Pressure | Temperature | Level | Force | Flow | Calibration

Standard product portfolio

Alexander Wiegand
Chairman andCEO, WIKA

About us
The WIKA Group is a global market leader in pressure and temperature measurement. The company also sets the standard in the measurement of level, force andflow, and in calibration technology.
The broad portfolio of high-precision instruments, IIoT solutions and comprehensive services makes WIKA a strong and reliable partner for all the requirements ofindustrial measurement technology.
The family-run business, founded in 1946, has a global presence with 11,200
employees. This includes our own subsidiaries, production sites and development
departments, such as the Innovation Center in Klingenberg. There alone, over
100 engineers work on smart sensing solutions that provide answers to global
challenges. WIKA’s unique experience and know-how make sensing technology
smarter, add more value and prepare it for a sustainable future:
Smart in sensing.

2

Contents

You will find more info
on our industry-specificproducts on page 118.

PRESSURE

FORCE

Display

Pressure gauges
Digital pressure gauges

412Compression force transducers

74

Transmit

Process transmitters
Pressure sensors
Pressure gauges with output signal

14
16
19

Tension/Compression force transducers

75

Bending/Shear beams

76

Load cells

77

SwitchContact pressure gauges
Pressure switches

21
23

Load pins

78

Additional
products andaccessories

Diaphragm seal systems
Valves and mounting accessories
Electrical accessories

26
27
29

Ring force transducers, tension links

79

Special force transducers

80Electronics

81

TEMPERATURE

FLOW

Display

Dial thermometers
Digital indicators

30
34

Transmit + Record

Thermocouples
Resistance thermometers
Temperature transmitters

36
42
47

Temperature switches
Thermometers with switch
contacts
Temperature controllers

48
49

Thermowells / Protection tubes
Accessories

51
53

SwitchAdditional
products andaccessories

50

Primary flow elements

82

Flow switches

89

IIOT SOLUTIONSIIoT service

90

IIoT products

92

CALIBRATION
Pressure

LEVEL
Display

Bypass level indicators
External chambers
Glass level gauges

54
57
58

Transmit

Submersible pressure sensors
Continuous measurement with
float

60
61

SwitchFloat switches
Optoelectronic switches
Vibrating level switches

66
70
73

Temperature

Digital pressure gauges
Hand-helds, calibrators
Precision pressure measuring
instruments
Pressure controllers
Pressure balances
Calibration software
Pressure generation

94
95
97
98
100
103
104

Reference thermometers
106
Hand-helds
107
Calibration baths
108
Portable temperature calibrators
109
Resistance thermometry bridges 110
Standard reference resistors, AC/DC 111

Additional products and accessories

112

Engineered solutions

113

SERVICE
Service116

Pressure ‫ ׀‬Display

Bourdon tube pressure gauges
Copper alloy

These pressure gauges are suitable for liquid and gaseous media,
so long as they are not highly viscous or crystallising and do notattack copper alloy parts. The scale ranges cover pressures from
0.6 ... 1,000 bar. These instruments are manufactured in accordance with the European standard EN837-1 (except for model
111.11 and 111.12 in NS 27).

111.10, 111.12

111.11

Nominal size

27, 40, 50, 63, 80, 100, 160 mm

Nominal size

40, 50, 63 mm

Scale range-1 … 0 to 0 … 400 bar

Scale range

0 … 0.6 to 0 … 400 bar

Accuracy class

2.5, 1.6 optional
NS 27: 4.0

Accuracy class
Data sheet

Standard version

Data sheet

Plastic case, liquid filling

Welding gauge ISO 5171

Nominal size

40, 50, 63 mm, model 111.26 also
80 mm

2.5

Scale range-1 … 0 to 0 … 400 bar

PM 01.03

Accuracy class

2.5

Data sheet

PM 01.10

PM 01.01, PM 01.17

113.13

4

Welding gauge ISO 5171

111.16, 111.26

214.11

Edgewise panel design

PG81, PG91

DirectDrive pressure gauge

Nominal size

40, 50, 63 mm

Nominal size

96 x 96, 72 x 72

Nominal size

36, 41 mm

Scale range-1 … 0 to 0 … 400 bar

Scale range■ NS 96 x 96:

Scale range

0 ... 6 to 0 ... 450 bar

Accuracy class

2.5

Accuracy class

4.0

Data sheet

PM 01.04

Data sheet

PM 01.50

0 … 0.6 to 0 … 1,000 bar■ NS 72 x 72: 0 … 0.6 to 0 … 400 bar

Accuracy class

1.6, 1.0

Data sheet

PM 02.07

PressurePressure ‫ ׀‬Display

212.20

Stainless steel case213.40

Heavy-duty version, case filling

Nominal size

100, 160 mm

Nominal size

63, 80, 100 mm

Scale range-1 ... 0 to 0 ... 1,000 bar

Scale range-1 … 0 to 0 … 1,000 bar

Accuracy class

1.0

Accuracy class

1.0, 1.6 (NS 63, 80)

Data sheet

PM 02.01

Data sheet

PM 02.06

113.53, 213.53

Stainless steel case, case filling

Nominal size

■ 113.53: 40, 80, 100 mm

Scale range-1 … 0 to 0 … 600 bar(213.53: to 1,000 bar)

Accuracy class

113.53: 1.6 (NS 80, 100), 2.5
213.53: 1.0 (NS 63, 100), 1.6 (NS 50)

Data sheet

PM 01.08, PM 02.12■ 213.53: 50, 63, 100 mm

Thermomanometers

MFTWith capillaries, forpressure and temperature measurement

THM10

Eco version, for pressure andtemperature measurement

100.02

For pressure and temperature
measurement

Nominal size

40, 42, 52 mm

Nominal size

63, 80 mm

Nominal size

63, 80 mm

Scale range■ Pressure: 0 … 4 bar

Scale range■ Pressure: 0 … 4 to 0 … 10 bar

Scale range■ Pressure: 0 … 1 to 0 … 16 bar

Accuracy class■ Pressure: 2.5 (EN 837-1)

Connection location Lower mount or back mount

Data sheet■ Temperature: 0 … 120 °C■ Temperature: 2.5■ Temperature: 0 … 100 to 0 … 150 °C

Connection location Lower mount or back mountAccuracy class■ Pressure: 2.5 (EN 837-1)

Accuracy class■ Pressure: 2.5 (EN 837-1)

Data sheet

PM 01.24

Data sheet

PM 01.23

PM 01.20

Further information at www.wika.com■ Temperature: 0 … 120 °C■ Temperature: 2 (EN 13190)■ Temperature: ±2.55Pressure ‫ ׀‬Display

Bourdon tube pressure gauges
Stainless steel
The wetted parts of these pressure gauges are manufactured
entirely from stainless steel. Thus they are suitable for gaseousand liquid aggressive media that are not highly viscous or
crystallising, also in aggressive environments. They are suitable forscale ranges from 0 ... 0.6 to 0 ... 7,000 bar.

131.11Compact version

232.50, 233.50

For the process industry,
standard version

232.30, 233.30

For the process industry,safety version

Nominal size

40, 50, 63 mm

Nominal size

63, 100, 160 mm

Nominal size

63, 100, 160 mm

Scale range■ NS 40, 50: 0 … 1 to 0… 600 bar

Scale range■ NS 63: 0 … 1 to 0 … 1,000 bar

Scale range■ NS 63: 0 … 1 to 0… 1,000 bar■ NS 63: 0 … 1 to 0 … 1,000 bar■ NS 100: 0 … 0.6 to 0 … 1,000 bar■ NS 160: 0 … 0.6 to 0 … 1,600 bar■ NS 100: 0 … 0.6 to 0 … 1,000 bar■ NS 160: 0 … 0.6 to 0 … 1,600 bar

Accuracy class

2.5

Ingress protection

IP54

Accuracy class

1.0 (NS 100, 160), 1.6 (NS 63)

Accuracy class

1.0 (NS 100, 160), 1.6 (NS 63)

Data sheet

PM 01.05

Ingress protection

IP65

Ingress protection

IP65

Data sheet

PM 02.02

Data sheet

PM 02.04

232.36, 233.36

High overload safety to 4 times thefull scale value, safety version

6

Depending on the pressure range and the instrument model,
overload safety of to a maximum of 5 x full scale value is possible.
To this point, the measurement accuracy is maintained. Liquid
filling the case ensures a precise instrument display, even withhigh dynamic pressure loads and vibrations.

232.34, 233.34

Process pressure gauge XSEL®,
safety version per ASME B40.100

Nominal size

100, 160 mm

Nominal size

4 ½", 6"

Scale range

0 … 0.6 to 0 … 40 bar

Scale range

0 … 0.6 bar to 0 … 2,000 bar

Overload safety

Up to 4 times the measuring range

Accuracy class

Grade 2A

Accuracy class

1.0

Ingress protection

IP54, with liquid filling IP65

Data sheet

PM 02.15

Data sheet

PM 02.10

Pressure ‫ ׀‬Display

Pressure

Test gauges
For highest accuracy
Depending on the instrument model, accuracies of 0.1, 0.25 or 0.6% of full scale value can be measured.
The pressure ranges cover from 0 ... 6 mbar to 0 ... 1,600 bar andare suitable for calibration tasks. For each of the pressure gauges
specified here, a DAkkS calibration certificate can be provided.

312.20

Copper alloy, class
0.6

332.50, 333.50

Stainless steel, standard
version, class 0.6

332.30, 333.30

Stainless steel, safety
version, class 0.6

Nominal size

160 mmNominal size

160 mmNominal size

160 mmScale range

0 … 0.6 to 0 … 600 bar

Scale range

0 … 0.6 to 0 … 1,600 bar

Scale range

0 … 0.6 to 0 … 1,600 bar

Accuracy class

0.6

Accuracy class

0.6

Accuracy class

0.6

Ingress protection

IP54

Ingress protection

IP65

Ingress protection

IP65

Data sheet

PM 03.01

Data sheet

PM 03.06

Data sheet

PM 03.05

342.11

From class 0.1, withtransport case andacceptance test certificate

610.20, 630.20

For low pressure ranges
from 10 mbar, class 0.6

Nominal size

250 mmNominal size

160 mmScale range

0 … 1 to 0 … 1,600 bar

Scale range

0 … 10 to 0 … 600 mbar

Accuracy class■ 0.1 for scale ranges < 400 bar

Accuracy class

0.6

Ingress protection

IP54

Data sheet

PM 06.09■ 0.25 for scale ranges ≥ 400 bar

Ingress protection

IP54

Data sheet

PM 03.03

Further information at www.wika.com

7

Pressure ‫ ׀‬Display

Diaphragm pressure gauges

The application areas for diaphragm pressure gauges are very
versatile. They are the specialists in the process industry whenit comes to critical measuring requirements such as with highly
corrosive or viscous media or when it comes to low pressures andhigh overload.
The scale ranges are from as low as 0 … 16 mbar to typically0 … 25 to 0 … 40 bar. Depending on the pressure range and the
instrument model, overload safety of 3 x or 5 x full scale value ispossible as standard.

422.12, 423.12

Grey cast iron case

For special designs, an overload safety of up to 400 bar is possible, with the measurement accuracy maintained.
Diaphragm pressure gauges are even suitable for highly viscous orcontaminated media by using an open connecting flange (per DIN/
ASME). For measuring particularly aggressive media, the complete wetted surface can be lined with a large selection of special
materials (e.g. PTFE, Hastelloy, tantalum, and many more).

432.50, 433.50,
432.30, 433.30,
452.50, 453.50,
452.30, 453.30

For the process industry, high overload
safety up to 10 times the full scale value,
max. 40 bar

8432.56, 433.56,
432.36, 433.36

For the process industry, high overload
safety to 40, 100 or 400 bar

Nominal size

100, 160 mm

Nominal size

100, 160 mm

Nominal size

100, 160 mm

Scale range

0 … 16 mbar to 0 … 40 bar

Scale range

0 … 16 mbar to 0 … 25 bar

Scale range

0 … 16 mbar to 0 … 40 bar

Accuracy class

1.6

Accuracy class

1.6

Accuracy class

1.6

Ingress protection

IP54, with liquid filling IP65

Ingress protection

IP54, with liquid filling IP65

Ingress protection

IP54, with liquid filling IP65

Data sheet

PM 04.02

Data sheet

PM 04.03

Data sheet

PM 04.07

Pressure ‫ ׀‬Display

Pressure

Capsule pressure gauges
For very low pressures
These measuring instruments are particularly suited to gaseous media. The scale ranges are between 0 … 2.5 mbar and0 … 1,000 mbar in accuracy classes from 0.1 to 2.5.
Capsule pressure gauges consist of two circular, corrugateddiaphragms, joined together around the edge with a pressure-tightseal. Overload protection is possible in certain cases.

611.10Standard version

These capsule pressure gauges are used mainly in medical,
vacuum, environmental and laboratory technology for contents
measurement and filter monitoring.

611.13612.20

Plastic case

Stainless steel case

Nominal size

50, 63 mm

Nominal size

50, 63 mm

Nominal size

63, 100, 160 mm

Scale range

0 … 25 to 0 … 600 mbar

Scale range

0 … 60 to 0 … 600 mbar

Scale range

0 … 6 to 0 … 600 mbar

Accuracy class

1.6

Accuracy class

2.5

Accuracy class

1.6

Ingress protection

IP54

Ingress protection

IP53

Ingress protection

IP54

Data sheet

PM 06.01

Data sheet

PM 06.12

Data sheet

PM 06.02

614.11, 634.11

Edgewise panel design632.50For the process industry

632.51For the process industry,high overload safety

Nominal size

72 x 72, 96 x 96, 144 x 144, 144 x 72 mm

Nominal size

63, 100, 160 mm

Nominal size

100, 160 mm

Scale range■ NS 72 x 72: 0 … 25 to 0 … 600 mbar

Scale range■ NS 63: 0 … 40 to 0 … 600 mbar

Scale range

0 … 2.5 mbar to 0 … 100 mbar

Accuracy class

1.6

Ingress protection

IP54

Data sheet

PM 06.06■ NS 96 x 96: 0 … 10 to 0 … 600 mbar

■ NS 144 x 144: 0 … 6 to 0 … 600 mbar

Accuracy class
Data sheet■ NS 144 x 72: 0 … 4 to 0 … 600 mbar

■ NS 100: 0 … 16 to 0 … 600 mbar

■ NS 160: 0 … 2.5 to 0 … 600 mbar

1.6

Accuracy class

1.6

PM 06.05

Ingress protection

IP54, with liquid filling IP65

Data sheet

PM 06.03

Further information at www.wika.com

9

Pressure ‫ ׀‬Display

Differential pressure gauges

Differential pressure gauges work with a wide range of pressure
elements. With this variety, measuring ranges from 0 ... 0.5 mbar
to 0 ... 1,000 bar and static overlay pressures up to 400 bar are
possible.

700.01, 700.02

With magnetic piston
or with magnetic pistonand separating diaphragm







the pollution degree in filter systems
the level in closed containers
the overpressure in clean rooms
the flow of gaseous and liquid media
and they control pumping plants

711.12, 731.12

With parallel entry,
copper alloy or stainless steel

DPG40
With integrated working pressure indication (DELTA-plus)

Nominal size

80 mm

Nominal size

100, 160 mm

Nominal size

100 mmScale range■ 700.01: 0 … 400 mbar to 0 … 10 bar

Scale range

0 … 0.6 to 0 … 1,000 bar

Scale range

0 … 0.16 to 0 … 10 bar

Accuracy class

1.6

Accuracy class

2.5

Ingress protection

IP33

Ingress protection

IP65

Data sheet

PM 07.02

Data sheet

PM 07.20

Accuracy class■ 700.02: 0 … 160 mbar to 0 … 2.5 bar■ 700.01: ±3 %■ 700.02: ±5 %with increasing differential pressure
Ingress protection

IP54

Data sheet

PM 07.14

716.11, 736.11

For very low differential pressuresfrom 2.5 mbar, copper alloy orstainless steel

10These measuring instruments monitor

732.51, 733.51,
732.31, 733.31

For the process industry,all-metal media chamber

732.14, 733.14,
762.14, 763.14

For the process industry, high overload
safety to 650 bar

Nominal size

100, 160 mm

Nominal size

100, 160 mm

Nominal size

100, 160 mm

Scale range■ NS 100: 0 … 10 to 0 … 250 mbar

Scale range

0 … 16 mbar to 0 … 40 bar

Scale range■ 0 ... 60 to 0 ... 250 mbar (measur-

Ambient temperature

To -70 °C■ NS 160: 0 … 2.5 to 0 … 250 mbar

ing cell DN 140)

■ 0 … 0.25 to 0 … 40 bar

Accuracy class

1.6

Ingress protection

IP66

Accuracy class

1.6

Accuracy class

1.6

Data sheet

PM 07.07

Ingress protection

IP54, with liquid filling IP65

Ingress protection

IP54, with liquid filling IP65

Data sheet

PM 07.05

Data sheet

PM 07.13(measuring cell DN 82)

Pressure ‫ ׀‬Display

Absolute pressure gauges are used when measured pressures are
independent of the natural fluctuations in atmospheric pressure.
The pressure of the measured media is determined against a reference pressure, which corresponds to the absolute pressure zero
point. For this, the reference chamber is completely evacuated, sothat there is a near-perfect vacuum in it.
Applications for these high-precision measuring instruments

Pressure

Absolute pressure gauges

are, for example, monitoring of vacuum pumps and vacuum
packaging machines. They are also used in laboratories, in orderto monitor condensation pressures or to determine the vapour
pressure of liquids.

532.52, 533.52,
532.53, 533.53,
532.54, 533.54

High overload safety
Nominal size

100, 160 mm

Scale range

0 … 25 mbar to 0 … 25 bar abs.,
high overload safety

Accuracy class

1.0 or 1.6 or 2.5

Ingress protection

IP54, with liquid filling IP65

Data sheet

PM 05.02

Further information at www.wika.com

11

Pressure ‫ ׀‬Display

Digital pressure gauges

CPG1200

Digital pressure gauge

CPG1500

Precision digital pressure gauge

Measuring range-1 ... 1,000 bar

Measuring range

Accuracy

Down to 0.25 % FS

Accuracy

Down to 0.025 % FS

Special feature■ Integrated data logger

Special feature■ Integrated data logger■ WIKA-Cal compatible

Data sheet

12

CT 10.20■ WIKA-Cal compatible■ Data transmission via Bluetooth®

■ Data transmission via USB or

Bluetooth®
■ Robust case, IP65

0 … 10,000 bar■ Password protection possible■ Robust case, IP65

Data sheet

CT 10.51

Pressure ‫ ׀‬Transmit

Pressure

Pressure sensor assemblies
and modules
Customer-specific electronic pressure measurement solutions

We see ourselves not only as a provider of top quality measurement technology, but also as a highly competent partner that is able
to create individually designed solutions together with you. In close cooperation with you, we are ready to develop products that are
tailor-made to cater for your individual needs. Create your perfect pressure sensor solution together with us. Here, the experience from amultitude of completed projects is incorporated – thus we can refer back to numerous proven solutions and components. As required, we
will adapt our systems to your individual application or develop new ones.
Talk to us – we are happy to provide you with advice!

MPR-1

Pressure sensor module

MTF-1

Pressure sensor module

Non-linearity(± % of span)≤ 0.125 or 0.25

Non-linearity(± % of span)≤ 0.125 or 0.25

Measuring range■ 0 … 0.4 to 0 … 25 bar

Measuring range■ 0 ... 10 to 0 ... 1,000 bar

Special feature■ 19 mm spanner width for limited mount-

Special feature■ Compact design

SignalData sheet■ 0 … 0.4 to 0 … 25 bar abs.

ing space■ No calibration necessary, due tocompensated output signal■ Low energy consumption■ Additional temperature indication

Analogue and digital

SignalPE 81.64

Data sheet

Further information at www.wika.com■ -1 ... +9 to -1 ... +24 bar■ Dry, welded measuring cell

Analogue and digital
PE 83.01

13

Pressure ‫ ׀‬Transmit

Process transmitters

Process transmitters differentiate themselves from pressure
sensors through their increased range of functionality: They feature
integrated displays, offer high measurement accuracies and freely
scalable measuring ranges, communicate via digital bus signals
and can be delivered with a multitude of case variants. Throughconnection to diaphragm seals, WIKA process transmitters are also
suitable for the harshest operating conditions.

Process transmitters are suitable for many industrial measuring
requirements in the widest variety of applications. They monitor
pumps, detect the level in containers or calculate quantities forflow measurement in pipelines.

®

SIL

IEC 61508 / IEC 61511

UPT-20

Universal process transmitterwith standard connection,
Ex intrinsically safe

UPT-21

Universal process transmitterwith flush process connection,
Ex intrinsically safe

DPT-EL

Electronic differential
pressure transmitter inprimary and secondary circuits

Non-linearity(% of span)≤ 0.1

Non-linearity(% of span)≤ 0.1

Non-linearity(% of span)

Output signal

4 … 20 mA, HART®

Output signal

4 … 20 mA, HART®

Output signal

Measuring range■ 0 … 0.4 to 0 … 4,000 bar

Measuring range■ 0 … 0.4 to 0 … 600 bar

4 … 20 mA, HART® protocol (optional),
PROFIBUS® PA, FOUNDATION™ Fieldbus

Measuring range■ 0 … 0.1 to 0 … 1,000 bar■ 0 … 1.6 to 0 … 40 bar abs.

■ -0.2 … +0.2 to -1 … +40 bar

Special feature■ Multi-functional display■ Freely scalable measuring range■ -0.2 … +0.2 to -1 … +40 bar

Special feature■ Simple menu navigationent designs

Special feature

hygienic applications■ Freely scalable measuring range■ Conductive plastic case or stainlesssteel case
■ Large LC display, rotatablesteel case
■ Large LC display, rotatable
PE 86.05

Data sheet■ Simple, uncomplicated installation■ Mounting possible without diaphragmseal■ Elimination of capillaries, that caneasily kink

■ For applications to SIL 2 (SIL 3)■ Can be combined with two differ-

ent designs of transmitters frommodel IPT-2x and/or model CPT-2x

PE 86.05
Data sheet

14■ 0 … 1.6 to 0 … 40 bar abs.

■ -0.05 … +0.05 to -1 … +40 bar■ Hygienic process connections in differ■ Electropolished stainless steel case for■ Conductive plastic case or stainless

Data sheet■ 0 … 1.6 to 0 … 40 bar abs.

≤ 0.05 … 0.1

PE 86.23

PressurePressure ‫ ׀‬Transmit

SIL

®

SIL

®

SIL

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

IPT-20, IPT-21

Process pressure transmitter withwelded metal measuring cell

Non-linearity(% of span)≤ 0.075 … 0.1

CPT-20, CPT-21

DPT-20

Process pressure transmitter withcapacitive ceramic measuring cell

Differential pressure transmitter,
intrinsically safe or with flameproof enclosure

Non-linearity(% of span)≤ 0.05

Non-linearity(% of span)

Output signal

4 … 20 mA, HART® protocol (optional),

Output signal

4 … 20 mA, HART® protocol (optional),

Output signal

4 … 20 mA, HART® protocol (optional),
PROFIBUS® PA, FOUNDATION™ Fieldbus

Measuring range■ 0 … 0.1 to 0 … 4,000 bar

Measuring range■ 0 … 0.025 to 0 … 100 bar abs.

Measuring range

0 … 10 mbar to 0 … 16 bar

Special feature■ Particularly robust, ceramic measuring

Special feature■ Freely scalable measuring ranges

PROFIBUS® PA, FOUNDATION™ Fieldbus■ 0 … 0.1 to 0 … 40 bar abs.
■ -1 … 0 to -1 … +40 bar

Special feature

PROFIBUS® PA, FOUNDATION™ Fieldbus

cell■ Dry ceramic measuring cell with variablesealing concept■ Freely scalable measuring ranges■ Case from plastic, aluminium or stainless steel■ Flush process connection (optional)■ Freely scalable measuring ranges■ Case from plastic, aluminium or stain-

less steel■ Flush process connection (optional)■ With integrated display and instrument
mounting bracket for wall/pipe mounting(optional)■ Process temperature ranges to 200 °C
Data sheet
Data sheet■ -1 … 0 to -1 … +100 bar

PE 86.07≤ 0.065 … 0.1■ Static load 160 bar, optionally 400 bar■ Case from plastic, aluminium or stain-

less steel■ With integrated display and instrument

mounting bracket for wall/pipe mounting(optional)■ 3- or 5-way valve optional■ SIL 2 per IEC 61508
Data sheet

PE 86.22

PE 86.06

Pressure measuring instruments with self-monitoring pressure indication®

®
TYPE EL
CLASS ITYPE EL
CLASS IFEBRUARY 2021

FEBRUARY 2021

DMS-FP

Diaphragm monitoring systemwith clamp connection

DMSU21SA

Diaphragm monitoring systemwith HART® protocol

DMSU22SAIn-line process transmitter

Non-linearity(% of span)≤ 0.1 %

Non-linearity(% of span)■ 0.1 %
■ 0.5 %

Non-linearity(% of span)

1 % (at process temperature)

Output signal■ 4 ... 20 mA

Output signal■ 4 ... 20 mA with HART® signal

Output signal■ 4 ... 20 mA with HART® signal■ 4 ... 20 mA with a superimposed

HART® communication signal(option: SIL qualification)
HART® specification:
7.3 FOUNDATIONTM Fieldbus PROFIBUS® PA

Measuring range< 40 barSpecial feature■ Double-diaphragm system to ensure

Data sheet

DS 95.20

Measuring range■ -1 ... +1.5 to -1 ... +24 bar

Special feature■ Double-diaphragm system prevents

the separation of the process and thepressure measuring instrument■ Clamp connection easy to open forcleaning and seal replacement■ Suitable for SIP and CIP

Further information at www.wika.com(HART® rev. 7)

Data sheet■ -14.5 ... 20 to -14.5 ... +350 psi

contamination of process and environment■ Hygienic process connections in different designs■ Signal transmission and configuration
with only one cable per measuringlocation■ Minimum installation costs, even withretrofitting(HART® rev. 7)■ 4 ... 20 mA■ 4 ... 20 mA

Measuring range■ 1 ... +15 bar■ 0 ... 16 bar abs.

■ 14.5 ... +200 psi

Special feature■ Dead-space free hygienic design with

Data sheet

DS 95.03

DS 95.11

thick-walled sensor tube from stainlesssteel■ In-line pressure measurement withsensor tube without system fill fluid■ Continuous sensor monitoring of the
double-tube system prevents contamination of process and environment■ Suitable for SIP and CIP■ EHEDG-certified and 3-A marked

15

Pressure ‫ ׀‬Transmit

Pressure sensors®

A-10

For general industrial applications

S-20

For demanding industrial applications

Flush diaphragmNon-linearity(± % of span)≤ 0.25 or 0.5 BFSL

Non-linearity(± % of span)≤ 0.125, 0.25 or 0.5 BFSL

Non-linearity(± % of span)≤ 0.2 BFSL

Measuring range■ 0 … 0.05 to 0 …1,000 bar

Measuring range■ 0 … 0.4 to 0 … 1,600 bar

Measuring range■ 0 … 0.1 to 0 … 600 bar■ 0 … 0.1 to 0 … 25 bar abs.
■ -0.05 … 0 to -1 … +24 bar

Special feature■ Excellent quality■ Exceptionally large variety

Data sheet■ Particularly cost-efficient

PE 81.60

IS-3

Intrinsic safety Ex ia■ 0 … 0.4 to 0 … 40 bar abs.
■ -0.4 … 0 to -1 … +59 bar

Special feature■ Availability at short notice■ Extreme variety■ High accuracy

Data sheet■ Special versions available

PE 81.61

E-10, E-11

Flameproof enclosure Ex d

■ 0 … 0.25 to 0 … 16 bar abs.
■ -0.1 … 0 to -1 … +24 bar

Special feature■ Proven technology■ Flush process connection■ Medium temperature to 150 °C■ Comprehensive stocks

Data sheet

PE 81.02

A-1200With IO-Link, PNP or
NPN switching output

Non-linearity(± % of span)≤ 0.2 BFSL

Non-linearity(± % of span)≤ 0.2 BFSL

Accuracy(± % of span)≤ 0.5 or ≤ 1

Measuring range■ 0 … 0.1 to 0 … 6,000 bar

Measuring range■ 0 … 0.4 to 0 … 1,000 bar

Measuring range■ 0 … 0.4 to 0 … 1,000 bar■ 0 … 0.25 to 0 … 25 bar abs.
■ -1 … 0 to -1 … +24 bar

Special feature■ Ignition protection type Ex ia■ Large selection of approvals

Data sheet■ 0 … 0.4 to 0 … 16 bar abs.
■ -1 … 0 to -1 … +25 bar

Special feature■ Exceptionally large variety■ Low-power version■ For sour gas applications (NACE)■ Excellent quality, proven in the field■ Further worldwide Ex approvals

PE 81.58

PE 81.27■ 0 … 0.4 to 0 … 25 bar abs.
■ 1 … 0 to -1 … +24 bar

Special feature■ Flush process connection (optional)

Data sheet

16

S-11

■ IO-Link version 1.1■ Medium temperature to +125 °C■ Multicolour 360° LED status display

Data sheet

PE 81.90

PressurePressure ‫ ׀‬Transmit

HP-2

For highest-pressure applications
to 15,000 bar

M-10, M-11

Spanner width 19 mm

Accuracy(± % of span)≤ 0.25 or 0.5

Non-linearity(± % of span)

Measuring range

0 … 1,600 to 0 … 15,000 bar

Measuring range

0 … 10 to 0 … 1,000 bar

Special feature■ Very high long-term stability

Special feature■ Small spanner width 19 mm

Data sheet

PE 81.25■ Excellent load cycle stability■ Cavitation protection (optional)

Data sheet≤ 0.2 BFSL

■ Flush connection G ¼ available

PE 81.53

P-30, P-31

For precision measurements

Non-linearity(± % of span)≤ 0.04 BFSL

Measuring range■ 0 … 0.25 to 0 … 1,000 bar■ 0 … 0.25 to 0 … 25 bar abs.
■ -1 … 0 to -1 … +15 bar

Special feature■ No additional temperature error in therange 10 ... 60 °C■ Flush process connection (optional)■ Analogue, CANopen® or USB

Data sheet

PE 81.54

Further information at www.wika.com

17

Pressure ‫ ׀‬Transmit

OEM pressure sensors

O-10

For industrial applications

MH-4

For mobile working machines

Non-linearity(± % of span)≤ 0.5 BFSL

Non-linearity(per IEC 62828-1)

Measuring range■ 0 … 6 to 0 … 600 bar

Special feature■ -1 … +5 to -1 … +59 bar

Non-linearity(per IEC 62828-1)

Measuring range

0 ... 6 to 0 ... 1,000 bar

Measuring range

0 … 40 to 0 … 600 bar

Special feature■ For extreme operating conditions

Special feature■ For extreme operating conditions■ Reliable and accurate■ Customer-specific solutions■ Excellent long-term stability

Data sheet

PE 81.63■ Signal stability thanks to CANopen®
■ Customer-specific solutions■ High production capacities

Data sheet

PE 83.02

PE 81.65

MH-3-HY

For mobile hydrogen applications

MG-1

For medical gases

R-1

For heating and refrigeration
technology

Accuracy(± % of span)≤1

Non-linearity(± % of span)≤ 0.5 BFSL

Accuracy(± % of span)≤2

Measuring range

0 … 20 to 0 … 600 bar

Measuring range

0 … 6 to 0 … 400 bar

Measuring range■ 0 … 6 to 0 … 160 bar

Special feature■ Approval per EC79/2009

Special feature

Cleaned, packed and marked for oxygen
per international standards

Special feature■ Special case design for the best possi-■ Compact and robust design■ Diagnostic function (optional)

Data sheet

PE 81.59

Data sheet■ -1 … +7 to -1 … +45 bar

ble condensation tightness■ Resistant to all common refrigerants

PE 81.44■ Wetted parts from stainless steel

Data sheet

18≤ ±0.25 % of span (BFSL)■ Reliable and accurate■ High production capacities■ Consistent quality

Data sheet

For mobile working machines,
CANopen®/J1939≤ ±0.25 % of span (BFSL)■ Customer-specific solutions■ Good delivery performanceMH-4-CAN

PE 81.45

Pressure ‫ ׀‬Transmit

The multi-functional intelliGAUGEs present a cost-effective
and, at the same time, reliable solution for nearly all pressure
measurement applications. They combine the analogue indicationof a mechanical pressure gauge, needing no auxiliary power, withthe electrical output signal of a pressure sensor. These hybrid
instruments are available with all commonly used electrical signals.
The sensor works in a non-contact way, without any influence on
the measuring signal. Many instruments are available in versions
for use in hazardous areas.

PGT21

Bourdon tube,
stainless steel case

Depending on the pressure gauge, the following electrical
output signals are possible:






0.5 … 4.5 V ratiometric
4 … 20 mA, 2-wire
4 ... 20 mA, 2-wire with Ex approvals
0 … 20 mA, 3-wire
0 … 10 V, 3-wire

For pressure gauges with nominal sizes 100 and 160 mm, theelectrical output signals can also be combined with switch
contacts.

PGT23.063

Bourdon tube, for the processindustry, safety version

PGT23.100,
PGT23.160

Bourdon tube, for the process industry,
standard or safety version

Nominal size

50, 63 mm

Nominal size

63 mm

Nominal size

100, 160 mm

Scale range

0 … 1.6 to 0 … 400 bar

Scale range

0 … 1 to 0 … 1,000 bar

Scale range

0 … 0.6 to 0 … 1,600 bar

Accuracy class

2.5

Accuracy class

1.6

Accuracy class

1.0

Ingress protection

IP65 (IP67 optional)

Ingress protection

IP54, filled IP65

Ingress protection

IP54, filled IP65

Data sheet

PV 11.03

Data sheet

PV 12.03

Data sheet

PV 12.04

PGT43

Diaphragm element, for theprocess industry, high overload
safety up to 10 times the full scalevalue, max. 40 bar

PGT43HP

Diaphragm element, for theprocess industry, high overload
safety to 40, 100 or 400 bar

PGT63HP

Capsule element, forthe process industry, high overload
safetyNominal size

100, 160 mm

Nominal size

100, 160 mm

Nominal size

100, 160 mm

Scale range

0 … 16 mbar to 0 … 25 bar

Scale range

0 … 16 mbar to 0 … 40 bar

Scale range

2.5 … 100 mbar

Accuracy class

1.6

Accuracy class

1.6

Accuracy class

1.6

Ingress protection

IP54, with liquid filling IP65

Ingress protection

IP54, with liquid filling IP65

Ingress protection

IP54

Data sheet

PV 14.03

Data sheet

PV 14.07

Data sheet

PV 16.06

Further information at www.wika.com

Pressure

Pressure gauges with output signal

19

Pressure ‫ ׀‬Transmit

Pressure gauges with output signal

DPGT43Differential pressure,for the process industry,all-metal media chamber

Differential pressure,for the process industry,high overload safety to 650 bar

Nominal size

100, 160 mm

Nominal size

100, 160 mm

Scale range

0 … 16 mbar to 0 … 40 bar

Scale range

0 … 60 mbar to 0 … 40 bar

Accuracy class

1.6

Accuracy class

1.6

Ingress protection

IP54, filled IP65

Ingress protection

IP54, filled IP65

Data sheet

PV 17.05

Data sheet

PV 17.13

DPGT40Differential pressure, with integrated
working pressure indication (DELTAtrans)

20

DPGT43HP

APGT43Absolute pressure,for the process industry

Nominal size

100 mmNominal size

100, 160 mm

Scale range

0 … 160 mbar to 0 … 10 bar

Scale range

0 … 25 mbar to 0 … 25 bar abs.

Accuracy class2.5 (1.6 optional)

Accuracy class

2.5

Ingress protection

IP65

Ingress protection

IP54, with liquid filling IP65

Data sheet

PV 17.19

Data sheet

PV 15.02

Pressure ‫ ׀‬Display + Switch

Control systems are gaining more and more importance inindustrial applications. Consequently, mere pressure indicationon the measuring instrument itself is no longer sufficient, rather
the measured value must be transferred to the control system via
an electrical signal, e.g. by closing or opening of a circuit. WIKA
is focusing on its contact pressure gauges in order to satisfy thistrend.
All instruments with inductive contacts are certified in accordancewith ATEX Ex ia.

PGS21

Bourdon tube,
stainless steel case

PressureContact pressure gauges

Depending on the model the following contacts are built-in: Magnetic snap-action contact, e.g. model 821, for general






applications
Inductive contact model 831, for hazardous areas
Electronic contact model 830 E, for PLC
Reed contact model 851, for general applications and PLC
Microswitch model 850
Transistor output NPN or PNP

PGS25

Bourdon tube, withelectronic pressure switch,stainless steel case

PGS21.100,
PGS21.160

Bourdon tube, stainless steel case

Nominal size

40, 50, 63 mm

Nominal size

50, 63 mm

Nominal size

100, 160 mm

Scale range

0 … 2.5 to 0 … 400 bar

Scale range

0 … 1.6 to 0 … 400 bar

Scale range

0 … 0.6 to 0 … 600 bar

Accuracy class

2.5

Accuracy class

2.5

Accuracy class

1.0

Ingress protection

IP65

Ingress protection

IP65

Ingress protection

IP54

Special feature

Version with VdS or LPCB approval
possible

Data sheet

PV 21.04

Data sheet

PV 22.01

Data sheet

PV 21.02

PGS23.100,
PGS23.160

Bourdon tube, for the process industry,
standard or safety version

PGS23.063

Bourdon tube, for the processindustry, safety version

PGS43.100,
PGS43.160

Diaphragm element, for the process industry,high overload safety up to 10 times thefull scale value, max. 40 bar

Nominal size

100, 160 mm

Nominal size

63 mm

Nominal size

100, 160 mm

Scale range

0 … 0.6 to 0 … 1,600 bar

Scale range

0 … 4 to 0 … 400 bar

Scale range

0 … 25 mbar to 0 … 25 bar

Accuracy class

1.0

Accuracy class

1.6

Accuracy class

1.6

Ingress protection

IP65 or IP66

Ingress protection

IP54

Ingress protection

IP54, with liquid filling IP65

Data sheet

PV 22.02

Data sheet

PV 22.03

Data sheet

PV 24.03

Further information at www.wika.com

21

Pressure ‫ ׀‬Display + Switch

Contact pressure gauges
432.36,432.56 with 8xx

Diaphragm element, for the processindustry, high overload safety to 100 or
400 bar532.53 with 8xx

Absolute pressure, for theprocess industry, high overload safety632.51 with 8xx

Capsule element, for theprocess industry, high overload safety

Nominal size

100, 160 mm

Scale range

0 … 25 mbar to 0 … 40 bar

Accuracy class

1.6

Ingress protection

IP54, with liquid filling IP65

Data sheet

PV 24.07

DPGS40Differential pressure, withmicroswitches, with integrated
working pressure indication(DELTA-comb)

Nominal size

100, 160 mm

Nominal size

100, 160 mm

Nominal size

100 mmScale range

0 … 25 mbar to 0 … 25 bar abs.

Scale range

0 … 2.5 to 0 … 100 mbar

Scale range

0 … 250 mbar to 0 … 10 bar

Accuracy class

1.6

Accuracy class

1.6

Accuracy class2.5 (1.6 optional)

Ingress protection

IP54, with liquid filling IP65

Ingress protection

IP54

Ingress protection

IP65

Data sheet

PV 25.02

Data sheet

PV 26.06

Data sheet

PV 27.20

SIL

IEC 61508 / IEC 61511

DPGS40TA

Differential pressure,with microswitches,with integrated working pressure
indication (DELTA-comb), withcomponent testing

22

DPGS43Differential pressure,for the process industry,all-metal media chamber

DPGS43HP

Differential pressure,for the process industry,high overload safety to 400 bar

Nominal size

100 mmNominal size

100, 160 mm

Nominal size

100, 160 mm

Scale range

0 … 250 mbar to 0 … 10 bar

Scale range

0 … 16 mbar to 0 … 40 bar

Scale range

0 … 60 mbar to 0 … 40 bar

Accuracy class2.5 (1.6 optional)

Accuracy class

1.6

Accuracy class

1.6

Ingress protection

IP65

Ingress protection

IP54, filled IP65

Ingress protection

IP54, filled IP65

Data sheet

PV 27.22

Data sheet

PV 27.05

Data sheet

PV 27.13

Pressure ‫ ׀‬Switch

Pressure

Pressure switches
Electronic pressure switches

PSD-4

Electronic pressure switch
with display

PSD-4-ECO

Electronic pressure switch
with display

Accuracy(± % of span)≤ 0.5

Accuracy(± % of span)≤ 1.0

Measuring range■ 0 … 0.4 to 0 … 1,000 bar

Measuring range■ 0 … 0.4 to 0 … 1,000 bar■ 0 … 0.4 to 0 … 25 bar abs.
■ -1 … 0 to -1 … +24 bar

Special feature■ Condition monitoring via IO-Link■ Reduction of variants■ -1 … 0 to -1 … +24 bar

Special feature■ Easy installation, good readability■ Good/Bad indication through parame-

terisable digital display (red/green)■ Compact size enables easy installationin confined spaces■ Parameterisation via 3 buttons

Data sheet■ 0 … 0.4 to 0 … 25 bar abs.

■ Optimised design makes OEM machine

integration easier

PE 81.86■ Designed for rough demands to

50g shock and -40 … +125 °C[-40 … +257 °F]

Data sheet

PE 81.69

A-1200With IO-Link, PNP or
NPN switching output

Accuracy(± % of span)≤ 0.5 or ≤ 1

Measuring range■ 0 … 0.4 to 0 … 1,000 bar■ 0 … 0.4 to 0 … 25 bar abs.
■ 1 … 0 to -1 … +24 bar

Special feature■ IO-Link version 1.1■ Medium temperature to +125 °C■ Multicolour 360° LED status display

Data sheet

PE 81.90

Further information at www.wika.com

23

Pressure ‫ ׀‬Switch

Pressure switches
Mechanical pressure switches for industrial applications

PSM01

Compact pressure switch

PSM02

Compact pressure switch,
settable hysteresis

Setting range-0.85 … -0.15 bar
0.2 … 2 bar to 30 … 320 bar

Setting range-0.85 … -0.15 bar
0.2 … 2 bar to 30 … 320 bar

Switching function

Normally open, normally closed, changeover contact

Switching function

Normally open, normally closed, changeover contact

Material

Galvanised steel or stainless steel

Material

Galvanised steel or stainless steel

Switching power■ 2 A, AC 48 V

Switching power■ 2 A / 4 A, AC 250 V

Data sheet

PV 34.81

Data sheet

PV 34.82■ 1 A / 2 A, DC 24 V

PSM-520

Pressure switch,
settable hysteresis
Setting range
Switching function
Material■ -0.4 … +7 bar■ 0 … 5 bar to 6 … 30 bar

Normally open, normally closed, changeover contact■ Bellows: Copper alloy CuSn6 per EN

1652

PSM-550

Pressure switch, for demanding
industrial applications
Setting range

24

Data sheet

PV 35.01■ 0 … 300 mbar

Switching function

Change-over contact (SPDT)

Material■ Bellows / Process connection:

EN1A per EN 10277-3, tin-plated

10 A / 6 A, AC 230 V■ -1 … 0 and -0.8 … +5 bar■ 0.1 … 1.1 bar to 10 … 30 bar■ Process connection: free cutting steel

Switching power■ 2 A / 4 A, DC 24 V

Copper alloy CuSn6 per EN 1652 or
stainless steel 1.4401■ With NBR diaphragm:
Process connection: free cutting steel
EN1A per EN 10277-3, tin-plated

Switching power

4 A / 10 A, AC 230 V

Data sheet

PV 35.03

PSM-700

Pressure switch, high adjustabilityof switch differential
Setting range■ -1 … 1.5 bar■ 0.2 … 1.6 bar, 7 … 35 bar

Switching function

Change-over contact (SPDT and DPDT)

Material■ Measuring element: Stainless steel

316L■ Process connection: Stainless steel

316L■ Case: Aluminium

Switching power

Up to AC 250 V/15 A

Data sheet

PV 35.05

PressurePressure ‫ ׀‬Switch

Mechanical pressure switches for the process industry
Due to the use of high-quality microswitches, the mechanicalpressure switches are notable for their high precision and longterm stability. Furthermore, the direct switching of electrical loads
up to AC 250 V/20 A is enabled, while simultaneously ensuring ahigh switch point reproducibility.

SIL3

The instruments come with a SIL certificate and are thus
particularly suited for safety-critical applications. In addition,with their ‘intrinsically safe’ and ‘flameproof enclosure’ ignition
protection types the pressure switches are ideally suited forpermanent use in hazardous areas.
All mechanical pressure switches for the process industry areavailable with EAC certificate and technical passport.

SIL

SIL3

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

PXS, PXA

Miniature pressure switch

PCS, PCA

Compact pressure switch

MW, MA

Diaphragm
pressure switch

Setting range

1 … 2.5 to 200 … 1,000 bar

Setting range-1 … -0.2 to 200 … 1,000 bar

Setting range

0 … 16 mbar to 30 … 600 bar

Ignition protection
type

Ex ia or Ex d

Ignition protection
type

Ex ia or Ex d

Ignition protection
type

Ex ia or Ex d

Switch

1 x SPDT or DPDT

Switch

1 x SPDT or DPDT

Switch1 or 2 x SPDT or 1 x DPDT

Switching power■ AC 250 V/5 A

Switching power■ AC 250 V/15 A

Switching power■ AC 250 V/20 A

Data sheet

PV 34.36, PV 34.38

PV 33.30, PV 33.31

Data sheet

PV 31.10, PV 31.11■ DC 24 V/5 A

Data sheet

SIL

SIL

BWX, BA

Bourdon tube pressure switch

Differential pressure switch

0 … 2.5 to 0 … 1,000 bar

Ignition protection
type

Ex ia or Ex d

Switch1 or 2 x SPDT or 1 x DPDT

Ignition protection
type

Switching power■ AC 250 V/20 A■ DC 24 V/2 A

PV 32.20, PV 32.22

Further information at www.wika.com

IEC 61508 / IEC 61511

DW, DA

Setting range

Setting range■ DC 24 V/2 A

SIL

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

Data sheet■ DC 24 V/2 A

APW, APA

Absolute pressure switch

0 … 16 mbar to 0 … 40 bar,static pressure to 160 bar

Setting range

0 … 25 mbar to 0 … 1.5 bar abs.

Proof pressure

11 bar abs.

Ex ia or Ex d

Ignition protection
type

Ex ia or Ex d

Switch1 or 2 x SPDT or 1 x DPDT

Switching power■ AC 250 V/20 A

Switch1 or 2 x SPDT or 1 x DPDT

Data sheet

PV 35.49, PV 35.48

Data sheet

PV 35.42, PV 35.43, PV 35.50■ DC 24 V/2 A

25

Pressure ‫ ׀‬Diaphragm seal systems

Diaphragm seal systems
These combinations of diaphragm seals and pressure gauges
or pressure sensors feature fast availability. They are particularly
suitable for demanding measuring requirements in thepharmaceutical and biotechnology industries, food and beverageindustries, and through to the oil and gas, chemical, petrochemicaland semiconductor industries.
The diaphragm seal systems can be used for processes withgases, compressed air or vapour, with liquid, pasty, powdery andWith flange connection

DSS26M

With pressuregauge per EN 837-1, internal
diaphragm

With threaded connection

DSS34M

Diaphragm seals

With pressure gauge
per EN 837-1, welded design

Applications with small flange process connections in theprocess industry

Applications with high requirements in the chemical,
petrochemical and water treatment industries

PN max.

40 bar

PN max.

60 bar

System fill fluid

KN2 for general applications

System fill fluid

KN2 for general applications

Data sheet

DS 95.09

Data sheet

DS 95.15

DSS26T

With high-quality pressure
sensor, internal diaphragm

26

crystallising media and also with aggressive, adhesive, corrosive,highly viscous, environmentally hazardous or toxic media.
The diaphragm seal is directly welded to the pressure gauge orpressure sensor. The diaphragm made of stainless steel provides
for the separation from the medium. The pressure is transmitted
to the measuring instrument via the system fill fluid which is insidethe diaphragm seal system.

DSS34T

With high-quality pressure
sensor, welded design

Applications with small flange process connections in theprocess industry

Applications with high requirements in the chemical,
petrochemical and water treatment industries

PN max.

40 bar

PN max.

60 bar

System fill fluid

KN2 for general applications

System fill fluid

KN2 for general applications

Data sheet

DS 95.10

Data sheet

DS 95.16

Diaphragm seals –combinations andaccessories

Extensive information can be found in our
brochure “Diaphragm seals – combinations
and accessories” at www.wika.com.

Order information

Diaphragm seal systems
with short delivery times

Extensive information can be found in our
brochure “Diaphragm seal systems withshort delivery times” at www.wika.com.

Pressure ‫ ׀‬Valves and mounting accessories

Valves and protective devices for increased safety and service
life. Via cocks, shut-off valves, valve manifolds or monoflanges,
pressure measuring instruments can be securely separated fromthe process during commissioning, maintenance or calibration.
Protective devices, such as syphons, overpressure protectors

IV10, IV11

Needle valve andmultiport valve
Application

For shutting off pressure measuring instruments with threaded connection

Version

Needle valve and multiport valve

Material

Stainless steel

Nominal pressure

To PN 420 (6,000 psi)
Option: to PN 680 (10,000 psi)

Data sheet

AC 09.22

IVM

Monoflanges

Application

For shutting off and venting pressure
measuring instruments with flange
connection

VersionFlange connection per ASME or EN

Material

Stainless steel

Nominal pressure

To PN 420 (6,000 psi)

Data sheet

AC 09.17

Block-and-bleed valve,square or flat form

VersionFor shutting off and venting pressure
measuring instruments with threaded
connection
Block-and-bleed valve

Material

Stainless steel

Nominal pressure

To PN 420 (6,000 psi)
Option: to PN 680 (10,000 psi)

Data sheet

AC 09.19

IBM, IBF

Application

IV30, IV31,
IV50, IV51

Valve manifold for differential pressure
measuring instruments
Application

For shutting off, pressure equalising aswell as purging and venting differential
pressure measuring instruments

Version

3-way and 5-way valve

Material

Stainless steel

Nominal pressure

To PN 420 (6,000 psi)
Option: to PN 680 (10,000 psi)

Data sheet

AC 09.23

910.10, 910.11

Monoblock

Stopcock and
DIN shut-off valve
Direct connection of pressure measuring
instruments to pipelines or containers
without interface valves. Control panels,
lubrication systems, dry gas seals

Application

For shutting off pressure measuring instruments with threaded connection

VersionMaterial

Stainless steel

Material

910.10: per DIN 16261, DIN 16262,
DIN 16263
910.11: per DIN 16270, DIN 16271,
DIN 16272

Nominal pressure

BF: class 150 … class 2,500,
in line with ASME B16.5
PN 16 … PN 100, in line with EN 1092-1
IBM: 6,000 ... 10,000 psi (420 ... 690 bar)

Nominal pressure

910.10: to 25 bar
910.11: to 400 bar

Data sheet

AC 09.01, AC 09.02

Version

Data sheet

Further information at www.wika.com

and snubbers, increase the service life and expand the range ofapplications for pressure measuring instruments. In addition to theextensive selection of instrumentation valves and accessories,
WIKA also offers the qualified assembly of various individual parts
to form a complete measuring assembly (“instrument hook-up”).

IV20, IV21

Application

Pressure

Valves and mounting accessories

Flange/threaded, flange/flange or
threaded/threaded

Brass, steel, stainless steel

AC 09.24, AC 09.25

27

Pressure ‫ ׀‬Valves and mounting accessories

Valves and mounting accessories

BV

HPNV

Ball valve

High-pressure needle valve

Application

First shut-off valve for pressure tap to local
instrument installation, media distribution,
drain or vent in pipelines

Application

VersionProcess and instrument version

VersionMaterial

Stainless steel 316L

Nominal pressure

To PN 420 (6,000 psi)
Option: to PN 680 (10,000 psi)

Data sheet

AC 09.28

910.15

Syphons and connecting
pipes

28

For injection systems, test benches, hydraulic power packs, blow-out protection,
blasting/cutting with water, high-pressurecleaning

2-way valve, straight or angled bore; 3-way
valve, one or two pressure connections

Material

Stainless steel

Nominal pressure

15,000 ... 60,000 psi [1,034 ... 4,136 bar]
Option: To PN 680 (10,000 psi)

Data sheet

AC 09.27

HPFA
High-pressure fittings and accessories

Application

For the protection of pressure measuring
instruments from excessive pulsationand heat

Application

For high-pressure applications in control
panels, where the space is restricted, or
test benches

VersionU-form, trumpet form, compact form,
standard

VersionMaterial

Steel, stainless steel

Nominal pressure

To 160 bar

Elbow fitting, T-fitting, cross fitting, bulkhead connector, anti-vibration gland, glandcollar, threaded connection, nipple, sealingcap, blind plug

Data sheet

AC 09.06

Material

Stainless steel

Nominal pressure

15,000 ... 60,000 psi [1,034 ... 4,136 bar]

Data sheet

AC 09.32

910.12

Snubber

Application

For the protection of pressure measuring
instruments from pressure surges andpulsations

Material

Brass, steel, stainless steel

Nominal pressure

To 400 bar

Data sheet

AC 09.03

Extensive information can be foundin our brochure “Instrumentation
valves and mounting accessories”
at www.wika.com.

From process to measurement

Instrumentation valves andmounting accessories

Pressure ‫ ׀‬Electrical accessories

A-AI-1,
A-IAI-1

Pressure

Electrical accessories

M12 x 1 cable

Preassembled cables M12 x 1

LCD attachable indicator, 50 x 50 mm
Input

4 … 20 mA, 2-wire■ Circular connector M12 x 1, 4- and 5-pin

Auxiliary power

Supply from the 4 … 20 mA current loop

■ Straight and angled version

Special feature

Model A-IAI-1 intrinsically safe per ATEX

Data sheet

AC 80.07■ 2, 5 or 10 m cable

905

Contact protection relay
for model 821 switch contacts■ Ingress protection IP67

904

Control unit
for inductive contacts model 831

Application

For optimal contact protection and highest
switching reliability

Application

For operating measuring instruments withinductive switch contacts

Data sheet

AC 08.05

Data sheet

AC 08.04

Further information at www.wika.com

29Temperature ‫ ׀‬Display

Dial thermometers

Diverse connection designs, stem diameters and individual
stem lengths enable a flexible measuring point design. Dial
thermometers with capillaries are particularly versatile.

Our dial thermometers work on the bimetal, expansion or gas
actuation principle. This enables scale ranges of -200 ... +700 °C
in different class accuracies, response times and resilience toenvironmental influences.

All thermometers are suited for operation in a thermowell ifnecessary.

Bimetal thermometers

A43

Heating technology

A50Standard version

Refrigeration andair-conditioning
applications

Nominal size

63, 80, 100 mm

Nominal size

63, 80, 100, 160 mm

Nominal size

63, 80, 100, 160 mm

Scale range-30 … +120 °C

Scale range-30 … +120 °C

Scale range-30 … +200 °C

Wetted parts

Copper alloy

Connection

Data sheet

TM 48.01

Removable thermowell withretainer screw

Permissible operating pres- Max. 6 bar
sure at thermowell/stem
Wetted parts

Copper alloy

Wetted parts

Copper alloy

Data sheet

TM 43.01

Data sheet

TM 50.03

A52, R52

Industrial series, axial and radial

Nominal size
Scale range

TG53

Process version per
ASME B40.200

TG54Process version per EN 13190

25, 33, 40, 50, 63, 80,
100, 160 mm

Nominal size

3, 4, 5, 6"

Nominal size

63, 80, 100, 160 mm

Scale range-70 … +70 to 0 … +600 °C

Scale range-70 … +70 to 0 … +600 °C-30 … +50 to 0 … +500 °C

Wetted parts

Stainless steel

Wetted parts

Stainless steel

OptionLiquid damping to max. 250 °C (case andprobe)

OptionLiquid damping to max. 250 °C (case andprobe)

Data sheet

TM 53.02

Data sheet

TM 54.02

Permissible operating pres- Max. 25 bar
sure at thermowell/stem

30

A48

Wetted parts

Stainless steel

Data sheet

TM 52.01

Temperature

Temperature ‫ ׀‬Display

Bimetal thermometers

Machine glass thermometer®
TYPE EL
CLASS IOCTOBER 202255

High-quality process version
per EN 13190

TG58SA

Bimetal thermometerfor sanitary applications

32

V shapeNominal size

63, 100, 160 mm

Nominal size

63, 80, 100, 130 mm

Nominal size

110, 150, 200 mm

Scale range-70 … +70 to 0 … 600 °C

Scale range-50 … 50 °C to -20 … 200 °C

Scale range-30 … +200 °C

Wetted parts

Stainless steel

Wetted parts

Stainless steel 316L

Wetted parts

Copper alloy

OptionLiquid damping to max. 250 °C (case andprobe)

Option■ Case filling with FDA-approved silicone

Option■ Dual scale °F/°C

Data sheet

TM 32.02

Data sheet

oil■ Certification packages for food andTM 55.01
Data sheet

pharmaceutical applications■ 2 variants: straight and 90°

TM 58.01

Expansion thermometers

TF58, TF59

With capillary, edgewise panel design

With capillary,stainless steel version
Nominal size

63, 100, 160 mm

Scale range-60 … +400 °C

Copper alloy

Wetted parts

Stainless steel■ Vertical arrangement

Option■ Liquid damping (case)TM 80.02

Data sheet

TM 81.01

Nominal size

58 x 25 mm, 62 x 11 mm

Scale range-50 … 250 °C

Wetted parts
OptionData sheet

70■ Special scales

Further information at www.wika.com■ Indication accuracy class 1

IFC

With capillary, standard version

Nominal size

52, 60, 80, 100 mm
48 x 48, 72 x 72, 96 x 96 mm

Scale range-100 … +400 °C

Wetted parts

Copper alloy

Option■ Square case version

Data sheet

TM 80.01■ Other case materials

31Temperature ‫ ׀‬Display

Dial thermometersGas-actuated thermometers

R73, S73, A73

Axial and radial,adjustable stem and dial

F73

With capillary

75

Highly vibration-resistantNominal size

100, 160 mm

Nominal size

100, 160 mm

Nominal size

100 mmScale range-200 … +100 to 0 … +700 °C

Scale range-200 … +100 to 0 … +700 °C

Scale range

0 … +700 or -50 … +650 °C

Wetted parts
OptionData sheet

Stainless steel■ Liquid damping (case)■ Contact bulb

Wetted parts
Option

TM 73.01
Data sheet

Stainless steel■ Armoured or coated capillary(PVC coating)■ Liquid damping (case)■ Contact bulb

Wetted parts

Stainless steel

OptionVarious neck tube and insertion lengths

Data sheet

TM 75.01

TM 73.01

Thermomanometers

MFTWith capillaries, forpressure and temperature measurement

Eco version, forpressure and temperature measurement

100.02

For pressure and temperature
measurement

Nominal size

40, 42, 52 mm

Nominal size

63, 80 mm

Nominal size

63, 80 mm

Scale range■ Pressure: 0 … 4 bar

Scale range■ Pressure: 0 … 4 to 0 … 10 bar

Scale range■ Pressure: 0 … 1 to 0 … 16 bar

Accuracy class■ Pressure: 2.5 (EN 837-1)

Connection location Lower mount or back mountAccuracy class■ Pressure: 2.5 (EN 837-1)

Data sheet

PM 01.23

Data sheet

32

THM10

■ Temperature: 0 … 120 °C■ Temperature: 2.5■ Temperature: 0 … 120 °C

Accuracy class■ Pressure: 2.5 (EN 837-1)

Data sheet

PM 01.24

PM 01.20■ Temperature: 2 (EN 13190)■ Temperature: 0 … 100 to 0 … 150 °C■ Temperature: 2.5 °C

Temperature

Temperature ‫ ׀‬Display

Dial thermometers with output signal

TGT70

Expansion thermometerwith output signal

TGT73

Gas-actuated thermometerwith output signal

Nominal size

63, 100 mm

Nominal size

100, 160 mm

Scale range-40 … +60 to 0 … 250 °C

Scale range-200 … +100 to 0 … 700 °C

Wetted parts
OptionData sheet

Stainless steel■ Capillary■ Output signals 4 … 20 mA or

0.5 … 4.5 V■ Other connection designs
TV 18.01

Further information at www.wika.com

Wetted parts
OptionStainless steel■ Capillary■ Liquid damping (case)■ Output signal 4 … 20 mA or 0 … 10 V

Data sheet

TV 17.10

33

Temperature ‫ ׀‬Display

Digital indicators

DI10

For panel mounting, current loop
display, 96 x 48 mm

For panel mounting, 96 x 96 mm

Input

4 … 20 mA, 2-wire

Input

Alarm output

2 electronic contacts (optional)

Alarm output

2 relaysSpecial feature

Wall-mounting case (optional)

Special feature■ Integrated transmitter power supply

Auxiliary power

Supply from the 4 … 20 mA current loop
Auxiliary power

AC 230 V or AC 115 V

Data sheet

AC 80.05

Data sheet

AC 80.06

DI32-1

For panel mounting, 48 x 24 mm

34

DI30

Input

Multi-function input for resistance thermometers, thermocouples and standard
signals

Alarm output

2 electronic contacts

Auxiliary power

DC 9 … 28 V

Data sheet

AC 80.13

Standard signals■ Wall-mounting case (optional)

DI35

For panel mounting, 96 x 48 mm

Input■ Multi-function input for resistancethermometers, thermocouples andstandard signals■ Alternatively double input for standard
signals with calculation function (+ - x /)for two transmitters

Alarm output

2 or 4 relays (optional)

Special feature■ Integrated transmitter power supply

Auxiliary power■ AC/DC 100 … 240 V

Data sheet

AC 80.03■ Analogue output signal (optional)■ DC 10 … 40 V, AC 18 … 30 V

Temperature

Temperature ‫ ׀‬Display

SIL

IEC 61508 / IEC 61511

DIH10

Connection head with digitalindicator

DIH50, DIH52

For current loops with
HART® communication

TF-LCD

Temperature probe for heating
and refrigeration technology, withdigital indicatorInput

4 … 20 mA

Dimensions

150 x 127 x 127 mm

Measuring range-40 … +120 °C

Auxiliary power

Supply from the 4 … 20 mA current loop

Case

Aluminium, stainless steel

Special feature■ Dust and waterproof case, IP68

Special feature■ Adjustment of display range and unit via

Data sheet

AC 80.11

Further information at www.wika.com

HART® communication■ Model DIH52 additionally suitable
for multidrop operation and with local
master function

Approval■ Intrinsically safe

Data sheet

AC 80.10■ Battery or solar operation■ Extremely long service life

Data sheet

TE 85.01■ Flameproof enclosure

35Temperature ‫ ׀‬Transmit + Record

Thermocouples

Thermocouples generate a voltage directly dependent
on temperature. They are particularly suitable for hightemperatures to 1,700 °C (3,092 °F) and for very high oscillatingstresses. For thermocouples, the accuracy classes per
IEC 60584-1 and ASTM E230 apply.

In our range of products you will find all market-standard
instrument versions. If required, a temperature transmitter can beinstalled in the connection head.

TC10-A

Measuring insert

NAMUR

NAMUR

NAMUR

TC10-B

TC10-CFor additional thermowell

Threaded, with protection tube

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, E, N or T

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Data sheet

Data sheet

Process connection Mounting thread

TE 65.01

TE 65.02

Data sheet

TE 65.03

NAMUR

TC10-D

Threaded, miniature design36

TC10-F

Flanged thermocouple, withprotection tube

NAMUR

TC10-H

Without thermowell

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, E, N or T

Measuring range-40 … +600 °C, -40 … +1,112 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Process connection Mounting thread

Process connection Flange

Process connection Mounting thread

Data sheet

Data sheet

Data sheet

TE 65.04

TE 65.06

TE 65.08

Temperature

Temperature ‫ ׀‬Transmit + Record

TC10-K

Measuring insert,
for installation in TC10-L

TC10-L

Flameproof enclosure,for additional thermowell

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, E, N or T

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Data sheet

Data sheet

TE 65.11

TC12-A

Measuring insert forprocess thermocouple

TE 65.12

TC12-B

Process thermocouple,
for additional thermowellTC12-M

Process thermocouple,
basic module

Sensor element

Types K, J, N or T

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, E, N or T

Measuring range-40 … +1,200 °C,-40 … +2,192 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

OptionEx i, Ex d

OptionEx i, Ex d

Data sheet

Data sheet

TE 65.17

Data sheet

TE 65.17

TE 65.16

Further information at www.wika.com

37Temperature ‫ ׀‬Transmit + Record

Thermocouples

TC40

Cable thermocouple

Sensor element

Types K, J, E, N or T

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring location Ungrounded or grounded
Cable

Silicone, PTFE/PFA, fibreglass

Data sheet

TE 65.40

TC46

Hot runner thermocouple

Sensor element

Types J or K

Measuring range-25 … +400 °C, -13 … +752 °F

Measuring location Ungrounded or groundedSpecial feature■ Probe diameter 0.5 … 3.0 mm

Data sheet

TE 65.46■ Plastic-moulded transition

TC47

Thermocouple forplastics machinery
Sensor element

Types J or K

Measuring range-25 … +400 °C, -13 … + 752 °F

Measuring location Ungrounded or groundedSpecial feature

Surface thermocouple

Bayonet thermocouple

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, N, E or T

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring range-40 … +1,200 °C, -40 … +2,192 °F

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Process connection Surface mounting

Special feature■ Single and dual thermocouple

Data sheet

TE 65.53

Data sheet

38

TC53

TE 65.50■ Explosion-protected versions■ Connection lead fibreglass with stain-

less steel braid

Data sheet

TC50■ Various process connections

TE 67.20

Temperature

Temperature ‫ ׀‬Transmit + Record

TC59-T

TC59-E

TC59-V

Tubeskin thermocouple assembly

Tubeskin thermocouple assembly

Tubeskin thermocouple assembly

TEFRACTO-PAD®

eTEFRACTO-PAD®

V-PAD®

Sensor element

Types K, J, N, E

Sensor element

Types K, J, N, E

Sensor element

Types K, J, N, E

Measuring range

0 ... 1,260 °C, 32 ... 2,300 °F

Measuring range

0 ... 1,260 °C, 32 ... 2,300 °F

Measuring range-25 … +400 °C, -13 … + 752 °F

Measuring location Ungrounded or grounded

Measuring location Ungrounded or grounded

Measuring location Ungrounded

Process connection Surface mounting, welded/shielded

Process connection Surface mounting, extractable/shielded

Process connection Surface mounting, welded

Data sheet

Data sheet

TE 65.60

Further information at www.wika.com

TE 65.61

Data sheet

TE 65.59

39

Temperature ‫ ׀‬Transmit + Record

Thermocouples

TCC

Linear sensor for
hot spot detection

High-temperature thermocouple

TC81

For flue gas temperature measurements

Sensor element

Type K thermocouple wires

Sensor element

Types S, R, B, K, N or J

Sensor element

Types K, N or J

Measuring range

0 … 400 °C, 32 … 752 °F

Measuring range

0 … 1,700 °C, 32 … 3,092 °F

Measuring range

0 … 1,200 °C, 32 … 2,192 °F

Special feature■ Continuous monitoring

Measuring location Ungrounded

Measuring location Ungrounded or grounded

Process connection Stop flange, threaded bushing

Process connection Stop flange, threaded bushing

Data sheet

Data sheet■ Self-restoring■ Passive element

Data sheet

TE 65.80

TE 65.81

TE 64.40

TC82

High-temperature thermocouple

40

TC80

TC83

Sapphire-design thermocouple

TC84

Sapphire-design thermocouple

Sensor element

Types K, J, E, N, S, R or B

Sensor element

Types K, N, S, R or B

Sensor element

Types S, R, B

Measuring range

0 … 1,700 °C, 32 … 3,092 °F

Measuring range

0 … 1,700 °C, 32 … 3,092 °F

Measuring range

0 … 1,700 °C, 32 … 3,092 °F

Thermowell

C610, C799

Thermowell

Sapphire (monocrystalline)

Thermowell

Sapphire (monocrystalline)

Data sheet

TE 65.82

Data sheet

TE 65.83

Case

Highest safety thanks to 2-chambersystemData sheet

TE 65.84

Temperature

Temperature ‫ ׀‬Transmit + Record

TC90

High-pressure thermocouple

TC95

Multipoint thermocouple

TC96-R

Flexible multipoint thermocouple

Sensor element

Types K, J or E

Sensor element

Types K, J, E, N or T

Sensor element

Types K, J, E or N

Measuring range

0 … 350 °C, 32 … 662 °F

Measuring range

0 … 1,200 °C, 32 … 2,192 °F

Measuring range

0 … 1,200 °C, 32 … 2,192 °F

Measuring tip

Ungrounded or grounded

Measuring tip

Ungrounded or grounded

Measuring tip

Ungrounded or grounded

Process connection Various high-pressure connections

Process connection Various process connections

Process connection Various process connections

Data sheet

Data sheet

Data sheet

TE 65.90

Further information at www.wika.com

TE 70.01

TE 70.10

41

Temperature ‫ ׀‬Transmit + Record

Resistance thermometers

Resistance thermometers are equipped with platinum sensor
elements which change their electrical resistance as a functionof temperature. In our range of products you will find resistancethermometers with connected cable as well as versions with connection head. A temperature transmitter can be installed directly inthe connection head.

Resistance thermometers are suitable for applications between-196 … +600 °C, -320 … +1,112 °F (dependent on instrument
model, sensor element, accuracy class and wetted materials).
Resistance thermometers are available in classes AA,
A and B in accordance with IEC 60751.

NAMUR

TR10-A

Measuring insert,
MIMS cable

NAMUR

NAMUR

TR10-B

TR10-CFor additional thermowell

Threaded, with protection tube

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Measuring range-196 … +600 °C, -320 … +1,112 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Connection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wireMeasuring insert

MIMS cable

Measuring insert

MIMS cable

Process connection Mounting thread

Data sheet

TE 60.01

Data sheet

TE 60.02

Data sheet

TE 60.03

NAMUR

TR10-D

Threaded, miniature design

42

TR10-F

Flanged resistance thermometer,with protection tube

NAMUR

TR10-H

Without thermowell

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Measuring range-196 … +500 °C, -320 … +932 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Connection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wire

Process connection Mounting thread

Process connection Flange

Process connection Mounting thread

Data sheet

Data sheet

Measuring insert

MIMS cable

Data sheet

TE 60.08

TE 60.04

TE 60.06

Temperature

Temperature ‫ ׀‬Transmit + Record

TR10-J

Threaded, withperforated thermowell
Sensor element

1 x Pt100, 2 x Pt100

Measuring range-196 … +600 °C, -320 … +1,112 °F

Connection method 2-, 3- and 4-wireMeasuring insert

MIMS cable

Process connection Mounting thread
Data sheet

TR11-A

Measuring insert, tubular design

TR10-K

Measuring insert,
for installation in TR10-L

TE 60.10

TR10-L

Flameproof enclosure,for additional thermowell

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Measuring range-50 … +250 °C, -58 … +482 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Connection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wireMeasuring insert

Tubular design

Measuring insert

MIMS cable

Measuring insert

MIMS cable

Data sheet

TE 60.13

Data sheet

TE 60.11

Data sheet

TE 60.12

TR12-A

Measuring insert for processresistance thermometer TR12-B

TR12-B

Process resistance thermometer,
for additional thermowellTR12-M

Process resistance thermometer,
basic module

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Measuring range-196 … +600 °C,-320 … +1,112 °F

Measuring range-196 … +600 °C, -320…+1,112 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Connection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wireMeasuring insert

MIMS cable

Measuring insert

MIMS cable

Measuring insert

MIMS cable

OptionEx i, Ex d

OptionEx i, Ex d

Data sheet

TE 60.16

Data sheet

TE 60.17

Data sheet

TE 60.17

Further information at www.wika.com

43Temperature ‫ ׀‬Transmit + Record

Resistance thermometers
TFT35

Threaded temperatureprobe with integrated
transmitter
Measuring range-50 … +200 °C

Special feature■ Output signal 4 … 20 mA, 0 … 10 V,

0.5 … 4.5 V■ Factory configured■ Measuring insert exchangeable■ Electr. connection via plug connection

Data sheet

TR36

Compact version

OEM miniature design

TR33

Miniature design, standard
versionSensor element

1 x Pt100, 1 x Pt1000

Sensor element

1 x Pt100, 1 x Pt1000

Sensor element

1 x Pt100, 1 x Pt1000

Measuring range-50 … +250 °C, -58 … +482 °F

Measuring range-50 … +250 °C, -58 … +482 °F

Measuring range-50 … +250 °C, -58 … +482 °F

OutputPt100, 4 … 20 mA

OutputPt100, Pt1000, 4 … 20 mA

OutputPt100, Pt1000, 4 … 20 mA

Data sheet

TE 60.36

CSA

Ordinary and hazardous locations

CSA

Ordinary locations

Data sheet

TE 60.31

Data sheet

TE 60.33

TR34

Miniature design, explosionprotected

44

TR31

TE 76.18

TR40

Cable resistance thermometer,
MIMS cable

TR41

Cable resistance thermometer
Tubular design

Sensor element

1 x Pt100, 1 x Pt1000

Sensor element

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Measuring range-50 … +250 °C, -58 … +482 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Measuring range-60 ... +250 °C, -76 ... +482 °F

OutputPt100, Pt1000, 4 … 20 mA

Connection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wire

CSA

Hazardous locations

Cable

Silicone, PTFE, PFA

Cable

Silicone, PTFE, PFA

Data sheet

TE 60.34

Data sheet

TE 60.40

Data sheet

TE 60.41

Temperature

Temperature ‫ ׀‬Transmit + Record

TR50

Surface resistance thermometer

TR55

With spring-loaded tip

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100, 2 x Pt100

Measuring range-196 … +600 °C, -320 … +1,112 °F

Measuring range-196 … +400 °C, -320 … +752 °F

Connection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wire

Process connection Surface mounting

Process connection Bayonet

Data sheet

Data sheet

TE 60.50

TR57-M

Pipe surface resistancethermometer for clamping

1 x Pt100, 2 x Pt100

Sensor element

1 x Pt100

Measuring range-196 … +500 °C, -320 … +932 °F

Measuring range-20 … +150 °C, -4 … +302 °F

Connection method 2-, 3- and 4-wireConnection method Pt100 3-wire, 4 … 20 mA

Process connection Compression fitting

Data sheet

TE 60.57

TE 60.55

TR75

DiwiTherm® with digital indicator

Bayonet resistance thermometer

Sensor element

Sensor element

Data sheet

TR53

TR81

For flue gas temperature
measurements

TE 60.53

TR60

Indoor and outdoor resistancethermometer

Sensor element

1 x Pt100, 2 x Pt100

Measuring range-40 … +80 °C, -40 … +176 °F

Connection method 2-, 3- and 4-wire
Process connection Wall mounting
Data sheet

TE 60.60

TR95

Multipoint resistancethermometer-40.0 … +199.9 °C, +200 … +450 °C
with automatic measuring rangechangeover (autorange)

Sensor element

1 x Pt100, 2 x Pt100

Sensor

Pt100

Measuring range-196 … +600 °C, -320 … +1,112 °F

Measuring range-196 … +600 °C, -320 … +1,112 °F

Connection method 2-, 3- and 4-wireConnection method 2-, 3- and 4-wirePower supply

Battery operation

Thermowell

Metal

Process connection Various process connections

Data sheet

TE 60.75

Data sheet

TE 60.81

Data sheet

Measuring range

Further information at www.wika.com

TE 70.01

45

Temperature ‫ ׀‬Transmit + Record

Resistance thermometers

TF35

Threaded temperatureprobe with plug connection
Measuring range-50 … +250 °C

Measuring element Pt1000, Pt100, NTC, KTY
Special feature■ Very high vibration resistance■ Compact design

TF37

Threaded temperatureprobe with connection lead
Measuring range
Special feature■ Electrical connection via plug connec-

Data sheettion

TE 67.10

TF-2000

Cable temperature probe
for heating and refrigeration
technology
Measuring range-50 … +120 °C

Measuring element Pt100, Pt1000, NTC
Special feature

Data sheet

46■ Permanently protected against conden-

sation■ Cost savings thanks to quick assembly■ Delivery reliability, even for large orders
TE 67.40-50 … +260 °C

Measuring element Pt100, Pt1000, NTC, KTY, Ni1000

Data sheet■ High vibration resistance■ Connection lead made of PVC, silicone,

PTFE■ Brass or stainless steel thermowellTE 67.12

TF44

Cable temperature probe for tubeskinmeasurement
Measuring range-50 … +200 °C

Measuring element Pt100, Pt1000, NTC, KTY
Special feature■ Connection lead from PVC, silicone■ Aluminium probe sleeve

TF41

Temperature probe for outdoor temperature measurement
Measuring range
Special feature

Data sheet■ Protected against dust and water jets,■ Clip-on sun protector

TE 67.17

TF45

Cable temperature probe forgeneral industrial applications
Measuring range-50 … +260 °C

Measuring element Pt100, Pt1000, NTC, KTY, Ni1000
Special feature■ Connection lead made of PVC, silicone,

PTFE■ Probe sleeve from stainless steel

IP65■ Protected against dust and water jets,

IP65■ With quick-mounting bracket

TE 67.14■ Smallest case design, UV-resistant

IP65■ Protected against dust and water jets,

Data sheet-40 … +100 °C

Measuring element Pt100, Pt1000, NTC

Data sheet

TE 67.15

Temperature ‫ ׀‬Transmit + Record

Temperature

Temperature transmitters

SIL

IEC 61508 / IEC 61511

T15

Digital temperature transmitter forresistance sensors

T16

Digital temperature transmitter forthermocouples

NAMUR

T38

HART® temperature transmitter

Input

Universal use for 1 or 2 sensors:resistance thermometers (up to 2 x
3-wire), thermocouples, potentiometers,
reed chains

Input

Resistance thermometers, potentiometers

Input

All commercially available thermocouples

Accuracy< 0.1 %

Accuracy

Typical < 2 K

Output4 … 20 mA

Output4 … 20 mA

Special feature

The fastest and simplest configuration on
the market

Special feature

The fastest and simplest configuration on
the market

Accuracy
Output4 … 20 mA, HART® protocol

TE 15.01

Data sheet

TE 16.01

Special featureTÜV-certified SIL version (full assessment), True Drift Detection technology

Data sheet

TE 38.01

Data sheet

SIL

IEC 61508 / IEC 61511< 0.1 %

NAMUR

T32

HART® temperature transmitter

Input

Resistance thermometers, thermocouples,
potentiometers

Accuracy< 0.1 %

Output4 … 20 mA, HART® protocol

Special featureTÜV-certified SIL version (full assessment)

Data sheet

TE 32.04

Further information at www.wika.com

T91

Analogue temperature transmitter 3-wire, 0 … 10 V
Input

Resistance thermometers, thermocouples

Accuracy< 0.5 or < 1 %Output0 … 10 V, 0 … 5 V

Special feature

Fixed measuring range

Data sheet

TE 91.01, TE 91.02

TIF50, TIF52

HART® field temperaturetransmitter
Input

Resistance thermometers, thermocouples,
potentiometers

Accuracy< 0.1 %

Output4 … 20 mA, HART® protocol

Special feature

PC configurable

Data sheet

TE 62.01

47

Temperature ‫ ׀‬Switch

Temperature switches
Temperature switches for industrial applications

TSD-30Electronic temperature switch
with display

TFS35

Bimetal temperature switch
for switching voltages to 48 V

TFS135

Bimetal temperature switch
for switching voltages to
250 V

Measuring range-20 … +80 °C, -20 … +120 °C,
0 … 150 °C

Switching tempera- 50 … 155 °C, fixedture

Switching tempera- 50 … 130 °C, fixedture

Output■ Switching outputs PNP or NPN

Special featureSpecial feature■ 4 … 20 mA■ IO-Link 1.1

TE 67.03■ Compact version: Normally closed (NC),normally open (NO)■ Electr. connection via plug connection■ 0 … 10 V

Data sheet■ Switching voltage to AC 48 V, DC 24 V

Data sheet■ Switching voltages to AC 250 V■ Contact version normally closed (NC)■ Electr. connection via plug connection■ 1 or 2 switch contacts■ Option: With measuring element Pt1000

TV 35.01/ Pt100

Data sheet

TV 35.02

Temperature switches for the process industry

SIL

TXS, TXA

Mini temperature switches

48

SIL

SIL

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

TCS, TCA

Compact temperatureswitches

TWG, TAG

Heavy-duty version

Setting range-15 … +20 to 180 … 250 °C

Setting range-30 … +10 to 160 … 250 °C

Setting range-30 … +70 to 0 … 600 °C

Ignition protection
type

Ex ia or Ex d

Ignition protection
type

Ex ia or Ex d

Ignition protection
type

Ex ia or Ex d

Switch

1 x SPDT

Switch

1 x SPDT or 1 x DPDT

Switch1 or 2 SPDT or 1x DPDT

Switching power

AC 220 V/5 A
DC 24 V/5 A

Switching power

AC 250 V/15 A
DC 24 V/2 A

Switching power

AC 250 V/20 A
DC 24 V/2 A

Data sheet

TV 31.70, TV 31.72

Data sheet

TV 31.64, TV 31.65

Data sheet

TV 31.60, TV 31.61

Temperature ‫ ׀‬Switch

SC15

Expansion thermometerwith microswitch, indicating
temperature controller

SB15

Expansion thermometerwith microswitch, safety temperature
limiter

Nominal size

60, 80, 100 mm
72 x 72, 96 x 96 mm

Nominal size

60, 80, 100 mm
72 x 72, 96 x 96 mm

Scale range-100 … +400 °C

Scale range

0 … 400 °C

Wetted parts

Copper alloy

Wetted parts

Copper alloy

OptionSheet steel version

OptionSheet steel version

Data sheet

TV 28.02

Data sheet

TV 28.03

TGS55

Bimetal thermometer,
stainless steel version
Nominal size

100 mmScale range-70 … +30 to 0 … 600 °C

Wetted parts

Stainless steel

OptionData sheet

TGS73

Gas-actuated thermometer,
stainless steel version

70 with 8xx

Expansion thermometerwith microswitch

Nominal size

100, 160 mm

Nominal size

100 mmScale range-200 … +100 to 0 … 700 °C

Scale range-60 … +40 to 0 … 250 °C

Wetted parts

Stainless steel

Wetted parts

Stainless steel

Liquid damping to max. 250 °C (case andprobe)

Option■ Capillary

OptionVarious contact versions

TV 25.01

Data sheet

TV 28.01

Data sheet

Further information at www.wika.com

Temperature

Thermometers with switch contacts■ Liquid damping (case)TV 27.01

49

Temperature ‫ ׀‬Switch ‫ ׀‬Accessories

Temperature controllers

CS4R

For rail mounting, 22.5 x 75 mm

CS6S, CS6H, CS6L

For panel mounting, 48 x 48,
48 x 96, 96 x 96 mm

Input

Multi-function input for resistance thermometers, thermocouples and standard
signals

Input

Multi-function input for resistance thermometers, thermocouples and standard
signals

Control mode

PID, PI, PD, P, ON/OFF (configurable)

Control mode

PID, PI, PD, P, ON/OFF (configurable)

Monitoring output

Relay or logic level DC 0/12 V to control anelectronic switch relay (SSR) or analogue
current signal 4 … 20 mA

Monitoring output

Auxiliary power■ AC 100 … 240 V

Relay (AC 250 V, 3A, (R) or 1A (L)) or
logic level DC 0/12 V for 3-point control tocontrol an electronic switch relay (SSR) or
analogue current signal 4 … 20 mA

Auxiliary power■ AC 100 … 240 V

Data sheet

AC 85.05
Data sheet

AC 85.08■ AC/DC 24 V

■ AC/DC 24 V

Accessories

TND

Display forT38 temperature transmitter

Installation rods

PP82

Purge-gas control panel

Input

4 … 20 mA■ For installation of high-temperature thermocouples■ Heavy-duty stainless steel version

Dimensions∅ 44 mm■ Suited for horizontal and vertical installation■ High mechanical stability through side protection

Scale range

13.5 x 28 mm■ Can be used with flanged thermocouples■ For wall and pipe mounting, 2"

Special feature■ 5-digit main measured value display■ Combines high mechanical stability and low weight■ Pressure gauge with liquid damping

Data sheet

50

IR80

■ 2 variants: with clip-on adapter or

remote for mounting in the connection
head cover

TE 38.01■ Data sheet AC 80.19

Temperature ‫ ׀‬Accessories

Temperature

Accessories
PU-548Programming unit for temperaturetransmitters■ LED status display■ Compact design■ No further voltage supply needed, neither for the program-ming unit nor for the transmitter■ Due to the magWIK quick connector, fast connection to thetransmitter possible■ Data sheet AC 80.18

magWIK

Magnetic quick connector

905

Contact protection relay
for model 821 switch contacts

904

Control unit for
inductive contacts■ For accelerated connection for all configuration and calibra-

Application

For optimal contact protection and highest
switching reliability

Application

For operating measuring instruments withinductive contacts■ Connection of 2-mm plug contacts or

Data sheet

AC 08.05

Data sheet

AC 08.04

tion processes

4-mm plug contacts with adapter■ Data sheet AC 80.15

Coupler connector

Further information at www.wika.com

Fittings

Wires and cables

51Temperature ‫ ׀‬Thermowells / Protection tubes

Thermowells / Protection tubes

Whether in aggressive or abrasive process media, whether inhigh- or low-temperature ranges: For electrical or mechanicalthermometers, to prevent direct exposure of their temperatureprobes to the medium, thermowells / protection tubes that suiteach application are available. Thermowells / Protection tubes
can be machined from bar stock material or assembled from tube
sections and can either be screw-, weld- or flange-fitted.
They are offered in standard and special materials such as
stainless steel 1.4571, 316L, Hastelloy® or titanium. Each version,
depending on its construction type and its mounting to theprocess, has certain advantages and drawbacks with respect to its
load limits and the special materials that can be used.

TW10

Thermowell with flange

Thus, only the wetted parts of the thermowell / protection tube
are manufactured from special materials, whereas the non-wetted flange is made of stainless steel and is welded to the special
material.
This design is used both for protection tubes and thermowells.
With tantalum as special material a removable jacket is used,which is slid over the supporting thermowell / protection tube fromstainless steel.

TW15

Threaded thermowellTW20

Weld-in thermowell for welding socket

Thermowell form

Tapered, straight or stepped

Thermowell form

Tapered, straight or stepped

Thermowell form

Tapered, straight or stepped

Nominal width■ ASME 1 … 4 inch DIN/EN

Head design

Hexagon, round with hexagon, or
round with spanner flats

Welding diameter

1.050, 1.315 or 1.900 inch
(26.7, 33.4 or 48.3 mm)

Pressure rating

ASME to 2,500 lb (DIN/EN to PN 100)

Process connection ½, ¾ or 1 NPT

Pressure rating

3,000 or 6,000 psi

Data sheet

TW 95.10, TW 95.11, TW 95.12

Data sheet

Data sheet

TW 95.20■ DN 25 … 100

TW25

Weld-in thermowell52

In order to manufacture thermowells / protection tubes for flange
mounting at low cost from special materials, the designs useddiffer from standard thermowells/protection tubes in accordancewith DIN 43772.

TW 95.15

TW30

Vanstone, thermowell for lap flanges

Thermowell form

Tapered, straight or stepped

Thermowell form

Tapered, straight or stepped

Bar diameter

Up to 2 inch (50.8 mm)

Nominal width

ASME 1, 1½ or 2 inch

Data sheet

TW 95.25

Pressure rating
Data sheet

TW31

Vanstone designin accordance with petrochemical
standard
Thermowell form

In accordance with Shell drawings
S38.113 and S38.114

ASME up to 2,500 lbs

Material

Stainless steel, special alloys

TW 95.30

FlangeSlip-on flanges per ASME B16.5

Data sheet

TW 95.31

Temperature

Temperature ‫ ׀‬Thermowells / Protection tubes

ScrutonWell®

Thermowells
in ScrutonWell® design
Thermowell form

Bar stock material or with welded-on helix

Process connection Flange, threaded or weld-in

TW35

Threaded protection
tube (DIN 43772 form 2, 2G, 3, 3G)

TW40

Protection tube with flange(DIN 43772 form 2F, 3F)

Protection tube form Form 2, 2G, 3 or 3G

Protection tube form Form 2F or 3F

Material

Stainless steel

Nominal width

Connection tothermometerM24 x 1.5 rotatable

Data sheet

TW 95.35

TW50

Threaded thermowell (DIN 43772 form
6, 7, 9)■ DIN/EN DN 25 … 50
■ ASME 1 … 2 inch

Pressure rating■ DIN/EN up to PN 100

Data sheet

TW 95.40

Stainless steel or special materials

Data sheet

SP 05.16

TW45

Threaded protection tube (DIN 43772
form 5, 8)
Protection tube form Form 5 or 8
Material

Stainless steel

Data sheet

TW 95.45■ ASME up to 1,500 psig

TW55

Thermowell for weld-in orwith flange (DIN 43772 form 4, 4F)

Thermowell form

Form 6, 7 or 9

Thermowell form

Form 4 or 4F

Data sheet

TW 95.50

Nominal width■ DIN/EN DN 25 … 50

Pressure rating■ DIN/EN up to PN 100

Data sheet

TW 95.55

Further information at www.wika.com

Material■ ASME 1 … 2 inch

■ ASME up to 2,500 psig

SWT52G, SWT52S

Protection tube for model 52

Connection tothermometer

Suitable for thermometers with plain
connection (without thread), collar ∅ 18mm, stem 8 and 13 mmProtection tube
material

Copper alloy, St35 or stainless steel

Process connection G ½ B thread
Max. processtemperature, process pressure■ 160 °C with copper alloy as protection

Data sheet

TW 90.11

tube material (6 bar stat.)■ 500 °C with St35 stainless steel asprotection tube material (25 bar stat.)53

Level ‫ ׀‬Display

Bypass level indicators
Continuous level measurement via visual indication of the level without auxiliary power
Applications

 Continuous level indication without auxiliary power Indication of the level proportional to height Individual design and corrosion-resistant materials make theproducts suitable for a broad range of applications

 Chemical industry, petrochemical industry, oil and natural gas

extraction (on- and offshore), shipbuilding, machine building,power generating equipment, power plants
 Process water and drinking water treatment, food and beverageindustry, pharmaceutical industry
Special features

 Process- and procedure-specific production
 Operating limits: □ Operating temperature: T = -196 … +450 °C□ Operating pressure: P = vacuum to 400 bar1)□ Limit density: ρ ≥ 340 kg/m3

 Wide variety of different process connections and materials
 Mounting of level transmitters and magnetic switches possible

as an option

 Explosion-protected versions1) Individual limit values. For application limits, the joint consideration of temperatureand pressure is required.

BNA-SStandard version

Chamber■ ∅ 60.3 x 2 mmMaterial■ Stainless steel 1.4571/316Ti■ ∅ 60.3 x 2.77 mm

High-pressure version

ChamberPressure

Max. 100 bar

Temperature-196 … +450 °C

Data sheet

LM 10.01■ Ø 60.3 x 3.91 mm■ Ø 60.3 x 5.54 mm

BNA-X

Special materials

Chamber∅ 60.3 x 2 mm
∅ 60.3 x 2.77 mm∅ 60.3 x 3.91 mm∅ 60.3 x 5.54 mm

Material■ Titanium 3.7035■ Ø 73 x 7.01 mm■ Ø 76.1 x 5 mm

■ 1.4401/1.4404 (316/316L)

Process connection ■ Flange DIN, ANSI, EN■ Thread■ Weld stub

54

BNA-H

■ Ø 71 x 7.5 mm■ Ø 76 x 10 mmMaterial■ 1.4401/1.4404 (316/316L)

Process connection ■ Flange DIN, ANSI, EN■ Thread■ Weld stubPressure

Max. 385 bar

Temperature-196 … +450 °C

Data sheet

LM 10.01■ Hastelloy C276
■ 6Mo 1.4547■ Monel■ Inconel

Process connection ■ Flange DIN, ANSI, EN■ Thread■ Weld stubPressure

Max. 250 bar

Temperature-196 … +450 °C

Data sheet

LM 10.01

LevelLevel ‫ ׀‬Display

BNA-P

Plastic version

Chamber

Ø 60.3 x 3 mmMaterial■ PVDF

BNA-L

Liquid/KOplus version

Chamber■ ∅ 88.9 x 2 mm

■ ∅ 88.9 x 2.9 mm

BNA-SD, BNA-HD DUplus
Standard/High-pressure
versionChamber■ 114 x 2

■ PP

■ 114 x 3.6■ 114 x 4.5

Process connection Flange DIN, ANSI, EN■ 114 x 6.3

Pressure

Max. 6 bar

Temperature-10 … +100 °C

Material

Data sheet

LM 10.01

Process connection ■ Flange DIN, ANSI, EN■ Thread■ Weld stub1.4401/1.4404 (316/316L)■ BNA-SD: ∅ 60.3 x 2 mm

∅ 60.3 x 2.77 mm■ BNA-HD: ∅ 60.3 x 3.91 mm

Material■ 1.4401/1.4404 (316/316L)

Process connection ■ Flange DIN, ANSI, EN■ Thread■ Weld stubPressure■ BNA-SD: max. 100 bar■ BNA-HD: max. 160 bar

Pressure

Max. 63 bar

Temperature-196 … +450 °C

Temperature-196 … +450 °C

Data sheet

LM 10.01

Data sheet

LM 10.01

Accessories for bypass level indicators

BLR

Reed level transmitter

BMD

Magnetic indication

Material

Stainless steel

Material

Meter runMax. 6,000 mm

Indication elements Plastic rollers, stainless steel flaps

Temperature-100 … +350 °C, depending on version

Cover

Polycarbonate, glass

Output signal

4 … 20 mA, HART®, PROFIBUS® PA

Length180 … 6,000 mm

Temperature-200 … +450 °C

Data sheet

LM 10.03

Data sheet

or FOUNDATION™ Fieldbus
LM 10.03

Further information at www.wika.com

Aluminium, anodised, stainless steel

BFT

Float

Material

Stainless steel, titanium,
various special materials

Pressure

To 450 bar

Temperature-200 … +450 °C

Density> 340 kg/m³

Data sheet

LM 10.02

55

Level ‫ ׀‬Display

Accessories for bypass
Combines the tried-and-trusted bypass with further independent measurement principles

BLM-SI, BLM-SD

Magnetostrictive level transmitter,
intrinsically safe (Ex i)

BLM-SF-FM

Magnetostrictive leveltransmitter with FM approval

Top-mounted level indicator

Chamber■ ∅ 42.4 x 2 mm (standard)

Material

Stainless steel 1.4404

Material

Stainless steel

Guide tube lengthMax. 5,800 mm

Guide tube lengthMax. 4,000 mm

Temperature-60 … +185 °C

Temperature-200 … +180 °C

Output signal

4 … 20 mA, HART®

Output signal

4 … 20 mA, HART®

Material

Data sheet

LM 10.05

Data sheet

LM 10.05(316/316L)
Process connection Flange DIN, ANSI, EN

SIL

SIL

IEC 61508 / IEC 61511

IEC 61508 / IEC 61511

BLM-TA

High-temperature version

BLM-TAI

High-temperature version,
intrinsically safe

Material

Stainless steel

Material

Guide tube lengthMax. 6,000 mm

Guide tube lengthMax. 6,000 mm

Temperature■ -40 ... +125 °C

Temperature■ -40 ... +125 °C■ -90 ... +125 °C■ -45 ... +250 °C

Output signal
Data sheet

56

UTN

■ -45 ... +450 °C

Stainless steel■ -40 ... +250 °C■ -40 ... +450 °C

4 ... 20 mA, HART® v6

Output signal

4 ... 20 mA, HART® v6

LM 10.05

Data sheet

LM 10.05■ ∅ 42.2 x 2.77 mm■ ∅ 60.3 x 2 mm

■ ∅ 60.3 x 2.77 mm■ Stainless steel 1.4571/316Ti■ Stainless steel 1.4401/1.4404

Pressure

Max. 40 bar

Temperature-196 … +300 °C

Data sheet

LM 11.02

Level ‫ ׀‬Display

The external chamber model BZG consists of an external chamber vessel that is mounted laterally to a container using at least 2
process connections (flange, thread or weld stub). Through thistype of arrangement, the level in the external chamber vessel cor-

Level

External chambers
responds to the level in the container. The level is measured by ameasuring instrument inserted additionally in the external chamber
vessel, for example model FLR or FLS, or by a guided wave radar.

Applications

 Level detection for almost all liquid media
 Individual design and corrosion-resistant materials make theproducts suitable for a broad range of applications

 Chemical industry, petrochemical industry, oil and natural gas

extraction (on- and offshore), shipbuilding, machine building,power generating equipment, power plants

Special features
Process- and procedure-specific production
Operating limits: □ Operating temperature: T = -196 … +450 °C□ Operating pressure: P = Vacuum to 400 bar 1)
 Wide variety of different process connections and materials
 Mounting of level transmitters and guided wave radars possible
as an option1) Individual limit values. For application limits, the joint consideration of temperatureand pressure is required.

BZG-H

External chamber,high-pressure version

Material■ Stainless steel 1.4571 (316Ti)■ Stainless steel 1.4401/1.4404

BZG-K

External chamber,steel version

Material■ Steel 1.0345/1.0460■ Steel 1.5415 (16Mo3)

External chamber,standard version

Material

Process connection Flange■ DIN EN 1092-1
DN 10 … DN 50,
PN 16 … PN 400■ DIN DN 10 … DN 50, PN 16 … PN 400■ ANSI B16.5 ½" … 4", class 150 … 2,500
Pressure

Max. 255 bar (material-dependent)

Pressure

400 barTemperature-10 … +425 °C (material-dependent)

Temperature-196 … +450 °C

Data sheet

LM 11.01

Data sheet

LM 11.01■ Stainless steel 1.4401/1.4404

Process connection Flange■ DIN EN 1092-1 DN 10 … DN 100,
PN 6 … PN 63■ DIN DN 10 … DN 100, PN 6 … PN 64■ ANSI B16.5 ½" … 4", class 150 … 600
Pressure

64 barTemperature-196 … +450 °C

Data sheet

LM 11.01

BZG-X

External chamber,special material version

Material■ Stainless steel 6Mo 1.4547(UNS S31254)■ Stainless steel 1.4306 (304L)■ A350 LF2/A333 Gr. 6

Process connection Flange■ DIN EN 1092-1
DN 10 … DN 100,
PN 100 … PN 400■ DIN DN 10 … DN 100,
PN 100 … PN 400■ ANSI B16.5 ½" … 4",class 600 … 2,500■ Stainless steel 1.4571 (316Ti)(316/316L)■ A105/A106 Gr. B(316/316L)

Further information at www.wika.com

BZG-S■ Duplex 1.4462 (UNS S31803)■ Super Duplex 1.4410 (UNS S3850)■ Titanium 3.7035 (grade 2)■ Hastelloy C276 (2.4819)

Processconnection

Flange■ DIN EN 1092-1 DN 10 … DN 100,
PN 63 … PN 400■ DIN DN 10 … DN 100,
PN 64 … PN 400■ ANSI B16.5 ½" … 4", class 600 … 2,500

Pressure

Max. 430 bar (material-dependent)

Temperature-196 … +450 °C (material-dependent)

Data sheet

LM 11.01

57

Level ‫ ׀‬Display

Glass level gauges
Direct level indication without auxiliary power
Applications

 Continuous level indication without auxiliary power
 Direct indication of the level Individual design and corrosion-resistant materials make theproducts suitable for a broad range of applications

 Chemical industry, petrochemical industry, oil and natural gas

extraction (on- and offshore), shipbuilding, machine building,power generating equipment, power plants
 Oil and gas, heat transfer and refrigeration systems, plants forcryogenics
Special features

 Process- and procedure-specific production
 Operating limits: □ Operating temperature:

T = -196 … +374 °C 1)□ Operating pressure: vacuum to 250 bar 1)
 Wide variety of different process connections and materials
 Illumination optional
 Heating and/or insulation optional
1) Individual limit values. For application limits, the joint consideration of temperatureand pressure is required.

LGG-E

Compact version

Carbon-Line version

LGG-RE, LGG-TE
Standard version

Type of indication

ReflexType of indication

Reflex/transparent

Type of indication

Reflex/transparent

Material■ Steel 1.0460

Material

Steel A350 LF2Material■ Steel 1.0570, A350 LF2■ A105, 1.0570

Process connection Flange DIN, ANSI, EN

58

LGG-RP, LGG-TP

Process connection ■ Flange DIN, ANSI, EN■ Male thread ½" NPT, ¾" NPT■ Weld stub ½", ¾"

Pressure

Max. 40 bar

Temperature-10 … +243 °C (steam)Pressure

Max. 100 bar

Glass size

2 … 11

Temperature■ -40 … +243 °C (steam)

Number of segments

1…3

Data sheet

LM 33.01■ -40 … +300 °C■ Stainless steel 1.4404/316L

Process connection ■ Flange DIN, ANSI, EN■ Male thread ½" NPT, ¾" NPT■ Weld stub ½", ¾"
Pressure

Max. 160 bar

Temperature■ -196 … +243 °C (steam)■ -196 … +300 °C

Glass size

4…9

Number of segments

1…5

Glass size

2 … 11

Data sheet

LM 33.01

Number of segments

1 … 5 (others on request)

Data sheet

LM 33.01

LevelLevel ‫ ׀‬Display

LGG-RI, LGG-TI

High-pressure version

LGG-M

Refraction version

Type of indication

Reflex/transparent

Type of indication

Refraction

Material■ Steel 1.5415

Material

Steel 1.5415■ Stainless steel 1.4404/316L

Process connection ■ Flange DIN, ANSI, EN■ Male thread ½" NPT, ¾" NPT■ Weld stub ½", ¾"
Pressure

Max. 250 bar

Temperature-196 … +100 °C

Glass size

2…9

Number of segments

1…5

Data sheet

LM 33.01

Further information at www.wika.com

Process connection ■ Flange DIN, ANSI, EN■ Male thread G ½, G ¾, ½" NPT, ¾" NPT■ Weld stub ½", ¾"
Pressure

Max. 250 bar

Temperature-10 … +374 °C

Glass size

2 … 11

Number of segments

1…9

Data sheet

LM 33.01

59

Level ‫ ׀‬Transmit

Submersible pressure
sensors
Hydrostatic level measurement

Applications

 Level measurement in rivers and lakes
 Control of sewage lift and pumping

stations

 Monitoring of sewage, settling andrainwater retention basins Level measurement in vessel and stor-

age systems for oils and fuelsSpecial features

LS-1000

For general applications

 Slimline and hermetically sealed design

Accuracy(± % of span)Measuring range

0 ... 0.1 to 0 ... 1 bar
0 ... 1.25 to 0 ... 2 bar abs.

Special feature■ Levels from 1 to 10 m

to 300 m water column
Highly resistant versions available
Explosion protection per ATEX, IECEx,
FM and CSA
Drinking water conformity per KTW and
ACS
Temperature output, HART® andlow-power output signal for battery
operation

LF-1

For demanding applications
Accuracy(± % of span)
Measuring range
Special feature

Data sheet

60

GL≤ 0.5 or ≤ 1
0 … 0.1 to 0 … 6 bar
0 … 1.6 to 0 … 6 bar abs.

LM 40.04

For applications in hazardous areas

Accuracy(± % of span)■ ≤ 0.5 or ≤ 0.25 (only applies to

Measuring range

0 … 0.1 to 0 … 25 bar

Special feature■ Suitable for all level measurements in

Data sheet

LM 40.05

LH-10

High-performance

Accuracy(± % of span)

Measuring range: < 0.25 bar: ≤ ±0.50 %Measuring ranges: ≥ 0.25 bar: ≤ ±0.25 %Measuring range■ 0 … 0.1 to 0 … 25 bar

Special feature■ Precise and reliable■ Integrated temperature measurement(option)■ Design from Hastelloy and FEP cablefor especially high resistance (optional)

Data sheet

PE 81.09

measuring ranges ≥ 0.25 bar (3.6 psi))hazardous areas■ Explosion protection per IECEx, ATEX

■ Permanently leak-tightand CSA■ Low total cost of ownership■ Suitable for measurements incontaminated and aggressive media■ An optimised discharge behaviour
and a large pressure port prevent the
instrument from clogging and ensure aminimum maintenance effort■ Can be used in explosion-protectedareas■ Developed for wireless applications≤ 0.5 or ≤ 1

IL-10■ Shipbuilding approval in accordancewith GL

Data sheet

PE 81.23

Level ‫ ׀‬Transmit

Level

Continuous measurement with float forindustrial applications
With reed measuring chain
Applications

 Level measurements of liquids in machine building Control and monitoring tasks for hydraulic power packs,

compressors and cooling systems

Special features

 Media compatibility: Oil, water, diesel, refrigerants and otherliquids

 Permissible medium temperature: -30 … +120 °C
 Output signals for level and temperature (optional) as resist-ance output signal or 4 ... 20 mA current output

 Accuracy, resolution: 24, 12, 10, 6 or 3 mm

RLT-1000

Stainless steel version

RLT-2000

Plastic version

RLT-3000

Stainless steel version withtemperature output signal

Accuracy24, 20, 12, 10, 6 or 3 mm

Accuracy24, 20, 12, 10, 6 or 3 mm

Accuracy24, 20, 12, 10, 6 or 3 mm

Output signal■ Resistance signal

Output signal■ Resistance signal

Level output signal■ 4 ... 20 mA, 2-wire■ 4 ... 20 mA, 2-wire■ 0 ... 5 V, 3-wireTemperature■ 0 ... 10 V, 3-wire■ 0 ... 5 V, 3-wire-30 … +80 °C(-30 … +120 °C optional)

Temperature

Guide tube length

150 … 1,500 mm

Data sheet

LM 50.02

Further information at www.wika.com■ 4 ... 20 mA, 2-wire■ 0 ... 10 V, 3-wire■ 0 ... 5 V, 3-wire■ 0 ... 10 V, 3-wire-10 … +80 °C(-30 … +120 °C optional)

Output signal
Temperature

Pt100 or Pt1000

Guide tube length

150 … 1,500 mm

Temperature-30 … +100 °C

Data sheet

LM 50.01

Guide tube length

150 … 1,500 mm

Data sheet

LM 50.05

61

Level ‫ ׀‬Transmit

Continuous measurement with float forthe process industry
Magnetostrictive
Applications

FLM-S High-accuracy level detection for almost all liquid media
 Chemical industry, petrochemical industry, natural gas,

offshore, shipbuilding, machine building, power generating
equipment, power plants
 Process water and drinking water treatment, food and beverageindustry, pharmaceutical industry
Special features
 Process- and procedure-specific solutions possible
 Operating limits:□ Operating temperature: T = -90 … +450 °C□ Operating pressure: P = vacuum to 100 bar□ Limit density: ρ ≥ 400 kg/m3
 Resolution < 0.1 mm
 Wide variety of different electrical connections, processconnections and materials
 Explosion-protected versions

Stainless steel version

Process connection ■ Mounting thread■ Flange: DIN, ANSI
Guide tube lengthMax. 6,000 mm

Pressure

0 … 200 bar

Temperature-90 … +450 °C

Density≥ 400 kg/m³

Ingress protection

IP66/68 per IEC/EN 60529

Data sheet

LM 20.01

SIL

IEC 61508 / IEC 61511

FLM-CAI

Compact version, intrinsically safe

62

FLM-CM

Compact version forindustrial applications

Process connection ■ Mounting thread downwards- G ½" … G 2"
- NPT ½" … NPT 2"■ Mounting flange- ANSI ½" … 2 ½", class 150 … 600- EN DN 20 … DN 65, PN 6 … PN 100- DIN DN 20 … DN 65, PN 6 … PN 100

Process connection ■ Mounting thread downwards- G ½" … G 2"
- NPT ½" … NPT 2"

Guide tube length

Density≥ 680 kg/m3

Ingress protection

IP68 per IEC/EN 60529

Data sheet

LM 20.05■ 100 … 1,000 mm (Ø 6 mm guide tube)■ 100 … 3,000 mm (Ø 12 mm guide tube)

Guide tube length

100 … 1,000 mm (Ø 6 mm guide tube)

Pressure

Vacuum to 40 bar

Temperature-40 … +125 °C

FLM-CA

Compact version for process applications
Process connection ■ Mounting thread downwards- G ½" … G 2"
- NPT ½" … NPT 2"■ Mounting flange- ANSI ½" … 2 ½", class 150 … 600- EN DN 20 … DN 65, PN 6 … PN 100- DIN DN 20 … DN 65, PN 6 … PN 100
Guide tube length■ 100 … 1,000 mm (Ø 6 mm guide tube)■ 100 … 3,000 mm (Ø 12 mm guide tube)

Pressure

Vacuum to 40 bar

Pressure

Vacuum to 40 bar

Temperature-40 … +250 °C

Temperature-40 … +250 °C

Density≥ 580 kg/m3

Density≥ 580 kg/m3

Ingress protection

IP68 per IEC/EN 60529

Ingress protection

IP68 per IEC/EN 60529

Data sheet

LM 20.04

Data sheet

LM 20.04

LevelLevel ‫ ׀‬Transmit

®

FLM-P

FLM-H

Plastic version

Hygienic version, forsanitary applications

Process connection ■ Mounting thread■ Flange DIN, ANSI
Guide tube lengthMax. 5,000 mm

Pressure

0 … 16 bar

Temperature-10 … +100 °C

Density≥ 800 kg/m³

Process connection ■ Clamp ISO 2852
■ Clamp DIN 32767■ Aseptic thread DIN 11864-1■ Aseptic liner DIN 11864-1■ Aseptic flange DIN 11864-2■ Aseptic clamp DIN 11864-3■ VARIVENT®
■ BioConnect®

Ingress protection

IP68 per IEC/EN 60529

Material

1.4435 (316L) or 1.4404 (316L)

Data sheet

LM 20.01

Guide tube lengthMax. 6,000 mm

Pressure

10 barTemperature-40 … +250 °C

Density≥ 770 kg/m³

Data sheet

LM 20.01

SIL

IEC 61508 / IEC 61511

FLM-TAI

High-temperature version,
intrinsically safe
Process connection ■ Mounting thread downwards- G ½" … G 2"
- NPT ½" … NPT 2"■ Mounting flange- ANSI ½" … 2 ½",
class 150 … 600- EN DN 20 … DN 65,
PN 6 … PN 100- DIN DN 20 … DN 65,
PN 6 … PN 100
Guide tube length

100 … 3,000 mm (Ø 12 mm guide tube)

Pressure

Vacuum to 40 bar

Temperature-40 ... +450 °C

Density≥ 400 kg/m³

Output signal

4 ... 20 mA, HART® v6

Ingress protection

IP68 per IEC/EN 60529

Data sheet

LM 20.01

Further information at www.wika.com

63Level ‫ ׀‬Transmit

Continuous measurement with float forthe process industry
With reed measuring chain
Applications

 Level detection for almost all liquid media
 Chemical industry, petrochemical industry, natural gas,

offshore, shipbuilding, machine building, power generating
equipment, power plants
 Process water and drinking water treatment, food and beverageindustry, pharmaceutical industry
Special features

 Process- and procedure-specific solutions possible
 Operating limits: □ Operating temperature: T = -80 … +200 °C□ Operating pressure: P = vacuum to 80 bar□ Limit density: ρ ≥ 400 kg/m3

 Wide variety of different electrical connections, processconnections and materials

 Optionally with programmable and configurable head-mounted

transmitter for 4 … 20 mA field signals, HART®, PROFIBUS®
PA and FOUNDATION™ Fieldbus
 Explosion-protected versions

FLR-SA, FLR-SB

Stainless steel version

FLR-SAI, FLR-SBI
Intrinsically safe

Process connection ■ Mounting thread■ Flange DIN,
ANSI, EN

Process connection ■ Mounting thread■ Flange DIN,
ANSI, EN

Guide tube lengthMax. 6,000 mm

Guide tube lengthMax. 6,000 mm

Pressure

0 … 100 bar

Pressure

0 … 100 bar

Temperature-80 … +200 °C

Temperature-80 … +200 °C

Density≥ 400 kg/m³

Density≥ 400 kg/m³

Ingress protection

To IP66/IP68 per IEC/
EN 60529

Ingress protection

To IP66/IP68 per IEC/
EN 60529

LM 20.02

Data sheet

Data sheet

FLR-F

Reed level transmitterfor food applications
Process connection ■ Threaded pipe connection DIN11851, downwards, DN 50 ...
DN 150■ Clamp pipe connection
DIN 32676, DN 25 … DN 100
or 1" … 4"
■ Clamp pipe connection ISO 2852,
DN 25 ... DN 150■ Others on request
Guide tube length■ Max. 1,500 mm(guide tube diameter 12 mm)■ Max. 3,500 mm(guide tube diameter 14 mm)

LM 20.02■ Max. 6,000 mm(guide tube diameter 18 mm)

64

Pressure

0 ... 25 bar

Temperature

Normal temperature: -20 ... +120 °C
High temperature: +120 ... +200 °C
Low temperature: -80 ... -20 °C

Density≥ 400 kg/m³

Ingress protection

To IP66/IP68 per IEC/EN 60529

Data sheet

LM 20.06

LevelLevel ‫ ׀‬Transmit®

FLR-PA, FLR-PB

Plastic version, PP, PVDF, PP

Process connection ■ Mounting thread■ Flange DIN,
ANSI, EN

FLR-HA3

Hygienic version, forsanitary applications-10 … +100 °C

Process connection ■ Clamp ISO 2852
■ Clamp DIN 32767■ Aseptic thread DIN 11864-1■ Aseptic liner DIN 11864-1■ Aseptic flange DIN 11864-2■ Aseptic clamp DIN 11864-3■ VARIVENT®
■ BioConnect®

Density≥ 800 kg/m3

Material

1.4435 (316L) or 1.4404 (316L)

Data sheet

LM 20.02

Guide tube lengthMax. 6,000 mm

Pressure

10 barTemperature-40 … +250 °C

Density≥ 770 kg/m³

Ingress protection

To IP66/IP68 per IEC/EN 60529

Data sheet

LM 20.02

Guide tube lengthMax. 5,000 mm

Pressure

0 … 3 barTemperature

Further information at www.wika.com

65Level ‫ ׀‬Switch

Float switches for industrial applications

Applications

 Level measurement of liquids in machine building Control and monitoring tasks for hydraulic power packs, compressors and coolingsystems

Special features

 Media compatibility: Oil, water, diesel, refrigerants and other liquids Permissible medium temperature range: -30 ... +150 °C
 Up to 4 switching outputs freely definable as normally open, normally closed orchange-over contact

 Optional temperature output signal, selectable as preconfigured bimetal switch oreither Pt100 or Pt1000

RLS-1000

Stainless steel version

Switching output

Up to 4
(normally closed, normally open,change-over contact)

Medium temperature -30 … +80 °C(-30 … +150 °C optional)

66

Guide tube length

60 … 1,500 mm

Data sheet

LM 50.03

RLS-2000

Plastic version

Switching output

RLS-3000

Stainless steel version, withtemperature output signal

Up to 4
(normally closed, normally open, change-overcontact)

Switching output

Up to 3
(normally closed, normally open, change-overcontact)

Medium temperature -10 … +80 °C(-30 … +120 °C optional)

Temperature output

Guide tube length

70 … 1,500 mm

Data sheet

LM 50.04

Medium temperature -30 … +80 °C(-30 … +150 °C optional)

Normally closed, normallyopen, Pt100, Pt1000

Guide tube length

60 … 1,500 mm

Data sheet

LM 50.06

LevelLevel ‫ ׀‬Switch

T YPE
APPROVED

RLS-4000Intrinsic safety Ex i

RLS-5000

For the shipbuilding industry(bilge water tanks)

Switching output

Up to 4
(normally closed, normally open,change-over contact)

Switching output

Temperature output(optional)

Normally closed, normallyopen, Pt100, Pt1000

Medium temperature -40 … +80 °C

Medium temperature -30 … +80 °C(-30 … +150 °C optional)
Guide tube length

60 … 1,500 mm

Data sheet

LM 50.07

RLS-7000

Miniature design,vertical installation

Normally closed, normally open, change-overcontact

Density≥ 1,000 kg/m3

Optional

Guide tube length

150 … 1,000 mm

Data sheet

LM 50.08

Data sheet

LM 50.09

Medium temperature -10 … +60 °C

RLS-8000

Miniature design,
horizontal installation
Normally closed, normally open, changeover contact

Medium temperature-25 ... +80 °C(-25 ... 100 °C optional)

Medium temperature-25 ... +80 °C(-25 ... 100 °C optional)

Wetted material:■ Polypropylene (PP)

Wetted material:■ Polypropylene (PP)■ Polyamide PA12 (on request)

Further information at www.wika.com

Normally closed, normally open, change-overcontact

Test device

Switching output

LM 50.11

Switching output

Marine cable, IP68

Normally closed, normally open, changeover contact

Data sheet

For water and wastewaterElectrical output

Switching output■ Polyamide PA6.6

RLS-6000■ Polyamide PA6.6■ Polyamide PA12 (on request)

Data sheet

LM 50.1267

Level ‫ ׀‬Switch

Float switches for the process industry
Robust switches for liquid media
Applications

 Level detection for almost all liquid media
 Pump and level control and monitoring of distinct filling levels
 Chemical industry, petrochemical industry, natural gas,

offshore, shipbuilding, machine building, power generating
equipment, power plants
 Process water and drinking water treatment, food and beverage industry
Special features

 Large range of application due to the simple, proven functional

principle

 For harsh operating conditions, long service life Operating limits: □ Operating temperature: T = -50 … +350 °C□ Operating pressure: P = vacuum to 40 bar□ Limit density: ρ ≥ 300 kg/m³

 Wide variety of different electrical connections, processconnections and materials

 Explosion-protected versions

FLS-SA, FLS-SB

Stainless steel version,for vertical installation
Switch points

Max. 8 switch points

Process connection ■ Mounting thread■ Flange DIN, ANSI, EN

68

Guide tube lengthMax. 6,000 mm

Pressure

0 … 40 bar

Temperature-50 … +300 °C

Density≥ 390 kg/m³

Data sheet

LM 30.01

FLS-PA, FLS-PB

Plastic version, for vertical installation

Switch points

Max. 8 switch points

Process connection ■ Mounting thread■ Flange DIN, ANSI, EN
Guide tube lengthMax. 5,000 mm

Pressure

0 … 3 barTemperature-10 … +100 °C

Density≥ 400 kg/m³

Data sheet

LM 30.01

LevelLevel ‫ ׀‬Switch

ELS-S

For lateral mounting
with external chamber
External chamber

Stainless steel

ELS-A

For lateral mounting
with external chamber
External chamber

Aluminium

Process connection Threaded pipe connection GE10-LR
galvanised steel

Process connection Threaded pipe connection GE10-LR
galvanised steel

Pressure

To 6 barPressure

Max. 1 bar

Temperature-30 … +300 °C

Temperature-30 … +150 °C

Data sheet

LM 30.03

Data sheet

LM 30.03

HLS-P

Plastic version, for horizontal installation

HLS-S

Stainless steel version,for horizontal installation

Process connection Flange DIN, ANSI, EN

Process connection Flange DIN, ANSI, EN

Pressure

0 … 3 barPressure

0 … 232 bar

Temperature-10 … +80 °C

Temperature-196 … +350 °C

Density≥ 750 kg/m³

Density≥ 600 kg/m³

Material

PP

Material

Stainless steel, titanium

Data sheet

LM 30.02

Data sheet

LM 30.02

Further information at www.wika.com

HLS-M1,
HLS-M2

Plastic or stainless steel version, withcable outlet
Process connection ■ ½" NPT(installation in the tank from outside)■ G ¼" (installation from inside, PP
version)■ G E" (installation from inside, stainlesssteel version)
Pressure■ HLS-M1: 1 bar

Temperature■ HLS-M1: -10 … +80 °C

Material■ HLS-M1: PP

Electricalconnection■ HLS-M1: Cable

Data sheet

LM 30.06■ HLS-M2: 5 bar■ HLS-M2: -40 … +120 °C■ HLS-M2: Stainless steel 1.4301■ HLS-M2: Cable or connector

HLS-SBIEx i

Intrinsically safe stainless steel version
for horizontal installation
Process connection ■ Mounting flange:
DIN DN 50 … DN 100, PN 6 … 160
EN 1092 DN 50 … DN 100,
PN 6 … PN 160
ANSI 2" … 4", class 150 … 900■ Square flange:
DN 80 and DN 92(other flanges on request)
Pressure

0 ... 100 bar (180 bar on request)

Temperature class

T2

Process temperature

180 °C 160 °C 108 °C 80 °C

T3

T4

Ambient temperature at case

80 °C

Density

600 kg/m³

Material

Stainless steel 1.4571

Data sheet

LM 30.02

T5

T6
65 °C

69

Level ‫ ׀‬Switch

Optoelectronic switches for the process industry
Applications
Chemical industry, petrochemical industry, natural gas, offshore
Shipbuilding, machine building, refrigerator units
Power generating equipment, power plants
Process water and drinking water treatment
Wastewater and environmental engineering







Special features
Temperature ranges from -269 … +400 °C
Versions for pressure ranges from vacuum to 500 bar
Special versions: High pressure, interface measurement
Signal processing is made using a separate model OSA-S
switching amplifier






OLS-S, OLS-H

Standard and high-pressure
versionMaterial

Stainless steel, Hastelloy, KM-glass,quartz glass, sapphire, graphite

Process connection ■ G ½ A■ ½ NPT

70

Pressure

0 … 500 bar

Temperature-269 … +400 °C

Data sheet

LM 31.01

OSA-S

Switching amplifier,for models OLS-S, OLS-H

OLS-C20

Compact design, high-pressure version

Output1 signal relay, 1 failure relay

Material

FunctionHigh or low alarmTime delay

To 8 sVoltage supply

AC 24/115/120/230 V
DC 24 V

Process connection ■ M16 x 1.5■ G½A■ ½ NPT

Stainless steel, quartz glass

Data sheet

LM 31.01

Insertion length

24 mm

Pressure

0 … 50 bar

Temperature-30 … +135 °C

Data sheet

LM 31.02

Level ‫ ׀‬Switch

Applications






Special features
For liquids such as oils, water, distilled water, aqueous media
Compact design
Mounting position as required
Accuracy ±2 mm
No moving components

Limit detection of liquids
Machine tools
Hydraulics
Machine building
Water technology

OLS-C01

Standard version

Material

Stainless steel, borosilicate glass







OLS-C02

With selectable switch length

Material

Stainless steel, borosilicate glass

OLS-C05

High-temperature version

Material

Stainless steel, borosilicate glass

Process connection G ⅜", G ½" or M12 x 1

Process connection G ½"

Process connection G ½"

Pressure

Max. 25 bar

Pressure

Max. 25 bar

Pressure

Max. 25 bar

Temperature-30 … +100 °C

Temperature-30 … +100 °C

Temperature-40 … +170 °C

Switching output

1 x PNP

Switch length

65 … 1,500 mm

Switching output

1 x PNP

Data sheet

LM 31.31

Switching output

1 x PNP

Data sheet

LM 31.33

Data sheet

LM 31.32

Further information at www.wika.com

Level

Optoelectronic switches for industrial
applications

71

Level ‫ ׀‬Switch

Optoelectronic switches for industrial
applications

OLS-C51

Intrinsic safety Ex i

Material

Stainless steel, borosilicate glass

OLS-C04

For refrigeration
technology
Material

Steel, nickel-plated; melted glass

OLS-5200

For the shipbuilding
industry
Material

Process connection G ½", ½" NPT

Process connection Male thread G ½" or M18 x 1.5

Pressure

Max. 40 bar

Pressure

Max. 40 bar

Pressure

Max. 25 bar

Temperature-30 … +135 °C

Temperature-40 … +100 °C

Temperature-40 … +130 °C

Output signal

4 … 20 mA low/high as switching output

Switching output

1 x PNP

Switching output

1 x PNP

Data sheet

LM 31.04

Data sheet

LM 31.34

Vibration resistance 10 … 5,000 Hz, 0 … 60g
Data sheet

72

Stainless steel, borosilicate glass

Process connection G ½"

LM 31.06

Level ‫ ׀‬Accessories

TLS-S

TLS-C

Ex version

Material (wetted)

Compact version■ Stainless steel 1.4404 (316L)■ Stainless steel 1.4404, 1.4435 (316L),

electropolished■ Stainless steel 1.4404 (316L) with PFA
coating■ Stainless steel 1.4404 (316L) with
ECTFE coating■ Hastelloy C-276■ Hastelloy C-276, electropolished
Process connection ■ G ¾" ... 2"■ ¾" … 2" NPT■ DN 25 ... DN 100 per DIN EN 1092-1■ 1" ... 4" per ASME B16.5 / ASME BPE

Material (wetted)■ Stainless steel 1.4404 (316L)■ Stainless steel 1.4404, 1.4435 (316L),

Process connection ■ G ¾" ... 2"■ ¾" … 2" NPT■ 1" ... 4" per ASME BPE■ Stainless steel 1.4404, 1.4435 (316L),

electropolished■ Hastelloy C-276■ Hastelloy C-276, electropolished

Process connection ■ G ¾" ... 2"■ ¾" … 2" NPT■ 1" ... 4" per ASME BPE-1 … +64 bar

Temperature-40 … +150 °C

Temperature-40 … +150 °C

Density≥ 500 ... 2,500 kg/m³

Output signal■ SPST relay output

Data sheet

LM 30.10

Temperature-40 … +200 °C

Output signal■ SPST relay output

Data sheet

LM 30.10■ PNP transistor output■ Stainless steel 1.4404 (316L)

Pressure≥ 500 ... 2,500 kg/m³≥ 500 ... 2,500 kg/m³

Material (wetted)-1 … +64 bar

Density■ DPDT relay output

Hygienic design version

Pressure-1 … +100 bar

Output signal

TLS-H

electropolished■ Hastelloy C-276■ Hastelloy C-276, electropolished

PressureDensity

Level

Vibrating level switches■ PNP transistor output■ PNP transistor output■ NAMUR (8.2 V)

Data sheet

LM 30.10

Further information at www.wika.com

73Force ‫ ׀‬Force transducers

Compression force transducers

Compression force transducers are designed for determining
compression forces and are suitable for static and dynamic
measurements in the direct force flow. WIKA force transducers are
manufactured from stainless steel and other high-quality materials,
are robust and are notable for their reliability and high quality evenin complex applications. Our compression force transducers areavailable in different rated loads.
They cover a wide range of application areas: For instance, these

F1106, F1119, F1136

Hydraulic compression force
transducer, clamping force test
instrument up to 500 kN

F1102

Hydraulic compression force
transducer, welding tongs test
instrument up to 36 kN

F1103, F1112, F1122

Hydraulic compression force
transducer, 3-jaw clamping force
test instrument up to 1,000 kN

Rated force Fnom0 … 160 N to 0 … 500 kN

Rated force Fnom0 … 100 N to 0 … 36 kN

Rated force Fnom

0 ... 1.1 kN to 0 ... 1,000 kN

Relative linearity
error■ Analogue ≤ ±1.6 % Fnom
■ Digital ≤ ±0.5 % Fnom

Relative linearity
error■ Analogue ≤ ±1.6 % Fnom
■ Digital ≤ ±0.5 % Fnom

Relative linearity
error■ Analogue ≤ ±1.6 % Fnom

■ Analogue indication

Display■ Analogue indication

Display■ Analogue indication

Display■ Digital display■ Digital display■ Pressure sensor■ Pressure sensor

Ingress protection

IP65, digital IP67

Ingress protection

IP65, IP67

Data sheet

FO 52.13, FO 52.10, FO 52.27

Data sheet

FO 52.16

F1201

Compression force
transducer up to
36 kN
Rated load Fnom

74

force transducers are employed in machine building or in theautomation of plants to determine the pressing and joining forces,
as well as for detecting weight in many industrial applications.
You can select the pertinent technical and regional approvals as
options.

F1222

Miniature compression force
transducer from 10 N

0 … 5 t to 0 … 30 t

Rated force Fnom

Relative linearity
error≤ ±0.05 % Fnom

Relative linearity
error

Output signal

2.0 ±0.2 mV/V

Output signal

Ingress protection

IP68

Data sheet

FO 51.71■ Digital ≤ ±0.5 % Fnom
■ Digital display■ Pressure sensor

Ingress protection

IP65, IP67

Data sheet

FO 52.24, FO 52.25, FO 52.26

F1861

Compression force transducer
with bilateral spherical force
introduction up to 50 t

0 … 10 N to 0 … 5,000 N

Rated load Fnom0 … 10 t to 0 … 50 t

±1 % Fnom

Relative linearity
error≤ 0.03 % Fnom

±0.1 mV/V (10 N)±0.2 mV/V (20 N to 5 kN)

Output signal

2.0 ±0.2 mV/V

Ingress protection

IP67

Ingress protection

IP65

Data sheet

FO 51.61

Data sheet

FO 51.11

Force ‫ ׀‬Force transducers

WIKA offers tension/compression force transducers in different
designs and versions. They are available in miniature designs, as
traditional s-type, as transducers with different thread forms or
as low-profile force transducers. Transducers in miniature design
are used for small mounting spaces and also for detecting small
forces. The s-type with female thread, which is very well suitedfor this purpose, features a particularly high accuracy and is used

F2220, F2221

Miniature tension/compression
force transducer from 10 N

in rated load ranges of up to 50 kN. For measuring high forces,
tension/compression force transducers in compact design are thefirst choice. For low-profile force transducers, the force is transmitted via the centrical female thread. They are highly dynamic andpossess a high fatigue strength.

F2222

Tension/Compression
force transducer
up to 2,200 kN

F2226

Tension/Compression force
transducer, male thread up to
3,300 kN

Rated force Fnom0 … 10 N to 0 … 50 kN

Rated force Fnom

0 … 22 N up to 0 … 2,200 kN

Rated force Fnom

0 …10 kN to 0 … 3,300 kN

Relative linearity
error

from ±0.15 % Fnom

Relative linearity
error±0.1 % Fnom

Relative linearityerror■ ≤ ±0.15 % Fnom (≤ 200 kN)

Output signal

1.5 ±0.15 or 2.0 ±0.2 mV/V

Output signal■ ≤ 25 lb: 2 mV/V

Output signal

2 mV/V

Ingress protection

IP65

Ingress protection

IP66

Data sheet

FO 51.16, FO 51.26

Data sheet

FO 51.51

F2301, F23C1,
F23S1

Tension/Compression
force transducer withthin-film technology up to
500 kN

Data sheet

FO 51.29

F2802

Tension/Compression
force transducer, S-type
up to 50 kN

F2808

Tension/Compression force transducer
from 5 N

Rated force Fnom0 … 0.5 kN to 0 … 50 kN

Rated force Fnom

0 … 5 N to 0 … 2,000 N±0.5 % Fnom

■ Steel ±0.03 % Fnom

■ Stainless steel ±0.05 % Fnom

Relative linearity
error±0.15 % Fnom

■ 4 … 20 mA, 2-wire/3-wire

Output signal

2.0 ±5 % mV/V

Output signal

2.0 ±10 % mV/V

Ingress protection

IP65 (< 5 kN), IP67 (≥ 5 kN)

Ingress protection

IP66

Data sheet

FO 51.48

Data sheet

FO 51.68

0 … 1 kN to 0 … 500 kN

Relative linearity
error
Output signal■ 0 … 10 V, 3-wire■ CANopen®

Data sheet

IP66■ ≤ ±0.20 % Fnom (> 200 kN)

Relative linearity
error dlin

Rated force Fnom

Ingress protection■ > 50 lb: 3 mV/V

Ingress protection■ Redundant versions available

Force

Tension/Compression force transducers

IP66, IP67, IP68, IP69, IP69K
FO 51.17

Further information at www.wika.com

75Force ‫ ׀‬Bending/shear beams

Bending/Shear beams
Bending beams and shear beams are used for the determinationof (shear) forces and are suitable for both static (weighing
technology) and dynamic (machine building) measurement
projects. To determine how strong the force is in the application,strain gauges or thin-film sensors are used, which are attached on
or in the measuring body.

F3201,
F3831

Shear beam up to 10 t

F3203,
F3833

Bending beam up to 500 kg

Rated load Fnom

0 … 500 kg to 0 … 10,000 kg

Rated load Fnom

0 … 5 kg to 0 … 500 kg

Relative linearity
error

From ±0.017 % Fnom

Relative linearity
error

From ±0.017 % Fnom

Output signal

2.0 ±0.2 mV/V

Output signal

2.0 ±0.2 mV/V

Ingress protection

IP65, IP67, IP68, IP69K, depending on
versionIngress protection

IP68, IP69, depending on version

Data sheet

FO 51.22, FO 51.73

Data sheet

FO 51.21, FO 51.72

AZK02

Mounting kit
for shear beams F3201 and F3831
Data sheet

76

The application areas of the bending beam and shear beam aremany and varied. Thus, these load cells are very often used inindustrial weighing technology as well as in the areas of special
machine building, manufacturing automation and gravimetric levelmeasurement. In addition, they are used in the laboratory andprocess industry for the indirect determination of torques.

FO 51.21

AZK03

Mounting kit for bending beams
F3203 and F3833
Data sheet

FO 51.22

Force ‫ ׀‬Load cells

Force

Load cells
Load cells are designed as a special form of force transducers foruse in weighing equipment. They enable very high measurement
accuracies between 0.01 % and 0.05 % Fnom.
Typical and widely used load cell geometries are single point load

F4801

Single point load cellup to 250 kg

cells, bending and shear beam load cells, s-type load cells, pendulum load cells and compression force load cells. In addition, there
are corresponding mounting kits and complete weighing modules
available.

F4802

Single point load cellup to 10 kg

F4818

Single point load cellup to 500 kg

Rated load Fnom

0 … 3 to 0 … 250 kg

Rated load Fnom0 … 0.3 kg to 0 … 10 kg

Rated load Fnom

0 … 20 kg to 0 … 500 kg

Relative linearity
error

0.02 % Fnom

Relative linearity
error

0.02 % Fnom

Relative linearity
error

0.02 % Fnom

Output signal

2.0 ±10 % mV/V

Output signal■ 1.0 ±10 % mV/V (0.3 ... 0.5 kg)

Output signal

2.0 ±10 % mV/V

Ingress protection

IP65

Ingress protection

IP65

Data sheet

FO 53.10

Data sheet

FO 53.14

F4881

Load cellfor multihead weighers■ 2.0 ±10 % mV/V (1 ... 10 kg)

Ingress protection

IP65

Data sheet

FO 53.13

F4882, F4883,
F4884, F4885

Load cells for checkweighers

Rated load Fnom

0 … 2 kg to 0 … 30 kg

Rated load Fnom0 … 1 kg to 0 … 635 kg

Relative linearity
error

0.02 % Fnom

Relative linearity
error≤ 0.02 % Fnom

Output signal

2.0 ±0.2 mV/V

Output signal

2.0 ±0.2 mV/V

Ingress protection

IP67

Ingress protection

IP66 or IP67

Data sheet

FO 53.16

Data sheet

FO 53.17, FO 53.18, FO 53.19, FO 53.20

Further information at www.wika.com

77Force ‫ ׀‬Load pins

Load pins
Load pins represent one of the most important components formeasuring forces. Existing retention bolts can easily be replacedby these products in existing applications. The application areas
range from construction machinery and cranes to manufacturing
automation. These force transducers are often used by designers
because, due to their design, they can be directly integrated into
the force flow, without taking up space.

individual, the exact layout is important. With WIKA, you will havespecialists by your side who already have lots of experience inforce measurement.

Since the design requirements for the use of load pins are very

F5308, F53C8,F53S8F5301,
F53C1

F5802

Load pin, heavy-duty version,
thin-film technology from 10 kN

Load pin with thin-film technology
up to 200 kN

Load pin from 20 kN

Rated force Fnom

From 10 kN

Rated force Fnom

0 … 5 kN to 0 … 200 kN

Rated force Fnom

20 ... 10,000 kN

Relative linearity
error±1 % Fnom / ±1.5 % Fnom

Relative linearity
error±1 % Fnom

Relative linearity
error

0.5 % ... 1 % Fnom

Output signal■ 4 … 20 mA, 2-wire/3-wire

Output signal■ 4 … 20 mA, 2-wire/3-wire

Output signal■ 4 … 20 mA, 2-wire/3-wire■ 0 … 10 V, 3-wire■ CANopen® redundant versions available■ 0 … 10 V, 3-wire■ CANopen® redundant versions available■ 0 … 10 V, 3-wire■ CANopen® redundant versions available

Ingress protection■ Unplugged state IP66, IP67

Ingress protection■ Unplugged state IP66, IP67

Ingress protection■ Unplugged state IP66, IP67

Data sheet

FO 51.43

Data sheet

FO 51.18

Data sheet

FO 51.55■ Plugged-in state IP68, IP69, IP69K■ Plugged-in state IP68, IP69, IP69K■ Plugged-in state IP68, IP69, IP69K

Tension links
Large lifting equipment and cranes generally move high to very
high loads. In (container) ports, in offshore applications or on
construction sites, (failure) safety in the movement of goods andloads is important. Man and machine must be protected equally
and a smooth process must be ensured. Among other things,
when moving loads, tension links, which are placed directly in theforce flow, ensure safe operation in order to prevent overloading of
the machinery. These force transducers are available in very small
dimensions up to very large formats. Tension links from WIKA withproven thin-film technology guarantee maximum safety in theirapplication thanks to their first-rate quality.

78

F7301,
F73C1, F73S1

Tension link withthin-film technology from 5 kN
Rated force Fnom

From 0 … 5 kN

Relative linearity
error±0.5 % Fnom

Output signal■ 4 … 20 mA, 2-wire/3-wire■ 0 … 10 V, 3-wire■ CANopen® redundant versions available

Ingress protection

Unplugged state IP66, IP67
Plugged-in state IP68, IP69, IP69K

Data sheet

FO 51.19

Force ‫ ׀‬Force transducers

Force

Ring force transducers
These force transducers are extremely robust and are suitable forthe detection of very high (static) forces. Furthermore, they aresuitable for many installation situations. The ring geometry is usedin force measurement for a wide variety of spatial conditions. The
main fields of application are found in spindle presses, in screw
force measurement or even in geotechnology.
WIKA offers electrical and hydraulic ring force transducers in diameters from 12 millimetres up to 430 millimetres as well as in various
installation heights.
Discover our portfolio now.

F6212

F6215

Ring force transducer
up to 100 kN

Ring force transducer
up to 1,500 kN

Rated force Fnom

0 … 2 to 0 … 100 kN

Rated force Fnom0 … 15 to 0 … 1,500 kN

Relative linearity
error≤ 0.5 % Fnom

Relative linearity
error≤ ±1 % Fnom

Output signal

0.8 ... 1.2 ±0.1 mV/V

Output signal

0.8 ... 1.2 ±0.1 mV/V

Ingress protection

IP65

Ingress protection

IP65

Data sheet

FO 51.27

Data sheet

FO 51.28

F6116

F6154

Hydraulic ring force transducer
up to 120 kN

Hydraulic ring force transducer,
heavy-duty version up to 1,500 kN

Rated force Fnom

0 ... 320 N to 0 ... 120 kN

Rated force Fnom

0 ... 25 kN to 0 ... 1,500 kN

Relative linearity
error

Analogue ≤ ±1.6 % Fnom
Digital ≤ ±0.5 % Fnom

Relative linearity
error

Analogue ≤ ±1.0 % Fnom
Digital ≤ ±0.5 % Fnom

Display■ Analogue indication

Display■ Analogue indication■ Digital display■ Pressure sensor■ Digital display■ Pressure sensor

Ingress protection

IP65, IP67

Ingress protection

IP65, IP67

Data sheet

FO 52.18

Data sheet

FO 52.17

Further information at www.wika.com

79Force ‫ ׀‬Force transducers

Special force transducers
We refer to force transducers that do not fit into any standard
design as special force transducers. Due to the specification of
the requirement, in some cases design-engineered solutions
must be considered. As a long-standing manufacturer of force
measurement technology, WIKA brings this expertise into playand can find the best and, at the same time, most cost-effective
solution for the customer. Among our special force transducers

F9204

Wire rope force
transducer up to 40 t

F9302

Strain transducer up
to 1,000 µε

FRKPS

Chain hoist
test set for checking friction
clutchesRated load Fnom0 … 1 to 0 … 40 t

Strain Fnom

0 … ±200, 0 … ±500, 0 … ±1,000 µε

Rated force Fnom

40 … 3,500 kg

Relative linearity
error±3 % Fnom

Relative linearity
error≤ ±2 % Fnom

Relative linearity
error

0.5 % Fnom

Output signal

4 … 20 mA, 2-wire

Output signal

4 … 20 mA, 3-wire

Output signal

4 … 20 mA

Ingress protection

IP66

Ingress protection

IP67

Ingress protection■ Force transducer IP67

Data sheet

FO 51.25

Data sheet

FO 54.10
Data sheet

FO 51.69

F9846

Strain transducer up
to 1,000 µε
Nominal strain Fnom 0 … 200 με to max. 0 … 1,000 με80are, for example, strain transducers that enable components tomeasure or force transducers for checking rope tension (wirerope force transducers). The applications in which special force
transducers are used are wide-ranging and always require great
experience in their engineering. You can count on this when you
trust in the right solution from WIKA.

Relative linearity
error±1 % Fnom

Output signal

1.0 ±0.1 mV/V

Ingress protection

IP65

Data sheet

FO 54.17■ Display instrument IP40

Force ‫ ׀‬Electronics

Force

Electronics
Many force measurement applications can be complemented byelectronic components. To ensure that all system-relevant components come from a single source, WIKA continuously expands
its product range with useful electronics. WIKA offers controllers,
amplifiers, limit switches, hand-held measuring instruments,
digital displays and electronic accessories that ensure trouble-free

B1940

Analogue cable amplifier
for strain gauge measuring
bridges
Input

Strain gauge measuring bridge, 4- or
6-wireOutput

0/4 … 20 mA, DC 0 … 10 V

Special feature■ High accuracy■ Cable length between amplifier andread-out unit: to 100 m are possible■ Compact design■ Ingress protection IP67
Auxiliary power

DC 12 … 28 V

Data sheet

AC 50.09

E1930, E1931, E1932

Multi-function display
for industrial mV/V andanalogue measuring instruments
5/6-digit digital display with high accuracy
Ingress protection

IP65

Data sheet

FO 58.05, FO 58.06, FO 58.07

Further information at www.wika.com

operation. With the help of electronics matched to the measuringcomponents, set limit values are maintained and checked with thereading instruments. Amplifiers are available with analogue anddigital output signals. The LED or LC displays are available with 4
or 6 digits.

ELMS1

Safety electronics
PLe in accordance withDIN EN ISO 13849-1
Input■ 8 safe 4 … 20 mA analogue inputs■ 8 safe digital inputs■ Fieldbus

OutputDigital limit switch

Input■ 0/4 … 20 mA

Output■ Two potential-free relay contacts■ 6 safe, positive-switching solid-state■ Galvanic isolation, line break (LB) and

Auxiliary power■ DC 20 … 90 V

Data sheet

AC 50.01■ Certified safety electronics, certified in

Auxiliary power

DC 24 V

Data sheet

AC 50.06

FE430

Weighing indicator

output (0 … 20 mA)

Special feature■ Fieldbus

accordance with DIN EN ISO 13849-1,
PLe■ Certified system solution incl. force
measurement, certified in accordancewith DIN EN 13849-1 cat. 3, PLd(change-over) with status LED■ One freely programmable analogue■ 2 safe relay outputs

outputs

Special feature

EGS80

short-circuit (SC) monitoring to SIL 2 per
IEC 61508■ AC 48 … 253 V

B6578

Junction box for load cells

Accuracy class in accordance with OIML with EU type approval
6000e

Number of loadcellsMax. 4Ingress protection

IP65

Ingress protection

IP67

Data sheet

AC 50.14

Data sheet

FO 58.02

81

Flow ‫ ׀‬Primary flow elements

Orifice plates and assemblies

Orifice plates represent the most common primary flow elementsin the world due to their proven technology and ease of installationand maintenance.

Versions

 Square edge orifice plates (standard version) This design isintended for general applications in clean liquids and gases.

Main characteristics
Maximum operating temperature to 800 °C
Maximum operating pressure to 400 bar
Suitable for liquid, gas and steam flow measurement
Accuracy: Uncalibrated ±0.5 … 2.5 %Repeatability of measurement 0.1 %







β: 0.2 … 0.75

Detail Quarter circle and conical entrance orifice plates

The best choice for measurement of liquids with low Reynolds
number.

FLC-OP

Orifice plate

Standards■ ISO 5167-2

Pipe size■ ≥ 2"

β

Depending on version

Accuracy 1)

Uncalibrated ±0.5 … 2.5 %Data sheet

FL 10.01■ ASME MFC3M■ ≥ 50 mm

Conical entrance:
Quarter circle:

β 0.10 … 0.316
β 0.245 … 0.60

Detail Segmental orifice plates

For measurements with two-phase, dirty and solids-containing
media.

β: 0.35 … 0.8

Detail Eccentric orifice

FLC-CO

Compact orifice plate for the
direct mounting of differential
pressure transmitters

82

Standards■ ISO 5167-2

Pipe size■ 2 … 14"βDepending on version

Accuracy≤ ±0.5 %Data sheet

FL 10.10

plates The application areas are similar to the segmental version. However, an eccentric orifice plate is the better solution
for smaller pipe diameters.

■ ANSI/ASME B16.5■ DN 50 … 350
β: 0.46 … 0.84

Detail1) The actual measuring deviation is specified during the engineering phase

Flow

Flow ‫ ׀‬Primary flow elements

Orifice flanges are intended for use insteadof standard pipe flanges when an orificeplate or flow nozzle must be installed.
Pairs of pressure tappings are machined
into the orifice flange, making separate
orifice carriers or tappings in the pipe wall
unnecessary.
Main characteristics

 Wide range of materials available The number and type of pressure

tapping (flange tap or corner tap)
can be manufactured to customer
requirements
 Special assemblies can be designed onrequestFLC-FL

FLC-MP

Orifice flangeStandards■ ISO 5167-2

Pipe size■ ≥ 2"

β

Depending on version

Accuracy 1)
Data sheet■ ASME B16.36■ ≥ 50 mm

Multi-hole orifice plate

Standards■ AGA Report number 3■ ASME B16.36

Pipe size■ 50 … 600 mm [2" ... 24"]

Uncalibrated ±0.5 … 2.5 %β

0.2 ... 0.65

FL 10.12

Accuracy 1)

1 ... 2 % depending on beta ratio andReynolds number

Data sheet

FL 10.15

Annular chambers are designed to be mounted between standard pipe flanges. Versions
are available to suit all common flange standards, including DIN and ANSI B16.5.
Main characteristics
 Standard material is 316/316L stainless steel, but a wide range of alternative materials isavailable
 Seals are included in the scope of delivery (as standard, 4.4 mm thick spiral-woundsealing 316/graphite filler, unless requested otherwise)

Further information at www.wika.com■ ISO 5167-2■ Larger versions on request

FLC-AC

Annular chamberStandards

ISO 5167-2

Pipe size■ ≥ 2"■ ≥ 50 mm

βDepending on version

Accuracy 1)

Uncalibrated ±0.5 … 2.5 %Data sheet

FL 10.13

83

Flow ‫ ׀‬Primary flow elements

Meter runs
To ensure high accuracy in the flow measurement of liquids, gases
and steam the primary flow element is supplied as an assembly incorporating the upstream and downstream pipe sections requiredby ISO 5167-1:2003. This assembly is known as a “meter run”.
Main characteristics

 Nominal width < 1 ½"
 Nominal pressure rating 300 … 2,500

depending on model/version
 Wide range of materials available

A calibration of the instrument can be performed if higher accuracy
is required.
An integral orifice plate is normally selected when thepipe diameter is 1 ½" or smaller and the medium is clean.
An extremely compact installation can be ensured as thepressure sensor can be mounted directly onto the meterrun. Without a calibration, an accuracy of ±1 … 2 % can be expected, the actual values will be confirmed during the engineering
phase.

Special assemblies

FLC-HHR-PP

HHR ProPak™ flow meterfor oil and gas

FLC-HHR-FP

HHR FlowPak® flow meter

FLC-MR

Meter runStandards

ISO 5167-2

Pipe size■ ½ … 1½ in

β0.2 … 0.75

Accuracy

Uncalibrated ±1 … 2 %Data sheet

FL 10.02■ 12 … 40 mm

FLC-FC

Cone flow meterStandards

ISO 5167-5

Pipe size

2 ... 64"

β and pipe length

0.45/0.6/0.75

Special feature

Low requirements for straight upstream
and downstream pipes

Data sheet

FL 10.11

FLC-WG
Wedge flow meter for slurries andhighly viscous media

Pipe size2", 3", 4", 6" or 8"

Pipe size

3 … 48"

Standards

ISO 5167-6

β and pipe length

0.75 or 0.40

β and pipe length

0.40 ... 0.70

Pipe size

1 … 24"

Special feature

No need for straight upstream anddownstream pipes

Special feature

No need for straight upstream anddownstream pipes

H/D ratios

0.2/0.3/0.4/0.5

Special feature■ Low maintenance through robust design

FL 10.07

Data sheet

Data sheet

FL 10.09■ For very high and very low Reynolds

numbers■ Bidirectional measurement possible

Data sheet

84

FL 10.08

Flow ‫ ׀‬Primary flow elements

Flow

Flow nozzles

A flow nozzle consists of a convergent section with a rounded profile and a cylindricalthroat. This design is generally selected for steam flow measurement at high velocity.

Flow nozzle for in-pipe installation

To reduce pressure loss an axisymmetric solution, called a Venturi nozzle, can be offered.
It combines the standard features of a flow nozzle with a divergent section.
Main characteristics
Suitable for liquid, gas and steam flow measurement
Optimum solution for measuring the flow of steam
Accuracy: Uncalibrated ±0.8 … 2 %Repeatability of measurement 0.1 %Lower pressure loss compared to orifice plate family.







Venturi nozzle

FLC-FN-PIP

FLC-FN-FLN

Flow nozzle for in-pipe installation

FLC-VN

Flow nozzle for flange assembly

Venturi nozzle

Pipe size■ ≥ 2 in

Pipe size■ ≥ 2 in

Pipe sizeβ

0.2 … 0.8

β

0.3 … 0.8

β0.316 ... 0.775

Accuracy 1)

Uncalibrated ≤ ±1 %Accuracy 1)

Uncalibrated ±0.8 %Accuracy 1)

Uncalibrated ±1 %Data sheet

FL 10.03

Data sheet

FL 10.03

Data sheet

FL 10.03■ ≥ 50 mm

Further information at www.wika.com■ ≥ 50 mm

1) The actual measuring deviation is specified during the engineering phase■ ≥ 2 in■ ≥ 50 mm

85

Flow ‫ ׀‬Primary flow elements

Venturi tubes

A Venturi tube is a reliable and easily managed and maintained
instrument that can measure a wide range of clean liquids andgases.
The main advantage of a Venturi tube over other differential
pressure flow measuring instruments is the higher pressure
recovery and the lower upstream and downstream straight tube
length requirements.
Main characteristics
In accordance with ISO 5167-4 and ASME MFC-3M standards
Fabricated from plate or machined from bar/forgings
Flanged or weld-in construction
Wide range of materials available
Pipe sizes from 50 ... 1,200 mm
Wide variety of pressure tappings available
Calibration possible on request
Accuracy: Uncalibrated ±0.5 … 1.5 %










FLC-VT-BAR

FLC-VT-WS

Venturi tube, bar body

Pipe size

86

Venturi tube, welded sheet■ 2 … 10 in

Pipe sizeβ

0.4 … 0.75

β

0.4 … 0.7

Accuracy 1)

Uncalibrated ≤ ±0.5 %Accuracy 1)

Uncalibrated ±1.5 %Data sheet

FL 10.04

Data sheet

FL 10.04■ 50 … 250 mm

Venturi tube, bar body■ ≥ 14 in■ 200 … 1,200 mm

1) The actual measuring deviation is specified during the engineering phaseFlow ‫ ׀‬Primary flow elements

FloTec (a multi-port, averaging pitot tube) measures the difference
between the static pressure and the dynamic pressure of themedia in the pipe. The volumetric flow is calculated from that difference using Bernoulli’s principle and taking into account the pipe
inner diameter. Using four dynamic ports this instrument is able
to evaluate a better velocity profile inside the pipe. This ensures ahigher accuracy in the flow measurement.

Flow

FloTec (averaging pitot tubes)

Vortex shedding frequency
Depending on the inner diameter, the medium characteristics andthe Reynolds number, a vortex will be generated around the pitot
tube. A support mounted on the opposite side of the pipe can besupplied should the natural frequency of the pitot coincide with thevortex shedding frequency. The necessity test is performed during
the design phase.

Main characteristics
Low installation costs
Long-term accuracy
Minimal permanent pressure loss
Fixed and extractable versions available






Vortex generation

FLC-APT-E

FloTec, extractable

FLC-APT-F
FloTec, fixed

Pipe size■ ≥ 3 in

Pipe size■ ≥ 3 inAccuracy

Uncalibrated ±1 %Accuracy

Uncalibrated ±1 %Data sheet

FL 10.05

Data sheet

FL 10.05■ ≥ 50 … 1,800 mm

Further information at www.wika.com■ ≥ 50 … 1,800 mm

87

Flow ‫ ׀‬Primary flow elements

Restriction orifices
When a reduction of pressure or a limitation of the flow rate is required, a restriction
orifice must be inserted into the pipeline.
Our technical department will produce thecorrect design for the restriction orifice,depending on customer requirements andflow conditions.
If a high pressure drop is required, phasechanges or sound problems can occur,
so that a more complex design might
be needed. The solution in these cases
is to decrease the differential pressure
in several steps, avoiding all the issues
created by these factors. This solution iscalled multi-step restriction orifice.

FLC-RO-ST

Single-step restriction
orifice

FLC-RO-MS

Multi-step restriction
orifice

Nominal size

½ … 24"

Nominal size

½ … 24"

Special feature■ Suitable for liquids, gases and steam

Special feature■ Suitable for liquids, gases and steam

Data sheet■ Single-bore or multi-hole versions

FL 10.06■ Special design with variable sections,with extension for high pressure drop
requirements in gas applications

Data sheet

FL 10.06

Main characteristics

 Multi-step restriction orifices to reduce

cavitation or undesired choking of theflow Multi-hole designs to reduce the noiselevel

Ultrasonic flow meter
For custody transfer of gases
By calculating velocity ratios between
two or more ultrasonic paths, the model
FLC-UFL provides reliable gas flow
measurement. Additional measured
variables, such as sound velocity, signal-tonoise ratio or signal strength, are available
for condition monitoring. For applications
requiring integrated volume conversion,pressure and temperature sensors can beconnected.

FLC-UFL

Ultrasonic flow meter

Pressure range
Accuracy

To 153 bar [2,250 psi]
FLC-UFL 1:■ 2 % (> 3 m/s)

■ 3 % (0.1 ... 3 m/s)
FLC-UFL 2:■ 1.5 % (> 3 m/s)
■ 2 % (0.1 ... 3 m/s)
FLC-UFL 3■ 1 % (Qt – Qmax)■ 2 % (Qmin – Qt)
FLC-UFL 4■ 0.5 % (Qt – Qmax)■ 1 % (Qmin – Qt)

Data sheet

88

FL 40.01

Flow ‫ ׀‬Flow switches

Flow

Flow switches
For monitoring liquid media

FSD-4

For liquid media

Measuring range■ Flow: 0 ... 3 m/s

Output signal■ 1 or 2 switching outputs plus optional■ Temperature: -20 ... 85 °C

analogue output■ Switching outputs PNP or NPN
adjustable■ Analogue output 4 ... 20 mA or
0 ... 10 V adjustable■ Optional IO-Link

Process connection ■ G ¼ A, G ½ A■ ¼ NPT, ½ NPT■ M18 x 1.5■ Various compression fittings optional
Data sheet

FSM-6100

For industrial heatexchangers
Measuring range■ Max. operating flow: 150 … 3,200 l/min

Output signal■ 2 switch points■ Medium temperature: -20 … +100 °C

Flow switch for wet sprinkler systems

Flow rate / Sensitivity

4 ... 10 GPM [15 ... 38 LPM]

Switching function

2 x SPDT (single pole double throw),form C

Data sheet

FL 50.01■ Switch point repeatability: ±5 % of span

Process connection ■ 1" NPT male per ASME B1.20.1■ 1" BSPT male per ISO 7
Data sheet

FSFD

FL 60.01

FL 80.02

Electromagnetic flow meters

FLC-608

Hybrid signal converter
for electromagnetic flow meters
Special featureStandards■ Available in different versions of installa-tion and power supply■ HART® protocol and module forpressure and temperature readoutavailable

For water-cycle andprocess applications

Retractable electromagnetic insertion
sensor■ 15 … 2,000 mm

DN■ 50 … 2,600 mm

Flow tube lining
material■ PTFE − standard for pipe diameters

Special feature■ “Hot-tapping” installation possible (drill-■ EN 61326 emission (group 1, class B)

Standards

FL 20.05■ 0.5 … 80 in

DN 15 ... DN 100, on request also for
DN > 100■ Hard rubber (ebonite) − for diameter≥ DN 125■ IECEx (option for separated version)■ MID MI-001 and OIML R49 for custody

Data sheet■ 2 … 104 in

ing of pipelines under pressure)■ Pressure gauge connection available■ 1" GAS or 1" NPT stopcock■ No moving parts and no pressure loss

■ ATEX (option for separated version)transfer

Further information at www.wika.com

FLC-1222DN■ EMC directive

and immunity (industrial application)

Data sheet

FLC-2200EL

FL 20.01

Standards■ ATEX (option for separated version)

Data sheet

FL 20.07■ IECEx (option for separated version)89

IIoT solutions

New opportunities for growth throughholistic IIoT solutions
From measured value to added value

With our innovative complete solutions, we
support our customers to become future-proof
by offering new added value through the
combination and use of digital measured data
across the entire value chain.

Real-time monitoring
Predictive algorithms identify potential problems in advance, keep
your employees up-to-date and trigger alarms in the event ofcritical values. This enables automatic or manual interventions to
avoid production downtime.

Data security
WIKA attaches great importance to the protection of your data.
With complete end-to-end encryption, bidirectional communication and a cloud solution hosted in the EU, we consistently implement the highest security standards.

Automation of maintenance
Maintenance actions are initiated automatically, eliminating theneed for manual reading and estimating. This allows your team tofocus on priority tasks.

90

Team productivity
IIoT solutions from WIKA enable the automation of menial,
time-consuming tasks to improve the efficiency of your employees.
This minimises errors or failures that can arise from human error in
repetitive, monotonous tasks.

Diagnostics and documentation
All measured data is archived to comply with internal and legal
requirements. The seamless collection of data allows existingprocess weaknesses to be identified and eliminated with the helpof diagnostic algorithms.

Cost reduction
Thanks to a precise, automatic evaluation of the measured datacollected, all processes along your value chain can be optimised
and unnecessary cost drivers eliminated.

IIoT solutions

IIoT solutions

Sensors

Connectivity

 Radio units for measuring instruments
 Measuring instruments with IIoT radio unit
 Mobile device app “myWIKA wireless device”

LoRa, mioty, LTE/NB-IoT and 5G networks

SolutionYour applications

Sensors

Connectivity

Cloud

Software

Cloud and on-premise

Applications

Tailor-made IIoT solutions from WIKA

Perfectly scalable for your business

Added value

Strong partner in IIoT ecosystems
WIKA is a founding member of the mioty alliance, and not only
drives the development of pioneering technologies, but also
supports industrial standards such as LoRaWAN® and OPC UA.
For WIKA, technological leadership has been the key to opening
up new markets and applications for over 75 years.

cloud solutions are hosted within the European Union. Our
comprehensive IIoT offering, based on the latest industry
standards, preserves the integrity of your data by encrypting itfrom end to end.

In order to fully meet the requirements of our customers and to beable to offer flexible solutions that are as compatible as possible,
WIKA cooperates with leading technical organisations andcompanies.
Data security has the highest priority – which is why all WIKA

Our partners

Further information at www.wika.com

91

IIoT products

IIoT products

NETRIS®1Radio unit with LoRaWAN® for WIKA
measuring instruments

Ingress protection

IP65, IP67

Accuracy■ RTD: ≤ ±0.1 % of span

■ RTD, potentiometer: ≤ ±0.10 % of span
■ Analogue: ≤ ±0.1 % of span

Transmission range 10 km [6 mi]
Function■ Register■ Configuration of measuring andtransmission rate
■ Sending measured values■ Alarm management■ Battery status
Data sheet

PEW-1000

Pressure sensor with wirelesstransmission for general industrial
applications
Ingress protection

IP54, IP67 and IP65

Accuracy≤ ±0.5 % of span

Transmission range 10 km [6 mi]
Measuring range

Data sheet■ 0 … 0.4 to 0 … 1,000 bar[0 … 1 to 0 … 15,000 psi] as well as
vacuum and ± measuring ranges

TRW

Resistance thermometer withwireless transmission for generalindustrial applications
Ingress protection

IP67

Accuracy≤ 0.1 % of span

Transmission range 10 km [6 mi]
Measuring range-196 … +500 °C [-321 … 932 °F]

Data sheet

TE 63.04

PE 87.23

AC 40.01

PGW23.100,
PGW26.100

Bourdon tube pressure gauge withwireless transmission, safety version
Ingress protection

IP54, IP65 (case filling)

Accuracy class

1.0

Transmission range 10 km [6 mi]92

Measuring range

0 … 0.6 to 0 … 1,600 bar[0 … 10 to 0 … 20,000 psi] as well as
vacuum and ± measuring ranges

Data sheet

PV 42.02

LoRaWAN® is a trademark used under licence from LoRa-Alliance®.
The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG, Inc. and any use of such marks by WIKA is under licence.

IIoT solutions

IIoT products

TYPE EL
CLASS IJANUARY 2023

NETRIS®3

Radio unit with LoRaWAN®

Ingress protection

IP65

Transmission range 10 km [6 mi]
Function■ Registration

Bourdon tube pressure gauge forconnection to NETRIS®3 radio unit,
safety version

Gas-actuated thermometerfor connection to NETRIS®3 radio unit

Pressure transmitterfor connection to
NETRIS®3 radio unit

IP54, IP65 (case filling)

Ingress protection

IP66/IP67

Accuracy class

1.0

Accuracy■ ≤ 0.1 % of span

Measuring range

0 … 0.6 to 0 … 1,600 bar[0 … 10 to 0 … 20,000 psi] as well as
vacuum and ± measuring ranges

Data sheet

TRU

Miniature resistance thermometer forconnection to NETRIS®3 radio unit

■ ≤ 0.5 % of span at > 1,000 bar

Process connection for hygienic applications
Measuring range■ 0 … 1 to 0 … 1,600 bar

Data sheet

PE 87.24

PV 42.03

AC 40.03

TGU73.100

PEU-20, PEU-21

Ingress protection■ Configuration of measuring andtransmission rate
■ Sending measured values■ Alarm management
Data sheet

PGU23.100,
PGU26.100[0 … 15 to 0 … 20,000 psi] as well as
vacuum and ± measuring range

FLRU

Reed level transmitter for connection
to NETRIS®3 radio unit

Ingress protection

IP65

Ingress protection

IP66, IP67

Ingress protection

IP66, IP68

Accuracy class

2.0

Accuracy■ Measuring range ≤ 300 °C: ±1.9 °K

Accuracy■ 2.7 mm [0.11 in]

Measuring range-170 … +600 °C [-274 ... +1,112 °F]

Data sheet

TV 17.13

Measuring range-196 … +500 °C [-321 … 932 °F]

Data sheet

TE 63.03■ Measuring range ≤ 300 °C: ±2.9 °K■ 7.5 mm [0.30 in]

Guide tube length
Data sheet

Further information at www.wika.com■ 5.5 mm [0.22 in]■ 9 mm [0.35 in]Max. 6,000 mm [236.22 in]
LM 20.13

93

Calibration ‫ ׀‬Pressure

Digital pressure gauges
High-quality digital pressure gauges from WIKA
Precision digital pressure gauges are suitable for stationary andalso mobile measurement and indication of pressures.
In addition, a digital pressure gauge can be used as a pressurereference and enables the easy testing, adjustment and calibrationof other pressure measuring equipment directly on-site. Throughefficient measuring cells with electronic linearisation of thecharacteristic curve, a high accuracy is achieved.

CPG1200

Digital pressure gauge

CPG1500

Precision digital pressure gauge

Measuring range-1 ... 1,000 bar

Measuring range

Accuracy

Down to 0.25 % FS

Accuracy

Down to 0.025 % FS

Special feature■ Integrated data logger

Special feature■ Integrated data logger■ WIKA-Cal compatible

Data sheet■ Password protection possible■ Robust case, IP65

Data sheet

CT 10.51

CT 10.20

DTK1X00

DMK1200

Test case incl. digital pressure gauge
model CPG1200 or CPG1500

Measuring case incl. digital pressuregauge model CPG1200■ Simple data transmission via USB or Bluetooth®

■ Simple data transmission via USB or Bluetooth®

■ Optional data logger for up to 1 million data points■ Optional data logger for up to 1 million data points■ Precise adjustment through the fine adjustment valve■ Supports the series MINIMESS® 1620 or MINIMESS® 1215

■ Clear and simple operation■ The right hand test pump for every pressure range■ Data sheet: CT 93.03

94■ WIKA-Cal compatible■ Data transmission via Bluetooth®

■ Data transmission via USB or

Bluetooth®
■ Robust case, IP65

0 … 10,000 bar

quick coupling systems■ Data sheet: CT 93.04

Calibration ‫ ׀‬Pressure

Hand-helds are portable calibration instruments for mobile use
for the accurate measurement and recording of pressure profiles.
There are interchangeable pressure sensors with measuring
ranges of up to 10,000 bar available for the instruments. Throughthis, hand-helds are particularly suitable as test instruments for a

CPH6200, CPH62I0

Hand-held pressure indicatorlarge variety of applications in the widest range of industries. Data
recorded in the hand-held can be evaluated via PC software, some
instruments document calibrations in the internal memory, which
are later read on a PC. Optionally, a calibration certificate can begenerated with our calibration software WIKA-Cal.

CPH6300

Hand-held pressure indicator

Measuring range-0.025 … +0.025 to -1 … 1,000 bar

Measuring range-0.025 … +0.025 to -1 … 1,000 bar

Accuracy

0.2 %, 0.1 % (optional)

Accuracy

0.2 %, 0.1 % (optional)

Special feature■ Integrated data logger

Special feature■ Robust and waterproof case with IP65,

Data sheet■ Differential pressure measurement

Completetest and service
cases
These cases can be assembled exactly to your requirements.
Thus you will be fully equipped on-site!

IP67■ Integrated data logger(optional)■ Ex version: Model CPH62I0 (optional)■ Differential pressure measurement(optional)

CT 11.01, CT 11.02

Data sheet

Further information at www.wika.com

Calibration

Hand-helds, calibrators

CT 12.01

95

Calibration ‫ ׀‬Pressure

Hand-helds, calibrators

CPH7000,
CPH7000-Ex

Portable process calibrator
Measuring range

CPH8000

Portable multi-function
calibrator-1 … 25 bar(-1 … 10,000 bar with CPT7000)

Measuring range■ -1 ... 700 bar

Accuracy

0.025 % FS

Accuracy

0.025 % FS

Special feature■ Large display with touchscreen

Special feature■ Integrated pressure generation

function■ Measurement of pressure, temperature,■ Measurement and simulation ofcurrent, voltage, ambient conditions

temperature, current, voltage,
resistance, frequency, pressure■ HART® communication■ Data transmission via WIKA-Wireless■ Supply of pressure, current and voltage■ Calibration function, data logger, switch

test

Data sheet96

CT 15.51■ Integrated data logger and calibration

Data sheet

CT 18.03

WIKA-Cal

Calibration software,accessories for digital pressure gauges■ Creation of calibration certificates for mechanical andelectronic pressure measuring instruments■ Fully automatic calibration with pressure controllers■ For the recording of certificate-relevant data in combination

with the CalibratorUnits of the CPU6000 series■ Determination of the required mass loads for pressurebalances■ Calibration of gauge pressure measuring instruments withabsolute pressure references and vice versa

Data sheet: CT 95.10

Calibration ‫ ׀‬Pressure

Precision pressure measuring instruments are electrical measuring systems which convert pressure into an electrical signal andoptionally visualise it. Precise pressure transmitters and processtransmitters are used for the monitoring and control of particularly
sensitive processes.

CPT2500

USB pressure sensor

Calibration

Precision pressure measuring
instruments
Due to the low, DAkkS-accredited measurement uncertainty ofdown to 0.008 % of the entire measuring chain, the particularly
accurate instruments find their primary applications as a factory/working standard for testing and/or calibrating a variety of pressure
measuring instruments.

CPT6030

Analogue pressure transducer

CPT6100, CPT6140,
CPT6180
Precision pressure sensor

Measuring range

0 … 0.025 to 0 … 1,000 bar

Measuring range

0 … 0.025 to 0 … 1,000 bar

Measuring range

0 … 0.025 to 0 … 400 bar

Accuracy

0.2 %, 0.1 % (optional)

Accuracy

0.025 %

Accuracy

0.01 %, 0.025 % (for CPT6140)

Special feature■ Recording interval adjustable from 1 ms

MediumNon-corrosive gases, liquids > 350 mbar

MediumNon-corrosive gases, liquids > 1 bar

Special feature■ Comp. temperature range -20 … +75 °C

Special feature■ RS-232 or RS-485 connection

Data sheet... 10 s■ No external voltage supply required■ Data storage and evaluation directly
via PC■ 4 … 20 mA■ DC 15 … 28 V

■ Barometric measuring range:■ Ingress protection IP67

CT 05.01
Data sheet

552 … 1,172 mbar abs., 0.01 % ofreading■ Measuring rate of 4 ms at CPT6140

CT 25.14
Data sheet

CPT9000,
CPT6020

CPG2500

Precision pressure sensor

Precision pressure
measuring instrument

Measuring range

0 … 0.025 to 0 … 1,000 bar

Measuring range

0 … 0.025 to 0 … 2,890 bar

Accuracy■ CPT9000: 0.008 %Accuracy

0.014 %, 0.01 % and 0.008 %MediumNon-corrosive gases, liquids > 1 bar

Medium■ CPT6020: 0.02 %Non-corrosive gases, liquids > 350 mbar
Special featureSpecial feature■ Comp. temperature range 0 … 50 °C■ RS-232 or RS-485■ Barometric measuring range:

552 … 1,172 mbar abs., 0.008 % ofreading■ Resolution 100 ppb or better
Data sheet

Precision air data test
indicator
Measuring range■ Altitudes to 100,000 ft■ Speeds to 1,150 knots

Accuracy

0.01 %, 0.009 %Special feature■ RVSM-compliant■ Ps, Qc, Ps/Pt or Ps/Qc configuration

with virtual channels

and 1 external sensor ofmodel CPT9000 or CPT6100■ Delta and leak test available

Data sheet

CT 25.10, CT 25.11

CPA2501■ Up to 2 exchangeable, internal sensors■ Barometric reference (optional)■ Measuring rate 20 ms■ Analogue output (optional)■ Altitude and airspeed rate indication

Data sheet

CT 29.02

CT 25.02

CPT9000: CT 25.12
CPT6020: CT 25.13

Further information at www.wika.com

97

Calibration ‫ ׀‬Pressure

Pressure controllers
WIKA pressure controllers: Always the right calibration solution
Pressure controllers are electronic controllers which quickly andautomatically provide a stable pressure reference. Due to the highaccuracy and control stability, pressure controllers are especiallysuitable as references for production lines and laboratories, inorder to carry out automatic testing and/or calibration of all typesof sensors.

CPC2000

Low-pressure version

With pneumatic ranges from 1 mbar to 700 bar and hydraulic ranges to 1,600 bar, the pressure controllers cover a wide range.
Each controller represents a breakthrough in control and measurement technology to provide first-class measurement accuracy andhighly stable pressure control.

CPC4000

Industrial series

CPC6050

Modular version

Measuring range

0 … 1 to 0 … 1,000 mbar

Measuring range

0 … 0.35 to 0 … 210 bar

Measuring range

0 … 0.025 to 0 … 210 bar

Accuracy

0.1 % / 0.3 % (for 0 … 1 mbar)

Accuracy

0.02 %

Accuracy

0.008 %

MediumAmbient air

Control stability

0.005 %

Control stability

0.003 %

Special feature■ Integrated pressure generation

MediumDry, clean air or nitrogen

MediumDry, clean air or nitrogenSpecial feature■ Up to 2 sensors

Special feature■ Up to 2 control/measuring channels with

Data sheet■ Integrated rechargeable battery

CT 27.51■ Fast control speed■ Sensors exchangeable(optional)■ Up to 24 internal programmablesequences■ Auto-channel for both controllers■ Automatic contamination protection

Data sheet

98

2 sensors each

■ Leak test function

CT 27.40■ Switch test function■ Automatic contamination protection(optional)

Data sheet

CT 27.62

Calibration

Calibration ‫ ׀‬Pressure

CPC8000

Pneumatic pressure controllers,premium version

CPC7000

Pneumatic pressure controllers,
high-pressure version

CPC8000-H

Hydraulic pressure controller,
high-pressure version

Measuring range

0 … 0.35 to 0 … 400 bar

Measuring range

0 … 100 bar to 0 … 700 bar

Measuring range

0 … 100 to 0 … 2,895 bar

Accuracy

0.01 … 0.008 %Accuracy

0.01 %

Accuracy

0.014 % … 0.01 %Control stability

0.002 %

Control stability

0.008 %

Control stability

0.005 %

MediumDry, clean air or nitrogen

MediumNitrogenMediumHydraulic oil or waterSpecial feature■ Excellent control stability and pressure

Special feature■ Robust and low-wear valve technology

Special feature■ High stability

control without overshooting■ Up to three interchangeable sensors■ Optional barometer for automaticconversion of the pressure type

Data sheet■ Automatic flooding■ High-pressure safety

Data sheet

CT 27.63■ Up to two interchangeable reference

sensors■ 6 x digital I/O■ Control performance can be matched

to application

with long-term stability■ Up to three interchangeable sensors■ Hydraulic liquids available, e.g.

Sebacate, Shell Tellus 22, Krytox, FC77

Data sheet

CT 28.05

CT 28.01

For aviation
An air data test set is a an electronic controller which
provides a pressure at a variable and adjustable rate.
Air data test sets are specifically developed to convert thepressure to be controlled into a height or rate of climb and velocity.
As a result of the high accuracy, control stability and ability tosimulate altitude and velocity, an air data test set is particularly
suitable as a reference for aircraft workshops and also for
instrument manufacturers and calibration laboratories in theaviation industry, in order to make calibrations on sensors anddisplays.

CPA8001

Air data test set

Measuring range■ Altitudes to 100,000 ft

Accuracy

0.01 % … 0.009 %Control stability

0.002 %■ Speeds to 1,150 knots

MediumDry, clean air or nitrogenSpecial feature■ Excellent control stability, even with

rate control■ Overshoot-free control■ RVSM compatible■ Configurations Ps/Pt, Ps/Qc

Data sheet

Further information at www.wika.com

CT 29.01

99

Calibration ‫ ׀‬Pressure

Pressure balances
Industrial seriesCompact and competitively priced dead-weight testers for use on-site or for maintenance and service
The compact dimensions and low weight are key features of these dead-weight testers for their daily use in service and maintenance.
With their integrated pressure generation and purely mechanical measurement principle, they are also specifically suited to on-site
applications.

CPB3500

Pneumatic compact
version

CPB3800

Hydraulic compact version

Compact, high-pressure
version with dual-range pistoncylinder system

Measuring range

0.015 … 1 to 1 … 120 bar

Measuring range

1 … 120 to 10 … 1,200 bar

Measuring range

1 … 2,600 bar

Accuracy

0.015 … 0.006 %Accuracy

0.05 … 0.025 %Accuracy

0.025 … 0.007 %MediumNon-corrosive gases

MediumSpecial oil

MediumSpecial oil or others on request

Special feature■ Compact dimensions and low weight

Special feature■ Compact dimensions and low weight

Special feature■ Dual-range piston-cylinder systems with■ 1 bar piston can be used for positive

and negative overpressure

Data sheet

CT 31.22

CPU6000 series
CalibratorUnit■ Determination of the required mass loads or the reference

pressure for calibration with pressure balances■ Recording of certificate-relevant data
■ Calibration of gauge pressure measuring instruments withabsolute pressure references and vice versa■ Easy calibration of pressure transmitters through the voltage

supply and multimeter function

Data sheet: CT 35.02

100

CPB3800HP

Data sheet■ Instrument base can now also be

combined with the CPB5800 piston-cylinder systems

CT 31.06

automatic changing between ranges■ Compact dimensions and low weight

Data sheet

CT 31.07

Calibration

Calibration ‫ ׀‬Pressure

Laboratory version
High-performance primary standards with excellent running characteristics for use in calibration laboratories
Through modern instrument design with excellent equipment features, the highest demands of operator convenience and performance
are fulfilled. The selection of dual-range piston-cylinder systems with automatic changing between ranges can ensure this measurement
uncertainty over a large pressure range, even with a single measuring system.

CPB5000

Pneumatic version

CPB5000HP
High-pressure
version

CPB5800

Hydraulic version withdual-range piston-cylinder
systems

Measuring range -0.03 … -1 to 0.4 … 100 bar

Measuring range 25 … 2,500 to 25 … 6,000 bar

Measuring range 1 … 120 to 1 … 1,400 bar

Accuracy

0.015 … 0.008 %Accuracy

0.025 … 0.02 %Accuracy

MediumNon-corrosive gases

MediumSpecial oil

MediumSpecial oil or others on request

Special feature

Patented system for fast piston-cylinder
exchange

Special feature

Robust instrument base with integrated
high-pressure generation

Special feature■ Dual-range piston-cylinder systems with

CT 31.01

Data sheet

Data sheet

Differential pressure
versionMeasuring range 0.03 … 2 to 25 … 1,600 bar
Accuracy

0.015 … 0.008 %MediumNon-corrosive gases or special oil

Special feature

Two complete pressure balances within one
case for real differential pressure measurements under static pressure

Data sheet

CT 31.56

Further information at www.wika.com

automatic changing between ranges■ Instrument base can now also be

CT 31.51

combined with the CPS5000 piston-cylinder systems

Data sheet

CPB5600DP

0.015 … 0.006 %CT 31.11

CPS5000

Hydraulic single-range
piston-cylinder systems
Special feature■ For the highest demands on accuracy andperformance■ Can be combined with the CPB5800

Data sheet

instrument baseCT 31.01

101

Calibration ‫ ׀‬Pressure

Pressure balances
High-end version
High-accuracy and high-performance primary standards with excellent operating characteristics, based on the physicalprinciple of Pressure = Force/Area
The direct measurement of the pressure (p = F/A), as well as the use of high-quality materials enable this small measurement uncertainty,
in conjunction with an excellent long-term stability (recommended recalibration interval of five years in accordance with the German
Calibration Service DKD/DAkkS). Furthermore, an automatic mass handling system and pressure generation ensure fully automatic
calibration. The pressure balance has therefore been used for years in factory and calibration laboratories in industry, national institutes
and research laboratories, and also in production by sensor and transmitter manufacturers.

CPB6000

Highest-accuracy primary
standard

CPB6000DP

Primary standard for
differential pressure

CPD8500

Digital pressure balance

Measuring range

4 … 5,000 bar

Measuring range

30 … 800 bar

Measuring range

1 … 500 bar (abs. and gauge)

Accuracy

0.0035 … 0.0015 %Accuracy

0.005 … 0.002 %Accuracy

0.005 … 0.0035 %MediumDry, clean air, nitrogen or special oil

MediumNon-corrosive gases

MediumNon-corrosive, dry gases

Special feature

For differential pressure measurementsfrom 10 Pa to 800 bar

Special feature■ Unique principle of operation basedSpecial feature

Different instrument variants for the highest
demands

Data sheet

CT 32.01

Data sheet

on SI units■ Intuitive operator interface

CT 32.02■ Automatic calibrations, no mass

handling needed■ Automatic compensation of the ambientconditions

Data sheet102

CT 32.05

Calibration ‫ ׀‬Pressure

Calibration

Calibration software
Easy and fast creation of a high-quality calibration certificate

WIKA-Cal

WIKA-Cal calibration software enables an automated calibration
process with the subsequent creation of calibration certificates(Cal-Template) or logger protocols (Log-Template) for pressure
measuring instruments. It is available as a demo version for freedownload from the homepage. Alongside the simple operationof the software, WIKA-Cal supports the user in the document
creation process.

Calibration software,accessories for digital pressure gauges■ Creation of calibration certificates for mechanical andelectronic pressure measuring instruments■ Fully automatic calibration with pressure controllers■ For the recording of certificate-relevant data in combination

with the CalibratorUnits of the CPU6000 seriesWith the purchase of a USB dongle with the desired licence, therange of functions of the demo version is automatically extended
while the USB dongle is plugged in and these functions areavailable so long as the USB dongle is connected to the computer.

■ Determination of the required mass loads for pressurebalances■ Calibration of gauge pressure measuring instruments withabsolute pressure references and vice versa

Data sheet: CT 95.10

In addition to the demo version, three WIKA-Cal licences are available in connection with a precision pressure measuring
instrument
The WIKA-Cal calibration software is available for online calibrations together with a PC. The scope of software functions depends on theselected licence. Several licences can be combined on one USB dongle.
Cal-Template (demo version)

Cal-Template (light version)

Cal-Template (full version)■ Fully automatic calibration■ Semi-automatic calibration■ Fully automatic calibration■ Limitation to two measuring points■ No limitation of the measuring points■ No limitation of the measuring points■ Creating calibration certificates per DIN EN 10204■ Calibration reports can be exported to Excel® template or XML file■ Calibration of gauge pressure measuring instruments with absolute pressure references and vice versa

Log-Template (demo version)

Log-Template (full version)■ Limitation to five measuring points■ No limitation of the measuring points■ Live measured value recording for a certain period of time with selectable interval, duration and start time

■ Creation of logger protocols with graphic and/or tabular representation of the measuring results in PDF format■ Possibility of exporting measuring results as CSV fileMulticalibration
The “Multicalibration” licence available for an additional charge canbe ordered in addition to Cal Light or Cal. With this, it is possible tocalibrate, incl. documentation, up to 16 test items simultaneously.
The prerequisite is that the test items are of the same instrument
model, measuring range and accuracy.
For pressure sensors, it is possible to use either several
multimeters (such as model CPU6000-M, for example) or amultiplexer to which all multimeters will be connected.

CPU6000-M
CPU6000-M

MUX

CPU6000-M

Pressure sensors, model CPU6000-M multimeter, multiplexer
and PC with WIKA-Cal software

Further information at www.wika.com

103

Calibration ‫ ׀‬Pressure

Pressure generation
Portable pressure generation
Hand test pumps serve as pressure generators for the testing,adjustment and calibration of mechanical and electronic pressure
measuring instruments through comparative measurements.
These pressure tests can take place in the laboratory or workshop,
or on-site at the measuring location.

CPP10-H

Pneumatic hand testpump

Pneumatic hand testpump

CPP700-H,
CPP1000-H

Hydraulic hand test pump

Measuring range-0.85 ... +10 bar

Measuring range-950 mbar … +35 bar

Measuring range

MediumAmbient air

MediumAmbient air

MediumOil or waterSpecial feature■ Switchable pressure/vacuum generation

Special feature■ Pressure and vacuum generation

Special feature■ Integrated medium reservoir

Data sheet

CT 91.07■ Precise adjustment through the fineadjustment valve■ Simple and ergonomic handling■ Compact dimensions and low weight
Data sheet

CT 91.10

CPP1000-M,
CPP1000-L

Hydraulic hand spindle pump
Measuring range

0 … 1,000 bar

MediumOil or waterSpecial feature■ Smooth-running, internally operating

precision spindle■ Compact dimensions

Data sheet

104

CPP30

CT 91.05

switchable■ Compact dimensions

Data sheet

CT 91.06

0 … 700 or 0 … 1,000 bar■ Ergonomic handlingCalibration

Calibration ‫ ׀‬Pressure

Pressure generation in the laboratory
Comparison test pumps serve as pressure generators or controllers for the testing, adjustment and calibration of mechanical andelectronic pressure measuring instruments.
Due to their stable case, these test pumps are particularly suitable
for stationary use in laboratories or workshops.

CPP120-X

Pneumatic comparison
test pump

CPP1200-X

Hydraulic comparison testpump

Measuring range

0 … 120 bar

Measuring range

MediumClean, dry, non-corrosive gases

MediumOil or waterSpecial feature■ Accurate pressure settingSpecial feature■ Integrated tank

■ Robust industrial series■ External initial pressure supply

Data sheet

necessaryCT 91.03

CPP4000-X

Hydraulic comparison testpump

0 … 1,200 bar■ Dual-area spindle pump
■ Robust industrial series

Data sheet

CT 91.08

CPP1000-X,
CPP1600-X
Hydraulic comparison test pump

CPP7000-X

Hydraulic comparison testpump

Measuring range

0 … 1,200 bar

Measuring range

0 … 1,000 to 0 … 1,600 bar

Measuring range

MediumOil or water

MediumOil or water

MediumSebacate oil

Special feature■ Integrated tankSpecial feature■ Integrated tankSpecial feature■ Integrated tank

■ Dual-area spindle pumpData sheet■ Robust laboratory version with primingpump■ Compact industrial series with primingpump■ Robust industrial series

CT 91.09
Data sheet

0 … 7,000 bar■ Robust laboratory version with priming

pump

Data sheet

CT 91.13

CT 91.12

105

Calibration ‫ ׀‬Temperature

Reference thermometers
Highly accurate temperature measurement with reference thermometers
Reference thermometers (standard thermometers) are, due totheir excellent stability and their geometrical adaptations, ideallysuited for applications in industrial laboratories. They enable easycomparative calibration in baths, in tube furnaces and in drywell calibrators. The advantage of reference thermometers is thewide temperature range, and with this, their flexible operation.
Furthermore, with their low drift, a long service life is ensured.

CTP1500

Digital thermometer

Platinum resistancethermometer

Measuring range-50 ... +250 °C

Measuring range-200 … +450 °C

Accuracy■ ±0.05 K at -20 ... +180 °C

Stability< 50 mK after 100 h at 450 °C■ ±0.15 K in remainder of range

Dimensions

Depending on version

Special feature■ Digital display in the handle

Data sheet

CT 61.05■ Battery powered

CTP5000

Reference thermometer

106

CTP2000

Dimensions

Ø 4 mm, l = 500 mm

Special feature■ 4-wire connection

Data sheet

CT 61.10■ Ends with 4 mm banana plugs

CTP6000

Reference thermometer

CTP9000
Thermocouple

Measuring range-196 … +660 °C

Measuring range-200 ... +420 °C

Measuring range

0 … 1,600 °C

Probe type

Pt100, Pt25

Probe type

Pt100

Thermocouple

Type S per IEC 584, class 1

Dimensions

Depending on version

Dimensions

Depending on version

Dimensions

Depending on version

Special feature■ Flying leads

Special feature■ Flying leads

Special feature■ Cold junction optional

Data sheet

CT 61.20

Data sheet

CT 61.30

Data sheet

CT 61.10■ DIN or SMART connector■ DIN or SMART connector■ 2,000 mm cableCalibration ‫ ׀‬Temperature

Calibration

Hand-helds

Hand-helds are portable calibration instruments for mobile use forthe accurate measurement and recording of temperature profiles.
For the instruments there are various designs of thermometers
available. Through this, hand-helds are particularly suitable as test
instruments for a large variety of applications in the widest range ofindustries.

CTH6200

Hand-held thermometer

Data recorded in the hand-held can be evaluated via PC software,some instruments document calibrations in the internal memory,which are later read on a PC. Optionally, a calibration certificate
can be generated with our calibration software WIKA-Cal.

CTH6300,
CTH63I0

Hand-held thermometer

Measuring range-50 … +250 °C

Measuring range-200 … +1,500 °C

Accuracy< 0.2

Accuracy

0.1 … 1 K

Probe type

Pt100

Probe type

Pt100, TC

Special feature

Integrated data logger

Special feature

Data sheet

CT 51.01
Data sheet

CTH7000

Hand-held thermometer

Hand-held thermometer
Measuring range-200 … +1,500 °C

Accuracy

0.03 … 0.2 K

Probe type

Pt100, TC■ 2 channels (optional)

Special feature■ Integrated data logger (optional)

CT 51.05

Data sheet■ Ex version: Model CTH63I0

Ex version: Model CTH65I0
CT 55.10

CTR1000

Infrared hand-heldthermometer

Measuring range-200 … +962 °C

Measuring range-60 … +1,000 °C

Accuracy

0.015 K

Accuracy

2 K or 2 % of reading

Probe type

Pt100, Pt25 and NTC

Special feature

Thermocouple connection (optional)

Special feature

Integrated data logger

Data sheet

CT 55.21

Data sheet

CT 55.50

Further information at www.wika.com

CTH6500,
CTH65I0107

Calibration ‫ ׀‬Temperature

Calibration baths

Calibration baths are electronic controllers which automatically,
quickly and with the help of a liquid supply a temperature. Due tothe high reliability, accuracy and exceptional homogeneity in themeasuring chamber, calibration baths are particularly suitable as afactory/working standard for the automatic testing and/or calibration of the widest range of temperature probes - independent of
diameter. A special micro calibration bath design enables on-site
applications.

CTB9100

Micro calibration bath

Measuring range-35 … +255 °C

Accuracy±0.2 … 0.3 K

Stability±0.05 K

Special feature■ Short heating and cooling times

Data sheet■ Easy to use

CT 46.30

CTB9500

Calibration bath,
low measuring range

108

Measuring range-45 … +200 °C

Stability±0.02 K

Immersion depth

200 mmMediumWater, oil or similar media

Data sheet

CT 46.20

CTM9350-165

Multi-function calibrator,
premium version
Measuring range
Accuracy

CTB9400

Calibration bath,medium measuring range-35 … +165 °C depending on theapplication

Measuring range

28 … 300 °C±0.07 … 0.5 K depending on the application

Stability±0.02 K

Immersion depth

200 mmMediumWater, oil or similar media

Data sheet

CT 46.20

Immersion depth

150 mmSpecial feature

Use as a dry-well calibrator, micro calibration bath, surface temperature calibrator
and infrared black body

Data sheet

CT 41.41

Calibration ‫ ׀‬Temperature

Calibration

Portable temperature
calibrators
CTD9100
Portable temperature calibrators (dry-well calibrators) are electronic controllers which
automatically, quickly and dryly supply a temperature. Due to the high reliability, accuracy
and simple operation, portable temperature calibrators are particularly suitable as a factory/working standard for the automatic testing and/or calibration of temperature measuring
instruments of all types.

CTD4000

Temperature dry-wellcalibrator

CTD9100-1100

High-temperature dry-wellcalibrator

Temperature dry-wellcalibrator
Measuring range-55 … +650 °C

Accuracy±0.15 … 0.8 K

Stability±0.01 … 0.05 K

Immersion depth

150 mmData sheet

CT 41.28

CTD9350-165,
CTD9350-700
Temperature dry-well calibrator,
premium version

Measuring range-24 … 650 °C

Measuring range

200 … 1,100 °C

Measuring range-35 … +700 °C

Accuracy

0.25 … 0.5 K

Accuracy±3 K

Accuracy±0.1 K

Stability

0.1 … 0.3 K

Stability±0.3 K

Stability

Immersion depth

104 mm/150 mm

Immersion depth

220 mm, bore depth 155 mm±0.008 ... 0.1 K depending on thereference

Data sheet

CT 41.10

Data sheet

CT 41.29

Immersion depth

150 mmData sheet

CT 41.39

CTD9100-375

Compacttemperature dry-well calibrator

CTI5000

Infrared calibrator

Measuring range

tamb … 375 °C

Measuring range

50 … 500 °C

Accuracy±0.5 … 0.8 K

Stability±0.1 … 0.4 K

Stability±0.05 K

Special feature

Large diameter of measuring surface

Immersion depth

100 mmData sheet

CT 41.42

Data sheet

CT 41.32

Further information at www.wika.com

CTM9350-165

Multi-function calibrator,
premium version
Measuring range-35 … +165 °C depending on theapplication

Accuracy±0.07 … 0.5 K depending on the application

Immersion depth

150 mmSpecial feature

Use as a dry-well calibrator, micro
calibration bath, surface temperature
calibrator and infrared black body

Data sheet

CT 41.41

109

Calibration ‫ ׀‬Temperature

Resistance thermometry bridges

By using built-in or external standard resistors, resistance thermometry bridges measure resistance ratios with high accuracy, which areindicative of the temperature, among other things. These instruments are not only used in the field of temperature measurement, but −due to their high accuracy − also in electrical laboratories.

CTR2000

Precision thermometer

CTR3000

Multi-functional precision
thermometer

Multiplexer

Measuring range-200 … +850 °C

Measuring range-210 … +1,820 °C

Measuring range-210 … +1,820 °C

Accuracy■ 0.01 K (4-wire)

Accuracy■ ±0.005 K (4-wire)

Accuracy■ ±0.005 K (4-wire)■ 0.03 K (3-wire)

Probe type

Pt100, Pt25

Special feature■ 3-wire measurement (optional)■ Up to 8 channels integrated in the

instrument (optional)

Data sheet

DC resistancethermometry bridge■ ±0.03 K (3-wire)■ ±0.004 % + 2 µV for thermocouples

Probe type

Pt100, Pt25, thermocouples

Probe type

Pt100, Pt25, thermocouples

Special feature■ Versatile applications by measuring

Special feature■ No loss of accuracy

CT 60.10

CTR6000■ ±0.03 K (3-wire)■ ±0.004 % + 2 µV for thermocouples

Data sheet

thermocouples and resistance thermometers■ Logger and scan functions■ Up to 44 channels possible
CT 60.15

CTR6500

AC resistancethermometry bridge■ Various coupler connectors connectable■ Complete automatic calibration routines

controllable

Data sheet

AC 87.01

CTR9000

Primary-standard
resistance thermometry
bridgeMeasuring range-200 … +962 °C

Measuring range-200 … +962 °C

Measuring range

0 … 260 Ω

Accuracy±3 mK (full range)

Accuracy

Accuracy

0.01 K, optional 0.005 K

Probe type

PRT, thermistors or fixed resistors

0.1 … 1.25 mK depending on resistanceratio

Probe type

SPRT, PRT or fixed resistors

Special feature■ Expendable to 60 channels (optional)

Probe type

SPRT, PRT or fixed resistors

Special feature■ Expendable to 60 channels (optional)

Special feature■ Expendable to 60 channels (optional)■ Internal resistors 25 Ω, 100 Ω,

10 kΩ, 100 kΩ

Data sheet

CT 60.30

CT 60.40■ 4 selectable standby currents possible(optional)■ Internal resistors 25 Ω, 100 Ω■ AC technology■ AC technology

Data sheet

110

CTS3000

Data sheet

CT 60.80

Standard reference resistors, AC/DC

Calibration

Calibration ‫ ׀‬Temperature

Electrical comparison standard
Reference resistors with high-accuracy, fixed resistance values, which are used in connection with resistance thermometry bridges. They
are also used as standards in accredited electrical laboratories.

CER6000-RR

CER6000-RW

Reference resistor

Resistance valueStandard reference resistor

1, 10, 25, 100, 300, 400, 500, 1,000 and
10,000 Ω

Resistance value

1, 10, 25, 100, 300, 400, 500, 1,000 and
10,000 ΩLong-term stability< ±5 ppm per year

Long-term stability

Special feature■ Low temperature coefficient±2 ppm per year
(HS version 0.5 ppm per year)

Special feature■ Low temperature coefficient

Data sheet

CT 70.30
Data sheet

CT 70.30■ Rugged stainless steel construction■ Rugged stainless steel construction

Connections of the reference resistor, model CER6000-RR

Red V AI A RedWhite V B

I B White

Reference resistor, model CER6000-RR with 100 Ω

Reference resistor, model CER6000-RR
with different resistance range

Further information at www.wika.com

111

Calibration ‫ ׀‬Accessories

Accessories
From individual components ... to complete turnkey kits
The following accessory components are the ideal complement
to the individual calibration instruments. Thus a complete solution
is not only quickly and easily configured, but can also be installedin the same manner. The various packages complete the productprogramme for calibration technology and can be used in manydifferent applications.
Customer-specific drilled inserts, silicone oil suited for calibrationin micro calibration baths and interface cables complete theproduct portfolio for temperature.
You can find a detailed description in our catalogue “Accessories
for calibration technology”.

112

Calibration technology

Accessories
for calibration technology

Pressure supply case

Pressure and vacuum supply
packages

Connection components

Pressure control

Calibration andadjustment tools

Temperature accessories

Calibration ‫ ׀‬Accessories

Calibration

Engineered solutions

We have been developing systems for use in our own group ofcompanies for years and can draw on our own process knowledge
to continually develop our systems further.
We offer robust and compact turnkey machinery from a single
source, with our own fixture construction and customer-specificsolutions as well as many application possibilities.

Test and calibration systems for workshops and laboratories
For the fitting-out of calibration laboratories, we offer individuallydesigned test workstations. Here we integrate proven calibration
systems from our extensive product range into ergonomic workstations. These can be individually equipped and combined with thefollowing components:

Further information at www.wika.com

 19" calibration racks in modular design for pressure sensors
 Connection columns with quick-release fasteners for test items

and references with exchangeable threaded inserts

 Electric and pneumatic power strips with 230-V voltage supply

and compressed air with air blow gun connection includingpressure regulator
 Work panel for setting the operating pressure with inlet pressure gauge, outlet pressure gauge and alternative pressure
supply PC workstations

113

Calibration ‫ ׀‬Engineered solutions

Test and calibration systems for production
The complete solutions are available in the widest range of
automation levels incl. tempering units, workpiece transportsystems, workpiece fixtures and electrical and pressure-side
contacting.
The focus is on the precise interaction of measurement
technology, testing system mechanics and control components. In
addition, the actual testing and adjustment processes can also be
combined with mounting and labelling processes.

Leak and pressure function test systems for production
We offer individual and turnkey solutions in various degrees
of automation for a wide variety of applications, from simple
test device through semi-automatic test benches to fully
automatic testing systems.
The testing processes can also be combined with assembly
processes, laser marking and automated parts handling(infeed/outfeed); in addition, the chaining of several stations
is possible.
Pneumatic or helium leak testing
on fittings, valves, hoses, coolers, pumps, filters and manyother test parts.
Pressure function tests or setting procedures
among other things for Control pressure of pressure reducers or thermostat
control valves The cracking pressure of safety relief valves Switch points of pressure switches and control valves
 Pressure containment of different components
Test methods

 Integral vacuum methods
 Accumulation methods (under atmosphere) Sniffing test

114

Calibration

Calibration ‫ ׀‬Engineered solutions

Customer-specific laser welding machines for production
Core elements of our turnkey concept for laser welding systems
are a modular axis system, both easily serviced and upgradeable,
as well as our own user-friendly, Windows-based control software,
for which no programming knowledge is required.

Your benefits

 We have strong and reliable partners for the laser sources with

continuous product development.

 Our systems are equipped with operator software for simple

and intuitive operation without needing CNC programming
knowledge.
 Our low-service axis concept can also be upgraded at a later
date thanks to the modular design of the axes.
 We can be at your side from as early as the preparation of yourrequirements specification document and offer you the opportunity to influence the entire development phase.

The GHP series features numerous functions and options: Camera systems can be integrated to check component posi






Model GHP-100
from the GHP series

tioning
External interfaces
CNC axes with servo drives
Automatic loading possible
2- to 5-axis kinematics
Automatic force-displacement controlled joining function
Automatic fixture recognition
Connection to the customer’s ERP system

Other models of the GHP series offer further special features.

Service for customer-specific systems Immediate help in case of faults

For the shortest response times andefficient problem analysis we offer
a remote service via smart glasses.
Using smart glasses, our specialists
can efficiently analyse the problem andquickly take targeted corrective action,
so you benefit from reduced downtime
and costs.

 Preventive maintenanceThrough regular system maintenance,premature wear can be prevented andthe risk of system downtime can be
minimised. We are happy to advise
you regarding the ideal maintenanceintervals and to design an individual
maintenance package for you.
Service hotline: +49 9372 132 5049

Testing & Automation Technology

Prüfstände und
Kalibrieranlagen

Extensive information can be found in our
brochure “Test benches and calibration
systems” at www.wika.com.

115

Service

Lasting impressions with reliable services

Choose from our comprehensive setof services

116

Installation and commissioning

Maintenance and repair

Analysis and support

WIKA’s field installation experts go to
the customers’ sites to provide tailoredsolutions that result in a short downtime.
Process reliability is guaranteed by expert
installation. Installations include multipoint
thermometers in reactors, thermocouples
in furnaces and level measurement instrumentation.

You can count on WIKA to do repairs– from diaphragm seal systems up tohighly accurate calibration instruments. We
support you in optimising your operational
processes. Benefit from our know-how forsolutions that are tailored to your needs.

WIKA offers reliable consultation services,
both analytical and technical, for a wide
range of industries. Our qualified service
technicians support in solving problems
and ensure that your measuring instrument
is back in working order in the shortest
possible time.

Calibration

Inspection and testing

WIKA provides its calibration services
on-site at your premises or in our laboratory, for WIKA as well as other instruments.
Pressure, temperature, mass, electrical,
force, dimensional, flow and torque aresome of the other calibrations and adjustments that we provide on shortest delivery
times.

You can rely on WIKA for on-site verification and function testing that is non-invasive and non-destructive. Our expertise
also includes in-situ verifications of multipoint thermometers.

Service

Service

Scan for more

Oil and gas

information

Chemical and petrochemical

For over 75 years, WIKA has helped industries all over theworld meet set industry benchmarks. With time, we took it uponourselves and our services engagement to not just meet but
also exceed those set benchmarks and expectations.
Power generation

Food and pharmaceutical

IIoT and digitalisation

Basic materials

We will always want to exceed your own expectations byproviding the best-in-class service. Moreover, our quality ofwork is backed by the strength of our OEM manufacturingexpertise.
To help you do more than just the best, we also ensure global
consistency To help you do more than just the best, we also
ensure global consistency, which means that you can count on
us for any service engagement, be it generic or customised innature, all over the world.

Down the street – around the world

Growing team with over 50 field service
technicians and supervisors worldwide
and already 15 mobile calibration vans
deployed in various countries around theworld.

With our ISO 17025 accredited calibrationlaboratory presence in over 20 countries,
we assure you that we will leave no stone
unturned to ensure global consistency andstandard of excellence.

We ensure high quality by professional
training and certification of our service
technicians. The observance of health andsafety aspects is very important to us.

117Industry-specific products
In our segment brochures you will find industry-specific know-how and special products
explicitly developed for specific application areas.

118

Ventilation and air-conditioning technology

Sanitary applications

Sensing technology for ventilation andair-conditioning
Our mechanical and electronic instruments areused for measuring and monitoring pressure,air flow, temperature, humidity and air quality.

Sanitary applications
Our measuring instruments optimally fulfil
requirements in terms of highest processreliability, hygienic design and the integration ofsensing technology into production plants.

SF6 gas solutions

High purity and ultra high purity

Power transmission and distribution
industry
WEgrid Solutions offers customised complete
solutions for plants filled with SF₆ gas.

Measurement solutions for semiconductor,
solar and light-emitting technologies
High purity, media resistance, leak tightness
and accuracy all make up the basic requirements for the development and production of
our measuring instruments for the semiconductor industry.
Further information at www.wika.com

Website and social media
Visit us on our website, in our online shop and on our social media channels.

WebsiteOnline shop

Blog

wika.comshop.wika.com

blog.wika.com

Find out about our wide range ofmeasurement technology and services,or market sectors. Download 3D drawings, technical documents or informative brochures.And please register for
our free newsletter!

Easy, quick and safe:
Directly select the right product for you
from our standard product portfolio.
Or adapt the instrument you want
exactly to your requirements with ourconfigurator.

In our blog, you can expect many
interesting articles on the theme of
measurement technology. Furthermore,
there are various insights into the worldof the WIKA Group.

WIKA on LinkedIn

WIKA on YouTube

WIKA on Instagram

linkedin.com/company/wikagroup

youtube.com/wikagroup

instagram.com/careeratwika/

Follow us on LinkedIn.
Don’t just follow our news on products
and applications, but also on important
events within the WIKA Group.

We are happy to welcome you to ourYouTube channel.
Here we don’t just promote ourcompany, but also present complex
technical contents, explained in a
simple and understandable way.

Follow us on Instagram to stay up todate on exciting career opportunities,
events, lotteries and much more.

Notes on the use of registered trademarks■ LoRaWAN® is a trademark used under licence from LoRa Alliance®.
■ The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG, Inc. and any
use of such marks by WIKA is under licence.
■ mioty® is a registered trademark of Frauenhofer IIS.
■ VARIVENT® is a registered trademark of the company GEA.
■ BioConnect® is a registered trademark of the company NEUMO.
Other brands and trademarks are the property of their respective owners.

119

WIKA worldwide
EuropeAustria
WIKA Messgerätevertrieb
Ursula Wiegand GmbH & Co. KG
Tel. +43 1 8691631
info@wika.at / www.wika.at
Benelux
WIKA Benelux
Tel. +31 475 535500
info@wika.nl / www.wika.nl
Bulgaria
WIKA Bulgaria EOOD
Tel. +359 2 82138-10
info@wika.bg / www.wika.bg
Croatia
WIKA Croatia d.o.o.
Tel. +385 1 6531-034
info@wika.hr / www.wika.hr
Denmark
WIKA Danmark A/S
Tel. +45 4581 9600
info@wika.as / www.wika.as
Finland
WIKA Finland Oy
Tel. +358 9 682492-0
info@wika.fi / www.wika.fi
France
WIKA Instruments s.a.r.l.
Tel. +33 1 71 68 10 00
info@wika.fr / www.wika.fr

Poland
WIKA Polska spółka z ogranizoną
odpowiedzialnością sp. k.
Tel. +48 54 230110-0
info@wikapolska.pl
www.wikapolska.pl
Romania
WIKA Instruments Romania S.R.L.
Tel. +40 21 4048327
info@wika.ro / www.wika.ro
RussiaAO “WIKA MERA”
Tel. +7 495-648018-0
info@wika.ru / www.wika.ru
Serbia
WIKA Merna Tehnika d.o.o.
Tel. +381 11 2763722
info@wika.rs / www.wika.rs
Spain
Instrumentos WIKA S.A.U.
Tel. +34 933 9386-30
info@wika.es / www.wika.es
Switzerland
WIKA Schweiz AG
Tel. +41 41 91972-72
info@wika.ch / www.wika.ch

Türkiye
WIKA Instruments
Endüstriyel Ölçüm Cihazları Tic. Ltd. Şti.
Tel. +90 216 41590-66
Germany
info@wika.com.tr
WIKA Alexander Wiegand SE & Co. KG www.wika.com.tr
Tel. +49 9372 132-0
info@wika.de / www.wika.de
UkraineTOV WIKA Prylad
IrelandTel. +38 044 496 83 80
WIKA Instruments Ireland Limited
info@wika.ua / www.wika.ua
Tel. +35 386 1449 360
info@wika.ie / www.wika.co.uk
United Kingdom
WIKA Instruments Ltd
Italy
Tel. +44 1737 644-008
WIKA Italia S.r.l. & C. S.a.s.
info@wika.co.uk / www.wika.co.uk
Tel. +39 02 93861-1
info@wika.it / www.wika.it

North America

Asia

Canada
WIKA Instruments Ltd.
Tel. +1 780 4637035
info@wika.ca / www.wika.ca

China
Botswana
WIKA Instrumentation Suzhou Co., Ltd. WIKA Instruments Botswana (Pty.) Ltd.
Tel. +86 512 6878 8000
Tel. +267 3110013
info@wika.cn / www.wika.com.cn
info@wika.co.bw / wika.co.bw

USA
WIKA Instrument, LP
Tel. +1 770 5138200
info@wika.com / www.wika.us

India
WIKA Instruments India Pvt. Ltd.
Tel. +1800-123-101010
info@wika.co.in / www.wika.co.in

Egypt
WIKA Near East Ltd.
Tel. +20 2 240 13130
info@wika.com.eg / www.wika.com.eg

Gayesco-WIKA USA, LP
Tel. +1 713 4750022
info@wikahouston.com
www.wika.us

Japan
WIKA Japan K. K.
Tel. +81 3 5439-6673
info@wika.co.jp / www.wika.co.jp

Namibia
WIKA Instruments Namibia Pty Ltd.
Tel. +26 4 61238811
info@wika.com.na / www.wika.com.na

Mensor Corporation
Tel. +1 512 3964200
sales@mensor.com
www.mensor.com

Kazakhstan
TOO WIKA Kazakhstan
Tel. +7 727 225 9444
info@wika.kz / www.wika.kz

Nigeria
WIKA WEST AFRICA LIMITED
Tel. +234 17130019
info@wika.com.ng / www.wika.ng

Latin America

Korea
WIKA Korea Ltd.
Tel. +82 2 869-0505
info@wika.co.kr / www.wika.co.kr

Saudi Arabia
WIKA Saudi Arabia Llc
Tel. +966 53 555 0874
info@wika.sa / www.wika.sa

Malaysia
WIKA Instrumentation (M) Sdn. Bhd.
Tel. +60 3 5590 6666
info@wika.my / www.wika.my

South Africa
WIKA Instruments Pty. Ltd.
Tel. +27 11 62100-00
sales@wika.co.za / www.wika.co.za

Philippines
WIKA Instruments Philippines Inc.
Tel. +63 2 234-1270
info@wika.ph / www.wika.ph

United Arab Emirates
WIKA Middle East FZE
Tel. +971 4 883-9090
info@wika.ae / www.wika.ae
Australia
Australia

Argentina
WIKA Argentina S.A.
Tel. +54 11 5442 0000
ventas@wika.com.ar
www.wika.com.ar
Brazil
WIKA do Brasil Ind. e Com. Ltda.
Tel. +55 15 3459-9700
vendas@wika.com.br
www.wika.com.br
Chile
WIKA Chile S.p.A.
Tel. +56 9 4279 0308
info@wika.cl / www.wika.cl
Colombia
Instrumentos WIKA Colombia S.A.S.
Tel. +57 601 7021347
info@wika.co / www.wika.co
MexicoInstrumentos WIKA Mexico S.A. de C.V.
Tel. +52 55 50205300
ventas@wika.com / www.wika.mx

Singapore
WIKA Instrumentation Pte. Ltd.
Tel. +65 6844 5506
info@wika.sg / www.wika.sg
Taiwan
WIKA Instrumentation Taiwan Ltd.
Tel. +886 3 420 6052
info@wika.tw / www.wika.tw
Thailand
WIKA Instrumentation Corporation(Thailand) Co., Ltd.
Tel. +66 2 326 6876
info@wika.co.th / www.wika.co.th

Africa/Middle East

Australia
WIKA Australia Pty. Ltd.
Tel. +61 2 88455222
sales@wika.com.au / www.wika.com.au
New Zealand
WIKA Instruments Limited
Tel. +64 9 8479020
info@wika.co.nz / www.wika.co.nz

Uzbekistan
WIKA Instrumentation FE LLC
Tel. +998 71 205 84 30
info@wika.uz / www.wika.uz

WIKA Alexander Wiegand SE & Co. KG
Alexander-Wiegand-Straße 30 | 63911 Klingenberg | Germany
Tel. +49 9372 132-0 | info@wika.de | www.wika.de

You can find further
information here!

12393852 03/2024 EN

www.wika.com