SERVOFLEX

Micro i.s. (5100 i.s.)
HIGH PERFORMANCE OXYGEN, CARBON MONOXIDE AND CARBON DIOXIDE PORTABLE GASANALYSIS, DESIGNED FOR USE IN HAZARD RATED LOCATIONS INCLUDING ZONE 0/DIVISION 1

FlexibleIntrinsically Safe (i.s.) design
permits use in any hazard rated
location including Zone 0/Division 1	Pumped or non-pumped(AFCD) functions
IP65 rating	Gas analysis for O2, CO or CO2

EASY TO USE
Intuitive, engineer-friendlyinterface and icons

SERVOFLEX
Micro i.s. (5100 i.s.)
Designed to meet the demands ofthe most dangerous environments,
Micro i.s. delivers ultra-sensitive
percentage measurements for oxygen,carbon monoxide and carbon dioxide.
A tough, life-proof design andfeature-rich performance ensuresmaximum operational flexibility.
Micro i.s. is Intrinsically Safe (i.s.) and isable to analyze both flammable and toxic
samples. This intelligent design means thatno purge is required prior to device use.
Micro i.s. also provides an intuitive userinterface, the ability to store up to
200 measurements and up to 18 hours
battery run time.

E rgonomic design
with carry strap

S tores up to 200 measurements
for subsequent download

Low cost of ownership
Uses non-depleting sensor
technologies that reduce ongoing
maintenance costs	Advanced Li-Ion rechargeable
batteries as standard(up to 18 hour run time)

Unrivalled performance ses industry-leading patented
U
Paramagnetic and IR technologies
and algorithms for ultra-sensitive
and accurate measurements

 anufactured by Servomex - overM
60 years’ experience innovating
and pioneering gas analysis withthousands of units used in thefield every year

Benchmark compliance TEX: Ex ia (Intrinsically Safe)
A_ Ta<_ +50°C)
IIC T4 Ga (-10°C<_ Ta<_ +122°F)(+14°F<IEC: Ex ia (Intrinsically Safe)_ Ta<_ +50°C)
IIC T4 Ga (-10°C<_ Ta<_ +122°F)(+14°F<FM: IS/I/1/ABCD/T4;_ Ta<_ +50°C: IP65 (Division 1)-10°C<Learn more about the SERVOFLEX Micro i.s. (5100 i.s.)
VISIT SERVOMEX.COM

F M: IS/I/0/AEx ia/IIC/T4;_ Ta<_ +50°C: IP65 (Zone 0)-10°C<CSA: IS/I/1/ABCD/T4;_ Ta<_ +50°C: IP65 (Division 1)-10°C< SA: IS/I/0/Ex ia/IIC/T4;
C_ Ta<_ +50°C: IP65 (Zone 0)-10°C<PRODUCT OVERVIEW: Micro i.s. (5100 i.s.)

Enhanced safety for the mostdangerous locations

ALTERNATIVE PRODUCTS
The SERVOFLEX portable product range
features a number of options designed tomeet your application needs.

When you work in environments where potentially explosive
atmospheres may be present, you need the most robust analyticalsolutions that enhance safety and provide efficient, engineer-friendly
gas measurements.
In applications like catalytic regeneration, decoke cycle, combustion
optimization and hazardous area process monitoring, i.s. certified
solutions help reduce costs and improve efficiency. No matter what yourapplication monitoring requirements are, you’ll want a device that offers
feature-rich performance, long battery runtime, low operational costs,
simplified ongoing maintenance and ease of use. And we don’t believe
you should have to compromize.

MiniMP(5200)

MiniMP combines high powered CO2 andO2 monitoring capabilities with ease of useand minimum maintenance requirements.
(EN15267-3 MCERTs approved), providing ahigh performance solution suited to quality
testing in safe areas.

A no compromiZe solution
The Micro i.s. combines intuitive user interaction and ansafety-enhanced i.s. design with ultra-sensitive, industry-leading
O2, CO and CO2 monitoring capabilities, providing the ideal portable
gas analysis solution for hazardous applications.
With flexible options including pumped and non-pumped formats
and a range of features designed to further simplify sample testing,the Micro i.s. doesn’t just meet requirements: it adapts perfectly todeliver the efficiency and usability your job demands.

MiniHD(5200)

Works as hard as you do
Optimized to deliver a tough and hard-working solution withmaximized uptime, the Micro i.s. comes with high-grade, long-life
Li-Ion rechargeable batteries as standard. An integrated digital LCD
notepad also allows up to 200 measurements to be stored on the device,permitting you to work and test with maximized efficiency.

Designed to withstand harsh conditions, thisCO2, CO and O2 analysis unit has an IP65
rating, protective, durable overmoulding andleading edge detection capabilities for safearea locations.

KEY APPLICATIONSHazardous area combustion optimization
Natural gas processingRefineries – catalytic cracker regeneration
HyCO applications

TECHNOLOGIES racker decoke cycle onC
petrochemical plantProcess monitoring
Inerting applications

PRODUCT DATA: Micro i.s. (5100 i.s.)

OPTIONSDESCRIPTIONSPECIFICATION

Rechargeable batteries

Advanced Lithium Ion (Li-Ion): Li-ion (02) 5110 pumped

16.5 hours run time (50°C/122°F)
14.8 hours run time (20°C/68°F)
10.5 hours run time (-5°C /23°F)
9.7 hours run time (-10°C /14°F)

Rechargeable batteries

Advanced Lithium Ion (Li-Ion): Li-ion (IR) 5110 pumped

10.5 hours run time (50°C/122°F)
9.7 hours run time (20°C/68°F)
7.4 hours run time (-5°C /23°F)
7 hours run time (-10°C /14°F)

Rechargeable batteries

Advanced Lithium Ion (Li-Ion): Li-ion (O2) 5111 AFCD

18 hours run time (50°C/122°F)
18 hours run time (20°C/68°F)

Rechargeable batteries

Advanced Lithium Ion (Li-Ion): Li-ion (IR) 5111 AFCD

11.4 hours run time (50°C/122°F)
11.4 hours run time (20°C/68°F)
9 hours run time (-10°C /14°F)

Power supply*

Device charging is only permitted in a safe area

100-240V charger supplied as standard (device is notdesigned to operate from main power source)

Alarms*

Flexible, user-configurable options

Concentration alarms indicated by a LED, icon display
and audible sounder

ACCESSORIESDESCRIPTIONSPECIFICATION

QuickConnect barb fitting

Simplify connection of gases with QuickConnect

Zero and span gas, 1/4” NPTSampling probe

Facilitates simplified sample drawing for continuous
flow rate

25cm with protective tip 180°C / 356°F max temp.)
100 cm untipped

Optional connector

Permits quick connection

1/8” NPT thread fitting

2 years spares packTwo years hassle-free operation

5x filters, 1 filter O-ring

Carry case

Robust and life-proof case for added protection

Black finish, ergonomic design with strap

Transport caseFor use in safe areas only

Keeps your device safe in an impact resistant case

MONITORING PERFORMANCESensor

O2

O2

COCO2

Technology

Paramagnetic

Paramagnetic

InfraredInfraredVariant

Standard (1 dp)

High Accuracy (2 dp)––

Full scale and Min. output

0-100%
0-10% (min.)

0-100%0-1% (min.)

0-5% CO

0-10%, 25%, 50%,100% CO2

316 Stainless Steel

316 Stainless Steel

316 Stainless Steel

316 Stainless SteelAccuracy (intrinsic error)±0.1% O2

±0.05% O2±2% FSR±2% FSR

Zero drift/week±0.4% O2

±0.2% O2

±4% FSR±4% FSR<15<15<15<15-10 to +50°C(14 to 122°F)-10 to +50°C(14 to 122°F)-10 to +50°C(14 to 122°F)-10 to +50°C(14 to 122°F)

Cell construction

T90 in secs.
Operating Temp.

Please note: Samples in excess of 21% Oxygen should not be measured in hazardous area. We recommend a calibration of the analyser after each power up.
Allow 1 hour warm up to meet performance specification. Please contact us for further information. *Standard feature

PRODUCT DATA:
Micro i.s. (5100 i.s.)

SAMPLE FOR MEASUREMENTS
Sample gas

Clean and dry at ambient temperature and free from particulates <0.6µm

Flow Control

AFCD maximizes measurement stability over the range – 1.5 – 6 ltr (0.05 – 0.2 ft3)

Sample inlet connection

QuickConnect barb fitting – 5mm OD stub for 6.3mm (1/4”) ID tube or adaptor to 1/4” NPT fitting option

Sample outlet connection

5mm OD stub (sample and bypass)

Inlet pressure

Pumped: 80kPa absolute (16.0psia) minimum / 110kPA absolute (16.0psia)

DEVICE SPECIFICATION

DEVICE SCHEMATIC

Size: 	
	160mm (6.3”) Width x 185mm(7.2”) High x 140mm (7.2”) Deep
Weight:	
1.8kg – 2.3kg (4lbs – 5.1lbs)
Certifications: 	ATEX: Ex ia (Intrinsically Safe)_ Ta<_ +50°C)
IIC T4 Ga (-10°C<_ Ta<_ +122°F)(+14°F< 	IEC: Ex ia (Intrinsically Safe)_ Ta<_ +50°C)
IIC T4 Ga (-10°C<_ Ta<_ +122°F)(+14°F< 	FM: IS/I/1/ABCD/T4;_ Ta<_ +50°C: IP65 (Division 1)-10°C< 	FM: IS/I/0/AEx ia/IIC/T4;_ Ta<_ +50°C: IP65 (Zone 0)-10°C<

PBMicro i.s. Rev.0 Date: 08/14

 	CSA: IS/I/1/ABCD/T4;_ Ta<_ +50°C: IP65 (Division 1)-10°C<
 	CSA: IS/I/0/Ex ia/IIC/T4;_ Ta<_ +50°C: IP65 (Zone 0)-10°C< marked and in compliance
CEwith ATEX, EMC and WEE
Directives scope UL approved andCE marked 100-240V / 43-70Hz
AC power supply

These analyzers are not intended for any form of use on humans and are not medical
devices as described in the Medical Devices Directive 93/42EEC.
Please note: This document was updated in August 2014. While every effort has been made toensure accuracy, no responsibility can be accepted for errors or omissions. Data may change, as wellas legislation, and you are strongly advised to obtain copies of the most recently issued regulations,standards and guidelines. This document is not intended to form the basis of a contract.

AMERICAS BUSINESS CENTER
Tel: +1 281 295 5800
Email: americas_sales@servomex.com

EUROPEAN BUSINESS CENTRE
Tel: +31 (0) 79 330 1580
Email: europe_sales@servomex.com

INDIA BUSINESS CENTER
Tel: +91 22 3934 2700
Email: MEI_sales@servomex.com

ASIA PACIFIC BUSINESS CENTRE
Tel: +86 (0)21 6489 7570
Email: asia_sales@servomex.com

LATIN AMERICA BUSINESS CENTER
Tel: +55 11 5188 8166
Email: brazil_sales@servomex.com

MIDDLE EAST BUSINESS CENTER
Tel: +971 6552 8073
Email: MEI_sales@servomex.com

Servomex has a policy of constant product improvement and reserves the right to change specifications without notice.
© Servomex Group Limited. 2014. A Spectris company. All rights reserved.