McDANIEL
Liquid Fillable, All Stainless Steel
Utility Gauges
The rugged construction of ourSeries 9 All Stainless Steel gauges
provides resistance to corrosive
media and harsh environments.
The McDaniel Series 9 gauges
feature 316 Stainless Steel wettedparts and a 304 Stainless Steel casewith crimped on bezel ring.
Thecase and socket joint is welded.
Liquid fillable, in the field, orfactory.
This series is ideally suited forO.E.M. and corrosive applications
that require an all stainless steel
gauge at an economical price.

SPECIFICATIONS
Accuracy:
ASME B40.1 standard - Grade A (2-1-2%)

Dial:
White enameled aluminum. Black figures. Sizes:
2 1/2 inches (63mm) - Model K94 inches (100mm) - Models E9 and G9

Case:
304 Stainless Steel, filled or fillable.

Ring:
304 Stainless Steel crimped on. (Polished on Panel,
U-Clamp, and Back connected configurations)

Bourdon Tube:
316 Stainless Steel(C Form 1000 PSI & Below).
(Spiral 1500 PSI & Above).
Pressure Utilization:
300% full scale pressure w/o rupture of the bourdon tube.
130% full scale pressure without loss of accuracy.

Stem and Socket:

316 Stainless Steel. E9 ¹⁄₄“ NPT and G9 ¹⁄₂“ NPT
K9 ¹⁄₄“ NPT standard (¹⁄₈“ NPT by special order)
Removable restrictor screw standard on K9, optional on E9and G9 models. One piece welded case and stem.

Ranges Available

Pointer:

Vacuum, Compound, Rec 3-15, up to 10,000 psi

Black enameled aluminum.

Configurations Available:

Lens:Polycarbonate.
(Laminated Safety Glass by special order.)Precision Movement
Stainless Steel

Bottom connected standard(L) = Lower-back connected - K9(C) = Center-back connected - E9 and G9(U) = U-Clamp mounted, Lower-back connected
on K9, Center-back on E9 and G9(P) = Front Flange, Lower-back connected on K9,Center-back on E9 and G9(R) = Rear flange for Back or Bottom connections

McDaniel Controls, Inc.
P. O. Box 187, Luling, LA 70070 • 14148 Highway 90, Boutte, LA 70039 U.S.A.
(985) 758-2782 • New Orleans (504) 467-1333 • Fax (985) 758-1688
www.mcdanielcontrols.com

D I M E N S I O N A L D ATAAAIJBack Connection:Lower Back (L) - K9Center Back (C) - E9 & G9CL
FFU-Clamp (U)
CB

FS
T

GDKS

E H1200P
D H

HPanel Mount (P)Bottom Connection

MGP

GG

H= Models E9 & G9 are Center BackN

H

ModelK9E9G9Nominal
Size
63
2.48
100
3.94
100
3.94

(1) = 33.5±1 (mm)(2) = 116±0,1 (mm)(3) = 33.5±1 (mm)(4) = 64±0.5 (mm)Case
Diametermm
inchesmminches
mminches

A
68.5
2.68
106
4.17
106
4.17(5) = Ø 100.6 (mm)(6) = 99±0,2 (mm)(7) = 3 x 1200
(8) = 88±1 (mm)GB
CDEF
57.5316264(4) 7
2.24 1.20 2.44 2.51 0.27
78 33.5 (1) 99 (6) 100.6 (5) 7
3.07 1.32 3.89 3.96 0.27
88 (8) 33.5 (1) 99 (6) 100.6 (5) 7
3.48 1.32 3.89 3.96 0.27

G
120.47120.47
15
0.59

H
14
0.55
14
0.55
220.87

I
85
3.35
Ø
132
5.20
Ø
132
5.20

JK
L
M
75 3.53134
2.95 0.14 1.20 1.34(2)(7)(3)
116 4.8 33.5
32
4.57 0.19 1.32 1.26(2)(7)(3)
116 4.8 33.5
32
4.57 0.19 1.32 1.26

N
20.07
3.50.143.50.14

P
190.74
––––

S
25
0.98
27
1.06
40
1.57

T70
2.76
102
4.02
102
4.02

* See website for dimensional data and downloadable AutoCAD files: www.mcdanielcontrols.com

HOW TO ORDER
1 - Select the Model number (K9=2¹⁄₂” dial, ¹⁄₄” NPT. E9=4” dial, ¹⁄₄” NPT. G9=4” dial, ¹⁄₂” NPT.)

Range
code

Range
psi

K9
Minor
Increments

E9/G9
Minor
Increments

2 - Specify Range Code from list at left.
3 - Specify Type of Mounting:Special Configuration Adder Codes
B=psi/Bar (dual scale) (nonstock)

K = psi/kPa (dual scale) (nonstock)A
0 - 15.5.25(Bottom connection is standard, no code)B0 - 301.5
O = O2 (oxygen cleaned) (nonstock)C0 - 60
2
1“L” = (Lower-back connection - K9)¹⁄₈ = ¹⁄₈” NPT (nonstock)D
0 - 100
2
1
NH3 = ammonia scale“C” = (Center-back connected - E9 and G9)
E
0 - 160
52
F
0 - 200
4
2“U” = (U-Clamp, Lower-back connection on K9, Center-back connected on E9 and G9)
G
0 - 300
10
5“P” = (Front Flange, Lower-back connected on K9, Center-back on E9 and G9)
H0 - 400
105“R” = (Rear Flange for Back or Bottom connection - not shown)
I0 - 600
2010J0 - 800
20104 - Specify any other Special Configuration or options. (special NPT size, dual-scales, Ammonia-NH3, etc.)
K0 - 1000
20105 - Add Type of Filling, if appropriate. (GF=Glycerin Filled)L0 - 1500
5025A 2¹⁄₂” all stainless steel
M0 - 2000
4020Range
Type ofSpecial
Type of utility gauge with ¹⁄₄” NPT bottom
N0 - 3000
100504k0 - 4000
100
Model Code Mounting Configuration Filling connection standard, range of 100 psi,
O0 - 5000
100
50
filled at the factory with glycerin.
K9D––––
GF
6k0 - 6000
200100(K9D–GF)
P0 - 10000
200100VACUUM
VACUUM
VACUUM
VACUUM
Warning Glycerin filled gauges should not be used in applications involving strong
S
VAC 30”Hg - 01.5
oxidizing agents such as chlorine, nitric acid or hydrogen peroxide. (Refer to ASME B40.1)
RECEIVERRECEIVER RECEIVER
RECEIVER
T
REC 3 - 15**22** Receiver gauge is dual scale. (0 - 100% primary, 0 - 10 square root secondary)
COMPOUND COMPOUND COMPOUND COMPOUND
U30”Hg + 15
2-0-1
2-0-1
V30”Hg + 30
2-0-1
2-0-1
W30”Hg + 60
5-0-2
1-0-1
X30”Hg + 100
5-0-2
5-0-2
Y30”Hg + 160
5-0-5
5-0-5
McDaniel warrants our utility gauges to be free of defects in material and workmanship: Replacement of any defective
2k30”Hg + 200
10-0-510-0-5gauge
will be made at no cost to the purchaser. Gauge failures determined to be caused by over-range and incompatibility withZ30”Hg + 300
10-0-1010-0-5

Order:

Example:Limited Warranty

environment/product media or abuse will not be considered under this warranty.

DISTRIBUTED BY:
It is recommended that users of pressure gauges become
familiar with American National Standard ASME B40.1, entitledGauges, Pressure and Vacuum – Indicating Dial Type – Elastic
Element. This specification is available from:
American Society of Mechanical Engineers
3 Park Avenue
New York, New York 10016

Phone: 212-591-7722
K9E9G91201
George A. Rau & Co. Gretna, LA 70053 - 504-367-7822