S30SCURRENT SENSORS

1/44 1909

Servo system Current-output type

S30S D24 SERIESIntroduction

RoHS

ABSOLUTE MAXIMUM RATINGSUnitValue

Remarks

Maximum Supply voltage

Vcc

V± 25.2

Primary conductor temperture

TB℃

100

Symbol

Unit

Value

VdkVrms

AC6700V，for 1minute（Sensing current 0.5mA）

Primary ⇔ Secondary

Fluxgate system

SymbolISOLATION CHARACTERISTICS
Parameters
lnsulation voltage

Remarks

Vw

kV23

Primary ⇔ Secondary
Input waveform：
・Front time 1.2µs
・Time to half value 50µs・single

Insulation resistance

RIS―≧ 500M Ω（at DC500V）

Primary ⇔ Secondary

Clearance distancedCI―

min : 12.1
（min : 30.4）

Primary ⇔ Secondary（Busbar ⇔ ConnectorPWB）

Creepage distancedCp―

min : 35.5
（min : 33.0）

Primary ⇔ Secondary（Busbar ⇔ ConnectorPIN）

Case material――

UL94 V-0

Filler material――

UL94 V-0

CaseCTIV

400 ～ 599（Group II）

FillerCTI

V

600（Group Ⅰ）――

2.475kVa.c ,CAT Ⅲ , PD2――

1kVa.c,CAT Ⅲ , PD2
1.5kVd.cComparative Tracking Index；（CTI）

Application example

Closed loop

Impulse withstand voltage

Reinforced isolation, non uniform
field according to EN50178：1997
Reinforced isolation, non uniform fieldaccording to EN62477-1：2012 andEN62477-1：2012/A11 2014

ENVIRONMENTAL AND MECHANICAL CHARACTERISTICS
Parameters

SymbolUnit

Ambient operating temperature

TA

Ambient storage temperature
MassValue
MIN

TYP

MAX℃

－ 40―

＋ 85

TS℃

－ 40―

＋ 90mkg

―

1.3―

125

Open loop

Parameters

RemarksS30S

CURRENT SENSORSSPECIFICATIONSParameters

Introduction

TA=+25℃, RM=1Ω, Vcc=±24V

SymbolUnit

IPN

Remarks

MAXA

－2000

－IPM

A

3500－－at Ta=+85℃ ,Vcc= ± 22.8Vmin,
RM=1ohm, t=2sec

Measuring resistance ＊ 1RMΩ0－－See Fig1

Conversion ratio

KN－－

1 : 5000－

Output current @IPN

IomA－400－－ 0.2

0.0＋ 0.2

TA=25℃～ 85℃ ,Without Iof.

－ 0.3

0.0＋ 0.3

TA= － 40℃～ 85℃ ,Without Iof.

mA

－ 0.20.0＋ 0.2

at IP = 0A.

Accuracy @IPN
X

Offset current ＊ 3

Fluxgate system

Value
TYP

Primary nominal current

Open loop

4 1909MIN

Primary current, measuring range ＊ 1,2

Iof%

Io = IPN /5000. Without Iof.

Linearity error（0A ～ IPN）

εL％－ 0.1

0.0＋ 0.1

Hysteresis error

IOH

mA－ 0.2

0.0＋ 0.2

at Ip = 0A → IPN → 0A

Supply voltage

Vcc

VConsumption current

Icc

mA－45－

at Ip= 0A.
Icc = 45 ＋ Ip /5000.

Response time @90% of IPN ＊ 4trµs－0.5－

di/dt=100A/µs
See Fig2, at very low current± 15（± 5%）± 24（± 5%）

Frequency bandwidth（− 3dB）＊ 5BWkHz－150－

Temperature coefficient of Iof ＊ 3TcIofµA/℃－ 4.80＋ 4.8

at Ip= 0A

Secondary coil resistanceRsΩ－－24.5TA = ＋ 85℃＊1 C
 urrent sensor has limited operating time depending on the measured resistance and maximum current.
Internal circuits can become corrupted if you used beyond the limited time.
＊ 2 The value of measured current which indicates an output with a greater than ± 5% deviation from the theoretical output value.
＊ 3 Offset current is measured after removal of the hysteresis.
＊ 4 Measurement condition : Primary conductor cross sectional area is as same as through hole, and penetration with 1turn in through hole.
＊ 5 High fundamental frequency primary current and/or harmonic current may result in excessive heating in magnetic core (Silicon steel).

STANDARDSEN50178:1997, EN62477-1：2012 and EN62477-1：2012/A11 2014, UL508（No.E243511）, CSA22.2 No.14-13
Closed loop

2/4

※ Please refer to the another sheet about conditions of UL Recognition.

126

S30S

CURRENT SENSORS

3/4

4 1909TYPICAL CHARACTERISTIC CURVES

140
22.8V&85℃
14.25V&85℃

12080

Introduction

RM[Ω]1006040200500

1000

1500

2000

2500

3000

3500

IPN[A]

Fluxgate system

Figure 1： Maximum Measureing ResistanceOpen loop

Frequency characteristic

Measurement conditions:
TA=+25℃ , RM=2.4 Ω, IP=3A ＊ 20T, Vcc= ± 24V

Closed loop

Figure 2：Frequency response curve

CONNECTION

Figure 3：Block diagram

127

S30S

CURRENT SENSORS

4/44 1909

DIMENSIONS (mm)Terminal number

Introduction

S30S2T0D24Z
242
1

43

1 Vcc（＋24V）
2 Iout3 ーVcc（−24V）
4 N.C
Tolerance : ±0.5
Unit : mm3A
1Detail
2Connector

4Detail A

Maker

PartsNo.

old parts No.

3Molex39-28-8040

5566-04A-210

1

Fluxgate system

Example of Name Plate
Detail A
Example of Name Plate

Revision code(5 figure), Lot №（5 figure)Hall-Element position

SerialPlate№(4 figure), Factory code(2 figure)
Example of Name

Hall-Element position

Serial №(4 figure), Factory code(2 figure)

Revision code(5 figure), Lot №（5 figure)

S30S2T0D24ZM

Terminal number
Revision code(5figure),
Lot №（5 figure)
1 Vcc（＋24V）

Serial №(4 figure), Factorycode(2 figure)2 IoutHall-Element position

Open loop

3 ーVcc（ー24V）

Detailil A
D

Tolerance : ±0.5
Unit : mmConnector

Detailil A
DMakerPartsNo.

old parts No.

Molex

38-00-6293

AE-6410-03C(197)

Closed loop

Detailil A
DS30S2T0D24ZJ
Terminal number

Detail A
Detail A

1 ーVcc（−24V）
2 Iout3 Vcc（＋24V）
Tolerance : ±0.5
Unit : mm

Detail A

Connector

128

Maker

PartsNo.

old parts No.

JST

BH3P-VH-1ーNotices

CURRENT SENSORS

1/1

3 1510

Important Notice
7. This product is not designed to resist radiation.

1. The content of this information is subject to change without

vehicle equipment, safety devices, and so on) in which・U se in liquids such as water, oil, chemical solutions, ororganic solvents, and use in locations where the product will
be exposed to such liquids.
・U se that involves exposure to direct sunlight, outdoor
exposure, or dusty conditions.
・Use in locations where corrosive gases such as sea winds,
Cl2, H2S, NH3, SO2, or NO2, are present. (Some productimproves durability)・U se in environments with strong static electricity orelectromagnetic radiation.
・Use that involves placing inflammable material next to theproduct.
・Use of this product either sealed with a resin filling or coatedwith resin.
・Use of water or a water soluble detergent for flux cleaning.
・Use in locations where condensation is liable to occur.

extremely high quality and high reliability is required, or if

8. D o not use or otherwise make available the TAMUTA

the malfunction or failures of product could be cause loss ofproducts or the technology described in this document for any

human life, bodily injury.

military purposes, including without limitation, for the design,prior notice for the purpose of improvements, etc. Ensure thatyou are in possession of the most up-to-date information whenusing this product.
2. T his product is intended to be used in general electronics
applications (electric home appliances, business equipment,
information equipment, communication terminal equipment,
measuring devices, industrial equipment, and so on). This
product is neither intended nor warranted for use in following
equipment or devices:
Special application (such as for medical devices,transportation equipment, traffic signal control equipment,
fire and crime prevention equipment, aeronautics andspace devices, nuclear power control, fuel control, in-Tamura Corporation shall not be held responsible for any

development, use, stockpiling or manufacturing of mass

damage incurred by customers or any third party when products

destruction weapons (e.g. nuclear, chemical, or biologicalare used in special application, unless specifically permitted inweapons or missile technology products). When exporting

this document.

and re-exporting the products or technology described in thisdocument, you should comply with the applicable export control

3. Tamura Corporation constantly strives to improve quality andlaws and regulations and follow the procedures required byreliability, but malfunction or failures are bound to occur withsuch laws and regulations including, without limitation, Japansome probability in current sensor. To ensure that failures do-Foreign Exchange and Foreign Trade Control Law and U.S.-

not cause accidents resulting in injury or death, fire accidents,

Export Administration Regulations. The TAMURA productssocial damage, and so on, users are to thoroughly verify theand related technology should not be used for or incorporated

safety of their designs in devices and/or systems.

into any products or systems whose manufacture, use, or saleis prohibited under any applicable domestic or foreign laws or

4. The operation examples and circuit examples shown in thisregulations.

information are for reference purposes only, and Tamura
Corporation disclaims all responsibility for any violations of

9. Please contact your TAMURA sales office for details as toindustrial property rights, intellectual property rights and any

environmental matters such as the RoHS compatibility ofother rights owned by Tamura Corporation or third parties that

Product. Please use TAMURA products in compliance with allthese may entail.

applicable laws and regulations that regulate the inclusion or5. T he circuit examples and part constants listed in theseuse of controlled substances, including without limitation, thespecifications are provided as reference for the verification

EU RoHS Directive. TAMURA assumes no liability for damagesof characteristics. The user is to perform design, verification,or losses occurring as a result of your noncompliance withand judgment under his or her own responsibility, taking into

applicable laws and regulations.

account the various conditions.

10. T AMURA assumes no liability for damages or losses incurred by
you or third parties as a result of unauthorized use of TAMURA

6. T he products are designed for use in environments whereproducts.

consumer electronics are commonly used. It is not designed foruse in special environments such as listed below, and if such

11. T his document and any information herein may not be

use is considered, the user is to perform thorough safety andreproduced in whole or in part without prior written permission

reliability checks under his/her responsibility.

from TAMURA.

9Appli note

CURRENT SENSORS

1/1

3 1709Application notes<General Considerations><Open loop>

1. T he sensor uses polar electronic components. When the

1. High frequency primary current may result in excessive heating

polarity of the power supply is mistaken, the sensor is damaged.

in iron magnetic core and cause damage to internal circuitry;

2. Static electricity or excessive voltage can increase an offset

for high frequency applications select current sensor with ferrite

voltage in the Hall element, and cause offset voltage to change.

core material.
2. If the measured current exceeds the rated current, magnetic

Please exercise care in handling and application.

core saturation will occur and the output voltage signal will not

3. In order to prevent the influence of noise, the use of twisted

be linearly proportional to the measured current.

cable or shielded cable for the output line is recommended
4. If using this device within a magnetic field generated by otherdevices, the specified accuracy may not be obtainable.

<Closed Loop>

5. O ur products（several models are excluded ）are adjusted

1. For closed loop current sensors please insure the power supply

with the trimming method by the measurement condition

voltage is balanced, symmetrical, and, applied simultaneously（Load resistance, Power supply voltage）of specification

to avoid potential increase in DC offset error.

sheets. Therefore, characteristics（Offset, Output, etc.）and its

2. Maximum rated current measurement duration is timedependent.

deviation may be changed in different circuit conditions from

Maximum rated current applied in excess of the time limit canthe measurement condition. All change characteristic items are

result in damage to internal electronic circuitry; please consult

not indicated on specification sheets.

Tamura for assistance.

6. The performance of current sensors with through-hole（aperture）

3. When using a measurement resistor to convert current output

is dependent on the position of the primary conductor. Tamura

to voltage output select a resistor with stable temperature

specifications are based on a primary conductor completely

characteristic to insure accuracy of the output voltage.

filling the through-hole（aperture）area.

4. Compensation current supplied to the secondary winding varies

7. T he current sensor rated current in DC Amps.

in proportion to the measured current based on the conversion

8. Please use mating connector with equivalent terminal plating

ratio.（If/KN; KN ＝ secondary turns）Please insure the PSU

material to insure proper operation and avoid possibility ofhas required current capacity to supply compensation current‘galvanic corrosion’.

to the secondary winding.

9. P lease do not store in high-temperature and high-humidity
storage environment. Please use it after confirming soldering<Flux-Gate>

when it is kept for six months or more.（product soldered with

1. Compensation current supplied to the secondary winding variessubstrate）
10. We recommend performing a zero offset adjustment by measuring

in proportion to the measured current. Please insure the PSU

the offset voltage at startup. In continuously operation for a few

has required current capacity to supply compensation current
to the secondary winding.

months, or at change of ambient temperature or humidity is large,

2. T here is 450kHz ripple voltage present on the output andwe recommend regularly performing a zero offset adjustment atreference output voltage signals . An external capacitor maybe

being idling（it is clear that the current is not apply）.

added if necessary.

11. The current sensor doesn't have built-in protection circuit(devices and fuses, etc.). As a failure mode of the sensor,
there is a short circuit and open state. In the case of a shortcircuit state, the abnor-mal temperature rise of the internal
parts is assumed, and there is a possibility to smoke and toignite. If it is used in safety critical circuit blocks, please takeappropriate measures by protection devices, protection circuits,etc. For closed loop –type sensors and flux gate (closed looptype) sensors, the consumption current of the secondary power
supply varies in proportion to the measurement current.

7