Elcometer 6910
elcometer.com

Flash Point1

Flash Point
Elcometer 6910

Setaflash ‘Series 8’ Closed Cup Tester
The Elcometer 6910 Setaflash ‘Series 8’ is an automated closed cup tester with atemperature range of ambient to 300°C (ambient to 572°F).
Test parameters, instrument status and flash point results are shown on the brightcolour LCD display - all easily accessed via menus, navigated by the keypad, which
guides the operator through a straightforward testing process.
The instrument uses an electric hot wire ignitor and is designed with two modes offlash point operation; Flash/No Flash and Ramp. Ramp Mode increases the sample
temperature at the rate of 2°C/min and automatically dips the ignitor every 1°C up
to 100°C (every 2°C above 100°C), until either a flash is detected or the end of test
temperature is reached.
During the test the automatic flash point is set to ignore halo and other false flash
effects reducing the risk of inaccurate test results and enhancing repeatability.
At the end of the flash point test, the sample cup is cooled by forced air to reducethe cycle time between tests.

STANDARDS:

Up to 64 results can be stored in the memory which can be downloaded to a PC orprinter via USB for further reporting.

ASTM D3278, ASTM D3828,
ASTM D7236, ASTM E502,
BS 2000-523, CLP Regulations EC
No 1272/2008, EPA 1020 A & B,
ISO 3679, ISO 3680,
UN Class 3 Non-viscous FlammableLiquids•••••

Small sample size, 2ml (0.067fl oz) or 4ml (0.135fl oz)
Flash/No Flash & Ramp Modes
Electric ignitor
Automatic dipping and flash detection
Forced air cooling for rapid test cycling

Technical Specification
Part NumberDescription

K6910M020

Elcometer 6910 Setaflash ‘Series 8’ Closed Cup Tester

Temperature Range

Ambient to 300°C (Ambient to 572°F)

Sample Size

2ml (0.067fl oz) or 4ml (0.135fl oz) according to method

Test Modes

Rapid Equilibrium and Ramp

Test Duration
Rapid Equilibrium Mode

1 minute below 100°C (212°F),
2 minutes above 100°C (212°F)

Test Duration Ramp Mode

Typically 7 minutes

Cool Down Time

80°C to 20°C (176°F to 68°F) in 4 minutes

Cup Material

AluminiumDimensions

300 x 340 x 380mm (11.8 x 13.4 x 15.0”)

Weight8kg (17.6lb)

Packing List

Elcometer 6910 Setaflash ‘Series 8’ Closed Cup Tester, mains power cable, 2ml (0.067fl oz)
syringe, programming plug, 2mm Allen key, 2.5mm Allen key, ‘O’ ring fitted, certified flash pointmaterial (50ml/1.7fl oz), quality assurance certificate and operating instructions

Accessories
KT006910N009

Certified Flash Point Material (50ml/1.7fl oz)● Calibration Certificate supplied as standard.

2Certificate●Flash Point
Setaflash ‘Series 8’ Active-Cool Closed Cup Tester

Elcometer 6910

The Elcometer 6910 Setaflash ‘Series 8’ Active-Cool Closed Cup Tester offers thesame features as the Setaflash ‘Series 8’ Closed Cup Tester but is designed witha temperature range of -30°C to 135°C (-22°F to 275°F) with no external coolingrequired and operates with an electric ignition.
This Active Cool Closed Cup Tester uses the Peltier cell technology for heating andcooling, ensuring precise temperature control and rapid cool down after a test iscompleted.
••••
•
•

Small sample size, 2ml (0.067fl oz) or 4ml (0.135fl oz)
Flash/No Flash & Ramp ModesElectric ignitor
Automatic dipping and flash detection
Active-Cool electronic Peltier cooling
1GB internal memory & intergrated USB port

STANDARDS:ASTM D3278, ASTM D3828,
ASTM D7236, ASTM E502,
BS 2000-523,
CLP Regulations EC No 1272/2008,
EPA 1020 A & B, ISO 3679, ISO 3680,
UN Class 3 Non-viscous Flammable
Liquids

Technical Specification
Part Number

Description

Certificate

K6910M021

Elcometer 6910 Setaflash ‘Series 8’ Active-Cool Closed Cup Tester

Temperature Range-30°C to 135°C (-22°F to 275°F)●Sample Size

2ml (0.067fl oz) or 4ml (0.135fl oz) according to method

Test Modes

Rapid Equilibrium and Ramp

Test Duration
Rapid Equilibrium Mode

1 minute below 100°C (212°F), 2 minutes above 100°C (212°F),
FAME 1 minute or user defined 1 to 99 minutes

Test Duration Ramp Mode

Typically 7 minutes

Cool Down Time

80°C to 20°C (176°F to 68°F) in 4 minutes

Cup Material

Aluminium

Dimensions

300 x 340 x 380mm (11.8 x 13.4 x 15.0”)

Weight8kg (17.6lb)

Packing List

Elcometer 6910 Setaflash ‘Series 8’ Active-Cool Closed Cup Tester, 2ml (0.067fl oz) syringe, ignitor,mains power cable, programming plug, 2mm Allen key, 2.5mm Allen key, certified flash point material(50ml/1.7fl oz), certificate of calibration, warranty registration form and operating instructions

Accessories
KT006910N009

Certified Flash Point Material (50ml/1.7fl oz)● Calibration Certificate supplied as standard.

elcometer.com

3

Flash Point
Elcometer 6910

Setaflash ‘Series 8’ Active-Cool (Corrosion Resisting)
The Elcometer 6910 Setaflash ‘Series 8’ Active-Cool (Corrosion Resisting) offers
the same feature as the Active Cool Closed Cup Tester but comes with a corrosion
resisting insert in the cup for testing corrosive samples.
It has been designed to carry out flash/no flash tests with a temperature range of-30°C to 135°C (-22°F to 275°F) with no external cooling required.
There is also the Ramp Mode which increases the sample temperature at
the rate of 2°C/min and automatically dips the ignitor every 1°C up to 100°C(every 2°C above 100°C), until either a flash is detected or the end of test temperatureis reached.
•••••••

STANDARDS:Small sample size, 2ml (0.067fl oz) or 4ml (0.135fl oz)
Flash/No Flash & Ramp Modes
Electric ignitor
Automatic dipping and flash detection
ActiveCool electronic Peltier cooling
1GB internal memory & intergrated USB port
Corrosion resisting insert

ASTM D3278, ASTM D3828,
ASTM D7236, ASTM E502,
BS 2000-523,
CLP Regulations EC No 1272/2008,
EPA 1020 A & B, ISO 3679,
ISO 3680, UN Class 3 Non-viscous
Flammable Liquids

Technical Specification
Part Number

Description

K6910M022

Elcometer 6910 Setaflash ‘Series 8’ Active-Cool (Corrosion Resisting)Temperature Range-30°C to 135°C (-22°F to 275°F)

Sample Size2ml (0.067fl oz) or 4ml (0.135fl oz) according to method

Test Modes

Rapid Equilibrium and Ramp

Test Duration
Rapid Equilibrium Mode

1 minute below 100°C (212°F), 2 minutes above 100°C (212°F),
FAME 1 minute or user defined 1 to 99 minutes

Test Duration Ramp Mode

Typically 7 minutes

Cool Down Time

80°C to 20°C (176°F to 68°F) in 4 minutes

Cup Material

Corrosion Resistant Steel Insert, Aluminium CupDimensions

300 x 340 x 380mm (11.8 x 13.4 x 15.0”)

Weight8kg (17.6lb)

Packing List

Elcometer 6910 Setaflash ‘Series 8’ Active-Cool (Corrosion Resisting), 2ml (0.067fl oz) syringe,
flash detector, mains power cable, programming plug, 2mm Allen key, 2.5mm Allen key, certified
flash point material (50ml/1.7fl oz), certificate of calibration, warranty registration form andoperating instructions

Accessories
KT006910N009

Certified Flash Point Material (50ml/1.7fl oz)● Calibration Certificate supplied as standard.

4Certificate●elcometer.be • elcometer.fr • elcometer.deelcometer.nl • elcometer.jp • elcometer.com.sg© Elcometer Limited, 2020. Alle rechten voorbehouden. Niets van dit Document mag worden gereproduceerd, overgedragen, getranscribeerd, opgeslagen (in een retrievalsysteem of anderszins) of vertaald in enige taal, in enige vorm of door
enig middel, zonder de voorafgaande schriftelijke toestemming van Elcometer Limited.

elcometer.com

5