5On the web: https://phoenixmicron.com

Email us at: sales@phoenixmicron.com

Exceptional retinal microscopy
The all-new Phoenix MICRON 5 delivers aperformance breakthrough for small animal
ocular imaging.
Designed to enhance efficiency andrepeatability, this new generation ofMICRON technology boasts a range of corefeatures, including interchangeable imaging
sensors and software-controlled camera
operation, expanded filter capabilities, andnew depth of field adjustments. Endless
opportunities at your fingertips.
New LT2 Lens Technology
Objective lenses slide on to the camera andmagnetically snap into place, making iteven faster and easier to switch from one
imaging modality to another with precisionalignment. This is ideal for workflows suchas verifying a laser burn with OCT.
Designed for in vivo small animal
ophthalmic research
Designed specifically for the challenges ofrodent eye and eye-brain research, MICRON
cameras have been used to image subjects
from zebrafish to small rabbits.

© 2023 Phoenix-Micron, Inc. All rights reserved. MICRON is a registered trademark of Phoenix-Micron, Inc.
MICRON products are covered by one or more issued or pending patents, including US patent 7,993,000

5On the web: https://phoenixmicron.com

Email us at: sales@phoenixmicron.com

Features:• Mouse retina resolution below 3 μm
• Live, real-time fundus view
• Captures still images and videos• Adjust aperture to tune depth of field• Control exposure for dark and light retinas
with light intensity, aperture, digital gain,
and frame rate
• Software-controlled focus, illumination,
aperture and filter selection• Record and manage animal subject details• Add user-defined tags to subject animals,
imaging sessions, and images for enhanced
search and analysis• Extract still images from videos• Review, zoom, compare, and export
images• Wide rage of excitation and emission
filters for fluorescent imaging• Easily switch between modalities andobjectives lenses with new LT2 slide-on /slide-off lenses© 2023 Phoenix-Micron, Inc. All rights reserved. MICRON is a registered trademark of Phoenix-Micron, Inc.
MICRON products are covered by one or more issued or pending patents, including US patent 7,993,000

5On the web: https://phoenixmicron.com

Email us at: sales@phoenixmicron.com

High-resolution, High-contrast Images

Color Fundus
Angiography
Fluorescence© 2023 Phoenix-Micron, Inc. All rights reserved. MICRON is a registered trademark of Phoenix-Micron, Inc.
MICRON products are covered by one or more issued or pending patents, including US patent 7,993,000

5Powered by MICRON Software Suite

On the web: https://phoenixmicron.com

Email us at: sales@phoenixmicron.com

Share the MICRON 5 Across Labs or Teams
Share a single MICRON imaging system with multiple labs while
controlling which users can access data from each labLive Fundus View; Capture Still Images and Video
Real-time fundus view with ability to capture videos, extract frames
from videos, and capture still images

Tag, Search, Review, and Export
Rich data and image management tools improve data and image analyticvalue© 2023 Phoenix-Micron, Inc. All rights reserved. MICRON is a registered trademark of Phoenix-Micron, Inc.
MICRON products are covered by one or more issued or pending patents, including US patent 7,993,000

5On the web: https://phoenixmicron.com

Email us at: sales@phoenixmicron.com

Supports a Wide Range of MICRON Imaging
Modality Add-ons

Image-Guided OCT2

Image-Guided Laser

Image-Guided Focal ERG

Slit Lamp

45014+17

Published papers thatincorporate MICRON data

Years of experienceinnovating patented
small animal imaging
technology

Integrated multimodality system

Imaging modalities,
designed for exacting
small animal ophthalmicresearch© 2023 Phoenix-Micron, Inc. All rights reserved. MICRON is a registered trademark of Phoenix-Micron, Inc.
MICRON products are covered by one or more issued or pending patents, including US patent 7,993,000

5On the web: https://phoenixmicron.com

Email us at: sales@phoenixmicron.com

Specification

Details

Retinal imaging resolution

3 μm or better (mouse) 6 μm (rat)

Depth of focus

Controlled by variable aperture settings

Range of focus

Retinal surface to crystalline lensField of view

50 degrees; 1.8mm (mouse) 3.6mm (rat)Dynamic imaging rate

30 fps to 2 fps

Imaging dynamic range

60 dB

Image formats

JPEG, TIFF, AVI

Light source

Xenon, 400 nm to 850 nm

Filters

4 filter positions in excitation and emission filter wheels. Includes
GFP filter set. Wide range of available filters sets.

Animal stage

2 degrees of rotation; 3 degrees of translation

Camera head standX-foot platform with vertical and tilt camera adjustment

Operating system

Windows 11Monitor

Wide screen 22” LCD

PowerAuto-switching 120 VAC / 230 VAC, 50 Hz or 60 Hz

Accessories

Keyboard, mouse, foot switch, dust cover, mouse pad, lenscleaning tissue, water bottle, coupling gel starter pack,Phoenix-Micron, Inc. | 543 NW York Drive, Suite 100 | Bend, OR 97703 | USA | +1.541.668.7539© 2023 Phoenix-Micron, Inc. All rights reserved. MICRON is a registered trademark of Phoenix-Micron, Inc.
MICRON products are covered by one or more issued or pending patents, including US patent 7,993,000