EA00111D/06/A2/04.18
71414170
2018-10-01

Products

Solutions

Installation Instructions
Replacing small partst-mass A, B, T 150

EN Installation Instructions: EA00111D →  3
DE Einbauanleitung: EA00111D →  15

Services

EA00111D

Replacement of small partst-mass A, B, T 150
Table of contents
1

Overview of spare part sets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

3

Personnel authorized to carry out repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

5

Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

6Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

7

Replacing the cover, seal for cover, electronics module, filter, neck set screw, neckseal, sensor seal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 9

8

Replacing the compression fitting . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11

9

Installing the Threadolet . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12

10

Reassembling the measuring device . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12

Endress+Hauser

3EA00111D

1

Overview of spare part sets

The Installation Instructions apply to the following spare part sets:
Order structure

Original spare part set

Contents

71158893

Set, 10 terminal block, 8-pin

10 × 8-pin plug-in terminal

71158894

Set, aluminum cover, window, seal

1 × aluminum cover, window
1 × O-ring (88.50 × 3.53 mm / 3.5 × 0.14 in)

71158895

Set, aluminum cover, blind version,seal

1 × aluminum cover, blind version
1 × O-ring (88.50 × 3.53 mm / 3.5 × 0.14 in)

71158896

Set, housing seals

1 × O-ring (88.50 × 3.53 mm / 3.5 × 0.14 in)
1 × O-ring (36.10 × 3.53 mm / 1.4 × 0.14 in)

71158897

Set, transmitter housing parts

1 × PTFE filter (10 × 2 mm / 0.39 × 0.08 in) with sealingring
1 × pressure compensation plug
1 × neck set screw M8×1271158898

Set, sealing ring, HNBR, 3/4"

1 × HNBR sealing ring(35.0/26.7 × 2.0 mm / 1.37/1.05 × 0.08 in)

71158899

Set, sealing ring, HNBR, 1"

1 × HNBR sealing ring(42.2/33.7 × 20 mm / 1.66/1.32 × 0.08 in)

71158900

Set, sealing ring, EPDM, 3/4"

1 × EPDM sealing ring(35.0/26.7 × 2.0 mm / 1.37/1.05 × 0.08 in)

71158901

Set, sealing ring, EPDM, 1"

1 × EPDM sealing ring(42.2/33.7 × 20 mm / 1.66/1.32 × 0.08 in)

71240317

Set, compression fitting G3/4", HNBR,
PEEK

1 × pipe union G3/4"
1 × nut
1 × front clamping ring, PEEK 2.2
1 × rear clamping ring, PEEK 2.2
1 × HNBR sealing ring(35.0/26.7 × 2.0 mm / 1.37/1.05 × 0.08 in)

71240318

Set, compression fitting G3/4", EPDM, 1 × pipe union G3/4"
PEEK
1 × nut
1 × front clamping ring, PEEK 2.2
1 × rear clamping ring, PEEK 2.2
1 × EPDM sealing ring(35.0/26.7 × 2.0 mm / 1.37/1.05 × 0.08 in)

71240319

Set, compression fitting 3/4", NPT,PEEK

1 × pipe union G3/4"
1 × nut
1 × front clamping ring, PEEK 2.2
1 × rear clamping ring, PEEK 2.2

XPD0006–

Set, electronics module t-mass A, B
150

1 × electronics module

XPD0022–Set, electronics module t-mass T 150

1 × electronics module

4

Endress+Hauser

EA00111D

Order structure

Original spare part set

Contents

DK6002–TP1

Set, mounting kit t-mass T 150

1 × Threadolet, stainless steel
1 × PEEK clamping ring

DK6002–TP2Set, mounting kit t-mass T 150(1 × Threadolet Alloy C22, 1 × PEEK clamping ring

• The order number of the spare parts set (on the product label on the package) candiffer from the production number (on the label directly on the spare part)!
• You can find the order number of the relevant spare parts set by entering the
production number of the spare part in the spare parts search tool.
• We recommend that you keep the Installation Instructions and packaging together at
all times.

Endress+Hauser

5EA00111D

2

Designated use

The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type. Only original parts from Endress+Hauser may be used. Only
spare parts kits designed by Endress+Hauser for the measuring device can be used at any
time.
Inspection is performed using the W@M Device Viewer. The procedure involved is describedas follows.
A spare parts overview is located in the interior of some measuring devices. If the spareparts kit is specified on this list, no inspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

Order code

4.

=

Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for productdata and click Spare Parts.

3.

All spare parts for the measuring device aredisplayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order Thenumber of the spare parts kit corresponds to anorder number on the spare parts list.

3

Personnel authorized to carry out repairs

Authorization to carry out repairs depends on the measuring device's approval type. The tablebelow shows the authorized group of people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out
safely and to the required quality standard. He/she must also guarantee the safety of thedevice following repair.

1)

6

Measuring device approval

Personnel authorized to perform repairs 1)

Without approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Endress+Hauser

EA00111D

4

Safety instructions• Check whether the spare part matches the labeling on the measuring device as described on
the cover page.
• The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type.
Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.
• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:– Specialized technical staff must be trained in instrument safety.
– They must be familiar with the individual operating conditions of the devices.
– In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is de-energized.
• When using Ex-certified measuring devices: Open device only when in a de-energized state(allow 10 minutes to elapse after switching off the power supply) or in environments which
do not have a potentially explosive atmosphere.
• In the case of measuring devices in safety-related applications in accordance withIEC 61508 or IEC 61511: Commission in accordance with Operating Instructions after
repair. Document the repair procedure.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Hot surfaces! Risk of injury! Before commencing work: allow the system and measuring
device to cool down to a touchable temperature.
• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removing the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn the measuring device off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open the housing for a brief period. Avoid the penetration of foreign bodies, moisture
or contaminants.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.

Endress+Hauser

7EA00111D• If spacing is reduced or the dielectric strength of the measuring device cannot be
guaranteed during repair work, perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service plug:– Do not connect in explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your Endress+Hauser service organization.

5

Symbols used

5.1

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

6

Series of steps

Tools list
Pin
4 mm

8

T10

0.5 × 3.5 mm

30 mm

Endress+Hauser

EA00111D

7

Replacing the cover, seal for cover, electronics module,
filter, neck set screw, neck seal, sensor seal

1→

2→

3→

ES

C

4→

5→

1.

2.
ES
C

ESC

6→

7→

8→

2.
ES

C

T10

Endress+Hauser

1.

T10

9

EA00111D

9→

10 →

11 →

26 25

24 23

22

2 1

1.

27

26 25

24 23

22

22 1

27

2.

0.5 × 3.5

12 →

13 →

4 mm
5 Nm
(3.7 lbf ft)
14 →

15 →

45°

T10

10

Endress+Hauser

EA00111D

16 →

17 →

18

1.
1.
2.
2.

7.1

Reassembling the transmitter and sensor

Reassembly is carried out in reverse order.

8

Replacing the compression fitting

Proceed as described in Section 7 →  9 and as illustrated in the diagram below.
1

8.1

Reassembling the compression fitting

Reassembly is carried out in reverse order.

Endress+Hauser

11

EA00111D

9

Installing the Threadolet
1D

90° (± 3°)

90° (± 3°)

D = 20.0 mm ± 0.5 mm (0.79 in ± 0.02 in)
For detailed information on mounting the measuring device, see the "Installation" sectionof the Technical Information for the device.

10

Reassembling the measuring device

Reassembly is carried out in reverse order.
Note the following:

LCAUTION
Risk of injury!
Danger due to improper process sealing!
‣ Ensure that the seals are clean and undamaged.
‣ Ensure that the correct sealing material has been used (e.g. Teflon tape for NPT ¾").
‣ Secure the seals correctly.

12

Endress+Hauser

EA00111D

1.

9

1.

Ensure that the direction of the arrow on
the sensor matches the flow direction of
the medium. Ensure the insertion depth
and alignment are correct.

2.

Tighten coupling nut fingertight.

3.

Tighten coupling nut with x turns (1).

4.

Mounting for the first time:

230
220
210

8

200
190
180

7

x
2.

3.

30 mm

Tighten coupling nut with 1 ¼ turns (1).
Repeat mounting:Tighten coupling nut with 1 turn (1).

1xEngineering unit mm (in)Number of turns to tighten

vibrations can be expected, tighten
 Ifthestrong
coupling nut with 1½ turns (1) when
mounting for the first time.

Turn the connector in the clockwise direction 3 times, plug into the electronics moduleand insert the electronics module into the transmitter. This ensures that the cablecannot be damaged (Figure 1 below).
1

2.
1.

3×

When replacing the electronics module: in the "Spare part" section of the service plate forthe new electronics module, enter all the missing spare part numbers based on theelectronics module that has been removed. (Fig. 1 below).
1

Endress+Hauser

13

EA00111D

For detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

14

Endress+Hauser

EA00111D

Austausch Kleinteile
t-mass A, B, T 150
Inhaltsverzeichnis
1Übersicht Ersatzteilsets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

2

Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

3

Reparaturberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 19

5

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20

6Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20

7Austausch Deckel, Dichtung für Deckel, Elektronikmodul, Filter, Madenschraube
Hals, Halsdichtung, Dichtung Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21

8Austausch der Pressverschraubung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 23

9Einbau Anschweißstutzen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24

10

Zusammenbau Messgerät . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 24

Endress+Hauser

15

EA00111D

1Übersicht Ersatzteilsets

Die Einbauanleitung ist für folgende Ersatzteilsets gültig:
Bestellstruktur

Original Ersatzteilset

Inhalt71158893

Set 10 Klemmenblock, 8-Pol

10 × Klemme steckbar, 8-Pol

71158894

Set Deckel Alu, Schauglas, Dichtung

1 × Deckel Alu, Schauglas
1 × O-Ring (88,50 × 3,53 mm / 3.5 × 0.14 in)

71158895

Set Deckel Alu, Blindversion,
Dichtung

1 × Deckel Alu, Blindversion
1 × O-Ring (88,50 × 3,53 mm / 3.5 × 0.14 in)

71158896

Set Gehäusedichtungen

1 × O-Ring (88,50 × 3,53 mm / 3.5 × 0.14 in)
1 × O-Ring (36,10 × 3,53 mm / 1.4 × 0.14 in)

71158897

Set Gehäuseteile Messumformer

1 × Filter PTFE (10 × 2 mm / 0,39 × 0, 08 in) mit
Dichtungsring
1 × Druckausgleichstopfen
1 × Madenschraube Hals M8×1271158898

Set Dichtungsring, HNBR, 3/4"

1 × Dichtring HNBR
(35,0/26,7 × 2,0 mm / 1,37/1,05 × 0,08 in)

71158899

Set Dichtungsring, HNBR, 1"

1 × Dichtring HNBR
(42,2/33,7 × 20 mm / 1,66/1,32 × 0,08 in)

71158900

Set Dichtungsring, EPDM, 3/4"

1 × Dichtring EPDM
(35,0/26,7 × 2,0 mm / 1,37/1,05 × 0,08 in)

71158901

Set Dichtungsring, EPDM, 1"

1 × Dichtring EPDM
(42,2/33,7 × 20 mm / 1,66/1,32 × 0,08 in)

71240317

Set Pressverschraubung G3/4", HNBR, 1 × Rohrverschraubung G3/4"
PEEK
1 × Mutter
1 × Klemmring vorne PEEK 2.2
1 × Klemmring hinten PEEK 2.2
1 × Dichtring HNBR
(35,0/26,7 × 2,0 mm / 1,37/1,05 × 0,08 in)

71240318

Set Pressverschraubung G3/4", EPDM, 1 × Rohrverschraubung G3/4"
PEEK
1 × Mutter
1 × Klemmring vorne PEEK 2.2
1 × Klemmring hinten PEEK 2.2
1 × Dichtring EPDM
(35,0/26,7 × 2,0 mm / 1,37/1,05 × 0,08 in)

71240319

Set Pressverschraubung 3/4", NPT,PEEK

1 × Rohrverschraubung G3/4"
1 × Mutter
1 × Klemmring vorne PEEK 2.2
1 × Klemmring hinten PEEK 2.2

XPD0006–

Set Elektronikmodul t-mass A, B 150

1 × Elektronikmodul

XPD0022–

Set Elektronikmodul t-mass T 150

1 × Elektronikmodul

DK6002–TP1

Set Montageset t-mass T 150

1 × Anschweißstutzen rostfreier Stahl
1 × PEEK-Klemmring

DK6002–TP2

Set Montageset t-mass T 150(1 × Anschweißstutzen Alloy C22, 1 × PEEK-Klemmring

16

Endress+Hauser

EA00111D• Die Bestellnummer des Ersatzteilsets (auf dem Produktaufkleber der Verpackung)kann sich von der Produktionsnummer (auf dem Aufkleber direkt auf dem Ersatzteil)unterscheiden!
• Durch Eingabe der Produktionsnummer des Ersatzteiles im Ersatzteilfindetool kanndie Bestellnummer des ensprechenden Ersatzteilsets ermittelt werden.
• Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

Endress+Hauser

17

EA00111D

2

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

3

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

Reparaturberechtigte Personen

Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die
Sicherheit während der Arbeiten, die Qualität der Ausführung und die Sicherheit desGeräts nach der Reparatur.

1)

18

Zulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

ohne Zulassung

1, 2, 3

mit Zulassung (z.B. IECEx)

1, 2, 3Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Endress+Hauser

EA00111D

4

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosem Zustand (nach Berücksichtigung
einer Wartezeit von 10 Minuten nach Abschalten der Energiezufuhr) oder in Umgebungenöffnen, die keine explosionsfähige Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.

Endress+Hauser

19

EA00111D• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

5

Verwendete Symbole

5.1

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

6

Handlungsschritte

Werkzeugliste
Stift
4 mm

20

T10

0,5 × 3,5 mm

30 mm

Endress+Hauser

EA00111D

7

Austausch Deckel, Dichtung für Deckel,
Elektronikmodul, Filter, Madenschraube Hals,
Halsdichtung, Dichtung Sensor

1→

2→

3→

ES

C

4→

5→

1.

2.
ES
C

ESC

6→

7→

8→

2.
ES

C

T10

Endress+Hauser

1.

T10

21

EA00111D

9→

10 →

11 →

26 25

24 23

22

2 1

1.

27

26 25

24 23

22

22 1

27

2.

0.5 × 3.5

12 →

13 →

4 mm
5 Nm
(3.7 lbf ft)
14 →

15 →

45°

T10

22

Endress+Hauser

EA00111D

16 →

17 →

18

1.
1.
2.
2.

7.1

Zusammenbau Messumformer und Messaufnehmer

Der Zusammenbau erfolgt in umgekehrter Reihenfolge.

8Austausch der Pressverschraubung

Vorgehen wie in Kap. 7. →  21 und wie im Bild unten.
1

8.1

Zusammenbau Pressverschraubung

Der Zusammenbau erfolgt in umgekehrter Reihenfolge.

Endress+Hauser

23

EA00111D

9

Einbau Anschweißstutzen
1D

90° (± 3°)

90° (± 3°)

D = 20,0 mm ± 0,5 mm (0,79 in ± 0,02 in)
Detaillierte Angaben zur Montage des Messgerätes: Kapitel "Montage", Technische
Information zum Gerät.

10

Zusammenbau Messgerät

Der Zusammenbau erfolgt in umgekehrter Reihenfolge.
Folgendes ist zu beachten:

LVORSICHT
Verletzungsgefahr!
Gefahr durch mangelnde Prozessdichtheit!
‣ Darauf achten, dass die Dichtungen unbeschädigt und sauber sind.
‣ Darauf achten, dass korrektes Dichtmaterial benutzt wird (z.B. Teflonband bei NPT ¾").
‣ Dichtungen korrekt befestigen.

24

Endress+Hauser

EA00111D

1.

9

1.

Sicherstellen, dass die Pfeilrichtung auf
dem Messaufnehmer mit derDurchflussrichtung des Messstoffsübereinstimmt. Korrekte Einstecktiefe und
Ausrichtung sicherstellen.

2.

Überwurfmutter fingerfest anziehen.

3.

Überwurfmutter mit x Umdrehungenanziehen (1).

4.

Erstmontage:

230
220
210

8

200
190
180

7

x
2.

2
x

3.

Maßeinheit mm (in)Anzahl Umdrehungen für das Anziehen

30 mmÜberwurfmutter mit 1 ¼ Umdrehungen (1)anziehen.
Wiederholmontage:
Überwurfmutter mit 1 Umdrehung (1) anziehen.
mit starken Vibrationen zu rechnen
 Wennist, dann bei der Erstmontage dieÜberwurfmutter mit 1½ Umdrehungen (1)anziehen.

Den Stecker 3 × im Uhrzeigersinn drehen, in das Elektronikmodul einstecken und das
Elektronikmodul in den Messumformer einsetzen. Auf diese Weise kann das Kabel nichtbeschädigt werden (Bild 1 unten).
1

2.
1.

Endress+Hauser

3×25

EA00111D

Beim Austausch Elektronikmodul: In der Rubrik Spare part auf dem Serviceschild desneuen Elektronikmoduls alle fehlenden Ersatzteilnummern auf Basis des ausgebauten
Elektronikmoduls übertragen. (Bild 1 unten).
1

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

26

Endress+Hauser

*71414170*
71414170

www.addresses.endress.com