Install Manual

ApisteFSV-1100/1200
Starter Kit for LabVIEW

VI Reference

株式会社イー・アイ・ソル

初 版：2011 年 9 月 14 日
このマニュアルは Microsoft WindowsXP 上で Microsoft Word 2003 を使用し作成したものです。

版権
本書の内容の一部、または全部を株式会社イー・アイ・ソルの許可なく変更することを禁じます。

ライセンス・商標について
Windows および Word は米国 Microsoft Corporation の登録商標です。
LabVIEW , NI-Vision , NI-IMAQ は、National Instruments の登録商標です。
その他、一般に会社名、製品名は各社の商標または登録商標です。

目次
概要説明

５

VI 一覧

５

各VIの説明７

Initialize および Close

７

Initialize ...................................................................................................................... ７
Close ............................................................................................................................ ８

Acquisition

８Snap............................................................................................................................. ８Start............................................................................................................................. ９Stop.............................................................................................................................. ９
Get Data ...................................................................................................................... １０

Utility１２

Temperature ................................................................................................................ １２
ColorBar ...................................................................................................................... １４
Read FSX..................................................................................................................... １５Control

１６Scale............................................................................................................................. １６Htemp .......................................................................................................................... １６Ltemp .......................................................................................................................... １７
Color ............................................................................................................................ １７Zoom ............................................................................................................................ １７Iso ................................................................................................................................ １８Isotemp ........................................................................................................................ １８Trig ...............................................................................................................................１８Areasize........................................................................................................................１９Areaview ......................................................................................................................１９Valfocus ........................................................................................................................２０Valhold .........................................................................................................................２０Anaarea........................................................................................................................２１Anahold........................................................................................................................２１Anaave .........................................................................................................................２２
Anaview .......................................................................................................................２２Digitarg ........................................................................................................................２３Digicomp ......................................................................................................................２３
Digihys .........................................................................................................................２４
Display .........................................................................................................................２４
Sto (FSV-1200 Only) ....................................................................................................２４
Correctmd ....................................................................................................................２５Range ...........................................................................................................................２５Lens (FSV-1200 Only) .................................................................................................２５Curing (FSV-1200 Only)..............................................................................................２６Emis .............................................................................................................................２６Airtemp ........................................................................................................................２６Unit ..............................................................................................................................２７Save (Write Only) ........................................................................................................２７Correct (Write Only)....................................................................................................２７Freez ............................................................................................................................２８Getline (Read Only).....................................................................................................２８Getpos (Read Only)......................................................................................................２８Getarea (Read Only)....................................................................................................２９Getareax (Read Only)..................................................................................................２９Getlensnum (FSV-1200 Only) (Read Only) ................................................................３０
Getlens (FSV-1200 Only) (Read Only)........................................................................３０
Getcuringnum (FSV-1200 Only) (Read Only).............................................................３０
Getcuring (FSV-1200 Only) (Read Only) ....................................................................３１
Comparater..................................................................................................................３１Getana (Read Only).....................................................................................................３２Reset (Write Only) .......................................................................................................３２

付加情報３３

概要説明
Apiste FSV-1200 Starter Kit for LabVIEW はアピステ社製 赤外線サーモグラフィ
FSV-1100 および FSV-1200 の LabVIEW 用ドライバおよびサンプル集です。本ドライバを用いることにより、カメラの設定コマンド等の書式に煩わされることなく、アプリケーションを構築できます。

VI 一覧

VI 名

内容ページ

VI Tree

本ドライバに含まれる VI がダイアグラムに配置されています。

Initialize

初期化を行い、接続 ID を取得します。

7

Close

FSV-1200 操作 VI の使用を終了し、接続 ID を開放します。

7

Snap

ポーリング送信

熱画像データを 1 フレーム分取得します。

8Start指定周期送信指定周期で熱画像の測定を開始します。

9Stop

指定周期送信を停止します。

9Get Data

指定周期で測定開始後、熱画像データを 1 フレーム分取得します。

10

Utility Temperature

Digit 値を温度に変換します。

12

Utility ColorBar

強度グラフ、IMAQ Image Display 用のカラーテーブル

14

Utility Read FSX

FSX 形式ファイルの読み込み

15

Scale

温度スケール（種別）

16

Htemp

温度スケール（上限温度）

16

Ltemp

温度スケール（下限温度）

17

Color

カラーバー

17Zoomズーム

17Iso

ISOTHERM 機能（ON ／ OFF）

18

Isotemp

ISOTHERM 機能（条件温度）

18

Trigger

トリガ

18

Areasize

エリアのサイズ

19

Areaview

エリアの表示／非表示

19

Embedded & Instrumentation５

VI 名内容ページValfocus

エリア計測対象切り替え

20Valhold

エリア計測のモード切り替え

20

Anaarea

アナログ出力の演算式設定

21

Anahold

アナログ出力のモード切り替え

21

Anaaveアナログ出力の平均化設定

22

Anaview

アナログ出力の表示切り替え

22

Digitarg

判定出力の条件切り替え

23

Digicomp

判定出力の閾値

23

Digihys

判定出力のヒステリシス

24

Display

画面表示切り替え

24

Sto

出力用ストローブ信号の出力機器切り換え

24

Correctmd

自動温度補正

25

Range

温度レンジ

25

Lens

カメラレンズの変更

25

Curing

接写リングの変更

26

Emis

放射率入力

26

Airtemp

周辺温度設定

26

Unit

温度単位

27

Save

設定内容の保存27

Correct

温度補正実行

27Freez

静止画像

28

Getline

ラインの温度読み出し

28

Getpos

ポイントの温度読み出し

28

Getarea

既定エリアの計測値読み出し

29

Getareax

指定エリアの計測値読み出し

29

Getlensnum

対応カメラレンズ数読み出し

30

Getlens

対応カメラレンズ仕様の読み出し

30

Getcuringnum

対応接写リング数読み出し

30Getcuring

対応接写リング長さの読み出し

31

Comparater

判定出力の読み出し／書き出し

31Getana

アナログ出力値読み出し

32

Reset

設定値の初期化

32

Embedded & Instrumentation
６

各 VI の説明
各 VI の説明をカテゴリー別に説明します。VI Tree には、本スタータキットに含まれるドライバ VI がダイアグラムにカテゴリー別に配置されています。カテゴリーは Initialize＆Close および、以下の 3 つに分類されます。
Acquisition （データ取得 VI）

Control （制御コマンド VI）

Utility （ユーティリティ VI）

FSV-1100/1200 を LabVIEW で使用する場合、まず Initialize を行い、接続 ID を取得した後に、Control VI を使用してカメラの制御を行います。Acquisition VI を使用して任意のタイミングまたは指定の周期で熱画像データを取得します。FSV-1200 Starter
Kit for LabVIEW には、簡単なサンプル VI も含まれておりますので、サンプル VI も参考にしてください。プログラムを終了させるときには、取得した接続 ID を開放する為に、Close してください。

Initialize および Close
Initialize

IP Address

FSV-1100/1200 の IP アドレス （例:169.254.182.119）

Port

FSV-1100/1200 のポート番号 （例:5000）
IP アドレスのデフォルト値は 「169.254.182.119」
ポート番号のデフォルト値は 「5000」 です。「

Embedded & Instrumentation
７

接続 ID以降はこの接続 ID を使用してカメラを制御します。

Close

接続 ID

接続 ID を接続して、セッションを開放します

Acquisition
赤外線サーモグラフィより熱画像データを取得するために使用する、データ取得用 VIです。データの取得方法には、任意のタイミングでデータを要求するポーリング送信（Snap） と、あらかじめサーモグラフィ側に設定した送信周期毎にデータを取得する指定周期送信（Start と Get Data）の 2 種類があります。

Snap
任意のタイミングでデータを要求（ポーリング送信）するためのデータ取得用 VI です。

2DTempArray

熱画像データの 2 次元配列

付加情報 1

別表参照 （本マニュアル 33 ページ）

Amb/H/L Temp

Amb Temp 周辺温度
H /L Temp 温度スケールの上下限温度

Embedded & Instrumentation
８

Start指定の周期で熱画像データを要求（指定周期送信）する際の、送信周期の設定と測定のスタートを行います。測定スタート後は Get Data で熱画像データを取得します。

送信周期60 fps≒ 16.7 msec 毎

30 fps≒ 33.3 msec 毎

15 fps≒ 66.7 msec 毎

12 fps≒ 83.3 msec 毎

10 fps≒ 100 msec 毎

6 fps≒ 167 msec 毎

5 fps≒ 200 msec 毎

4 fps≒ 250 msec 毎

3 fps≒ 333 msec 毎

2 fps≒ 500 msec 毎

1 fps＝ 1 sec 毎

0.5 fps＝ 2 sec 毎

0.2 fps＝ 5 sec 毎

0.1 fps＝ 10 sec 毎

Trigger

トリガ自動送信 トリガ画像のみ送信します。

Stop
指定周期送信を停止します。

Embedded & Instrumentation
９Get Data
指定周期送信スタート後に熱画像データを取得します。Get Data には２つのタイプがあります。

Get Data （Temp）
熱画像データを温度データとして取得する場合にはこの VI を使用します。FSV-1200
内部で保持している周辺温度値や放射率を使用して、全画素分を温度情報に変換済
みのデータを取得します。放射率を無効にすると、FSV-1200 本体の放射率設定を無視し、放射率 1.0 で計算された温度データが出力されます。

タイムアウト ms

タイムアウト（単位はミリ秒）

Trigger Mode?

送信周期設定をトリガ自動送信にした場合は True を設定
トリガ待ち状態でタイムアウトが発生した場合、Get Data VI を抜けますが、タイムアウトエラーにはなりません。

放射率有効?

FSV-1200 内部の放射率を使用する場合は True を設定
False の場合は FSV-1200 内部の放射率設定を無視し、放射率 1.0 で計算された温度データを出力します。

2DTempArray

熱画像データの 2 次元配列

付加情報 1

別表参照 （本マニュアル 33 ページ）

付加情報 2

FSV-1200 固有の付加情報 (FSV-1100 はデータ無し)

Embedded & Instrumentation
１０

Get Data （Digit）
熱画像データを温度データ変換前のデータとして取得する場合にはこの VI を使用します。FSV-1200 内部では、この Digit 値の状態で保持されており、FSV-1200 本体コントローラー側で行われる平均などの各種演算値は、Digit 値を基に演算を行った後の値に対して温度変換を行っている為、FSV-1200 の MEASUREMENT 機能等で出力される演算値と同じ値にしたい場合は、LabVIEW 側でも Digit 値を基に演算を行ってから温度へと変換する必要があります。また、FSV-1200 本体コントローラーより出力されるビデオ画像も、この Digit 値に対してカラーマッピングを行った映像が出力される為、FSV-1200 本体コントローラーより出力されるビデオ出力映像と見た目が同じようにLabVIEW で表示したい場合も、Digit 値を使用して表示を行う必要があります。Digit
値を温度に変換するには、Utility VI の Temperature VI を使用します。

タイムアウト ms

タイムアウト（単位はミリ秒）

Trigger Mode?

送信周期設定をトリガ自動送信にした場合は True を設定
トリガ待ち状態でタイムアウトが発生した場合、Get Data VI を抜けますが、タイムアウトエラーにはなりません。

2DDigitArray

Digit 値の 2 次元配列

付加情報 1

別表参照 （本マニュアル 33 ページ）

付加情報 2

FSV-1200 固有の付加情報 (FSV-1100 はデータ無し)

Embedded & Instrumentation１１

Utility
Temperature
Digit 値を温度に変換します。Digit 値の入力に対し、スカラ値と 2DArray 値の接続が可能な多態性 VI となっています。Digit 値から温度を求める際に必要なパラメータは付加情報 1 の中に含まれています。Get Data で取得した付加情報 1 も合わせてTemperature VI に渡す必要があります。放射率の設定も可能です。

Temperature （Scalar）

Digit

Get Data (Digit) で取得した Digit 値

付加情報 1

別表参照 （本マニュアル 33 ページ）

放射率設定

無効

FSV-1200 内部の設定を無視し放射率 1.0 で温度を算出

有効（FSV-1200 設定値）
FSV-1200 内部の設定を使用し温度を算出
有効（任意指定）LabVIEW 側で設定した放射率を使用し温度を算出
放射率

放射率設定が 「有効（任意指定）」 のときに、この放射率が有効となります。範囲は 0.1〜1.0 です。

Temp

温度データHLtempDigit

温度スケールの上下限値温度に相当する Digit 値
熱画像データを画面に表示する際、強度グラフの Z スケール最大最小値や IMAQ Image Display の最大・最小値に与えます。Digit
値に対してカラーマッピングを行う際に使用します。

Embedded & Instrumentation
１２

Temperature （2DArray）

2DDigitArray

Get Data (Digit) で取得した Digit 値

付加情報 1

別表参照 （本マニュアル 33 ページ）

放射率設定

無効

FSV-1200 内部の設定を無視し放射率 1.0 で温度を算出

有効（FSV-1200 設定値）
FSV-1200 内部の設定を使用し温度を算出
有効（任意指定）LabVIEW 側で設定した放射率を使用し温度を算出
放射率

放射率設定が 「有効（任意指定）」 のときに、この放射率が有効となります。範囲は 0.1〜1.0 です。

2DTempArray

温度データ

2DDigitArray (放射率計算後)
放射率の演算を行った後の Digit 値
画面表示の為に Digit 値を利用する場合、Get Data より取得したDigit 値のままでは、放射率の設定が反映されていません。放射率を利用する場合は、設定された放射率より再計算された Digit 値を使用する必要があります。
HLtempDigit

温度スケールの上下限値温度に相当する Digit 値
熱画像データを画面に表示する際、強度グラフの Z スケール最大最小値や IMAQ Image Display の最大・最小値に与えます。Digit値に対してカラーマッピングを行う際に使用します。

Embedded & Instrumentation
１３

ColorBar
強度グラフや IMAQ Image Display に熱画像データを表示する際のカラーテーブルを出力します。カラーバーの種類は 6 種類あります。

Color bar

RAINBOW1RAINBOW2RAINBOW3HOTIRON
HOTWHITE
HOTBLACK

強度グラフに熱画像を出力する際に使用します。

Color Table

強度グラフのプロパティ：カラーテーブルに接続します。

Palette:User Palette

IMAQ Image Display に熱画像を出力する際に使用します

IMAQ Image Display のプロパティ：Palette：User Palette に接続します。その際は、プロパティ：Palette：Palette Type は「User Defined」 に設定してください。

Embedded & Instrumentation
１４

Read FSX
アピステ社製ソフトウェアで保存される熱画像ファイル FSX 形式のファイルを読み込みます。

ファイルパス

FSX ファイルのパスを指定します。
ファイルパスが空の場合はファイルダイアログが現れます。

2DTempArray

温度データ

付加情報 1

別表参照 （本マニュアル 33 ページ）

付加情報 2

FSV-1200 固有の付加情報 (FSV-1100 はデータ無し)

FSX 付加情報

FSX ファイル固有の付加情報

Embedded & Instrumentation
１５

ControlFSV-1100/1200 の制御を行う為の制御コマンドに対応した VI です。赤外線サーモグ
ラフィの設定を変更する際は制御コマンド VI を使用します。制御コマンドによるサーモ
グラフィ本体の動作については、付属の取扱説明書をご覧ください。制御コマンド VI を使用する際は「接続 ID」と、読み出しか書き込みのどちらかを「種類」に接続して使用してください。一部のコマンドは読み込みのみ、または書き込みのみのものがあります。コ
マンド別の入力値はコマンド別の説明をご覧ください。入力値と出力値は同じデータタ
イプです。

接続 ID

接続 ID

種類０：読み出し

１：書き込みScale
温度スケールを変更／確認します。

温度スケールAuto Level / Auto Scale / Manual

Htemp
温度スケールの上限温度を設定／確認します。

温度スケール上限

温度スケール上限値

Embedded & Instrumentation
１６

Ltemp
温度スケールの上限温度を設定／確認します。

温度スケール下限

温度スケール下限値

Color
カラーバーの色を変更／確認します。

カラーバー

GRAY / RAINBOW1 / RAINBOW2

Zoom
デジタルズーム倍率を変更／確認します。

ズーム

1倍 /2倍 /4倍Embedded & Instrumentation
１７IsoISOTHERM 機能を活用／確認します。

ISO THERM

OFF / ONIsotemp
ISOTHERM 機能の条件温度を変更／確認します。

温度

閾値となる温度Trig
トリガの種類を変更／確認します。

トリガ種類

ONE-SHOT / LEVEL / CONTINUE / PAUSE

Embedded & Instrumentation
１８

Areasize
処理領域の位置とサイズを変更／確認します。

エリア No

0 〜 3

エリア領域

X1
Y1
X2
Y2

Areaview
処理領域を表示／非表示にする。または設定を確認します。

エリア No0 〜 3

表示?

非表示 / 表示

Embedded & Instrumentation
１９

Valfocus
処理領域から求める計算値を変更／確認します。

エリア No

0 〜 3

計測対象

計測値 MAX / MIN / AVE / PIX
計測値 PIX 時の条件温度

Valhold
処理領域から求める計測値の計測モードを変更／確認します。

エリア No

0 〜 3

計測モード

OFF / SAMPLE / PEAK / BOTTOM / P-P / AVERAGE

Embedded & Instrumentation
２０Anaarea
アナログ出力値を求める演算式を変更／確認します。

アナログ出力 No

0 〜 3

演算式

例： 1+2Anahold
アナログ出力値の計測モードを変更／確認します。

アナログ出力 No

0 〜 3

計測モード

OFF / SAMPLE / PEAK / BOTTOM / P-P / AVERAGE

Embedded & Instrumentation
２１Anaave
アナログ出力値の平均化を行う回数を変更／確認します。

アナログ出力 No

0 〜 3

平均化回数

1 〜 5000

Anaview
アナログ出力の表示形態を変更／確認します。

アナログ出力 No

0 〜 3

表示形態

FOCUS / VOLT / OFF

Embedded & Instrumentation
２２Digitarg
判定出力を条件指定する機器を変更／確認します。

デジタル出力 No

0 〜 3

出力条件

COMP UP / COMP DOWN / PC

Digicomp
出力条件の「MODE」で設定した出力条件に対する閾値を変更／確認します。

デジタル出力 No

0 〜 3

出力条件

MODE

温度 / ピクセル

閾値

出力条件に対する閾値

Embedded & Instrumentation
２３

Digihys
出力条件の「MODE」で設定した出力条件に対するヒステリシスを変更／確認します。

デジタル出力 No

0 〜 3

出力条件

MODE

温度 / ピクセル

ヒステリシス値

出力条件に対するヒステリシス値

Display
画面の表示レイアウトを切り換え／確認します。

画面表示

UNDER SCAN / OVER SCAN / OFF

Sto (FSV-1200 Only)
出力用ストローブ信号の出力機器を変更／変更します。

出力用ストローブ信号

Embedded & Instrumentation
２４

COMP / PCCorrectmd
自動温度補正の設定を変更／確認します。サーモグラフィの機種により選択できる項
目が違うところを注意してください。

温度補正方法OFF / ON (FSV-1100)
OFF / Auto / Manual (FSV-1200)

Range
温度レンジを切り換え／確認します。

温度レンジ

LOW / HIGH / AUTO

Lens (FSV-1200 Only)
カメラレンズを変更／確認します。

カメラレンズ

Embedded & Instrumentation
２５Curing (FSV-1200 Only)
接写リングを変更／確認します。

接写リング

Emis
サーモグラフィ本体の放射率設定値を変更／確認します。

放射率0.1 〜 1.0

Airtemp
周囲温度を変更／確認します。

周辺温度

Embedded & Instrumentation
２６MODE

AUTO / MANUAL

任意値

Manual の際の周辺温度値

Unit
温度単位の設定を変更／確認します。

温度単位

℃ / Ｆ

Save (Write Only)
サーモグラフィ本体の現在の設定内容を保存します。

Correct (Write Only)
赤外線サーモグラフィの温度補正を行います。

Embedded & Instrumentation
２７Freez
撮像画像を静止／動画に切り換えます。または撮像画像の表示状態を確認します。

動作

動画 / 静止

Getline (Read Only)
指定した X 軸座標上の温度を読み出します。

Y 座標

0 〜 239

読み出し対象MAX / MIN / AVE

温度(Out)温度

Getpos (Read Only)
指定した座標の温度を読み出します。

指定位置

X 座標

温度(Out)

温度

Embedded & Instrumentation
２８

Y 座標

Getarea (Read Only)
設定された処理領域の計測値を読み出します。

エリア No0 〜 3

読み出し条件温度(Out)

読み出し対象MAX / MIN / AVE / PIX

条件温度

PIX 時の条件温度

温度 or ピクセル数

Getareax (Read Only)
指定したエリアの計測値を読み出します。

エリア領域

X1
Y1
X2
Y2

読み出し条件温度(Out)読み出し対象

MAX / MIN / AVE / PIX

条件温度

PIX 時の条件温度

温度 or ピクセル数

Embedded & Instrumentation
２９

Getlensnum (FSV-1200 Only) (Read Only)
取り付け可能なカメラレンズの種類数を読み出します。

種類数(Out)

取り付け可能なカメラレンズの数

Getlens (FSV-1200 Only) (Read Only)
使用中のカメラレンズ名と接写リングの接続を読み出します。

カメラレンズ番号
カメラレンズ名称(Out)
接写リング接続可否

Getcuringnum (FSV-1200 Only) (Read Only)
取り付け可能な接写リングの種類数を読み出します。

種類数(Out)

取り付け可能な接写リングの数

Embedded & Instrumentation
３０

Getcuring (FSV-1200 Only) (Read Only)
使用中の接写リングの長さを読み出します。

接写リング番号
接写リングの長さ(Out)Comparater
判定出力値 (デジタル OUT 0〜3） の読み出し／書き込みを行います。デジタル
OUT 0〜3 へ書き込みを行う場合は、Digitarg コマンドを使用して、判定出力の条件を指示する機器を PC に設定してください。

デジタル出力 No

0 〜 3

出力OFF / ON

Embedded & Instrumentation
３１

Getana (Read Only)
アナログ出力値を読み出します。

アナログ出力 No

0 〜 3

出力値(Out)

アナログ出力電圧値 ( 0.000 〜 5.000 )Reset (Write Only)
サーモグラフィ本体の設定を初期化します。ネットワーク設定は初期化されません。

Embedded & Instrumentation
３２

付加情報

付加情報１
項 目

内 容
Amb Temp

撮像時の、赤外線サーモグラフィの周囲温度を表します。

Shutter

温度補正の設定値を表します。（0：OFF，1：Manual，2：Auto）

Range

温度レンジの設定値を表します。（0：Low，1：High）

Brightness

赤外線サーモグラフィの輝度を表します。

Contrast

赤外線サーモグラフィのコントラストを表します。

Emissivity

測定対象物の放射率を表します。（0.1 〜 1）

Unit

温度単位を表します。（0：℃，1：°F）

Htemp

温度スケールの上限温度を表します。

Ltemp

温度スケールの下限温度を表します。
温度スケールのカラーバーを表します。

Color（0：GRAY，1：RAINBOW1，2：RAINBOW2）
Trig

トリガの入った熱画像かどうかを表します。（0：OFF，1：ON）

Amb Mode

周囲温度の設定値を表します。（0：Sensor，1：Manual）

Amb Temp Manual「Amb Mode」項目が“1：Manual” 時の、周囲温度の設定値を表します。

zArea(0〜3)Area

熱画像への Area0〜3 の表示／非表示を表します。（0：OFF，1：ON）
Area0〜3 で計測する内容を表します。

種別（0：最高温度，1：最低温度，2：平均温度，3：指定温度を超えた部分の総ピクセル数）

Start X

Area0〜3 の開始点（X 座標）を表します。

Start Y

Area0〜3 の開始点（Y 座標）を表します。

End X

Area0〜3 の終了点（X 座標）を表します。

End Y

Area0〜3 の終了点（Y 座標）を表します。

Value

撮像時の、Area0〜3 の計測値を表します。
赤外線サーモグラフィのズーム設定値（倍率）を表します。

Zoom（0：1 倍，1：2 倍，2：4 倍）

Embedded & Instrumentation
３３

Iso Therm

ISOTHERM 機能の利用状態を表します。（0：OFF，1：ON）

Iso Therm Temp

ISOTHERM 機能の閾値を表します。（ASCII 表示）

zAna Area(0〜3)
Ana Area

アナログ出力(0〜3) の演算式を表します。（22 通り)

zAna View(0〜3)アナログ出力(0〜3) の表示方法を表します。
Ana View（0：温度／ピクセル，1：電圧値（0 〜 5V）に変換，2：非表示）
Range MAX

撮像時の、温度スケールの上限温度を表します。

Range MIN

撮像時の、温度スケールの下限温度を表します。

Data Count

温度− Digit 値の計測ポイント数を表します。（最大 31）

zTempTemp

計測ポイント(0〜最大 30) の温度（℃）を表します。

Digit

計測ポイント(0〜最大 30) に対応する Digit 値を表します。

zDigitzAnalog Output(0〜3)
Analog Output

撮像時の、アナログ出力(0〜3) の出力値を表します。（0 〜 4095）

zOut(0〜3)
Out

判定結果を表します。（0：OFF，1：ON）

zOut Mode(0〜3)
判定出力（0 〜 3）の出力条件を設定する機器を表します。
Out Mode（0：COMP1，1：COMP2，2：PC）
zOut Comp(0〜3)
出力条件値を表します。「Out Mode(0〜3)」項目が
Out Comp“0：COMP1” または“1：COMP2” のときに使用されます。
赤外線サーモグラフィの撮像タイミングの設定値を表します。
Trigeer Mode
（0：PAUSE，1：ONE-SHOT，2：LEVEL，3：CONTINUE）
Error Code

エラーコードを表します。

Embedded & Instrumentation
３４