SHF Communication Technologies AGWilhelm-von-Siemens-Str. 23 • Aufgang D • 12277 Berlin – Marienfelde • Germany
Phone ++49 30 / 772 05 10 • Fax ++49 30 / 753 10 78
E-Mail: sales@shf.biz • Web: http://www.shf.biz

Application Note
AN-BT65-1
Impact of Bias Tees on
Communication Signals

Resonance free transmission performance from 20 kHz to over 65 GHz
Innovative construction – Patent pending

Selecting a bias tee for your application looks pretty straightforward, but watch out - sometimes
important data are not given to enable an appropriate decision.
Here is a summary of the specifications you should consider:
Bandwidth and frequency response
It is obvious that a low insertion loss, the amount of bandwidth and the frequency response of S21 areimportant. If you want to transmit 44 Gbps signals, a flat frequency response and a bandwidth of 65
GHz are required to transmit all relevant spectral components of the signal. The limiting factor in thebandwidth of the SHF BT65 is the performance of the V connectors. Figure 1 shows a sharp spike at~71GHz, which corresponds to the moding frequency of the V connectors. Aside from this, theperformance is very good to over 100GHz.

Fig.1:
Frequency Response ofan SHF BT65 Bias Tee
Measured with:
Agilent 8510XF NetworkAnalyzer

Fig. 1 shows a typical frequency response of a 65 GHz SHF bias tee. Note that the data also includes
additional attenuation due to 1mm to 1.85mm adaptors between the bias tee and VNA.
As pseudo random bit streams show a spectrum with a [sin (x)/x]2 envelope which tolerates theomission of spectral lines close to DC only to a limited extent. It is therefore important to have a lowfrequency cut off and a linear transmission of the low frequency spectral components as well (Fig. 2).
0Fig. 2:Frequency Response ofan SHF BT65 Bias Teeat low Frequencies

S21 [dB]-1-2

Measured with:
Agilent 4395A Network
Analyzer

-3-4

-5
0.01

0.05

0.10

0.50

1.00

5.00 10.00

Frequency [MHz]Linear phase and group delay
When pulses or communication signals are to be transmitted, not only the amplitude of the transfer
function but also its phase is quite important. If the phase of the bias tee is not linear, distortions willoccur.
SHF reserves the right to change specifications and design without notice – AN-BT65-1 – Rev. 1.1 – 9/FEB/2004

Page 2/7

Fig. 3 explains this effect: A square wave (simulated up to its 5th harmonic) is shown after thetransmission through a network with a linear and a network with a non-linear phase response.

Fig. 3: Influence of a Non-Linear Phase Response
At higher frequencies, the amplitude slope of even small components might be very steep and thus anon-linearity would be difficult to see. Because of this, another specification is more commonly used:group delay.
The group delay is the derivative of the phase versus frequency. More mathematically,tg = −dϕ∆ϕ≈−dω
2π∆f.

Modern network analyzers determine the group delay by measuring the phase at discrete frequencies
and calculating the phase difference at two different frequencies in their CPU.
The step value ∆f is called the aperture and can be selected by the operator of the network analyzer –creating a new problem: Without specifying the aperture, the group delay specification is meaningless.
If two group delay measurements are to be compared they have to be made with the same aperture
because larger aperture values tend to “average” the ripple, making it appear as though the groupdelay is smaller than it really is.
Fig. 4 shows a group delay measurement from one of our bias tees compared with the bias tee ofanother vendor:
200

Fig.4:
Group Delay

100

SHFBT65
Other Vendor

GD [ps]0-100

Measured with:
Anritsu 37397C NetworkAnalyzer-200

Aperture: 40MHz-300
0.0

0.5

1.0

1.5

2.0

2.5

3.0

3.5

4.0

4.5

5.0

Frequency [GHz]SHF reserves the right to change specifications and design without notice – AN-BT65-1 – Rev. 1.1 – 9/FEB/2004

Page 3/7

Clearly, the group delay of the SHF bias tee is flatter at low frequencies. The important point to notehere is that the data sheet of the other vendor displays a group delay measurement with a variation of
only ± 2ps. This measurement is not wrong but an aperture of around 500 MHz was used. Our
measurement shown above was taken with an aperture of 40 MHz. If a bigger aperture is chosen theripple will be smoothed out and cannot be detected any more.
To determine whether a given group delay ripple will have any impact on your signal, you have to lookat the eye diagram. Fig 5 shows the results. As you can see, a group delay ripple causes abroadening of top and base line.
Electrical input signal at 2.5 Gbps
Generated by SHF BPG 44 Opt. LJ
NRZ, PRBS 223 –1,measured withDigital Communications Analyzer
Agilent 86100B / 83484A
50 cm Sucoflex 102EA + 3 dB V-Goldattenuator between the generator and thesampling head input

Other Vendor
Electrical output signal at 2.5 Gbps
measured with:
Digital Communications Analyzer
Agilent 86100B / 83484A
50 cm Sucoflex 102EA between thegenerator and the bias tee
3 dB V-Gold attenuator before the samplinghead input
Note: The low frequency group delay
ripple introduces line broadening.

SHF BT65
Electrical output signal at 2.5 Gbps
measured with:
Digital Communications Analyzer
Agilent 86100B / 83484A
50 cm Sucoflex 102EA between thegenerator and the bias tee
3 dB V-Gold attenuator before thesampling head input
Note: No additional broadening of top andbase line.

Fig. 5: Eye diagrams at 2.5Gbps
SHF reserves the right to change specifications and design without notice – AN-BT65-1 – Rev. 1.1 – 9/FEB/2004

Page 4/7

Electrical input signal
at 44 Gbps
Generated by SHF BPG 44 Opt LJ
NRZ, PRBS 223 –1,
measured with
Digital Communications Analyzer
Agilent 86100B / 83484A
50 cm Sucoflex 102EA + 3 dB V-Goldattenuator between the generator andthe sampling head input
Q-Factor: 16.29
Jitter: 356 fs
Eye amplitude: 626 mV
Other Vendor
Electrical output signal
at 44 Gbps
measured with
Digital Communications AnalyzerAgilent 86100B / 83484A
50 cm Sucoflex 102EA between thegenerator and the bias tee
3 dB V-Gold attenuator before thesampling head input
Q-factor: 15.06
Jitter: 444 fs
Eye amplitude: 583 mVSHF BT65 Electrical output signal
at 44 Gbps
measured with
Digital Communications Analyzer
Agilent 86100B / 83484A
50 cm Sucoflex 102EA between thegenerator and the bias tee
3 dB V-Gold attenuator before thesampling head input
Q-factor: 16.41
Jitter: 359 fs
Eye amplitude: 582 mVFig 6: Eye diagrams at 44 Gbps
A high quality bias tee shows negligible performance degradation with 44 Gbps data signals.

SHF reserves the right to change specifications and design without notice – AN-BT65-1 – Rev. 1.1 – 9/FEB/2004

Page 5/7

Influence of DC voltage and current on the low frequency cut off
It is obvious that DC parameters like maximum voltage and current are also important specifications.
Rather less known is the effect that capacitors change their value depending on the DC voltage
applied and that inductors might go into saturation according to the amount of DC current flowingthrough. This of course will change the low frequency cut-off point.
We evaluated this influence with a setup according to Fig. 7. If you also want to characterize a bias tee
versus DC voltage or current make sure that the attenuator can handle the currents involved.

VNA

Attenuator

DUT

20dB 9W
Power Supply
0…400mA

Fig 7: Characterising the DC response of abias tee

PCGPIBThe measurement reveals the behavior that the low frequency cut off increases with increasing bias
voltage and current. This surprising result shows that a bias tee that seems to be superior at 0 V bias
(normally shown in data sheets) is worse under more realistic operating conditions. Figure 8 shows thelow frequency response of bias tees with no bias applied and the performance with applied voltage.
1100

Response [dB]

Response [dB]-1
Measurement
setup:-2-1-2-3-3-4-4-5
02550

75

100

125

150

175

200

Frequency [MHz]

225

250

-502550

75

100

125

150

175

200

225

250

Frequency [KHz]

SHF BT65
Low Frequency Cut Off versus Bias@ 0mA-3dB @ 18 kHz@ 200mA -3dB @ 27 kHz@ 400mA -3dB @ 45 kHz

Other Vendor
Low Frequency Cut Off versus Bias
@ 0mA-3dB @ 15 kHz@ 200mA -3dB @ 48 kHz@ 400mA -3dB @ 85 kHz

Measured with:
HP4395A Network Analyzer,
Agilent E3646A Power Supply,
20dB/9W Power Attenuator (SHF)

Measured with:
HP4395A Network Analyzer,
Agilent E3646A Power Supply,
20dB/9W Power Attenuator (SHF)

Fig 8: Comparison of low frequency cut-off frequency with different bias voltages for the SHF BT65
and a bias tee from another manufacturer

SHF reserves the right to change specifications and design without notice – AN-BT65-1 – Rev. 1.1 – 9/FEB/2004

Page 6/7

Isolation
For some measurements (e.g. if you want to measure very small DC currents) it is important that there
is no RF signal present at the DC port. The isolation of the bias tee characterizes this. SHF bias tees
have a specified minimum isolation of 40 dB, so the maximum RF signal at the DC port would be only
one ten-thousandth of the original signal.

-50

Isolation (dB)

-60

-70

-80

-90

-100
0102030405060Frequency (GHz)Fig 9 Isolation measured between the bias port and HF-in port.
And finally…
The inductance of our bias tee at the rated current is > 0.4 mH. This is important information if youconnect a capacitive load to the DC port of the bias tee and it has no internal bypass capacitor toground. The external capacitor will create a series resonating circuit in connection with the inductanceof the bias tee. The resonant frequency is given by the formulaf Re s =12π LC.

If the resonant frequency of this circuit lies above the lower cut off frequency of the bias tee, this will
generate a narrow band notch in the transmission characteristic.
Knowing the inductance of the bias tee helps you to determine whether the series resonance might
affect your measurement and also shows you how much AC signal current will be bypassed to groundin the audio frequency range.
Conclusions
To determine which bias tee is best for your application requires consideration of a whole set ofparameters. Low frequency cut off, group delay and aperture and eye diagrams have to be comparedas well as the influence of DC voltage and current on the low frequency transmission characteristic.
For some applications, the isolation of the bias tee and the value of the internal inductor are also
important.

SHF reserves the right to change specifications and design without notice – AN-BT65-1 – Rev. 1.1 – 9/FEB/2004

Page 7/7