Brief Operating Instructions

Proline t-mass B 150
Thermal Mass Flow Measuring System

These Instructions are Brief Operating Instructions; they do not replace the
Operating Instructions included in the scope of supply.
For detailed information, refer to the Operating Instructions and otherdocumentation on the CD-ROM provided or visit "www.endress.com/
deviceviewer".

KA01104D/06/EN/01.12
71165082

Table of contents

Proline t-mass B 150

Table of contents
1

Document information . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3

1.1 Document conventions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3

2

Basic safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

2.1
2.2
2.3
2.4
2.5

Requirements for the personnel . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Workplace safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Operational safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Product safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

3

Product description . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

5
5
6
6
6

3.1 Product design . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

4Incoming acceptance and product identification . . . . . . . . . . . . . . . . . . . . . 8

4.1 Incoming acceptance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8
4.2 Product identification . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 95Storage and transport . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10

5.1 Storage conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
5.2 Transporting the product . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10
5.3 Packaging disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 10

6

Installation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11

6.1 Installation conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11
6.2 Mounting the measuring device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16
6.3 Post-mounting check . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

7

Electrical connection . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20

7.1
7.2
7.3
7.4

Connection conditions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Connecting the measuring device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Ensuring the degree of protection . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Post-connection check . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

8Operation options . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24

20
21
22
23

8.1 Structure and function of the operating menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
8.2 Access to the operating menu via the local display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26
8.3 Access to the operating menu via the operating tool . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 32

9Commissioning . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 33

9.1
9.2
9.3
9.4
9.5
9.6

Function check . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Switching on the measuring device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Setting the operating language . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Configuring the measuring device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Defining the tag name . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Protecting settings from unauthorized access . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

33
33
34
34
35
36

10 Diagnostic information and remedial measures . . . . . . . . . . . . . . . . . . . . . 37

2

Endress+Hauser

Proline t-mass B 150

Document information

1

Document information

1.1

Document conventions

1.1.1

Safety symbols
SymbolMeaning

DANGERDANGER!
This symbol alerts you to a dangerous situation. Failure to avoid this situation will result in serious
or fatal injury.

A0011189-EN

WARNING
A0011190-EN

CAUTION
A0011191-EN

NOTICE
A0011192-EN

1.1.2

WARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result in serious
or fatal injury.
CAUTION!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result in minor
or medium injury.
NOTICE!
This symbol contains information on procedures and other facts which do not result in personal
injury.

Electrical symbols

SymbolMeaning
Direct current
A terminal to which DC voltage is applied or through which direct current flows.

A0011197

A0011198

Alternating current
A terminal to which alternating voltage (sine-wave) is applied or through which alternating current flows.

)

Ground connection
A grounded terminal which, as far as the operator is concerned, is grounded via a grounding system.

*

Protective ground connection
A terminal which must be connected to ground prior to establishing any other connections.

A0011200

A0011199

A0011201

1.1.3

Equipotential connection
A connection that has to be connected to the plant grounding system: This may be a potential equalizationline or a star grounding system depending on national or company codes of practice.

Tool symbols

A0011219

Phillips head screwdriverEndress+Hauser

A0011220

Flat blade screwdriver

A0011221

Allen key

A0011222

Hexagon wrench3Document information

1.1.4

Proline t-mass B 150

Symbols for certain types of information

SymbolMeaning
AllowedIndicates procedures, processes or actions that are allowed.

A0011182

Preferred
Indicates procedures, processes or actions that are preferred.
A0011183

Forbidden
Indicates procedures, processes or actions that are forbidden.
A0011184

Tip
Indicates additional information.
A0011193

Reference to documentation
Refers to the corresponding device documentation.
A0011194

Reference to page
Refers to the corresponding page number.
A0011195

Reference to graphic
Refers to the corresponding graphic number and page number.
A0011196,,

Series of steps…

Result of a sequence of actions

Ã

1.1.5

Symbols in graphics
SymbolMeaning

1, 2, 3,...

Item numbers,,

Series of steps…

A, B, C, ...

Views

A-A, B-B, C-C, ...

Sections
Flow direction

A0013441-

A0011187

Hazardous area
Indicates a hazardous area.
Safe area (non-hazardous area)
Indicates a non-hazardous location.

.
A0011188

4

Endress+Hauser

Proline t-mass B 150

2

Basic safety instructions

2.1

Requirements for the personnel

Basic safety instructions

The personnel must fulfill the following requirements for its tasks:► Trained, qualified specialists must have a relevant qualification for this specific function andtask► Are authorized by the plant owner/operator► Are familiar with federal/national regulations► Before beginning work, the specialist staff must have read and understood the instructions in
the Operating Instructions and supplementary documentation as well as in the certificates(depending on the application)► Following instructions and basic conditions

2.2

Designated use

Application and media
The measuring device described in these Instructions is intended only for flow measurement ofgases.
To ensure that the measuring device remains in proper condition for the operation time:► Only use the measuring device in full compliance with the data on the nameplate and thegeneral conditions listed in the Operating Instructions and supplementary documentation.
► Based on the nameplate, check whether the ordered device is permitted for the intended use
in the hazardous area (e.g. explosion protection, pressure vessel safety).
► Use the measuring device only for media against which the process-wetted materials areadequately resistant.
Incorrect use
The manufacturer is not liable for damage caused by improper or non-designated use.
If the sensor gland is opened, the accuracy specifications of the measuring device no longer apply.
In such situations, the measuring device must be removed and returned for recalibration and tothe manufacturer.

!WARNING
Risk of injury if the process connection and sensor gland are opened under pressure.
► The process connection and the sensor gland should only be opened in an unpressurized state.
NOTICE
Dust and moisture can enter the transmitter when the transmitter housing is opened.
► Only open the transmitter housing briefly, ensuring that no dust or moisture enters the
housing.
NOTICE
Danger of breakage of the sensor due to corrosive or abrasive fluids!
► Verify the compatibility of the process fluid with the sensor material.
► Ensure the resistance of all fluid-wetted materials in the process.
► Observe the specified maximum process pressure.
Endress+Hauser

5Basic safety instructions

Proline t-mass B 150

Verification for borderline cases:► For special fluids and fluids for cleaning, Endress+Hauser is glad to provide assistance inverifying the corrosion resistance of fluid-wetted materials, but does not accept any warranty
or liability.
Residual risks
The external surface temperature of the housing can increase by max. 15 K due to the power
consumption of the electronic components. Hot process fluids passing through the measuring
device will further increase the surface temperature of the housing. The surface of the sensor,
in particular, can reach temperatures which are close to the fluid temperature.
Possible burn hazard due to fluid temperatures!
► For elevated fluid temperature, ensure protection against contact to prevent burns.

2.3

Workplace safety

For work on and with the device:► Wear the required personal protective equipment according to federal/national regulations.
For welding work on the piping:► Do not ground the welding unit via the measuring device.

2.4

Operational safety

Risk of injury.
► Operate the device in proper technical condition and fail-safe condition only.
► The operator is responsible for interference-free operation of the device.

2.5

Product safety

This measuring device is designed in accordance with good engineering practice to meet stateof-the-art safety requirements, has been tested, and left the factory in a condition in which they
are safe to operate.
It meets general safety standards and legal requirements. It also complies with the EC directives
listed in the device-specific EC Declaration of Conformity. Endress+Hauser confirms this by
affixing the CE mark to the device.

6

Endress+Hauser

Proline t-mass B 150

Product description

3

Product description

3.1

Product design
1

2

3

4

5

6

2

7

2

6

2

5

2

4

2

3

2

2

2
1

Es

c

8

7

A0017196

1
2
3
4
5
6
7
8

Electronics compartment cover
Display module
Terminal block
Electronics module
Cable glandTransmitter housing
SensorS-DATEndress+Hauser

7

Incoming acceptance and product identification

Proline t-mass B 150

4Incoming acceptance and product identification

4.1

Incoming acceptance

A0015502

1
+
2

1
+
2
A0013843

A0013695

A0015502

A0013698

A0015502

A0013699

8

Endress+Hauser

Proline t-mass B 150

Incoming acceptance and product identification

A0015502

A0013697

If one of the conditions does not comply, contact your Endress+Hauser distributor.

4.2

Product identification

The following options are available for identification of the measuring device:• Nameplate specifications• Order code with breakdown of the device features on the delivery note
• Enter serial numbers from nameplates in W@M Device Viewer(www.endress.com/deviceviewer): All information about the measuring device is displayed.
For an overview of the scope of the Technical Documentation provided, refer to the following:
The W@M Device Viewer : Enter the serial number from the nameplate(www.endress.com/deviceviewer)

12Order code:
Ser. no.:Ext. ord. cd.:

3

A0014053

å1
1
2
3

Example of a nameplate
Order code
Serial number (Ser. no.)Extended order code (Ext. ord. cd.)

For detailed information about interpreting the nameplate specifications, refer to the
Operating Instructions for the device on the CD-ROM provided

Endress+Hauser

9Storage and transport

5

Storage and transport

5.1

Storage conditions

Proline t-mass B 150

Observe the following notes for storage:• Store in original packaging.
• Do not remove the protection cap mounted on the transducer.
• Protect from direct sunlight.
• Select a storage location where moisture cannot collect in the measuring device.
• Store in a dry and dust-free place.
• Do not store outdoors.
• Storage temperature(® ä 15)

5.2

Transporting the product

Observe the following notes during transport:• Transport the measuring device to the measuring point in the original packaging.
• Do not remove the protection cap mounted on the transducer. It prevents mechanical damage
and contamination.

5.3

Packaging disposal
For detailed information about disposing of the packaging materials, refer to the Operating
Instructions for the device on the CD-ROM provided

10

Endress+Hauser

Proline t-mass B 150

6Installation

6.1

Installation conditions

Installation

For mechanical reasons and to protect the pipe, support is recommended for heavy sensors (e.g.
with a hot-tap retractable assembly).
6.1.1

Mounting position

Orientation
The direction of the arrow on the sensor helps you to install the sensor according to the flow
direction.
Orientation

Recommendation
1) 2)

Vertical orientation

A0017337

Horizontal orientation, transmitter head up

A0015589

3)

Horizontal orientation, transmitter head
down

A0015590

4)aInclined mounting position, transmitter head
down

A0015773

1)
2)
3)
4)

In the case of saturated or unclean gases, upward flow in a vertical pipe section is preferred to minimize condensationor contamination.
Not recommended in the case of extreme vibrations or unstable installations.
Suitable only for clean and dry gases. If buildup or condensate are always present: Mount the sensor in an inclined
position.
Select inclined mounting position (a = approx. 135°) if the gas is very wet or saturated with water.

Requirement for pipework
The measuring device must be professionally installed, and the following points must be
observed:

Endress+Hauser

11

Installation

Proline t-mass B 150

Further information ® ISO standard 14511

A0005103

Correctly aligned flanges and seals

A0005105

Incorrectly sized seal

A0005106

A0005104

Incorrectly aligned flanges and seals Internal diameter of pipe does not correspondto internal diameter of sensor

Choosing the sensor length►

12

Determining the dimensions A, B, C1 and C2

Endress+Hauser

Proline t-mass B 150

Installation

9
9

230
220

230

9

230
220

220
210
210
8

8

210

200

200
190

190
180
7

200
190
180

7

A

A

B

B

C1 C2

8

C1 C2

180
7

A0015768

A
B
C1
C2

Internal pipe diameter DN (circular pipe) or internal dimension (rectangular duct)
Thickness of pipe wall or of duct wall
Length of mounting set
Length of sensor compression fitting

Determining C1 and C2 (Endress+Hauser original parts only)
DK6MB-BXA mounting boss G1A

C1 + C2 = 99 mm (3.90 in)

DK6MB-DXA mounting boss G3/4A

C1 + C2 = 99 mm (3.90 in)

DK6MB-AXA mounting boss 1" NPT

C1 + C2 = 107 mm (4.21 in)

DK6MB-CXA mounting boss 3/4" NPT

C1 + C2 = 102 mm (4.02 in)Determining C1 and C2 (not limited to Endress+Hauser original parts)
C1

Length of pipe connection used

C2 (compression fitting with G1A thread)

39 mm (1.54 in)C2 (compression fitting with G3/4A thread)

39 mm (1.54 in)C2 (compression fitting with 1" NPT thread)

47 mm (1.85 in)C2 (compression fitting with 3/4" NPT thread)

42 mm (1.65 in)►

Calculating insertion depth(0.3 · A) + B + (C1 + C2)

Endress+Hauser

13Installation

Proline t-mass B 150

Mounting conditions for mounting boss

D

90°

90°

A0011843

D= 31.0 mm ± 0.05 mm (1.22 in ± 0.02 in)►When installing in rectangular ducts with thin walls:
Ã Use suitable support brackets.

Align the insertion version with the direction of flow.

90° (±3°)90°(±7°)A0015746

Check and ensure that the sensor on the pipe/duct is aligned at a 90° angle to the direction of
flow. Rotate the sensor so that the arrow marking on the sensor body corresponds to the directionof flow. The line marking on the body used to adjust the insertion depth must be aligned withthe direction of flow.
Inlet and outlet runs
For the dimensions and installation lengths of the device, see the "Technical Information"
document, "Mechanical construction" section

14

Endress+Hauser

Proline t-mass B 150

Installation

Recommended inlet and outlet runs

1

2
20 × DN

5 × DN

25 × DN

5 × DN4

3
20 × DN

5 × DN

50 × DN

5 × DN

20 × DN

5 × DN

40 × DN

5 × DN

6

5A0016943

1
2
3
4
5
6

reduction
expansion
90° elbow or T-piece
2 × 90° elbow
Control valve
2 × 90° elbow (3-dimensional)

Outlet run for pressure or temperature transmitterPTTT

2…5 × DN

A0015603

PT
TT

Pressure measuring device
Temperature measuring device

6.1.2

Requirements from environment and process

Ambient temperature range
Measuring device–40 to +60 °C (–40 to +140 °F)

Local display–20 to +60 °C (–4 to +140 °F), the readability of the display may be impaired attemperatures outside the temperature range.

►

If operating outdoors:

Endress+Hauser

15

Installation

Proline t-mass B 150

Avoid direct sunlight, particularly in warm climatic regions.
System pressure
SensorDepending on the version, please note the details on the name plate .
Max. 20 bar g (290 psi g)
Thermal insulation

A0015763

6.2

Mounting the measuring device

6.2.1

Required tools

For transmitter
For turning the transmitter housing (in increments of 90°): Allen screw 4 mm (0.15 in)
For sensor
For the sensor gland: Corresponding mounting tools
6.2.2

Preparing the measuring device

1.

Remove all remaining transport packaging.

2.

Remove the protective cap from the sensor.

3.

Remove stick-on label on the electronics compartment cover.

6.2.3

Mounting the measuring device!WARNING
Danger due to improper process sealing!
► Ensure that the gaskets are clean and undamaged (G 1 A, G ¾ A).
► Ensure that the correct sealing material has been used (e.g. Teflon tape for NPT 1", NPT ¾").
► Install the gaskets correctly.
16

Endress+Hauser

Proline t-mass B 150

1.

Installation

Ensure that the direction of the arrow on the sensor matches the flow direction of themedium.

2.
9

230
220
210

8

200
190
180

7

1½
30 mm

A0017331

Ensure the insertion depth and alignment are correct. Thread adapter nut: when mounting
for the first time, tighten it fingertight + 1½ turns. For all other mounting routines, tighten
it fingertight and 1 additional turn3.

Install the measuring device or turn the transmitter housing so that the cable entries do
not point upwards.
Ã

A0013964

6.2.4

Turning the transmitter housing

To provide easier access to the connection compartment or display module, the transmitter
housing can be turned clockwise or counterclockwise to 4 indexed positions by a maximum of
2 x 90°:Endress+Hauser

17Installation

Proline t-mass B 150

max. 2 × 90°

4 mm

4 mm
A0017227

6.2.5

Turning the display moduleEscA0017228

6.3

Post-mounting check

Is the device damaged (visual inspection)?

q

Does the measuring device conform to the measuring point specifications?
For example:• Process temperature• Process pressure (refer to the chapter on "Material load curves" of the "Technical Information"
document on the CD-ROM provided)• Ambient temperature range (® ä 15)• Measuring range

18

qEndress+Hauser

Proline t-mass B 150

Installation

Has the correct orientation for the sensor been selected (® ä 11)?
• According to sensor type
• According to medium properties• According to medium temperature• According to process pressureq

Does the arrow on the sensor match the direction of flow of the medium through the piping(® ä 11)?

q

Have sufficient inlet and outlet runs been provided upstream and downstream of the measuring point?

q

Correctly aligned in the direction of flow?

q

Correct sensor immersion depth?

q

Is the device adequately protected from precipitation and direct sunlight?

q

Is the device protected against overheating?

q

Is the device protected against excessive vibrations?

q

Check gas property (e.g. purity, dryness, cleanness).

q

Are the measuring point identification and labeling correct (visual inspection)?

qEndress+Hauser

19Electrical connection

Proline t-mass B 150

7Electrical connection

7.1

Connection conditions

7.1.1

Required tools• For cable entries: Use corresponding tools• Wire stripper• When using stranded cables: Crimping tool for wire end ferrule• Flat blade screwdriver£ 3 mm (0.12 in)
7.1.2

Requirements for connecting cable

The connecting cables provided by the customer must fulfill the following requirements.
Electrical safety
In accordance with applicable federal/national regulations.
Cable specification
Permitted temperature range:• –40 °C (–40 °F)...³ 80 °C (176 °F)• Minimum requirement for cable temperature range: ambient temperature +20 K
Current output
For 4-20 mA HART: Shielded cable recommended. Observe grounding concept of the plant.
Pulse/frequency/switching output
Standard installation cable is sufficient
Cable diameter• Included cable glands: M20 × 1.5 with cable Æ 6 to 12 mm (0.24 to 0.47 in)• Core cross-sections 0.5 to 1.5 mm2 (21 to 16 AWG)
7.1.3

Requirements for the supply unit

Device supply voltage
DC 24 V (18 to 30 V)
The power supply circuit must comply with SELV/PELV requirements.
Pulse/frequency/status supply voltage
An external power supply is required for each output.
Order characteristic for"Output"

Maximum terminal voltage

Option B, K

DC 30 VLoad
0 to 750 W, depending on the external supply voltage of the power supply unit

20

Endress+Hauser

Proline t-mass B 150

7.1.4

Electrical connection

Terminal assignment

The terminal assignment for the electrical connection can be found on the nameplate of theelectronics module.

1 L+
2 L–
22 +
23 –
24 +
25 –
26 +
27 –

Power Supply
DC 24 V

Pulse/freq./switch output
4…20 mA HART

A0017222

7.1.5

Preparing the measuring device

1.

Remove dummy plug if present.

2.

NOTICE! Insufficient sealing of the housing. Operational reliability of the measuring
device could be defeated. Use suitable cable glands corresponding to the degree ofprotection.
If measuring device is delivered without cable glands:
Provide suitable cable gland for corresponding connecting cable .(® ä 20)

3.

If measuring device is delivered with cable glands:
Observe cable specification (® ä 20).

7.2

Connecting the measuring device

NOTICE
Limitation of electrical safety due to incorrect connection!
► SELV/PELV-compliant 24 V DC (18 to 30 V) power supply.
► 4 to 20 mAHART active► Maximum output values: DC 24V, 22 mA, load 0 to 750 Ω

Endress+Hauser

21

Electrical connection

7.2.1

Proline t-mass B 150

Connecting the cables

20 mm

1
2
22
23
24
25
26
27

3 (0.12)
1
2
22
23
24
25
26
27

–

+

E

10 (0.4)mm (in)A0017250►

NOTICE! Housing degree of protection voided due to insufficient sealing of the housing.
Screw in the thread without using any lubricant. The threads on the cover are coated witha dry lubricant.
Reverse the removal procedure to reassemble the transmitter.
For HART communication: When connecting the cable shielding to the ground terminal,
observe the grounding concept of the facility.

7.3

Ensuring the degree of protection

The measuring device fulfills all the requirements for the IP66 and IP67 (Type 4X enclosure)degree of protection.
To guarantee IP 66 and IP 67 degree of protection (Type 4X enclosure), carry out the following
steps after the electrical connection:
1.

Check whether the housing seals of the connection and electronics compartment are clean
and inserted correctly. Dry, clean or replace the seals if necessary.

22

Endress+Hauser

Proline t-mass B 150

Electrical connection

2.

Tighten all housing screws and screw covers.

3.

Firmly tighten the cable glands.

4.

To ensure that moisture does not enter the cable entry, route the cable so that it loops
down before the cable entry ("water trap").
Ã

5.

Insert dummy plugs into unused cable entries.

A0013960

7.4

Post-connection check

Are cables or the device undamaged (visual inspection)?

q

Are the power supply and signal cables correctly connected?

q

Does the supply voltage correspond to the specifications in the connection diagram?

q

Do the cables comply with the requirements (® ä 20)?

q

Do the cables have adequate strain relief? Are they routed securely?

q

Is the cable route completely isolated? Without loops and cross-overs?

q

Are all the screw terminals firmly tightened?

q

Are all the cable glands installed, firmly tightened and leak-tight? Cable run with "water trap" (® ä
20) ?

q

Does the supply voltage match the specifications on the transmitter nameplate (® ä 20)?

q

Is the terminal assignment correct (® ä 20)?

q

If supply voltage is present, is the device ready for operation and do values appear on the display module?

q

Are all housing covers installed and firmly tightened?

qEndress+Hauser

23

Operation options

Proline t-mass B 150

8Operation options

8.1

Structure and function of the operating menu

8.1.1

Structure of the operating menu

Language
Display/operat.

Setup

task-oriented

Operator

MaintenanceOperating menu for operators and maintenances

Diagnostics

Expertfunction-oriented

ExpertOperating menu for experts

A0014058-EN

24

Endress+Hauser

Proline t-mass B 150

8.1.2

Operation options

Operating philosophy

The individual parts of the operating menu are assigned to certain user roles. Each user rolecontains typical tasks within the device lifecycle.
For detailed information about the operating philosophy of the instrument, refer to the
Operating Instructions for the device on the CD-ROM provided

Endress+Hauser

25

Operation options

8.2

Proline t-mass B 150

Access to the operating menu via the local display

A0014013

1
1.1
1.2
1.3
1.4
1.5
1.6
1.7
2
2.1
2.2
2.3
2.4
2.5
2.6
3
3.1
3.2
4
5

26

Operational display with measured value shown as "1 value, max." (example)
Device tag
Display area for measured values (4-line)
Explanatory symbols for the measured value: measured value type, measuring channel number,symbol for event behavior
Status area
Measured value
Unit for the measured value
Operating elements
Operational display with measured value shown as "1 bar graph + 1 value" (example)
Bar graph display for measured value 1
Measured value 1 with unit
Explanatory symbols for measured value 1: measured value type, measuring channel number
Measured value 2
Unit for measured value 2
Explanatory symbols for measured value 2: measured value type, measuring channel number
Navigation view: picklist of a parameter
Navigation path and status area
Display area for navigation: Ã designates the current parameter value
Editing view: text editor with input mask
Editing view: numeric editor with input mask

Endress+Hauser

Proline t-mass B 150

8.2.1

Operation options

Operational display

Status area
Status signals

A0013956

FailureA0013959

Function check

A0013958

Out of specification

Diagnostic behavior

Locking

A0013961

Alarm

A0013962

Warning

A0013957

Maintenance required
Communication

A0013965

A0013963

Device locked

Remote operation enabled

Display area
Measured variables
SymbolMeaning
Corrected volume flow, FAD
A0013711

Mass flow
A0013710

Temperature
A0013947

Totalizer
A0013943

Current output
A0013945

Symbols for measurement channel numbers
Measurement channel 1 to 4
A0016325

The measurement channel number is displayed only if more than one channel is present for the same measured variable
type.
Symbols for diagnostic behavior
The diagnostic behavior pertains to a diagnostic event that is relevant to the displayed measured variable.
For more information about the symbols, refer to the "Status area" section (® ä 27).

8.2.2

Navigation view

Status area
The following appears in the status area of the navigation view in the top right corner:• Of the submenu– The direct access code for the parameter you are navigating to (e.g. 0022-1)– If a diagnostic event is present, the diagnostic behavior and status signal• In the wizard
If a diagnostic event is present, the diagnostic behavior and status signal
Endress+Hauser

27

Operation options

Proline t-mass B 150

Display area
Icons for menus

A0013974

A0013973

Display/operat.

A0013975

Setup

Diagnostics

Icons for submenus, wizards, parameters

A0013967

8.2.3

Lock symbols

A0013968

Submenu

A0013966

ExpertWizardA0013972

Parameters within a wizard

A0013963

Parameter locked

Editing view

Input mask
Operating symbols in the numeric editor

A0016621

A0013985

Confirms selection.

Moves the input position one position
to the left.

A0013986

Exits the input without applying thechanges.

–.
A0016619

A0016620

Inserts decimal separator at the input Inserts minus sign at the input position.
position.

A0014040

Clears all entered characters.

Operating symbols in the text editor

A0013985

Confirms selection.

A0013987

Switches to the selection of thecorrection tools.

A0013986

Exits the input without applying thechanges.

Aa1
A0014040

Clears all entered characters.

A0013981

Toggle• Between upper-case and lower-caseletters• For entering numbers• For entering special characters

Correction symbols underA0013989

A0013990

A0013991

A0013988

Clears all entered characters. Moves the input position one Moves the input position one
Deletes one character
position to the left.
position to the right.
immediately to the left of the
input position.

28

Endress+Hauser

Proline t-mass B 150

8.2.4

Operation options

Operating elements

Key

Meaning
Minus key
In a menu, submenu
Moves the selection bar upwards in a choose list.
A0013969

With a Wizard
Confirms the parameter value and goes to the previous parameter.
With a text and numeric editor
In the input mask, moves the selection bar to the left (backwards).
Plus key
In a menu, submenu
Moves the selection bar downwards in a choose list.

A0013970

With a Wizard
Confirms the parameter value and goes to the next parameter.
With a text and numeric editor
Moves the selection bar to the right (forwards) in an input screen.
Enter key
For operational display• Pressing the key briefly opens the operating menu.
• Pressing the key for 2 s opens the context menu.

A0013952

In a menu, submenu• Pressing the key briefly:– Opens the selected menu, submenu or parameter.
– Starts the wizard.
– If help text is open, closes the help text of the parameter.
• Pressing the key for 2 s for parameter:
If present, opens the help text for the function of the parameter.
With a Wizard
Opens the editing view of the parameter.
With a text and numeric editor• Pressing the key briefly:– Opens the selected group.
– Carries out the selected action.
• Pressing the key for 2 s confirms the edited parameter value.
Escape key combination (press keys simultaneously)
In a menu, submenu• Pressing the key briefly:– Exits the current menu level and takes you to the next higher level.
– If help text is open, closes the help text of the parameter.
• Pressing the key for 2 s returns you to the operational display ("home position").

+
A0013971

With a Wizard
Exits the wizard and takes you to the next higher level.
With a text and numeric editor
Closes the text or numeric editor without applying changes.

Endress+Hauser

29

Operation options

Proline t-mass B 150

Key

Meaning
Minus/Enter key combination (press the keys simultaneously)+
A0013953

Reduces the contrast (brighter setting).
Plus/Enter key combination (press and hold down the keys simultaneously)+
A0013954

Increases the contrast (darker setting).
Minus/Plus/Enter key combination (press the keys simultaneously)++
A0013955

8.2.5

For operational display
Enables or disables the keypad lock.

Opening the context menu

Using the context menu, the user can call up the following three menus quickly and directlyfrom the measured value display:• Setup• Conf. backup disp.
• Simulation
Calling up and closing the context menu
The user is in the measured value display.
1.

2 s Press F forÃ The context menu opens.

XXXXXXXXXX
Setup
Conf.backup disp
Simulation

20.50
kg/h
A0014003-EN

2.

Press S + O simultaneously.
Ã The context menu is closed and the measured value display appears.

Calling up the menu via the context menu
1.

Open the context menu.

2.

Press O to navigate to the desired menu.

3.

Press F to confirm the selection.
Ã The selected menu opens.

8.2.6

Calling up help text

Help text is available for some parameters and can be called up from the navigation view. The
help text provides a brief explanation of the parameter function and thereby supports swift andsafe commissioning.
30

Endress+Hauser

Proline t-mass B 150

Operation options

Calling up and closing the help text
The user is in the navigation view and the selection bar is on a parameter.
1.

2 s Press F forÃ The help text for the selected parameter opens.

2.

Press S + O simultaneously.
Ã The help text is closed.

8.2.7

User roles and related access authorization

The two user roles "Operator" and "Maintenance" have different write access to the parameters
if the customer defines a user-specific access code. This protects the device configuration via thelocal display from unauthorized access (® ä 36).
Access authorization to parameters
User role

Read access

Write accessWithout accesscode(from the factory)

With access code

Without accesscode(from the factory)

With access code

Operator

Ã

Ã

Ã

-- 1)

Maintenance

Ã

Ã

Ã

Ã

1)

Despite the defined access code, certain parameters can always be modified and thus are excepted from the write
protection, as they do not affect the measurement (e.g. "Format display").

For detailed information about parameters which can always be modified, refer to the Operating
Instructions for the device on the CD-ROM provided
If an incorrect access code is entered, the user obtains the access rights of the "Operator" role.
The user role with which the user is currently logged on is indicated by the Access status
display parameter. Navigation path: Display/operation ® Access status display
8.2.8

Disabling write protection via access code

If the -symbol appears on the local display in front of a parameter, the parameter is writeprotected by a user-specific access code and its value cannot be changed at the moment using
the local display(® ä 36).
The locking of the write access via local operation can be disabled by entering the customerdefined access code via the respective access option.
1.

After you press F, the input prompt for the access code appears.

2.

Enter the access code.
Ã The -symbol in front of the parameters disappears; all previously write-protectedparameters are now re-enabled.

Endress+Hauser

31Operation options

8.2.9

Proline t-mass B 150

Enabling and disabling the keypad lockThe keypad lock makes it possible to block access to the entire operating menu via local operation.
As a result, it is no longer possible to navigate through the operating menu or change the valuesof individual parameters. Users can only read the measured values on the operational display.
The keypad lock is enabled and disabled in the same way:
The user is in the operational display.
►

By simultaneously pressing the S + O + F keys.
Ã After enabling the keypad lock:

XXXXXXXXXX

20.50

Keylock on

kg/hA0014000-EN

8.3

Access to the operating menu via the operating tool
For detailed information about access to the operating menu via operating tool, refer to the
Operating Instructions for the device on the CD-ROM provided

8.3.1

Via HART protocol

This communication interface is present in the following device version:• Order characteristic for "Output", option A: 4-20 mA HART
• Order characteristic for "Output", option B: 4-20 mA HART, pulse/frequency/switching
output32

Endress+Hauser

Proline t-mass B 150

Commissioning

7
4

2

1

6

3

5
A0017373

å2
1
2
3
4
5
6
7

Options for remote operation via HART protocol
Control system (e.g. PLC)
Field Communicator 475
Computer with operating tool (e.g. FieldCare, AMS Device Manager, SIMATIC PDM)
Commubox FXA195 (USB)
Field Xpert SFX100
VIATOR Bluetooth modem with connecting cable
Transmitter

9

Commissioning

9.1

Function check

Before commissioning the device, make sure that the post-installation and post-connection
checks have been performed.
• "Post-mounting check" checklist (® ä 18)• "Post-connection check" checklist (® ä 23)

9.2

Switching on the measuring device

After a successful function check, switch on the measuring device.
After a successful startup, the local display switches automatically from the startup display to themeasured value display.
If nothing appears on the local display or a diagnostic message is displayed, refer to thedevice's operating instructions which can be found on the CD-ROM supplied with thedevice.
Endress+Hauser

33

Commissioning

9.3

Proline t-mass B 150

Setting the operating language

Factory setting: English or ordered local language

XXXXXXXXX

20.50
mA
Main menu
Language

0104-1EnglishDisplay/operat.
Setup
LanguageÃ English
DeutschEspañol
Français0104-1

LanguageÃ English
DeutschEspañol
Français0104-1

Hauptmenü
Sprache0104-1
DeutschAnzeige/Betrieb
Setup
A0013996

9.4

Configuring the measuring device

The Setup menu and its guided wizards enable fast commissioning of the measuring device. The
wizards guide you systematically through all parameters that are required for configuration, suchas parameters for measurement, outputs and the local display.
Overview of the "Setup" menu
Setup

34

®
Select gas type®

Process pressure®

Temperature®
Endress+Hauser

Proline t-mass B 150

Commissioning

Pipe shape®

Pipe inner diameter®

Duct internal height®

Duct internal width®

Installation factor®

Assign current output®

4mA value®

20mA value®

Operating mode®
Pulse
Frequency
Switching output

Options

Meaning

Select gas typeDefine the gas type

Process pressure

Define the process pressure

Temperature

Display the current temperature

Pipe shapeEnter whether the pipe is circular or rectangular

Pipe inner diameter

If the pipe is circular, enter the internal diameter

Duct internal height

If the pipe is rectangular, enter the height

Duct internal width

If the pipe is rectangular, enter the width

Installation factor

Factor for correcting the mass flow

Current output

Select the current output assignment

4mA value

Define a measured variable for the 4mA value

20mA value

Define a measured variable for the 20mA value

Operating mode

Select the pulse/frequency/switch output

9.5

Defining the tag nameTo enable fast identification of the measuring point within the system, you can enter a uniquedesignation using the Device tag parameter and thus change the factory setting.
Navigation path
Setup ® Advanced setup ® Device tag

Endress+Hauser

35Commissioning

Proline t-mass B 150

User entry
Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /)In the factory configuration, the tag name appears at the top left in the header.

9.6

Protecting settings from unauthorized access

The following options exist for protecting the configuration of the measuring device fromunintentional modification after commissioning:• Write protection via access code (® ä 36)• Write protection via lock switch (® ä 36)• Write protection via keypad lock (® ä 29)
9.6.1

Write protection via access code

Using the customer-specific access code, the parameters for the measuring device configuration
are write-protected and their values can no longer be changed via local operation.
Define access code1.

Navigating to the "Define access code" parameter: Setup ® Advanced setup ® Def. access
code

2.

Define a max. 4-digit numeric code as an access code.
Ã The -symbol appears in front of all write-protected parameters.

The device automatically locks the write-protected parameters again if a key is not pressed for
10 minutes in the navigation and editing view. The device locks the write-protected parameters
automatically after 60 s if the user skips back to the measured value display mode from thenavigation and editing view.
If write access is activated via access code, it can be also be deactivated only via the accesscode (® ä 31).
9.6.2

Write protection via lock switch

Unlike write protection via user-specific access code, this allows write access to the entire
operating menu - other than the Contrast display parameter - to be locked.
The values of the parameters are still visible, but can no longer be changed (except for Contrast
display), either via the local display, the service interface (CDI) or HART protocol.

36

Endress+Hauser

Proline t-mass B 150

Diagnostic information and remedial measures

1

1

2

2

ON

–

+

OFF

E

A0017255

1.

Unscrew the electronics compartment cover.

2.

Pull out the display module with a gentle rotational movement.

3.

To make it easier to access the lock switch, attach the display module to the edge of theelectronics compartment.

4.

Setting the lock switch (WP) on the electronics module to the ON position enables the
hardware write protection.
Ã The symbol appears in the header of the measured value display and in thenavigation view in front of the parameters.

5.

Feed the ribbon cable into the gap between the housing and electronics module and plug
the display module into the electronics compartment in the desired direction until it
engages.

6.

Screw down the electronics compartment cover

10

Diagnostic information and remedial measures

Faults detected by the self-monitoring system of the measuring device are displayed as adiagnostic message in alternation with the operational display. The message on the remedial
measures can be called up from the diagnostic message, and contains important information on
the fault.

Endress+Hauser

37

Diagnostic information and remedial measures

XXXXXXXXX

Proline t-mass B 150S

XXXXXXXXX

20.50

S

S441
Curr. output 1
i

mA

1
2
5

(ID:153)
Curr. output 1
S441 2d11h05m55s
1. Check process
2. Check current output settings

3
4

A0013940-EN

å3
1
2
3
4
5

Message for remedial measures
Short text
Diagnostic behavior with diagnostic code
Service ID
Operation time of occurrence
Remedial measures

The user is in the diagnostic message.
1.

Press O ( symbol).
Ã The message for the remedial measures for the diagnostic event opens.

2.

Press S + O simultaneously.
Ã The message about the remedial measures closes.

38

Endress+Hauser

KA01104D/06/EN/01.12
71165082