B/W CCD Camera
CleverDragon series
CSCS20BC2
Operation Manual
Thank you for purchasing our B/W CCD camera. This operation manual contains many important informationsuch as how to use this product correctly and safely. Please read through this manual carefully. After reading,
keep this manual by the side of this product for your future reference.

BEFORE USE - GENERAL SAFETY INSTRUCTIONS
Read the following safety precautions carefully before using this product. These instructions containvaluable information on safe and proper use that will prevent harm and damage to the operator and otherpersons. Make sure that you fully understand the following details (indications, graphic symbols) before
proceeding to the remaining sections in this manual.
OWNER’S RECORD
Please fill in the blank below the model name and product serial number, which is found on bottom chassisof your device. Keep this number for your record.
Model Name
Serial No.
Indication definitions
Indication

WARNINGCAUTION

Meaning
This indicates the existence of a hazard that death or catastrophic bodily injury(*1)
may result from improper use.
This indicates the existence of a hazard that bodily injury (*2) or property damage(*3)
may result from improper use.

Notes
*1 Catastrophic bodily injury means loss of eyesight, burns (high and low temperature), shock, fracture,poisoning, etc. which leaves a sequela and require hospitalization or prolonged treatment.
*2 Bodily injury means injuries, burns and electric shock which does not require hospitalization or
prolonged treatment.
*3 Property damage means extended harm to home, household effects, domesticated animals, and pets.
Graphic symbol definitions
Indication

Meaning
This mark indicates a prohibited action that must not be carried out. The actual prohibited
action is indicated in the symbol or nearby graphically or described in text.
This mark indicates a mandatory action that must not be carried out. The actual instruction isindicated in the symbol or nearby graphically or described in text.
Handling Precautions

WARNING
Stop operation immediately when any abnormality or defect occurs.
Use during an abnormal condition; such as emitting smoke, burning odors, damage fromdropping invasion of foreign objects, etc. may cause fire and/or electric shock. Be always sure
to disconnect the connection cable from the camera connector at once and contact your dealer.
Do not operate in places with possibility of becoming wet.
This may cause fire and/or electric shock.
Do not repair, disassemble and/or modify by yourself.
This may cause fire and/or electric shock. Be always sure to contact your dealer for internal
repair, check and cleaning of the product.
Don’t place things or materials on the unit.
Ingress of foreign materials such as metallic things and liquid into the unit may cause a fire oran electric shock.
Do not put the product in an unstable, slanting and/or vibrated place.
Drop and/or fail of the product may cause injury.
Do not touch the connection cables during a thunderstorm.
This might cause electric shock.
Use the specified power supply.
Use of an unspecified power supply may result in fire or electric shock.
Do not be handled roughly, damaged, fabricated, bent forcefully, pulled, twisted, bundled,placed under heavy objects or heated the connection cable.
Otherwise, fire or electric shock may result.

CAUTION
Note the following instructions when installing.
-Do not wrap the product in an inflammable material, such as cloth.
-Do not put the product in a narrow space, since the heat generated from the product may bedifficult to emanate.
If you do not follow the above, the heat generated by the product may cause fire.
Avoid setting in humid, smoky, vaporized or dusty places. A fire or an electric shock may
occur in such places.
This may cause fire and/or electric shock.
Do not put the product in direct sunshine and/or high temperature.
The temperature inside the product may cause fire.
Use the specified connection cable.
Otherwise, a fire or an electric shock may occur.
Turn OFF the power in the case of connection.
Turn OFF the power in the case of connection of cable.
Otherwise, an electric shock or a malfunction may occur.
Do not expose its camera head to any intensive light (such as direct sunlight).
Otherwise, its inner image pickup device might get damaged.
Avoid short-circuiting signal output.
Otherwise, a malfunction may occur.
Avoid giving a strong shock against the camera body.
It might cause a breakdown or damage.
If your camera is used in a system where its camera connector is subjected to strong repetitive
shocks, its camera connector is possible to break down. If you intend to use your camera insuch a situation, if possible, bundle and fix a camera cable in the place near the camera, and do
not transmit a shock to the camera connector.
Ask your dealer to perform a periodical check and internal cleaning (approx. once every
five years).
Dust inside the product may cause fire and/or trouble. For check and cleaning cost, please
consult your dealer.

DISCLAIMER (LIMITED WARRANTY)
We disclaim any responsibility and shall be held harmless for any damages or losses incurred by the user inany of the following cases;
 Fire, earthquake or any other act of God; acts by third parties; misuse by the user, whether intentional or
accidental; use under extreme operating conditions.
 Malfunction or non-function resulting in indirect, additional or consequential damages, including but notlimited to loss of expected income and suspension of business activities.
 Incorrect use not in compliance with instructions in this instruction specifications and manual.
 Malfunctions resulting from misconnection to other equipment.
 Repairs or modifications made by the user or caused to be made by the user and carried out by an
unauthorized third party.
 Notwithstanding the foregoing, Teli’s liabilities shall not, in any circumstances, exceed the purchaseprice of the product.
 About the item which does not have a publication in the specifications and manual of this product, it
considers as the outside for a guarantee.

RESTRICTION FOR USE
 Should the equipment be used in the following conditions or environments, give consideration to safetymeasures and inform us of such usage:
1. Use of the equipment in the conditions or environment contrary to those specified, or use outdoors.
2. Use of the equipment in applications expected to cause potential hazard to people or property, which
require special safety measures to be adopted.
 This product can be used under diverse operating conditions. Determination of applicability ofequipment or devices concerned shall be determined after analysis or testing as necessary by the designerof such equipment or devices, or personnel related to the specifications. Such designer or personnel shallassure the performance and safety of the equipment or devices.
 This product is not designed or manufactured to be used for control of equipment directly concernedwith human life (*1) or equipment relating to maintenance of public services/functions involving factorsof safety (*2). Therefore, the product shall not be used for such applications.
(*1): Equipment directly concerned with human life refers to.
- Medical equipment such as life-support systems, equipment for operating theaters.
- Exhaust control equipment for exhaust gases such as toxic fumes or smoke.
- Equipment mandatory to be installed by various laws and regulations such as the Fire Act or
Building Standard Law- Equipment related to the above.
(*2): Equipment relating to maintenance of public services/functions involving factors of safety refersto.
- Traffic control systems for air transportations, railways, roads, or marine transportation- Equipment for nuclear power generation- Equipment related to the above

CAUTIONS ON USE
 Carefully handle the units
Do not drop, or give a strong shock or vibration to the camera. This may cause problems. Treat thecamera cables carefully to prevent cable problems, such as cable breakdown and loosened connections.
 Operating ambient temperature and humidity
Do not use the camera in places where temperature and humidity exceed the specifications. Picture
quality will lower and internal parts may be damaged.
Be particularly careful when using in places exposed to direct sunlight. When shooting in hot places,
depending on the conditions of the object and the camera (for example when the gain is increased), noisein the form of vertical strips or white dots may occur. This is not a malfunction.
 Restriction for the lens combination
This camera might form a ghost to image area depending on the combination of a lens and an
illumination with this camera. But this is not a failure of this camera. Therefore, please check the
combination of the lens and the illumination with this camera when use.
When mounting a lens, take extra caution so that the lens is not tilted, nor does flaw exist at thelens-mount-screw part. Also check to confirm that no dirt nor other foreign object is put inside
Improper mounting might cause the parts to become locked.
 Do not shoot under intense light
Avoid intense light such as spot light on part of the screen because it may cause blooming or smears. If
intense light falls on the screen, vertical stripes may appear on the screen, but this is not a malfunction.
 Do not expose the camera’s image-pickup-plane to sunlight or other intense light directly.
Its inner CCD (charge-coupled device) might be damaged.
 MoireWhen thin stripe patterns are shot, stripe patterns that are not actually there (moire) may appears asinterference stripes. This is not a malfunction.
 Undesirable noise
If the camera or the cables are located near something which emit strong magnetism or near somethingwhich emit strong electric wave, undesirable noise may appear on the screen. In such a case, try tochange the location of the camera or the cable wiring.
 Handling of the protection cap
When the camera is not in use, put a lens-cap onto the camera head for protection of theimage-pickup-plane.
 When not using the camera for a longtime
Stop supplying power for safety.
 When cleaning the camera
Always turn off the power and clean with a piece of soft dry cloth.
To remove stubborn stains, use a soft cloth soaked in diluted acid-free detergent. Do not use alcohol,
benzine, thinner, etc. If used, coating and printed letters may be discolored.
In case the image-pickup-plane should be settled with fine dust, dirt, or scratched, ask your dealer fortechnical advice.
 Wastes of this product should be separated and discarded in compliance with the various national andlocal ordinances.

1. PRODUCT DESCRIPTION
CleverDragon series CSCS20BC2 is an integrated type B/W CCD camera with a SXGA format
all-pixel-data readout CCD. The model is suited for high-resolution image processing use. Its compact,
light-weight body is ideal for system integration.
2. FEATURES(1) All-pixel reading
The all-pixel reading system allows the CSCS20BC2 to read all pixels in just 1/19.5 second.
CSCS20BC2 is equipped with a full-frame shutter that allows all-pixel reading even during shutter
operations.
(2) Full-frame shutter
Since all pixels are output even by a random trigger shutter operation, high resolution can be achieved,
without deteriorating the vertical resolution.
(3) Tetragonal lattice layout
The tetragonal lattice layout of CCD pixels facilitates computation for image processing.
(4) Camera Link interface (power supply type)
By using a Camera Link-capable frame grabber board to which power can be supplied, high-speedtransfer of captured images to a PC as well as various types of camera control from the PC are allowed.
Power can also be supplied to the camera with only one cable.
(5) Random trigger shutter function
CSCS20BC2 is equipped with a random trigger shutter, which starts exposure synchronized withexternal trigger signals. Fast-moving objects can thus be captured in place, which ensures accurate
image processing.
(6) Restart-Reset
Images can be shot and fetched at arbitrary timing based on external VD signal input.
(7) Partial scan
Further speed-up is possible because ranges except the range of the image output that the user set are
not read.
(8) Ultra-compact and lightweight main unit
The space-saving ultra-compact and lightweight camera has excellent resistance against vibration andimpact.
(9) RoHS compliant
CleverDragon series are complied with EU RoHS.

7-1. Explanation of Each Function(1) Readout mode
Address: 0x90, Bit: 0, Value:0 to 1
Video is output from the camera link connector. The output video can be grabbed by the frame grabberboard. The frame rate and resolution of output images that this model supports are as follows: All pixel read out Partial Scan

19.5 fps (maximum frame rate) / 1360(H)1024(V)
19.5 fps to 34.8fps (depend on Video output width)
Horizontal resolution: 1360 (fixed)
Partial Scan Video Start Line: 0 to 512
Partial Scan Video Width: 512 to 1024(1-1) All pixel read out
Address: 0x90, Bit: 0, Value: 0
As all pixels are read out in approx. 1/19.5s, you will get images with the higher V resolution (you haveto change the shutter speed to 1/19.5s or shorter).
Vertical Timing (maximum speed on all pixel readout mode)A = approx. 271.1us
B = approx. 50.9ms (1024H-336CLK)
1frame = A + B + C = approx. 51.3ms*1H = 1790CLK

C = approx. 136.0us

FVAL
LVAL
DVAL

3. CONFIGURATION(1)Camera body ······························ 1(2)Accessory
Manual ································ 1

DATA OUT
A

B
C

Horizontal Timing
Same timing

4. OPTION PARTS(1)Camera-mounting kit ······················ Model name: CPT8560*Contact your dealer / distributor for details of option units.
*Application software is not supplied as a standard item.

FVAL
a
LVAL
DVAL

5. INTERFACE
Video output/controlling/power supply connector: (Camera Link Base Configuration) CAMERA LINK
Outputs video signals and VALID, based on the camera link standard LVDS.
This connector is connected to the frame grabber board, image processing device and others.
And it is possible to supply the power to the camera, by using an exclusive Camera Link cable and frame
grabber board complied with Power over Camera Link standard.
Connector model: HDR-EC26FDTG2+ (Manufactured by Honda Connectors)

Pin #
1
2
3
4
5
6
7
8
9
10
11
12
13

I/O
O
O
O
O
O
I
O
I
I
I
I
-

Signal name
+12V
TxOUT0TxOUT1TxOUT2TxCLK OUTTxOUT3SerTC(RxD)+
SerTFG(TxD)CC1(TRIG/VD)CC2+
CC3CC4+
GND

Pin #
14
15
16
17
18
19
20
21
22
23
24
25
26

I/O
O
O
O
O
O
I
O
I
I
I
I
-

Signal nameGND
Tx OUT0+
Tx OUT1+
Tx OUT2+
Tx CLK OUT+Tx OUT3+
SerTC(RxD)SerTFG(TxD)+
CC1(TRIG/VD)+
CC2CC3+
CC4+12V*Please confirm the power supply of the camera cuts when the connector is connected or pulls out. It causes
the breakdown etc.

DATA OUT
b

d
e

c

f

a = 336CLK

b = 22CLK

c = 16CLK

d = 1360CLK

e = 16CLK

f = 40CLK(1-2) Partial Scan
Address: 0x90, Bit: 0, Value: 1
Ranges except the range of the image output that the user set are not read.
Partial scan setting procedure is following: Video start line
Address: 0xC4, Bit: 0 to 9, Value: 0 to 512
 Video output width
Address: 0xC8, Bit: 0 to 10, Value: 512 to 1024
 Partial scan update
Address: 0xC0, Bit: 0, Value: 1
 If you want maximum frame rate, you must shorten the shutter speed.
ex. 1) Video Start Line =128, Video Width = 768(maximum speed on all pixel readout mode)

FVAL
LVAL
DVAL
DATA OUT
A

B

C

D

A = approx. 996.4us

B = approx. 38.2ms ( 768H-336CLK )ex. 2) Video Start Line=256,Video Width=512

C = approx. 861.3us

D = approx. 40.0ms(maximum speed on all pixel readout mode)

FVAL
LVAL

6. CONNECTION EXAMPLESDVALDATA OUT

CCD camera
CSCS20BC2

A

Power-over Camera Link Cable

AC100V
50/60HzB = approx. 25.5ms (512H-336CLK) C = approx. 1586.7us

C

D = approx. 28.8ms(2) Setup Addition Value
Address: 0x70, Bit: 0 to 7, Value: 0 to 255
You can add the offset level to the reference black level.
Setup Addition
Setting Range
Calculation formula(calculated value)(a)+0 to +255 [digit]
0x00 to 0xFF
+ a [digit](10bit)(0 to 255)
(10bit)+0 to +63 [digit]
0x00 to 0xFF
+ a/4 [digit](8bit)(0 to 255)
(8bit)

DC+12V
from ATX power suplyetc.

PC etc.

B
D

A = approx. 1721.8us

This device complies with Part 15 of the FCC rules. Operation is subject to the following two conditions:(1) This device may not cause harmful interference, and (2) This device accept any interference received,
including interference that may cause undesired operation.
This equipment has been tested and found to comply with the limits for a class A digital device, pursuant toPart 15 of the FCC Rules. These limits are designed to provide reasonable protection against harmfulinterference when the equipment is operated in a commercial environment. This equipment generates, uses,
can radiate radio frequency energy and, if not installed and used in accordance with the instruction manual,
may cause harmful interference to radio communications. Operation of this equipment in a residential area
is likely to cause harmful interference in which case the user will be required to correct the interference at
his own expense.

7. FUNCTIONS
By accessing the camera register published on the camera link I/F, you can control/set each function.
Since access to the camera register is performed via the frame grabber board, the controlling and setting
methods differ depending on the frame grabber board you use. For details, refer to the instruction manual of
the relevant frame grabber board or contact our sales representative.
This instruction manual describes the specifications in the case where the camera register is directly
connected by serial transmission over the camera link interface.
For details of the control and setting of functions, refer to “Interface Specification”. Please ask your
distributor or a sales representative about “Interface Specification”.

(3) Gain
Address：0x76, Bit：0 to 6, Value:0 to 90
You can set Gain (video gain).
Gain
Setting Range
Calculation formula(calculated value)(b)0x00 to 0x5A
0 to approx. +12dB
0.132dB  b [dB]
(0 to 90)

Frame Grabber Board
compiled with PoCL

Notes on gain setting:
It is possible to set a maximum of +12 dB (Calculation value) but the warranty range
for this product is 0 to +10 dB. When using this product, be sure to set a gain value
within the warranty range.
And, setting a too high gain value can increase noise. When you adjust the brightnessof the shot image, you are responsible for finally confirming the image quality by using
the entire machine/equipment.

D3004086D

(4) Electronic shutter
The exposure time has the format (numerator/denominator).
The numerator and denominator can be set separately in respective registers.
1 frame length depends on the shutter speed.
Numerator
Denominator

Address: 0xA4, Bit: 0 to 7, Value: 1 to 255
Address: 0xA0, Bit: 0 to 14, Value: 1 to 20000*When you set the exposure time longer than approximately 1 second, white spots andthe unevenness in highlight portion might occasionally be observed on screen. Thisphenomenon is due to the characteristics of the CCD image-pickup device, and do notreflect performance error in the pickup device or CCD Camera itself.

(1) Writing to the register
To write data in a register, send a command, as follows. (Address' max-length is 2 bytes, and Data's
max-length is 8 bytes)
Address
2nd byte

Address
1st byte‘,’(0x2C)

Data
8th byte

Data
7th byte(5-1) Fixed mode
Address: 0x92, Bit: 0, Value: 0
The exposure time is determined by the setting value for the shutter speed.
exposure time

TRIG
CCD
exposureData
5th byte

Data
4th byte

Data
3rd byte

Data
2nd byte

Data
1st byte[CR](0x0D)For example, to write data 0x38 to address 0x76, send a command, as follows:‘7’(0x37)‘6’(0x36)‘,’(0x2C)‘3’(0x33)‘8’(0x38)[CR](0x0D)

The camera responds to the write command with No Error (ACK) or Error (NAK), as follows:
No Error (ACK):[ACK](0x06)[CR](0x0D)

Error (NAK):(5) Random trigger shutter
Address: 0x91, Bit: 0, Value:1
In the random trigger shutter mode, you can shoot and grab an image at an arbitrary timing by trigger
signal input from the external.
 External trigger signals can be input from the camera link I/F CC1.
 If polarity is set to negative polarity, exposure starts at the falling edge of the trigger.
Address: 0x93, Bit: 0, Value:0 is Negative Polarity, Value:1 is Positive Polarity
 The random trigger shutter of this camera can be operated in two types of mode: fixed mode andpulse width mode. How to determine the exposure time differs depending on the mode.

Data
6th byte[NAK](0x15)[CR](0x0D)(2) Reading the register
To read data from a register, send ', (comma)', 'R', 'Q' and [CR] code following the address. For example,
to read data in address 0x91, send a command, as follows:‘9’(0x39)‘1’(0x31)‘,’(0x2C)‘R’(0x52)‘Q’(0x51)

Data
7th byte

Data
6th byte

Data
5th byte

Data
4th byte

Data
3rd byte[Optical axis accuracy][CR](0x0D)

Optical axis accuracy

The camera responds to the read request, as follows (Data's max-length is 8 bytes):Data
8th byte[Mechanical spec](1) Lens mountC-mount*Depending on the lens you use, the performance of the camera may not be brought out fully due to thedeterioration in resolution and brightness in the peripheral area, occurrence of the ghost, aberration andothers. When you check the combination between the lens and camera, be sure to use the lens you
actually use.
*Install a next lens; the C mount lens, its dimension of protrusion from flange is equal to or less than 7.9mm. If a lens does not stand to this condition, it might not be installed to this camera.
(2) Flange back17.526 mm(3) Dimensions
29mm(W)  29mm(H)  26.5mm(D)* Not including protrusion(4) Mass
Approx. 40g(5) Camera body grounding:
Conductive between circuit GND and camera bodyData
2nd byte

Data
1st byte

Position accuracy ofeffective pixels
（Ｘ）
（Ｙ）
±25um
±25um

Flange focal distances(Flange back)( for 17.526mm )±50um

Rotation accuracy ofeffective pixels(θ)±0.12°

9. BEFORE DETERMINING IT AS BEING A FAULT
If any trouble occurs in use, check the following first.
If the trouble persists, contact your distributor or our sales representatives.
Phenomena

Check item

Cannot turn on power Check the connecttion of the CameraLink(PoCL) frame grabber board and CameraLink
cable. the connecttion of the CameraLink(PoCL) frame grabber board and CameraLink
Shooting image is not Check
displayed

cable.
Check that the camera register are correct.
Check that lens aperture is not closed.
Check that the CameraLink(PoCL) grabber board is installed and set up correctly.

Frame drop pccurs on If more than one boards are installed in the PCI slots, remove the other boards.
shooting image
Shooting image
remains still

Check that the camera is not in the random trigger mode.
Check the setting of the CameraLink(PoCL) grabber board.

Check the connecttion of the CameraLink(PoCL) frame grabber board and CameraLink
Cannot control camera cable.
Check the connection of the CameraLink cable.
from PC
Check that the CameraLink(PoCL) grabber board is installed and set up correctly.

10. EXTERNAL VIEW DRAWING[CR](0x0D)

Actually, the camera responds to the read request as minimum data length: For example, to read data
0x10 to address 0x91, the camera responds as follows:‘1’(0x31)*exposure time = set time of E-shutter speed‘0’(0x30)[CR](0x0D)

7-2. Error Status
If NAK is returned to the sent command, you can obtain detailed information on the error by accessing
the status register (Address:0x69) and the expansion status register (Address:0x6A).

approx.
3.5usec

approx. 260.6usec (all pixel readout)

Details of error

Status[0x69]

Expansion
Status[0x6A]

0x00

0x00

0x01

0x01
0x02
0x03

LVAL
No Error

DVAL
DATA OUT
Communication Error(5-2) Pulse width mode
Address: 0x92, Bit: 0, Value: 1
The exposure time is determined by the pulse width.
exposure time

General Accessing Error*exposure time = pulse width + approx. 1.8usec

TRIG

FlashRom Accessing Error

approx. 3.5usec

approx. 5.3usec

User Setting Read Error

CCD
exposureapprox. 260.6usec (all pixel readout)

Partial Mode Related Error

FVAL
LVAL
Shutter Related Error

DVAL
DATA OUT(6) Restart Reset
Address: 0x91, Bit: 0 to 1, Value:2
The restart / reset function is available with the ext. VD signal. You can get an arbitrary slower shutter
speed than normal shutter and random trigger shutter.
 External VD signals can be input from the camera link I/F CC1.
 If polarity is set to negative polarity, exposure starts at the falling edge of the trigger.
Address: 0x93, Bit: 0, Value:0 is Negative Polarity, Value:1 is Positive Polarity
 The shutter speed (exposure time) is determined by ext. VD signal interval.

No errorA packet format error was detected.
An overrun error was detected.
An framing error was detected.
The data length of the command is beyond thespecified upper limit.
The reserved area was accessed.
You attemped a read access to a Write Only Register.
You attemped a write access to a Read Only Register.
The setting value is out of the specified range.
FlashRom access error
You load your saved setting when any saved setting
data is not available.
You changed registers related partial scan mode when
you select all pixel read-out mode.
You attempted to set a shutter speed beyond thespecified range.
You changed Random Trigger shutter mode when
you select Normal Shutter mode or Restart-Reset
You changed trigger polarity when you select Normal
Shutter mode.

8. SPECIFICATIONS[Basic specification](1) Image sensor
Total pixels
Active pixelVideo output pixels
Scanning area
Unit cell size(2) Scan method(3) Synchronization method(4) Aspect ratio(5) Video Output
Data
Readout mode
All pixel readout (default):
Partial Scan (ex.1)(ex.2)

Vertical Timing
VD
CCD
exposure

DATA OUTFVAL

Horizontal Timing
Ext. VD input
more than 1H

CCD
exposure
DATA OUT
FVAL
when the external VD inputs to the camera while video output(7) Output bit
Address: 0x87, Bit: 0 to 3, Value: 8 or 10
You can set gray scale per pixel. Value:8 is 8bit , Value:10 is 10bit

7-2. Command Communication Protocol
The command communication protocol is the teli standard method (method in which parameters are set in
the registers in the camera).
In command send/receive operation, hexadecimal address and data are converted to ASCII data.
All ASCII alphabetic characters used are uppercase characters.

0x04

0x03

0x01
0x02
0x03
0x04
0x01

0x04

0x01

0x05

0x01

0x02

0x01
0x06

0x02
0x03

all-pixel-data-readout interline transfer CCD
1434(H)  1050(V)1392(H)  1040(V)
1360(H)  1200(V)6.47mm(H)  4.84mm(V)(= Equivalent to 1/2” type CCD size)
4.65m(H)  4.65m(V)Non-interlace
Internal synchronization
4:3
Compliant with CameraLink standard version 1.2
10 / 8 bit switching (factory default: 8bit）

1360(H)  1024(V) [19.5fps]
1360(H)  768(V) [25.0fps] (MAX)1360(H)  512(V) [34.8fps] (MAX)minimum video output lines: 512, minimum step: 1 line
(6) Sensitivity
500 lx, F5.6
(7) Minimum subject illuminance 5lx F1.4 (GAIN:10dB, video level 50 %)(8) Gain0 to Approx. +12 dB [factory default:0dB](9) Setup Level(factory default)
10 ± 5 [digit] (8bit)(addition value)+0 to +63 [digit] (8bit)(10) Gamma correction
OFF ( = 1.0 fixed)(11) Power supply voltage
DC12V ± 10% (ripple 50mVP-P or less)(12) Power consumption
Approx. 1.8W
[Electrical shutter specification](1) Shutter Speed
2 to 1/20,000 [sec](2) Random Trigger Shutter
ON / OFF switching (factory default: OFF)
Fixed mode
The exposure time depends on the shutter speed setting
Pulse width mode
The exposure time depends on the pulse width.
(3) Restart-Reset
ON / OFF switching (factory default: OFF)The exposure time depends on the period of Ext. VD.
[Internal sync signal specification](1) Driving frequency
36.000 MHz (1 CLK) ±100ppm(2) Horizontal sync frequency
20.11 kHz (1H = 1790CLK)(3) Vertical sync frequency
19.49Hz (maximum frequency on all pixel readout mode)[Input signal specification](1) TRIG/VD
Camera Link interface input: CC1
Polarity
Positive/Negative switching (factory default: negative)
Pulse width
8s to 2s
Notes of trigger mode:*When the interval of the input trigger signal is extremely shot, or when the trigger signal is noisy, there isa possibility of causing the malfunction. In this case, please input a proper trigger signal.

[Operating ambient conditions](1) Performance assurance
Temperature:0C to +40C
Humidity:
10% to 90% (no condensation)(2) Operation guaranteed
Temperature:-5C to +45C
Humidity:
90% or less (no condensation)(3) Storage
Temperature:-20C to +60C
Humidity:
95% or less (no condensation)(4) EMC conditions (Electro-Magnetic Compatibility)
EMI (Electro-Magnetic Interference): EN61000-6-4 (Conformity)
EMS (Electro-Magnetic Susceptibility): EN61000-6-2 (Conformity)(5) FCCFCC Part 15 Subpart B class A (Conformity)*About the conformity of EMC standard of this machine, it has guaranteed in the conditions combinedwith our system condition. When used combined parts other than specification of our company, I ask you
to have final EMC conformity checked of a visitor with a machine and the whole equipment.

[ UNIT : mm ]Following information is only for EU-member states:
The use of the symbol indicates that this product may not be treated as household
waste. By ensuring this product is disposed of correctly, you will help prevent
potential negative consequences for the environment and human health, which
could otherwise be caused by inappropriate waste handling of this product.
For more detailed information about the take-back and recycling of this product, pleasecontact your supplier where you purchased the product.

”This symbol is applicable for EU member states only”[Communication specification](1) Communication speed(2) Start bit(3) Data bit(4) Parity(5) Stop bit(6) Handshake

9600 bps (fixed)
1
8
None
1
None

[Camera PSU input impedance]
Ci ≤ 57F, Rx = 10k ± 10%
PSU
TestCircuit

PoCL
Cable

Ia

CSCS20BC2
PSU

Pin#: 1, 26

Vb

Rx

Cx

CameraPSU

Pin#: 13, 14
Ia = 52uA, Vb = 0.52V, Cx < 50uF

[Typical Spectral Response]*The lens characteristics and light source characteristics are not reflected in table.

1.0
0.9
0.8
Relative Response

Error type

FVAL

0.7
0.6
0.5
0.4
0.3
0.2
0.1
0.0
400

500

600

700
800
Wave Length[nm]

900

1000

D3004086D