EA00040D/06/A2/14.19
71430332
2019-04-01

Products

Solutions

Installation Instructions
Replacing transmitter parts
Promag 10, Prosonic Flow 91 W

EN Installation Instructions: EA00040D →  3
DE Einbauanleitung: EA00040D →  13

Services

EA00040D

Replacing the transmitter parts
Promag 10, Prosonic Flow 91 W
Table of contents
1

Overview of spare part sets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

3Personnel authorized to carry out repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

5Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

6Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

7

Replacing the O-ring, cover, display, board carrier, main board, fuse and terminal . . . . . 9

8

Replacing the housing seal, connection board, cable gland and wall bracket . . . . . . . . . 10

Endress+Hauser

3EA00040D

1

Overview of spare part sets

The Installation Instructions apply to the following spare part sets:
Order number

Original spare part set

Content

50104745

Display kit 10/91W, with keyboard

1 × display module, mechanical keyboard,
complete

50104748

Board/display carrier kit

1 × board carrier, electronics module
1 × cover for board carrier, electronics module
1 × wire, yellow/green
1 × screw M4 × 16/6
1 × spring washer M450104749

Set of 10 fuses, slow-blow, 1.0 A/250

10-fuse board TR5 1 A, slow-blow

50104750

Set of 10 fuses, slow-blow, 1.6 A/250

10-fuse board ,1.6 A, slow-blow

50104751

Main board kit, 85 - 250VAC

1 × main board, L44 Promag 10, 85 - 250 VAC
1 × 4-pin plug-in terminal
1 × 2-pin plug-in terminal

50104752

Main board kit, 20 - 28VAC, 11 - 40VDC

1 × main board L49 Promag10, 24 VAC/VDC
1 × 4-pin plug-in terminal
1 × 2-pin plug-in terminal

50104753

Housing seal kit, field housing

2 × O-ring 113.90 × 3.53 mm /4.5 × 0.14 in
1 × O-ring 47.00 × 5.34 mm /1.9 × 0.2 in

50104754

Connection board kit, FS, electronics module

1 × connection board, L50, FS, transmitter, EMF,
complete

50106219

Set of 10 fuses, T 2A00/125

10 × SMD fuses, slow-blow, 2A 125V

50108138Set, terminal

1 × 4-pin plug-in terminal
1 × 2-pin plug-in terminal

71022832

Cable gland kit, wall bracket

1 × cable gland M20 × 1.5

71022835

Wall bracket kit, complete

1 × wall bracket for housing, UDM
2 × screw M6 × 20
1 × O-ring 52.39 × 3.53 mm / 2.06 × 0.14 in

71023273

Housing seal kit, field housing

1 × O-ring 113.90 × 3.53 mm / 4.5 × 0.14 in
1 × O-ring 52.39 × 3.53 mm / 2.06 × 0.14 in

71098237

Main board kit, 85-250VAC HE

1 × main board, L44 Promag 10, 85 - 250V HEC
1 × 4-pin plug-in terminal
1 × 2-pin plug-in terminal

71098239

Main board kit, 20-28VAC 11-40VDC HE

1 × main board, L49 Promag 10, 24VAC/VDC
HEC
1 × 4-pin plug-in terminal
1 × 2-pin plug-in terminal

71098240

Connection board kit, FS, electronics module,HE

1 × connection board, L50, FS, transmitter EMF
HEC

4Endress+Hauser

EA00040D

Order number

Original spare part set

Content

71138304

Main board kit, 85-250VA

1 × main board, L57, 85-250AC
1 × 4-pin plug-in terminal
1 × 2-pin plug-in terminal

71138309

Main board kit, 20-28VAC 11-40VDC

1 × main board, L57, 24VAC/DC
1 × 4-pin plug-in terminal
1 × 2-pin plug-in terminal• The order number of the spare parts set (on the product label on the package) candiffer from the production number (on the label directly on the spare part)!
• You can find the order number of the relevant spare parts set by entering the
production number of the spare part in the spare parts search tool.
• We recommend that you keep the Installation Instructions and packaging together at
all times.

2

Designated use

The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type. Only original parts from Endress+Hauser may be used. Only
spare parts kits designed by Endress+Hauser for the measuring device can be used at any
time.
Inspection is performed using the W@M Device Viewer. The procedure involved is describedas follows.
A spare parts overview is located in the interior of some measuring devices. If the spareparts kit is specified on this list, no inspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for productdata and click Spare Parts.

3.

All spare parts for the measuring device aredisplayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order Thenumber of the spare parts kit corresponds to anorder number on the spare parts list.

Endress+Hauser

5EA00040D

3

Personnel authorized to carry out repairs

Authorization to carry out repairs depends on the measuring device's approval type. The tablebelow shows the authorized group of people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out
safely and to the required quality standard. He/she must also guarantee the safety of thedevice following repair.

1)

4

Measuring device approval

Spare parts kit

Personnel authorized toperform repairs 1)

Without approvalWith approval (e.g. IECEx)

50104745
50104749
50104750
50104751
50104752
50104754
50108138
71022832
71098237
71098239
71098240
71138304
71138309

1, 2, 3

Without approvalWith approval (e.g. IECEx)

50104748
50104753
71022835
71023273

2, 3

For custody transfer

All

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Safety instructions• Check whether the spare part matches the labeling on the measuring device as described on
the cover page.
• The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type.
Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.

6

Endress+Hauser

EA00040D• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:– Specialized technical staff must be trained in instrument safety.
– They must be familiar with the individual operating conditions of the devices.
– In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is de-energized.
• When using Ex-certified measuring devices: Open device only when in a de-energized state(allow 10 minutes to elapse after switching off the power supply) or in environments which
do not have a potentially explosive atmosphere.
• In the case of measuring devices in safety-related applications in accordance withIEC 61508 or IEC 61511: Commission in accordance with Operating Instructions after
repair. Document the repair procedure.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Hot surfaces! Risk of injury! Before commencing work: allow the system and measuring
device to cool down to a touchable temperature.
• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removing the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn the measuring device off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open the housing for a brief period. Avoid the penetration of foreign bodies, moisture
or contaminants.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.
• If spacing is reduced or the dielectric strength of the measuring device cannot be
guaranteed during repair work, perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service plug:– Do not connect in explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your Endress+Hauser service organization.

Endress+Hauser

7EA00040D

5

Symbols used

5.1

Safety symbols
SymbolMeaning

DANGERDANGER!
This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.

WARNINGWARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.

CAUTIONCAUTION!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.

NOTICENOTE!
This symbol contains information on procedures and other facts which do not result inpersonal injury.

5.2

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

6

Series of steps

Tools list
Acid-free, non3 mm

8T20

0.5 × 3.5 mm

8 mm, 24 mm

hardening lubricant

Endress+Hauser

EA00040D

7

Replacing the O-ring, cover, display, board carrier,main board, fuse and terminal

LWARNING
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is de-energized.
1→

2→

3→

2.
1.

1.

4→

5→

6→

3.
2.

1.

T20

7

Endress+Hauser

9EA00040D

7.1

Reassembling the transmitter housing

Reassembly is carried out in reverse order. Note the following:1→

2‣ Acid-free, non-hardening
lubricant8

Replacing the housing seal, connection board, cablegland and wall bracket

LWARNING
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is de-energized.
1→

2→

3→

2.
1.

1.

10

Endress+Hauser

EA00040D

4→

5→

6→

1.
2.
12 Nm

1.

8 mm

3 (0.12)

3.
4.

3.

4.

2.

‣ Special screw! Do not loosen thescrew completely or replace itwith another one. Only use
original parts from Endress+Hauser.

7→

8→

9

2.
24 mm

1.
T20

2.

2.
1.

8.1

1.

Reassembling the transmitter housing

Reassembly is carried out in reverse order. Note the following:1→

2‣ Acid-free, non-hardening
lubricantEndress+Hauser

11

EA00040D

Austausch Messumformerteile
Promag 10, Prosonic Flow 91 W
Inhaltsverzeichnis
1Übersicht Ersatzteilsets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

2

Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

3

Reparaturberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

5

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

6

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

7Austausch O-Ring Deckel, Anzeige, Printträger, Hauptprint, Sicherung undAnschlussklemme . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 19

8Austausch Gehäusedichtung, Anschlussprint, Kabelverschraubung und
Wandhalterung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 20

Endress+Hauser

13

EA00040D

1Übersicht Ersatzteilsets

Die Einbauanleitung ist für folgende Ersatzteilsets gültig:
Bestellnummer

Original Ersatzteilset

Inhalt50104745

Set Anzeige 10/91W mit Tastatur

1 × Anzeigemodul mechanische Tastatur
komplett

50104748

Set Print/Displayträger

1 × Printträger Elektronik
1 × Abdeckung zu Printträger Elektronik
1 × Litze gelb/grün
1 × Schraube M4 × 16/6
1 × Federring M450104749

Set 10 Sicherungen träge 1,0 A/250

10 × Sicherung-Print TR5 1 A träge

50104750

Set 10 Sicherungen träge 1,6 A/250

10 × Sicherung-Print 1,6 A träge

50104751

Set Hauptprint, 85 - 250VAC

1 × Hauptprint L44 Promag 10, 85 - 250 VAC
1 × Klemme steckbar 4 Pol
1 × Klemme steckbar 2 Pol

50104752

Set Hauptprint, 20 - 28VAC, 11 - 40VDC

1 × Hauptprint L49 Promag10, 24 VAC/VDC
1 × Klemme steckbar 4 Pol
1 × Klemme steckbar 2 Pol

50104753

Set Gehäusedichtung, Feldgehäuse

2 × O-Ring 113,90 × 3,53 mm /4,5 × 0,14 in
1 × O-Ring 47,00 × 5,34 mm /1,9 × 0,2 in

50104754

Set Anschlussprint, FS Elektronik

1 × Anschlussprint L50 FS Umformer, MID,
komplett

50106219

Set Set 10 Sicherungen T 2A00/125

10 × SMD-Sicherungen, träge, 2A 125V

50108138

Set Anschlussklemme

1 × Klemme steckbar 4 Pol
1 × Klemme steckbar 2 Pol

71022832

Set Kabelverschraubung Wandhalter

1 × Kabelverschraubung M20 × 1,571022835

Set Wandhalter, komplett

1 × Wandhalter für Gehäuse UDM
2 × Schraube M6 × 20
1 × O-Ring 52,39 × 3,53 mm / 2,06 × 0,14 in

71023273

Set Gehäusedichtung, Feldgehäuse

1 × O-Ring 113,90 × 3,53 mm / 4,5 × 0,14 in
1 × O-Ring 52,39 × 3,53 mm / 2,06 × 0,14 in

71098237

Set Hauptprint, 85-250VAC HE

1 × Hauptprint L44 Promag 10, 85 - 250V HEC
1 × Klemme steckbar 4 Pol
1 × Klemme steckbar 2 Pol

71098239

Set Hauptprint, 20-28VAC 11-40VDC HE

1 × Hauptprint L49 Promag 10, 24VAC/VDC HEC
1 × Klemme steckbar 4 Pol
1 × Klemme steckbar 2 Pol

71098240

Set Anschlussprint, FS Elektronik HE

1 × Anschlussprint L50 FS Umformer MID HEC

14

Endress+Hauser

EA00040D

Bestellnummer

Original Ersatzteilset

Inhalt71138304

Set Hauptprint, 85-250VA

1 × Hauptprint L57, 85-250AC
1 × Klemme steckbar 4 Pol
1 × Klemme steckbar 2 Pol

71138309

Set Hauptprint, 20-28VAC 11-40VDC

1 × Hauptprint L57, 24VAC/DC
1 × Klemme steckbar 4 Pol
1 × Klemme steckbar 2 Pol• Die Bestellnummer des Ersatzteilsets (auf dem Produktaufkleber der Verpackung)kann sich von der Produktionsnummer (auf dem Aufkleber direkt auf dem Ersatzteil)unterscheiden!
• Durch Eingabe der Produktionsnummer des Ersatzteiles im Ersatzteilfindetool kanndie Bestellnummer des ensprechenden Ersatzteilsets ermittelt werden.
• Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

2

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Endress+Hauser

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

15

EA00040D

3

Reparaturberechtigte Personen

Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die
Sicherheit während der Arbeiten, die Qualität der Ausführung und die Sicherheit desGeräts nach der Reparatur.

1)

4

Zulassung des Messgeräts

Ersatzteilset

Reparaturberechtigter
Personenkreis 1)

ohne Zulassung
mit Zulassung (z.B. IECEx)

50104745
50104749
50104750
50104751
50104752
50104754
50108138
71022832
71098237
71098239
71098240
71138304
71138309

1, 2, 3ohne Zulassung
mit Zulassung (z.B. IECEx)

50104748
50104753
71022835
71023273

2, 3

Bei eichfähigem Verkehr

alle

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
16

Endress+Hauser

EA00040D• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosem Zustand (nach Berücksichtigung
einer Wartezeit von 10 Minuten nach Abschalten der Energiezufuhr) oder in Umgebungenöffnen, die keine explosionsfähige Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

Endress+Hauser

17

EA00040D

5

Verwendete Symbole

5.1

Warnhinweissymbole
SymbolBedeutung

GEFAHRGEFAHR!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen wird.

WARNUNGWARNUNG!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen kann.

VORSICHTVORSICHT!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu leichter oder mittelschwerer Körperverletzung führen kann.

HINWEISHINWEIS!
Dieser Hinweis enthält Informationen zu Vorgehensweisen und weiterführenden
Sachverhalten, die keine Körperverletzung nach sich ziehen.

5.2

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

6

Handlungsschritte

Werkzeugliste
säurefreies, nicht
3 mm

18

T20

0,5 × 3,5 mm

8 mm, 24 mm

härtendes FettEndress+Hauser

EA00040D

7

Austausch O-Ring Deckel, Anzeige, Printträger,Hauptprint, Sicherung und Anschlussklemme

LWARNUNG
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
1→

2→

3→

2.
1.

1.

4→

5→

6→

3.
2.

1.

T20

7

Endress+Hauser

19

EA00040D

7.1

Zusammenbau Messumformergehäuse

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
1→2‣ säurefreies, nicht härtendes Fett

8

Austausch Gehäusedichtung, Anschlussprint,
Kabelverschraubung und Wandhalterung

LWARNUNG
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
1→

2→

3→

2.
1.

1.

20

Endress+Hauser

EA00040D

4→

5→

6→

1.
2.
12 Nm

1.

8 mm

3 (0.12)

3.
4.

3.

4.

2.

‣ Spezialschraube! Schraube nicht

vollständig lösen oder durchandere Schraube ersetzen. Nur
Originalteile von Endress+Hauser
verwenden.

7→

8→

9

2.
24 mm

1.
T20

2.

2.
1.

8.1

1.

Zusammenbau Messumformergehäuse

Der Zusammenbau erfolgt in umgekehrter Reihenfolge. Folgendes ist zu beachten:
1→2‣ säurefreies, nicht härtendes FettEndress+Hauser21

*71430332*
71430332

www.addresses.endress.com