DYNAMIC
GD4200-USB
USBU
SB PRESSURE TRA
TRANSDUCER
TR N• Sample rate software selection up to 1,000 Hz• Silicon-on-Sapphire pressure sensor technology• Choice of pressure ranges from vacuum to 5,000 bar• Accuracy (NLHR) ±0.15% of span BFSL
• Auto detect and conﬁguration• USB 2.0 compatible• ESI-USB© downloadable software with auto update• Measure & record up to 16 pressure inputs together• Create customised test certiﬁcate• Automatic temperature compensation• Support for easy integration with applications
created by C#, VB, Labview and Excel VBA(api dll library)• 2m lead & carry case included
Vers. 20/1/Eng

DYNAMIC GD4200-USB

USB Pressure Transducer

Technical DataDescription
The GD4200-USB© Digital Pressure Transducer
has been designed to measure, analyse and recordpressure directly on your PC without the need forcostly I/O interface boards. It allows the user tomeasure up to 16 pressure inputs simultaneously
and easily create customised test certiﬁcates.
The transducer is USB 2.0 compatible and connects via adetachable USB lead. The data is captured by the ESI-USB©
GSR½KYVEFPI;MRHS[W -RXIVJEGI WSJX[EVI ERH TVIWIRXIH SR
WGVIIRMRVIEPXMQI[MXLSYXPSWWSJEGGYVEG]SVFERH[MHXL-X
MW GSQTEXMFPI [MXL;MRHS[W   ERH  ERH GSR½KYVIWEYXSQEXMGEPP] [MXL ]SYV HIWOXST PETXST SV XEFPIX ZME 97&TVSXSGSP -X LEW MRWXERX GSRRIGXMSR ERH [MPP EYXSQEXMGEPP]
HIXIGX [LIR E WIRWSV MW GSRRIGXIH8LI WEQTPI VEXI IREFPIWH]REQMGTVIWWYVIWXSFIQIEWYVIH[MXLYTXSFMXVIWSPYXMSREXYWIVWIPIGXEFPIWTIIHWYTXS,^(EXEGERFIHMWTPE]IH
MRKVETLMGEPSVXEFYPEVJSVQ[MXLEGLSMGISJTVIWWYVIYRMXWERHJYPP]EHNYWXEFPIWGEPIW(EXEGERFIWEZIHXSE½PISVI\TSVXIHXS)\GIP4(*ERHGWZ

8LI)7-97&WSJX[EVILEWQER]JIEXYVIWMRGPYHMRKGIVXM½GEXIKIRIVEXMSR GYWXSQMWEFPI GIVXM½GEXI XIQTPEXIW PIEO XIWXMRKQIEWYVIVIGSVHERHZMI[YTXSWIRWSVWEXXLIWEQIXMQIZMVXYEP HMJJIVIRXMEP TVIWWYVI QIEWYVIQIRX WIX EYXSQEXMG XIWXWXEVXERHWXSTXMQIWERHEPEVQPIZIPWJSVIEGLMRHMZMHYEPWIRWSV
;I EVI GSRXMRYEPP] IRLERGMRK XLI )7-97& WSJX[EVI ERH XLIEYXSYTHEXI JIEXYVI QIERW XLEX ]SY [MPP RIZIV QMWW E RI[version.
8LI YRMUYI 7MPMGSRSR7ETTLMVI WIRWSV XIGLRSPSK] TVSZMHIWSYXWXERHMRK TIVJSVQERGI ERH KMZIW I\GIPPIRX WXEFMPMX] SZIV E
[MHI XIQTIVEXYVI VERKI )\GIPPIRX QIEWYVIQIRX EGGYVEG]
TVSZMHIW LMKL VIWSPYXMSR [MXL E TVIGMWMSR KVIEXIV XLER  MR
2MRITVIWWYVIVERKIWLEZIFIIRGEVIJYPP]WIPIGXIHXS
enable the user to cover any pressure that the application
VIUYMVIWJVSQZEGYYQYTXSFEVZMEXLIYWISJXLI)7-97&HMKMXEPP]WIPJWGEPMRKWSJX[EVI)EGLYRMXVIUYMVIWJVIIHS[RPSEHSJXLI)7-97&WSJX[EVIERH
MWWYTTPMIH[MXLQ97&PIEH VEXIHXS-4 ERHEGSRZIRMIRX
carry case.

Dimensions(in mm)ESI-USB Software

Ø28~110Graph Screen

19 A/F
HEX
122
1/4" BSPM (G1/4) OR 1/4“ NPTM OR F250-C

Monitor Screen

DYNAMIC GD4200-USB

USB Pressure Transducer

Technical Data
TypeSensor Technology:

GD4200-USB

Silicon-on-Sapphire (SoS)

Output Signal:

USB 2.0 compatible

Supply Voltage:

5 VDC via USB bus

Pressure Reference:

Standard Pressure Ranges:

Standard Pressure Ranges (other):

Overpressure Safety:

Gauge (default); Absolute reference input by user

-1 to 2.5 bar; 0 – 16 bar; 0 – 100 bar; 0 – 400 bar; 0 – 1,000 bar; 0 – 1,500 bar; 0 – 2,000 bar; 0 – 4,000 bar, 0 – 5,000bar

User selectable for psi and other measurement units

2x up to 400 bar; 1.5x for 1,000 bar; 1.1x for 1,500 bar; 1.5x for 2,000 bar; 1.25x for 4,000 bar; 1.2x for 5,000 bar;

Accuracy NLHR:≤ ±0.15 % of span BFSL

Sample Rate:

User selectable to 1,000 samples per second (1,000 Hz)
Resolution: 21 bits for ≤5 Hz; 16 bits for >5 - 1,000 Hz

Operating Ambient Temperature:-20 °C to +85 °C (-4 °F to +185 °F)

Operating Media Temperature:-50 °C to +125 °C (-58 °F to +257 °F)

Storage Temperature:+5 °C to +40 °C (+41 °F to +104°F) Recommended Best PracticeTemperature Effects:±1.5 %FS total error band for -10 °C to +80 °C. Typical thermal zero and span coefficients ±0.015 %FS/ °C

Electromagnetic Compatibility:

EN61326-1, EN61326-2-3 (Laboratory equipment)

Wetted Parts:

Titanium alloy

Pressure Media:

All fluids compatible with titanium alloy

Pressure Connection:

1/4” BSP male (G1/4); 1/4” NPT male or F250-C (Autoclave)

Electrical Connection:

Mating to USB mini B socket for cable connection to PC. Supplied with 2m USB lead rated to IP68 as standard.

Software compatibility:

Windows 8, Windows 8.1 and Windows 10

Net. Weight (Kg):

0.4 Kg

DYNAMIC GD4200-USB

USB Pressure Transducer

Order Matrix

Output

Type

Electrical Connection

Dynamic (1,000 Hz)

GD4200-USB

Pressure Range

Process Connection

Electrical Connection
Mating to USB mini B socket-

Pressure Range in bar-1 to 2.5 bar

02.50-16 bar0016

0-100 bar

0100

0-400 bar

0400

0-1,000 bar

1000

0-1,500 bar

1500

0-2,000 bar

2000

0-4,000 bar

4000

0-5,000 bar

5000

Process Connection
1/4” BSP male (G1/4)

AB

1/4” NPT maleAM

Autoclave F-250-C female (for pressures above 1,500bar)

DE

Order Number Example

GD4200-USB1500ABFor options not listed please contact the sales teamDISCLAIMER : ESI Technology Ltd operates a policy of continuous product development. We reserve the right to change specification without
prior notice.All products manufactured by ESI Technology Ltd are calibrated using precision calibration equipment, traceable to national
measurement standards.

t. +44(0)1978 262 255
e. sales@esi-tec.com

www.esi-tec.com