EA01163D/06/A2/05.23-00
71613668
2024-01-01

Products

Solutions

Installation Instructions
Alteration with Display andoperating module DKX001
Proline 300

Services

EA01163D

Alteration with remote display and operating module
DKX001
Proline 300
Table of contents
1

Overview of alteration kit . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Authorized installation personnel . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

3Checking the configuration of the alteration kit supplied . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

5

Safety symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

6Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

7Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

8Mounting the remote display and operating module DK001 . . . . . . . . . . . . . . . . . . . . . . . . . 8

9Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12

Endress+Hauser

3

EA01163D

1

Overview of alteration kit

The conversion instructions apply to the following alteration kits:
Product structure

Conversion procedure

Alteration kit (contents)

DKX001–**A

Continue to use existing display module

Remote display module, 4-line,
illuminated, touch control
1 × housing with display module
1 × conversion instructions

DKX001–**F

Display module included in the delivery

Remote display module, 4-line,
illuminated, touch control
1 × housing with display module
1 × conversion instructions

2

Authorized installation personnel

Authorization to carry out installation depends on the measuring device's approval type. The
table below shows the authorized group of people in each case.

1)

3

Measuring device approval

Persons authorized to carry out installation 1)

Without approval

1, 2, 3

With approval (e.g. IECEx)

1, 2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/conversion must be performed under supervision.

Checking the configuration of the alteration kit
supplied

The alteration kit and Installation Instructions are used to upgrade certain functions of anEndress+Hauser measuring device. Only original parts from Endress+Hauser may be used.
Only alteration kits designated by Endress+Hauser for the measuring device may be used.

4

Endress+Hauser

EA01163D

The configuration of the supplied alteration kit can be checked via W@M Device Viewer.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code

Description

=

?

1.

www.endress.com/deviceviewer

2.

Enter the serial number of the alteration kit,search for the product data and click "Order code".

3.

The order structure of the alteration kit isdisplayed.

4.

Check the order number of the alteration kit.

• Check the serial number against the serial number ofthe original device.
• Does the configuration of the alteration kit match theordered configuration?
= YES, the alteration kit may be used for the measuring
device.
= NO, the alteration kit may not be used for themeasuring device.

4

Safety instructions• Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and alteration.
• The specialized technical staff involved in the mounting, electrical installation,commissioning, maintenance, installation and alteration of the measuring devices mustmeet the following requirements:• Specialized technical staff must be trained in instrument safety.
• Specialized technical staff must be familiar with the individual operating conditions of thedevices.
• In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is de-energized.
• For measuring devices intended for use in hazardous locations, please observe theguidelines in the Ex documentation (XA).
• For measuring devices in safety-related applications in accordance with IEC 61508 or
IEC 61511: following alteration, re-commission the device in accordance with the Operating
Instructions. Document the alteration.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Risk of injury from hot surfaces! Before commencing work: allow the system and measuring
device to cool to a touch-safe temperature.
• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging electronic components! Ensure you have a working environment
protected from electrostatic discharge.
Endress+Hauser

5

EA01163D• After removing the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn the measuring device off before removing internal covers.
• Only open the housing for a short period. Avoid foreign objects, moisture or dirt entering
the housing.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.
• If, during alteration, spacing is reduced or the dielectric strength of the measuring device
cannot be guaranteed, perform a test on completion of the work (e.g. high-voltage test inaccordance with the manufacturer's instructions).
• Service connector:• Do not connect in potentially explosive atmospheres.
• Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your responsible www.addresses.endress.com.

5

Safety symbols
DANGER

This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.
WARNING

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.
CAUTION

This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.
NOTICE

This symbol contains information on procedures and other facts which do not result inpersonal injury.

6

Endress+Hauser

EA01163D

6

Symbols

6.1

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

7

Series of steps

Tools list

Allen key
3 mmEndress+Hauser

Slotted headscrewdriver
0.5 x 3.5 mm

Hexagon wrench
20 mm, 22 mm

7

EA01163D

8

Mounting the remote display and operating module
DK001

8.1

Wall mounting
78 (3.07)

136 (5.35)
114 (4.49)!103 (4.06)
7.0 (0.27)

7
0.4

)

(
12

86 (3.39)

!

1

8.2

Unit mm (in)Pipe mounting with mounting bracket
1

2

8

2

3

1 = mounting bracket, 2 = Allen screws, 3 = hose clip

Endress+Hauser

EA01163D

A

3

B

A = cable entry parallel to pipe, B = cable entry perpendicular to pipe

8.3

Opening the transmitter housing with a display module andremoving existing display module

LCAUTION
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is deenergized.
1→

2→

1.
2.

Nicht

n
fne
öf

un
te r

n
f ne
öf

Nicht

un
t er

3.

ES
C

3 mm

E

1.

e

ar

1.

+
ar

e

–

2.

3.

3

Endress+Hauser

9EA01163D

8.4

Installing existing display module

1→

2→

3

1.

2.

3.

1.

2.

C

S

E
C

S

E

2.
3 mm

1.

8.5

3.

3 mm

Opening the transmitter housing without a display module

1

2.
un
te r

n
f ne
öf

Nich
Sp t unPowean
Do nu ter r
en not ng öff
op ne
Neergizeenso pas d wh n
us ouenten vri
sio r
n

Nicht

I/O

e

3.
ar

3 mm
1.

10

2.5 Nm (1.8 lbf ft)

Endress+Hauser

EA01163D

8.6

Connecting the remote display and operating module DKX001

LWARNING
Explosion Hazard!
‣ Do not connect 2 displays simultaneousely in hazardous location!
LCAUTION
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is deenergized.
Only one display or operation unit may be connected to the transmitter at any one time.
Existing integrated display module: disconnect electrical connection.
1→

2→

3→

1.

3.

Nicht

un
t er

n
fne
öf

3 (0.12)

e

2.

ar

er
Powr ente fn nun g öf whe
enichtnnunNpat op
d vrir
S nozeDoergis ouionene patens
N us
so
I/O

4→

5→

e

Nicht

un
t er

n
fne
öf

ar

er

Powr ente fn nun g öf whe
enichtnnunNpat op
d vrir
S nozeDoergis ouionen patens
Neus
so
I/O

Endress+Hauser

11

EA01163D

6

2.

e

Nicht

un
t er

n
fne
öf

ar

1.

3 mm

3.

For detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

9

Disposal

If required by the Directive 2012/19/EU on waste electrical and electronic equipment(WEEE), the product is marked with the depicted symbol in order to minimize the disposalof WEEE as unsorted municipal waste. Do not dispose of products bearing this marking as
unsorted municipal waste. Instead, return them to the manufacturer for disposal under theapplicable conditions.

12

Endress+Hauser

EA01163D

Nachrüsten mit abgesetztem Anzeige-und Bedienmodul
DKX001
Proline 300
Inhaltsverzeichnis
1Übersicht Nachrüstsatz . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

2

Einbauberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14

3Überprüfung der Konfiguration des gelieferten Nachrüstsatzes . . . . . . . . . . . . . . . . . . . . . 14

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

5Warnhinweissymbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 16

6

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17

7

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 17

8

Abgesetztes Anzeige- und Bedienmodul DK001 montieren . . . . . . . . . . . . . . . . . . . . . . . . 18

9Entsorgung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22

Endress+Hauser

13

EA01163D

1Übersicht Nachrüstsatz

Die Umbauanleitung ist für folgende Nachrüstsätze gültig:
Bestellstruktur

Umbauvorgang

Nachrüstsatz (Inhalt)

DKX001–**A

Bestehendes Anzeigemodul weiterverwenden

Getrenntes Anzeigemodul 4-zeilig,
beleuchtet, Touch Control
1 × Gehäuse mit Anzeigemodul
1 × Umbauanleitung

DKX001–**F

Anzeigemodul im Lieferumfang

Getrenntes Anzeigemodul 4-zeilig,
beleuchtet, Touch Control
1 × Gehäuse mit Anzeigemodul
1 × Umbauanleitung2

Einbauberechtigte Personen

Die Berechtigung zur Durchführung eines Einbaus ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Zulassung des Messgeräts

Einbauberechtigter Personenkreis 1)

Ohne Zulassung

1, 2, 3

mit Zulassung (z.B. IECEx)

1, 2, 3Bei eichfähigem Verkehr

4

1)

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

3Überprüfung der Konfiguration des gelieferten
Nachrüstsatzes

Nachrüstsatz und Einbauanleitung dienen dazu bestimmte Funktionen eines Endress+Hauser
Messgerätes zu erweitern. Es dürfen nur Originalteile von Endress+Hauser verwendet werden.
Grundsätzlich dürfen nur Nachrüstsätze verwendet werden, die von Endress+Hauser für das
Messgerät vorgesehen sind.

14

Endress+Hauser

EA01163D

Die Überprüfung der Konfiguration des gelieferten Nachrüstsatzes kann via W@M Device
Viewer durchgeführt werden.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer des Nachrüstsatzes eingeben,
Produktdaten suchen und auf "Bestellcode" klicken.

3.

Anzeige der Bestellstruktur des Nachrüstsatzes.

4.

Bestellnummer des Nachrüstsatzes prüfen.

• Seriennummer mit Seriennummer des Originalgerätsüberprüfen.
• Stimmt die Konfiguration des Nachrüstsatzes mit derbestellten Konfiguration überein?
= JA, der Nachrüstsatz darf für das Messgerät
verwendet werden.
= NEIN, der Nachrüstsatz darf für das Messgerät nichtverwendet werden.

4

Sicherheitshinweise• Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Nachrüsten einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung, Einbau und Nachrüsten der Messgeräte müssen erfüllt sein:• In Gerätesicherheit ausgebildet.
• Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
• Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Messgeräten für den explosionsgefährdeten Bereich: Hinweise in der ExDokumentation (XA) beachten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach dem Nachrüsten Neuinbetriebnahme gemäß Betriebsanleitung
durchführen. Nachrüsten dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
Endress+Hauser

15

EA01163D• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.
• Wenn bei den Nachrüstarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:• Nicht in explosionsfähiger Atmosphäre anschließen.
• Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige www.addresses.endress.com.

5

Warnhinweissymbole
GEFAHR

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen wird.
WARNUNG

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen kann.
VORSICHT

Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu leichter oder mittelschwerer Körperverletzung führen kann.
HINWEIS

Dieser Hinweis enthält Informationen zu Vorgehensweisen und weiterführenden
Sachverhalten, die keine Körperverletzung nach sich ziehen.

16

Endress+Hauser

EA01163D

6

Verwendete Symbole

6.1

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

7

Handlungsschritte

Werkzeugliste

Innensechskant-

Schlitzschrauben-schlüssel

zieher3 mm

0.5 x 3.5 mm

Endress+Hauser

Sechskantschlüssel
20 mm, 22 mm

17EA01163D

8

Abgesetztes Anzeige- und Bedienmodul DK001
montieren

8.1

Wandmontage
78 (3.07)

136 (5.35)
114 (4.49)!103 (4.06)
7.0 (0.27)

7
0.4

)

(
12

86 (3.39)

!

4

8.2

Maßeinheit mm (in)Rohrmontage mit Montagebügel
1

5

18

2

3

1 = Montagebügel, 2 = Innensechskantschrauben, 3 = Schlauchschelle

Endress+Hauser

EA01163D

A

6

B

A = Kabeleinführung parallel zum Rohr, B = Kabeleinführung senkrecht zum Rohr

8.3Öffnen Messumformergehäuse mit Anzeigemodul und Ausbau
bestehendes Anzeigemodul

LVORSICHT
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
1→

2→

1.
2.

Nicht

n
fne
öf

un
te r

n
f ne
öf

Nicht

un
t er

3.

ES
C

3 mm

E

1.

e

ar

1.

+
ar

e

–

2.

3.

3

Endress+Hauser

19

EA01163D

8.4

Einbau bestehendes Anzeigemodul

1→

2→

3

1.

2.

3.

1.

2.

C

S

E
C

S

E

2.
3 mm

1.

8.5

3.

3 mm

Öffnen Messumformergehäuse ohne Anzeigemodul

1

2.
un
te r

n
f ne
öf

Nich
Sp t unPowean
Do nu ter r
en not ng öff
op ne
Neergizeenso pas d wh n
us ouenten vri
sio r
n

Nicht

I/O

e

3.
ar

3 mm
1.

20

2.5 Nm (1.8 lbf ft)

Endress+Hauser

EA01163D

8.6

Abgesetztes Anzeige- und Bedienmodul DKX001 anschließen

LWARNUNG
Explosionsgefahr!
‣ Keine gleichzeitige Verwendung von 2 Anzeigemodulen in explosionsgefährdetem Bereich!
LVORSICHT
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
Es darf immer nur eine Anzeige oder Bedienung am Messumformer angeschlossen sein.
Vorhandenes integriertes Anzeigemodul: Elektrischen Anschluss trennen.
1→

2→

3→

1.

3.

Nicht

un
t er

n
fne
öf

3 (0.12)

e

2.

ar

er
Powr ente fn nun g öf whe
enichtnnunNpat op
d vrir
S nozeDoergis ouionene patens
N us
so
I/O

4→

5→

e

Nicht

un
t er

n
fne
öf

ar

er

Powr ente fn nun g öf whe
enichtnnunNpat op
d vrir
S nozeDoergis ouionen patens
Neus
so
I/O

Endress+Hauser

21

EA01163D

6

2.

e

Nicht

un
t er

n
fne
öf

ar

1.

3 mm

3.

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

9

Entsorgung

Gemäß der Richtlinie 2012/19/EU über Elektro- und Elektronik-Altgeräte (WEEE) ist das
Produkt mit dem abgebildeten Symbol gekennzeichnet, um die Entsorgung von WEEE als
unsortierten Hausmüll zu minimieren. Gekennzeichnete Produkte nicht als unsortierter
Hausmüll entsorgen, sondern zu den gültigen Bedingungen an den Hersteller zurückgeben.

22

Endress+Hauser

*71613668*
71613668

www.addresses.endress.com