MATERIAL DATA SHEETDENTAL

PolyJet™ Dental Materials
Developed for the DentaJet Series multi-material 3D printing product line, our materials are engineered toproduce highly accurate and consistent dental parts at scale.
MED610™ Bio-compatible material
Rigid, transparent dental resin featuring high dimensional stability. Intended for printing surgical guides andremoval partial denture casting patterns.
VeroGlaze™ MED620
Biocompatible VeroGlaze™ MED620 is a rigid, opaque dental resin featuring high dimensional stability andaccuracy. Intended for printing dental models.
MED625FLX™ biocompatible material
Flexible, biocompatible material for temporary in-mouth placement. Ideal for gingival masks.
SUP711S™
Gel like support material was specially designed to uphold overhangs and complicated geometries on DentaJet
Series 3D printing LED curing process.
TrueDent™
FDA cleared (Class II) light-curable resin indicated for the fabrication of dental appliances. Compatible only with
the J5 DentaJet® 3D printer. Indicated for the fabrication of dental appliances including but not limited toremovable full dentures and temporaries*.

Property

Standard/Procedure

Clear Biocompatible MED610

Biocompatible VeroGlaze MED620

Tensile Strength (MPa)D-638-03

50 – 65

55 – 65

Elongation at break (%)

D-638-05

10 – 25

15 – 25

Modulus of elasticity (MPa)D-638-04

2,000 – 3,000

2,300 – 3,300

Flexural Strength (MPa)D-790-03

75 – 110

80 – 100

Flexural Modulus (MPa)D-790-04

2,200 – 3,200

2,300 – 3,200

HDT 0.45 MPa (ºC)

D-648-06

45 – 50

45 – 50

HDT 1.82 MPa (ºC)

D-648-07

45 – 50

45 – 50

Izod Notched Impact (J/M)

D-256-06

20 – 3020 – 30

Water Absorption (%)

D-570-98 24HR

1.1 – 1.5

.1 – 1.5

Tg (ºC)

DMA E

52 – 54

52 – 54

Shore Hardness (D)Scale D

83 – 86

83 – 86

Rockwell Hardness (scale M)

Scale M

73 – 76

73 – 76

Polymerized density (gr/cm )ASTM D792

1.17 – 1.18

1.17 – 1.18

Bio-compatibility

DIN EN ISO 10993 -1:2009

Approved

Approved

Support Removal Type–

WaterJet

WaterJet3

2

Dental Materials
Property

Standard/ Procedure

MED625FLX

Tensile Strength (MPa)D-412

3–5Elongation at break (%)

D-412

45 – 55%

Compressive Set (%)

D-395

0.5 – 1.5%

Shore Hardness (Scale A)

D-224073 – 77

Tensile Tear Resistance (kg/cm)

D-624

8 – 12Polymerized density (gr/cm )D-792

1.16 – 1.17 g/cm3

Bio-compatibility

EN ISO 10993-1

Approved

Support Removal Type–

WaterJet

3

TrueDent Specifications
Property

Requirement

Value

Compliance

Ultimate flexural strength (MPa)

≥65
≥50

≥85
≥94

ISO 20795-1:2013
ISO 10477:2020

Flexural modulus (MPa)

≥2000

≥2300

ISO 20795-1:2013

Water Sorption (µg/mm^3)

≤32
≤40

≤32
≤28

ISO 20795-1:2013
ISO 10477:2020

Solubility (µg/mm^3)≤1.6≤1.1ISO 20795-1:2013

Part NumberQuantity

Product Description

OBJ-09164

Pack of 1

TDM100, TrueDent Clear, 1.1 KG

OBJ-09165

Pack of 1

TDM120, TrueDent Cyan, 1.1 KG

OBJ-09166

Pack of 1

TDM130, TrueDent Magenta, 1.1 KG

OBJ-09168

Pack of 1

TDM110, TrueDent White, 1.1 KG

OBJ-09169

Pack of 1

TDM140, TrueDent Yellow, 1.1 KG

OBJ-09167

Pack of 1

TDS100, TrueDent Support, 1.1 KG* For a full list of indications please refer to the TrueDent brochure.

Stratasys Headquarters
7665 Commerce Way,
Eden Prairie, MN 55344+1 800 801 6491 (US Toll Free)+1 952 937-3000 (Intl)+1 952 937-0070 (Fax)

1 Holtzman St., Science Park,PO Box 2496
Rehovot 76124, Israel+972 74 745 4000
+972 74 745 5000 (Fax)stratasys.com
ISO 9001:2015 Certified© 2024 Stratasys. All rights reserved. Stratasys is a registered trademark of Stratasys Inc. PolyJet, MED610, MED620, SUP711, MED625FLX, TrueDent,
VeroGlaze, VeroDent and VeroDentPlus are trademarks of Stratasys, Inc. All other trademarks are the property of their respective owners, and Stratasys assumes
no responsibility with regard to the selection, performance, or use of these non-Stratasys products. Product specifications subject to change without notice.
Printed in the USA. MDS_PJD_DentalMaterials_0524a