BA00470C/07/EN/15.13
71214527

Products

Solutions

Valid as of:Softwareversion 01.04.00

Operating Instructions
Liquiport 2010 CSP44
Automatic sampler for liquid media
Maintenance & diagnostics

ServicesAbout this manualThis manual describes all the tasks you must perform for maintenance, diagnostics and repair.
A description of the following is provided here:• General troubleshooting• Overview of diagnostic messages• Description of the information in the "Diagnostics" menu
– Diagnostics list
– Logbooks– System information– Output status– Systemtest/Reset– Sensor change– Manual hold– Term information– Simulation– Sensor information• Cleaning and maintenance• Accessories and spare parts
This manual does not include the following:• Setup/General settings--> Operating Instructions BA00465C "Commissioning"• Display/Operation--> Operating Instructions BA00465C "Commissioning"• Inputs--> Operating Instructions BA00492C "Operation & settings"• Outputs--> Operating Instructions BA00492C "Operation & settings"• Sampling programs--> Operating Instructions BA00492C "Operation & settings"• Additional functions--> Operating Instructions BA00492C "Operation & settings"• Data management--> Operating Instructions BA00492C "Operation & settings"• Calibration--> Operating Instructions BA00493C "Calibration"• Expert--> Internal Service ManualLiquiport 2010 CSP44

Table of contents
1

Diagnostics andtroubleshooting . . . . . . . . . . . . . 4

1.1 General troubleshooting . . . . . . . . . . . . . . 4
1.2 Diagnostic information on local display 11
1.3 Adjusting diagnostic information . . . . . 11
1.4 Overview of diagnostic information . . . 14
1.5 Pending diagnostic messages . . . . . . . . . 33
1.6 Diagnostic list . . . . . . . . . . . . . . . . . . . . . . 33
1.7 Logbooks . . . . . . . . . . . . . . . . . . . . . . . . . . 34
1.8 Device information . . . . . . . . . . . . . . . . . 41
1.9 Resetting the measuring device . . . . . . . 42
1.10 Runtime info . . . . . . . . . . . . . . . . . . . . . . 44
1.11 Status of inputs/outputs . . . . . . . . . . . . . 44
1.12 Sensor replacement (for version withsensors with the Memosens protocol) . 45
1.13 Manual hold (for version with sensors with
the Memosens protocol) . . . . . . . . . . . . . 45
1.14 Firmware history . . . . . . . . . . . . . . . . . . . 46

2

Maintenance . . . . . . . . . . . . . . . 47

2.1
2.2
2.3
2.4
2.5
2.6

Recommended maintenance . . . . . . . . . 47
Replacing the pump tube . . . . . . . . . . . . 48
Cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Replacing the storage batteries . . . . . . . 53
Calibration . . . . . . . . . . . . . . . . . . . . . . . . 54
Simulation . . . . . . . . . . . . . . . . . . . . . . . . 54

3

Repair. . . . . . . . . . . . . . . . . . . . . 56

3.1
3.2
3.3

Spare parts . . . . . . . . . . . . . . . . . . . . . . . . 56
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . 57

4

Accessories . . . . . . . . . . . . . . . . 58

4.1
4.2
4.3

Accessories for Liquiport 2010 CSP44 . 58
Measuring cable . . . . . . . . . . . . . . . . . . . . 58
Sensors . . . . . . . . . . . . . . . . . . . . . . . . . . . 59

Index. . . . . . . . . . . . . . . . . . . . . . 63

Endress+Hauser

Diagnostics and troubleshooting

1

Liquiport 2010 CSP44

Diagnostics and troubleshooting

The sampler continuously monitors its own functions.
The color of the display background changes to red if a diagnostics message for error category"F" occurs.
The LED beside the display flashes red if a diagnostics message for error category "M" occurs.

1.1

General troubleshooting

1.1.1 Troubleshooting
A diagnostic message appears on the display, measured values are not plausible or you
encounter a problem.
1. See the Diagnostics menu for details on the diagnostic message.
 Follow the instructions to rectify the problem.
2. If this does not help: Search for the diagnostic message under "Overview of diagnostic
information" () in this manual. Use the message number as a search criterion. Ignore theletters indicating the Namur error category.
 Follow the troubleshooting instructions provided in the last column of the error tables.
3. If the measured values are implausible, the onsite display is faulty or you encounter otherproblems, search under "Process errors without messages" ( ä 4) or "Device-specific errors"( ä 10).
 Follow the recommended measures.
4. Contact the Service Department if you cannot rectify the error yourself. Only cite the error
number.
1.1.2 Process errors without messages
pH/ORP measurement
ProblemPossible cause

Tests and/or remedial measures

Display values deviate
from reference
measurement

Incorrect calibration

Repeat the calibration.
Where necessary, check and repeat the calibration
with the reference device.

Sensor fouled

Clean the sensor.

Temperature measurement

Check the temperature measured values of bothdevices.

Temperature compensation

Check the settings for temperature compensation
and adjustment for both devices.

4

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

ProblemPossible cause

Tests and/or remedial measures

Measuring chain
zero-point cannot be
adjustedContaminated reference system

Test with new sensor

Junction clogged

Clean or grind junction

Asymmetric sensor voltage too high Clean junction or test with another sensor
No change or subtle
change in display– Sensor fouled– Sensor old– Sensor defective (reference lead)

Clean the sensor.

Reference has low level of KCl

Check KCl supply: 0.8 bar (12 psi) over medium
pressure

Measuring chain slope:– Cannot be adjusted– To low– No slope

Device input defective

Check device directly.

– Sensor old– Hair-line crack in the glass
membrane

Renew sensor.

Constant, incorrect
measured value

Sensor does not immerse properlyor protection cap not removed

Check installation position, remove protection cap.

Incorrect temperature
value

Fluctuations in measuredvalue

No current output signal

Air pockets in assembly

Check assembly and orientation.

Ground fault at or in device

Perform test measurement in isolated vessel, withbuffer solution if applicable.

Hair-line crack in the glass
membrane

Renew sensor.

Device in impermissible operating
condition (does not respond when
key pressed)

Switch off device and switch it on again.

Sensor failure

Replace sensor

Interference on signal output cable

Check cable routing, route cable separately ifnecessary.

Interference potential in medium

Eliminate source of interference or ground medium
as close as possible to sensor.

Cable disconnected or shortcircuited Disconnect cable and measure directly at device.
Output defective

See "Device-specific errors" section.

Fixed current output
signalCurrent simulation active

Switch off simulation.

Incorrect current output
signalTotal load in current loop too highMeasure the load and reduce it to the permitted
value if necessary (--> Technical data, BA"Commissioning").

EMC (interference coupling)

Check wiring. Identify and eliminate cause ofinterference.

Endress+Hauser

5Diagnostics and troubleshooting

Liquiport 2010 CSP44

Conductivity measurement
ProblemPossible cause

Tests and/or remedial measures

Display values deviate
from reference
measurement

Incorrect calibration

Repeat the calibration.
Where necessary, check and repeat the calibration
with the reference device.

Sensor fouled

Clean the sensor.

Temperature measurement

Check the temperature measured values of bothdevices.

Temperature compensation

Check the settings for temperature compensation
and adjustment for both devices.

Polarization fields

Use suitable sensor:• Larger cell constant• Graphite instead of stainless steel (observe
material resistance properties)

Short-circuit/moisture in sensor

Check sensor.

Short-circuit in cable or socket

Check cable and socket.

Disconnection in sensor

Check sensor.

Disconnection in cable or socket

Check cable and socket.

Incorrect cell constant setting

Check cell constant.

Incorrect output assignment

Check assignment of measured value to current
signal.

Air pockets in assembly

Check assembly and orientation.

Ground fault at or in device

Measure in isolated vessel.

Device in impermissible operating
condition (does not respond when
key pressed)

Switch off device and switch it on again.

Sensor failure

Replace sensor

Display values deviate
from reference
measurement

Implausible measured
values:– Measured value
constantly 000– Measured value too low– Measured value too
high– Measured value frozen– Current output value
not as expected

Incorrect temperature
value

6

Endress+Hauser

Liquiport 2010 CSP44

ProblemDiagnostics and troubleshooting

Possible cause

Tests and/or remedial measures

No/incorrect temperaturecompensation

ATC: select type of compensation; if linear, set
suitable coefficients.
MTC: set process temperature.

Incorrect temperature measurement Check temperature measured value.
Bubbles in medium

Suppress formation of bubbles by:– Gas bubble trap
– Creating counterpressure (orifice plate)– Measurement in bypass

Flow rate too high (can lead tobubble formation)

Reduce flow rate or select less turbulent mounting
location.

Voltage potential in medium (onlyfor conductive)

Ground medium close to sensor.

Sensor fouling or buildup on sensor

Clean sensor (see "Cleaning the conductivity sensors"section).

Interference on signal output cable

Check cable routing, route cable separately ifnecessary.

Interference potential in medium

Eliminate source of interference or ground medium
as close as possible to sensor.

Interference on measuring cable

Connect cable shield as per wiring diagram.

Measured value in processincorrect

Fluctuations in measuredvalue

No current output signal

Cable disconnected or shortcircuited Disconnect cable and measure directly at device.
Output defective

See "Device-specific errors" section.

Fixed current output
signalCurrent simulation active

Switch off simulation.

Incorrect current output
signalTotal load in current loop too highMeasure the load and reduce it to the permitted
value if necessary (--> Technical data, BA"Commissioning").

EMC (interference coupling)

Check wiring. Identify and eliminate cause ofinterference.

Oxygen measurement
Problem

Display value - - - -Endress+Hauser

Possible cause

Tests and/or remedial measures

Sensor failure

Test with new sensor

Sensor cable disconnected

Check cable or cable extension.

Incorrect sensor connection

Check the connection at input module (--> BA"Commissioning", "Wiring" section).

Electronics module defective

Replace the module.

7

Diagnostics and troubleshooting

Liquiport 2010 CSP44

ProblemPossible cause

Tests and/or remedial measures

No change or subtle
change in display– Sensor fouled– Sensor old (membrane)• Clean the sensor.
• If necessary:– Change electrolyte, change membrane cap(amperometric sensor)– Change fluorescence cap (optical sensor)

Constant, incorrect
measured value

Device in impermissible operating
condition (does not respond when
key pressed)

Switch off device and switch it on again.

Membrane fouled

Clean or replace cap

Electrolyte exhausted orcontaminatedChange electrolyte

Measured value too low

Measured value too highAnode coating worn

Repolarize sensor

Black anode coating

Regenerate sensor in factory

Air pocket under membrane

Clean sensor, optimize installation where necessary

Polarization not complete

Wait until polarization time elapses (--> Technicaldata in sensor operating manual)

Incorrect temperature measurement Check/correct value.
Implausible measured
value

Incorrect altitude setting
Incorrect air pressure

Incorrect temperature
value

Fluctuations in measuredvalue

No current output signal

Incorrect sensor connection

Incorrect calibration
Reset and repeat calibration.
Check the connection at input module (--> BA"Commissioning", "Wiring" section).

Temperature sensor defective

Replace sensor

Interference on signal output cable

Check cable routing, route cable separately ifnecessary.

Interference potential in medium

Eliminate source of interference or ground medium
as close as possible to sensor.

Cable disconnected or shortcircuited Disconnect cable and measure directly at device.
Output defective

See "Device-specific errors" section.

Fixed current output
signalCurrent simulation active

Switch off simulation.

Incorrect current output
signalTotal load in current loop too highMeasure the load and reduce it to the permitted
value if necessary.

EMC (interference coupling)

Disconnect both output cables and measure directly
at device.

8

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

Turbidity and nitrate measurement
ProblemPossible cause

Tests and/or remedial measures

Sensor failure

Test with new sensor

Sensor cable disconnected

Check cable or cable extension.

Incorrect sensor connection

Check the connection at input module (--> BA"Commissioning", "Wiring" section).

Electronics module defective

Replace the module.

No change or subtle
change in display

Sensor fouled

Clean the sensor.

Constant, incorrect
measured value

Device in impermissible operating
condition (does not respond when
key pressed)

Switch off device and switch it on again.

Sensor not calibrated or incorrectly
calibrated

Calibration with original sample might be necessary
for concentration or solids content.

Sensor fouled

Clean sensor

Sensor installed in "dead zone" or air
pocket in assembly or flange

Check installation position, move sensor to area thatreceives good flow. Pay attention when mounting inhorizontal pipes

Incorrect sensor orientation

Align sensor:• Normal media:
Direct flow to measuring window• For high solids content:
Align measuring window at angle of 90˚ to flow

Incorrect sensor connection

Check the connection at input module (--> BA"Commissioning", "Wiring" section).

Temperature sensor defective

Replace sensor

Interference on signal output cable

Check cable routing, route cable separately ifnecessary.

Irregular flow / turbulence / airbubbles / large solid particles

Select a better mounting location or reduceturbulence, use a large measured value damping
factor if necessary
Set gas bubble threshold to 100 %Display value - - - -Implausible measured
value

Incorrect temperature
value

Fluctuations in measuredvalue

No current output signal

Cable disconnected or shortcircuited Disconnect cable and measure directly at device.
Output defective

See "Device-specific errors" section.

Fixed current output
signalCurrent simulation active

Switch off simulation.

Incorrect current output
signalTotal load in current loop too highMeasure the load and reduce it to the permitted
value if necessary.

EMC (interference coupling)

Disconnect both output cables and measure directly
at device.

Air bubbles

Do not mount sensor above aeration discs

Value switches to zero andback to measured value

Endress+Hauser

9Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.1.3 Device-specific errors
ProblemPossible cause

Tests and/or remedial measures

The device cannot beswitched on/display remains
dark

No supply voltage

Check if voltage supplied

Basic module defective

Replace basic module

Values appear on display
but:– Display does not change
and / or– Device cannot be
operated

Module not wired correctly

Check modules and wiring

Impermissible operating system
condition

Switch off device and switch it on again

Control signals are notaccepted or outputs do notswitchIncorrect program setting

Check program setting

Incorrect wiring

Check wiring

Electronics failure

Replace basic module

Siphon in sampling hose

Check sampling hose

Connection not air-tight/sampling
hose taking in air– Check tubes/connections– Check the sampling hose is routed correctly

Bottles not being filled correctly– Incorrect distribution selected in the controls– Calibrate the distribution arm

Distribution arm does not move

– Incorrect distribution selected in the controls– Check distribution arm connection– Distributor defective, replace distributor or– Have repaired by E+H Service

Incorrect bottle filled– Incorrect distribution selected in the controls

Wrong pump tube

Only use genuine pump tube

Faulty sensory mechanism

Replace sensory mechanism (contact E+H
Service)

Sample not representative

No sampling

Connection leaking

Check tubes/connections for leaks

Sampling hose taking in air

Check the sampling hose is routed correctlyWrong pump tube

Only use genuine pump tube

Faulty sensory mechanism

Replace sensory mechanism (contact E+H
Service)

Implausible measured values Inputs defective(only for version withsensors with the Memosens
protocol)First perform tests and take measures as
outlined in "Process-specific errors" section

Current output, incorrect
current value

Check with integrated current simulation,connect mA meter directly to current output.

Incorrect adjustment
Load too large

Measuring input test:– Connect a resistor instead of the
conductivity sensor

Shunt / short to ground in current loop

10

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

ProblemPossible cause

Tests and/or remedial measures

No current output signal

Basic module defective

Check with integrated current simulation,connect mA meter directly to current output.

1.2

Diagnostic information on local display

Up-to-date diagnostic events are displayed along with their status category, diagnostic code andshort text. By clicking the navigator, you can call up additional information and tips on remedial
measures.

1.3

Adjusting diagnostic information

1.3.1 Classification of diagnostics messages
More detailed information on the current diagnostics messages displayed is provided in the
DIAG/Diagnostics list menu.
In accordance with Namur specification NE 107, the diagnostics messages are characterized by:• Message number• Error category (letter in front of the message number)– F = Failure. A malfunction has been detected.
The cause of the malfunction is to be found in the sampling point/measuring point. Any
controller connected should be set to manual mode.
– M = Maintenance required. Action should be taken as soon as possible.
The device still measures/takes a sample correctly. Immediate measures must not be taken.
However, proper maintenance efforts would prevent a possible malfunction in the future.
– C = Function check, (no error)
Maintenance work is being performed on the device. Wait until the work has been
completed.
– S = Out of specification. The measuring point is being operated outside specifications.
Operation is still possible. However, you run the risk of increased wear, shorter operating
life or lower accuracy levels. The cause of the problem is to be found outside the measuring
point.
• Message text
If you contact the Service Department, please cite the message number only. Since you can
individually change the assignment of an error to an error category, the Service Department
cannot use this information.

Endress+Hauser

11Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.3.2 Adjusting the device behavior
All the diagnostics messages are assigned to specific error categories at the factory. Since othersettings might be preferred depending on the application, error categories and the effect errors
have on the measuring point can be configured individually. Furthermore, every diagnostics
message can be disabled.
ExampleThe controller returns diagnostics message 531 "Logbook full". You want to change this message
so that an error is not indicated on the display for example.
1. Go to:– Menu/Setup/General settings/Extended setup/Diagnostics/Device behavior
for device-specific diagnostics messages (as in this example)– Menu/Setup/Inputs/<Sensor type>/Extended setup/Diagnostics settings/Diag.
behavior
for sensor-specific diagnostics messages.
2. Select the diagnostics message and press the navigator button.
3. Decide:a. Should the message be deactivated?
b. Do you want to change the error category?
c. Should an error current be output?
d. Do you want to trigger a cleaning program?
4.

Deactivate the message, for example (Diagnostics message to "Off").

Configuration options
The list of diagnostic messages displayed depends on the path selected. There are
device-specific messages, and messages that depend on what sensor is connected.
Path:... /Extended setup/Diagnostics settings/Diag. behavior (optional)

Function

Options

List of diagnostic messages

Select the message to be changed. Only then can you
make the settings for this message.

Diag. code

Read only

Diagnostic message

Options• On• Off
Factory setting
Depends on the message

12

Info

You can deactivate or reactivate a diagnostics message
here.
Deactivating means:• No error message in the measuring mode
• No error current at the current output

Endress+Hauser

Liquiport 2010 CSP44

Path:

Diagnostics and troubleshooting... /Extended setup/Diagnostics settings/Diag. behavior (optional)

FunctionError current

Options

Info

Options• On• Off

Decide whether an error current should be output at thecurrent output if the diagnostic message display isactivated.

Factory setting
Depends on the message

Status signal

Options• Maintenance (M)• Out of specification (S)• Function check (C)• Failure (F)Factory setting
Depends on the message

Diag. output

Options• None• Binary output
Factory setting
None

Cleaning
program(optional)

Options• None• Cleaning 1• Cleaning 2• Cleaning 3• Cleaning 4

If general device errors occur, the error current is
output at all the current outputs.
In the case of channel-specific errors, the error
current is only output at the particular current
output.
The messages are divided into different error categories
in accordance with NAMUR NE 107.
--> BA00470C "Maintenance & diagnostics"
Decide whether you want to change a status signal
assignment for your application.

You can use this function to select an output to which
the diagnostic message should be assigned.
For sensors with the Memosens protocol:
You first have to configure a relay output for"Diagnostics" (Menu/Setup/Outputs, assign "Diagnostics"
function and set Operating mode to "as assigned") before
being able to assign the message to an output.
--> BA00492C "Operation & settings"
Decide whether the diagnostic message should trigger acleaning program.
You can define the cleaning programs under:
Menu/Setup/Additional functions/Cleaning.

Factory setting
None
Detail information

Endress+Hauser

Read only

Here you can find more information on the diagnostic
message and instructions on how to resolve theproblem.

13

Diagnostics and troubleshooting

1.4

Liquiport 2010 CSP44

Overview of diagnostic information

1.4.1 Device-specific, general diagnostics messages
No.

Message

Factory settings

Tests or remedial measures

Cat.

Diag.
on/offError
current

202

Selftest active

F

On

Off

Wait for self-test to be finished

216

Hold active

C

On

Off

Output values and status of the channel are on
hold

241

Device error

F

On

On

242

Software incomp.

F

On

On

243

Device error

F

On

On

Internal device error
1. Update the software
2. Contact the Service Department
3. Replace the backplane (Service)261

Electr. module

F

On

On

Electronics module defective
1. Replace the module
2. Contact the Service Department

262

Module connect.

F

On

On

Electronics module not communicating
1. Check cable connection, replace if necessary
2. Check the power supply of the sampling
control module
3. Contact the Service Department

263

Electr. module

F

On

On

Wrong kind of electronics module
1. Replace the module
2. Contact the Service Department284

Firmware update

M

On

Off

Update completed successfully

285

Update error

F

On

On

Firmware update failed
1. Repeat update
2. SD card error --> use another card
3. Incorrect firmware --> repeat with suitable
firmware4. Contact the Service Department

302

Battery low

M

On

Off

Buffer battery of real time clock is low
The date and time are lost if the power isinterrupted.
--> Contact the Service Department (battery
replacement)

304

Module data

F

On

On

At least 1 module has incorrect configurationdata
1. Check the system information
2. Contact the Service Department

305

Power consum.

F

On

On

Total power consumption too high
1. Check installation
2. Remove sensors/modules

306

Software error

F

On

On

Internal firmware error--> Contact the Service Department

14

Endress+Hauser

Liquiport 2010 CSP44

No.

311

MessageTemp. sensor

Diagnostics and troubleshooting

Factory settings

Tests or remedial measures

Cat.

Diag.
on/offError
current

F

On

On

Temperature sensor PT2 in sample compartment
is defective.
• Sample temperature measurement notpossible• A sampling program will not be canceled-> Replace sensor

314

No sample flow

F

On

On

Negative pressure cannot be created in theperistaltic pump.
-> Check pump tube for leaks-> Immerse suction line in medium

322

Read sub-program

F

On

On

Selected subprogram cannot be read from theprogram memory--> Create new subprogram

323

Write sub-prg.

F

On

On

Subprogram created cannot be saved-> Hardware error-> Contact the Service Department

324

Delete sub-prg.

F

On

On

Selected subprogram cannot be deleted from theprogram memory-> Reset the software

325

Readsub-prg.list

F

On

On

Subprogram list cannot be read from the program
memory-> Reset the software

328

Distribution arm

F

On

On

Distribution arm zero point not found during testrun-> Perform distribution arm test underDiagnostics/System test/Distribution arm-> Contact the Service Department

331

Peristaltic pump

F

On

On

• Peristaltic pump defective• Motor cable broken-> Contact the Service Department

332

Peristaltic pump

F

On

On

Control of peristaltic pump defective--> Contact the Service Department

333

Pressure sensor

F

On

On

Medium detection not possible
No sampling possible.
• Suction line not drained before sampling• Pressure sensor defective-> Check suction line, where necessary purge itusing the pump test under Diagnostics/Systemtest/Pump test-> Contact the Service Department

Endress+Hauser

15

Diagnostics and troubleshootingNo.

Message

Liquiport 2010 CSP44

Factory settings

Tests or remedial measures

Cat.

Diag.
on/offError
current

337

Pump tubing

M

On

Off

Pump tube operating time reached shortly
Display under Diagnostics/Runtime info/Tubingage-> Schedule replacement-> After replacement, reset the operating time
under Diagnostics/Runtime info

338

Pump tubing

M

On

Off

Pump tube operating time reached
Display under Diagnostics/Runtime info/Tubingage-> Change the pump tubing->After replacement, reset the operating time
under Diagnostics/Runtime info

343

Power supply

M

On

Off

Power supply failure

344

Program pause

C

On

Off

Sampling program paused

345

Time changeover

M

On

Off

Daylight saving time/winter time setting
Normal time (winter time) active.

346

Time changeover

M

On

Off

Daylight saving time/winter time setting
Daylight saving time active

347

Sample confirm.

F

On

On

Sampling command has not been processed-> Check internal cable to 1IF-> Reset the software

348

Read program

F

On

On

Selected program cannot be read from theprogram memory--> Create new program

349

Write program

F

On

On

Program created cannot be saved
Hardware error has occurred--> Contact the Service Department

351

Delete prog.

F

On

On

Selected program cannot be deleted from theprogram memory--> Reset the software

352

Read Prog.list

F

On

On

Program list cannot be read from the program
memory--> Perform device reset underDiagnostics/System test/Reset/Device reset

353

Overfill check

F

On

Off

Total capacity of bottle reached• No further sampling to current bottle istriggered-> Adjust the sampling program under Program
selection if desired

354

Bottle check

F

On

Off

No empty bottles available for current program• No further sampling-> Check the program settings under Program
selection

16

Endress+Hauser

Liquiport 2010 CSP44

No.

Message

Diagnostics and troubleshooting

Factory settings

Tests or remedial measures

Cat.

Diag.
on/offError
current

355

Start time over

M

On

Off

Start time entered is in the past
• Enter a new start time356

Overfill check

F

On

Off

The total sample volume does not fit in thesample bottle-> Change the sample volume

357

Sampling faulted

M

On

Off

• Sample discarded• There are too many sampling requests pending-> Adjust the sampling program under Program
selection

358

Configuration

F

On

On

Program configuration does not match thecurrent device configuration-> Adjust the configuration

359

Emptying error

F

On

On

• Error when draining• Draining and sampling program cancelled-> Check connection to FMSY1 module-> Check 4R module, replace if necessary-> Restart the software under
Setup/Diagnostics/System test/Restart

366

Module connect.

F

On

On

No communication with actuator module-> Check internal connecting cable to module 1IF

370

Intern. Voltage

F

On

On

Internal voltage outside the valid range--> Check supply voltage

373

Electr. temp.

M

On

Off

High electronics temperature--> Check ambient temperature and energy
consumption

374

Sensor check

F

On

Off

No measurement signal from sensor-> Check sensor connection-> Check sensor, replace if necessary375

No 4R module

F

On

On

No connection to 4R module-> Check 4R module, replace if necessary-> Restart the software under
Setup/Diagnostics/System test/Restart

401

Reset to default

F

On

On

Factory reset is performed

406

Param. active

C

Off

Off

--> Wait for configuration to be finished

407

Diag. active

C

Off

Off

--> Wait for maintenance to be finished

412

Writing backup

F

On

Off

--> Wait for the write process to be finished

413

Reading backup

F

On

Off

--> Wait

Endress+Hauser

17

Diagnostics and troubleshootingNo.

Message

Liquiport 2010 CSP44

Factory settings

Tests or remedial measures

Cat.

Diag.
on/offError
current

460

Curr. under-run

S

On

Off

461

Current exceeded

S

On

Off

Reasons• Sensor in air• Air pockets in assembly• Sensor fouled• Incorrect flow to sensor
Measures
1. Check sensor installation
2. Clean sensor
3. Adjust assignment of current outputs

462

Output Deviation

S

On

Off

When the current output is read back, the value
deviates from the target value displayed.
Possible reasons: Current load outside
specification, short-circuit or open current loop,
module defective
1. Check installation of current loop
2. Check module
3. Contact the Service Department502

No text catalog

F

On

On

--> Contact the Service Department

503

Language change

M

On

Off

Language change failed--> Contact the Service Department

530

Logbook at 80%

M

On

Off

1.

531

Logbook full

M

On

Off

2.
3.

532

License error

M

On

Off

--> Contact the Service Department

540

Parameter save

M

On

Off

Configuration saving has failed, --> repeat

Save the logbook to the SD card and thendelete the logbook in the device
Set memory to ring memory
Deactivate logbook

541

Parameter load

M

On

Off

Configuration successfully loaded

542

Parameter load

M

On

Off

Configuration loading has failed, --> repeat543

Parameter load

M

On

Off

Configuration loading aborted

544

Parameter reset

M

On

Off

Factory default successful

729

Filter candle

M

On

Off

• Filter cartridge change necessary• The limit value for operating hours has been
exceeded-> Replace the filter cartridge for the sample
preparation system and reset the operating hours
counter in the Diagnostics menu

18

Endress+Hauser

Liquiport 2010 CSP44

No.

730

Message

Clean. solution

Diagnostics and troubleshooting

Factory settings

Tests or remedial measures

Cat.

Diag.
on/offError
current

M

On

Off

• Warning for level of cleaning solution forsample preparation system• Depending on the cleaning duration, thecleaning interval and external events, theremaining quantity is sufficient for a few hours
or days-> Top up cleaning solution for sample
preparation system-> Check level switch for cleaning solution

910

Limit switch

S

On

Off

Limit switch activated921

Pump bracket

F

On

On

The pump bracket is detected as open.
• Pump bracket open
• Reed contact defective-> Close pump bracket-> Contact the Service Department

969

Modbus Watchdog

S

Off

Off

The device did not receive a Modbus telegram
from the master within the specified time. The
status of Modbus process values received is set toinvalid.

970

Input Overload

S

On

On

Current input overloaded
The current input is switched off from 23 mA due
to overload and reactivated automatically when anormal load is present.

971

Input low

S

On

On

Current input too low
At 4 to 20 mA, the input current is less than thelower error current.
--> Check the input for short-circuits.

972

Input > 20 mA

S

On

On

Current output range exceeded

973

Input < 4 mA

S

On

On

Current output range undershot

974

Diag. confirmed

C

Off

Off

User has acknowledged the message displayed in
the measuring menu.

975

Device reset

C

Off

Off

Device Reset991

CO2 conc. range

F

On

On

CO2 concentration (degassed conductivity)
outside the measuring range992

pH calc range

F

On

On

pH calculation outside the measuring range993

rH calc range

F

On

On

rH calculation outside the measuring range

994

Dual cond range

F

On

On

Dual conductivity outside the measuring range

Endress+Hauser

19Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.4.2 Sensor-specific diagnostics messages
Abbreviations used for sensor types• P ... pH/ORP (general)– P (glass) ... glass electrode– P (ISFET) ... ISFET sensor• C ... Conductivity (general)– C (cond.) ... Conductive sensor– C (ind.) ... Inductive sensor• O ... Oxygen (general)– O (opt.) ... Optical sensor– O (amp.) ... Amperometric sensor• N ... Nitrate• T ... Turbidity and solids• S ... SAC• I ... ISE• Cl ... Chlorine
No.

002

Message

Sensor unknown

Factory settings
Cat.

Diag. Error
curren
t

F

On

On

Sensor type Tests or remedial measures

All

Replace sensor

004

Sensor problem

F

On

On

All

005

Sensor data

F

On

On

All

Invalid sensor data
1. Check the firmware compatibility for thesensor and transmitter, load suitable
firmware if necessary
2. Reset sensor to factory setting,disconnect sensor and connect it again
3. Update transmitter date
4. Replace sensor

010

Sensor scanning

F

Off

On

All

Wait for initialization to be finished012

Writing data

F

On

On

All

Could not write sensor data
1. Repeat write process
2. Replace sensor

013

Sensor type

F

On

On

All

Replace sensor, making sure correct sensor
type is used018

Sensor not ready

F

On

On

All

Sensor communication blocked
1. Sensor fails tag check. Replace.
2. Internal software error. Contact Service
Department022

Temp. sensor

F

On

On

P, C, O, I, Cl

Temperature sensor defective
Replace sensor

061

Sensor electr.

F

On

On

All

Sensor electronics defective
Replace sensor

20

Endress+Hauser

Liquiport 2010 CSP44

No.

Message

Diagnostics and troubleshooting

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

062

Sensor connect.

F

On

On

All

1.
2.

081

Initialization

F

On

On

All

Wait for initialization to be finished100

Sensor comm.

F

On

On

All

Sensor not communicating
1. Check sensor connection
2. Check sensor plug
3. Contact the Service Department

101

Sensor incompat.

F

On

On

All

1.
2.
3.

102

Calib. Timer

M

On

Off

All

Calibration interval elapsed. Measurement
can still take place.
Calibrate sensor

103

Calib. timer

M

On

Off

All

Calibration interval will elapse soon.
Measurement can still take place.
Calibrate sensor

104

Calib. validity

M

On

Off

All

Validity of last calibration expired.
Measurement can still take place.
Calibrate sensor

105

Calib. validity

M

On

Off

All

Validity of last calibration will expire soon.
Measurement can still take place.
Calibrate sensor

106

Sensor TAG

F

On

On

All

Sensor has invalid tag or tag group
1. Replace sensor
2. Use new sensor with identical design
3. Deactivate tag check

107

Calib. active

C

On

Off

P, C, O, I, Cl

Wait for calibration to be finished

108

Sterilization

M

On

Off

P, C, OSpecified number of sterilizations will soon bereached. Measurement can still take place.
Replace sensor

109

Sterilizat. cap

M

On

Off

O (amp.)Specified number of sterilizations for the cap
is reached. Measurement can still take place.
Replace membrane cap

110

Channel init.

F

On

On

All

Initialization of channel failed, operation is
not possible--> Contact the Service Department

114

Temp.offset high

M

On

Off

115

Temp. offset low

M

On

Off

All except U Calibration alarm: Limit values fortemperature offset exceeded
All except U 1. Check temperature sensor
2. Replace sensor

Endress+Hauser

Check sensor connection
Contact the Service Department

Update sensor firmware
Replace sensor
Contact the Service Department

21

Diagnostics and troubleshootingNo.

Message

Liquiport 2010 CSP44

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

116

Temp. slope high

M

On

Off

117

Temp. slope low

M

On

Off

All except U Calibration alarm: Limit values fortemperature slope exceeded
All except U Sensor old or defective
1. Repeat calibration
2. Replace sensor

118

Sensor glass

F

On

On

P (glass)

119

Sensor check

M

On

Off

P (glass)

120

Sensor reference

F

On

On

P (glass)

121

Sensor reference

M

On

Off

P (glass)

122

Sensor glass

F

On

On

P (glass)

123

Sensor glass

M

On

On

P (glass)

124

Sensor glass

M

On

Off

P (glass)

125

Sensor glass

F

On

On

P (glass)

126

Sensor check

M

On

Off

P (glass)

Sensor condition check (SCC), poor sensor
condition
Glass membrane fouled or dry, junction
blocked
1. Clean sensor, regenerate
2. Replace sensor

127

Sensor check

M

On

Off

P (glass)

Sensor condition check (SCC), adequate
sensor condition

128

Sensor leakage

F

On

On

P (ISFET),O (amp.)

ISFET leak current alarm
Defective due to gate abrasion or damage
Replace sensor

129

Sensor leakage

F

On

Off

P (ISFET),O (amp.)

ISFET leak current warning
Measuring can continue until the alarm
occurs130

Sensor supply

F

On

On

P, O, I, Cl

Poor sensor power supply
1. Check sensor connection
2. Replace sensor

22

Glass breakage warning, impedance of pH
glass too low
Measuring can continue until the alarm (118)
occurs.
1. Inspect sensor for hair-line cracks andbreakage
2. Check medium temperature
3. Replace sensor
Reference warning, impedance of reference
too low
Measuring can continue until the alarm (120)
occurs1. Check reference forclogging/contamination
2. Clean reference/junction
3. Replace sensor
Impedance limit values exceeded/undershot
Measuring can continue until the alarm (122,124) occurs.
1. Inspect sensor for hair-line cracks andbreakage
2. Check limit values and change where
necessary
3. Replace sensor

Endress+Hauser

Liquiport 2010 CSP44

No.

Message

Diagnostics and troubleshooting

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

131

Sensor calib.

M

On

Off

O (opt.)132

Sensor calib.

M

On

Off

O (opt.)133

Sensor signal

F

On

On

O (opt.)No signal (fluorescence decay)
1. Replace sensor cap
2. Contact the Service Department

134

Sensor signal

M

On

Off

O (opt.)Low signal amplitude. Measurement can stilltake place.
1. Replace sensor cap
2. Contact the Service Department

135

Sensor temp.

S

On

On

O

136

Sensor temp.

S

On

On

O

Temperature outside specification
1. Check process
2. Check installation

137

Sensor LED

F

On

On

O (opt.)Sensor LED: no voltage
Contact the Service Department

138

Sensor LED

F

On

On

O (opt.)Sensor LED: no power
Contact the Service Department

140

Sensor check

F

On

On

O

Sensor range errors
Contact the Service Department

141

Polarization

F

On

On

C (cond.)Polarization warning
The measured value is distorted at highconductivity levels.
Use a sensor with a larger cell constant142

Sensor signal

F

On

On

C

No conductivity displayed
Reasons: sensor in air, sensor defective
1. Check installation
2. Replace sensor

143

Sensor check

F

On

Off

C

Sensor self-test error
1. Replace sensor
2. Contact the Service Department

144

Cond. out of rng

S

Off

On

C

Conductivity outside the measuring range
Use a sensor with a suitable cell constant146

Sensor temp.

S

Off

Off

C, N, T, S

Temperature outside specification
1. Check temperature
2. Check measuring chain
3. Replace sensor type

Endress+Hauser

Limit values for sensor relaxation time
(fluorescence decay time)exceeded/undershot
Reasons: high oxygen content, incorrect
calibration
1. Repeat calibration
2. Replace sensor cap
3. Replace sensor

23

Diagnostics and troubleshootingNo.

Message

Liquiport 2010 CSP44

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

147

Sensor check

F

On

On

C (ind.)Coil transmission current too highReasons: transmission coil short-circuit,inductance too low
1. Replace sensor
2. Contact the Service Department

148

Sensor check

F

On

On

C (ind.)Coil transmission current too low
Reasons: transmission coil interrupted,
inductance too high
1. Replace sensor
2. Contact the Service Department

149

Sensor LED

F

On

On

T

Sensor LED failure
1. Replace sensor
2. Contact the Service Department

151

Sensor buildup

F

On

On

T

Buildup, high level of contamination
1. Clean sensor
2. Replace sensor
3. Contact the Service Department

152

Sensor data

M

Off

Off

C (ind.)No calibration data
Perform air set calibration

153

Sensor failure

F

On

On

N, T, S

Sensor strobe lamp defective
Reasons: deterioration, end of operating life,mechanical interference/vibration
1. Replace sensor
2. Contact the Service Department

154

Sensor data

M

Off

Off

C

Factory calibration is used
Perform calibration

155

Sensor failure

F

On

On

N, T, S

Sensor failure
Error with analog evaluation
1. Replace sensor
2. Contact the Service Department

156

organ. pollution

M

On

On

N, T, S

Excessive organic fouling
Reasons: sensor fouling, high organic content,
incorrect orientation
1. Clean sensor
2. Install automatic cleaning
3. Check application

157

Filter change

M

On

Off

N, S

Optical filter must be replaced
Reasons: long period of operation, moisture insensor
1. Replace sensor
2. Contact the Service Department

158

Sensor check

F

On

On

N, T, S

Invalid measured value
1. Check sensor power supply
2. Restart device
3. Contact the Service Department

24

Endress+Hauser

Liquiport 2010 CSP44

No.

Message

Diagnostics and troubleshooting

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

159

Sensor check

F

On

On

N, T, S

Uncertain measured value
Reasons: sensor fouling, incorrect application
1. Clean sensor
2. Check application

160

Sensor data

F

On

On

N, T, S, Cl

No calibration data
Reasons: data deleted
1. Select other data record
2. Use factory calibration
3. Contact the Service Department

161

Filter change

F

On

Off

N, T, S

Filter change necessary
Reasons: long period of operation, moisture insensor
1. Replace sensor
2. Contact the Service Department

162

Install.factor

M

On

Off

C (ind.)163

Install.factor

M

On

Off

C (ind.)Installation factor exceeded/undershot, alarm
Reason: distance between wall and sensor too
small (< 15 mm)
1. Check pipe diameter
2. Clean sensor
3. Calibrate sensor

164

Sensor data

M

Off

Off

C

No temperature calibration data
Factory calibration is used
1. Check process
2. Check sensor, replace if necessary

168

Polarization

S

On

Off

C (cond.)Polarization warning
The measured value is distorted at highconductivity levels.
Use a sensor with a larger cell constant

169 - 170:
Warning issued by hours of operation monitoring system. Measurement can still take place.
1. Replace sensor
2. Adjust monitoring limit
3. Deactivate monitoring
169

Operating time

M

On

Off

S

Operating hours, conc. > 200 mg/l

170

Operating time

M

On

Off

S

Operating hours, conc. < 50 mg/l

171

Lamp change

M

On

Off

N, T, S

Need to change lamp
1. Replace sensor
2. Contact the Service Department

172

Echo lost

F

On

On

U

Echo signal lost173

Sludge level

F

On

On

U

Incorrect interface measurement.
Replace sensor.

174

Turbid. failure

F

On

On

U

Incorrect turbidity measurement.
Replace sensor.

175

Wiper failure

F

On

On

U

Wiper not working.Clean or replace sensor.

Endress+Hauser

25

Diagnostics and troubleshootingNo.

Message

Liquiport 2010 CSP44

Factory settingsCat.

Sensor type Tests or remedial measures

Diag. Error
current

176 - 199:
Warning issued by hours of operation monitoring system. Measurement can still take place.
1. Replace sensor
2. Adjust monitoring limit
3. Deactivate monitoring
176

Operating time

M

On

Off

Cl

Operating hours > 100 nA

177

Operating time

M

On

Off

Cl

Operating hours > 20 nA

178

Operating time

M

On

Off

Cl

Operating hours > 15 ˚C

179Operating time

M

On

Off

P

Operating hours > 300 mV

180

Operating time

M

On

Off

P

Operating hours < -300 mV

181

Operating time

M

On

Off

O (opt.)Operating hours < 25 μS

182

Operating time

M

On

Off

O (opt.)Operating hours > 40 μS

183

Operating time

M

On

Off

O (amp.)Operating hours > 10 nA (COS51D)

184

Operating time

M

On

Off

O (amp.)Operating hours > 30 nA (COS22D)

185

Operating time

M

On

Off

O (amp.)Operating hours > 40 nA (COS51D)

186

Operating time

M

On

Off

O (amp.)Operating hours > 160 nA (COS22D)

187

Operating time

M

On

Off

C

Operating hours > 80 ˚C, 100 nS/cm

188

Operating time

M

On

Off

C, O

Operating hours < 5 ˚C

189

Operating time

M

On

Off

O

Operating hours > 5 ˚C

190Operating time

M

On

Off

O

Operating hours > 25 ˚C

191

Operating time

M

On

Off

O, I, Cl

Operating hours > 30 ˚C

192

Operating time

M

On

Off

O, I

Operating hours > 40 ˚C

193

Operating time

M

On

Off

P, C, OOperating hours > 80 ˚C

194Operating time

M

On

Off

P

Operating hours > 100 ˚C

195

Operating time

M

On

Off

C

Operating hours > 120 ˚C

196Operating time

M

On

Off

C

Operating hours > 125 ˚C

197

Operating time

M

On

Off

C

Operating hours > 140 ˚C

198Operating time

M

On

Off

C

Operating hours > 150 ˚C

199Operating time

M

On

Off

All except U Total operating hours

215

Simul. active

C

On

Off

All

26

Simulation active
End simulation by changing to measuring
mode

Endress+Hauser

Liquiport 2010 CSP44

No.

Message

Diagnostics and troubleshooting

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

408

Calib. aborted

M

Off

Off

P, C, O, I, Cl

Calibration aborted

500

Sensor calib.

M

On

Off

All

Calibration aborted, main measured value
fluctuates
Reasons: sensor too old, sensor occasionallydry, calibration value not constant
1. Check sensor
2. Check calibration solution

501

Sensor calib.

M

On

Off

All except U Calibration aborted, temperature measured
value fluctuates
Reasons: sensor too old, sensor occasionallydry, temperature of calibration solution notconstant
1. Check sensor
2. Regulate calibration solution
temperature

505 - 522:
Limit values of calibration monitoring system exceeded/undershot. Measuring can continue if a warning is issued.
Possible reasons: sensor old or defective, reference blocked, calibration solution too old or contaminated
1. Check sensor, replace if necessary
2. Check calibration solution, replace if necessary
3. Repeat calibration
505

Sensor calib.

M

On

Off

P, O, I, Cl

Max. zero point warning

507

Sensor calib.

M

On

Off

P, O, I, Cl

Min. zero point warning509

Sensor calib.

M

On

Off

P, O, I, Cl

Min. slope warning

511

Sensor calib.

M

On

Off

P, O, I, Cl

Max. slope warning

513

Zero Warn

M

On

Off

O (amp.), Cl

Zero point warning

515

Sensor calib.

M

On

Off

P (ISFET)Max. operating point warning

517

Sensor calib.

M

On

Off

P (ISFET)Min. operating point warning

518

Sensor calib.

M

On

Off

P, O, I, Cl

Delta slope warning

520

Sensor calib.

M

On

Off

P, O, I, Cl

Delta zero point warning

522

Sensor calib.

M

On

Off

P (ISFET)

Delta operating point warning

523

Sensor calib.

M

On

On

C

Invalid cell constant, max./min. value or
lower/upper warning value reached
1. Calibrate sensor
2. Replace sensor

524

Sensor calib.

M

On

On

C

526

Sensor calib.

M

On

Off

C

528

Sensor calib.

M

On

Off

C

Endress+Hauser

27

Diagnostics and troubleshootingNo.

Message

Liquiport 2010 CSP44

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

534

Sensor calib.

M

On

Off

Cl

Set limit for electrolyte consumption isreachedMeasurement can still take place.
1. Replace the electrolyte
2. Clear the electrolyte consumption
counter

535

Sensor check

M

On

Off

O (amp.), Cl

Specified number of cap calibrations isreachedMeasurement can still take place.
Replace sensor cap

550

Process temp.

S

On

On

C

551

Process temp.

S

On

On

C

Process temperature above/below
concentration table– Process value outside specification– Incomplete table--> Extend table

552

Process conduc.

S

On

On

C

553

Process conduc.

S

On

On

C

554

Process conc.

S

On

On

C

555

Process concent.

S

On

On

C

Process conductivity above/below
concentration table– Process value outside specification– Incomplete table--> Extend table
Process concentration above/below
concentration table– Process value outside specification– Incomplete table--> Extend table

556

Process temp.

S

On

On

C

557

Process temp.

S

On

On

C

Process temperature above/below
compensation table– Process value outside specification– Incomplete table--> Extend table

558

Process conduc.

S

On

On

C

559

Process conduc.

S

On

On

C

Process compensation above/below
compensation table– Process value outside specification– Incomplete table--> Extend table

560

Proc.cond.comp

S

On

On

C

561

Proc.cond.comp

S

On

On

C

720

Membrane change

M

On

Off

I

Conductivity compensation above/below
compensation table– Process value outside specification– Incomplete table--> Extend table

28

Membrane cap replacement necessary
1. Replace membrane cap
2. Reset timer

Endress+Hauser

Liquiport 2010 CSP44

No.

Message

Diagnostics and troubleshooting

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

722

Sensor reference

F

On

On

P

Alarm: impedance of reference membrane
too low.
1. Check sensor, replace if necessary
2. Check and correct reference limit value

723

Sensor reference

M

On

Off

I

Warning: impedance of reference membrane
too low. Measurement can continue until thealarm level.
1. Check sensor, replace if necessary
2. Check and correct reference limit value

724

Sensor reference

F

On

On

I

Alarm: impedance of reference membrane
too high.
1. Check sensor, replace if necessary
2. Check and correct reference limit value

725

Sensor reference

M

On

Off

I

Warning: impedance of reference membrane
too high. Measurement can continue until thealarm level.
1. Check sensor, replace if necessary
2. Check and correct reference limit value771

Lamp change

F

On

Off

N, T, S

Lamp change alarm• Configured operating time has beenreached-> Replace lamp-> Contact the Service Department

841

Operating range

S

Off

Off

All

Process value outside operational range
1. Check application
2. Check sensor

842

Process value

S

Off

Off

P

843

Process value

S

Off

Off

P

Process limit value exceeded/undershot
Reasons: sensor in air, air pockets inassembly, incorrect flow to sensor, sensor
defective
1. Change process value
2. Check measuring chain
3. Change sensor type

844

Process value

S

Off

Off

N, T, S

Measured value outside specified range
Reasons: sensor in air, air pockets inassembly, incorrect flow to sensor, sensor
defective
1. Increase process value
2. Check measuring chain
3. Change sensor type

904

Process check

F

On

On

All

Stagnating measuring signal
Reasons: sensor in air, sensor fouling,
incorrect flow to sensor, sensor defective
1. Check measuring chain
2. Check sensor
3. Restart device

Endress+Hauser

29

Diagnostics and troubleshootingNo.

Message

Liquiport 2010 CSP44

Factory settings
Cat.

Diag. Error
current

Sensor type Tests or remedial measures

914

USP/ EP alarm

M

On

Off

C

915

USP / EP warning

M

On

Off

C

934

Process temp.

S

Off

Off

All except U Process temperature high
1. Do not increase temperature
2. Check measuring chain
3. Change sensor type

935

Process temp.

S

Off

Off

All except U Process temperature low
1. Do not reduce temperature
2. Check measuring chain
3. Change sensor type

942

Process value

S

Off

Off

All except U Process value high
1. Do not increase process value
2. Check measuring chain
3. Change sensor type

943

Process value

S

Off

Off

All except U Process value low
1. Do not decrease process value
2. Check measuring chain
3. Change sensor type

944

Sensor range

S

On

Off

S

Measurement at the margin of sensor's
dynamic range
Reasons: Changes in the process to a higher
or lower measuring range
1. Check application
2. Use sensor suited to the measuring
range of the application

983

Sensor ISE check

F

On

On

I

Electrode or membrane defective
1. Check electrode, replace if necessary
2. Check membrane cap, replace ifnecessary

984

Process temp.

S

On

On

I

Temperature outside specification
1. Check process temperature
2. Check measuring chain

985

Sensor Interface

F

On

On

I

Sensor interface error
1. Check plug
2. Check cable, replace if necessary

987

Calib. req.

M

On

On

I

Electrode change--> Calibration required

30

USP limit values exceeded
Check process

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

1.4.3 Configuration options for troubleshooting (for version with sensors with the
Memosens protocol)
The table only lists the diagnostics messages that depend on your settings in the menu. The
path where you can change the settings is specified in the table.
The sensor type is also indicated in the path if the message only applies to one type of sensor.
All other settings affect several sensor types.
No.

Path to software function

102

Menu/Setup/Inputs/Extended setup/Calib. settings/Calibration timer

103

Menu/Setup/Inputs/Extended setup/Calib. settings/Calibration timer/Calibration timer

104

Menu/Setup/Inputs/Extended setup/Calib. settings/Calib. expiration date/Alarm limit

105

Menu/Setup/Inputs/Extended setup/Calib. settings/Calib. expiration date/Warning limit

108

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Sterilizations/Warning limit

109

Menu/Setup/Inputs/Oxygen (amp.)/Extended setup/Diagnostics settings/Number of cap sterilizations/Warning
limit

122

Menu/Setup/Inputs/pH Glass/Extended setup/Diagnostics settings/Glass impedance (SCS)/Lower alarm limit

123

Menu/Setup/Inputs/pH Glass/Extended setup/Diagnostics settings/Glass impedance (SCS)/Lower warning limit

124

Menu/Setup/Inputs/pH Glass/Extended setup/Diagnostics settings/Glass impedance (SCS)/Upper alarm limit

125

Menu/Setup/Inputs/pH Glass/Extended setup/Diagnostics settings/Glass impedance (SCS)/Upper warning limit

126

Menu/Setup/Inputs/pH Glass/Extended setup/Diagnostics settings/Sensor Condition Check

127

Menu/Setup/Inputs/pH Glass/Extended setup/Diagnostics settings/Sensor Condition Check

145

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Delta slope/Alarm limit

157

Menu/Setup/Inputs/Nitrate/Extended setup/Diagnostics settings/Limits operating hours/Filter change

168

Menu/Setup/Inputs/Cond c/Extended setup/Polarization detetected

169

Menu/Setup/Inputs/SAC/Extended setup/Diagnostics settings/Limits operating hours/Operation > 200 mg/l

170

Menu/Setup/Inputs/SAC/Extended setup/Diagnostics settings/Limits operating hours/Operation < 50 mg/l

176

Menu/Setup/Inputs/Chlorine/Extended setup/Diagnostics settings/Limits operating hours

178

Menu/Setup/Inputs/Oxygen (amp.)/Extended setup/Diagnostics settings/Number of cap sterilizations/Alarm
limit

179

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 300 mV

180

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation < -300 mV

181

Menu/Setup/Inputs/Extended setup/Oxygen (opt.)/Diagnostics settings/Limits operating hours/Operation < 25μs

182

Menu/Setup/Inputs/Oxygen (opt.)/Extended setup/Diagnostics settings/Limits operating hours/Operation > 40μs

183

Menu/Setup/Inputs/Oxygen (amp.)/Extended setup/Diagnostics settings/Limits operating hours/Operation >15 nA

Endress+Hauser

31

Diagnostics and troubleshooting

No.

Liquiport 2010 CSP44

Path to software function

184

Operating time

185

Menu/Setup/Inputs/Oxygen (amp.)/Extended setup/Diagnostics settings/Limits operating hours/Operation >
50 nA

186

Operating time

187

Menu/Setup/Inputs/Cond c/Extended setup/Diagnostics settings/Limits operating hours/Operation > 80°C <100nS/cm

188

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation < 5°C

190

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 25°C

192

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 40°C

193

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 80°C

194

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 100°C

195

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 120°C

196

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 125°C

197

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 140°C

198

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operation > 150°C

199

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours/Operating time

505

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Zero point/Upper warning limit

507

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Zero point/Lower warning limit

509

Menu/Setup/Inputs/Oxygen (amp.)/Extended setup/Diagnostics settings/Slope/Lower warning limit

511

Menu/Setup/Inputs/Oxygen (amp.)/Extended setup/Diagnostics settings/Slope/Upper warning limit

513

Menu/Setup/Inputs/Oxygen (amp.)/Extended setup/Diagnostics settings/Zero point/Warning limit

515

Menu/Setup/Inputs/pH ISFET/Extended setup/Diagnostics settings/Operating point/Upper warning limit

517

Menu/Setup/Inputs/pH ISFET/Extended setup/Diagnostics settings/Operating point/Lower warning limit

518

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Delta slope/Warning limit

520

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Delta zero point/Warning limit

522

Menu/Setup/Inputs/pH ISFET/Extended setup/Diagnostics settings/Delta operating point/Warning limit

842

Menu/Setup/Inputs/Redox/Extended setup/Diagnostics settings/Redox-Meas value/Upper alarm limit

843

Menu/Setup/Inputs/Redox/Extended setup/Diagnostics settings/Redox-Meas value/Lower alarm limit

904

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Process Check System

934

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours

935

Menu/Setup/Inputs/Extended setup/Diagnostics settings/Limits operating hours

942

Menu/Setup/Inputs/Redox/Extended setup/Diagnostics settings/Redox-Meas value/Upper warning limit

943

Menu/Setup/Inputs/Redox/Extended setup/Diagnostics settings/Redox-Meas value/Lower warning limit

32

Endress+Hauser

Liquiport 2010 CSP44

1.5

Diagnostics and troubleshooting

Pending diagnostic messages

The Diagnostics menu contains all the information on the device status.
Furthermore, various service functions are available.
The following messages are directly displayed every time you enter the menu:• "Most important message"
Diagnostics message recorded with the highest criticality level• "Past message"
Diagnostic message whose cause is no longer present.
All the other functions in the Diagnostics menu are described in the following chapters.
Diagnostics messages associated with sampling are deleted under the following conditions:• Diagnostics messages caused by sampling are deleted automatically with the next successful
sampling.
• Diagnostics messages caused by the level of medium in the bottle are deleted the next time
the bottle is changed.

1.6

Diagnostic list

All the current diagnostics messages are listed here.
A time stamp is available for each message. Furthermore, the system also displays theconfiguration and description of the message as saved in "Menu/Setup/General
settings/Diagnostics/Device behavior".
For this purpose, select the appropriate message and press the navigator.

Endress+Hauser

33

Diagnostics and troubleshooting

1.7

Liquiport 2010 CSP44

Logbooks

1.7.1 Available logbooks
Types of logbooks• Logbooks physically available (all apart from the overall logbook)• Database view of all logbooks (=overall logbook)
Logbook

Visible inMax.
entriesCan bedisabled

Logbook
can bedeletedEntriescan bedeletedCan beexported1)

Program logbook

Program logbook

5000

Yes

No

Yes

Yes

Overall logbook

All events

1000

Yes

No

Yes

No

Diagnostics logbook

Diagnostic events

250

(Yes)

No

Yes

Yes

Calibration logbook

Calibration events

75

(Yes)

No

Yes

Yes

Operation logbook

Configuration events

250

(Yes)

No

Yes

Yes

Version logbook

All events

50

No

No

No

Yes

Hardware version
logbookAll events

125

No

No

No

Yes

Data logbook

Data logbooks

150,000

Yes

Yes

Yes

Yes

Debugging logbook

Only accessible with thespecial activation code
(Service)

1000

Yes

No

Yes

Yes

1)

Data in brackets means this depends on the overall logbook

1.7.2 Logbooks menu
Diagnostics/Logbooks
Function

Options

Program logbook

Chronological list of the programming events.

Show

Events are displayed

Select a particular event to display more detailed
information.

Go to date

User input• Go to date
• TimeUse this function to go directly to a specific time in thelist. In this way, you avoid having to scroll through all
the information. The complete list is always visible,
however.

Read only

The bottle statistics for the sampler are displayed. The
statistics are displayed for each individual bottle when
the program is started.
Further information is provided in the "Bottle statistics"
section.

Show summary of
current program

34

Info

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

Diagnostics/Logbooks
Function

Options

Info

Summary of inputs

Read only

The counters configured for the analog and binary input
are displayed.
Max. 8 lines

Delete all entries

ActionYou can delete all the program logbook entries here.

All events

Chronological list of all the logbook entries, withinformation on the type of event.

Show

Events are displayed

Select a particular event to display more detailed
information.

Go to date

User input• Go to date
• TimeUse this function to go directly to a specific time in thelist. In this way, you avoid having to scroll through all
the information. The complete list is always visible,
however.

Calibration events

Chronological list of the calibration events.

Show

Events are displayed

Select a particular event to display more detailed
information.

Go to date

User input• Go to date
• TimeUse this function to go directly to a specific time in thelist. In this way, you avoid having to scroll through all
the information. The complete list is always visible,
however.

Delete all entries

ActionYou can delete all the calibration logbook entries here.

Configuration events

Chronological list of the configuration events.

Show

Events are displayed

Select a particular event to display more detailed
information.

Go to date

User input• Go to date
• TimeUse this function to go directly to a specific time in thelist. In this way, you avoid having to scroll through all
the information. The complete list is always visible,
however.

Delete all entries

ActionYou can delete all the operation logbook entries here.

Diagnostic events

Chronological list of the diagnostics events.

Show

Events are displayed

Select a particular event to display more detailed
information.

Go to date

User input• Go to date
• TimeUse this function to go directly to a specific time in thelist. In this way, you avoid having to scroll through all
the information. The complete list is always visible,
however.

Delete all entries

ActionYou can delete all the diagnostics logbook entries here.

Endress+Hauser

35

Diagnostics and troubleshooting

Liquiport 2010 CSP44

You can view your data logbook entries graphically on the display ("Show plot").
You can also adapt the display to suit your individual requirements:• If you press the navigator button in the graphic display, you are given additional options suchas the zoom function and x/y movement of the graph.
• Furthermore, you can also define a cursor. If you select this option, you can move along thegraph with the navigator and view the logbook entry (data stamp/measured value) in text
form for every point in the graph.
• Simultaneous display of two logbooks ("Select 2nd plot" and "Show plot"),  å 1:
– A small cross marks the currently selected graph for which the zoom can be changed or acursor used, for example.
– You can select the other graph in the context menu (by pressing the navigator button), andthen apply the zoom function, a movement or a cursor to this graph.
– Using the context menu, you can also select both graphs simultaneously. This allows you to
use the zoom function simultaneously on both graphs, for example.

Fig. 1: Two graphs displayed simultaneously. The top graph is "selected".

36

a0016688

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

Diagnostics/Logbooks
Function

Options

Info

Data logbooks

Chronological list of the data logbook entries.

Data logbook 1 <Logbook
name>This submenu is available for each data logbook that you
have set up and activated.

Source of data

Read only

Displays the input or the mathematical function

Measured value

Read only

Displays the measured value that is recorded

Log time left

Read only

Displays the days, hours and minutes until the logbook
is full. Please note the instructions regarding theselection of the memory type in the General
settings/Logbooks menu (--> BA "Operation andsettings").

Show

Events are displayed

Select a particular event to display more detailed
information.

Show plot

Graphic display of thelogbook entries

The entries are displayed according to your settings in
the General settings/Logbooks menu.

Select 2nd plot

Choice of another datalogbookYou can view a second logbook at the same time as thecurrent logbook.

Go to date

User input• Go to date
• TimeUse this function to go directly to a specific time in thelist. In this way, you avoid having to scroll through all
the information. The complete list is always visible,
however.

Delete all entries

ActionYou can delete all the data logbook entries here.

Options
• CSV
• FDM

Save the logbook in the preferred file format.
You can then open the file you saved (.csv) on the PC andprocess it in MS-Excel for example.1).
You can import the FDM files into Fieldcare and archivethem so they are tamper-proof.

The action commences assoon as the option is
selected

Save the logbook in the preferred file format. You canthen open the CSV file you saved on the PC and process
it in MS-Excel for example.
You can import the FDM files into Fieldcare and archivethem so they are tamper-proof.

Save logbooks
File format

Program logbook
All data logbooks
Data logbook 1...nAll event logbooks
Calibration logbook
Diagnostic logbook
Configuration logbook
HW version logbook
Version logbook

The file name is made up of the "Logbook ident" (Menu/Setup/General settings/Logbooks), an abbreviation forthe particular logbook and a time stamp.
1)

CSV files use international number formats and separators. Therefore they must be imported into MS Excel asexternal data with the correct format settings. If you double-click the file to open it, the data are only displayedcorrectly if MS Excel is installed with the US country setting.

Endress+Hauser

37

Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.7.3 Program logbook
The following table shows an overview of the exported program logbook and explains the mostimportant terms in the program logbook.
Entry

ExampleInfo

Timestamp

05.05.2010 12:40

Time stamp - the start time in the case of sampling

Event

BasicPrgStart

Power on--> Time the device was started
Power failure--> Time the power failed (to the minute)
BasicPrgStart, StdPrgStart--> Time the program was started
BasicSampling, StdSampling--> Entry made during sampling
PrgPartStart, PrgPartStop--> Time a subprogram is enabled and disabled
PrgStop--> Time the program was ended

Name

Program1

In the case of BasicPrgStart, StdPrgStart, BasicSampling or
PrgStop--> the name of the program appears
In the case of StdSampling, PrgPartStart or PrgPartStop--> the name of the subprogram appears

Bottle configuration

12x+6x PE/glass plate
distribution

The selected bottle configuration is displayed

Left bottle volume

1000

Right bottle volume

3000

The bottle volume is displayed--> "Right bottle volume" remains empty for bottle
configurations with different volumes

Sampling mode

Time-paced CTCV

Time-paced CTCV --> in proportion to timeFlow-paced VTCV --> in proportion to volume
Time/flow-paced CTVV --> in proportion to flow
Single sample --> single sample
Sample table --> single sample--> the sampling mode is displayed

Sampling interval/unit

10 min--> The interval and unit are displayed

Samples/bottle

4With bottle change--> Number of samples per bottle

Bottles/sample

0

Multiple bottles

Sampling volume/unit

100 mlSample volume when sampling

38

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

Entry

ExampleInfo

Start mode

Immediate

Field only populated for PrgPartStart, BasicPrgStart andStdPrgStart:--> The program start setting is displayed- Immediate --> immediately- Date/time --> after date/time- Volume --> with a volume- Event --> when an event occurs- Interval --> after an interval- Individual dates --> individual timetable- Multiple date --> multiple dates

Start date

05.05.2010

Field only populated if Start mode = Date/Time:--> The start date is displayed

Stop mode

Program end

The program stop setting is displayed- Program end --> when the program ends
- Continuous --> continuous operation- Bottles full --> when bottles are full
- Date/time --> after date/time- Event --> when an event occurs

Stop date

06.05.2010

Field only populated if Program end = Date/Time:--> The time the program was stopped is displayed

Start flow sum/unit

100 m³

Field only populated if Start mode = Volume:--> The starting volume is displayed

Bottle number

1

Field only populated in the case of BasicSampling or
StdSampling:--> The bottle which was filled with the sample is displayed

Sample nbr

2

Number of samples transferred to the current bottle

Sampling result

Sampling Ok

Sampling Ok --> sampling ok
Sampling nOk --> sampling failed--> For detailed diagnostics messages, see the diagnostics
logbook

Running sample number

1

Running sample number in the current program

Flow sum since last
sampling

1

For flow-paced and time/flow-paced sampling:--> Flow since the last sampling
For all other types of sampling:--> Display: 0

Endress+Hauser

39

Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.7.4 Bottle statistics
In "Menu/Diagnostics/Logbooks/Logbook program", select the "Show summary of current
program" item to display the bottle statistics for the sampler. The statistics are displayed for eachindividual bottle when the program is started. This gives you detailed feedback on the last
sampling operations.
The statistics are deleted when the following event occurs:• Program is started
The statistics are selectively overwritten when the following event occurs:• When the 1st bottle is reached in situations where "Continuous operation" is configured
as the end of the program in the program settings.
The statistics are displayed as follows:

Column

Display

Info

1

hh:mm

The time the first sample was transferred to the bottle is displayed.

2bt

The bottle number is displayed.

3

Smp

Displays how often sampling was triggered per bottle.

4

n.s.

Indicates the number of times a sample was not taken even though sampling was triggered. Thiscan occur if the maximum permissible fill volume for the bottle has been reached but the systemis still supposed to transfer samples to the bottle. The "Overfill sensor" message is displayed
while the program is active.

5

n.f.

The value indicates how often sampling was canceled since the system was unable to take in any
medium, or enough medium, into the dosing chamber to cover the LF1 probe.

6ml

The sampling volume collected per bottle is displayed.

7

Q

The total flow for every bottle is displayed (if connected).

40

Endress+Hauser

Liquiport 2010 CSP44

1.8

Diagnostics and troubleshooting

Device information

1.8.1 System information
Diagnostics/System information
Function

Options

Info

Device tag

Read only

Individual device tag, --> "General settings"

Order code

Read only

You can order identical hardware with this code.
This code changes on account of changes to thehardware and you can enter the new code you received
from the manufacturer here1).

To find out what device version you have, enterthe order code in the search screen at the following address:www.products.endress.com/order-ident
Orig. order code ext.

Read only

Complete order code for the original device, resulting
from the product structure.

Current order code ext.

Customized text

Current code, taking into account changes to thehardware. You must enter this code yourself.

Serial number

Read only

The serial number allows you to access device data anddocumentation on the Internet:
www.products.endress.com/device-viewer

Software version

Read only

Current version

Sw version FMSY1

Read only

Current version

FMSY1-proj. version

Read only

Current version

ENP version

Read only

Version of the electronic nameplate

Read only• Description• Serial number• Order code
• Hardware version• Software version

This information is provided for every electronics
module available. Specify the serial numbers and ordercodes when servicing, for example.

Read only• Description• Serial number• Order code
• Hardware version• Software version

This information is provided for every sensor available.
Specify the serial numbers and order codes whenservicing, for example.

System modules
Depends on the electronics
module available, e.g.:
BaseSensors
Depends on the sensors
connected

1)

Provided you give the manufacturer all the information about changes to the hardware.

Endress+Hauser

41

Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.8.2 Sensor information
Select the channel you want from the list of channels.
Information in the following categories is displayed:• Extreme values
Extreme conditions to which the sensor has been exposed up to now, e.g. min./max.
temperatures1)• Operating timeSensor operation under extreme conditions• Calibration information
Calibration data of the last calibration• Sensor specifications
Measuring range limits for main measured value and temperature• General information
Sensor identification information
The specific data that are displayed depends on what sensor is connected.

1.9

Resetting the measuring device

Diagnostics/Systemtest/Reset
Function

Options

Info

Power supply

Read only

The current supply voltage is displayed.

Manual sampling
Bottle configuration

Read only

Bottle volume

Read only

Distributor position

Options• Bottle 1...

Select which bottle should be filled with the sample.

10 to 10000 ml

You can change the sample volume.

Sample volume

Factory setting
100 mlStart sampling

ActionPeristaltic pump

1)

42

Pump purge

ActionPump purge, to stoppress ESC

Read only

Current pump run time

Read only

Not available for all sensor types.

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

Diagnostics/Systemtest/Reset
Function
Power supply

Options

Info

Read only

The current supply voltage is displayed.
With AC power supply: 24 V ±0.5 V
With DC power supply: 22 to 28 V

Motor current

Read only

The current consumption of the pump is displayed.

VacuumRead only

The vacuum is an indicator of the suction height.
-> 100 mbar is equivalent to a suction height of approx.
1 m.

Medium detected

Read only

Yes: the medium was detected
No: no medium was detected

Pump suction

ActionPump suction, to stoppress ESC

Read only

Current pump run time

Read only

Power supply

Read only

The current supply voltage is displayed.
With AC power supply: 24 V ±0.5 V
With DC power supply: 22 to 28 V

Motor current

Read only

The current consumption of the pump is displayed.

VacuumRead only

The vacuum is an indicator of the suction height.
-> 100 mbar is equivalent to a suction height of approx.
1 m.

Medium detected

Read only

Yes: the medium was detected
No: no medium was detected

Distribution arm

ActionOnly for bottle configurations with more than one
bottle.

Test distribution arm

Read only

Position

Read only

When the menu item is activated, the distribution arm
undergoes a test run. Afterwards, the system moves toeach position in succession and the position is displayed.
In the case of plate distribution, the arm moves left andright to ensure the bottles are numbered consecutively.
Calibrate the distribution arm if the arm is not
positioned precisely over the bottles.

Device reset

Options• OK• ESC

Restart and keep all the settings

Factory default

Options• OK• ESC

Restart with factory settings
Settings that have not been saved are lost.

Power supply

Read only• Digital Supply 1: 1.2V
• Digital Supply 2: 3.3V
• Analog Supply: 12.5V• Sensor Supply: 24V• Temperature

Detailed list of power supply to instrument.

Endress+Hauser

The actual values can vary without a malfunction
having occurred.

43

Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.10 Runtime info
The following information is displayed:• Operating hours device:
Displays the total operating hours of the device in days, hours and minutes• Sample totalizer :Number of all samples taken and sample errors• Pump tube life :Displays how old the tube is in days, hours and minutes
This counter must be reset when a tube is replaced.
Set the specific counter reading to zero with "Reset".

1.11 Status of inputs/outputs
Path:Display/Operation/Measurement
The following measured values are listed (read only):• Binary inputs
Current function state: on or off• Current inputs
Actual current values of all the current inputs available• Binary outputs
Current function state: on or off• Temperature sensors
Current value is displayed:• Current outputs (for version with sensors with the Memosens protocol)
Actual current values of the current outputs

44

Endress+Hauser

Liquiport 2010 CSP44

Diagnostics and troubleshooting

1.12 Sensor replacement (for version with sensors with the Memosens
protocol)Diagnostics/Sensor change
Function

Options

Info

List of channels

Options• On• Off

If you set the function to "On", the measured value
at the current output is set to hold. In this way you
avoid an error being reported at the process control
system if the sensor is replaced on site.

Factory settingOff
All channels sensor change on Action
All channels sensor change
offActionYou can set the hold individually for every
Memosens channel. Alternatively, you can set all
the Memosens channels simultaneously to hold, or
cancel the hold.
Once you have replaced the sensor, you have todisable the hold at the same point.

1.13 Manual hold (for version with sensors with the Memosens
protocol)Diagnostics/Manual hold
Function

Options

Info

List of channels

Options• On• Off

If you set the function to "On", the measured value atthe current output is set to hold.

Factory settingOff
All channels manual hold on

ActionAll channels manual hold off

ActionEndress+Hauser

You can set the hold individually for every
Memosens channel. Alternatively, you can set all
the Memosens channels simultaneously to hold, or
cancel the hold.
Once you have completed the maintenance task,
you have to disable the hold at the same point.

45

Diagnostics and troubleshooting

Liquiport 2010 CSP44

1.14 Firmware history
Date

VersionChanges to software

04/2013 01.04.00 Extension• Conductivity:– Measuring range switch– Temperature compensation ISO 7888 at 20 ˚C• Keylock with password protection• pH:– Icon for manual and automatic temperature compensation(ATC/MTC+MED)– Monitoring of the upper and lower limit of the glass SCS
value can be switched on/off independently of eachanother• ISE– Simultaneous calibration of two parameters– User-defined electrode type
– Raw measured values can be selected for current output– Timer for membrane replacement• Logbooks remain intact after the firmware update

Documentation: edition
BA00465C/07/EN/15.13
BA00470C/07/EN/15.13
BA00492C/07/EN/15.13
BA00493C/07/EN/15.13
SD01068C/07/EN/01.12

Improvement• Offset icon only for pH or ORP• Turbidity: autorranging can be switched off• Export Print (xml): export file revised and style sheet added
for better legibility.
• Overview of input with counter function• Input menu accessible via program creation• External signal for basic programs• Quick programming via start screen
07/2012 01.03

Extension• USP/EP (United States Pharmacopeia and European
Pharmacopeia) and TDS (Total Dissolved Solids) for
conductivity

BA00465C/07/EN/14.12
BA00470C/07/EN/14.12
BA00492C/07/EN/14.12
BA00493C/07/EN/14.12

Improvement• Adapted factory settings• SAC: Factory calibration in the field incl. filter operating time
reset and lamp change• ISFET leak current visible in measuring screen• Multiselect for limit switch and cleaning cycles
04/2011 01.02

Extension• Support for additional sensors:– Chlorine
– ISE
– SAC
– Interface• Mathematics functions

BA465C/07/EN/13.11
BA470C/07/EN/13.11
BA492C/07/EN/13.11
BA493C/07/EN/13.11

Improvement• Modified software structures• Adapted factory settings• User-defined measuring screens
06/2010 01.00

46

Original software

BA465C/07/EN/06.10
BA470C/07/EN/06.10
BA464C/07/EN/04.10
BA467C/07/EN/04.10

Endress+Hauser

Liquiport 2010 CSP44

2

Maintenance

Maintenance! WARNING
Process pressure and temperature, contamination, electrical voltage
Risk of serious or fatal injury‣ De-energize the device and disconnect the battery connection.
‣ If a sensor has to be removed during maintenance work, avoid hazards posed by pressure,temperature and contamination.

NOTICE
Electrostatic discharge (ESD)
Risk of damaging the electronic components‣ Take personal protective measures to avoid ESD, such as discharging beforehand at PE or
permanent grounding with a wrist strap.
‣ For your own safety, only use genuine spare parts. With genuine parts, the function,accuracy and reliability are also ensured after maintenance work.

2.1

Recommended maintenance

Maintenance work has to be carried out at regular intervals to ensure the efficient operation of
the sampler.
The maintenance work comprises:• Replacing the wear parts• Cleaning the device
The cleaning intervals depend heavily on:• The medium• The ambient conditions of the sampler (dust etc.)• The programming intervals
For this reason, adapt the cleaning intervals to your specific requirements but always ensurethat these cleaning tasks are performed regularly.
Replacing wear parts
Wear parts are replaced by Endress+Hauser Service at one- and two-year intervals. Pleasecontact your local sales center in this regard.
Endress+Hauser offers its customers a maintenance contract. With a maintenancecontract, you can increase your level of operational safety and relieve your operating staff
of some of their workload. Ask your Endress+Hauser Service Organization for detailed
information on maintenance contracts.

Endress+Hauser

47

Maintenance

2.2

Liquiport 2010 CSP44

Replacing the pump tube

! CAUTION
Danger of injury due to rotating parts‣ Take the sampler out of service before opening the peristaltic pump.
‣ Safeguard the sampler against accidental operation while working on the open peristaltic
pump.

2

2
3
4
5
6
7

3
4

Pump tube
Fastening clip
Pump bracket
Pump head cover
Positioning pin
Knurled head screw

5

7
6
Fig. 2: Opening the peristaltic pumpa0014202

Open the peristaltic pump as follows:
1. Take the sampler out of service by pausing a program that is currently running.
2. Open the fastening clip (item 3) and push the pump bracket (item 4) upwards.
3. Remove the knurled head screw (item 7) and open the pump head cover (item 5)
downwards.

48

Endress+Hauser

Liquiport 2010 CSP44

Maintenance

2

9

2
8
9
10

10

Pump tube
Clamp
Marking ring
Roller8

Fig. 3: Replacing the pump tube

1.
2.
3.
4.
5.
6.
7.
8.

a0014203

Remove the clamp (item 8) and remove the pump tube (item 2) from the pump.
Remove any silicone deposits on the roller (item 10) and the flexible pump bracket.
Make sure the roller turns smoothly and evenly.
Apply some lubricant to the roller.
Secure the new pump tube to the pressure sensor with the clamp (item 8).
Guide the pump tube around the roller and insert the marking ring into the groove(item 9).
Close the pump head cover and screw it tight. Close the pump bracket.
Under Menu/Diagnostics/Term information/Pump tube life reset the tube life to zero byselecting "Reset".
Calibrate the sample volume each time you replace a pump tube.
--> See Operating Instructions BA00493C "Calibration".

NOTICE
Incorrect sample volume‣ Only reset the tube life counter to zero once you have successfully replaced the pump tube
in order to avoid incorrect dosing of the medium.

Endress+Hauser

49

Maintenance

2.3

Liquiport 2010 CSP44

Cleaning

2.3.1 Housing
Clean the housing with commercially available cleaning agents.
NOTICE
Prohibited cleaning agents
Damage to the housing surface or housing seal
‣ For cleaning purposes, never use concentrated mineral acids or bases.
‣ Never use organic cleaners such as benzyl alcohol, methanol, methylene chloride, xylene orconcentrated glycerol cleaner.
‣ Never use high-pressure steam for cleaning purposes.
2.3.2 Wetted parts
Peristaltic pump

1
Fig. 4: Peristaltic pump
1
2

2a0014215

Suction line
Liquid detector

Clean the wetted parts as follows:
1. Connect a container containing clear water to the suction line (item 1).
2. Remove the bottles from the sample compartment.
3. Rinse the wetted parts with clear water by taking a manual sample or by performing apump test (under Menu/Diagnostics/Systemtest/Reset/Peristaltic pump -> Pump
purge/Pump suction).
4. Release the couplings to the left and right of the pressure sensor (item 2). Clean the tube
piece carefully with a bottle brush.
5. Reconnect the sample supply to the tube connection and put the bottles back in the sample
compartment.

50

Endress+Hauser

Liquiport 2010 CSP44

MaintenanceInterior of peristaltic pump
! WARNING
Danger of injury due to rotating parts‣ Take the sampler out of service before opening the peristaltic pump.
‣ Safeguard the sampler against accidental operation while working on the open peristaltic
pump.
1
2
3

1

Pump rotor
RollerPump tube

2

3
Fig. 5: Interior view of the peristaltic pump

1.
2.
3.
4.
5.

a0014029

Take the sampler out of service by pausing a program that is currently running.
Open the peristaltic pump as described in the "Replacing the pump tube" section.
Remove the pump tube.
Remove any silicone deposits on the roller and the flexible pump bracket.
Make sure the roller turns smoothly and evenly.

Cleaning the distribution arm
Make sure the distribution arm is seated correctly! The distribution arm must be locked asotherwise the rotation movement could be blocked or the system might no longer
approach the bottles correctly.
Clean the distribution arm as follows:
1. Separate the upper from the lower compartment of the unit by opening the lockablelatches at the side. Turn the top device section 90˚.
2. Screw off the distribution arm.
3. Clean these parts with water or soapsuds. Use a bottle brush if necessary.
4. Reinstall the cleaned distribution arm.

Endress+Hauser

51

Maintenance

Liquiport 2010 CSP44

2.3.3 Sample compartment
The sample compartment has a fully integrated plastic inner lining.
Clean the sample compartment as follows:
1. Remove the bottles.
2. Spray-clean the sample compartment with a water hose.
You can wash the PE and glass bottles in a dishwasher at 60 ˚C.
2.3.4 Digital sensors (for version with sensors with the Memosens protocol)
1.

2.
3.
4.
5.
6.

If an error occurs or the maintenance schedule stipulates that the sensor has to be
replaced, use a new sensor, or a sensor that has been precalibrated in the laboratory.
A sensor is calibrated in the laboratory under optimum external conditions, thereby
ensuring better quality of measurement.
Remove the sensor to be serviced and install the new sensor.
You must perform calibration if you use a sensor that is not precalibrated.
The sensor data are automatically accepted by the transmitter. A release code is not
required.
Measurement is resumed.
Take the used sensor back to the laboratory. In the laboratory you can get the sensor readyfor reuse while ensuring the availability of the measuring point.
– Clean the sensor. For this purpose, use the cleaning agent specified in the sensor
manual.
– Inspect the sensor for cracks or other damage.
– If no damage is found, regenerate the sensor. Where necessary, store the sensor in aregeneration solution (--> sensor manual).
– Recalibrate the sensor for reuse.

2.3.5 Assemblies (for version with sensors with the Memosens protocol)
Refer to the assembly operating manual for information on servicing and troubleshooting theassembly. The assembly operating manual describes the procedure for mounting anddisassembling the assembly, replacing the sensors and seals, and contains information on thematerial resistance properties, as well as on spare parts and accessories.

52

Endress+Hauser

Liquiport 2010 CSP44

2.4

MaintenanceReplacing the storage batteriesFirst remove the cover of the battery compartment to replace the batteries.
! WARNING
Device is energized
Improper connection can cause injury or death.
‣ If a power supply unit or a charger is connected, disconnect it from the mains.

1

2

1

Fig. 6: Cover of battery compartment
1
2

Fixing screws
Cover of battery compartment

1.
2.
3.
4.
5.

Release the screws (item 1).
Remove the cover of the battery compartment (item 2).
Remove the old batteries and disconnect them.
Connect the new batteries. Make sure the polarity is correct.
Place the new batteries into the battery compartment and replace the battery
compartment cover.

a0014207

Batteries must be replaced every 3 years by the following type of battery:Panasonic LC-R127R2PG1.

Endress+Hauser

53

Maintenance

2.5

Liquiport 2010 CSP44

Calibration

Distribution arm
The position of the distribution arm is set at the factory.
Sample volume
The sample volume of the peristaltic pump is also calibrated at the factory.
For version with sensors with the Memosens protocol:
Users must decide whether the process conditions present require calibration during initialcommissioning.
Additional calibration is not required in many standard applications.
Sensors with Memosens protocol are calibrated at the factory.
Calibrate the sensors at sensible intervals depending on the process.
All information on calibration is provided in BA00493C "Calibration".

2.6

Simulation

You can simulate values at inputs and outputs for testing purposes:• Current values at current outputs• Measured values at inputs
Only current values are simulated. It is not possible to use the simulation function to
calculate the totalized value for the flow or rainfall.
The inputs and outputs must be activated in the Setup menu prior to simulation.
Diagnostics/Simulation
Function

Options

Current output x:y
Simulation

Info
Simulation of an output current
This menu appears once for each current output.

Options• On• Off

If you simulate the value at the current output, this isindicated on the display by a simulation icon in front of
the current value.

Factory setting
Off
Current

2.4 to 23.0 mA

Set the desired simulation value.

Factory setting
4 mA

54

Endress+Hauser

Liquiport 2010 CSP44

MaintenanceDiagnostics/Simulation
Function

Options

Alarm relay
Relay x:y
Simulation

Info
Simulation of a relay stateThis menu appears once for each relay.

Options• On• Off

If you simulate the relay state, this is indicated on thedisplay by a simulation icon in front of the relay display.

Factory setting
Off
State

Options• Low• HighFactory setting
Low

Meas. inputs

Simulation of a measured value
This menu appears once for each measuring input.

Channel : parameterSim. main value

Set the desired state.
The relay switches in accordance with your setting when
you switch on the simulation. The display shows "On"(="Low") or "Off" (="High") for the simulated relay state.

Options• On• Off

If you simulate the measured value, this is indicated on
the display by a simulation icon in front of the measured
value.

Factory setting
Off
Main value

Depends on the sensor

Set the desired simulation value.

Factory setting
Depends on the sensor
Sim. temperature

Options• On• Off

If you simulate the temperature measured value, this isindicated on the display by a simulation icon in front of
the temperature.

Factory setting
Off
Temperature-50.0 to +250.0 ˚C(-58.0 to 482.0 ˚F)

Set the desired simulation value.

Factory setting
20.0 ˚C (68.0 ˚F)Endress+Hauser

55

Repair

Liquiport 2010 CSP44

3

Repair

3.1

Spare parts
Contact your Endress+Hauser Service if you have any questions regarding spare parts.

1
2
3

8

4
5
6

7

a0014211

Fig. 7: Spare parts

Item No.

Designation and contents

Order number
Spare parts kit

1

Lid for housing upper complete

71119023

2

Rechargeable battery pack71119018

3

Housing cover with display CM44

71119035

Pump tubing, 2 pcs.

71114701

Pump tubing, 25 pcs.

71114702

Peristaltic pump: pump housing

71119029

Lockable latch with keys

71119017

Keys

71119017

7

Housing base71119022

8

Peristaltic pump: pump head

71119008

4
5
6

56

Endress+Hauser

Liquiport 2010 CSP44

Repair910a0014212

Fig. 8: Spare parts

Item No.

Designation and contents

Order number
Spare parts kit

9Peristaltic pump: pump motor

71119030

10

Downholder complete

71119013

Distribution arm

71119007

Distribution arm

71119025

Seal set for peristaltic pump:O-ring ID=12.42 W=1.78 OD=15.98 EPDM, 2 pcs
O-ring ID=20.92 W=2.62 OD=25.53 EPDM, 2 pcs
O-ring ID=13.00 W=4.00 OD=21.00 NBR, 1 pc

71110928

3.2

ReturnThe sampler is repaired on site.
Contact your Endress+Hauser Service Department.

3.3

Disposal

The device contains electronic components and must therefore be disposed of in accordancewith regulations on the disposal of electronic waste.
Please comply with local regulations and guidelines.
You can return the 12V storage batteries for disposal.

Endress+Hauser

57

Accessories

4

Liquiport 2010 CSP44

Accessories
The most important accessories available at the time this document went to print are listed
below.
Contact your Service Department or sales center for accessories that are not listed here.

4.1

Accessories for Liquiport 2010 CSP44

71111864
71111866
71111867
71111868
71111870
71112221
71111178
71111176
71111874
71111878
71111880
71111233
71111234
71111482
71111484
71111184
71114701
71114702
71111881
71111872
71111882
71111883
71111884
51516983
71129799
71111879

4.2

Base, complemented
CSP44 base + 1 x 20 liter (5.28 US gal.), PE
CSP44 base + 12 x 2 liter (0.53 US gal.), PE
CSP44 base + 12 x 0.7 liter (0.18 US gal.), glass
CSP44 base + 24 x 1 liter (0.26 US gal.), PE
CSP44 base + 12 x 1 liter (0.26 US gal.) + 6 x 2 liter (0.53 US gal.), PE
Bottles + caps
20 liter (5.28 US gal.) PE + cap, 1 pc.
2 liter (0.53 US gal.) PE wedge shaped bottle + cap, 12 pcs.
1 liter (0.26 US gal.) PE wedge shaped bottle + cap, 24 pcs.
0.7 liter (0.18 US gal.) glass + cap, 12 pcs.
Accessories base
Kit CSP44 base cover, transporting
Kit CSP44 freezer cartridge
Suction line
Suction line ID 10 mm (3/8"), reinforced fabric, PVC, clear, length 10 m (33 ft), suction head V4A
Suction line ID 10 mm (3/8"), EPDM, black, length 10 m (33 ft), suction head V4A... m; suction line coil ID 10 mm (3/8"), PVC... m; suction line coil ID 10 mm (3/8"), EPDM
Strainer V4A 316(x) for ID 10 mm (3/8"), 1 pc.
Tubing customized
Pump tubing, 2 pcs.
Pump tubing, 25 pcs.
Installation
CSP44 suspension kit, for use in 500 to 600 mm diameter manhole
Power supply
Lead-acid battery 24 V DC
Kit CSP44 charger adapter cable, battery for power unit
Kit CSP44 power unit/charger for indoor use, 100 to 120/200 to 240 V AC ±10 %, 50/60 Hz
Kit CSP44 power unit/charger for outdoor use, IP 65, 100 to 120/200 to 240 V AC ±10 %, 50/60 Hz
Communication; software
Commubox FXA291 + FieldCare Device Setup
Field Data Manager software; 1 license
Retrofit kits
Kit CSP44 retrofit distribution system (distribution arm, distribution drive)

Measuring cable

Memosens data cable CYK10• For digital sensors with Memosens technology:
pH, ORP, oxygen (amperometric), chlorine, conductivity (conductive)• Order as per product structure (--> Online Configurator, www.products.endress.com/cyk10)

58

Endress+Hauser

Liquiport 2010 CSP44

Accessories

Measuring cable CYK81• Unterminated cable for extending sensor cables (e.g. Memosens)• 2 x 2 cores, twisted with shielding and PVC sheath (2 x 2 x 0.5 mm2 + shielding)• Material sold by the meter, Order No.: 51502543

4.3

Sensors
You can only connect sensors with M12 plug.

4.3.1 Glass electrodes
Orbisint CPS11D• pH sensor with Memosens technology• Dirt-repellent PTFE junction• Order as per product structure (--> Online Configurator, www.products.endress.com/cps11d)• Technical Information TI028C/07/EN
Ceraliquid CPS41D• pH sensor with Memosens technology• Ceramic junction and KCl liquid electrolyte• Order as per product structure (--> Online Configurator, www.products.endress.com/cps41d)• Technical Information TI079C/07/EN
Ceragel CPS71D• pH sensor with Memosens technology• Twin-chamber reference system and integrated bridge electrolyte• Order as per product structure (--> Online Configurator, www.products.endress.com/cps71d)• Technical Information TI245C/07/EN
Orbipore CPS91D• pH sensor with Memosens technology• Open aperture junction for media with high potential for fouling• Order as per product structure (--> Online Configurator, www.products.endress.com/cps91d)• Technical Information TI375C/07/EN
Orbipac CPF81D• pH compact sensor for installation or immersion operation in industrial water and wastewater• Order as per product structure (--> Online Configurator, www.products.endress.com/cpf81d)• Technical Information TI191C/07/EN
4.3.2 Pfaudler sensors
Ceramax CPS341D• Electrode with pH sensitive enamel• Meets highest demands of measuring accuracy, pressure, temperature, sterility and durability• Order as per product structure (--> Online Configurator,
www.products.endress.com/cps341d)• Technical Information TI468C/07/EN

Endress+Hauser

59

Accessories

Liquiport 2010 CSP44

4.3.3 ORP sensors
Orbisint CPS12D• ORP sensor with Memosens technology• Dirt-repellent PTFE junction;• Order as per product structure (--> Online Configurator, www.products.endress.com/cps12d)• Technical Information TI367C/07/EN
Ceraliquid CPS42D• ORP sensor with Memosens technology• Ceramic junction and KCl liquid electrolyte• Order as per product structure (--> Online Configurator, www.products.endress.com/cps42d)• Technical Information TI373C/07/EN
Ceragel CPS72D• ORP sensor with Memosens technology• Twin-chamber reference system and integrated bridge electrolyte• Order as per product structure (--> Online Configurator, www.products.endress.com/cps72d)• Technical Information TI374C/07/EN
Orbipac CPF82D• ORP compact sensor for installation or immersion operation in industrial water andwastewater• Order as per product structure (--> Online Configurator, www.products.endress.com/cpf82d)• Technical Information TI191C/07/EN
Orbipore CPS92D• ORP sensor with Memosens technology• Open aperture junction for media with high potential for fouling• Order as per product structure (--> Online Configurator, www.products.endress.com/cps92d)• Technical Information TI435C/07/EN
4.3.4 pH-ISFET sensors
Tophit CPS471D• Sterilizable and autoclavable ISFET sensor with Memosens technology• For food and pharmaceutical industry, process engineering, water treatment andbiotechnology• Order as per product structure (--> Online Configurator,
www.products.endress.com/cps471d)• Technical Information TI283C/07/EN
Tophit CPS441D• Sterilizable ISFET sensor with Memosens technology• For low-conductivity media, with liquid KCl- electrolyte• Order as per product structure (--> Online Configurator,
www.products.endress.com/cps441d)• Technical Information TI352C/07/EN

60

Endress+Hauser

Liquiport 2010 CSP44

Accessories

Tophit CPS491D• ISFET sensor with Memosens technology• Open aperture junction for media with high potential for fouling• Order as per product structure (--> Online Configurator,
www.products.endress.com/cps491d)• Technical Information TI377C/07/EN
4.3.5 Inductive conductivity sensors
Indumax CLS50D• Inductive conductivity sensor with very good resistance properties for standard, Ex andhigh-temperature applications• Memosens protocol• Order as per product structure (--> Online Configurator, www.products.endress.com/cls50d)• Technical Information TI182C/07/EN
4.3.6 Conductive conductivity sensors
Condumax CLS15D• Conductive conductivity sensor for pure water, ultrapure water and applications in hazardous
areas• Order as per product structure (--> Online Configurator, www.products.endress.com/cls15d)• Technical Information TI109C/07/EN
Condumax H CLS16D• Hygienic, conductive conductivity sensor for pure water, ultrapure water and applications inhazardous areas• With EHEDG and 3A approval• Order depending on version, see Technical Information TI227C/07/en
Condumax W CLS21D‣ Two-electrode sensor in plug-in head and fixed cable version‣ Order as per product structure, see Technical Information TI085C/07/en
4.3.7 Oxygen sensors
Oxymax COS51D• Amperometric sensor for dissolved oxygen, with Memosens technology• Order as per product structure (--> Online Configurator, www.products.endress.com/cos51d)• Technical Information TI413C/07/EN
Oxymax COS61D• Optical oxygen sensor for drinking water and industrial water measurement• Measuring principle: quenching• Memosens protocol• Material: stainless steel 1.4571 (AISI 316Ti)• Order as per product structure (--> Online Configurator, www.products.endress.com/cos61d)• Technical Information TI387C/07/EN

Endress+Hauser

61

Accessories

Liquiport 2010 CSP44

4.3.8 Ion-selective sensors
ISEmax CAS40D• Ion-selective sensors• Order as per product structure (--> Online Configurator, www.products.endress.com/cas40d)• Technical Information TI491C/07/EN
4.3.9 Turbidity sensors
Turbimax CUS51D• For nephelometric turbidity and solids measurement in wastewater• 4-beam alternating light method based on scattered light• With Memosens protocol• Order as per product structure (--> Online Configurator, www.products.endress.com/cus51d)• Technical Information TI461C/07/EN
4.3.10 SAC and nitrate sensors
Viomax CAS51D• SAC and nitrate measurement in drinking water and wastewater• With Memosens protocol• Order as per product structure (--> Online Configurator, www.products.endress.com/cas51d)• Technical Information TI459C/07/EN

62

Endress+Hauser

Liquiport 2010 CSP44

Index

A

M

Accessories
Measuring cable . . . . . . . . . . . . . . . . . . . . . . 58
Sensors . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Adjusting diagnostic information . . . . . . . . . . 11

Maintenance . . . . . . . . . . . . . . . . . . . . . . . . . . 47
Manual hold . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Measuring cable . . . . . . . . . . . . . . . . . . . . . . . 58

B

Process error . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
pH/ORP . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
turbidity and nitrate . . . . . . . . . . . . . . . . . . . . 9
Process errors
Conductivity . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Oxygen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7
Program logbook. . . . . . . . . . . . . . . . . . . . . . . 38

Bottle statistics . . . . . . . . . . . . . . . . . . . . . . . . . 40

C
Calibration . . . . . . . . . . . . . . . . . . . . . . . . . . . . 54
Cleaning
Assemblies . . . . . . . . . . . . . . . . . . . . . . . . . . 52
Digital sensors . . . . . . . . . . . . . . . . . . . . . . . 52
Housing . . . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Sample compartment . . . . . . . . . . . . . . . . . . 52
Wetted parts . . . . . . . . . . . . . . . . . . . . . . . . 50
Cleaning assemblies . . . . . . . . . . . . . . . . . . . . . 52
Cleaning the housing . . . . . . . . . . . . . . . . . . . . 50

D
Device behavior . . . . . . . . . . . . . . . . . . . . . . . . 12
Device information. . . . . . . . . . . . . . . . . . . . . . 41
Device-specific errors . . . . . . . . . . . . . . . . . . . . 10
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Adjusting device behavior . . . . . . . . . . . . . . 12
Device-specific diagnostics messages . . . . . 14
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . 33
Logbooks . . . . . . . . . . . . . . . . . . . . . . . . . . . 34
Manual hold . . . . . . . . . . . . . . . . . . . . . . . . . 45
Message classification . . . . . . . . . . . . . . . . . 11
Runtime info. . . . . . . . . . . . . . . . . . . . . . . . . 44
Sensor replacement . . . . . . . . . . . . . . . . . . . 45
System information . . . . . . . . . . . . . . . . . . . 41
Troubleshooting settings . . . . . . . . . . . . . . . 31
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 57

E
Error categories . . . . . . . . . . . . . . . . . . . . . . . . 11

L
Logbooks . . . . . . . . . . . . . . . . . . . . . . . . . . 34–37

Endress+Hauser

P

R
Recommended maintenance . . . . . . . . . . . . .
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Replacing the pump tube . . . . . . . . . . . . . . . .
Replacing the storage batteries . . . . . . . . . . .
Reset . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Resetting the measuring device . . . . . . . . . . .
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

47
56
48
53
42
42
57

S
Sensor information . . . . . . . . . . . . . . . . . . . . .
Sensor replacement. . . . . . . . . . . . . . . . . . . . .
Sensors . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Spare parts . . . . . . . . . . . . . . . . . . . . . . . . . . .
System information . . . . . . . . . . . . . . . . . . . .

42
45
59
56
41

T
Troubleshooting . . . . . . . . . . . . . . . . . . . . . . . . . 4

www.addresses.endress.com