Machining process with a tool bit

Bonding of sensors

ImageIR® 8300 hp

High-speed Thermography Camera – Allrounder in VGA Format
Detector Format
Large detector enables
highest sensivity
MicroScan(1,280 × 1,024) IR pixels bygenuine camera hardware
640×512355 Hz

IR-Frame Rate
Analysis of extreme temperaturechanges and gradients in full frame
Measurement Accuracy
Highly accurate andrepeatable measurements
10 GigE Interface
High-speed, long-distance interference
proof data transmission
HighSense
Flexible setting of temperature measurement ranges / integration times beyond calibration ranges
HDR
Facilitates the analysis of objects withextreme temperature gradients

With its ImageIR® 8300 hp, InfraTec introduces another top level
thermographic camera model belonging to the ImageIR® highend camera series. The implementation of a digitally interfaced(640 × 512) IR pixels MWIR detector (snapshot) allows 355 Hz
full-frame real-time imaging without compromising any thermal accuracy. The ImageIR® 8300 hp and its cooled focal-planearray photon detector reach an outstanding thermal resolution– better than 0.02 K – and extremely short integration times inthe microsecond range. The new version was developed formost demanding operations in research and development andprocess monitoring fields. Thanks to HighSense, ImageIR® users
have the option of setting up individual measuring ranges andintegration times while maintaining the factory calibration andthus aligning the camera performance with the respective task.
Its modular structure consisting of the optical, detector and interface section, makes the camera easily compatible to the related
applications and for tailored configurations. An integrated trigger interface guarantees a repeatable high-precision triggeringof quick procedures. Multiple configurable digital inputs andoutputs serve as control ports for the camera or as generator of
digital control signals for external devices. The optical channel
consists of the exchangeable infrared lens completed by application-specific apertures, filters and reference elements. All exchangeable ImageIR® 8300 hp standard lenses can be combined witha motorised focus unit easily operable from the camera‘s application software. As part of the autofocus function it allows precise,fast and remotely controlled motorised focusing.
www.InfraTec-infrared.com

www.InfraTec.eu

Technical Specifications
Spectral range

MCT: (1.5 … 5.5) µm
InSb: (1.5 … 5.7) µm

Pitch

15 µm

Detector

MCT or InSb

Detector format (IR pixels)(640 × 512)

Image format with opto-mechanical MicroScan (IR pixels)*(1,280 × 1,024)

Image acquisition

SnapshotReadout mode

ITR / IWR

Aperture ratio

f/3.0

Detector cooling

Stirling cooler

Temperature measuring range(-40 … 1,500) °C, up to 3,000 °C*

Measurement accuracy± 1 °C or ± 1%

Temperature resolution @ 30 °C

MCT: Better than 0.02 K; InSb: Better than 0.025 K

Frame rate (full / half / quarter / sub frame)*

MCT: Up to 232 / 828 / 2,300 / 3,725 Hz
InSb: Up to 355 / 670 / 1,200 / 5,000 Hz

Window mode

Yes

Focus

Manual, motorised or automatic*

Dynamic range

Up to 16 bit*

Integration time

(0.6 … 20,000) µs

Rotating filter wheel*

Up to 6 positions

Rotating aperture wheel*

Up to 5 positions

Interfaces

GigE, 10 GigE*, 2 × CAMLink*, HDMI*

Trigger

4 IN / 2 OUT, TTL

Analogue signals*, IRIG-B*

2 IN / 2 OUT, yes

Tripod adapter

1/4" and 3/8" photo thread, 2 × M5

Power supply

24 V DC, wide-range power supply (100 … 240) V AC

Storage and operation temperature(-40 … 70) °C, (-20 … 50) °C

Protection degree

IP54, IEC 60529

Dimensions; weight

MCT: (241 × 120 × 160) mm*; InSb: (235 × 120 × 160) mm*; 3.3 kg (without lens)

Further functions

Multi Integration Time*, HighSense*, HDR*

Analysis and evaluation software

IRBIS® 3, IRBIS® 3 view, IRBIS® 3 plus*, IRBIS® 3 professional*, IRBIS® 3 control*,IRBIS® 3 online*, IRBIS® 3 process*, IRBIS® 3 active*, IRBIS® 3 mosaic*, IRBIS® 3 vision** Depending on modelLenses

Focal length (mm)

FOV (°)

Wide-angle lens12(43.6 × 35.5)

1.3Standard lens25(21.7 × 17.5)

0.6

Telephoto lens

50

(11.0 × 8.8)

0.3

Telephoto lens

100

(5.5 × 4.4)

0.15

Telephoto lens

200

(2.7 × 2.2)

0.08

Minimum object
distance (mm)Object size
(mm)Pixel size(µm)

Close-up for telephoto lens 50 mm

300(58 × 46)

90

Close-up for telephoto lens 100 mm

500(48 × 38)75

Microscopic lens M=1.0×

40 / 195 / 300(9.6 × 7.7)

15

Microscopic lens M=3.0×22(3.2 × 2.6)

5

Microscopic lens M=8.0×14(1.2 × 0.96)

1.9

Macro andmicroscopic lenses

IFOV (mrad)

Airbag test

Impact of a steel ball

© InfraTec 04 / 2021 – All stated product names and trademarks remain in property of their respective owners. Design, specification and technical progress subject to change without prior notice.