Rev 1.0

High-Resolution CCD Color Camera
CleverDragon Series

CSCQS9CC23
Specification

Contents
RESTRICTION FOR USE ··················································1
CASES FOR INDEMNITY (LIMITED WARRANTY) ······2
NOTES ON USING THIS PRODUCT································3
1. Overview ·········································································5
2. Features ···········································································5
3. Configuration···································································5
4. Optional parts ··································································5
5. Specifications ··································································66. Serial communication ·····················································15
7. Functions ········································································15
8. Outline Drawing ·····························································20
9. Guarantee········································································21
10. Repair ···········································································21

D4168764A

Rev 1.0

RESTRICTION FOR USE! Should the equipment be used in the following conditions or environments, give consideration tosafety measures and inform us of such usage:
1. Use of the equipment in the conditions or environment contrary to those specified, or use outdoors.
2. Use of the equipment in applications expected to cause potential hazard to people or property, which
require special safety measures to be adopted.
! This product can be used under diverse operating conditions. Determination of applicability ofequipment or devices concerned shall be determined after analysis or testing as necessary by thedesigner of such equipment or devices, or personnel related to the specifications. Such designer or
personnel shall assure the performance and safety of the equipment or devices.
! This product is not designed or manufactured to be used for control of equipment directly
concerned with human life (*1) or equipment relating to maintenance of public
services/functions involving factors of safety (*2). Therefore, the product shall not be used forsuch applications.
(*1): Equipment directly concerned with human life refers to.
· Medical equipment such as life-support systems, equipment for operating theaters.
· Exhaust control equipment for exhaust gases such as toxic fumes or smoke.
· Equipment mandatory to be installed by various laws and regulations such as the Fire Act
or Building Standard Law· Equipment related to the above(*2): Equipment relating to maintenance of public services/functions involving factors of safety
refers to.
· Traffic control systems for air transportation, railways, roads, or marine transportation· Equipment for nuclear power generation· Equipment related to the above

1
D4168764A

Rev 1.0

CASES FOR INDEMNITY (LIMITED WARRANTY)
We shall be exempted from taking responsibility and held harmless for damage or losses incurred by the user
in the following cases.
! In the case damage or losses are caused by fire, earthquake, or other acts of God, acts by a third party,
deliberate or accidental misuse by the user, or use under extreme operating conditions.
! In the case of indirect, additional, consequential damages (loss of business interests, suspension ofbusiness activities) are incurred as result of malfunction or non-function of the equipment, we shallbe exempted from responsibility for such damages.
! In the case damage or losses are caused by failure to observe the information contained in the
instructions in this instruction manual and specifications.
! In the case damage or losses are caused by use contrary to the instructions in this instruction manual andspecifications.
! In the case damage or losses are caused by malfunction or other problems resulting from use of
equipment or software that is not specified.
! In the case damage or losses are caused by repair or modification conducted by the customer or
any unauthorized third party (such as an unauthorized service representative).
! Expenses we bear on this product shall be limited to the individual price of the product.
! About the item which does not have a publication in the specifications and manual of this product, it considers
as the outside for a guarantee.

2
D4168764A

Rev 1.0

NOTES ON USING THIS PRODUCT! Handle carefully
Do not drop the equipment or allow it to be subject to strong impact or vibration, as such action may cause
malfunctions. Further, do not damage the connection cable, since this may cause wire breakage.
! Environmental operating conditions
Do not use the product in locations where the ambient temperature or humidity exceeds the specifications.
Otherwise, image quality may be degraded or internal components may be adversely affected. In
particular, do not use the product in areas exposed to direct sunlight. Moreover, during shooting underhigh temperatures, vertical stripes or white spots (noise) may be produced, depending on the subject orcamera conditions (such as increased gain). However, such phenomena are not malfunctions.
! About lens mountAs for the Special-mount lens used combining this camera, the projection distance from bottom of the
Bottom ofscrew should use 9.5mm or less.
Special-mount lens

the screw

9.5mm or less

! Check a combination with the lens
Depending on the lens and lighting you use, an image is reflected as a ghost in the imaging area. However,this is not because of a fault the camera.
In addition, depending on the lens you use, the performance of the camera may not be brought out fully
due to deterioration in resolution and brightness in the peripheral area, aberration and others.
Be sure to check a combination with the camera by using the lens and lighting you actually use.
When installing a lens in the camera, make sure carefully that it is not tilted.
In addition, use a mounting screw free from defects and dirt. Otherwise, the camera may be unable to be
removed.
! About the difference between the right and on the left of an image output screen
As for CCD used with this camera, the left half and the right half of the screen are independently output(2ch output method). Therefore, It is not failure, although a level difference may be visible to an image oneither side or a boundary line may be visible to middle of the screen bordering on middle of the screenwith the setting mode of a camera.
! Defect of CCD
In transportation and keeping, it is not a breakdown though the defect might occur by accident in CCD.
! Do not shoot under intense light.
Avoid intense light such as spot lights on part of the screen because it may cause blooming or smears. If
intense light falls on the screen, vertical stripes may appear on the screen, but this is not a malfunction.
! Occurrence of moiré
If you shoot thin stripe patterns, moiré patterns (interference fringes) may appear. This is not a
malfunction.
! Occurrence of noise on the screen
If an intense magnetic or electromagnetic field is generated near the camera or connection cable, noise
may be generated on the screen. If this occurs, move the camera or the cable.

3
D4168764A

Rev 1.0! Handling of the protective cap
If the camera is not in use, attach the lens cap to the camera to protect the image pickup surface.
! Turn OFF the power in the case of connection
Turn OFF the power in the case of connection of connection camera cable.
Otherwise, an electric shock or a malfunction may occur.
! Disposal
When disposing of the camera, it may be necessary to disassemble it into separate parts, in accordancewith the laws and regulations of your country and/or municipality concerning environmental
contamination.
The use of the symbol indicates that this product may not be treated as household waste. By ensuring thisproduct is disposed of correctly, you will help prevent potential negative consequences for theenvironment and human health, which could otherwise be caused by inappropriate waste handling of thisproduct. For more detailed information about the take-back and recycling of this product, please contact
your supplier where you purchased the product.

”This symbol is applicable for EU member states only”

4
D4168764A

Rev 1.0

1. Overview
This CCD Color camera is a high-resolution camera that features all pixel readout mode 2/3 CCD.

2. Features(1) High resolution
Bayer array high pixel density CCD (number of effective pixels 5.05 M, number of total pixels 5.24 M) is used.
(2) Square grids
The CCD pixels arrayed in square grids facilitates computation for image processing.
(3) Full-frame shutter
Since all pixels are output even by shutter operation, high resolution can be achieved, without deteriorating thevertical resolution.
(4) Camera link interface
The interface of image output and the camera control adopts the Camera Link standard.

When you use frame grabber board for the camera link, high-speed image transfer and various controls to PC are
possible.
(5) All-pixel readout mode (normal mode)
All pixel signals (in the effective area) are output in approximately 1/9.3 second.
(6) Programmable partial scan mode
Partial scan within the range arbitrary from 100 lines to 2456 lines is possible.
(7) High-speed draft readout mode
By reading 4 lines from every 16 lines, all signals in the effective area are output in approximately in 1/31.3
second.
(8) Random trigger shutter
By external trigger signal input, the shot image can be grabbed at an arbitrary timing.
(9) Multiple-shutter
By external trigger signal input, the shot image can be grabbed at an arbitrary timing and the accumulated shot
images can be output at an arbitrary timing.

3. Configuration(1) Camera body ················································································· 1(2) Accessories
Instruction Manual (Japanese)···················································· 1
Instruction Manual (English)······················································ 1

4. Optional parts(1) I/O cable
Model name: CPRC3700-**(2) Camera Link cable
Model name: 14B26-SZLB-***-0LC(3) Camera adapter
Model name: CA170(4) Camera mounting kit
Model name: CPT4000F*NOTE: Application software is not supplied as a standard item.

Notes on optional parts and compliance with safety standard conditions:
We assure the compliance of this camera with the safety standard when it is used in combination with theoptional parts listed above.
If you use the camera in combination with parts other than specified by our company, you are responsible
for finally confirming the compliance with the safety standard by using the entire machine/equipment.

5
D4168764A

Rev 1.0

5. Specifications[Electrical specification](1) Imager
・Number of total pixels
・Number of effective pixels
・Number of picture output
effective pixels
・Pixel size
・Optical size(2) Scan method(3) Aspect ratio(4) Synchronization method(5) Standard subject illuminance(6) Minimum subject illuminance(7) Video output
・Data・Readout mode
All pixel readout
High-speed draft readout
Partial scanInterline CCD
2536 (H) x 2068 (V)2456 (H) x 2058 (V)2448 (H) x 2058 (V)3.45µm (H) x 3.45µm (V)
2/3 type
Progressive
6:5
Internal synchronization
2200 lx, F11, 5100 K
9 lx（F1.4, GAIN MAX, all pixel readout, video level 50 %,γ=0.65 equivalent）
Compliant with the camera link standard. Base configuration 1tap
RGB 24bit / RAW 12bit / RAW 10bit / RAW 8bit switching(initial factory setting: RGB)
Approximately 9.3 fps / 2448 (H) x 2058 (V)Approximately 31.3 fps / RGB 2448(H)×253(V), RAW 2448(H)×254(V)Approximately 9.3 fps to 47.1fps / 2448(H) x 100 to 2058(V)* About an image output:
As for CCD used with this camera, the left half and the right half of the screen are
independently output (2ch output method). Therefore, it is not failure, although a level
difference may be visible to an image on either side or a boundary line may be visible to middleof the screen boarding on middle of the screen with the setting mode of a camera.
* About a right and left level difference between TAP of the image output
In the test environment of the Teli standard, we measure the level difference by using
the gray scale chart of ITE-Ⅱ type. We measure a center white part of the chart.
- Shipment standard (difference of the right part and left part ): 105 [digit] or less (12bit)
In the following cases, the camera might not be able to correct the level difference
automatically.
- When the image level of a right and left center part is low- When the image level of a right and left center part is high- When the level difference of each pixel (right and left part of center) is large
When you cannot correct the level difference, please correct the level difference by thefollowing method.
- After the source of light and object are changed, one push correction is done.
- The level difference is corrected with manual.
In the shipment setting, the calibration mode is AUTO.
(8) Gain(9) Setup(10) White balance
Effective range
・OPWB
Effective area

0 to +6dB (61 levels) (initial factory setting: 0 dB)
0 to 33(528) LSB (RAW 12bit) (529 levels)(initial factory setting: 16(264) LSB (calculated value) )OPWB/MANUAL switching (initial factory setting: MANUAL)
2400 K to 9000 K
Full screen
6
D4168764A

Rev 1.0
・MANUAL
Preset value
User setting(11) Gamma correction(12) Masking correction(13) Calibration(14) Power supply voltage(15) Power consumption［Electronic shutter specification］］(1) Shutter speed
・Readout mode
All pixel readout
Partial scan
High-speed draft readout(2) Random trigger shutter
・Fixed mode・Pulse width mode(3) Multiple-shutter[Internal sync signal specification](1) Driving frequency(2) Scanning frequency
・Readout mode
All pixel readout
High-speed draft readout
Programmable partial scan

［Input signal specification］］(1) TRIG
・Signal level (I/O input)
・Polarity
・Pulse width(2) MULTI・Polarity
・Pulse width

3000 K, 3700 K, 4000 K, 4500 K, 5500 K, 6500 K
R-gain and B-gain can be set independently.
OFF(γ=1.0) / PRESET(γ=0.65) / USER(LUT) switching(initial factory setting: PRESET)
ON(Six points can be corrected)/OFF switching (initial factory setting: OFF)MANUAL / AUTO / ONEPUSH switching (initial factory setting: AUTO)
DC12 V (DC10V to DC13.2V) (ripple 50 mV(p-p) or less)Approximately 4.6 W

2 to 1/20,000 s
2 to 1/20,000 s
2 to 1/20,000 s
ON/OFF switching (initial factory setting: OFF)The exposure time depends on the shutter speed setting.
The exposure time depends on the pulse width.
ON/OFF switching (initial factory setting: OFF)
Exposure by TRIG input, readout by MULTI input* Enabled when random trigger shutter is ON.

49.0909 MHz

Horizontal
VerticalHorizontal
VerticalHorizontal
Vertical19.21 kHz9.30 Hz8.21 kHz31.14 Hz19.21 kHz
Arbitrariness

Camera link I/F and I/O connector input
TTL level
Positive/Negative polarity switching possible (initial factory setting: Negative)
52.1µs or more
Camera link I/F input
Negative polarity
52.1µs to 10 ms[Output signal specification](1) VDI/O connector output
・Signal level
4 V (p-p)
・Polarity
Negative polarity(2) EXPOSURE (Exposure period) I/O connector output
・Signal level
4 V (p-p)
・Polarity
Negative polarity

7
D4168764A

Rev 1.0［Dimensions］］(1) Lens mountC-mount

Notes on combination of C-mount lens:
Depending on the lens you use, the performance of the camera may not be brought out fully due
to the deterioration in resolution and brightness in the peripheral area, occurrence of a ghost,aberration and others.When you check the combination between the lens and camera, be sure to
use the lens you actually use.
As for the C-mount lens used combining this camera, the projection distance from bottom of thescrew should use 9.5mm or less.
Bottom ofthe screw

C-mount lens

9.5mm or less

(2)
(3)
(4)
(5)

Flange back17.526 mm
Dimensions
54 mm (W) x 43 mm (H) x 69 mm (D)MassApproximately 190 g
Camera body grounding: insulation status
Conductive between circuit GND and camera body

［Operating ambient conditions］］(1) Performance assurance(2) Operating assurance(3) Storage environment

Temperature
Humidity
Temperature
Humidity
Temperature
Humidity: 0 to 40°C
: 10 to 90% (no condensation): -5 to 45°C
: 10 to 90% (no condensation): -20 to 60°C
: 10 to 90% (no condensation)

8
D4168764A

Rev 1.0［Typical spectral response］］
The lens characteristics and light source characteristics is not reflected in table.

1.0

Relative Response

0.9

GREEN

BLUE

0.8

RED

0.7
0.6
0.5
0.4
0.3
0.2
0.1
0.0
400

450

500

550

600

650

700

Wave Length [nm]［Applicable safety standards］］(1) EMC （Electro-Magnetic Compatibility）
EMI（Electro-Magnetic Interference）
EMS（Electro-Magnetic Susceptibility）(2) FCC［Communication specification］］(1) Communication speed(2) Start bit(3) Data bit(4) Parity bit(5) Stop bit(6) Handshake:EN61000-6-4 / 2001
:EN61000-6-2 / 2001
:FCC Part 15 Subpart B class A

9600/19200/38400/57600 bps
1
8
None
1
None

9
D4168764A

Rev 1.0[Connector pin assignment](1) Video output/controlling connector (Camera Link Base Configuration) CAMERA LINK
Outputs video signals and VALID, based on the camera link standard LVDS.
This connector is connected to the frame grabber board, image processing device and others.
・Connector model: DR 26-PIN connector 10226-2210PE (manufactured by 3M).
Pin No.
1
2
3
4
5
6
7
8
9
10
11
12
13

I/O
O
O
O
O
O
I
O
I
I
I
I
-

Signal NameGND
Tx OUT0Tx OUT1Tx OUT2Tx CLK OUTTx OUT3-

Pin No.
14
15
16
17
18
19
20
21
22
23
24
25
26

Ser TC（RxD）+
Ser TFG（TxD）CC1（TRIG）CC2（MULTI）+
CC3CC4+
GND

I/O
O
O
O
O
O
I
O
I
I
I
I
-

Signal NameGND
Tx OUT0+
Tx OUT1+
Tx OUT2+
Tx CLK OUT+Tx OUT3+
Ser TC（RxD）Ser TFG（TxD）+
CC1（TRIG）+
CC2（MULTI）CC3+
CC4GND(2) Connector for power supply and sync signal input/output I/OThis is a terminal used for power supply to the camera. This connector is connected to the power supply unit.
In addition, this connector is used for sync signal (VD, Exposure signal) output and external trigger signal
input.
・Connector (Camera side) : HR10A-10R-12PB(71) (Manufactured by HIROSE DENKI)
・Plug (Cable side): HR10A-10P-12S(73) (Manufactured by HIROSE DENKI) or equivalents
Pin No.

I/O

Signal Name

1
2
3
4
5
6
7
8
9
10
11
12

I
-

GND
+12V
GND
N.C.
GND
N.C.
VD
GND
N.C.
EXPOSURE
TRIG
GND

O
O
-

１
２

９
８

10

３

11 12 ７
４

５

６

Rearview

10
D4168764A

Rev 1.0[Switch setting]
By using the DIP switches on the back surface of the camera body, you can set serial transmission speed andmemory readout for when the power supply is turned on.
If you change the switch setting after the power supply is turned on, the change is not reflected.

1
2
3

O
N

Not usedTransmission speed setting

4
5

Memory readout setting

6
7
8
9

Not used

10(1) Transmission speed setting
You can set the speed of serial transmission by camera link.
SW2
SW3
Transmission speed
OFFOFF
9600 bps
ON
OFF
19200 bps
OFFON
38400 bpsONON
57600 bps(2) Memory readout setting
You can set the number of the setting value saving memory bank to be called when the power supply is turned on.
The memory consists of 8 banks.
SW4
SW5
SW6
Memory number
OFF
OFF
OFF
1
ON
OFF
OFF
2
OFF
ON
OFF
3
ON
ON
OFF
4
OFF
OFF
ON
5
ON
OFF
ON
6
OFF
ON
ON
7
ON
ON
ON
8

11
D4168764A

Rev 1.0［Timing chart］］(1) Horizontal timing
1) All pixel readout, Partial scan
LVAL
DVAL
DATA OUT1DATA OUT2aa = 45CLK

c
d

b
e

b = 2448CLK

c = 44CLK

d = 18 or 19CLK

e = 2555 or 2556CLK

2) High-speed draft readout
LVAL
DVAL
DATA OUT1DATA OUT2
a

c

b

d

e

a = 45CLK

b = 2448CLK

c = 44CLK

d = 3443 or 3444CLK

e = 5980 or 5981CLK3) CLK ｒate
DATA CLOCKDVALDATA OUT

1

2

3

4

2445

2446

2447

2448

a
b

a = 20.37ns（ 49.0909MHz)b = 49.87us12
D4168764A

Rev 1.0(2) Vertical timing
1) All pixel readout＊ex） shutter mode：Shutter off
The period of A/C depend on the shutter speed.

2) High-speed draft readout＊ex） shutter mode：Shutter off
The period of A/C depend on the shutter speed.
FVAL
LVAL
DVAL
DATA OUT
A

B

C

D

E

A = approx. 894us (shutter off)

B = 1H(RAW) or 2H(RGB)

D = approx 69.2us

E = A+B+C+D

C = [254H(RAW) or 253H(RGB)]-3446CLK(1H: 5981CLK ; The period of A becomes long for 1/31.3s shutter speed or more. )13
D4168764A

Rev 1.0
3) Partial scan
·One step partial scanning·Two step partial scanning

14
D4168764A

Rev 1.0

6. Serial communication
By Camera link serial communication interface, it is possible to control the following functions.
(1) Memory control- Store- Readout- Reset(2) Video output mode
RGB 24bit / RAW 12bit / RAW 10bit / RAW 8bit
(3) Setup0 - 33LSB (528LSB:RAW12Bit)(4) Gain
0 - +6 dB(5) Shutter speed
2 - 1/20,000 s(6) Random shutter trigger ON / OFF- Trigger polarity
Positive / Negative- Random trigger mode
Pulse width mode / FIX mode(7) Multiple shutter
ON / OFF(8) High-speed draft readout
ON / OFF(9) Programmable partial scan
ON / OFF* High-speed draft readout mode must be disabled.
- Start position of effective line 0 – 1958 line (1line / 1step)- Number of effective line
100 – 2058 line (1line / 1step)(10) White balance
OPWB/MANUAL- MANU
Preset (6 data) / User manual (R / B gain)(11) Gamma
ON / OFF(12) Masking correction
ON / OFF (6 Parameter correction)(13) TAP correction of
the left half and the right half
AUTO / ONE PUSH / MANUAL

7. Functions
By accessing the camera register published on the camera link I/F, you can control/set each function.
Since access to the camera register is performed via the frame grabber board, the controlling and setting methods
differ depending on the frame grabber board you use. For details, refer to the instruction manual of the relevant
frame grabber board or contact our sales representative.

7.1 Explanation of Each Function(1) Readout mode
Video is output from the camera link connector. The output video can be grabbed by the frame grabber board.
The frame rate and resolution of output images that this model supports are as follows:
1) All pixel readout: Approximately 9.3 fps / 2448(H) x 2058(V)2) High-speed draft readout : Approximately 31.1 fps / RGB 2448(H)×253(V), RAW 2448(H)×254(V)3) Partial scan
: Approximately 47.1 to 9.3 fps / 2448(H) x 100(V) to 2058(V)1) All pixel readout
Reads out all pixels in about 1/9.3 second.

15
D4168764A

Rev 1.0
Approximately 2066H
FVAL
LVAL

1
2
3
4
5
6
7
8

2055
2056
2057
2058

DVAL

DATA OUT

2) High-speed draft readout
By reading 4 lines out of every 16 lines, reads out the whole valid area in approximately 1/31.3 seconds.
Control and setting of functions can be done by accessing the cam. When the frame rate of a partial scanning
exceeds it when the shutter mode is switched from normality (internal synchronization) or the partial
scanning mode to the draft mode, more high-speed than the setting of the speed of the shutter it is changed at
the speed of the shutter matched to the frame rate.
Approximately 264H
FVAL
LVAL

1+5
8+12
17+21
24+28
33+37
40+44

2025+2028
2033+2037

DVAL

DATA OUT

Reading 4 lines out of every 16 lines in
the inside of a total of 2058 lines

Effective lines : 254 lines

16
D4168764A

Rev 1.0
3) Programmable partial scan
A range arbitrary from 100 lines to 2058 lines can be read. The frame rate can be raised to 47.1 fps or less byskipping it at high speed excluding an effective area.
When the frame rate of a partial scanning exceeds it when the shutter mode is switched from normality(internal synchronization) to the partial scanning mode, more high-speed than the setting of the speed of the
shutter it is changed at the speed of the shutter matched to the frame rate.
･Starting position (Partial V Start1,2) of an effective line can be set.
･Number (Partial Height1,2) of effective lines can be set.
Example: (Partial V Start1, Partial Height1) = 100 effective lines and start 980 line eyes(center partial)(Partial V Start2, Partial Height2) = 0 effective lines and start 0 line eyes(center partial)
Approximately 596H
FVAL
LVAL

980
981
982
983
984
985
986

1077
1078
1079

DVAL

DATA OUT

Skipped lines ： 979 line

High-speed transfer ： 245 line＋
Nomal-speed transfer ： 3 line

Effective line : 100 line

Effective line : 100 line

High-speed transfer ： 248 line

Skipped lines ： 979 line

Example: (Partial V Start1, Partial Height1) = 100 effective lines and start 1 line eyes(top partial)(Partial V Start2, Partial Height2) = 100 effective lines and start 1959 line eyes(bottom partial)
Approximately 674H
FVAL
LVAL

2056
2057
2058

1959
1960
1961

96
97
98
99
100

1
2
3
4
5

DVAL

DATA OUT

Skipped lines ： 100 line
Effective line ： 100 line
Effective line ： 1858 line

High-speed transfer ： 467 line＋
Nomal-speed transfer ： 3 line
Effective line ： 100 line

Skipped lines ： 100 line

17
D4168764A

Rev 1.0
7.2 Rndom trigger shutter
In the random trigger shutter mode, you can shoot and grab an image at an arbitrary timing by trigger signal input
from the external.
・External trigger signals can be input either from the camera link I/F CC1 or I/O connector.
・If polarity is set to negative polarity, exposure starts at the falling edge of the trigger.
・The random trigger shutter of this camera can be operated in two types of mode: fixed mode and pulse
width mode. How to determine the exposure time differs depending on the mode.
1) Fixed mode
・The exposure time is determined by the setting value for the shutter speed.
exposure time

※ H ･･･ It differs depending on the image output mode.

TRIG
approx.
1.58 usCCD※5.6H （ 14,485CLK)
FVAL
LVAL
DVAL
DATA OUT

2) Pulse width mode
・The exposure time is determined by the pulse width (exposure time = pulse width + approximately 14µs).
・Set a pulse width of 1H (approximately 52.1µs) or more.
exposure time

※ H ･･･ It differs depending on the image output mode.

TRIG
approx.
1.58 usapprox.
15.2usCCD※5.6H （ 14,485CLK)
FVAL
LVAL
DVAL
DATA OUT

Notes of trigger mode:
When the interval of the input trigger signal is extremely short, or when the trigger signal is noisy, there
is a possibility of causing the malfunction. In this case, please input a proper trigger signal.

18
D4168764A

Rev 1.0
7.3 Multiple-shutter mode
In the multiple-shutter mode, video is output in sync with a MULTI signal from the external after the end of
exposure time.
・Valid only when the random trigger shutter mode is ON.
・MULTI signals can be input from the camera link I/F CC2.
・If exposure is executed several times before MULTI signal input, the images are output superposed.
・The exposure time is determined by the random trigger shutter mode setting and its determination method.
・The pulse width must be set to negative polarity and 52.1µs to 10 ms.
※ H ･･･ It differs depending on the image output mode.
TRIG
MULTI
A

B

CCD
※4.5H

（ 11,495CLK）

FVAL
LVAL
DVAL
DATA OUT
A+B

Notes on multiple-shutter:
An image has been stored to CCD device until a MULTI signal is inputted and it begins to read out
image after CCD is exposed. Therefore, an image may degenerate step by step when an image is stored
to CCD device for a long time.
And, Electric charge is superimposed in CCD device when the multiplex exposure. Therefore, CCD willflood with electric charge when electric charge is over superimposed in CCD device. Consequently
vertical stripes noise will appear. At that time please stop using spotlight and dimmer for example close
the iris.

19
D4168764A

Rev 1.0

8. Outline Drawing

20
D4168764A

Rev 1.0

9. Guarantee
The term of a guarantee is one year after the product delivery.
If by any chance trouble by responsibility of our company occurs before an above period, TELI repairs it free ofcharge.
During terms of a guarantee, when the trouble cause is the case of below, TELI charges the repair costs.
(1) Troubles and the damages that causes by misuse, unsuitable repair or remodeling.
(2) Distribution hazards like drops and vibrations after purchase. Troubles and damages by transportation.
(3) Troubles and damages by fire, natural calamity (earthquake, storm and flood damage, thunderbolt), damages
from salty breeze, gas harm, abnormal voltage.

10. Repair(1) Condition for repair
Basically, has to return it to our company when the user requests us to repair product.
Beside that, customer should pay these expenses (travel expenses, camera disassembly technology costs) of
both customer and end user. Also customer should pay in themselves costs for return camera to us.
(2) The period of repairing product· Repair free of charge ... Refer to Clause 9.
· Charged repair ............. Basically, repair period is 7 years after the last production end of products.

21
D4168764A

Rev 1.0

Head Office : 7-1, 4 chome, Asahigaoka, Hino-shi, Tokyo, 191-0065, Japan(International Business Department)
Phone : +81-42-589-8771
Fax: +81-42-589-8774
URL: http://www.toshiba-teli.co.jp/

Distributor• This product must be classified for disposal according to the laws of each country and municipal laws.
• Information contained in this document is subject to change without prior notice.

D4168764A