EA01136D/06/A2/03.20
71474676
2020-07-31

Products

Solutions

Services

Installation Instructions
Cold tap, atmospheric pressuret-mass 65I, t-mass B 150, t-mass T 150,
t-mass I 300, t-mass I 500-digital

EN Installation Instructions: EA01136D →  3
DE Einbauanleitung: EA01136D →  13

EA01136D

Cold tap, ambient pressuret-mass 65I, t-mass B 150, t- mass T 150, t-mass I 300, t-mass 500-digital
Table of contents
1

Overview of accessories . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Authorized installation personnel . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

3

Safety Instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

4Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

5Special safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

6

Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

7Installation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

8Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12

Endress+Hauser

3

EA01136D

1

Overview of accessories

The Installation Instructions apply to the following accessories:
Order structure

Device component

DK6ML-

Cold tap at ambient pressure, t-mass

The correct sensor length must be ordered depending on the measuring device.
For detailed information on the sensor length, see the Technical Information for thedevice.
Applicator: Via the Internet: https://portal.endress.com/webapp/applicator2

Authorized installation personnel

Authorization to carry out installation depends on the measuring device's approval type. The
table below shows the authorized group of people in each case.
Whoever carries out the installation has full responsibility to ensure that work is carriedout safely and to the required quality standard. He/she must also guarantee the safety ofthe device following installation.

1)

4

Measuring device approval

Personnel authorized to carry out installation 1)

Without approval

1, 2. 3

With approval (e.g. IECEx)

1, 2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Endress+Hauser

EA01136D

3

Safety Instructions• Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.
• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:• Specialized technical staff must be trained in instrument safety.
• They must be familiar with the individual operating conditions of the devices.
• In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is deenergized.
• For measuring devices intended for use in hazardous locations, please observe theguidelines in the Ex documentation (XA).
• When using measuring devices in safety-related applications in accordance with IEC 61508
or IEC 61511: following installation, re-commission the device in accordance with the
Operating Instructions. Document the installation.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Hot surfaces! Risk of injury! Before commencing work: allow the system and measuring
device to cool down to a touchable temperature.
• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removal of the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn instrument off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open housing for a brief period. Avoid the penetration of foreign bodies, moisture orcontaminants.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.

Endress+Hauser

5

EA01136D• If, during installation, spacing is reduced or the dielectric strength of the measuring device
cannot be guaranteed, perform a test on completion of the work (e.g. high-voltage test inaccordance with the manufacturer's instructions).
• Service plug:• Do not connect in explosive atmospheres.
• Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your Endress+Hauser service organization.

4

Symbols used

4.1

Safety symbols

DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

4.2
SymbolSymbols for certain types of information
Meaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

6

Series of steps

Endress+Hauser

EA01136D

5

Special safety instructions• Check the weld seams and threaded joints for leaks.
• NPT/BSP pipe thread: use suitable sealing material.
• Hot surfaces! Risk of injury! Before commencing work: allow the system and measuring
device to cool down to a touchable temperature.
• Wear protective goggles and safety gloves.
• If threads on parts of the mounting set are damaged or defective, the parts must be repairedor the defective parts must be replaced if necessary.

6

Tools list
Torque wrench (4Nm)

Sealing material for
NPT/BSP pipe
thread7

Installation

7.1

Installing the process connection
Before welding work can commence, always comply with the installation conditions
specified in the Operating Instructions for the measuring device concerned.

Endress+Hauser

7

EA01136D

Preparatory steps:
The pipe must be free from burrs.
The welding work must be carried out professionally and the welding seam must be
clean and without bulges.
The hole diameter for the cold tap should be Ø 31.0 mm ± 0.5 mm (Ø 1.22 in ± 0.019
in). The cold tap must be welded at an angle of 90° to the pipe.
If the pipe walls are thin, support brackets and a baseplate must be welded on forreinforcement to distribute the load. Otherwise, the mounting can be so unstable thatthe pipe is damaged.
1→

2

Ø 31.0 mm ± 0.5 mm(Ø 1.22 in ± 0.019 in)

90° ± 3°90° ± 3°‣ Drill or cut the hole to the correct
diameter.

‣ Align the process connection

correctly and weld it onto thepipe.

Material (process connection):
1.4404 as per EN 10272 and 316/316L as per A479

7.2

Cold tap for ambient pressure
The mounting set (DK6ML-) is a tool for installing and/or removing a t-mass insertion
device at ambient pressure.
Using an integrated ball valve, this tool makes it possible to seal off a pipe leak-tight sothat the process can be resumed.
The mounting set can be used for process pressures up to 20bar/290psi in two possible
ways: ‒ With the measuring device installed correctly. ‒ The mounting kit on its own
with the valve closed.
When installing and removing the device, the process must be stopped and the pipe
returned to ambient pressure before any work can commence.

8

Endress+Hauser

EA01136D

C

A
B

H

D

F

R

E

G

1

A

Dimensional drawing of the cold tap

Dimensions (mm/in)
A

B

C

D

E

F

G

H

R

mm

⌀ 40

G1", NPT1",
G¾", NPT¾"

90

86

25

201

⌀ 30.0

4.0

165

in

⌀ 1.6

G1", NPT1",
G¾", NPT¾"

3.5

3.4

1.0

7.9

⌀ 1.2

0.2

6.5

Endress+Hauser

9EA01136D

7.3

Determining the insertion depth for t-mass 65 I, B 150,
I 300, 500-digital
9

230
220
210

8

200
190
180

0.3 × AA = Internal pipe diameter for circular pipes
or channel height for rectangular ducts (sensor installed vertically)or channel width for rectangular ducts (sensor installed horizontally)(a = min. 80 mm or 3 in)
B = Thickness of pipe wall or channel wall
C = Height of weld-in nipple on pipe or channel including sensor pipe union and valve
unit

A

B

C

7

Determining the insertion depth for t-mass 65 I:
Insertion depth = (0.3 × A) + B + C + 2 mm
Insertion depth = (0.3 × A) + B + C + 0.079 in
Determining the insertion depth for t-mass B 150, t-mass 300, 500-digital:
Insertion depth = (0.3 × A) + B + C
Insertion depth = (0.3 × A) + B + C

10Endress+Hauser

EA01136D

7.4

Determining the insertion depth for t-mass T 150
clamping rings undergo Metalplastic deformation during the9initial installation.
As a result the insertion depth is fixed
after initial installation and theclamping rings can no longer be
replaced.

230
220
210

8

200
190
180

7

2.
C

330

‣

A

1.

B

‣ Pay attention to information on

preconditions and on determining theinsertion depth.
Check the insertion depth closely before
tightening the clamping rings.

A = Fixed insertion depth 8 mm (0.31 in)±2 mm (0.08 in)
B = Pipe wall thickness
C = Total height of cold tap on pipe including
sensor pipe union and valve unit

Determining the insertion depth before mounting for the first time:
Insertion depth = 8 + B + C -1
Checking the insertion depth after mounting:
For all nominal diameters = 8 + B + C

7.5

Installing the measuring device

LCAUTION
Danger!
At ambient pressure, the cold tap may be used for the installation and removal of devices if
pipes are unpressurized.
‣ When installed or when the valve is closed, the process pressure may not exceed 20bar/
290psi.
For detailed information on the installation of the device: see the "Installation" section of
the Operating Instructions for the device.

Endress+Hauser

11

EA01136D

8

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize the disposalof WEEE as unsorted municipal waste. Do not dispose of products bearing this marking as
unsorted municipal waste. Instead, return them to Endress+Hauser for disposal under theapplicable conditions.

12

Endress+Hauser

EA01136D

Ein- und Ausbauarmatur, Umgebungsdruck
t-mass 65I, t-mass B 150, t- mass T 150, t-mass I 300, t-mass 500-digital
Inhaltsverzeichnis
1Übersicht Zubehörteile . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

2

Einbauberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

3

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

4

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

5Spezielle Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

6

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

7

Montage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17

8

Entsorgung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21

Endress+Hauser

13

EA01136D

1Übersicht Zubehörteile

Die Einbauanleitung ist für folgende Zubehörteile gültig:
Bestellstruktur

Gerätekomponente

DK6ML-

Ein-und Ausbauarmatur bei Umgebungsdruck, t-mass

Je nach Messgerät muss die entsprechende Sensorlänge bestellt werden.
Detaillierte Angaben zur Sensorlänge: Technische Informationen zum Gerät.
Applicator: Über das Internet: https://portal.endress.com/webapp/applicator2

Einbauberechtigte Personen

Die Berechtigung zur Durchführung eines Einbaus ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die einen Einbau vornimmt, übernimmt die Verantwortung für die Sicherheit
während der Arbeiten, die Qualität der Ausführung und die Sicherheit des Geräts nachdem Einbau.

1)

3

Zulassung des Messgeräts

Einbauberechtigter Personenkreis 1)

Ohne Zulassung

1, 2. 3Mit Zulassung (z.B. IECEx)

1, 2, 3Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Sicherheitshinweise• Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:• In Gerätesicherheit ausgebildet.
• Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
• Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
14

Endress+Hauser

EA01136D• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Messgeräten für den explosionsgefährdeten Bereich: Hinweise in der ExDokumentation (XA) beachten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Einbau Neuinbetriebnahme gemäß Betriebsanleitung durchführen. Einbau
dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.
• Wenn bei den Einbauarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:• Nicht in explosionsfähiger Atmosphäre anschließen.
• Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

4

Verwendete Symbole

4.1

Warnhinweissymbole

GEFAHR

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen wird.
Endress+Hauser

15

EA01136DWARNUNG

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen kann.
VORSICHT

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu leichter oder mittelschwerer Körperverletzung führen kann.
HINWEIS

Dieser Hinweis enthält Informationen zu Vorgehensweisen und weiterführenden
Sachverhalten, die keine Körperverletzung nach sich ziehen.

4.2
SymbolSymbole für Informationstypen
Bedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

5

Handlungsschritte

Spezielle Sicherheitshinweise• Schweißnähte und Schraubverbindungen auf Leckage prüfen.
• NPT/BSP Rohrgewinde: geeignetes Dichtungsmaterial verwenden.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Schutzhandschuhe und Schutzbrille tragen.
• Defekte Gewinde an Teilen des Montagesets erfordern eine Instandsetzung der Teile, oderdie defekten Teile müssen, falls notwendig ersetzt werden.

6

Werkzeugliste
Drehmomentenschlüssel (4 Nm)

16

Dichtungsmaterial
für NPT/BSP
Rohrgewinde

Endress+Hauser

EA01136D

7Montage

7.1

Einbau Prozessanschluss
Bevor Schweissarbeiten beginnen können, müssen die Einbaubedigungen der
Betriebsanleitung des entsprechenden Messgerätes beachtet werden.

Vorbereitende Arbeiten:
Die Rohrleitung muss gratfei gemacht werden.
Die Schweissung soll fachmännisch, sauber und ohne Wülste durchgeführt werden.
Der Lochdurchmesser für die Ein-und Ausbauarmatur soll Ø 31,0 mm ± 0,5 mm (Ø 1.22
in ± 0.019 in) betragen. Die Ein-und Ausbauarmatur soll senkrecht mit einem 90°
Winkel zur Rohrleitung dicht verschweisst werden.
Bei dünnen Rohrwänden sind zur Verstärkung Haltewinkel sowie eine Grundplatteanzuschweißen, um so die Last zu verteilen. Andernfalls kann die Befestigung so instabilsein, dass die Rohrleitung beschädigt wird.
1→

2

Ø 31.0 mm ± 0.5 mm(Ø 1.22 in ± 0.019 in)

90° ± 3°90° ± 3°‣ Loch auf Durchmesser bohren
oder schneiden.

‣ Prozessanschluss korrekt

ausrichten und an das Rohranschweißen.

Material (Prozessanschluss):
1.4404 gemäß EN 10272 und 316/316L gemäß A479

Endress+Hauser

17

EA01136D

7.2

Ein-und Ausbauarmatur für UmgebungsdruckDas Montageset (DK6ML-) ist ein Werkzeug zum Ein-bzw. Ausbau eines t-mass
Einsteckgerätes bei Umgebungsdruck.
Dieses Werkzeug ermöglicht es anhand eines Kugelventils, eine Rohrleitung dicht zu
verschliessen, damit der Prozess wieder in Betrieb genommen werden kann.
Das Montageset kann bei Prozessdrücken bis zu 20bar/290psi auf zwei Arten verwendet
werden: ‒ Mit dem Messgerät in korrekt installiertem Zustand. ‒ Nur das Montageset
mit dem Ventil im geschlossenen Zustand.
Beim Ein-und Ausbau des Gerätes muss der Prozess beendet und die Rohrleitung auf
Umgebungsdruck gebracht werden, bevor die Arbeiten beginnen dürfen.

C

A
B

H

D

F

R

E

G

2

A

Maßzeichnung Ein-Ausbauarmatur

Maße (mm/in)

18

A

B

C

D

E

F

G

H

R

mm

⌀ 40

G1", NPT1",
G¾", NPT¾"

90

86

25

201

⌀ 30,0

4,0

165

in

⌀ 1,6

G1", NPT1",
G¾", NPT¾"

3,5

3,4

1,0

7,9

⌀ 1,2

0,2

6,5

Endress+Hauser

EA01136D

7.3

Bestimmung der Einstecktiefe für t-mass 65 I, B 150,
I 300, 500-digital
9

230
220
210

8

200
190
180

0.3 × AA = Rohrinnendurchmesser bei rundem Rohroder Kanalhöhe bei rechteckigem Kanal (Einbau Messaufnehmer senkrecht)oder die Kanalbreite bei rechteckigem Kanal (Einbau Messaufnehmer waagrecht)(a = min. 80 mm oder 3 in)
B = Rohrwandstärke oder Kanalwandstärke
C = Höhe des Einschweißstutzens am Rohr oder Kanal einschließlich MessaufnehmerRohrverschraubung und Ventileinheit

A

B

C

7

Bestimmung der Einstecktiefe für t-mass 65 I:
Einstecktiefe = (0,3 × A) + B + C + 2 mm
Einstecktiefe = (0,3 × A) + B + C + 0.079 in
Bestimmung der Einstecktiefe für t-mass B 150, t-mass 300, 500-digital:
Einstecktiefe = (0,3 × A) + B + C
Einstecktiefe = (0,3 × A) + B + C

Endress+Hauser

19

EA01136D

7.4

Bestimmung der Einstecktiefe für t-mass T 150
Klemmringe verformen
 Metallischesich plastisch bei der Erstmontage.
9

Damit ist die Einstecktiefe nach derErstmontage festgelegt und die
Klemmringe können nicht mehr
ausgetauscht werden.

230
220
210

8

200
190
180

7

2.
C

330

B

‣

A

1.

‣ Angaben zu den Vorbedingungen undzur Bestimmung der Einstecktiefe
beachten.
Einstecktiefe genau überprüfen, bevor
die Klemmringe festgezogen werden.

A = Fixe Einstecktiefe 8 mm (0,31 in)±2 mm (0,08 in)
B = Rohrwandstärke
C = Gesamthöhe der Ein-und Ausbauarmatur
am Rohr einschließlich MessaufnehmerRohrverschraubung und Ventileinheit

Bestimmung der Einstecktiefe vor Erstmontage:
Einstecktiefe = 8 + B + C -1
Kontrolle der Einstecktiefe nach Montage:
Für alle Nennweiten = 8 + B + C

7.5

Messgerät einbauen

LVORSICHT
Gefahr!
Die Einbauarmatur bei Umgebungsdruck darf zum Ein-und Ausbau von Geräten bei
drucklosen Rohren dienen.
‣ Im eingebauten Zustand oder mit geschlossenem Ventil darf der Prozessdruck 20bar/
290psi nicht überschreiten.
Detaillierte Angaben zum Einbau des Geräts: Kapitel "Montage", Betriebsanleitung zum
Gerät.

20

Endress+Hauser

EA01136D

8

Entsorgung

Gemäß der Richtlinie 2012/19/EG über Elektro- und Elektronik-Altgeräte (WEEE) ist das
Produkt mit dem abgebildeten Symbol gekennzeichnet, um die Entsorgung von WEEE als
unsortierten Hausmüll zu minimieren. Gekennzeichnete Produkte nicht als unsortierter
Hausmüll entsorgen, sondern zu den gültigen Bedingungen an Endress+Hauser
zurückgeben.

Endress+Hauser21

*71474676*
71474676

www.addresses.endress.com