0 … 10 V / 3-wirecharacteristics

adjustable housing

optionIS-version,
up to 2 freely configurable contacts

dimensions

47 x 47 x 68 mm (W x H x D)

with contacts

PA 440field display
display

4-digit LCD display

signal input

4 … 20 mA / 2-wire
0 … 10 V / 3-wireoptionIS-version,
up to 2 freely configurable contacts

dimensions

plastic housing
120 x 80 x 57 mm (W x H x D)
aluminium housing
125 x 80 x 57 mm (W x H x D)

with contacts

DISPLAYS

41

CIT 200

process display
display

4-digit LED display

signal input

0/4 … 20 mA, 0/1 … 5 V
0/2 … 10 V

optionsignal input PT 100 / PT 500 / PT 1000,
remote control

dimensions

front panel housing
72 x 36 x 77 (95) mm (W x H x D)

low-cost process display

CIT 250

process display
display

4-digit LED display

signal input

0/4 … 20 mA, 0/1 … 5 V
0/2 … 10 V

optionsignal input PT100 / PT 500 / PT 1000,
up to 2 freely configurable contacts

dimensions

front panel housing
72 x 36 x 95 mm (W x H x D)

with contacts

CIT 300

process display
display

4-digit LED display

signal input

0/4 … 20 mA, 0/1 … 5 V
0/2 … 10 V

optionsignal input PT 100 / PT 500 / PT 1000,
2 or 4 freely configurable contacts,
analogue output

dimensions

front panel housing
96 x 48 x 98 mm (W x H x D)

with contacts
and analogue output

42

DISPLAYS

CIT 350

process display
display

4-digit LED display,
multicolour 20 segment bargraphsignal output

0/4 … 20 mA, 0/1 … 5 V
0 /2 … 10 V

option2 or 4 freely configurable contacts,
analogue output

dimensions

front panel housing
48 x 96 x 98 mm (W x H x D)

with bargraph

CIT 400

process display
display

4-digit LED display

signal input

0/4 … 20 mA

optionIS-version,
up to 4 limit value relays and 1 alarm relay

dimensions

front panel housing
72 x 72 x 110 mm (W x H x D)
housing for hat rail and wall mounting
70 x 75 x 110 (W x H x D)

with contacts
and analogue output

CIT 600

multichannel process display LCD
displaygraphic LCD display
128 x 64 pixel, background illuminatedsignal input

2 / 4 / 8 channels: 0/4 … 20 mA

optionsignal input PT 100 / PT 500 / PT 1000,
lockable door

dimensions

front panel housing
96 x 96 x 98 mm (W x H x D)

characteristics

communication interface RS 485 (Modbus RTU),
software for parameterization display

with contacts

DATA LOGGER

43

The data logger can record up to 2 million measurement values

The software helps the user displaying (table, graphic), saving,with a maximum measuring rate of 1 Hz in internal memory

evaluating and exporting the recorded data as well as issuing(8 MB). By connecting a USB stick, the capacity can be enlarged toreports and configurations.

several GB.

CIT 650

multichannel process display, LCD
displaygraphic LC display
128 x 64 pixel, background illuminatedsignal input

max. 8 channels: 0/4 ... 20 mA
0/1/2 ... 5/10 V

optionsignal input PT 100 / PT 500 / PT 1000,
USB-host-port, internal memory 8 MB,lockable door

dimensions

front panel housing
96 x 96 x 100 mm (W x H x D)

with data logger
and contacts

CIT 700

multichannel process display, TFT
displaygraphic TFT-display,
CIT 700: 3,5“, 320 x 240 pixel,
CIT 750: 5,7“, 320 x 240 pixel,
touchscreen

characteristics

3 freely fittable slots,
16 different input / output modules,transducer supply 24 VDC,
communication interfaces:
RS-485 (Modbus RTU ) master / slave,USB-host-port, USB device connection

CIT 750

with contacts andanalogue outputs and data logger

optioncommunication interfaces:
3 x RS-485, 1 x RS-232, 2 x USB-host-port,
Ethernet 10 MB, RJ-45 (Modbus TCP, Java Applets)dimensions

96 x 96 x 100 mm (W x H x D) CIT 700
144 x 144 x 100 mm (W x H x D) CIT 750

functional range
standard- configuration of max. 60 channels via inputs,
outputs, mathematical / logical functions,controller, profiles or virtual outputs- allocation of clogged channels in 10 groups (max. 6 channels per group)- connection of channels via mathematical / logical functions- 8 integrated PD / PI / PID controller

data logger- data logging of max. 60 channels- freely selectable measuring rate (max. 10 Hz)- extensive trigger functions- internal memory 1.5 GB

44

SPECIAL DEVICES

SPECIAL DEVICES

SPECIAL DEVICES

45

Among high-quality standard sensors and pressure transmitters, BD|SENSORS
has always been a synonym for flexible customer-oriented product solutions.
There is seldom a challenge that we cannot meet – we want to offer the
products that you really need in your application.
You have the feeling that your requirements cannot be fulfilled by our current
product line? Please contact us!

strain gauge

oil and gas industry

nominal pressure

0… 5000 psi to 0 … 15000 psi

accuracy

0.5 % FSO

HU 300(according to IEC 60770)

electricalconnection

MIL/ Bendix connection,
cable outlet,
Glenair (on request)

pressure port

WECO® 2“ (1502)
WECO® 2“ (2002/2202)

application

hammer unionceramic sensor

marine / shipbuilding / offshore

nominal pressure

0 … 60 mbar to 0 … 20 bar

accuracy

0.2 % FSO

(according to IEC 60770)

pressure port

G ¼“, hose connection

characteristics

with communication interface
for offset, span and damping adjustment

application

air-bubbling

EP 500

46

NOTES

47

INDUSTRIES

MEDIA

plant and machine engineering

sewagechemical and biochemical industry

aggressive media

energy industry

colours

renewable energy

gases

semiconductor industry / cleanroom technology

fuels and oils

HVAC

pasty and viscous media

hydraulics

oxygenrefrigeration

water

calibration techniques

laboratory techniques

medical technology

food and beveragevehicles and mobile hydraulics

oil and gas industry

pharmaceutical industry

marine / shipbuilding / offshore

heavy industry

environmental industry

packaging and paper industry

48

INDUSTRIES

heavy industry
The heavy industry sector – in particular the mining, heavy chemical, iron andsteel industries – places high demands on the housing, the electronics and thesensor element. No problem for BD|SENSORS, as our pressure transmitters canwithstand even the roughest process conditions and are characterized by••••high mechanical stability (shock and vibration resistance)
ATEX approval (ia = intrinsically safe version, xd = flameproof enclosure)dust ATEX zone 20
SIL (construction of pressure transmitters acc. to international safety standard)

environmental industry - water and waste water
BD|SENSORS sets standards in industrial and domestic water treatment with its elaborate
selection of pressure and level measurement devices. Whether drinking water, sludge oraggressive waste water is concerned – the demands on our hydrostatic submersible probes
could not be more diverse.
By using specific sensor technologies and robust housing materials (PP, PVC, PVDF, stainless steel etc.) as well as seal and cable sheath materials in combination with many years ofexperience, we assist you in choosing the correct transmitter type for your application.

chemical and petrochemical industry
From the production of colors and varnishes to synthetic fabrics, from the distillation to thestorage in tanks – an accurate monitoring and exact dosage is essential for the safety andmaximum productivity in chemical and petrochemical plants.
Precise measurements of our pressure and level transmitters in crucial places result in asaving of time, a higher productivity and reliability in the later production stages. We answer
all your questions and provide individual solutions!

oil and gas industry
The oil and gas industry becomes more and more important, as resources areactivated, new oil and gas fields are exploited and existing plants are modified
in order to cover the worldwide demand for oil and natural gas. For improving
the efficiency of those plants, higher and higher requirements are placed tocomponent suppliers. BD|SENSORS offers safe electronic pressure and levelmeasurement devices by:• carefully selecting and using oil and seawater resistant metal alloys andcable sheath materials• globally accepted approvals as GL, DNV, ATEX, UL etc.
• high reliability (SIL certification)• abrasion-resistant pressure sensors• ingress protection rates of IP 68 and higher.

INDUSTRIES

49

energy industry / renewable energies
A high efficiency, reliability and economic efficiency are the fundamental requests that plant
construction companies have in energy management. Whether fossil fuels, wind or water
power, solar or geothermal energy – system components such as pressure and level transmitters with an outstanding life expectancy and precision are of importance here.

food and luxury food industry
Due to rising demands on the hygienic conditions in the food and luxury food sector, ourpressure and level transmitters are part of a hygienic concept and have to comply with thespecific process requirements such as materials, CIP/ SIP capable sensors, surface roughness, adaption and design of process connections acc. to 3A standard and EHEDG as well aselastomeric seals.
BD|SENSORS offers with its pressure transmitters, pressure switches and level probes thecomplete equipment for measurements under alternating conditions or for cyclic cleaningand sterilization processes.

plant and mechanical engineering
The plant and mechanical engineering sector is situated in a complex global environment.
To be successful in this branch, reliability and flexibility in the measurement task (pressure
ranges, accuracy, electrical and mechanical connections), on-time deliveries as well as theprocessing of output signals are in the center of interest.
As a reliable supplier of electronic pressure measurement devices, BD|SENSORS offers complete and practical solutions for companies and measurement tasks of all kind. With our customized products, we are able to convert your challenge into an efficient process control.

marine / shipbuilding / offshore
The electronic pressure and level transmitters such as DMK 458, DMP 457 and LMK 458 faceextreme mechanical and climatic conditions on board of ships, harbor terminals, wind power
stations, drilling rigs etc. BD|SENSORS offers with its wide product range solutions for requirements such as:• resistance against seawater• vibration resistance and long-term stability• high safety aspect also in hazardous areas as well as an overcharging protection
for chemicals and LPG•	stability at extreme temperature changes•	highest accuracy on draught measurement• marine approvals such as GL (Germanischer Lloyd) and DNV (Det Norske Veritas).

50

4 ARGUMENTS

COMPETENCEPRICE / PERFORMANCE

Industrial pressure measurement technology

pressure measurement at the highest levelfrom 0.1 mbar up to 6000 bar
pressure transmitters, electronic pressure

The concentration on electronic pressure transmitter

switches or hydrostatic level probes

has led to extraordinary efficiency and economical

OEM or high-end products

pricing.

standard products or customized solutions
is certain to be one of the mosthas the right pressure measuring

economical suppliers on the world market, given equal

device at the right price.

technical and commercial conditions.

RELIABILITYFLEXIBILITY

projectable delivery times and strict

We have special solutions for your individual

observance of deadlines

requirement.

Short delivery times and firm deadlines, even for

We solve your problem in industrial pressure

special designs, makea reliable

partner for our customers.

measurement quickly and economically, not only withlarge-scale production lines, but also for smallerrequirements.

reduces the level of your stockkeeping and increases your profitability.

is especially flexible when technicalsupport and quick assistance are required in service
case as well as for rush orders.

51

KNOW-HOW

A successful cooperation to our

Know-how is the foundation for successfully

customer´s full satisfaction is our

producing high-quality electronic pressure
measurement devices.

motivation - developing together

Modern equipment in development and pro-high-quality competitive pressure

duction together with reliable partners are
the basic units which make this foundationstrong.

and level transmitters.
Customer-specific solutions, relia->> ISO 9001 certified

bility and flexibility combined with>> state-approved metrology centeran excellent price/performance ra->> accredited calibration laboratory

tio make us a competent partner for>> EMC-lab for norm-conforming tests>> state-of-the-art CNC production>> CIM production

pressure measurement at thehighest level.“Venture Plaza Funabashi 222, 1-17-25 Kitahoncho,
Funabashi-shi, Chiba 273-0864, Japan
Tel: +81-47-489-5939 Fax:+81-47-489-5940
https://all-device.com info@all-device.com
DOWNLOAD THIS BROCHURE

Status 03/2018
Technical changes
reserved.