TI01287D/06/EN/09.22-00
71574742
2022-08-01

Products

Solutions

ServicesTechnical Information
Proline Promass Q 500
Coriolis flowmeter

Innovative specialist for challenging applications, as remote version with up
to 4 I/Os
Application

Your benefits• Measuring principle operates independently of physical fluid
properties such as viscosity or density• Highest measurement performance for custody transfer,density and under tough process conditions• Secured measuring quality – unmatched accuracy of mass
flow, volume flow and density• Optimized performance for liquids with entrained gas – MFT(Multi-Frequency Technology)• Fewer process measuring points – multivariablemeasurement (flow, density, temperature)• Space-saving installation – no in-/outlet run needs• Full access to process and diagnostic information –numerous, freely combinable I/Os and Ethernet• Reduced complexity and variety – freely configurable I/Ofunctionality• Integrated verification – Heartbeat Technology

Device properties• Mass/volume flow: measurement error ±0.05 %• Density: measurement error ±0.1 kg/m3• High turndown due to low pressure loss/zero point• Remote version with up to 4 I/Os
• Backlit display with touch control and WLAN access• Standard cable between sensor and transmitter

Proline Promass Q 500

Table of contents
About this document . . . . . . . . . . . . . . . . . . . . . . . .

4

Symbols . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

Function and system design . . . . . . . . . . . . . . . . . . . 5

Measuring principle . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Measuring system . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Equipment architecture . . . . . . . . . . . . . . . . . . . . . . . . .
Security . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

5
6
8
8

Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11

Measured variable . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11
Measuring range . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 11
Operable flow range . . . . . . . . . . . . . . . . . . . . . . . . . . . 12
Input signal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12

Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

14Power supply . . . . . . . . . . . . . . . . . . . . . . . . . . . . .

32

Output and input variants . . . . . . . . . . . . . . . . . . . . . . .
Output signal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Signal on alarm . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Load . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Ex connection data . . . . . . . . . . . . . . . . . . . . . . . . . . .
Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Galvanic isolation . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Protocol-specific data . . . . . . . . . . . . . . . . . . . . . . . . . .

14
16
22
24
24
26
26
26

Terminal assignment . . . . . . . . . . . . . . . . . . . . . . . . . . 32
Available device plugs . . . . . . . . . . . . . . . . . . . . . . . . . 34
Pin assignment, device plug . . . . . . . . . . . . . . . . . . . . . . 35
Supply voltage . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 37
Power consumption . . . . . . . . . . . . . . . . . . . . . . . . . . . 37
Current consumption . . . . . . . . . . . . . . . . . . . . . . . . . . 37
Power supply failure . . . . . . . . . . . . . . . . . . . . . . . . . . 37
Overcurrent protection element . . . . . . . . . . . . . . . . . . . 37
Electrical connection . . . . . . . . . . . . . . . . . . . . . . . . . . 37
Potential equalization . . . . . . . . . . . . . . . . . . . . . . . . . 51
Terminals . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Cable entries . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Cable specification . . . . . . . . . . . . . . . . . . . . . . . . . . . . 51
Overvoltage protection . . . . . . . . . . . . . . . . . . . . . . . . . 57

Performance characteristics . . . . . . . . . . . . . . . . . . 57

Reference operating conditions . . . . . . . . . . . . . . . . . . . 57
Maximum measured error . . . . . . . . . . . . . . . . . . . . . . . 57
Repeatability . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Response time . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Influence of ambient temperature . . . . . . . . . . . . . . . . . 60
Influence of medium temperature . . . . . . . . . . . . . . . . . . 60
Influence of medium pressure . . . . . . . . . . . . . . . . . . . . 60
Design fundamentals . . . . . . . . . . . . . . . . . . . . . . . . . . 61

Installation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62

Mounting location . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Orientation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Inlet and outlet runs . . . . . . . . . . . . . . . . . . . . . . . . . .
Mounting the transmitter housing . . . . . . . . . . . . . . . . .
Special mounting instructions . . . . . . . . . . . . . . . . . . . .

2

62
63
64
65
66

Environment . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 68

Ambient temperature range . . . . . . . . . . . . . . . . . . . . .
Storage temperature . . . . . . . . . . . . . . . . . . . . . . . . . .
Climate class . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Relative humidity . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Operating height . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Degree of protection . . . . . . . . . . . . . . . . . . . . . . . . . .
Vibration- and shock-resistance . . . . . . . . . . . . . . . . . . .
Interior cleaning . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Mechanical load . . . . . . . . . . . . . . . . . . . . . . . . . . . . .
Electromagnetic compatibility (EMC) . . . . . . . . . . . . . . .

68
69
69
69
69
69
69
70
70
70

Process . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 70

Medium temperature range . . . . . . . . . . . . . . . . . . . . . . 70
Density . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 71
Pressure-temperature ratings . . . . . . . . . . . . . . . . . . . . 71
Sensor housing . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 74
Rupture disk . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Flow limit . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
Pressure loss . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 75
System pressure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Thermal insulation . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Heating . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Vibrations . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 77

Custody transfer . . . . . . . . . . . . . . . . . . . . . . . . . . . 78
Mechanical construction . . . . . . . . . . . . . . . . . . . .

79

Dimensions in SI units . . . . . . . . . . . . . . . . . . . . . . . . . 79
Dimensions in US units . . . . . . . . . . . . . . . . . . . . . . . . . 95
Weight . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 103
Materials . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 103
Process connections . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Surface roughness . . . . . . . . . . . . . . . . . . . . . . . . . . . 106

Operability . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 106

Operating concept . . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Languages . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Local operation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Remote operation . . . . . . . . . . . . . . . . . . . . . . . . . . . 107
Service interface . . . . . . . . . . . . . . . . . . . . . . . . . . . . 113
Network integration . . . . . . . . . . . . . . . . . . . . . . . . . . 114
Supported operating tools . . . . . . . . . . . . . . . . . . . . . . 115
HistoROM data management . . . . . . . . . . . . . . . . . . . . 116

Certificates and approvals . . . . . . . . . . . . . . . . . .

118

CE mark . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 118
UKCA marking . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 118
RCM mark . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 118
Ex approval . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 118
Hygienic compatibility . . . . . . . . . . . . . . . . . . . . . . . . 121
Pharmaceutical compatibility . . . . . . . . . . . . . . . . . . . . 122
Functional safety . . . . . . . . . . . . . . . . . . . . . . . . . . . . 122
HART certification . . . . . . . . . . . . . . . . . . . . . . . . . . . 122
FOUNDATION Fieldbus certification . . . . . . . . . . . . . . . 122
Certification PROFIBUS . . . . . . . . . . . . . . . . . . . . . . . 122
EtherNet/IP certification . . . . . . . . . . . . . . . . . . . . . . . 122

Endress+Hauser

Proline Promass Q 500

Certification PROFINET . . . . . . . . . . . . . . . . . . . . . . . . 122
Certification PROFINET with Ethernet-APL . . . . . . . . . . 123
Pressure Equipment Directive . . . . . . . . . . . . . . . . . . . 123
Radio approval . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Measuring instrument approval . . . . . . . . . . . . . . . . . . 123
Additional certification . . . . . . . . . . . . . . . . . . . . . . . . 123
Other standards and guidelines . . . . . . . . . . . . . . . . . . 124

Ordering information . . . . . . . . . . . . . . . . . . . . . . 124
Application packages . . . . . . . . . . . . . . . . . . . . . . 125

Diagnostic functionality . . . . . . . . . . . . . . . . . . . . . . . 125
Heartbeat Technology . . . . . . . . . . . . . . . . . . . . . . . . 125
Concentration measurement . . . . . . . . . . . . . . . . . . . . 125
Advanced density function . . . . . . . . . . . . . . . . . . . . . 126
Premium density and extended density function . . . . . . . 126
Petroleum . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 126
Petroleum & locking function . . . . . . . . . . . . . . . . . . . . 126
OPC-UA Server . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 126

Accessories . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 126

Device-specific accessories . . . . . . . . . . . . . . . . . . . . . . 127
Communication-specific accessories . . . . . . . . . . . . . . . 128
Service-specific accessories . . . . . . . . . . . . . . . . . . . . . 129
System components . . . . . . . . . . . . . . . . . . . . . . . . . . 129

Documentation . . . . . . . . . . . . . . . . . . . . . . . . . . . 130

Standard documentation . . . . . . . . . . . . . . . . . . . . . . . 130
Supplementary device-dependent documentation . . . . . . 131

Registered trademarks . . . . . . . . . . . . . . . . . . . . . 132

Endress+Hauser

3

Proline Promass Q 500

About this document
Symbols

Electrical symbols
SymbolMeaning
Direct current
Alternating current
Direct current and alternating current
Ground connection
A grounded terminal which, as far as the operator is concerned, is grounded via agrounding system.
Potential equalization connection (PE: protective earth)Ground terminals that must be connected to ground prior to establishing any otherconnections.
The ground terminals are located on the interior and exterior of the device:• Interior ground terminal: potential equalization is connected to the supply network.
• Exterior ground terminal: device is connected to the plant grounding system.

Communication-specific symbols
SymbolMeaning
Wireless Local Area Network (WLAN)
Communication via a wireless, local network.
LED
Light emitting diode is off.
LED
Light emitting diode is on.
LED
Light emitting diode is flashing.

Symbols for certain types of information
SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Preferred
Procedures, processes or actions that are preferred.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.
Reference to documentationA

Reference to page
Reference to graphic

Visual inspection

4

Endress+Hauser

Proline Promass Q 500

Symbols in graphics
SymbolMeaning

1, 2, 3, ...

Item numbers

1. , 2. , 3. , …

Series of steps

A, B, C, ...

Views

A-A, B-B, C-C, ...

Sections-

Hazardous area.

Safe area (non-hazardous area)
Flow direction

Function and system design
Measuring principle

The measuring principle is based on the controlled generation of Coriolis forces. These forces arealways present in a system when both translational and rotational movements are superimposed.
Fc = 2 · ∆m (ν · ω)
Fc = Coriolis force∆m = moving mass
ω = rotational velocity
ν = radial velocity in rotating or oscillating system

The amplitude of the Coriolis force depends on the moving mass ∆m, its velocity ν in the system andthus on the mass flow. Instead of a constant rotational velocity ω, the sensor uses oscillation.
In the sensor, two parallel measuring tubes containing flowing fluid oscillate in antiphase, acting likea tuning fork. The Coriolis forces produced at the measuring tubes cause a phase shift in the tube
oscillations (see illustration):• At zero flow (when the fluid is at a standstill) the two tubes oscillate in phase (1).
• Mass flow causes deceleration of the oscillation at the inlet of the tubes (2) and acceleration at theoutlet (3).

1

2

3A0028850

The phase difference (A-B) increases with increasing mass flow. Electrodynamic sensors register the
tube oscillations at the inlet and outlet. System balance is ensured by the antiphase oscillation of thetwo measuring tubes. The measuring principle operates independently of temperature, pressure,viscosity, conductivity and flow profile.

Endress+Hauser

5Proline Promass Q 500

Density measurement
The measuring tube is continuously excited at its resonance frequency. A change in the mass andthus the density of the oscillating system (comprising measuring tube and fluid) results in acorresponding, automatic adjustment in the oscillation frequency. The resonance frequency is thus afunction of the medium density. The microprocessor utilizes this relationship to obtain a density
signal.
Volume measurement
Together with the measured mass flow, this is used to calculate the volume flow.
Temperature measurement
The temperature of the measuring tube is determined in order to calculate the compensation factor
due to temperature effects. This signal corresponds to the process temperature and is also available
as an output signal.
Multi-frequency technology (MFT)
Highly accurate measurement of two-phase flow (gaseous medium with suspended bubbles or micro
bubbles) using active compensation. Here, the two measuring tubes are excited simultaneously at
different resonance frequencies. By analyzing the resonance characteristics of both oscillations inrelation to the medium, measuring errors can be actively compensated.
Gas Fraction Handler (GFH)
The Gas Fraction Handler is a Promass software function that improves measurement stability andrepeatability. The function continuously checks for the presence of disturbances in single-phase flow,
i.e. gas bubbles in liquids or droplets in gas. In the presence of the second phase, flow and density
become increasingly unstable. The Gas Fraction Handler function improves measurement stability
with respect to the severity of the disturbances, without any effect under single-phase flow
conditions.
The Gas Fraction Handler is only available in device versions with HART, Modbus RS485,
PROFINET and PROFINET with Ethernet-APL.
For detailed information on the Gas Fraction Handler, see the Special Documentation for "Gas
Fraction Handler" →  131
The measuring system consists of a transmitter and a sensor. The transmitter and sensor are
mounted in physically separate locations. They are interconnected by connecting cables.

Measuring system

TransmitterTwo versions of the transmitter are available.
Proline 500 – digital 1)
For use in applications not required to meet special requirements due toambient or operating conditions.

Proline 500
For use in applications required to meet special requirements due toambient or operating conditions.

1

1
A
B

2
2
3

3
A
B
1
2
3

Non-hazardous area or Zone 2; Class I, Division 2
Non-hazardous area or Zone 2; Class I, Division 2 or Zone 1; Class I,
Division 1
Transmitter
Connecting cable: cable, separate, standard
Sensor connection housing with integrated ISEM

• Flexible and cost-effective separate installation.
• A standard cable can be used as the connecting cable.

6

1
2
3

Non-hazardous area or Zone 2; Class I, Division 2 or Zone 1; Class I,
Division 1
Transmitter with integrated ISEM
Connecting cable: cable, separate
Sensor connection housing

Application examples for sensors without electronics:
Strong vibrations at the sensor.

Endress+Hauser

Proline Promass Q 500

Proline 500 – digital 1)• Electronics in the transmitter housing, ISEM (intelligent sensor
electronics module) in the sensor connection housing• Signal transmission: digital
Order code for "Integrated ISEM electronics", option A "Sensor"

Proline 500• Electronics and ISEM (intelligent sensor electronics module) in thetransmitter housing• Signal transmission: analog
Order code for "Integrated ISEM electronics", option B "Transmitter"

Connecting cable (can be ordered in various lengths →  127 )• Length:• Zone 2; Class I, Division 2: max. 300 m (1 000 ft)• Zone 1; Class I, Division 1: max. 150 m (500 ft)• Standard cable with common shield (pair-stranded)• Length: max. 20 m (65 ft)• Cable with a common shield and individual shielded cores (3 pairs)• DN 150 to 250 (6 to 10 "): Two cables with common shield andindividual shielded cores (3 pairs)

Hazardous area
Use in: Zone 2; Class I, Division 2

Use in: Zone 1; Class I, Division 1 or Zone 2; Class I, Division 2

Mixed installation is possible:• Sensor: Zone 1; Class I, Division 1• Transmitter: Zone 2; Class I, Division 2
Housing versions and materials• Transmitter housing• Aluminum, coated: aluminum, AlSi10Mg, coated• Material: polycarbonate• Material of window in transmitter housing• Aluminum, coated: glass• Polycarbonate: plastic• Transmitter housing• Aluminum, coated: aluminum, AlSi10Mg, coated• Cast, stainless: cast, stainless steel, 1.4409 (CF3M) similar to 316L
• Window material: glass

Configuration• External operation via 4-line, illuminated graphic local display (LCD) with touch control and guided menus ("Make-it-run" wizards) for applicationspecific commissioning.
• Via service interface or WLAN interface:• Operating tools (e.g. FieldCare, DeviceCare)• Web server (access via Web browser, e.g. Microsoft Internet Explorer, Microsoft Edge)1)

Not available for devices with DN 150, 200, 250 (6", 8", 10")

Sensor connection housing
Different versions of the connection housing are available.
Order code for "Sensor connection housing", option A, "Aluminum, coated":
Aluminum, AlSi10Mg, coateddevice version is only available in conjunction with the Proline
 This
500 – digital transmitter.
Order code for "Sensor connection housing", option B, "Stainless":• Hygienic version, stainless steel 1.4301 (304)• Optional: order code for "Sensor feature", option CC "Hygienic version,
for maximum corrosion resistance": stainless steel 1.4404 (316L)
Order code for "Sensor connection housing", option C, "Ultra-compact
hygienic, stainless":
Hygienic version, stainless steel 1.4301 (304)
device version is only available in conjunction with the Proline
 This
500 – digital transmitter.
Order code for "Sensor connection housing", option L, "Cast, stainless":
1.4409 (CF3M) similar to 316L

Endress+Hauser

7

Proline Promass Q 500

SensorPromass Q• Nominal diameter range: DN 25 to 250 (1 to 10 ")• Bent dual-tube system: DN 25 to 100 (1 to 4 ")• Four-tube system: DN 150 to 250 (6 to 10 ")• Excellent performance across a wide range of applications• Simultaneous measurement of flow, volume flow, density andtemperature (multivariable)• Immune to process influences• Materials:• Sensor: stainless steel, 1.4404 (316L)• Measuring tubes: stainless steel, 1.4404 (316/316L)• Process connections: stainless steel, 1.4404 (316/316L)
A0030075

Equipment architecture

1

5
3

2

4
6

3
7

A0027512

1
1
2
3
4
5
6
7

Security

Possibilities for integrating measuring devices into a system
Control system (e.g. PLC)
Connecting cable (0/4 to 20 mA HART etc.)
Fieldbus
Coupler
Non-hazardous area
Hazardous area: Zone 2; Class I, Division 2
Hazardous area: Zone 1; Class I, Division 1

IT security
Our warranty is valid only if the product is installed and used as described in the Operating
Instructions. The product is equipped with security mechanisms to protect it against any inadvertentchanges to the settings.
IT security measures, which provide additional protection for the product and associated datatransfer, must be implemented by the operators themselves in line with their security standards.
Device-specific IT security
The device offers a range of specific functions to support protective measures on the operator's side.
These functions can be configured by the user and guarantee greater in-operation safety if usedcorrectly. An overview of the most important functions is provided in the following section:

8

Function/interface

Factory setting

Recommendation

Write protection via hardware write
protection switch →  9

Not enabled

On an individual basis following risk assessment

Access code(also applies for Web server login or
FieldCare connection) →  9

Not enabled(0000)

Assign a customized access code duringcommissioning

Endress+Hauser

Proline Promass Q 500

Function/interface

Factory setting

Recommendation

WLAN(order option in display module)

EnabledOn an individual basis following risk assessment

WLAN security mode

Enabled (WPA2PSK)Do not change

WLAN passphrase(password) →  9Serial number

Assign a customized WLAN passphrase duringcommissioning

WLAN mode

Access point

On an individual basis following risk assessment

Web server →  10

EnabledOn an individual basis following risk assessment

CDI-RJ45 service interface →  10–

On an individual basis following risk assessment

Protecting access via hardware write protection
Write access to the parameters of the device via the local display, Web browser or operating tool(e.g. FieldCare, DeviceCare) can be disabled via a write protection switch (DIP switch on the mainelectronics module). When hardware write protection is enabled, only read access to the parameters
is possible.
Hardware write protection is disabled when the device is delivered.
Protecting access via a password
Different passwords are available to protect write access to the device parameters or access to thedevice via the WLAN interface.
• User-specific access code
Protect write access to the device parameters via the local display, Web browser or operating tool(e.g. FieldCare, DeviceCare). Access authorization is clearly regulated through the use of a userspecific access code.
• WLAN passphrase
The network key protects a connection between an operating unit (e.g. notebook or tablet) and thedevice via the WLAN interface which can be ordered as an option.
• Infrastructure mode
When the device is operated in infrastructure mode, the WLAN passphrase corresponds to the
WLAN passphrase configured on the operator side.
User-specific access code
Write access to the device parameters via the local display, Web browser or operating tool (e.g.
FieldCare, DeviceCare) can be protected by the modifiable, user-specific access code.
WLAN passphrase: Operation as WLAN access point
A connection between an operating unit (e.g. notebook or tablet) and the device via the WLAN
interface, which can be ordered as an optional extra, is protected by the network key. The WLAN
authentication of the network key complies with the IEEE 802.11 standard.
When the device is delivered, the network key is pre-defined depending on the device. It can bechanged via the WLAN settings submenu in the WLAN passphrase parameter.
Infrastructure mode
A connection between the device and WLAN access point is protected by means of an SSID andpassphrase on the system side. Please contact the relevant system administrator for access.
General notes on the use of passwords• The access code and network key supplied with the device should be changed duringcommissioning.
• Follow the general rules for generating a secure password when defining and managing the accesscode or network key.
• The user is responsible for the management and careful handling of the access code and network
key.

Endress+Hauser

9Proline Promass Q 500

Access via Web server
The device can be operated and configured via a Web browser with the integrated Web server. The
connection is via the service interface (CDI-RJ45) or the WLAN interface. For device versions with
the EtherNet/IP and PROFINET communication protocols, the connection can also be established via
the terminal connection for signal transmission with EtherNet/IP, PROFINET (RJ45 plug) or
PROFINET with Ethernet-APL (two-wire).
The Web server is enabled when the device is delivered. The Web server can be disabled if necessary(e.g. after commissioning) via the Web server functionality parameter.
The device and status information can be hidden on the login page. This prevents unauthorizedaccess to the information.
For detailed information on device parameters, see:
The "Description of Device Parameters" document →  130
Access via OPC-UA
The "OPC UA Server" application package is available in the device version with the HART
communication protocol →  126.
The device can communicate with OPC UA clients using the "OPC UA Server" application package.
The OPC UA server integrated in the device can be accessed via the WLAN access point using the
WLAN interface - which can be ordered as an optional extra - or the service interface (CDI- RJ45) via
Ethernet network. Access rights and authorization as per separate configuration.
The following Security Modes are supported as per the OPC UA Specification (IEC 62541):• None• Basic128Rsa15 – signed• Basic128Rsa15 – signed and encrypted
Access via service interface (CDI-RJ45)
The device can be connected to a network via the service interface (CDI-RJ45). Device-specificfunctions guarantee the secure operation of the device in a network.
The use of relevant industrial standards and guidelines that have been defined by national andinternational safety committees, such as IEC/ISA62443 or the IEEE, is recommended. This includes
organizational security measures such as the assignment of access authorization as well as technicalmeasures such as network segmentation.
Transmitters with an Ex de approval may not be connected via the service interface (CDI-RJ45)!
Order code for "Approval transmitter + sensor", options (Ex de): BA, BB, C1, C2, GA, GB, MA,
MB, NA, NB
The device can be incorporated into a ring topology. The device is integrated via the terminal
connection for signal transmission (output 1) and the connection to the service interface (CDIRJ45) .

10

Endress+Hauser

Proline Promass Q 500

Input
Measured variable

Direct measured variables• Mass flow
• Density• Temperature
Calculated measured variables• Volume flow
• Corrected volume flow
• Reference density

Measuring range

Measuring range for liquids
DN
Measuring device

DN
Compatible pipe diameter

Measuring range full scalevalues min(F) to max(F)

[mm]

[in]

[mm]

[in]

[kg/h][lb/min]

25

1

25/40

1/1½

0 to 20 000

0 to 735

50

2

50/80

2/3

0 to 80 000

0 to 2 94080

380/1003/4

0 to 200 000

0 to 7 350100

4100/150

4/6

0 to 550 000

0 to 20 210150

6150/200

6/8

0 to 850 000

0 to 31 240200

8200/250

8/10

0 to 1 500 000

0 to 55 130

250

10

250/300

10/12

0 to 2 400 000

0 to 88 200

Measuring range for gases
The full scale value depends on the density and the sound velocity of the gas used. The full scale value can becalculated with the following formulas:max(G) = minimum of(max(F) · ρG : x ) and(ρG · (cG/2) · di2 · (π/4) · 3600 · n)

 max(G)Maximum full scale value for gas [kg/h]

 max(F)Maximum full scale value for liquid [kg/h]

 max(G) <  max(F) max(G) can never be greater than  max(F)ρG

Gas density in [kg/m³] at operating conditionsx

Limitation constant for max. gas flow [kg/m³]

cG

Sound velocity (gas) [m/s]diMeasuring tube internal diameter [m]

π

Pi

n=2

Number of measuring tubes for DN 25 to 100 (1 to 4 ")n=4Number of measuring tubes for DN 150 to 250 (6 to 10 ")DN

Endress+Hauser

x

[mm]

[in]

[kg/m3]

25

1

90

50

2

80

11

Proline Promass Q 500

DN

x

[mm]

[in]

[kg/m3]

80

3

83

100

4

180

150

6

200

200

8

200

250

10

200

To calculate the measuring range, use the Applicator sizing tool →  129
If calculating the full scale value using the two formulas:1.

Calculate the full scale value with both formulas.

2.

The smaller value is the value that must be used.

Recommended measuring range
Flow limit →  75

Operable flow range

Over 1000 : 1.
Flow rates above the preset full scale value do not override the electronics unit, with the result thatthe totalizer values are registered correctly.

Input signal

Output and input variants→  14External measured values
Various pressure and temperature measuring devices can be ordered from Endress+Hauser: see"Accessories" section →  129
It is recommended to read in external measured values to calculate the corrected volume flow.
HART protocol
The measured values are written from the automation system to the measuring device via the HART
protocol. The pressure transmitter must support the following protocol-specific functions:• HART protocol• Burst mode
Current input
The measured values are written from the automation system to the measuring device via thecurrent input →  12.
Digital communication
The measured values can be written by the automation system via:
• FOUNDATION Fieldbus• PROFIBUS DP• PROFIBUS PA• Modbus RS485• EtherNet/IP• PROFINET• PROFINET with Ethernet-APL
Current input 0/4 to 20 mA

12Current input

0/4 to 20 mA (active/passive)

Current span

• 4 to 20 mA (active)• 0/4 to 20 mA (passive)

Endress+Hauser

Proline Promass Q 500

Resolution

1 µA

Voltage drop

Typically: 0.6 to 2 V for 3.6 to 22 mA (passive)

Maximum input voltage≤ 30 V (passive)

Open-circuit voltage≤ 28.8 V (active)

Possible input variables• Pressure• Temperature• Density

Status input

Endress+Hauser

Maximum input values• DC –3 to 30 V
• If status input is active (ON): Ri >3 kΩ

Response time

Configurable: 5 to 200 ms

Input signal level• Low signal: DC –3 to +5 V
• High signal: DC 12 to 30 V

Assignable functions• Off• Reset the individual totalizers separately• Reset all totalizers• Flow override

13

Proline Promass Q 500

OutputDepending on the option selected for output/input 1, different options are available for the other
outputs and inputs. Only one option can be selected for each output/input 1 to 4. The following
tables must be read vertically (↓).

Output and input variants

Example: If the option BA "4–20 mA HART" was selected for output/input 1, one of the options A, B,
D, E, F, H, I or J is available for output 2 and one of the options A, B, D, E, F, H, I or J is available for
output 3 and 4.
Output/input 1 and options for output/input 2
Options for output/input 3 and 4 →  15

Order code for "Output; input 1" (020) →Possible options

Current output 4 to 20 mA HARTBA

Current output 4 to 20 mA HART Ex i passive↓

Current output 4 to 20 mA HART Ex i active

CA↓

FOUNDATION Fieldbus

CC↓

FOUNDATION Fieldbus Ex iSA↓

PROFIBUS DPTA↓PROFIBUS PALA↓PROFIBUS PA Ex i

GA↓

Modbus RS485

HA↓

EtherNet/IP 2-port switch integrated

MA↓

PROFINET 2-port switch integrated

NA↓

PROFINET with Ethernet-APL

RA↓

PROFINET with Ethernet-APL Ex i

RB
↓

RC

Order code for "Output; input 2" (021) →

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

Not assigned

A

A

A

A

A

A

A

A

A

A

A

A

A

Current output 4 to 20 mA

B

B

B

B

B

B

B

Current output 4 to 20 mA Ex i passive
User-configurable input/output

1)

Pulse/frequency/switch output
Double pulse output

2)

B
C

C

C

C

C

D

D

D

D

D

D

D

D

E

E

E

E

E

E

E

E

F

Pulse/frequency/switch output Ex i passive

F
G

G

G

G

G

Relay output

H

H

H

H

H

H

H

H

Current input 0/4 to 20 mA

I

I

I

I

I

I

I

I

Status input

J

J

J

J

J

J

J

J

1)
2)

A specific input or output can be assigned to a user-configurable input/output →  21.
If double pulse output (F) is selected for output/input 2 (021), only the double pulse output (F) option is available for selection for output/input 3(022).

14

Endress+Hauser

Proline Promass Q 500

Output/input 1 and options for output/input 3 and 4
Options for output/input 2 →  14

Order code for "Output; input 1" (020) →Possible options

Current output 4 to 20 mA HARTBA

Current output 4 to 20 mA HART Ex i passive↓

Current output 4 to 20 mA HART Ex i active

CA↓

FOUNDATION Fieldbus

CC↓

FOUNDATION Fieldbus Ex iSA↓

PROFIBUS DPTA↓PROFIBUS PALA↓PROFIBUS PA Ex i

GA↓

Modbus RS485

HA↓

EtherNet/IP 2-port switch integrated

MA↓

PROFINET 2-port switch integrated

NA↓

PROFINET with Ethernet-APL

RA↓

PROFINET with Ethernet-APL Ex i
Order code for "Output; input 3" (022), "Output; input 4" (023)
Not assigned
Current output 4 to 20 mA

1)

→

↓

RC

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

↓

A

A

A

A

A

A

A

A

A

A

A

A

A

B

B

B

B

B

B

Current output 4 to 20 mA Ex i passive

RB

2)

C

C

User-configurable input/output

D

D

D

D

D

D

Pulse/frequency/switch output

E

E

E

E

E

E

Double pulse output (slave)3)

Pulse/frequency/switch output Ex i passive

F
4)

F
G

G

Relay output

H

H

H

H

H

H

Current input 0/4 to 20 mA

I

I

I

I

I

I

Status input

J

J

J

J

J

J

1)
2)
3)
4)

The order code for "Output; input 4" (023) is only available for the Proline 500-digital transmitter, order code for "Integrated ISEM electronics",
option A.
The current output 4 to 20 mA Ex i passive (C) option is not available for input/output 4.
The double pulse output (F) option is not available for input/output 4.
The pulse/frequency/switch output Ex i passive (G) option is not available for input/output 4.

Endress+Hauser

15

Proline Promass Q 500

Output signal

Current output 4 to 20 mA HART
Order code

"Output; input 1" (20):
Option BA: current output 4 to 20 mA HART

Signal mode

Can be set to:• Active• Passive

Current range

Can be set to:• 4 to 20 mA NAMUR• 4 to 20 mA US• 4 to 20 mA• 0 to 20 mA (only if the signal mode is active)• Fixed current

Open-circuit voltage

DC 28.8 V (active)

Maximum input voltage

DC 30 V (passive)

Load

250 to 700 Ω

Resolution

0.38 µA

Damping

Configurable: 0 to 999.9 s

Assignable measured
variables• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Electronics temperature• Oscillation frequency 0• Oscillation damping 0• Signal asymmetry• Exciter current 0
range of options increases if the measuring device has one or more
 The
application packages.

Current output 4 to 20 mA HART Ex i

16

Order code

"Output; input 1" (20) choose from:• Option CA: current output 4 to 20 mA HART Ex i passive• Option CC: current output 4 to 20 mA HART Ex i active

Signal mode

Depends on the selected order version.

Current range

Can be set to:• 4 to 20 mA NAMUR• 4 to 20 mA US• 4 to 20 mA• 0 to 20 mA (only if the signal mode is active)• Fixed current

Open-circuit voltage

DC 21.8 V (active)

Maximum input voltage

DC 30 V (passive)

Load• 250 to 400 Ω (active)• 250 to 700 Ω (passive)

Resolution

0.38 µA

Endress+Hauser

Proline Promass Q 500

Damping

Configurable: 0 to 999.9 s

Assignable measured
variables• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Electronics temperature• Oscillation frequency 0• Oscillation damping 0• Signal asymmetry• Exciter current 0
range of options increases if the measuring device has one or more
 The
application packages.

FOUNDATION Fieldbus
FOUNDATION Fieldbus

H1, IEC 61158-2, galvanically isolated

Data transfer

31.25 kbit/s

Current consumption

10 mA

Permitted supply voltage

9 to 32 V

Bus connection

With integrated reverse polarity protection

PROFIBUS DP
Signal encoding

NRZ code

Data transfer

9.6 kBaud…12 MBaud

Terminating resistor

Integrated, can be activated via DIP switches

PROFIBUS PAPROFIBUS PA

In accordance with EN 50170 Volume 2, IEC 61158-2 (MBP), galvanicallyisolated

Data transmission

31.25 kbit/s

Current consumption

10 mA

Permitted supply voltage

9 to 32 V

Bus connection

With integrated reverse polarity protection

Modbus RS485
Physical interface

RS485 in accordance with EIA/TIA-485 standard

Terminating resistor

Integrated, can be activated via DIP switches

EtherNet/IP
Standards

In accordance with IEEE 802.3

PROFINET
Standards

Endress+Hauser

In accordance with IEEE 802.317Proline Promass Q 500

PROFINET with Ethernet-APL
Device use

Device connection to an APL field switch
The device may only be operated according to the following APL port
classifications:• If used in hazardous areas: SLAA or SLAC 1)• If used in non-hazardous areas: SLAX
Connection values of APL field switch (corresponds to APL port classification
SPCC or SPAA, for instance):• Maximum input voltage: 15 VDC• Minimum output values: 0.54 W
Device connection to an SPE switch
The device may only be operated according to the following PoDL power class: If
used in the non-hazardous area: PoDL power class 10
Connection values of SPE switch (corresponds to PoDL power class 10, 11 or 12):
• Maximum input voltage: 30 VDC• Minimum output values: 1.85 W

PROFINET

According to IEC 61158 and IEC 61784

Ethernet-APL

According to IEEE 802.3cg, APL port profile specification v1.0, galvanicallyisolated

Data transfer

10 Mbit/s

Current consumption

Transmitter• Max. 400 mA(24 V)• Max. 200 mA (110 V, 50/60 Hz; 230 V, 50/60 Hz)

Permitted supply voltage

9 to 30 V

Network connection

With integrated reverse polarity protection

1)

For more information on using the device in the hazardous area, see the Ex-specific Safety Instructions

Current output 4 to 20 mA

18Order code

"Output; input 2" (21), "Output; input 3" (022) or "Output; input 4" (023):
Option B: current output 4 to 20 mA

Signal mode

Can be set to:• Active• Passive

Current span

Can be set to:• 4 to 20 mA NAMUR• 4 to 20 mA US• 4 to 20 mA• 0 to 20 mA (only if the signal mode is active)• Fixed current

Maximum output values

22.5 mA

Open-circuit voltage

DC 28.8 V (active)

Maximum input voltage

DC 30 V (passive)

Load

0 to 700 Ω

Resolution

0.38 µA

Endress+Hauser

Proline Promass Q 500

Damping

Configurable: 0 to 999.9 s

Assignable measured
variables• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Electronics temperature• Oscillation frequency 0• Oscillation damping 0• Signal asymmetry• Exciter current 0
range of options increases if the measuring device has one or more
 The
application packages.

Current output 4 to 20 mA Ex i passive
Order code

"Output; input 2" (21), "Output; input 3" (022):
Option C: current output 4 to 20 mA Ex i passive

Signal mode

Passive

Current span

Can be set to:• 4 to 20 mA NAMUR• 4 to 20 mA US• 4 to 20 mA• Fixed current

Maximum output values

22.5 mA

Maximum input voltage

DC 30 VLoad

0 to 700 Ω

Resolution

0.38 µA

Damping

Configurable: 0 to 999 s

Assignable measured
variables• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Electronics temperature• Oscillation frequency 0• Oscillation damping 0• Signal asymmetry• Exciter current 0
range of options increases if the measuring device has one or more
 The
application packages.

Pulse/frequency/switch output
FunctionCan be configured as pulse, frequency or switch output

VersionOpen collector
Can be set to:• Active• Passive• Passive NAMUR Ex-i, passive

Endress+Hauser

Maximum input values

DC 30 V, 250 mA (passive)

Open-circuit voltage

DC 28.8 V (active)

Voltage drop

For 22.5 mA: ≤ DC 2 V

19

Proline Promass Q 500

Pulse output
Maximum input values

DC 30 V, 250 mA (passive)

Maximum output current

22.5 mA (active)

Open-circuit voltage

DC 28.8 V (active)

Pulse width

Configurable: 0.05 to 2 000 ms

Maximum pulse rate

10 000 Impulse/s

Pulse value

Configurable

Assignable measured
variables• Mass flow
• Volume flow
• Corrected volume flow

Frequency output
Maximum input values

DC 30 V, 250 mA (passive)

Maximum output current

22.5 mA (active)

Open-circuit voltage

DC 28.8 V (active)

Output frequency

Configurable: end value frequency 2 to 10 000 Hz(f max = 12 500 Hz)

Damping

Configurable: 0 to 999.9 s

Pulse/pause ratio

1:1

Assignable measured
variables• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Electronics temperature• Oscillation frequency 0• Oscillation damping 0• Signal asymmetry• Exciter current 0
range of options increases if the measuring device has one or more
 The
application packages.

Switch output
Maximum input values

DC 30 V, 250 mA (passive)

Open-circuit voltage

DC 28.8 V (active)

Switching behavior

Binary, conductive or non-conductive

Switching delay

Configurable: 0 to 100 s

Number of switchingcyclesUnlimited

Assignable functions

• Off
• On
• Diagnostic behavior• Limit value• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Totalizer 1-3• Flow direction monitoring• Status• Partially filled pipe detection• Low flow cut off
range of options increases if the measuring device has one or more
 The
application packages.

20

Endress+Hauser

Proline Promass Q 500

Double pulse output
Function

Double pulse

VersionOpen collector
Can be set to:• Active• Passive• Passive NAMUR

Maximum input values

DC 30 V, 250 mA (passive)

Open-circuit voltage

DC 28.8 V (active)

Voltage drop

For 22.5 mA: ≤ DC 2 V

Output frequency

Configurable: 0 to 1 000 Hz

Damping

Configurable: 0 to 999 s

Pulse/pause ratio

1:1

Assignable measured
variables• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature
range of options increases if the measuring device has one or more
 The
application packages.

Relay output
Function

Switch output

Version

Relay output, galvanically isolated

Switching behavior

Can be set to:• NO (normally open), factory setting• NC (normally closed)

Maximum switchingcapacity (passive)• DC 30 V, 0.1 A• AC 30 V, 0.5 A

Assignable functions

• Off
• On
• Diagnostic behavior• Limit value• Mass flow
• Volume flow
• Corrected volume flow
• Density• Reference density• Temperature• Totalizer 1-3• Flow direction monitoring• Status• Partially filled pipe detection• Low flow cut off
range of options increases if the measuring device has one or more
 The
application packages.

User-configurable input/output
One specific input or output is assigned to a user-configurable input/output (configurable I/O)
during device commissioning.

Endress+Hauser

21

Proline Promass Q 500

The following inputs and outputs are available for assignment:• Choice of current output: 4 to 20 mA (active), 0/4 to 20 mA (passive)• Pulse/frequency/switch output• Choice of current input: 4 to 20 mA (active), 0/4 to 20 mA (passive)• Status input
The technical values correspond to those of the inputs and outputs described in this section.
Signal on alarm

Depending on the interface, failure information is displayed as follows:
HART current output
Device diagnostics

Device condition can be read out via HART Command 48

PROFIBUS PA
Status and alarm
messages

Diagnostics in accordance with PROFIBUS PA Profile 3.02

Failure current FDE (Fault
Disconnection Electronic)

0 mA

PROFIBUS DP
Status and alarm
messages

Diagnostics in accordance with PROFIBUS PA Profile 3.02

EtherNet/IP
Device diagnostics

Device condition can be read out in Input Assembly

PROFINET
Device diagnostics

According to "Application Layer protocol for decentralized periphery", Version 2.3

PROFINET with Ethernet-APL
Device diagnostics

Diagnostics according to PROFINET PA Profile 4

FOUNDATION Fieldbus
Status and alarm
messages

Diagnostics in accordance with FF-891

Failure current FDE (Fault
Disconnection Electronic)

0 mA

Modbus RS485
Failure mode

22

Choose from:• NaN value instead of current value• Last valid value

Endress+Hauser

Proline Promass Q 500

Current output 0/4 to 20 mA
4 to 20 mA
Failure mode

Choose from:• 4 to 20 mA in accordance with NAMUR recommendation NE 43• 4 to 20 mA in accordance with US• Min. value: 3.59 mA• Max. value: 22.5 mA• Freely definable value between: 3.59 to 22.5 mA• Actual value• Last valid value

0 to 20 mA
Failure mode

Choose from:• Maximum alarm: 22 mA• Freely definable value between: 0 to 20.5 mA

Pulse/frequency/switch output
Pulse output
Failure mode

Choose from:• Actual value• No pulses

Frequency output
Failure mode

Choose from:• Actual value• 0 Hz• Defined value (f max 2 to 12 500 Hz)

Switch output
Failure mode

Choose from:• Current status• Open• Closed

Relay output
Failure mode

Choose from:• Current status• Open• Closed

Local display
Plain text display

With information on cause and remedial measures

Backlight

Red backlighting indicates a device error.

Status signal as per NAMUR recommendation NE 107

Endress+Hauser

23

Proline Promass Q 500

Interface/protocol• Via digital communication:• HART protocol• FOUNDATION Fieldbus• PROFIBUS PA• PROFIBUS DP• Modbus RS485• EtherNet/IP• PROFINET• PROFINET with Ethernet-APL• Via service interface• CDI-RJ45 service interface• WLAN interface
Plain text display

With information on cause and remedial measures

Additional information on remote operation →  107
Web browser
Plain text display

With information on cause and remedial measures

Light emitting diodes (LED)
Status information

Status indicated by various light emitting diodes
The following information is displayed depending on the device version:• Supply voltage active• Data transmission active• Device alarm/error has occurred• EtherNet/IP network available• EtherNet/IP connection established• PROFINET network available• PROFINET connection established• PROFINET blinking feature

Load

Output signal →  16

Ex connection data

Safety-related values
Order code for"Output; input 1"

Output type

Safety-related values"Output; input 1"
26 (+)

24

Option BA

Current output
4 to 20 mA HART

UN = 30 VDC
UM = 250 VAC

Option GA

PROFIBUS PA

UN = 32 VDC
UM = 250 VAC

Option LA

PROFIBUS DP

UN = 32 VDC
UM = 250 VAC

Option MA

Modbus RS485

UN = 30 VDC
UM = 250 VAC

Option SA

FOUNDATION Fieldbus

UN = 32 VDC
UM = 250 VAC

Option NA

EtherNet/IP

UN = 30 VDC
UM = 250 VAC

27 (–)

Endress+Hauser

Proline Promass Q 500

Order code for"Output; input 1"

Output type

Safety-related values"Output; input 1"
26 (+)

Option RA

PROFINET

Option RB

UN = 30 VDC
UM = 250 VAC

PROFINET with Ethernet- APL port profile SLAXAPLSPE PoDL classes 10, 11, 12
UN = 30 VDC
UM = 250 VAC

Order code for"Output; input 2";"Output; input 3""Output; input 4"

Output type

Option B

1)

27 (–)

Current output
4 to 20 mA

Safety-related values
Output; input 2

Output; input 3

Output; input
4 1)

24 (+)

22 (+)

20 (+)

25 (–)

23 (–)

21 (–)UN = 30 VDC
UM = 250 VAC

Option D

User-configurable input/ UN = 30 VDC
outputUM = 250 VAC

Option E

Pulse/frequency/switch
outputUN = 30 VDC
UM = 250 VAC

Option F

Double pulse output

UN = 30 VDC
UM = 250 VAC

Option H

Relay output

UN = 30 VDC
IN =100 mADC/500 mAAC
UM = 250 VAC

Option I

Current input
4 to 20 mA

UN = 30 VDC
UM = 250 VAC

Option J

Status input

UN = 30 VDC
UM = 250 VAC

The order code "Output; input 4" is only available for the Proline 500 – digital transmitter.

Intrinsically safe values
Order code
"Output; input 1"

Output type

Intrinsically safe values"Output; input 1"
26 (+)

Option CA

Current output
4 to 20 mA HART Ex i
passiveUi = 30 V
li = 100 mA
Pi = 1.25 W
Li = 0 µH
Ci = 6 nF

Option CC

Current output
4 to 20 mA HART Ex i
activeEx ia 1)U0 = 21.8 V
l0 = 90 mA
P0 = 491 mW
L0 = 4.1 mH (IIC)/15 mH(IIB)
C0 = 160 nF (IIC)/
1 160 nF (IIB)

27 (–)

Ex ic 2)U0 = 21.8 V
l0 = 90 mA
P0 = 491 mW
L0 = 9 mH (IIC)/39 mH(IIB)
C0 = 600 nF (IIC)/
4 000 nF (IIB)

Ui = 30 V
li = 10 mA
Pi = 0.3 W
Li = 5 µH
Ci = 6 nF

Endress+Hauser

25

Proline Promass Q 500

Order code
"Output; input 1"

Output type

Intrinsically safe values"Output; input 1"
26 (+)
1)

27 (–)
2)

Option HA

PROFIBUS PA Ex i
(FISCO Field Device)

Ex ia
Ui = 30 V
li = 570 mA
Pi = 8.5 W
Li = 10 µH
Ci = 5 nF

Ex ic
Ui = 32 V
li = 570 mA
Pi = 8.5 W
Li = 10 µH
Ci = 5 nF

Option TA

FOUNDATION Fieldbus
Ex i

Ex ia 1)
Ui = 30 V
li = 570 mA
Pi = 8.5 W
Li = 10 µH
Ci = 5 nF

Ex ic 2)
Ui = 32 V
li = 570 mA
Pi = 8.5 W
Li = 10 µH
Ci = 5 nF

Option RC

PROFINET with EthernetAPL Ex i

Ex ia 1)
2-WISE power load APL
port profile SLAA

Ex ic 2)
2-WISE power load APL
port profile SLAC

1)
2)

Only available for Proline 500 transmitter Zone 1; Class I, Division 1.
Only available for transmitter Zone 2; Class I, Division 2 and only for Proline 500 – digital transmitter

Order code for"Output; input 2";"Output; input 3";"Output; input 4"

Output type

Intrinsically safe values or NIFW values
Output; input 2

Output; input 3

Output; input 4

24 (+)

22 (+)

20 (+)

25 (–)

Option C

Current output
4 to 20 mA Ex i passive

Ui = 30 V
li = 100 mA
Pi = 1.25 W
Li = 0Ci = 0

Option G

Pulse/frequency/switch
output Ex i passive

Ui = 30 V
li = 100 mA
Pi = 1.25 W
Li = 0Ci = 01)

23 (–)

1)

21 (–)The order code "Output; input 4" is only available for the Proline 500 – digital transmitter.

Low flow cut off

The switch points for low flow cut off are user-selectable.

Galvanic isolation

The outputs are galvanically isolated:• from the power supply• from one another• from the potential equalization (PE) terminal

Protocol-specific data

HART
Manufacturer ID

0x11

Device type ID

0x3B

HART protocol revision7

Device description files(DTM, DD)

Information and files under:
www.endress.com

HART load

Min. 250 Ω

System integration

Information on system integration: Operating Instructions →  130.
• Measured variables via HART protocol• Burst Mode functionality

26

Endress+Hauser

Proline Promass Q 500

FOUNDATION Fieldbus
Manufacturer ID

0x452B48 (hex)

Ident number

0x103B (hex)

Device revision

1

DD revision

Information and files under:• www.endress.com• www.fieldcommgroup.org

CFF revisionInteroperability Test Kit (ITK) Version 6.2.0
ITK Test Campaign Number

Information:• www.endress.com• www.fieldcommgroup.org

Link Master capability (LAS)

Yes

Choice of "Link Master" and"Basic Device"

Yes
Factory setting: Basic Device

Node address

Factory setting: 247 (0xF7)

Supported functions

The following methods are supported:• Restart• ENP Restart• Diagnostic• Set to OOS• Set to AUTO
• Read trend data
• Read event logbook

Virtual Communication Relationships (VCRs)
Number of VCRs

44

Number of link objects in VFD 50
Permanent entries

1

Client VCRs

0

Server VCRs

10

Source VCRs

43

Sink VCRs

0

Subscriber VCRs

43

Publisher VCRs

43

Device Link Capabilities
Slot time4

Min. delay between PDU

8Max. response delay

16

System integration

Information regarding system integration: Operating Instructions →  130.
• Cyclic data transmission• Description of the modules• Execution times• Methods

PROFIBUS DP

Endress+Hauser

Manufacturer ID

0x11

Ident number

0x156F

Profile version

3.02

27

Proline Promass Q 500

Device description files (GSD,
DTM, DD)

Information and files under:• https://www.endress.com/download
On the device product page: PRODUCTS → Product Finder → Links• https://www.profibus.com

Supported functions• Identification & Maintenance
Simplest device identification on the part of the control system andnameplate• PROFIBUS upload/download
Reading and writing parameters is up to ten times faster with PROFIBUS
upload/download• Condensed status
Simplest and self-explanatory diagnostic information by categorizing
diagnostic messages that occur

Configuration of the device
address• DIP switches on the I/O electronics module• Via operating tools (e.g. FieldCare)

Compatibility withearlier model

If the device is replaced, the measuring device Promass 500 supports thecompatibility of the cyclic data with previous models. It is not necessary toadjust the engineering parameters of the PROFIBUS network with thePromass 500 GSD file.
Previous model:
Promass 83 PROFIBUS DP• ID No.: 1529 (hex)• Extended GSD file: EH3x1529.gsd• Standard GSD file: EH3_1529.gsd
of the function scope of compatibility: Description
Operating Instructions →  130.

System integration

Information regarding system integration: Operating Instructions →  130.
• Cyclic data transmission• Block model• Description of the modules

PROFIBUS PA

28

Manufacturer ID

0x11

Ident number

0x156D

Profile version

3.02

Device description files (GSD,
DTM, DD)

Information and files under:• https://www.endress.com/download
On the device product page: PRODUCTS → Product Finder → Links• https://www.profibus.com

Supported functions• Identification & Maintenance
Simplest device identification on the part of the control system andnameplate• PROFIBUS upload/download
Reading and writing parameters is up to ten times faster with PROFIBUS
upload/download• Condensed status
Simplest and self-explanatory diagnostic information by categorizing
diagnostic messages that occur

Configuration of the device
address• DIP switches on the I/O electronics module• Local display• Via operating tools (e.g. FieldCare)

Endress+Hauser

Proline Promass Q 500

Compatibility withearlier model

If the device is replaced, the measuring device Promass 500 supports thecompatibility of the cyclic data with previous models. It is not necessary toadjust the engineering parameters of the PROFIBUS network with thePromass 500 GSD file.
Earlier models:• Promass 80 PROFIBUS PA• ID No.: 1528 (hex)• Extended GSD file: EH3x1528.gsd• Standard GSD file: EH3_1528.gsd• Promass 83 PROFIBUS PA• ID No.: 152A (hex)• Extended GSD file: EH3x152A.gsd• Standard GSD file: EH3_152A.gsd
of the function scope of compatibility: Description
Operating Instructions →  130.

System integration

Information regarding system integration: Operating Instructions →  130.
• Cyclic data transmission• Block model• Description of the modules

Modbus RS485
ProtocolModbus Applications Protocol Specification V1.1

Response times• Direct data access: typically 25 to 50 ms• Auto-scan buffer (data range): typically 3 to 5 ms

Device type

Slave

Slave address range

1 to 247

Broadcast address range0

Function codes• 03: Read holding register• 04: Read input register• 06: Write single registers• 08: Diagnostics• 16: Write multiple registers• 23: Read/write multiple registers

Broadcast messages

Supported by the following function codes:• 06: Write single registers• 16: Write multiple registers• 23: Read/write multiple registers

Supported baud rate

• 1 200 BAUD
• 2 400 BAUD
• 4 800 BAUD
• 9 600 BAUD
• 19 200 BAUD
• 38 400 BAUD
• 57 600 BAUD
• 115 200 BAUD

Data transfer mode

• ASCII• RTU

Data access

Each device parameter can be accessed via Modbus RS485.

 For Modbus register information

Endress+Hauser

29

Proline Promass Q 500

Compatibility withearlier model

If the device is replaced, the measuring device Promass 500 supports thecompatibility of the Modbus registers for the process variables and the
diagnostic information with the previous model Promass 83. It is notnecessary to change the engineering parameters in the automation system.
of the function scope of compatibility: Description
Operating Instructions →  130.

System integration

Information on system integration: Operating Instructions →  130.
• Modbus RS485 information• Function codes• Register information• Response time
• Modbus data map

EtherNet/IP
Protocol• The CIP Networks Library Volume 1: Common Industrial Protocol• The CIP Networks Library Volume 2: EtherNet/IP Adaptation of CIP

Communication type

• 10Base-T• 100Base-TX

Device profile

Generic device (product type: 0x2B)

Manufacturer ID

0x000049E

Device type ID

0x103B

Baud rates

Automatic ¹⁰⁄₁₀₀ Mbit with half-duplex and full-duplex detection

Polarity

Auto-polarity for automatic correction of crossed TxD and RxD pairs

Supported CIP connections

Max. 3 connections

Explicit connections

Max. 6 connections

I/O connections

Max. 6 connections (scanner)

Configuration options formeasuring device• DIP switches on the electronics module for IP addressing• Manufacturer-specific software (FieldCare)• Add-on Profile Level 3 for Rockwell Automation control systems• Web browser• Electronic Data Sheet (EDS) integrated in the measuring device

Configuration of the EtherNet
interface• Speed: 10 MBit, 100 MBit, auto (factory setting)• Duplex: half-duplex, full-duplex, auto (factory setting)Configuration of the device
address• DIP switches on the electronics module for IP addressing (last octet)• DHCP• Manufacturer-specific software (FieldCare)• Add-on Profile Level 3 for Rockwell Automation control systems• Web browser• EtherNet/IP tools, e.g. RSLinx (Rockwell Automation)

Device Level Ring (DLR)

Yes

System integration

Information regarding system integration: Operating Instructions→  130.
• Cyclic data transmission• Block model• Input and output groups

PROFINET30

ProtocolApplication layer protocol for decentral device periphery and distributed
automation, Version 2.3

Communication type

100 MBit/s

Conformance Class

Conformance Class B

Netload Class

Netload Class 2 0 Mbps

Endress+Hauser

Proline Promass Q 500

Baud rates

Automatic 100 Mbit/s with full-duplex detection

Cycle times

From 8 ms

Polarity

Auto-polarity for automatic correction of crossed TxD and RxD pairs

Media Redundancy Protocol
(MRP)

Yes

System redundancy support

System redundancy S2 (2 AR with 1 NAP)

Device profile

Application interface identifier 0xF600
Generic device

Manufacturer ID

0x11

Device type ID

0x843B

Device description files (GSD,
DTM, DD)

Information and files under:• www.endress.com
On the product page for the device: Documents/Software → Device drivers• www.profibus.com

Supported connections• 2 x AR (IO Controller AR)• 1 x AR (IO-Supervisor Device AR connection allowed)• 1 x Input CR (Communication Relation)• 1 x Output CR (Communication Relation)• 1 x Alarm CR (Communication Relation)

Configuration options formeasuring device• DIP switches on the electronics module, for device name assignment (lastpart)• Asset management software (FieldCare, DeviceCare, Field Xpert)• Integrated Web server via Web browser and IP address• Device master file (GSD), can be read out via the integrated Web server ofthe measuring device.
• Onsite operation

Configuration of thedevice name

• DIP switches on the electronics module, for device name assignment (lastpart)• DCP protocol• Asset management software (FieldCare, DeviceCare, Field Xpert)• Integrated Web server

Supported functions• Identification & Maintenance, simple device identifier via:
• Control system• Nameplate• Measured value status
The process variables are communicated with a measured value status• Blinking feature via the local display for simple device identification andassignment• Device operation via asset management software (e.g. FieldCare,
DeviceCare, SIMATIC PDM)

System integration

Information regarding system integration: Operating Instructions →  130.
• Cyclic data transmission• Overview and description of the modules• Status coding• Startup configuration• Factory setting

PROFINET with Ethernet-APL

Endress+Hauser

ProtocolApplication layer protocol for decentral device periphery and distributed
automation, Version 2.4

Communication type

Ethernet Advanced Physical Layer 10BASE-T1L

Conformance Class

Conformance Class B (PA)

Netload Class

Netload Class 2 0 Mbps

Baud rates

10 Mbit/s Full-duplex

Cycle times

64 ms

31

Proline Promass Q 500

Polarity

Automatic correction of crossed "APL signal +" and "APL signal -" signal lines

Media Redundancy Protocol(MRP)

Not possible (point-to-point connection to APL field switch)

System redundancy support

System redundancy S2 (2 AR with 1 NAP)

Device profile

PROFINET PA profile 4 (Application interface identifier API: 0x9700)

Manufacturer ID

0x11

Device type ID

0xA43B

Device description files (GSD,
DTM, FDI)

Information and files under:• www.endress.com/download
On the device product page: PRODUCTS → Product Finder → Links• www.profibus.com

Supported connections• 2x AR (IO Controller AR)• 2x AR (IO Supervisor Device AR connection allowed)

Configuration options formeasuring device• DIP switches on the electronics module, for device name assignment (lastpart)• Asset management software (FieldCare, DeviceCare, Field Xpert)• Integrated Web server via Web browser and IP address• Device master file (GSD), can be read out via the integrated Web server ofthe measuring device.
• Onsite operation

Configuration of thedevice name

• DIP switches on the electronics module, for device name assignment (lastpart)• DCP protocol• Asset management software (FieldCare, DeviceCare, Field Xpert)• Integrated Web server

Supported functions• Identification & Maintenance, simple device identifier via:
• Control system• Nameplate• Measured value status
The process variables are communicated with a measured value status• Blinking feature via the local display for simple device identification andassignment• Device operation via asset management software (e.g. FieldCare,
DeviceCare, SIMATIC PDM with FDI package)

System integration

Information regarding system integration: Operating Instructions →  130.
• Cyclic data transmission• Overview and description of the modules• Status coding• Startup configuration• Factory setting

Power supply
Terminal assignment

Transmitter: supply voltage, input/outputs
HART
Supply voltage
1 (+)

2 (–)

Input/output
1
26 (+)

27 (–)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version ordered →  14.

32

Endress+Hauser

Proline Promass Q 500

FOUNDATION Fieldbus
Supply voltage
1 (+)

2 (–)

Input/output
1
26 (A)

27 (B)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version ordered →  14.

PROFIBUS DP
Supply voltage
1 (+)

2 (–)

Input/output
1
26 (B)

27 (A)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version ordered →  14.

PROFIBUS PA
Supply voltage
1 (+)

2 (–)

Input/output
1
26 (B)

27 (A)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version ordered →  14.

Modbus RS485
Supply voltage
1 (+)

2 (–)

Input/output
1
26 (B)

27 (A)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version ordered →  14.

EtherNet/IP
Supply voltage

Input/output
1

1 (+)

EtherNet/IP(RJ45 connector)

2 (–)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version
ordered →  14.

PROFINET
Supply voltage

Input/output
1

1 (+)

PROFINET(RJ45 connector)

2 (–)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version
ordered →  14.

PROFINET with Ethernet-APL

Endress+Hauser

Supply voltage

Input/output
1

1 (+)

EtherNet/IP(RJ45 connector)

2 (–)

Input/output
2
24 (+)

25 (–)

Input/output
3
22 (+)

23 (–)

Input/output
4
20 (+)

21 (–)The terminal assignment depends on the specific device version
ordered →  14.

33

Proline Promass Q 500

Transmitter and sensor connection housing: connecting cable
The sensor and transmitter, which are mounted in separate locations, are interconnected by aconnecting cable. The cable is connected via the sensor connection housing and the transmitter
housing.
Terminal assignment and connection of the connecting cable:• Proline 500 – digital→  37
• Proline 500 →  38
Device plugs may not be used in hazardous areas!

Available device plugs

Device plugs for fieldbus systems:
Order code for "Input; output 1"• Option SA "FOUNDATION Fieldbus" →  34
• Option GA "PROFIBUS PA" →  34
• Option NA "EtherNet/IP" →  34
• Option RA "PROFINET" →  35
• Option RB "PROFINET with Ethernet-APL" →  35
Device plug for connecting to the service interface:
Order code for "Accessory mounted"
Option NB, adapter RJ45 M12 (service interface) →  36
Order code for "Input; output 1", option SA "FOUNDATION Fieldbus"
Order code forCable entry/connection →  39

"Electrical connection"

2

3

M, 3, 4, 5

7/8" connector–

Order code for "Input; output 1", option GA "PROFIBUS PA"
Order code forCable entry/connection →  39

"Electrical connection"

2

3

L, N, P, U

Connector M12 × 1–

Order code for "Input; output 1", option NA "EtherNet/IP"
Order code for"Electrical connection"

2

3

L, N, P, U

Connector M12 × 1–

Connector M12 × 1

Connector M12 × 1

R
1)

2)

34

Cable entry/connection →  39

1) 2)

,S

1) 2)

,T

1) 2)

,V

1) 2)

Cannot be combined with an external WLAN antenna (order code for "Enclosed accessories", option P8) of
an RJ45 M12 adapter for the service interface (order code for "Accessories mounted", option NB) or of theremote display and operating module DKX001
Suitable for integrating the device in a ring topology.

Endress+Hauser

Proline Promass Q 500

Order code for "Input; output 1", option RA "PROFINET"
Order code for"Electrical connection"

2

3

L, N, P, U

Connector M12 × 1–

Connector M12 × 1

Connector M12 × 1

R
1)

2)

Cable entry/connection →  39

1) 2)

,S

1) 2)

,T

1) 2)

,V

1) 2)

Cannot be combined with an external WLAN antenna (order code for "Enclosed accessories", option P8) of
an RJ45 M12 adapter for the service interface (order code for "Accessories mounted", option NB) or of theremote display and operating module DKX001.
Suitable for integrating the device in a ring topology.

Order code for "Input; output 1", option RB "PROFINET with Ethernet-APL"
Order code

Cable entry/connection →  39

"Electrical connection"

2

3

L, N, P, U

M12 plug × 1–

Order code for "Accessory mounted", option NB "Adapter RJ45 M12 (service interface)"
Order code

Pin assignment, device plug

Cable entry/coupling →  39

"Accessory mounted"

Cable entry
2

Cable entry
3

NB

Plug M12 × 1–

FOUNDATION Fieldbus
Pin

Assignment

2

3

1

+

Signal +

1

4

2

-

Signal –3

Grounding4

Not assigned

CodingPlug/socket

A

Plug

CodingPlug/socket

A

Plug

PROFIBUS PA
Pin

2

3

1

1

4

2
3
4

Assignment+

PROFIBUS PA +Grounding-

PROFIBUS PA –Not assigned

plug: Recommended• Binder, series 713, part no. 99 1430 814 04• Phoenix, part no. 1413934 SACC-FS-4QO SH PBPA SCO

Endress+Hauser

35

Proline Promass Q 500

PROFINET
Pin

2

3

1

4
A0032047

Assignment

1

+

TD +

2

+

RD +

3

-

TD –

4

-

RD –

CodingPlug/socket

D

Socketplug:
 Recommended• Binder, series 825, part no. 99 3729 810 04• Phoenix, part no. 1543223 SACC-M12MSD-4Q

PROFINET with Ethernet-APL
PinAssignment

3

4

1

-

APL signal -

2

1

2

+

APL signal +3

Cable shield1

4

Not assigned

Metal plughousing

Cable shield

CodingPlug/socket

A

Socket1If a cable shield is usedplug:
 Recommended• Binder, series 713, part no. 99 1430 814 04• Phoenix, part no. 1413934 SACC-FS-4QO SH PBPA SCO

EtherNet/IP
Pin

2

3

1

4
A0032047

Assignment

1

+

Tx

2

+

Rx

3

-

Tx

4

-

Rx

CodingPlug/socket

D

Socketplug:
 Recommended• Binder, series 763, part no. 99 3729 810 04• Phoenix, part no. 1543223 SACC-M12MSD-4Q

Service interface
Order code for "Accessories mounted", option NB: Adapter RJ45 M12 (service interface)
Pin

2

3

1

Assignment

1

+

Tx

2

+

Rx

3

-

Tx

4

-

Rx

4
A0032047

36

Endress+Hauser

Proline Promass Q 500

CodingPlug/socket

D

Socketplug:
 Recommended• Binder, series 763, part no. 99 3729 810 04• Phoenix, part no. 1543223 SACC-M12MSD-4Q

Supply voltage

Order code for"Power supply"

Terminal voltage

Option D

DC 24 V

Option E

AC 100 to 240 V –15 to +10%

50/60 HzDC 24 V–

Option I

Power consumption

Frequency range
±20%

–

±20%

AC 100 to 240 V –15 to +10%

50/60 Hz

TransmitterMax. 10 W (active power)switch-on current

Current consumption

Max. 36 A (<5 ms) as per NAMUR Recommendation NE 21

Transmitter• Max. 400 mA (24 V)• Max. 200 mA (110 V, 50/60 Hz; 230 V, 50/60 Hz)

Power supply failure• Totalizers stop at the last value measured.
• Depending on the device version, the configuration is retained in the device memoryor in thepluggable data memory (HistoROM DAT).
• Error messages (incl. total operated hours) are stored.

Overcurrent protection
element

The device must be operated with a dedicated circuit breaker, as it does not have an ON/OFF switchof its own.
• The circuit breaker must be easy to reach and labeled accordingly.
• Permitted nominal current of the circuit breaker: 2 A up to maximum 10 A.

Electrical connection

Connection of connecting cable: Proline 500 – digital
1
2

3
5
6

+ – B A61 62 63 64

4

62 61 64 63

4

–

+

A

B

A0028198

1
2
3
4
5
6

Endress+Hauser

Cable entry for cable on transmitter housing
Terminal connection for potential equalization (PE)Connecting cable ISEM communication
Grounding via ground connection; in the version with a device plug, grounding is ensured through the plugitselfCable entry for cable or connection of device plug on sensor connection housing
Terminal connection for potential equalization (PE)37

Proline Promass Q 500

Depending on the device version of the sensor connection housing, the connecting cable is connected
via terminals or device plugs.
Sensor connection housing
Order code for "Housing"Connection onconnection housing
via

Connection ontransmitter housing
via

Option A: aluminum coatedTerminals

Terminals

Option B: stainless

Terminals

Terminals

Option C: ultra-compact, hygienic, stainlessDevice plug

Terminals

Option L: cast, stainless

Terminals

Terminals

Pin assignment, device plug
Device plugs are only available for device version, order code for "Housing":
Option C: ultra-compact, hygienic, stainless
For connection to sensor connection housing.
2

1

3

5
4

1)

Pin

Color 1)

1

Brown

+

2

White

A

3

Blue

B

4

Black

–

5

–

Assignment

Connection
to terminal

Supply voltage
ISEM communication

61
64
63

Supply voltage

62

–

–

CodingPlug/socket

A

Plug

Cable colors of connecting cable

A connecting cable with a device plug is optionally available.
Connection of the connecting cable: Proline 500
The number of connecting cables depends on the nominal diameter of the device. For devices
with nominal diameters DN ≥150 mm (6 in), an additional, second connecting cable is required.
First connecting cable for all nominal diameters

38

Endress+Hauser

Proline Promass Q 500

TM2 TT TT TM TM GND S2 S2 S1 S1
+
+
+
+
TM2 +
7 6 5 4
X4 X3 12 11 10 9 8

ER ER
+
42 41

7 6 5 4
X4 X3 12 11 10 9 8
TM2 +
+
+
+
+
TM2 TT TT TM TM GND S2 S2 S1 S1

42 41
+
ER ER

1
2
3
4
5

A0046886

1
2
3
4
5

Terminal connection for potential equalization (PE)
Cable entry for connecting cable on transmitter connection housing
Connecting cable
Cable entry for connecting cable on sensor connection housing
Terminal connection for potential equalization (PE)

Additional, second connecting cable for nominal diameters DN ≥150 mm (6 in)
S22 S22 S21 S21 E2 E2+++
X8 X7 X6 X5 X1 X2
12
3

X8 X7 X6 X5 X1 X2+++
S22 S22 S21 S21 E2 E2
A0046894

1
2
3

Cable entry for second connecting cable on transmitter connection housing
Second connecting cable
Cable entry for second connecting cable on sensor connection housing

Transmitter connection• Terminal assignment →  32
• Device plug pin assignment →  35

Endress+Hauser

39

Proline Promass Q 500

Transmitter connection: Proline 500 – digital

1

2 3 6 4

5
A0028200

1
2
3
4
5
6

Terminal connection for supply voltage
Terminal connection for signal transmission, input/output
Terminal connection for signal transmission, input/output
Terminal connection for connecting cable between sensor and transmitter
Terminal connection for signal transmission, input/output or terminal for network connection (DHCP client)
via service interface (CDI-RJ45); optional: terminal connection for external WLAN antenna
Terminal connection for potential equalization (PE)An adapter for RJ45 to the M12 plug is optionally available:
Order code for "Accessories", option NB: "Adapter RJ45 M12 (service interface)"
The adapter connects the service interface (CDI-RJ45) to an M12 plug mounted in the cableentry. Therefore the connection to the service interface can be established via an M12 plug
without opening the device.
Network connection (DHCP client) via service interface (CDI-RJ45) →  113
Connecting the transmitter: Proline 500

1
2
3
4

A0026781

1
2
3
4

Terminal connection for supply voltage
Terminal connection for signal transmission, input/output
Terminal connection for signal transmission, input/output or terminal for network connection (DHCP client)
via service interface (CDI-RJ45); optional: terminal connection for external WLAN antenna
Terminal connection for potential equalization (PE)An adapter for RJ45 to the M12 plug is optionally available:
Order code for "Accessories", option NB: "Adapter RJ45 M12 (service interface)"
The adapter connects the service interface (CDI-RJ45) to an M12 plug mounted in the cableentry. Therefore the connection to the service interface can be established via an M12 plug
without opening the device.
Network connection (DHCP client) via service interface (CDI-RJ45) →  113

40

Endress+Hauser

Proline Promass Q 500

Connecting in a ring topology
Device versions with EtherNet/IP and PROFINET communication protocols can be integrated into aring topology. The device is integrated via the terminal connection for signal transmission (output 1)
and the connection to the service interface (CDI-RJ45).
Transmitters with an Ex de approval may not be connected via the service interface (CDI-RJ45)!
Order code for "Approval transmitter + sensor", options (Ex de):BA, BB, C1, C2, GA, GB, MA, MB, NA, NB
Integrate the transmitter in a ring topology:• EtherNet/IP• PROFINET
Transmitter: Proline 500 – digital

1

2 3 6 4

5
A0028200

1
2
2
4
5
6

Terminal connection for supply voltage
Terminal connection for signal transmission, input/output
Terminal connection for signal transmission: PROFINET or EtherNet/IP (RJ45 plug)
Terminal connection for connecting cable between sensor and transmitter
Terminal connection to service interface (CDI-RJ45)
Terminal connection for potential equalization (PE)Transmitter: Proline 500

1
2
3
4

A0026781

1
2
3
4

Terminal connection for supply voltage
Terminal connection for signal transmission: PROFINET or EtherNet/IP (RJ45 plug)
Terminal connection to service interface (CDI-RJ45)
Terminal connection for potential equalization (PE)If the device has additional inputs/outputs, these are routed in parallel via the cable entry forconnection to the service interface.

Endress+Hauser

41

Proline Promass Q 500

Connection examples
Current output 4 to 20 mA HART
1

2

3
4...20 mA
6

4

5
A0029055

2
1
2
3
4
5
6

Connection example for 4 to 20 mA HART current output (active)
Automation system with current input (e.g. PLC)
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications →  51
Connection for HART operating devices →  107
Resistor for HART communication (≥ 250 Ω): observe maximum load →  16
Analog display unit: observe maximum load →  16
Transmitter

1

2

3

4

5
4...20 mA

A0028762

3
1
2
3
4
5

42

Connection example for 4 to 20 mA HART current output (passive)
Automation system with current input (e.g. PLC)
Power supply
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications →  51
Analog display unit: observe maximum load →  16
Transmitter

Endress+Hauser

Proline Promass Q 500

HART input
1

2

4

3

6
4...20 mA

5

2

3
A0028763

4
1
2
3
4
5
6

Connection example for HART input with a common negative (passive)
Automation system with HART output (e.g. PLC)
Active barrier for power supply (e.g. RN221N)
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
Analog display unit: observe maximum load →  16
Pressure transmitter (e.g. Cerabar M, Cerabar S): see requirements
Transmitter

PROFIBUS PA
1

2

3

4

5
6
6

6
5
6
6

6
8

7
A0028768

5
1
2
3
4
5
6
7
8

Endress+Hauser

Connection example for PROFIBUS PA
Control system (e.g. PLC)
PROFIBUS PA segment coupler
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
T-box
Measuring device
Local grounding
Bus terminator
Potential matching line43

Proline Promass Q 500

PROFIBUS DP
1

3

2

A

A

4
B

B

A
4
B

A0028765

6
1
2
3
4

Connection example for PROFIBUS DP, non-hazardous area and Zone 2/Div. 2
Control system (e.g. PLC)
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
Distribution box
Transmitter

If baud rates > 1.5 MBaud an EMC cable entry must be used and the cable shield must continue
as far as the terminal wherever possible.
EtherNet/IP
1

2

3

4

5

5
A0028767

7
1
2
3
4
5

44

Connection example for EtherNet/IP
Control system (e.g. PLC)
Ethernet switch
Observe cable specifications
Device plug
Transmitter

Endress+Hauser

Proline Promass Q 500

EtherNet/IP: DLR (Device Level Ring)
1

3

2

4

5

4

A0027544

1
2
3
4
5

Control system (e.g. PLC)
Ethernet switch
Observe cable specifications →  51
Transmitter
Connecting cable between the two transmitters

PROFINET
1

2

3

4

5

5
A0028767

8
1
2
3
4
5

Endress+Hauser

Connection example for PROFINET
Control system (e.g. PLC)
Ethernet switch
Observe cable specifications
Device plug
Transmitter45

Proline Promass Q 500

PROFINET with Ethernet-APL
6

1

2

A
B
S

A
B

3
5
A
B
S

A
B

4
A0047536

9
1
2
3
4
5
6

Connection example for PROFINET with Ethernet-APL
Cable shield
Measuring device
Local grounding
Potential equalization
Trunk or TCP
Field switch

PROFINET: MRP (Media Redundancy Protocol)
1

2

3

4

5

4

A0027544

1
2
3
4
5

46

Control system (e.g. PLC)
Ethernet switch
Observe cable specifications →  51
Transmitter
Connecting cable between the two transmitters

Endress+Hauser

Proline Promass Q 500

PROFINET: system redundancy S2
1

2

4

3
5

5

5

A0039553

 10
1
2
3
4
5

Connection example for system redundancy S2Control system 1 (e.g. PLC)
Synchronization of control systems
Control system 2 (e.g. PLC)
Industrial Ethernet Managed Switch
Transmitter

FOUNDATION Fieldbus
1

2

3

4

5
6
6

6
5
6
6

6
8

7
A0028768

 11
1
2
3
4
5
6
7
8

Endress+Hauser

Connection example for FOUNDATION Fieldbus

Control system (e.g. PLC)
Power Conditioner (FOUNDATION Fieldbus)
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
T-box
Measuring device
Local grounding
Bus terminator
Potential matching line

47

Proline Promass Q 500

Modbus RS485
1

3

2

A

A

4
B

B

A
4
B

A0028765

 12
1
2
3
4

Connection example for Modbus RS485, non-hazardous area and Zone 2; Class I, Division 2

Control system (e.g. PLC)
Cable shield provided at one end. The cable shield must be grounded at both ends to comply with EMC
requirements; observe cable specifications
Distribution box
Transmitter

Current output 4-20 mA
1

2

3
4...20 mA
A0028758

 13
1
2
3

Connection example for 4-20 mA current output (active)

Automation system with current input (e.g. PLC)
Analog display unit: observe maximum load →  16
Transmitter

1

2

3

4
4...20 mA

A0028759

 14
1
2
3
4

48

Connection example for 4-20 mA current output (passive)

Automation system with current input (e.g. PLC)
Active barrier for power supply (e.g. RN221N)
Analog display unit: observe maximum load →  16
Transmitter

Endress+Hauser

Proline Promass Q 500

Pulse/frequency output

1

2

3

12345
A0028761

 15
1
2
3

Connection example for pulse/frequency output (passive)

Automation system with pulse/frequency input (e.g. PLC with 10 kΩ pull-up or pull-down resistor)
Power supply
Transmitter: observe input values →  19

Switch output

1

2

3

A0028760

 16
1
2
3

Connection example for switch output (passive)

Automation system with switch input (e.g. PLC with a 10 kΩ pull-up or pull-down resistor)
Power supply
Transmitter: observe input values →  19

Double pulse output
1

3

2

4
A0029280

 17
1
2
3
4

Endress+Hauser

Connection example for double pulse output (active)

Automation system with double pulse input (e.g. PLC)
Transmitter: observe input values →  21
Double pulse output
Double pulse output (slave), phase-shifted49

Proline Promass Q 500

2
1

4

3

5
A0029279

 18
1
2
3
4
5

Connection example for double pulse output (passive)

Automation system with double pulse input (e.g. PLC with a 10 kΩ pull-up or pull-down resistor)
Power supply
Transmitter: observe input values →  21
Double pulse output
Double pulse output (slave), phase-shifted

Relay output

1

2

3

A0028760

 19
1
2
3

Connection example for relay output (passive)

Automation system with relay input (e.g. PLC)
Power supply
Transmitter: observe input values →  21

Current input
1

2

3

4

A0028915

 20
1
2
3
4

50

Connection example for 4 to 20 mA current input

Power supply
Terminal box
External measuring device (to read in pressure or temperature, for instance)
Transmitter

Endress+Hauser

Proline Promass Q 500

Status input

1

2

3

A0028764

 21
1
2
3

Potential equalization

Connection example for status input

Automation system with status output (e.g. PLC)
Power supply
Transmitter

Requirements
For potential equalization:• Pay attention to in-house grounding concepts• Take account of operating conditions like the pipe material and grounding• Connect the medium, sensor and transmitter to the same electrical potential• Use a ground cable with a minimum cross-section of 6 mm2 (0.0093 in2) and a cable lug forpotential equalization connections
For devices intended for use in hazardous locations, please observe the guidelines in the
Ex documentation (XA).

Terminals

Spring-loaded terminals: Suitable for strands and strands with ferrules.
Conductor cross-section 0.2 to 2.5 mm2 (24 to 12 AWG).

Cable entries• Cable gland: M20 × 1.5 with cable ⌀ 6 to 12 mm (0.24 to 0.47 in)• Thread for cable entry:• NPT ½"
• G ½"• M20• Device plug for digital communication: M12
Only available for certain device versions →  34.
• Device plug for connecting cable: M12
A device plug is always used for the device version with the order code for "Sensor connection
housing", option C "Ultra-compact, hygienic, stainless".

Cable specification

Permitted temperature range• The installation guidelines that apply in the country of installation must be observed.
• The cables must be suitable for the minimum and maximum temperatures to be expected.
Power supply cable (incl. conductor for the inner ground terminal)
Standard installation cable is sufficient.
Protective grounding cable for the outer ground terminal
Conductor cross-section < 2.1 mm2 (14 AWG)
The use of a cable lug enables the connection of larger cross-sections.
The grounding impedance must be less than 2 Ω.
Signal cable
Current output 4 to 20 mA HARTA shielded cable is recommended. Observe grounding concept of the plant.

Endress+Hauser

51

Proline Promass Q 500

PROFIBUS PA
Twisted, shielded two-wire cable. Cable type A is recommended .
For further information on planning and installing PROFIBUS networks see:
• Operating Instructions "PROFIBUS DP/PA: Guidelines for planning and commissioning"(BA00034S)• PNO Directive 2.092 "PROFIBUS PA User and Installation Guideline"• IEC 61158-2 (MBP)PROFIBUS DP
The IEC 61158 standard specifies two types of cable (A and B) for the bus line which can be used forevery transmission rate. Cable type A is recommended.
Cable typeA

Characteristic impedance

135 to 165 Ω at a measuring frequency of 3 to 20 MHz

Cable capacitance< 30 pF/mWire cross-section> 0.34 mm2 (22 AWG)

Cable type

Twisted pairs

Loop resistance≤110 Ω/km

Signal damping

Max. 9 dB over the entire length of the cable cross-section

ShieldCopper braided shielding or braided shielding with foil shield. When grounding
the cable shield, observe the grounding concept of the plant.

For further information on planning and installing PROFIBUS networks see:
• Operating Instructions "PROFIBUS DP/PA: Guidelines for planning and commissioning"(BA00034S)• PNO Directive 2.092 "PROFIBUS PA User and Installation Guideline"• IEC 61158-2 (MBP)
EtherNet/IP
The standard ANSI/TIA/EIA-568-B.2 Annex specifies CAT 5 as the minimum category for a cableused for EtherNet/IP. CAT 5e and CAT 6 are recommended.
For more information on planning and installing EtherNet/IP networks, please refer to the"Media Planning and Installation Manual. EtherNet/IP" of ODVA Organization
PROFINET
Standard IEC 61156-6 specifies CAT 5 as the minimum category for a cable used for PROFINET. CAT
5e and CAT 6 are recommended.
For more information on planning and installing PROFINET networks, see: "PROFINET Cabling
and Interconnection Technology", Guideline for PROFINETPROFINET with Ethernet-APL
The reference cable type for APL segments is fieldbus cable type A, MAU type 1 and 3 (specified in
IEC 61158-2). This cable meets the requirements for intrinsically safe applications according to IEC
TS 60079-47 and can also be used in non-intrinsically safe applications.
Cable typeA

Cable capacitance

45 to 200 nF/km

Loop resistance

15 to 150 Ω/km

Cable inductance

0.4 to 1 mH/km

Further details are provided in the Ethernet-APL Engineering Guideline(https://www.ethernet-apl.org).

52

Endress+Hauser

Proline Promass Q 500

FOUNDATION Fieldbus
Twisted, shielded two-wire cable.
For further information on planning and installing FOUNDATION Fieldbus networks see:
• Operating Instructions for "FOUNDATION Fieldbus Overview" (BA00013S)• FOUNDATION Fieldbus Guideline• IEC 61158-2 (MBP)
Modbus RS485
The EIA/TIA-485 standard specifies two types of cable (A and B) for the bus line which can be used
for every transmission rate. Cable type A is recommended.
Cable typeA

Characteristic impedance

135 to 165 Ω at a measuring frequency of 3 to 20 MHz

Cable capacitance< 30 pF/mWire cross-section> 0.34 mm2 (22 AWG)

Cable type

Twisted pairs

Loop resistance≤110 Ω/km

Signal damping

Max. 9 dB over the entire length of the cable cross-section

ShieldCopper braided shielding or braided shielding with foil shield. When grounding
the cable shield, observe the grounding concept of the plant.

Current output 0/4 to 20 mA• Standard installation cable is sufficient• For custody transfer measurement, use a shielded cable: tin-plated copper braid, optical cover≥ 85 %
Pulse /frequency /switch output• Standard installation cable is sufficient• For custody transfer measurement, use a shielded cable: tin-plated copper braid, optical cover≥ 85 %
Double pulse output• Standard installation cable is sufficient• For custody transfer measurement, use a shielded cable: tin-plated copper braid, optical cover≥ 85 %
Relay output
Standard installation cable is sufficient.
Current input 0/4 to 20 mA• Standard installation cable is sufficient• For custody transfer measurement, use a shielded cable: tin-plated copper braid, optical cover≥ 85 %
Status input• Standard installation cable is sufficient• For custody transfer measurement, use a shielded cable: tin-plated copper braid, optical cover≥ 85 %
Choice of connecting cable between the transmitter and sensor
Depends on the type of transmitter and the installation zones

Endress+Hauser

53

Proline Promass Q 500

4
1

1

A
3

2

1

5
A
3

B
1

C
3

B
3

3

2

6
C
3
A0032476

1
2
3
4
5
6
A

B

C

Proline 500 digital transmitter
Proline 500 transmitter
Sensor Promass
Non-hazardous area
Hazardous area: Zone 2; Class I, Division 2
Hazardous area: Zone 1; Class I, Division 1
Standard cable to 500 digital transmitter →  54
Transmitter installed in the non-hazardous area or hazardous area: Zone 2; Class I, Division 2 / sensor
installed in the hazardous area: Zone 2; Class I, Division 2Standard cable to 500 digital transmitter →  55
Transmitter installed in the hazardous area: Zone 2; Class I, Division 2 / sensor installed in the hazardous
area: Zone 1; Class I, Division 1
Signal cable to 500 transmitter →  57
Transmitter and sensor installed in the hazardous area: Zone 2; Class I, Division 2 or Zone 1;
Class I, Division 1

A: Connecting cable between sensor and transmitter: Proline 500 – digital
Standard cable
A standard cable with the following specifications can be used as the connecting cable.

54

Design4 cores (2 pairs); uninsulated stranded CU wires; pair-stranded with common
shieldShieldingTin-plated copper braid, optical cover ≥ 85 %Loop resistancePower supply line (+, –): maximum 10 Ω

Cable length

Maximum 300 m (900 ft), see the following table.

Cross-section

Cable length [max.]

0.34 mm2 (AWG 22)

80 m (240 ft)

0.50 mm2 (AWG 20)

120 m (360 ft)

0.75 mm2 (AWG 18)

180 m (540 ft)

1.00 mm2 (AWG 17)

240 m (720 ft)

1.50 mm2 (AWG 15)

300 m (900 ft)

Endress+Hauser

Proline Promass Q 500

Optionally available connecting cable
Design2 × 2 × 0.34 mm2 (AWG 22) PVC cable 1) with common shield (2 pairs,
uninsulated stranded CU wires; pair-stranded)

Flame resistanceAccording to DIN EN 60332-1-2

Oil-resistanceAccording to DIN EN 60811-2-1

ShieldingTin-plated copper braid, optical cover ≥ 85 %Operating temperature

When mounted in a fixed position: –50 to +105 °C (–58 to +221 °F); when cable
can move freely: –25 to +105 °C (–13 to +221 °F)

Available cable length

Fixed: 20 m (60 ft); variable: up to maximum 50 m (150 ft)

1)

UV radiation can impair the cable outer sheath. Protect the cable from direct sunshine where possible.

B: Connecting cable between sensor and transmitter: Proline 500 - digital
Standard cable
A standard cable with the following specifications can be used as the connecting cable.

Endress+Hauser

Design4, 6, 8 cores (2, 3, 4 pairs); uninsulated stranded CU wires; pair-stranded with
common shield

ShieldingTin-plated copper braid, optical cover ≥ 85 %Capacitance C

Maximum 760 nF IIC, maximum 4.2 µF IIB

Inductance L

Maximum 26 µH IIC, maximum 104 µH IIB

Inductance/resistanceratio (L/R)

Maximum 8.9 µH/Ω IIC, maximum 35.6 µH/Ω IIB (e.g. according to IEC
60079-25)

Loop resistancePower supply line (+, –): maximum 5 Ω

Cable length

Maximum 150 m (450 ft), see the following table.

55

Proline Promass Q 500

Cross-section

Cable length [max.]

Termination

2 x 2 x 0.50 mm2(AWG 20)50 m (150 ft)

2 x 2 x 0.50 mm2 (AWG 20)

BN WT YE GN

A
B
GY• +, – = 0.5 mm2• A, B = 0.5 mm2
3 x 2 x 0.50 mm2(AWG 20)100 m (300 ft)

3 x 2 x 0.50 mm2 (AWG 20)

BN WT GY PK YE GN

A
B
GY• +, – = 1.0 mm2• A, B = 0.5 mm2
4 x 2 x 0.50 mm2(AWG 20)150 m (450 ft)

4 x 2 x 0.50 mm2 (AWG 20)

BN WT GY PK RD BU

A
B
GY

YE GN

• +, – = 1.5 mm2• A, B = 0.5 mm2

Optionally available connecting cable
Connecting cable forZone 1; Class I, Division 1

Standard cable2 × 2 × 0.5 mm2 (AWG 20) PVC cable 1) with common shield (2 pairs, pairstranded)

Flame resistanceAccording to DIN EN 60332-1-2

Oil-resistanceAccording to DIN EN 60811-2-1

ShieldingTin-plated copper braid, optical cover ≥ 85 %Operating temperature

When mounted in a fixed position: –50 to +105 °C (–58 to +221 °F); when cable
can move freely: –25 to +105 °C (–13 to +221 °F)

Available cable length

Fixed: 20 m (60 ft); variable: up to maximum 50 m (150 ft)

1)56

UV radiation can impair the cable outer sheath. Protect the cable from direct sunshine where possible.

Endress+Hauser

Proline Promass Q 500

C: Connecting cable between sensor and transmitter: Proline 500
Design7 × 0.38 mm2 PUR cable 1) with individual shielded cores and common
copper shield
With order code for «Approval; transmitter; sensor», options AA, BS, CS,
CZ, GR, GS, MS, NS, UR, US:
7 × 0.38 mm2 PVC cable 1) with individual shielded cores and common
copper shield

Conductor resistance≤ 50 Ω/km (0.015 Ω/ft)

Capacitance: core/shield≤ 420 pF/m (128 pF/ft)

Cable length (max.)

20 m (60 ft)

Cable lengths (available fororder)

5 m (15 ft), 10 m (30 ft), 20 m (60 ft)

Cable diameter

11 mm (0.43 in) ± 0.5 mm (0.02 in)Operating temperature

Depends on the device version and how the cable is installed:• Standard version:• Cable - fixed installation: –40 to +105 °C (–40 to +221 °F)• Cable - movable: –25 to +105 °C (–13 to +221 °F)• Order code for "Test, certificate", option JP:• Cable - fixed installation: –50 to +105 °C (–58 to +221 °F)• Cable - movable: –25 to +105 °C (–13 to +221 °F)• Order code for "Test, certificate", option JQ:• Cable - fixed installation: –60 to +105 °C (–76 to +221 °F)• Cable - movable: –25 to +105 °C (–13 to +221 °F)

1)Overvoltage protection

UV radiation can impair the cable outer sheath. Protect the cable from direct sunshine where possible.

Mains voltage fluctuations→  37

Overvoltage category

Overvoltage category II

Short-term, temporary overvoltage

Up to 1200 V between cable and ground, for max. 5 s

Long-term, temporary overvoltage

Up to 500 V between cable and ground

Performance characteristics
Reference operating
conditions• Error limits based on ISO 11631• Water with +15 to +45 °C (+59 to +113 °F) at 2 to 6 bar (29 to 87 psi)
• Specifications as per calibration protocol• Accuracy based on accredited calibration rigs that are traced to ISO 17025.
To obtain measured errors, use the Applicator sizing tool →  129

Maximum measured erroro.r. = of reading; 1 g/cm³ = 1 kg/l; T = medium temperature
Base accuracy
Design fundamentals →  61
Mass flow and volume flow (liquids)• ±0.05 % o.r. (optional for mass flow: PremiumCal; order code for "Calibration flow", option D)• ±0.10 % o.r. (standard)
Mass flow (gases)±0.25 % o.r.

Endress+Hauser

57

Proline Promass Q 500

Mass flow (cryogenic liquids and gases under –100 °C (–148 °F))
±0.35 % o.r. (order code for "Measuring tube material", option LA)
Density (liquids)
Standard density• ±0.2 kg/m³ ( ±0.0002 g/cm³ )• Valid in density range: 0 to 2 000 kg/m³
Premium density (DN 25 (1"); order code for "Application package", option EI)• ±0.1 kg/m³• Valid in density range: 0 to 3 000 kg/m³
For additional information, see the Special Documentation on the advanced density function→  131
For highly accurate density measurement, the pitch and roll angle and pressure compensation must
be configured.
For highly accurate density measurement, avoid significant tensile stresses due to the installationand ensure the flow velocity in the nominal diameter is > 0.1 m/s (0.33 ft/s).
Density (cryogenic liquids and gases under –100 °C (–148 °F))
±0.03 g/cm³ (order code for "Measuring tube material", option LA)
Temperature±0.1 °C ± 0.003 · T °C (±0.18 °F ± 0.003 · (T – 32) °F)
Zero point stability
DN

Zero point stability

[mm]

[in]

[kg/h][lb/min]

25

1

0.36

0.013

50

2

1.3

0.048

80

3

4.4

0.162

100

4

11.5

0.42

150

6

16

0.59

200

8

24

0.88

250

10

50

1.84

Flow values
Flow values as turndown parameters depending on the nominal diameter.
SI units

58

DN

1:1

1:10

1:20

1:50

1:100

1:500

[mm]

[kg/h][kg/h][kg/h][kg/h][kg/h][kg/h]

25

20 000

2 000

1 000

400

200

40

50

80 000

8 000

4 000

1 600

800

160

80

200 000

20 000

10 000

4 000

2 000

400

100

550 000

55 00027 500

11 000

5 500

1 100

150

850 000

85 00042 50017 000

8 500

1 700

200

1 500 000

150 000

75 00030 000

15 0003 000

250

2 400 000

240 000

120 000

48 00024 0004 800

Endress+Hauser

Proline Promass Q 500

US units
DN

1:1

1:10

1:20

1:50

1:100

1:500

[inch][lb/min][lb/min][lb/min][lb/min][lb/min][lb/min]

1

735

73

37

15

7

1

2

2939

294

147

59

29

6

3

7349

735

367

147

73

15

4

20209

2021

1010

404

202

40

6

31232

3123

1562

625

312

62

8

55115

5511

2756

1102

551

110

10

88183

8818

4409

1764

882

176

Accuracy of outputs
The outputs have the following base accuracy specifications.
Current output
Accuracy±5 µA

Pulse/frequency output
o.r. = of reading
Accuracy

Repeatability

Max. ±50 ppm o.r. (over the entire ambient temperature range)o.r. = of reading; 1 g/cm3 = 1 kg/l; T = medium temperature
Base repeatability
Design fundamentals →  61
Mass flow and volume flow (liquids)±0.025 % o.r.
Mass flow (gases)±0.20 % o.r.
Mass flow (cryogenic liquids and gases under –100 °C (–148 °F))
±0.175 % % o.r. (order code for "Measuring tube material", option LA)
Density (liquids)• ±0.1 kg/m3 / ±0.0001 g/cm3• Premium density: ±0.02 kg/m3 / ±0.00002 g/cm3
Density (cryogenic liquids and gases under –100 °C (–148 °F))
±0.015 g/cm3 (order code for "Measuring tube material", option LA)
Temperature±0.05 °C ± 0.0025 · T °C (±0.09 °F ± 0.0015 · (T–32) °F)

Response time

Endress+Hauser

The response time depends on the configuration (damping).

59

Proline Promass Q 500

Influence of ambient
temperature

Current output
Temperature coefficientMax. 1 μA/°C

Pulse/frequency output
Temperature coefficient

Influence of medium
temperature

No additional effect. Included in accuracy.

Mass flow and volume flowo.f.s. = of full scale value
If there is a difference between the temperature at zero adjustment and the process temperature, theadditional measured error of the sensors is typically
DN 25 (1"): ±0.0001 % o.f.s./°C (±0.00005 % o.f.s./°F)
DN 50 to 250 (2 to 10 "): ±0.00015 % o.f.s./°C (±0.000075 % o.f.s./°F)
The influence is reduced when the zero adjustment is performed at process temperature.
Density
If there is a difference between the density calibration temperature and the process temperature, themeasured error of the sensors is typically ±0.015 kg/m3/°C (±0.0075 kg/m3/°F) outside of+20 to +60 °C (+68 to +140 °F).
Premium density (order code for "Application package", option EI)
If there is a difference between the reference temperature of 20°C and the process temperature, theadditional maximum measured error of the sensors is typically±0.0025 kg/m3/°C (±0.00139 kg/m3/°F) within the temperature calibration range.
Outside the calibrated temperature range, the influence of the process temperature is typically±0.005 kg/m3/°C (±0.00278 kg/m3/°F)
3[kg/m ]
4.0
3.5
3.0
2.5
1

2.0
1.5
1.0

2

0.5
0
-150

-100

-50

0

50

100

150

200 [°C]

300

400

[°F]
-200

-100

0

100

200

A0046818

12Standard density
Premium density

Temperature±0.005 · T °C (± 0.005 · (T – 32) °F)
Influence of medium
pressure60

The tables below show the effect that a difference in pressure between the calibration pressure andthe process pressure has on the accuracy in the case of the mass flow and density.

Endress+Hauser

Proline Promass Q 500

o.r. = of reading
It is possible to compensate for the effect by:• Reading in the current pressure measured value via the current input or a digital input.
• Specifying a fixed value for the pressure in the device parameters.
Operating Instructions →  130.
Mass flow
DN[% o.r./bar]±0.0005[% o.r./psi]±0.00003

[mm]

[in]

25

1

–0.0040–0.000276

50

2–0.0025–0.000172

80

3–0.0050–0.000345

100

4–0.0040–0.000276

150

6–0.0077–0.000531

200

8–0.0074–0.000510

250

10–0.0076–0.000524

[mm]

[in]

[% o.r./bar]±0.0006±0.0003 1)[% o.r./psi]±0.00004±0.00002 1)

25

1–0.0029–0.000200

50

2–0.0034–0.000234

80

3–0.0024–0.000166

100

4–0.0006–0.000041

150

6–0.0040–0.000276

200

8–0.0015–0.000103

250

10–0.0048–0.000331

Density
DN

1)Premium density

The values for the influence of the medium pressure are based on the density of water.

Design fundamentals

o.r. = of reading, o.f.s. = of full scale value
BaseAccu = base accuracy in % o.r., BaseRepeat = base repeatability in % o.r.
MeasValue = measured value; ZeroPoint = zero point stability
Calculation of the maximum measured error as a function of the flow rate
Flow rate

Maximum measured error in % o.r.

ZeroPoint³ BaseAccu ⋅ 100± BaseAccu
A0021339
A0021332

ZeroPoint< BaseAccu ⋅ 100±
A0021333

Endress+Hauser

ZeroPoint
MeasValue ⋅ 100
A0021334

61

Proline Promass Q 500

Calculation of the maximum repeatability as a function of the flow rate
Flow rate³

Maximum repeatability in % o.r.

½ ⋅ ZeroPoint
BaseRepeat ⋅ 100

A0021340
A0021335<½ ⋅ ZeroPoint
BaseRepeat ⋅ 100

ZeroPoint± ½ ⋅ MeasValue ⋅ 100
A0021337

A0021336

Example of maximum measured error
E [%]
2.5
2.0
1.5
1.0
0.5
0
0

10

20

30

40

50

60

70

80

90

100 Q [%]
A0028741

E
Q

Maximum measured error in % o.r. (example with PremiumCal)
Flow rate in % of maximum full scale value

Installation
Mounting location

A0028772

To prevent measuring errors arising from accumulation of gas bubbles in the measuring pipe, avoid
the following mounting locations in the piping:• Highest point of a pipeline.
• Directly upstream of a free pipe outlet in a down pipe.
Installation in down pipes
However, the following installation suggestion allows for installation in an open vertical pipeline.
Pipe restrictions or the use of an orifice with a smaller cross-section than the nominal diameter
prevent the sensor running empty while measurement is in progress.

62

Endress+Hauser

Proline Promass Q 500

1

2

3
4

5
A0028773

 22
1
2
3
4
5

Installation in a down pipe (e.g. for batching applications)

Supply tankSensorOrifice plate, pipe restriction
Valve
Batching tank

DN

Orientation

Ø orifice plate, pipe restriction

[mm]

[in]

[mm]

[in]

25

1

14

0.55

50

2

28

1.10

80

3

50

1.97

100

4

65

2.60

150

6

90

3.54

200

8

120

4.72

250

10

150

5.91

The direction of the arrow on the sensor nameplate helps you to install the sensor according to theflow direction (direction of medium flow through the piping).
Orientation
A

Recommendation
1)

Vertical orientation

A0015591

B

2)

Horizontal orientation, transmitter at
top

Exception:→  23,  64
A0015589

Endress+Hauser

63

Proline Promass Q 500

Orientation
C

Recommendation
3)

Horizontal orientation, transmitter atbottomException:→  23,  64
A0015590

D→  66 4)

Horizontal orientation, transmitter atside
A0015592

1)2)This orientation is recommended to ensure self-draining.
Applications with low process temperatures may reduce the ambient temperature. To maintain theminimum ambient temperature for the transmitter, this orientation is recommended.
Applications with high process temperatures may increase the ambient temperature. To maintain the
maximum ambient temperature for the transmitter, this orientation is recommended.
Not recommended for inhomogeneous media.

3)
4)

If a sensor is installed horizontally with a curved measuring tube, match the position of the sensor to
the fluid properties.

1

2
A0028774 23
1
2

Inlet and outlet runs

64

Orientation of sensor with curved measuring tube

Avoid this orientation for fluids with entrained solids: Risk of solids accumulating.
Avoid this orientation for outgassing fluids: Risk of gas accumulating.

No special precautions need to be taken for fittings which create turbulence, such as valves, elbows
or T-pieces, as long as no cavitation occurs →  76.

Endress+Hauser

Proline Promass Q 500

Mounting the transmitter
housing

Proline 500 – digital transmitter
Post mountingø 20…70(ø 0.79…2.75)

4x

3x

TX 25

SW 10

~1

0

~
2(

)
4.0

A0029051

 24

Engineering unit mm (in)Wall mounting
17 (0.67)==

L

211 (8.31)

5.8 (0.23)

5.8 (0.23)
149 (5.85)
A0029054

 25
LEngineering unit mm (in)Depends on order code for "Transmitter housing"Order code for "Transmitter housing"• Option A, aluminum, coated: L =14 mm (0.55 in)• Option D, polycarbonate: L = 13 mm (0.51 in)Endress+Hauser

65

Proline Promass Q 500

Proline 500 transmitter
Post mounting

LWARNING
Order code for "Transmitter housing", option L "Cast, stainless": cast transmitters are very
heavy.
They are unstable if they are not mounted on a secure, fixed post.
‣ Only mount the transmitter on a secure, fixed post on a stable surface.

! 20…70 (! 0.79 to 2.75)

1

4x

SW 13

A0029057

 26

Engineering unit mm (in)Wall mounting! 18 (0.71)! 8.6 (0.39)

20 (0.79)

100 (3.94)! 10 (0.39)

100 (3.94)
A0029068

 27

Special mounting
instructions

Engineering unit mm (in)Drainability
When installed vertically, the measuring tubes can be drained completely and protected against
buildup.
Hygienic compatibility
When installing in hygienic applications, please refer to the information in the "Certificates andapprovals/hygienic compatibility" section →  121

66

Endress+Hauser

Proline Promass Q 500

Rupture disk
Process-related information: →  75.

LWARNING
Danger from medium escaping!
Medium escaping under pressure can cause injury or material damage.
‣ Take precautions to prevent danger to persons and damage if the rupture disk is actuated.
‣ Observe the information on the rupture disk sticker.
‣ Make sure that the function and operation of the rupture disk is not impeded through the
installation of the device.
‣ Do not use a heating jacket.
‣ Do not remove or damage the rupture disk.
The position of the rupture disk is indicated by a sticker affixed beside it.
The transportation guard must be removed.
The existing connecting nozzles are not intended for the purpose of rinsing or pressure monitoring,but instead serve as the mounting location for the rupture disk.
In the event of a failure of the rupture disk, a drain device can be screwed onto the female thread of
the rupture disk in order to drain off any escaping medium.

i

RUPTURE DISK

1

2

3
A0030346

1
2
3

Rupture disk label
Rupture disk with 1/2" NPT female thread and 1" width across flats
Transport protection

For information on the dimensions, see the "Mechanical construction" section (accessories).
Pitch and roll angles
If the device is used to measure the density of liquids, the pitch and roll angles must be taken into
account during installation.
For correct measurement, the pitch angle and roll angle must be determined duringcommissioning (with a tolerance of ±10 °) and entered in the pitch angle and roll angle
parameters.
For detailed information on density measurement, see the Special Documentation for the device→  131
Pitch angle
The technically relevant pitch angle is the angle shaded gray = –90 to +90 °.
Example (blue): Installation of the device with a pitch angle α = +30 °Endress+Hauser

67

Proline Promass Q 500+90°

α = +30°
0°–90°

A0040032

 28

Side view with flow direction from left to right.

Roll angle
The technically relevant roll angle is the angle shaded gray = –180 to +180 °.
Example (blue): Installation of the device with a roll angle β = +45 °0°
β = +45°

–90°

+90°

+/–180°A0040033

 29

Top view in flow direction

Environment
Ambient temperature range

Measuring device• –40 to +60 °C (–40 to +140 °F)• Order code for "Test, certificate", option JP:–50 to +60 °C (–58 to +140 °F)• Order code for "Test, certificate", option JQ:• Sensor: –60 to +60 °C (–76 to +140 °F)• Transmitter: –50 to +60 °C (–58 to +140 °F)

Readability of thelocal display–20 to +60 °C (–4 to +140 °F)
The readability of the display may be impaired at temperatures outside the temperature
range.

Dependency of ambient temperature on medium temperature →  70

68

Endress+Hauser

Proline Promass Q 500‣ If operating outdoors:
Avoid direct sunlight, particularly in warm climatic regions.
You can order a weather protection cover from Endress+Hauser. →  127.

Storage temperature–50 to +80 °C (–58 to +176 °F)

Climate class

DIN EN 60068-2-38 (test Z/AD)

Relative humidity

The device is suitable for use in outdoor and indoor areas with a relative humidity of 4 to 95%.

Operating height

According to EN 61010-1• ≤ 2 000 m (6 562 ft)• > 2 000 m (6 562 ft) with additional overvoltage protection (e.g. Endress+Hauser HAW Series)

Degree of protection

Transmitter• IP66/67, Type 4X enclosure, suitable for pollution degree 4• When the housing is open: IP20, Type 1 enclosure, suitable for pollution degree 2• Display module: IP20, Type 1 enclosure, suitable for pollution degree 2
Sensor• IP66/67, Type 4X enclosure, suitable for pollution degree 4• When the housing is open: IP20, Type 1 enclosure, suitable for pollution degree 2
Optional
DN 25 to 100: order code for "Sensor options", option CM "IP69External WLAN antenna
IP67

Vibration- and shockresistanceVibration sinusoidal, in accordance with IEC 60068-2-6
Sensor: order code for "Meas. tube mat., wetted parts surface", option LA, SD, SE, SF, TH, TT, TU• 2 to 8.4 Hz, 3.5 mm peak
• 8.4 to 2 000 Hz, 1 g peakSensor: order code for "Meas. tube mat., wetted parts surface", option HA, SA, SB, SC• 2 to 8.4 Hz, 7.5 mm peak
• 8.4 to 2 000 Hz, 2 g peakTransmitter• 2 to 8.4 Hz, 7.5 mm peak
• 8.4 to 2 000 Hz, 2 g peak
Vibration broad-band random, according to IEC 60068-2-64
Sensor: order code for "Meas. tube mat., wetted parts surface", option LA, SD, SE, SF, TH, TT, TU• 10 to 200 Hz, 0.003 g2/Hz• 200 to 2 000 Hz, 0.001 g2/Hz• Total: 1.54 g rms
Sensor: order code for "Meas. tube mat., wetted parts surface", option HA, SA, SB, SC• 10 to 200 Hz, 0.01 g2/Hz• 200 to 2 000 Hz, 0.003 g2/Hz• Total: 2.70 g rms
Transmitter• 10 to 200 Hz, 0.01 g2/Hz• 200 to 2 000 Hz, 0.003 g2/Hz• Total: 2.70 g rms
Shock half-sine, according to IEC 60068-2-27

Endress+Hauser

69

Proline Promass Q 500• Sensor: order code for "Meas. tube mat., wetted parts surface", option LA, SD, SE, SF, TH, TT, TU
6 ms 30 g• Sensor: order code for "Meas. tube mat., wetted parts surface", option HA, SA, SB, SC
6 ms 50 g• Transmitter
6 ms 50 g
Rough handling shocks, according to IEC 60068-2-31
Interior cleaning• Cleaning in place (CIP)• Sterilization in place (SIP)
Options• Oil- and grease-free version for wetted parts, without declaration
Order code for "Service", option HA• Oil- and grease-free version for wetted parts as per IEC/TR 60877-2.0 and BOC 50000810-4, withdeclaration
Order code for "Service", option HB

Mechanical load

Transmitter housing and sensor connection housing:• Protect against mechanical effects, such as shock or impact• Do not use as a ladder or climbing aid

Electromagnetic
compatibility (EMC)• As per IEC/EN 61326 and NAMUR Recommendation 21 (NE 21)• Device version with PROFIBUS DP: Complies with emission limits for industry as per EN 50170
Volume 2, IEC 61784
The following applies for PROFIBUS DP: If baud rates > 1.5 MBaud, an EMC cable entry must be
used and the cable shield must continue as far as the terminal wherever possible.
Details are provided in the Declaration of Conformity.
This unit is not intended for use in residential environments and cannot guarantee adequate
protection of the radio reception in such environments.

Process
Medium temperature range
Standard version–50 to +205 °C (–58 to +401 °F)

Order code for "Measuring tube
mat., wetted surface", option SA, SB

Low-temperature version–196 to +150 °C (–320 to +302 °F)

Order code for "Measuring tube
mat., wetted surface", option LA

NOTICE
Material fatigue due to excessive
temperature difference!
‣ Maximum temperature
difference of media used: 300 K

70

Endress+Hauser

Proline Promass Q 500

Dependency of ambient temperature on medium temperature
Ta
A
B

Tm
A0031121

 30
Ta
Tm
A
B

Exemplary representation, values in the table below.

Ambient temperature
Medium temperature
Maximum permitted medium temperature Tm at Ta max = 60 °C (140 °F); higher medium temperatures Tm
require a reduction in the ambient temperature Ta
Maximum permitted ambient temperature Ta for the maximum specified medium temperature Tm of thesensorValues for devices that are used in the hazardous area:
Separate Ex documentation (XA) for the device →  131.
Not insulated

InsulatedA
Version 1)
Standard version
1)

B
Ta

Tm

60 °C (140 °F)

205 °C (401 °F)

A

Ta Tm
–

–

B
Ta

Tm

Ta

Tm

60 °C (140 °F)

150 °C (302 °F)

50 °C (122 °F)

205 °C (401 °F)

The values apply for Promass Q 500 - digital and Promass Q 500.

Density

0 to 5 000 kg/m3 (0 to 312 lb/cf)

Pressure-temperature
ratings

The following pressure/temperature diagrams apply to all pressure-bearing parts of the device andnot just the process connection. The diagrams show the maximum permissible medium pressure
depending on the specific medium temperature.
Pressure-temperature ratings with the +151 to +205 °C (+304 to +401 °F) temperature range
are only for measuring devices with the extended temperature version.

Endress+Hauser

71

Proline Promass Q 500

Flange according to EN 1092-1 (DIN 2501)[psi] [bar]
100
1400
90
1200
80
1000
800

PN 100

70
60

PN 63

50

600

40

400

30

PN 40

20
200
0

10

PN 16

0
-200

-100

-320

-50
-80

50

0
0

80

100
160

240

150
320

200

[°C]

400 [°F]
A0029636-EN

 31

With flange material 1.4404 (F316/F316L)

Flange according to ASME B16.5[psi] [bar]
100
1400
90
1200
80
1000
800

70
60
50

600

40

400

30
20

200
0

Class 600

10Class 300

Class 150

0
-200
-320

-50
-80

50

0
0

80

100
160

240

150
320

200

[°C]

400 [°F]
A0029638-EN

 32

72

With flange material 1.4404 (F316/F316L)

Endress+Hauser

Proline Promass Q 500

Flange JIS B2220[psi] [bar]1600 110

63K (≤ DN 80)

100

1400

63K (≥ DN 100)

90
1200

80

1000

70
40K

60

800

50

600

40

400

30

20K

20
200

10

10K

0

0

-200

50

0
80

-320

100
160

150

200 [°C]
400 [°F]

320

240

A0029639-EN

 33

With flange material 1.4404 (F316/F316L)

Flange DIN 11864-2 Form A[psi] [bar]
50
600
40
400

30

DN 8…40

20
200

10

≥ DN 50

0

0

-50
0

-80

50

0
80

100
160

240

150
320

200

250 [°C]

400

480 [°F]
A0028782-EN

 34

With connection material 1.4404 (316/316L)

Thread DIN 11851[psi] [bar]
50
600
40
400

DN 8…40

30
20

200
0

≥ DN 50

10
0
-50
-80

50

0
0

80

100
160

240

150
320

200

250 [°C]

400

480 [°F]
A0028794-EN

 35

Endress+HauserWith connection material 1.4404 (316/316L)

73

Proline Promass Q 500

DIN 11851 allows for applications up to +140 °C (+284 °F) if suitable sealing materials are used.
Please take this into account when selecting seals and counterparts, as these components can limit
the pressure and temperature range.
Thread ISO 2853
[psi] [bar]
400
20
200
0

10
0
-50
-80

50

0
0

80

100
160

240

150
320

200

250 [°C]

400

480 [°F]
A0028799-EN

 36

With connection material 1.4404 (316/316L)

Thread SMS 1145
[psi] [bar]
400
20
200
0

10
0
-50
-80

50

0
0

80

100
160

240

150
320

200

250 [°C]

400

480 [°F]
A0028800-EN

 37

With connection material 1.4404 (316/316L)

SMS 1145 allows for applications up to 16 bar (232 psi) if suitable sealing materials are used. Pleasetake this into account when selecting seals and counterparts, as these components can limit thepressure and temperature range.
Tri-Clamp[psi] [bar]
400
20
200
0

10
0
-50
-80

50

0
0

80

100
160

240

150
320

200

250 [°C]

400

480 [°F]
A0032216-EN

The clamp connections are suitable up to a maximum pressure of 16 bar (232 psi). Please observe
the operating limits of the clamp and seal used as they can be over 16 bar (232 psi). The clamp andseal are not included in the scope of supply.
Sensor housing

The sensor housing is filled with helium and protects the electronics and mechanics inside.
If a measuring tube fails (e.g. due to process characteristics like corrosive or abrasive fluids), thefluid will initially be contained by the sensor housing.
In the event of a tube failure, the pressure level inside the sensor housing will rise according to theoperating process pressure. If the user judges that the sensor housing burst pressure does not

74

Endress+Hauser

Proline Promass Q 500

provide an adequate safety margin, the device can be fitted with a rupture disk. This prevents
excessively high pressure from forming inside the sensor housing. Therefore, the use of a rupture
disk is strongly recommended in applications involving high gas pressures, and particularly inapplications in which the process pressure is greater than 2/3 of the sensor housing burst pressure.
If there is a need to drain the leaking medium into a discharge device, the sensor should be fittedwith a rupture disk. Connect the discharge to the additional threaded connection →  91.
If the sensor is to be purged with gas (gas detection), it should be equipped with purge connections.
Do not open the purge connections unless the containment can be filled immediately with adry, inert gas. The use of helium at low pressure is recommended for purging.
Maximum pressure: 0.5 bar (7.3 psi)
Burst pressure of the sensor housing
The following sensor housing burst pressures are only valid for standard devices and/or devices
equipped with closed purge connections (not opened/as delivered).
If a device fitted with purge connections (order code for "Sensor option", option CH "Purge
connection") is connected to the purge system, the maximum pressure is determined by the purge
system itself or by the device, depending on which component has the lower pressure classification.
If the device is fitted with a rupture disk (order code for "Sensor option", option CA "Rupture disk"), therupture disk trigger pressure is decisive .
The sensor housing burst pressure refers to a typical internal pressure which is reached prior tomechanical failure of the sensor housing and which was determined during type testing. The
corresponding type test declaration can be ordered with the device (order code for "Additional
approval", option LN "Sensor housing burst pressure, type test").
DN

Sensor housing burst pressure

[mm]

[in]

[bar]

[psi]

25

1

220

3 191

50

2

160

2 320

80

3

150

2 175

100

4

120

1 740

150

6

120

1740

200

8

100

1450

250

10

100

1450

For information on the dimensions: see the "Mechanical construction" section →  79
Rupture disk

To increase the level of safety, a device version with a rupture disk with a trigger pressure of
10 to 15 bar (145 to 217.5 psi)can be used (order code for "Sensor option", option "rupture disk").
For information on the dimensions: see the "Mechanical construction" section (accessories) →  91

Flow limit

Select the nominal diameter by optimizing between the required flow range and permissible pressureloss.
For an overview of the full scale values for the measuring range, see the "Measuring range"
section →  11
• The minimum recommended full scale value is approx. 1/20 of the maximum full scale value• In most applications, 20 to 50 % of the maximum full scale value can be considered ideal• A low full scale value must be selected for abrasive media (such as liquids with entrained solids):
flow velocity < 1 m/s (< 3 ft/s).
To calculate the flow limit, use the Applicator sizing tool →  129

Pressure loss

Endress+Hauser

To calculate the pressure loss, use the Applicator sizing tool →  129

75

Proline Promass Q 500

System pressure

It is important that cavitation does not occur, or that gases entrained in the liquids do not outgas.
This is prevented by means of a sufficiently high system pressure.
For this reason, the following mounting locations are recommended:• At the lowest point in a vertical pipe
• Downstream from pumps (no danger of vacuum)

A0028777

Thermal insulation

In the case of some fluids, it is important to keep the heat radiated from the sensor to thetransmitter to a low level. A wide range of materials can be used for the required insulation.
NOTICE
Electronics overheating on account of thermal insulation!
‣ Recommended orientation: horizontal orientation, sensor connection housing pointing
downwards.
‣ Do not insulate the sensor connection housing.
‣ Maximum permissible temperature at the lower end of the sensor connection housing:
80 °C (176 °F)‣ Thermal insulation with not isolated extended neck: We recommend that you do not insulate theextended neck in order to ensure optimum dissipation of heat.

A0034391

 38

Thermal insulation with not isolated extended neckLow-temperature version: It is generally not necessary to insulate the sensor connection
housing. If insulation is provided, the rules that apply are the same as those for thermal
insulation.
Heating

Some fluids require suitable measures to avoid loss of heat at the sensor.
Heating options• Electrical heating, e.g. with electric band heaters 1)• Via pipes carrying hot water or steam• Via heating jackets
NOTICE
Danger of overheating when heating‣ Ensure that the temperature at the lower end of the transmitter housing does not exceed
80 °C (176 °F).
‣ Ensure that sufficient convection takes place at the transmitter neck.
‣ When using in potentially explosive atmospheres, observe the information in the device-specific
Ex documentation. For detailed information on the temperature tables, see the separate
document entitled "Safety Instructions" (XA) for the device.

1)

76

The use of parallel electric band heaters is generally recommended (bidirectional electricity flow). Particular considerations must be made if asingle-wire heating cable is to be used. Additional information is provided in the document EA01339D "Installation instructions for electrical
trace heating systems" →  132

Endress+Hauser

Proline Promass Q 500

Vibrations

Endress+Hauser

The operational reliability of the measuring system is not affected by plant vibrations.

77

Proline Promass Q 500

Custody transfer
The measuring device is optionally tested in accordance with OIML R117/R81 and has an
EU type evaluation certificate which authorizes the use in EU type-examination certificates according
to Measuring Instruments Directive 2014/32/EU for service subject to legal metrological control("custody transfer") for liquids other than water and cryogenic liquids (Annex VII).
The measuring device is optionally tested according to OIML R137 and has an EU type-examination
certificate according to Measuring Instruments Directive 2014/32/EU for service subject to legal
metrological control ("custody transfer") as a gas meter (Annex IV).
The device is used with a legally controlled totalizer on the local display and optionally with legallycontrolled outputs.
Measuring devices subject to legal metrological control totalize in both directions, i.e. all the outputs
consider flow components in the positive (forward) and negative (reverse) flow direction.
Generally a measuring device subject to legal metrological control is secured against tampering byseals on the transmitter or sensor. These seals may normally only be opened by a representative of
the competent authority for legal metrology controls.
After putting the device into circulation or after sealing the device, operation is only possible to alimited extent.
Detailed ordering information is available from your local Endress+Hauser sales center for national
approvals, which are based on the OIML certificates, for applications with liquids other than water orcryogenic liquids.
More information is provided in the supplementary documentation.

78

Endress+Hauser

Proline Promass Q 500

Mechanical construction
Dimensions in SI units

Housing of Proline 500 – digital transmitter
Non-hazardous area or hazardous area: Zone 2; Class I, Division 2
A

F

Q

N

P

G

A0033789

Order code for "Transmitter housing", option A "Aluminum, coated" and order code for "Integrated ISEM
electronics", option A "Sensor"
A
[mm]

F
[mm]

G
[mm]

N
[mm]

P
[mm]

Q
[mm]

167

232

89

187

24

21

Order code for "Transmitter housing", option D "Polycarbonate" and order code for "Integrated ISEM
electronics", option A "Sensor"
A
[mm]

F
[mm]

G
[mm]

N
[mm]

P
[mm]

Q
[mm]

177

234

89

197

17

22

Housing of Proline 500 transmitter
Hazardous area: Zone 2; Class I, Division 2 or Zone 1; Class I, Division 1
A

T
C

G

Q

F

B

Q
A0033788

Endress+Hauser

79

Proline Promass Q 500

Order code for "Transmitter housing", option A "Aluminum, coated" and order code for "Integrated ISEM
electronics", option B "Transmitter"
A
[mm]

B
[mm]

C
[mm]

F 1)
[mm]

G
[mm]

Q
[mm]

T
[mm]

188

85

103

318

217

130

239

1)

DN ≥ 150: values +72 mm

Order code for "Transmitter housing", option L "Cast, stainless" and order code for "Integrated ISEM
electronics", option B "Transmitter"
A
[mm]

B
[mm]

C
[mm]

F 1)
[mm]

G
[mm]

Q
[mm]

T
[mm]

188

85

103

295

217

130

239

1)

DN ≥ 150: values +72 mm

Sensor connection housing
A

G
C

D

K

F

E

B

L

M
A0033787

Order code for "Sensor connection housing", option A "Aluminum, coated"
DN

A 1)

B

C

D

E

F

G

K

L

M

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

25

148

94

54

217

256.5

473.5

136

15.2

2)

73

50

148

94

54

408

277

685

136

28.0

2)

115

43.3

2)

169

136

68.9

2)

220

80

148

100
1)
2)

148

94
94

54
54

524
655

304

828

330

985

136

Depending on the cable gland used: values up to + 30 mm
Depending on the process connection

Order code for "Sensor connection housing", option B "Stainless"
DN

A 1)

B

C

D

E

F

G

K

L

M

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

15.2

2)

73

28.0

2)

115

43.3

2)

169

68.9

2)

220

25
50
80
100

80

137
137
137
137

78
78
78
78

59
59
59
59

217
408
524
655

251.5
272
299
325

468.5
680
823
980

134
134
134
134

Endress+Hauser

Proline Promass Q 500

DN

A 1)

B

C

D

E

F

G

K

L

M

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

58.5

2)

244

71.5

2)

323.9

93.8

2)

355.6

150
200
250
1)
2)

137
137
137

78
78
78

59
59
59

626
790
887

336
375
392

962
1 166
1 279

134
134
134

Depending on the cable gland used: values up to + 30 mm
Depending on the process connection

Order code for "Sensor connection housing", option C "Ultra-compact hygienic, stainless"
DN

A 1)

B

C

D

E

F

G

K

L

M

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

15.2

2)

73

28.0

2)

115

43.3

2)

169

68.9

2)

220

25

124

50

124

80

124

100
1)
2)

124

68
68
68
68

56
56
56
56

217
408
524
655

251.5

468.5

272

680

299

823

325

980

112
112
112
112

Depending on the cable gland used: values up to + 30 mm
Depending on the process connection

Order code for "Sensor connection housing", option L "Cast, stainless"
DN

A 1)

B

C

D

E

F

G

K

L

M

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

[mm]

15.2

2)

73

28.0

2)

115

43.3

2)

169

68.9

2)

220

58.5

2)

244
324
356

25

145

50

145

80

145

100
150

1)
2)

Endress+Hauser

145
145

86
86
86
86
86

59
59
59
59
59

217
408
524
655
626

280

497

300

708

327
353

851
1 008

363

989

136
136
136
136
136

200

145

86

59

790

403

1 193

136

71.5

2)

250

145

86

59

887

419

1 306

136

93.8

2)

Depending on the cable gland used: values up to + 30 mm
Depending on the process connection81Proline Promass Q 500

Flange connections

B
E

A

C

Fixed flange connections EN 1092-1, ASME B16.5, JIS B2220

L

D

A0015621

Length tolerance for dimension L in mm:+1.5 / –2.0
Flange according to EN 1092-1 (DIN 2501 / DIN 2512N): PN16
1.4404 (F316/F316L): order code for "Process connection", option D1SFlange with groove according to EN 1092-1 Form D (DIN 2512N): PN16
1.4404 (F316/F316L): order code for "Process connection", option D5S
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

100

220

180

8 × ⌀18

20

107.1

1 128

150

285

240

8 × ⌀2222159.3

1 136/1 330 1)

200

340

295

12 × ⌀22

24

206.5

1 343

250

405

355

12 × ⌀26

26

260.4

1 775

Surface roughness (flange): EN 1092-1 Form B1 (DIN 2526 Form C), Ra 3.2 to 12.5 µm
1)Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Flange according to EN 1092-1 (DIN 2501): PN16 with reduction in nominal diameter
1.4404 (F316/F316L)
FlangeDN[mm]

DeviceReduction
to DN
[mm]

Order code for"Processconnection",
option

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

100

80

DHS

220

180

8 × ⌀18

20

107.1

874

150

100

DJS

285

240

8 × ⌀22

22

159.3

1 167

200

150

DLS

340

295

12 × ⌀22

24

206.5

1 267

Surface roughness (flange): EN 1092-1 Form B1 (DIN 2526 Form C), Ra 3.2 to 12.5 µm

Flange according to EN 1092-1 (DIN 2501 / DIN 2512N): PN 40
1.4404 (F316/F316L): order code for "Process connection", option D2S
Flange with groove according to EN 1092-1 Form D (DIN 2512N): PN 40
1.4404 (F316/F316L): order code for "Process connection", option D6S

82

DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

115

85

4 × ⌀14

18

28.5

440

50

165

125

4 × ⌀18

20

54.5

715

Endress+Hauser

Proline Promass Q 500

Flange according to EN 1092-1 (DIN 2501 / DIN 2512N): PN 40
1.4404 (F316/F316L): order code for "Process connection", option D2S
Flange with groove according to EN 1092-1 Form D (DIN 2512N): PN 40
1.4404 (F316/F316L): order code for "Process connection", option D6S
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

80

200

160

8 × ⌀18

24

82.5

840

100

235

190

8 × ⌀22

24

107.1

1 128

150

300

250

8 × ⌀2628

159.3

1 176/1 370 1)

200

375

320

12 × ⌀30

34

206.5

1 395

250

450

385

12 × ⌀33

38

258.8

1 845

Surface roughness (flange): EN 1092-1 Form B1 (DIN 2526 Form C), Ra 3.2 to 12.5 µm
1)Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Flange according to EN 1092-1 (DIN 2501): PN 40 with reduction in nominal diameter
1.4404 (F316/F316L)
FlangeDN[mm]

DeviceReduction
to DN
[mm]

Order code for"Processconnection",
option

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

80

50

DGS

200

160

8 × ⌀18

24

82.5

840

100

80

DIS

235

190

8 × ⌀22

24

107.1

874

150

100

DKS

300

250

8 × ⌀26

28

159.3

1 167

200

150

DMS

375

320

12 × ⌀30

34

206.5

1 267

Surface roughness (flange): EN 1092-1 Form B1 (DIN 2526 Form C), Ra 3.2 to 12.5 µm

Flange according to EN 1092-1 (DIN 2501 / DIN 2512N): PN 63
1.4404 (F316/F316L): order code for "Process connection", option D3S
Flange with groove according to EN 1092-1 Form D (DIN 2512N): PN 63
1.4404 (F316/F316L): order code for "Process connection", option D7S
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

50

180

135

4 × ⌀22

26

54.5

724

80

215

170

8 × ⌀22

28

81.7

875

100

250

200

8 × ⌀26

30

106.3

1 128

150

345

280

8 × ⌀3336

157.1

1 216/1 410 1)

200

415

345

12 × ⌀36

42

204.9

1 439

250

470

400

12 × ⌀36

46

255.4

1 885

Surface roughness (flange): EN 1092-1 Form B1 (DIN 2526 Form C), Ra 3.2 to 12.5 µm
1)Endress+Hauser

Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

83

Proline Promass Q 500

Flange according to EN 1092-1 (DIN 2501 / DIN 2512N): PN 100
1.4404 (F316/F316L): order code for "Process connection", option D4S
Flange with groove according to EN 1092-1 Form D (DIN 2512N): PN 100
1.4404 (F316/F316L): order code for "Process connection", option D8S
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

140

100

4 × ⌀18

24

28.5

470

50

195

145

4 × ⌀26

28

53.9

740

80

230

180

8 × ⌀26

32

80.9

885

100

265

210

8 × ⌀30

36

104.3

1 128

150

355

290

12 × ⌀33

44

154.0

1 256/1 450 1)

200

430

360

12 × ⌀36

52

199

1 479

250

505

430

12 × ⌀39

60

248

1 949

Surface roughness (flange): EN 1092-1 Form B2 (DIN 2526 Form E), Ra 0.8 to 3.2 µm
1)Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Flange according to ASME B16.5: Class 150
1.4404 (F316/F316L)
Order code for "Process connection", option AAS
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

110

79.4

4 × ⌀15.7

14.2

26.7

440

50

150

120.7

4 × ⌀19.1

19.1

52.6

715

80

190

152.4

4 × ⌀19.1

23.9

78.0

840

100

230

190.5

8 × ⌀19.1

23.9

102.4

1 128

150

280

241.3

8 × ⌀22.2

25.9

154.1

1 203/1 398 1)

200

345

298.5

8 × ⌀22.2

29

202.7

1 423

250

405

362

12 × ⌀25.4

30.6

254.5

1 832

Surface roughness (flange): Ra 3.2 to 6.3 µm
1)Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Flange according to ASME B16.5: Class 150 with reduction in nominal diameter
1.4404 (F316/F316L)
FlangeDN[mm]

DeviceReduction
to DN
[mm]

Order code for"Processconnection",
option

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

80

50

AJS

190

152.4

4 × ⌀19.1

23.9

78.0

720

100

80

ALS

230

190.5

8 × ⌀19.1

23.9

102.4

874

150

100

ANS

280

241.3

8 × ⌀22.4

25.4

154.2

1 167

200

150

APS

345

298.5

8 × ⌀22.2

29

202.7

1 266

250

200

AVS

405

362

12 × ⌀25.4

30.6

254.6

1 408/
1 832 1)

300

250

AXS

485

431.8

12 × ⌀25.4

32.2

304.8

1 935

Surface roughness (flange): Ra 3.2 to 6.3 µm
1)84Installed length such as Promass F, DN 250 (order code for "Sensor option", option CO)

Endress+Hauser

Proline Promass Q 500

Flange according to ASME B16.5: Class 300
1.4404 (F316/F316L)
Order code for "Process connection", option ABS
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

125

88.9

4 × ⌀19.1

17.5

26.7

440

50

165

127

8 × ⌀19.1

22.3

52.6

715

80

210

168.3

8 × ⌀22.3

28.4

78.0

840

100

255

200

8 × ⌀22.3

31.7

102.4

1 128

150

320

269.9

12 × ⌀22.2

37

154.1

1 223/1 417 1)

200

380

330.2

12 × ⌀25.4

41.7

202.7

1 443

250

445

387.4

16 × ⌀28.5

48.1

254.5

1 863

Surface roughness (flange): Ra 3.2 to 6.3 µm
1)Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Flange according to ASME B16.5: Class 300 with reduction in nominal diameter
1.4404 (F316/F316L)
FlangeDN[mm]

DeviceReduction
to DN
[mm]

Order code for"Processconnection",
option

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

80

50

AKS

210

168.3

8 × ⌀22.3

28.4

78.0

732

100

80

AMS

255

200

8 × ⌀22.3

31.7

102.4

894

150

100

AOS

320

269.9

12 × ⌀22.3

36.5

154.2

1 187

200

150

AQS

380

330.2

12 × ⌀25.4

41.7

202.7

1 266

250

200

AWS

445

374.4

16 × ⌀28.6

48.1

254.6

1 439/
1 863 1)

300

250

AZS

520

450.8

16 × ⌀31.8

51.3

304.8

1 935

Surface roughness (flange): Ra 3.2 to 6.3 µm
1)Installed length such as Promass F, DN 250 (order code for "Sensor option", option CO)

Flange according to ASME B16.5: Class 600
1.4404 (F316/F316L)
Order code for "Process connection", option ACS
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

125

88.9

4 × ⌀19.1

23.9

24.3

490

50

165

127

8 × ⌀19.1

31.8

49.2

742

80

210

168.3

8 × ⌀22.2

40.0

73.7

904

100

275

215.9

8 × ⌀25.4

48.4

97.3

1 158

150

355

292.1

12 × ⌀28.5

54.7

154.1

1 273/1 467 1)

200

420

349.2

12 × ⌀31.8

62.6

202.7

1 499

250

510

431.8

16 × ⌀35

70.5

254.5

1 946

Surface roughness (flange): Ra 3.2 to 6.3 µm
1)Endress+Hauser

Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

85

Proline Promass Q 500

Flange JIS B2220: 10K
1.4404 (F316/F316L)
Order code for "Process connection", option NDS
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

50

155

120

4 × ⌀19

16

50

715

80

185

150

8 × ⌀19

18

80

832

100

210

175

8 × ⌀19

18

100

1 128

150

280

240

8 × ⌀2322150

1 160/1 354 1)

200

330

290

12 × ⌀23

22

200

1 379

250

400

355

12 × ⌀25

24

250

1 775

Surface roughness (flange): Ra 3.2 to 6.3 µm
1)Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Flange JIS B2220: 20K
1.4404 (F316/F316L)
Order code for "Process connection", option NES
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

125

90

4 × ⌀19

16

25

440

50

155

120

8 × ⌀19

18

50

715

80

200

160

8 × ⌀23

22

80

832

100

225

185

8 × ⌀23

24

100

1 128

150

305

260

12 × ⌀25

28

150

1 192/1 386 1)

200

350

305

12 × ⌀25

30

200

1 379

250

430

380

12 × ⌀27

34

250

1 845

Surface roughness (flange): Ra 1.6 to 3.2 µm
1)Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Flange JIS B2220: 40K
1.4404 (F316/F316L)
Order code for "Process connection", option NGS
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

130

95

4 × ⌀19

22

25

485

50

165

130

8 × ⌀19

26

50

760

80

210

170

8 × ⌀23

32

75

890

100

250

205

8 × ⌀25

36

100

1 168

150

355

295

12 × ⌀33

44

150

1 304/1 498 1)

200

405

345

12 × ⌀33

50

200

1 459

Surface roughness (flange): Ra 1.6 to 3.2 µm
1)86Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Endress+Hauser

Proline Promass Q 500

Flange JIS B2220: 63K
1.4404 (F316/F316L)
Order code for "Process connection", option NHS
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

140

100

4 × ⌀23

27

22

494

50

185

145

8 × ⌀23

34

48

775

80

230

185

8 × ⌀25

40

73

915

100

270

220

8 × ⌀27

44

98

1 168

150

365

305

12 × ⌀33

54

146

1 334/1 528 1)

200

425

360

12 × ⌀33

60

190.9

1 479

Surface roughness (flange): Ra 1.6 to 3.2 µm
1)Endress+Hauser

Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

87Proline Promass Q 500

Fixed flange DIN 11864-2

X

E

B

A

C

X

D
L
A0015627

 39

Detail X: Asymmetrical process connection; the part shown in blue is provided by the supplier.

Length tolerance for dimension L in mm:+1.5 / –2.0

Flange DIN11864-2 Form A, for pipe according to DIN11866 series A, flange with notch
1.4404 (316/316L)
Order code for "Process connection", option KCS
DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

25

70

53

4 × ⌀9

10

26

454

50

94

77

4 × ⌀9

10

50

720

80

133

112

8 × ⌀11

12

81

900

100

159

137

8 × ⌀11

14

100

1 128

3A version available: order code for "Additional approval", option LP in conjunction with
Ramax = 0.76 µm: order code for "Measuring tube material", option SB

Flange DIN11864-2 Form A, for pipe according to DIN11866 series A, flange with notch with nominal
diameter reduction
1.4404 (316/316L)
Order code for "Process connection", option KAS
FlangeDN[mm]

DeviceReduction
to DN
[mm]

A
[mm]

B
[mm]

C
[mm]

D
[mm]

E
[mm]

L
[mm]

40

25

82

65

4 × ⌀9

10

38

454

3A version available: order code for "Additional approval", option LP in conjunction with
Ramax = 0.76 µm: order code for "Measuring tube material", option SB

88

Endress+Hauser

Proline Promass Q 500

Clamp connections

B

A

Tri-Clamp

L
A0015625

Length tolerance for dimension L in mm:+1.5 / –2.0
Tri-Clamp for pipe according to DIN 11866 series C
1.4404 (316/316L)
Order code for "Process connection", option FTS
DN
[mm]

Clamp
[in]

A
[mm]

B
[mm]

L
[mm]

25

1

50.4

22.1

434

50

2

63.9

47.5

720

80

3

90.9

72.9

900

100

4

118.9

97.4

1 128

3A version available: order code for "Additional approval", option LP in conjunction with
Ramax = 0.76 µm: order code for "Measuring tube material", option SB

Tri-Clamp (1½), for pipe according to DIN 11866 series C with nominal diameter reduction
1.4404 (316L)
Order code for "Process connection", option FAS
Tri-Clamp
DN[mm]

DeviceReduction
to DN
[mm]

Clamp
[in]

A
[mm]

B
[mm]

L
[mm]

40

25

1½ 1)

50.4

34.80

434

3A version available: order code for "Additional approval", option LP in conjunction with
Ramax = 0.76 µm: order code for "Measuring tube material", option SB
1)

Endress+Hauser

The connection corresponds to the hygienic clamp dimensions according to ASME BPE.

89

Proline Promass Q 500

Compression fittings

B

A

Thread DIN 11851, DIN11864-1, SMS 1145

L
A0015628

Length tolerance for dimension L in mm:+1.5 / –2.0
Thread DIN 11851, for pipe according to DIN11866, series A
1.4404 (316/316L)
Order code for "Process connection", option FMW
DN
[mm]

A
[in]

B
[mm]

L
[mm]

25

Rd 52 × ¹⁄₆

26

434

50

Rd 78 × ¹⁄₆

50

720

80

Rd 110 × ¹⁄₄

81

900

100

Rd 130 × ¹⁄₄

100

1 128

3A version available: order code for "Additional approval", option LP in conjunction with
Ramax = 0.76 µm: order code for "Measuring tube material", option SB

Thread DIN11864-1 Form A, for pipe according to DIN11866, series A
1.4404 (316/316L)
Order code for "Process connection", option FLW
DN
[mm]

A
[in]

B
[mm]

L
[mm]

25

Rd 52 × ¹⁄₈

26

434

50

Rd 78 × ¹⁄₆

50

720

80

Rd 110 × ¹⁄₄

81

900

100

Rd 130 × ¹⁄₄

100

1 128

3A version available: order code for "Additional approval", option LP in conjunction with
Ramax = 0.76 µm: order code for "Measuring tube material", option SB

Thread SMS 1145
1.4404 (316/316L)
Order code for "Process connection", option SCS

90

DN
[mm]

A
[in]

B
[mm]

L
[mm]

25

Rd 40 × ¹⁄₆

22.6

434

50

Rd 70 × ¹⁄₆

48.6

720

80

Rd 98 × ¹⁄₆

72.9

900

Endress+Hauser

Proline Promass Q 500

Thread SMS 1145
1.4404 (316/316L)
Order code for "Process connection", option SCS
DN
[mm]

A
[in]

B
[mm]

L
[mm]

100

Rd 132 × ¹⁄₆

97.6

1 128

3A version available: order code for "Additional approval", option LP in conjunction with
Ramax = 0.76 µm: order code for "Measuring tube material", option SB

Accessories
Rupture disk/purge connections
45°
1

1;2

L

ca. 75 (approx. 2.95)

A

SW 1"
½" NPT½" NPT35 (1.38)

A

1

2A0030349 40
1
2

Endress+Hauser

Engineering unit mm (in)Connection nipple for purge connections: order code for "Sensor options", option CH "Purge connection"
Connection nipple with rupture disk: order code for "Sensor option", option CA "Rupture disk"

DN

A

L

[mm]

[mm]

[mm]

25

32

240

50

53

452

80

80

380

100

106

584

150

118.5

584

200

158.5

584

250

174.3

584

91

Proline Promass Q 500

203 (8.0)

243 (9.6)

213 (8.4)

39 (1.5)

Weather protection coverA0029552

 41

Weather protection cover for Proline 500 – digital; engineering unit mm (in)280 (11.0)
134 (5.3)

12 (0.47)

30 (1.18)

48 (1.9)

146 (5.75)

255 (10.0)

A0029553

 42

Weather protection cover for Proline 500; engineering unit mm (in)External WLAN antenna
The external WLAN antenna is not suitable for use in hygienic applications.
Proline 500 – digital

105 (4.1)

External WLAN antenna mounted on device

A0033607

 43

92

Engineering unit mm (in)Endress+Hauser

Proline Promass Q 500

External WLAN antenna mounted with cable

1500 (59.1)

70 (2.8)

72 (2.8)

The external WLAN antenna can be mounted separately from the transmitter if the transmission/
reception conditions at the transmitter mounting location are poor.

A0033606

 44

Engineering unit mm (in)Proline 500
External WLAN antenna mounted on device

105 (4.1)

68 (2.7)

173 (6.8)
A0028923

 45

Engineering unit mm (in)External WLAN antenna mounted with cable
The external WLAN antenna can be mounted separately from the transmitter if the transmission/
reception conditions at the transmitter mounting location are poor.

Endress+Hauser

93

1500 (59.1)

72 (2.8)

Proline Promass Q 500

70 (2.8)

A0033597

 46

94

Engineering unit mm (in)Endress+Hauser

Proline Promass Q 500

Dimensions in US units

Housing of Proline 500 – digital transmitter
Non-hazardous area or hazardous area: Zone 2; Class I, Division 2
A

F

Q

N

P

G

A0033789

Order code for "Transmitter housing", option A "Aluminum, coated" and order code for "Integrated ISEM
electronics", option A "Sensor"
A
[in]

F
[in]

G
[in]

N
[in]

P
[in]

Q
[in]

6.57

9.13

3.50

7.36

0.94

0.83

Order code for "Transmitter housing", option D "Polycarbonate" and order code for "Integrated ISEM
electronics", option A "Sensor"
A
[in]

F
[in]

G
[in]

N
[in]

P
[in]

Q
[in]

6.97

9.21

3.50

7.76

0.67

0.87

Housing of Proline 500 transmitter
Hazardous area: Zone 2; Class I, Division 2 or Zone 1; Class I, Division 1
A

T
C

G

Q

F

B

Q
A0033788

Order code for "Transmitter housing", option A "Aluminum, coated" and order code for "Integrated ISEM
electronics", option B "Transmitter"

1)

Endress+Hauser

A
[in]

B
[in]

C
[in]

F 1)
[in]

G
[in]

Q
[in]

T
[in]

7.40

3.35

4.06

12.5

8.54

5.12

9.41

DN ≥ 6": values +2.83 in

95

Proline Promass Q 500

Order code for "Transmitter housing", option L "Cast, stainless" and order code for "Integrated ISEM
electronics", option B "Transmitter"
A
[in]

B
[in]

C
[in]

F 1)
[in]

G
[in]

Q
[in]

T
[in]

7.40

3.35

4.06

11.6

8.54

5.12

9.41

1)

DN ≥ 6: values +2.83 in

Sensor connection housing
A

G
C

D

K

F

E

B

L

M
A0033787

Order code for "Sensor connection housing", option A "Aluminum, coated"
DN

A 1)

B 1)

C

D

E

F

G

K

L

M

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

1

5.83

3.70

2.13

8.54

10.1

18.64

5.35

0.60

2)

2.87

2

5.83

3.70

2.13

16.06

10.91

26.97

5.35

1.10

2)

4.53

1.70

2)

6.65

5.35

2.71

2)

8.66

3

5.83

4
1)
2)

5.83

3.70
3.70

2.13
2.13

20.63
25.79

11.97
12.99

32.6
38.78

5.35

Depending on the cable gland used: values up to +1.18 in
Depending on the process connection

Order code for "Sensor connection housing", option B "Stainless"
DN

A 1)

B

C

D

E

F

G

K

L

M

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

0.60

2)

2.87

1.10

2)

4.53

1.70

2)

6.65

2.71

2)

8.66

2.30

2)

9.61

2.81

2)

12.75

3.69

2)

14

1
2
3
4
6
8
10
1)
2)

96

5.39
5.39
5.39
5.39
5.39
5.39
5.39

3.07
3.07
3.07
3.07
3.07
3.07
3.07

2.32
2.32
2.32
2.32
2.32
2.32
2.32

8.54
16.06
20.63
25.79
24.65
31.10
34.92

9.9
10.71

18.44
26.77

11.77
12.8
13.23
14.76
15.43

32.4
38.58
37.87
45.91
50.35

5.28
5.28
5.28
5.28
5.28
5.28
5.28

Depending on the cable gland used: values up to +1.18 in
Depending on the process connection

Endress+Hauser

Proline Promass Q 500

Order code for "Sensor connection housing", option C "Ultra-compact hygienic, stainless"
DN

A 1)

B 1)

C

D

E

F

G

K

L

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

1

4.88

2.68

2.20

8.54

9.9

18.44

4.41

0.60

2)

2.87

1.10

2)

4.53

1.70

2)

6.65

2.71

2)

8.66

2

4.88

3

4.88

4

4.88

1)
2)

2.68

2.20

2.68

2.20

2.68

2.20

16.06
20.63

10.71

26.77

11.77

25.79

12.8

32.4
38.58

4.41
4.41
4.41

M

Depending on the cable gland used: values up to +1.18 in
Depending on the process connection

Order code for "Sensor connection housing", option L "Cast, stainless"
DN

A 1)

B

C

D

E

F

G

K

L

M

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

[in]

1

5.71

3.39

2.32

8.54

11.02

19.57

5.35

0.60

2)

2.87

2

5.71

3.39

2.32

16.06

11.81

27.87

5.35

1.10

2)

4.53

3

5.71

3.39

2.32

20.63

12.87

33.5

5.35

1.70

2)

6.65

2.71

2)

8.66

23.03

2)

9.61

28.15

2)

12.76

36.93

2)

14.02

4

5.71

6

5.71

8

5.71

10
1)
2)

5.71

3.39

2.32

3.39

2.32

3.39

2.32

3.39

2.32

25.79
24.65
31.10

13.9
14.29
15.87

34.92

16.5

39.69
38.94
46.97
51.42

5.35
5.35
5.35
5.35

Depending on the cable gland used: values up to +1.18 in
Depending on the process connection

B
E

A

C

Fixed flange connections ASME B16.5

L

D

A0015621

Length tolerance for dimension L in inch:+0.06 / –0.08
Flange according to ASME B16.5: Class 150
1.4404 (F316/F316L)
Order code for "Process connection", option AAS

Endress+Hauser

DN
[in]

A
[in]

B
[in]

C
[in]

D
[in]

E
[in]

L
[in]

1

4.33

3.13

4 × Ø0.62

0.56

1.05

17.32

2

5.91

4.75

4 × Ø0.75

0.75

2.07

28.15

3

7.48

6.00

4 × Ø0.75

0.94

3.07

33.07

97

Proline Promass Q 500

Flange according to ASME B16.5: Class 150
1.4404 (F316/F316L)
Order code for "Process connection", option AAS
DN
[in]

A
[in]

B
[in]

C
[in]

D
[in]

E
[in]

L
[in]

4

9.06

7.50

8 × Ø0.75

0.94

4.03

44.41

6

11.02

9.5

8 × ⌀0.87

1.02

6.07

47.36/55.04 1)

8

13.58

11.75

8 × ⌀0.87

1.14

7.98

56.02

10

15.94

14.25

12 × ⌀1

1.2

10.02

72.13

Surface roughness (flange): Ra 126 to 248 µin
1)Installed length such as Promass F, DN 6" (order code for "Sensor option", option CN)

Flange according to ASME B16.5: Class 150 with reduction in nominal diameter
1.4404 (F316/F316L)
FlangeDN[in]

DeviceReduction
to DN
[in]

Order code for"Processconnection",
option

A
[in]

B
[in]

C
[in]

D
[in]

E
[in]

L
[in]

3

2

AJS

7.48

6

4 × ⌀0.75

0.94

3.07

28.35

4

3

ALS

9.06

7.5

8 × ⌀0.75

0.94

4.03

34.41

6

4

ANS

11.02

9.5

8 × ⌀0.88

1

6.07

45.94

8

6

APS

13.58

11.75

8 × ⌀0.87

1.14

7.98

49.84

10

8

AVS

15.94

14.25

12 × ⌀1

1.2

10.02

55.43/
72.13 1)

12

10

AXS

19.09

17

12 × ⌀1

1.27

12

76.18

Surface roughness (flange): Ra 126 to 248 µin
1)Installed length such as Promass F, DN 10" (order code for "Sensor option", option CO)

Flange according to ASME B16.5: Class 300
1.4404 (F316/F316L)
Order code for "Process connection", option ABS
DN
[in]

A
[in]

B
[in]

C
[in]

D
[in]

E
[in]

L
[in]

1

4.92

3.50

4 × Ø0.75

0.69

1.05

17.32

2

6.50

5.00

8 × Ø0.75

0.88

2.07

28.15

3

8.27

6.63

8 × Ø0.88

1.12

3.07

33.07

4

10.04

7.87

8 × Ø0.88

1.25

4.03

44.41

6

12.6

10.63

12 × ⌀0.87

1.46

6.07

48.15/55.79 1)

8

14.96

13

12 × ⌀1

1.64

7.98

56.81

10

17.52

15.25

16 × ⌀1.12

1.89

10.02

73.35

Surface roughness (flange): Ra 126 to 248 µin
1)98Installed length such as Promass F, DN 150 (order code for "Sensor option", option CN)

Endress+Hauser

Proline Promass Q 500

Flange according to ASME B16.5: Class 300 with reduction in nominal diameter
1.4404 (F316/F316L)
FlangeDN[in]

DeviceReduction
to DN
[in]

Order code for"Processconnection",
option

A
[in]

B
[in]

C
[in]

D
[in]

E
[in]

L
[in]

3

2

AKS

8.27

6.63

8 × ⌀0.88

1.12

3.07

28.82

4

3

AMS

10.04

7.87

8 × ⌀0.88

1.25

4.03

35.2

6

4

AOS

12.6

10.63

12 × ⌀0.88

1.44

6.07

46.73

8

6

AQS

14.96

13

12 × ⌀1

1.64

7.98

49.84

10

8

AWS

17.52

14.74

16 × ⌀1.13

1.89

10.02

56.65/
73.35 1)

12

10

AZS

20.47

17.75

16 × ⌀1.25

2.02

12

76.18

Surface roughness (flange): Ra 126 to 248 µin
1)Installed length such as Promass F, DN 10" (order code for "Sensor option", option CO)

Flange according to ASME B16.5: Class 600
1.4404 (F316/F316L)
Order code for "Process connection", option ACS
DN
[in]

A
[in]

B
[in]

C
[in]

D
[in]

E
[in]

L
[in]

1

4.92

3.50

4 × Ø0.75

0.94

0.96

19.29

2

6.50

5.00

8 × Ø0.75

1.25

1.94

29.21

3

8.27

6.63

8 × Ø0.87

1.57

2.90

35.59

4

10.83

8.50

8 × Ø1.00

1.91

3.83

45.59

6

13.98

11.5

12 × ⌀1.12

2.15

6.07

50.12/57.76 1)

8

16.54

13.75

12 × ⌀1.25

2.46

7.98

59.02

10

20.08

17

16 × ⌀1.38

2.78

10.02

76.61

Surface roughness (flange): Ra 126 to 248 µin
1)Endress+Hauser

Installed length such as Promass F, DN 6" (order code for "Sensor option", option CN)

99

Proline Promass Q 500

Accessories
Rupture disk/purge connections
45°
1

1;2

L

ca. 75 (approx. 2.95)

1

A

SW 1"
½" NPT½" NPT35 (1.38)

A

2A0030349

 47
1
2

Engineering unit mm (in)Connection nipple for purge connections: order code for "Sensor options", option CH "Purge connection"
Connection nipple with rupture disk: order code for "Sensor option", option CA "Rupture disk"

DN

A

L

[in]

[in]

[in]

1

1.26

9.45

2

2.09

17.8

3

3.15

14.96

4

4.17

22.99

6

4.67

22.99

8

6.24

22.99

10

6.86

22.99

203 (8.0)

243 (9.6)

213 (8.4)

39 (1.5)

Weather protection coverA0029552

 48

100

Weather protection cover for Proline 500 – digital; engineering unit mm (in)Endress+Hauser

Proline Promass Q 500

280 (11.0)
134 (5.3)

12 (0.47)

30 (1.18)

48 (1.9)

146 (5.75)

255 (10.0)

A0029553

 49

Weather protection cover for Proline 500; engineering unit mm (in)External WLAN antenna
The external WLAN antenna is not suitable for use in hygienic applications.
Proline 500 – digital

105 (4.1)

External WLAN antenna mounted on device

A0033607

 50

Engineering unit mm (in)External WLAN antenna mounted with cable
The external WLAN antenna can be mounted separately from the transmitter if the transmission/
reception conditions at the transmitter mounting location are poor.

Endress+Hauser101

1500 (59.1)

70 (2.8)

72 (2.8)

Proline Promass Q 500

A0033606

 51

Engineering unit mm (in)Proline 500
External WLAN antenna mounted on device

105 (4.1)

68 (2.7)

173 (6.8)
A0028923

 52

Engineering unit mm (in)External WLAN antenna mounted with cable

1500 (59.1)

72 (2.8)

The external WLAN antenna can be mounted separately from the transmitter if the transmission/
reception conditions at the transmitter mounting location are poor.

70 (2.8)

A0033597

 53

102

Engineering unit mm (in)Endress+Hauser

Proline Promass Q 500

WeightAll values (weight exclusive of packaging material) refer to devices with EN/DIN PN 40 flanges.
Transmitter• Proline 500 – digital polycarbonate: 1.4 kg (3.1 lbs)
• Proline 500 – digital aluminum: 2.4 kg (5.3 lbs)
• Proline 500 aluminum: 6.5 kg (14.3 lbs)
DN ≥ 150 (6"): 9 kg (19.8 lbs)
• Proline 500 cast, stainless: 15.6 kg (34.4 lbs)
DN ≥ 150 (6"): 18.5 kg (40.8 lbs)
Sensor• Sensor with cast connection housing version, stainless: +3.7 kg (+8.2 lbs)
• Sensor with aluminum connection housing version:
Weight in SI units
DN[mm]

Weight [kg]

25

11

50

33

80

60

100

149

150

166

200

296

250

483

Weight in US units

Materials

DN[in]

Weight [lbs]

1

24

2

73

3

132

4

329

6

366

8

653

10

1 065

Transmitter housing
Housing of Proline 500 – digital transmitter
Order code for "Transmitter housing":• Option A "Aluminum coated": aluminum, AlSi10Mg, coated• Option D "Polycarbonate": polycarbonate
Housing of Proline 500 transmitter
Order code for "Transmitter housing":• Option A "Aluminum coated": aluminum, AlSi10Mg, coated• Option L "Cast, stainless": cast, stainless steel, 1.4409 (CF3M) similar to 316L
Window material
Order code for "Transmitter housing":• Option A "Aluminum, coated": glass• Option D "Polycarbonate": plastic• Option L "Cast, stainless": glass

Endress+Hauser

103

Proline Promass Q 500

Fastening components for mounting on a post
• Screws, threaded bolts, washers, nuts: stainless A2 (chrome-nickel steel)• Metal plates: stainless steel, 1.4301 (304)
Sensor connection housing
Order code for "Sensor connection housing":• Option A "Aluminum coated": aluminum, AlSi10Mg, coated• Option B "Stainless":• Stainless steel 1.4301 (304)• Optional: Order code for "Sensor feature", option CC "Hygienic version, for maximum corrosion
resistance": stainless steel, 1.4404 (316L)• Option C "Ultra-compact, stainless":• Stainless steel 1.4301 (304)• Optional: Order code for "Sensor feature", option CC "Hygienic version, for maximum corrosion
resistance": stainless steel, 1.4404 (316L)• Option L "Cast, stainless": 1.4409 (CF3M) similar to 316L
Cable entries/cable glands

1

2
3
4
A0028352

 54
1
2
3
4

Possible cable entries/cable glands

Female thread M20 × 1.5
Cable gland M20 × 1.5
Adapter for cable entry with female thread G ½" or NPT ½"
Device plug

Cable entries and adapters

Material

Cable gland M20 × 1.5

Plastic• Adapter for cable entry with female thread G ½"
• Adapter for cable entry with female thread NPT ½"

Nickel-plated brassavailable for certain device versions: •OnlyOrdercode for "Transmitter housing":• Option A "Aluminum, coated"• Option D "Polycarbonate"• Order code for "Sensor connection housing":• Proline 500 – digital:
Option A "Aluminum coated"
Option B "Stainless"
Option L "Cast, stainless"• Proline 500:
Option B "Stainless"
Option L "Cast, stainless"104

Endress+Hauser

Proline Promass Q 500

Cable entries and adapters

Material• Adapter for cable entry with female thread G ½"
• Adapter for cable entry with female thread NPT ½"

Stainless steel, 1.4404 (316L)

available for certain device versions: •OnlyOrdercode for "Transmitter housing":
Option L "Cast, stainless"• Order code for "Sensor connection housing":
Option L "Cast, stainless"
Adapter for device plug

Stainless steel, 1.4404 (316L)

plug for digital communication:
 Device
Only available for certain device versions →  34.

Device plug
Electrical connection

Material

Plug M12x1• Socket: Stainless steel, 1.4404 (316L)• Contact housing: Polyamide• Contacts: Gold-plated brass

Connecting cable
UV rays can impair the cable outer sheath. Protect the cable from exposure to sun as much as
possible.
Connecting cable for sensor - Proline 500 – digital transmitter
PVC cable with copper shield
Connecting cable for sensor - Proline 500 transmitter• PUR cable with copper shield• Devices with order code for «Approval; transmitter; sensor», options AA, BS, CS, CZ, GR, GS, MS,
NS, UR, US: PVC cable with copper shield
Measuring tubes
Stainless steel, 1.4404 (316/316L); manifold: stainless steel, 1.4404 (316/316L)
Process connections
Flanges according to EN 1092-1 (DIN 2501) / according to ASME B16.5 / as per JIS B2220:
Stainless steel, 1.4404 (F316/F316L)
Available process connections→  106
Seals
Welded process connections without internal seals
Accessories
Protective cover
Stainless steel, 1.4404 (316L)
External WLAN antenna• Antenna: ASA plastic (acrylic ester-styrene-acrylonitrile) and nickel-plated brass• Adapter: Stainless steel and nickel-plated brass• Cable: Polyethylene• Plug: Nickel-plated brass• Angle bracket: Stainless steel

Endress+Hauser

105

Proline Promass Q 500

Process connections

Fixed flange connections:• EN 1092-1 (DIN 2501) flange• EN 1092-1 (DIN 2512N) flange• ASME B16.5 flange• JIS B2220 flange
Process connection materials →  105

Surface roughness

All data refer to parts in contact with the medium. The following surface roughness categories canbe ordered.
• Not polished• Ramax = 0.76 µm (30 µin)
• Ramax = 0.38 µm (15 µin)

Operability
Operating concept

Operator-oriented menu structure for user-specific tasks• Commissioning• Operation• Diagnostics• Expert level
Fast and safe commissioning• Guided menus ("Make-it-run" wizards) for applications• Menu guidance with brief descriptions of the individual parameter functions• Access to the device via Web server• WLAN access to the device via mobile handheld terminal, tablet or smart phone
Reliable operation• Operation in local language• Uniform operating philosophy applied to device and operating tools• If replacing electronic modules, transfer the device configuration via the integrated memory(HistoROM backup) which contains the process and measuring device data and the event logbook.
No need to reconfigure.
Efficient diagnostic behavior increases measurement availability• Troubleshooting measures can be called up via the device and in the operating tools• Diverse simulation options, logbook for events that occur and optional line recorder functions

Languages

Can be operated in the following languages:• Via local operation
English, German, French, Spanish, Italian, Dutch, Portuguese, Polish, Russian, Turkish, Chinese,
Japanese, Korean, Vietnamese, Czech, Swedish• Via Web browser
English, German, French, Spanish, Italian, Dutch, Portuguese, Polish, Russian, Turkish, Chinese,
Japanese, Vietnamese, Czech, Swedish• Via "FieldCare", "DeviceCare" operating tool: English, German, French, Spanish, Italian, Chinese,
Japanese

Local operation

Via display module
Equipment:• Order code for "Display; operation", option F "4-line, illuminated, graphic display; touch control"• Order code for "Display; operation", option G "4-line, illuminated, graphic display; touch control +WLAN"
Information about WLAN interface →  113

106

Endress+Hauser

Proline Promass Q 500

1

2

A0028232

 55
1
2

Operation with touch control

Proline 500 – digital
Proline 500

Display elements• 4-line, illuminated, graphic display• White background lighting; switches to red in event of device errors• Format for displaying measured variables and status variables can be individually configured• Permitted ambient temperature for the display: –20 to +60 °C (–4 to +140 °F)
The readability of the display may be impaired at temperatures outside the temperature range.
Operating elements• External operation via touch control (3 optical keys) without opening the housing: , , 
• Operating elements also accessible in the various zones of the hazardous area
Remote operation

Via HART protocol
This communication interface is available in device versions with a HART output.

8
4

1

2

7

3

5

6
A0028747

 56
1
2
3

4
5
6
7
8

Endress+Hauser

Options for remote operation via HART protocol (active)

Control system (e.g. PLC)
Field Communicator 475
Computer with Web browser (e.g. Internet Explorer) for accessing the integrated device Web server orcomputer with operating tool (e.g. FieldCare, DeviceCare, AMS Device Manager, SIMATIC PDM) with COM
DTM "CDI Communication TCP/IP"
Commubox FXA195 (USB)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
VIATOR Bluetooth modem with connecting cable
Transmitter107

Proline Promass Q 500

10
3

2

6

1

4

9

5

7

8
A0028746

 57
1
2
3
4
5

6
7
8
9
10

Options for remote operation via HART protocol (passive)

Control system (e.g. PLC)
Transmitter power supply unit, e.g. RN221N (with communication resistor)
Connection for Commubox FXA195 and Field Communicator 475
Field Communicator 475
Computer with Web browser (e.g. Internet Explorer) for accessing the integrated device Web server orcomputer with operating tool (e.g. FieldCare, DeviceCare, AMS Device Manager, SIMATIC PDM) with COM
DTM "CDI Communication TCP/IP"
Commubox FXA195 (USB)
Field Xpert SFX350 or SFX370
Field Xpert SMT70
VIATOR Bluetooth modem with connecting cable
Transmitter

Via FOUNDATION Fieldbus network
This communication interface is available in device versions with FOUNDATION Fieldbus.
1

1

2

3
4
5
6

7

8

9

6

7

8

9
A0028837

 58
1
2
3
4
5
6
7
8
9

108

Options for remote operation via FOUNDATION Fieldbus network

Automation system
Computer with FOUNDATION Fieldbus network card
Industry network
High Speed Ethernet FF-HSE network
Segment coupler FF-HSE/FF-H1
FOUNDATION Fieldbus FF-H1 network
Power supply FF-H1 network
T-box
Measuring device

Endress+Hauser

Proline Promass Q 500

Via PROFIBUS DP network
This communication interface is available in device versions with PROFIBUS DP.
1

2

3

4

4

4
A0020903

 59
1
2
3
4

Options for remote operation via PROFIBUS DP network

Automation system
Computer with PROFIBUS network card
PROFIBUS DP network
Measuring device

Via PROFIBUS PA network
This communication interface is available in device versions with PROFIBUS PA.
1

2

3
4
5

6

7
A0028838

 60
1
2
3
4
5
6
7

Options for remote operation via PROFIBUS PA network

Automation system
Computer with PROFIBUS network card
PROFIBUS DP network
Segment coupler PROFIBUS DP/PAPROFIBUS PA network
T-box
Measuring device

Via Modbus RS485 protocol
This communication interface is available in device versions with a Modbus-RS485 output.

Endress+Hauser

109

Proline Promass Q 500

31

2A0029437

 61
1
2
3

Options for remote operation via Modbus-RS485 protocol (active)

Control system (e.g. PLC)
Computer with Web browser (e.g. Internet Explorer) for accessing the integrated device Web server or with
operating tool (e.g. FieldCare, DeviceCare) with COM DTM "CDI Communication TCP/IP" or Modbus DTM
Transmitter

Via EtherNet/IP network
This communication interface is available in device versions with EtherNet/IP.
Star topology
1

2

3

4

5

5

5
A0032078

 62
1
2
3
4
5

Options for remote operation via EtherNet/IP network: star topology

Automation system, e.g. "RSLogix" (Rockwell Automation)Workstation for measuring device operation: with Custom Add-On Profile for "RSLogix 5000" (Rockwell
Automation) or with Electronic Data Sheet (EDS)
Computer with Web browser (e.g. Internet Explorer) for accessing the integrated Web server or computerwith operating tool (e.g. FieldCare, DeviceCare) with COM DTM "CDI Communication TCP/IP"
Standard Ethernet switch, e.g. Scalance X204 (Siemens)
Measuring device

Ring topology
The device is integrated via the terminal connection for signal transmission (output 1) and theservice interface (CDI-RJ45).

110

Endress+Hauser

Proline Promass Q 500

1

2

3

4

5

5

5
A0033725

 63
1
2
3
4
5

Options for remote operation via EtherNet/IP network: ring topology

Automation system, e.g. "RSLogix" (Rockwell Automation)Workstation for measuring device operation: with Custom Add-On Profile for "RSLogix 5000" (Rockwell
Automation) or with Electronic Data Sheet (EDS)
Computer with Web browser (e.g. Internet Explorer) for accessing the integrated Web server or computerwith operating tool (e.g. FieldCare, DeviceCare) with COM DTM "CDI Communication TCP/IP"
Standard Ethernet switch, e.g. Scalance X204 (Siemens)
Measuring device

Via PROFINET network
This communication interface is available in device versions with PROFINET.
Star topology
1

3

4

2

4

4
A0026545

 64
1
2
3
4

Options for remote operation via PROFINET network: star topology

Automation system, e.g. Simatic S7 (Siemens)
Computer with Web browser (e.g. Internet Explorer) for accessing the integrated Web server or computerwith operating tool (e.g. FieldCare, DeviceCare, SIMATIC PDM) with COM DTM "CDI Communication TCP/IP"
Standard Ethernet switch, e.g. Scalance X204 (Siemens)
Measuring device

Ring topology
The device is integrated via the terminal connection for signal transmission (output 1) and theservice interface (CDI-RJ45).

Endress+Hauser

111

Proline Promass Q 500

1

2

3

4

4

4
A0033719

 65
1
2
3
4

Options for remote operation via PROFINET network: ring topology

Automation system, e.g. Simatic S7 (Siemens)
Computer with Web browser (e.g. Internet Explorer) for accessing the integrated Web server or computerwith operating tool (e.g. FieldCare, DeviceCare, SIMATIC PDM) with COM DTM "CDI Communication TCP/IP"
Standard Ethernet switch, e.g. Scalance X204 (Siemens)
Measuring device

Via APL network
1

2

3

4
5

6

6

6
A0046117

 66
1
2
3
4
5
6

112

Options for remote operation via APL network

Automation system, e.g. Simatic S7 (Siemens)
Ethernet switch, e.g. Scalance X204 (Siemens)
Computer with Web browser (e.g. Internet Explorer) for access to integrated Web server or computer with
operating tool (e.g. FieldCare or DeviceCare with PROFINET COM DTM or SIMATIC PDM with FDI-Package)
APL power switch (optional)
APL field switch
Measuring device

Endress+Hauser

Proline Promass Q 500

Service interface

Via service interface (CDI-RJ45)
A point-to-point connection can be established via onsite device configuration. With the housing
open, the connection is established directly via the service interface (CDI-RJ45) of the device.
An adapter for RJ45 to the M12 plug is optionally available:
Order code for "Accessories", option NB: "Adapter RJ45 M12 (service interface)"
The adapter connects the service interface (CDI-RJ45) to an M12 plug mounted in the cableentry. The connection to the service interface can be established via an M12 plug without
opening the device.
Proline 500 – digital transmitter

1
open
press

3

2

A0029163

 67
1

2
3

Connection via service interface (CDI-RJ45)

Computer with Web browser (e.g. Microsoft Internet Explorer, Microsoft Edge) for accessing the integrated
Web server or with "FieldCare", "DeviceCare" operating tool with COM DTM "CDI Communication TCP/IP" or
Modbus DTM
Standard Ethernet connecting cable with RJ45 plug
Service interface (CDI-RJ45) of the measuring device with access to the integrated Web server

Proline 500 transmitter

1
3

2

A0027563

 68
1

2
3

Connection via service interface (CDI-RJ45)

Computer with Web browser (e.g. Microsoft Internet Explorer, Microsoft Edge) for accessing the integrated
Web server or with "FieldCare", "DeviceCare" operating tool with COM DTM "CDI Communication TCP/IP" or
Modbus DTM
Standard Ethernet connecting cable with RJ45 plug
Service interface (CDI-RJ45) of the measuring device with access to the integrated Web server

Via WLAN interface
The optional WLAN interface is available on the following device version:
Order code for "Display; operation", option G "4-line, illuminated; touch control + WLAN"

Endress+Hauser

113

Proline Promass Q 500

3

4

3

4

1

1

2

2

5

6

3

3

4

4

7
A0034569

1
2
3
4
5
6
7

Transmitter with integrated WLAN antenna
Transmitter with external WLAN antenna
LED lit constantly: WLAN reception is enabled on measuring device
LED flashing: WLAN connection established between operating unit and measuring device
Computer with WLAN interface and Web browser (e.g. Microsoft Internet Explorer, Microsoft Edge) foraccessing the integrated device Web server or with operating tool (e.g. FieldCare, DeviceCare)
Mobile handheld terminal with WLAN interface and Web browser (e.g. Microsoft Internet Explorer, Microsoft
Edge) for accessing the integrated device Web server or operating tool (e.g. FieldCare, DeviceCare)
Smart phone or tablet (e.g. Field Xpert SMT70)

Function

WLAN: IEEE 802.11 b/g (2.4 GHz)
• Access Point with DHCP server (factory setting)• Network

Encryption

WPA2-PSK AES-128 (in accordance with IEEE 802.11i)

Configurable WLAN channels

1 to 11Degree of protection

IP67

Available antennas• Internal antenna• External antenna (optional)
In the event of poor transmission/reception conditions at the place ofinstallation.
Available as an accessory .

 Only 1 antenna is active at any one time!

Network integration

114

Range

• Internal antenna: typically 10 m (32 ft)• External antenna: typically 50 m (164 ft)

Materials (external antenna)• Antenna: ASA plastic (acrylic ester-styrene-acrylonitrile) and nickelplated brass• Adapter: Stainless steel and nickel-plated brass• Cable: Polyethylene• Plug: Nickel-plated brass• Angle bracket: Stainless steel

Network integration is only available for the HART communication protocol.

Endress+Hauser

Proline Promass Q 500

With the optional "OPC-UA Server" application package, the device can be integrated into an Ethernet
network via the service interface (CDI-RJ45 and WLAN) and communicate with OPC-UA clients. If
the device is used in this way, IT security must be considered.
Transmitters with an Ex de approval may not be connected via the service interface (CDI-RJ45)!
Order code for "Approval transmitter + sensor", options (Ex de):BA, BB, C1, C2, GA, GB, MA, MB, NA, NB
For permanent access to device data and for device configuration via the Web server, the device isincorporated directly in a network via the service interface (CDI-RJ45). In this way, the device can beaccessed any time from the control station. The measured values are processed separately via the
inputs and outputs through the automation system.
1

2

5

3

5

4

6
7
8

5

A0033618

1
2
3
4
5
6
7
8

Automation system, e.g. Simatic S7 (Siemens)
Ethernet switch
Edge Gateway
Cloud
Measuring device
Ethernet network
Measured values via inputs and outputs
Optional WLAN interface

The optional WLAN interface is available on the following device version:
Order code for "Display; operation", option G "4-line, backlit, graphic display; touch control +WLAN"Special Documentation for the OPC-UA Server application package →  131.

Supported operating tools

Endress+Hauser

Different operating tools can be used for local or remote access to the measuring device. Dependingon the operating tool used, access is possible with different operating units and via a variety ofinterfaces.
Supported operating
tools

Operating unit

Interface

Additional information

Web browser

Notebook, PC or tablet
with Web browser• CDI-RJ45 service
Special Documentation for theinterfacedevice →  131• WLAN interface• Ethernet-based
fieldbus (EtherNet/IP,
PROFINET)

DeviceCare SFE100

Notebook, PC or tablet• CDI-RJ45 service
with Microsoft Windowsinterfacesystem• WLAN interface• Fieldbus protocol→  129

FieldCare SFE500

Notebook, PC or tablet• CDI-RJ45 service
with Microsoft Windowsinterfacesystem• WLAN interface• Fieldbus protocol→  129115

Proline Promass Q 500

Supported operating
tools

Operating unit

Interface

Additional information

Field Xpert

SMT70/77/50• All fieldbus protocols• WLAN interface• Bluetooth• CDI-RJ45 service
interface

Operating Instructions
BA01202S

WLAN→  129

SmartBlue app

Smart phone or tablet
with iOs or Android

Device description files:
Use update function ofhandheld terminal

Other operating tools based on FDT technology with a device driver such as DTM/iDTM or
DD/EDD can be used for device operation. These operating tools are available from theindividual manufacturers. Integration into the following operating tools, among others, issupported:• FactoryTalk AssetCentre (FTAC) from Rockwell Automation → www.rockwellautomation.com• Process Device Manager (PDM) from Siemens → www.siemens.com• Asset Management Solutions (AMS) from Emerson → www.emersonprocess.com• FieldCommunicator 375/475 from Emerson → www.emersonprocess.com• Field Device Manager (FDM) from Honeywell → www.process.honeywell.com• FieldMate from Yokogawa → www.yokogawa.com• PACTWare → www.pactware.com
The related device description files are available: www.endress.com → Downloads
Web serverThanks to the integrated Web server the device can be operated and configured via a Web browser
and via the service interface (CDI-RJ45) or via the WLAN interface. The structure of the operating
menu is the same as for the local display. In addition to the measured values, device status
information is also displayed and allows users to monitor the status of the device. Furthermore thedevice data can be managed and the network parameters can be configured.
A device that has a WLAN interface (can be ordered as an option) is required for the WLAN
connection: order code for "Display; operation", option G "4-line, illuminated; touch control + WLAN".
The device acts as an Access Point and enables communication by computer or a mobile handheld
terminal.
Supported functions
Data exchange between the operating unit (such as a notebook for example) and the measuring
device:• Upload the configuration from the measuring device (XML format, configuration backup)• Save the configuration to the measuring device (XML format, restore configuration)• Export event list (.csv file)• Export parameter settings (.csv file or PDF file, document the measuring point configuration)• Export the Heartbeat verification log (PDF file, only available with the "Heartbeat Verification"
application package)• Flash firmware version for device firmware upgrade, for instance• Download driver for system integration• Visualize up to 1000 saved measured values (only available with the Extended HistoROM
application package →  125)Web server special documentation →  131

HistoROM data management

The measuring device features HistoROM data management. HistoROM data management
comprises both the storage and import/export of key device and process data, making operation andservicing far more reliable, secure and efficient.
When the device is delivered, the factory settings of the configuration data are stored as a
backup in the device memory. This memory can be overwritten with an updated data record, forexample after commissioning.

116

Endress+Hauser

Proline Promass Q 500

Additional information on the data storage concept
There are different types of data storage units in which device data are stored and used by the device:

Available data

HistoROM backup

T-DAT

S-DAT

• Event logbook such as diagnostic events forexample• Parameter data record backup• Device firmware package• Driver for system integration for exporting via Web
server, e.g:
• GSD for PROFIBUS DP• GSD for PROFIBUS PA• GSDML for PROFINET• EDS for EtherNet/IP• DD for FOUNDATION Fieldbus• Measured value logging("Extended HistoROM" order
option)• Current parameter data record(used by firmware at run time)• Maximum indicators (min/max
values)• Totalizer values• Sensor data: nominal diameteretc.
• Serial number• Calibration data
• Device configuration (e.g. SW
options, fixed I/O or multi I/O)

Attachable to the user interface
board in the connection
compartment

In the sensor plug in the transmitter
neck part

Storage location Fixed on the user interface board in the connection
compartment

Data backup
Automatic• The most important device data (sensor and transmitter) are automatically saved in the DAT
modules• If the transmitter or measuring device is replaced: once the T-DAT containing the previous device
data has been exchanged, the new measuring device is ready for operation again immediately
without any errors• If the sensor is replaced: once the sensor has been replaced, new sensor data are transferred fromthe S-DAT in the measuring device and the measuring device is ready for operation again
immediately without any errors• If exchanging the electronics module (e.g. I/O electronics module): Once the electronics module
has been replaced, the software of the module is compared against the current device firmware.
The module software is upgraded or downgraded where necessary. The electronics module isavailable for use immediately afterwards and no compatibility problems occur.
ManualAdditional parameter data record (complete parameter settings) in the integrated device memory
HistoROM backup for:• Data backup function
Backup and subsequent restoration of a device configuration in the device memory HistoROM
backup• Data comparison function
Comparison of the current device configuration with the device configuration saved in the device
memory HistoROM backup
Data transmission
Manual• Transfer of a device configuration to another device using the export function of the specificoperating tool, e.g. with FieldCare, DeviceCare or Web server: to duplicate the configuration or tostore in an archive (e.g. for backup purposes)• Transmission of the drivers for system integration via Web server, e.g.:• GSD for PROFIBUS DP• GSD for PROFIBUS PA• GSDML for PROFINET• EDS for EtherNet/IP• DD for FOUNDATION Fieldbus

Endress+Hauser

117

Proline Promass Q 500

Event list
Automatic• Chronological display of up to 20 event messages in the events list
• If the Extended HistoROM application package (order option) is enabled: up to 100 event
messages are displayed in the events list along with a time stamp, plain text description andremedial measures• The events list can be exported and displayed via a variety of interfaces and operating tools e.g.
DeviceCare, FieldCare or Web server
Data logging
ManualIf the Extended HistoROM application package (order option) is enabled:• Record up to 1 000 measured values via 1 to 4 channels• User configurable recording interval• Record up to 250 measured values via each of the 4 memory channels• Export the measured value log via a variety of interfaces and operating tools e.g. FieldCare,
DeviceCare or web server

Certificates and approvals
Current certificates and approvals that are available for the product can be selected via the ProductConfigurator at www.endress.com:

CE mark

1.

Select the product using the filters and search field.

2.

Open the product page.

3.

Select Configuration.

The device meets the legal requirements of the applicable EU Directives. These are listed in thecorresponding EU Declaration of Conformity along with the standards applied.
Endress+Hauser confirms successful testing of the device by affixing to it the CE mark.

UKCA marking

The device meets the legal requirements of the applicable UK regulations (Statutory Instruments).
These are listed in the UKCA Declaration of Conformity along with the designated standards. By
selecting the order option for UKCA marking, Endress+Hauser confirms a successful evaluation andtesting of the device by affixing the UKCA mark.
Contact address Endress+Hauser UK:
Endress+Hauser Ltd.
Floats Road
Manchester M23 9NF
United Kingdom
www.uk.endress.com

RCM mark

The measuring system meets the EMC requirements of the "Australian Communications and Media
Authority (ACMA)".

Ex approval

The measuring device is certified for use in hazardous areas and the relevant safety instructions areprovided in the separate "Safety Instructions" (XA) document. Reference is made to this document on
the nameplate.
The following devices have equipment protection level (EPL) Ga/Gb (Zone 0 in the measuring tube):• Device versions with the order code for "Integrated ISEM electronics", option A and the order codefor "Approval; transmitter; sensor", option BI, BJ, BM or BN.
• Device versions with the order code for "Integrated ISEM electronics", option B and the order codefor "Approval; transmitter; sensor", option BA, BB, BC or BD.
The separate Ex documentation (XA) containing all the relevant explosion protection data isavailable from your Endress+Hauser sales center.

118

Endress+Hauser

Proline Promass Q 500

Proline 500 – digital
ATEX/IECEx
Currently, the following versions for use in hazardous areas are available:
Ex ia
Transmitter

SensorCategory

Type of protection

Category

Type of protection

II(1)G[Ex ia] IIC

II1/2GEx ia IIC T6…T1 Ga/Gb
Ex ia IIB T6…T1 Ga/Gb

II(1)G[Ex ia] IIC

II2G

Ex ia IIC T6…T1 Gb
Ex ia IIB T6…T1 Gb

II3(1)G

Ex ec [ia Ga] IIC T5…T4 Gc

II1/2GEx ia IIC T6…T1 Ga/Gb
Ex ia IIB T6…T1 Ga/Gb

II3(1)G

Ex ec [ia Ga] IIC T5…T4 Gc

II2G

Ex ia IIC T6…T1 Gb
Ex ia IIB T6…T1 Gb

Ex tb
Transmitter

SensorCategory

Type of protection

Category

Type of protection

II(1)D[Ex ia] IIIC

II2D

Ex ia tb IIIC T** °C Db

Non-Ex / Ex ec
Transmitter

SensorCategory

Type of protection

Category

Type of protection

Non - Ex

Non-Ex

II3G

Ex ec IIC T5…T1 Gc

II3G

Ex ec IIC T5…T4 Gc

II3G

Ex ec IIC T5…T1 Gc

CCSAUS

Currently, the following versions for use in hazardous areas are available:
IS (Ex nA, Ex i)
Transmitter

SensorClass I Division 2 Groups A - D

Class I, II, III Division 1 Groups A-G

Class I Division 2 Groups A - D

Class I, II, III Division 1 Groups C-G

NI (Ex nA)
Transmitter

SensorClass I Division 2 Groups A - D

Ex nA / Ex i

Endress+Hauser

Transmitter

SensorClass I, Zone 2 AEx/ Ex nA [ia Ga] IIC T5…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIC T6…T1 Ga/Gb
Class I, Zone 1 AEx/ Ex ia IIB T6…T1 Ga/Gb

Class I, Zone 2 AEx/ Ex nA [ia Ga] IIC T5…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIC T6…T1 Gb
Class I, Zone 1 AEx/ Ex ia IIB T6…T1 Gb

119Proline Promass Q 500

Ex nA
Transmitter

SensorClass I, Zone 2 AEx/ Ex nA IIC T5...T4 Gc

Class I, Zone 2 AEx/ Ex nA IIC T5...T1 Gc

Ex tb
Transmitter

Sensor[AEx / Ex ia ] IIIC

Zone 21 AEx/ Ex ia tb IIIC T** °C Db

Proline 500
ATEX/IECEx
Currently, the following versions for use in hazardous areas are available:
Ex db eb
Transmitter

SensorCategory

Type of protection

Category

Type of protection

II2G

Ex db eb ia IIC T6…T4 Gb

II1/2GEx ia IIC T6…T1 Ga/Gb

II2G

Ex db eb ia IIB T6…T4 Gb

II1/2GEx ia IIB T6…T1 Ga/Gb

II2G

Ex db eb ia IIC T6…T4 Gb

II2G

Ex ia IIC T6…T1 Gb

II2G

Ex db eb ia IIB T6…T4 Gb

II2G

Ex ia IIB T6…T1 Gb

Ex db
Transmitter

SensorCategory

Type of protection

Category

Type of protection

II2G

Ex db ia IIC T6…T4 Gb

II1/2GEx ia IIC T6…T1 Ga/Gb

II2G

Ex db ia IIB T6…T4 Gb

II1/2GEx ia IIB T6…T1 Ga/Gb

II2G

Ex db ia IIC T6…T4 Gb

II2G

Ex ia IIC T6…T1 Gb

II2G

Ex db ia IIB T6…T4 Gb

II2G

Ex ia IIB T6…T1 Gb

Ex tb
Category

II2D

Type of protection
Transmitter

Sensor

Ex tb IIIC T85°C Db

Ex ia tb IIIC T** °C Db

Ex ec
Category

II3G

Type of protection
Transmitter

Sensor

Ex ec IIC T5…T4 Gc

Ex ec IIC T5…T1 Gc

CCSAUS

Currently, the following versions for use in hazardous areas are available:

120

Endress+Hauser

Proline Promass Q 500

IS (Ex i) and XP (Ex d)Transmitter

SensorClass I, II, III Division 1 Groups A-G
Class I, II, III Division 1 Groups C-G

NI (Ex nA)
Transmitter

SensorClass I Division 2 Groups ABCD

Ex deTransmitter

SensorClass I, Zone 1 AEx/ Ex de ia IIC T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIC T6…T1 Ga/Gb

Class I, Zone 1 AEx/ Ex de ia IIB T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIB T6…T1 Ga/Gb

Class I, Zone 1 AEx/ Ex de ia IIC T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIC T6…T1 Gb

Class I, Zone 1 AEx/ Ex de ia IIB T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIB T6…T1 Gb

Transmitter

SensorClass I, Zone 1 AEx/ Ex d ia IIC T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIC T6…T1 Ga/Gb

Class I, Zone 1 AEx/ Ex d ia IIB T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIB T6…T1 Ga/Gb

Class I, Zone 1 AEx/ Ex d ia IIC T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIC T6…T1 Gb

Class I, Zone 1 AEx/ Ex d ia IIB T6…T4 Gb

Class I, Zone 1 AEx/ Ex ia IIB T6…T1 Gb

Transmitter

SensorClass I, Zone 2 AEx/ Ex nA IIC T5...T4 Gc

Class I, Zone 2 AEx/ Ex nA IIC T5...T1 Gc

Transmitter

SensorZone 21 AEx/ Ex tb IIIC T85°C Db

Zone 21 AEx/ Ex ia tb IIIC T** °C Db

Ex d

Ex nA

Ex tb

Hygienic compatibility

Endress+Hauser• 3-A approval• Only measuring devices with the order code for "Additional approval", option LP "3A" have3-A approval.
• The 3-A approval refers to the measuring device.
• When installing the measuring device, ensure that no liquid can accumulate on the outside of
the measuring device.
A remote display module must be installed in accordance with the 3-A Standard.
• Accessories (e.g. heating jacket, weather protection cover, wall holder unit) must be installed inaccordance with the 3-A Standard.
Each accessory can be cleaned. Disassembly may be necessary under certain circumstances.
• EHEDG-tested
Only devices with the order code for "Additional approval", option LT "EHEDG" have been tested andmeet the requirements of the EHEDG.
To meet the requirements for EHEDG certification, the device must be used with processconnections in accordance with the EHEDG position paper entitled "Easy cleanable Pipe couplings
and Process connections" (www.ehedg.org).
• FDA• Food Contact Materials Regulation (EC) 1935/2004

121

Proline Promass Q 500

Pharmaceutical compatibility• FDA 21 CFR 177• USP <87>
• USP <88> Class VI 121 °C• TSE/BSE Certificate of Suitability• cGMP
Devices with the order code for "Test, certificate", option JG "Conformity with cGMP-derived
requirements, declaration" comply with the requirements of cGMP with regard to the surfaces ofparts in contact with the medium, design, FDA 21 CFR material conformity, USP Class VI tests and
TSE/BSE conformity.
A serial number-specific declaration is generated.

Functional safety

The measuring device can be used for flow monitoring systems (min., max., range) up to SIL 2(single-channel architecture; order code for "Additional approval", option LA) and SIL 3 (multichannel architecture with homogeneous redundancy) and is independently evaluated and certified inaccordance with IEC 61508.
The following types of monitoring in safety equipment are possible:• Mass flow
• Volume flow
• Density
Functional Safety Manual with information on the SIL device →  131

HART certification

HART interface
The measuring device is certified and registered by the FieldComm Group. The measuring systemmeets all the requirements of the following specifications:• Certified according to HART 7• The device can also be operated with certified devices of other manufacturers (interoperability)

FOUNDATION Fieldbus
certification

FOUNDATION Fieldbus interface

Certification PROFIBUSPROFIBUS interface

The measuring device is certified and registered by the FieldComm Group. The measuring systemmeets all the requirements of the following specifications:• Certified in accordance with FOUNDATION Fieldbus H1• Interoperability Test Kit (ITK), revision version 6.2.0 (certificate available on request)• Physical Layer Conformance Test
• The device can also be operated with certified devices of other manufacturers (interoperability)

The measuring device is certified and registered by the PNO (PROFIBUS Nutzerorganisation e.V./
PROFIBUS User Organization). The measuring system meets all the requirements of the following
specifications:• Certified according to PA Profile 3.02
• The device can also be operated with certified devices of other manufacturers (interoperability)
EtherNet/IP certification

The measuring device is certified and registered by the ODVA (Open Device Vendor Association).
The measuring system meets all the requirements of the following specifications:• Certified in accordance with the ODVA Conformance Test
• EtherNet/IP Performance Test
• EtherNet/IP PlugFest compliance• The device can also be operated with certified devices of other manufacturers (interoperability)

Certification PROFINETPROFINET interface
The measuring device is certified and registered by the PNO (PROFIBUS Nutzerorganisation e.V. /PROFIBUS User Organization). The measuring system meets all the requirements of the following
specifications:• Certified according to:• Test specification for PROFINET devices• PROFINET Security Level 2– Netload Class 2 0 Mbps
• The device can also be operated with certified devices of other manufacturers (interoperability)• The device supports PROFINET S2 system redundancy.

122

Endress+Hauser

Proline Promass Q 500

Certification PROFINET with
Ethernet-APL

PROFINET interface

Pressure Equipment
Directive

The devices can be ordered with or without a PED or UKCA approval. If a device with a PED or UKCA
approval is required, this must be explicitly stated in the order. For devices with nominal diameters
less than or equal to DN 25 (1"), this is neither possible nor necessary. A UK Ex approval must be
selected for UKCA.

The measuring device is certified and registered by the PNO (PROFIBUS Nutzerorganisation e.V. /PROFIBUS User Organization). The measuring system meets all the requirements of the following
specifications:• Certified according to:• Test specification for PROFINET devices• PROFINET PA Profile 4• PROFINET Security Level 2– Netload Class 2 0 Mbps
• APL conformance test
• The device can also be operated with certified devices of other manufacturers (interoperability)• The device supports PROFINET S2 system redundancy.

• With the marking:a) PED/G1/x (x = category) orb) UK/G1/x (x = category)on the sensor nameplate, Endress+Hauser confirms compliance with the "Essential Safety
Requirements"a) specified in Annex I of the Pressure Equipment Directive 2014/68/EU orb) Schedule 2 of Statutory Instruments 2016 No. 1105.
• Devices bearing this marking (PED or UKCA) are suitable for the following types of medium:• Media in Group 1 and 2 with a vapor pressure greater than, or smaller and equal to
0.5 bar (7.3 psi)
• Unstable gases• Devices not bearing this marking (without PED or UKCA) are designed and manufactured
according to sound engineering practice. They meet the requirements ofa) Art. 4 Para. 3 of the Pressure Equipment Directive 2014/68/EU orb) Part 1, Para. 8 of Statutory Instruments 2016 No. 1105.
The scope of application is indicateda) in diagrams 6 to 9 in Annex II of the Pressure Equipment Directive 2014/68/EU orb) Schedule 3, Para. 2 of Statutory Instruments 2016 No. 1105.
Radio approval

The measuring device has radio approval.
For detailed information on the radio approval, see the Special Documentation →  131

Measuring instrument
approval

The measuring device is approved as a component in measuring systems (MI-005) in service subject
to legal metrological control in accordance with the European Measuring Instruments Directive
2014/32/EU (MID).
The measuring device is qualified to OIML R117 and has an OIML Certificate of Conformity(optional).

Additional certification

CRN approval
Some device versions have CRN approval. A CRN-approved process connection with a CSA approvalmust be ordered for a CRN-approved device.
Tests and certificates• EN10204-3.1 material certificate, wetted parts and sensor housing• Pressure test, internal process, inspection certificate• PMI test (XRF), internal procedure, wetted parts, test report• Compliance with requirements derived from cGMP, Declaration• EN10204-2.1 confirmation of compliance with the order and EN10204-2.2 test report

Endress+Hauser

123

Proline Promass Q 500

Testing of welded connections
OptionTest standard

Component

ISO 23277 AL2x (PT)
ASME
ASME
NORSOK Measuring tube Process connection
ISO 10675-1 AL1 (RT, DR) B31.3 VIII Div.1
M-601
NFS Appx. 4+8
KF

x

KK

x

KP

x

KR
K1

x
x

K2

x

K3

x

K4

x

PT

RT

PT

RT

PT

RT

VT, PT

VT, RT

PT

DR

PT

DR

PT

DR

VT, PT

VT, DR

PT = penetrant testing, RT = radiographic testing, VT = visual testing, DR = digital radiography
All options with test report

Other standards andguidelines• EN 60529
Degrees of protection provided by enclosures (IP code)• IEC/EN 60068-2-6
Environmental influences: Test procedure - Test Fc: vibrate (sinusoidal).
• IEC/EN 60068-2-31
Environmental influences: Test procedure - Test Ec: shocks due to rough handling, primarily for
devices.
• EN 61010-1
Safety requirements for electrical equipment for measurement, control and laboratory use general requirements• IEC/EN 61326-2-3
Emission in accordance with Class A requirements. Electromagnetic compatibility (EMC
requirements).
• NAMUR NE 21
Electromagnetic compatibility (EMC) of industrial process and laboratory control equipment• NAMUR NE 32
Data retention in the event of a power failure in field and control instruments withmicroprocessors• NAMUR NE 43
Standardization of the signal level for the breakdown information of digital transmitters withanalog output signal.
• NAMUR NE 53
Software of field devices and signal-processing devices with digital electronics• NAMUR NE 80
The application of the pressure equipment directive to process control devices• NAMUR NE 105
Specifications for integrating fieldbus devices in engineering tools for field devices• NAMUR NE 107
Self-monitoring and diagnosis of field devices• NAMUR NE 131
Requirements for field devices for standard applications• NAMUR NE 132
Coriolis mass meter

Ordering information
Detailed ordering information is available from your nearest sales organizationwww.addresses.endress.com or in the Product Configurator at www.endress.com:

124

1.

Select the product using the filters and search field.

2.

Open the product page.

Endress+Hauser

Proline Promass Q 500

3.

Select Configuration.
Product Configurator - the tool for individual product configuration• Up-to-the-minute configuration data
• Depending on the device: Direct input of measuring point-specific information such asmeasuring range or operating language• Automatic verification of exclusion criteria• Automatic creation of the order code and its breakdown in PDF or Excel output format• Ability to order directly in the Endress+Hauser Online Shop

Application packages
Many different application packages are available to enhance the functionality of the device. Suchpackages might be needed to address safety aspects or specific application requirements.
The application packages can be ordered with the device or subsequently from Endress+Hauser.
Detailed information on the order code in question is available from your local Endress+Hauser sales
center or on the product page of the Endress+Hauser website: www.endress.com.
Detailed information on the application packages:
Special Documentation for the device →  131
Diagnostic functionality

Order code for "Application package", option EA "Extended HistoROM"
Comprises extended functions concerning the event log and the activation of the measured value
memory.
Event log:
Memory volume is extended from 20 message entries (standard version) to up to 100 entries.
Data logging (line recorder):• Memory capacity for up to 1000 measured values is activated.
• 250 measured values can be output via each of the 4 memory channels. The recording interval canbe defined and configured by the user.
• Measured value logs can be accessed via the local display or operating tool e.g. FieldCare,
DeviceCare or Web server.
For detailed information, see the Operating Instructions for the device.

Heartbeat Technology

Order code for "Application package", option EB "Heartbeat Verification + Monitoring"
Heartbeat Verification
Meets the requirement for traceable verification to DIN ISO 9001:2008 Chapter 7.6 a) "Control ofmonitoring and measuring equipment".
• Functional testing in the installed state without interrupting the process.
• Traceable verification results on request, including a report.
• Simple testing process via local operation or other operating interfaces.
• Clear measuring point assessment (pass/fail) with high test coverage within the framework of
manufacturer specifications.
• Extension of calibration intervals according to operator's risk assessment.
Heartbeat Monitoring
Continuously supplies data, which are characteristic of the measuring principle, to an external
condition monitoring system for the purpose of preventive maintenance or process analysis. These
data enable the operator to:• Draw conclusions - using these data and other information - about the impact the process
influences (e.g. corrosion, abrasion, formation of buildup etc.) have on measuring performanceover time.
• Schedule servicing in time.
• Monitor the process or product quality, e.g. gas pockets.
For detailed information, see the Special Documentation for the device.

Concentration measurement

Order code for "Application package", option ED "Concentration"
Calculation and outputting of fluid concentrations.

Endress+Hauser

125

Proline Promass Q 500

The measured density is converted to the concentration of a substance of a binary mixture using the"Concentration" application package:• Choice of predefined fluids (e.g. various sugar solutions, acids, alkalis, salts, ethanol etc.).
• Common or user-defined units (°Brix, °Plato, % mass, % volume, mol/l etc.) for standard
applications.
• Concentration calculation from user-defined tables.
For detailed information, see the Special Documentation for the device.

Advanced density function

Order code for "Application package", option EH "Advanced density function"
Advanced software functions for density measurement:• Easy integration into existing density applications with integrated time period signal (TPS).
• Two density values shown simultaneously on the local display.
• Advanced density coefficients for optimum recalibrations.
For detailed information, see the Special Documentation for the device.

Premium density andextended density function

Order code for "Application package", option EI " Premium density, +/- 0.1 kg/m3 + extended density
function"
Highest density measurement accuracy thanks to premium density calibration and extended
software functions for density measurement:• Easy integration into existing density applications with integrated time period signal (TPS).
• Two density values shown simultaneously on the local display.
• Advanced density coefficients for optimum recalibrations.
For detailed information, see the Special Documentation for the device.

Petroleum

Order code for "Application package", option EJ "Petroleum"
The most important parameters for the Oil & Gas Industry can be calculated and displayed with thisapplication package.
• Corrected volume flow and calculated reference density in accordance with the "API Manual of
Petroleum Measurement Standards, Chapter 11.1"• Water content, based on density measurement• Weighted mean of the density and temperature
For detailed information, see the Special Documentation for the device.

Petroleum & locking function

Order code for "Application package", option EM "Petroleum & locking function"
The most important parameters for the Oil & Gas Industry can be calculated and displayed with thisapplication package. It is also possible to lock the settings.
• Corrected volume flow and calculated reference density in accordance with the "API Manual of
Petroleum Measurement Standards, Chapter 11.1"• Water content, based on density measurement• Weighted mean of the density and temperature
For detailed information, see the Special Documentation for the device.

OPC-UA Server

Order code for "Application package", option EL "OPC-UA Server"
The application package provides an integrated OPC-UA server for comprehensive device services for
IoT and SCADA applications.
For detailed information, see the Special Documentation for the device.

Accessories
Various accessories, which can be ordered with the device or subsequently from Endress+Hauser, areavailable for the device. Detailed information on the order code in question is available from your126

Endress+Hauser

Proline Promass Q 500

local Endress+Hauser sales center or on the product page of the Endress+Hauser website:www.endress.com.
Device-specific accessories

For the transmitter
Accessories

Description

Transmitter• Proline 500 – digital• Proline 500

Transmitter for replacement or storage. Use the order code to define the following
specifications:• Approvals• Output• Input• Display/operation• Housing• Software
Proline 500 – digital transmitter: • Order
number: 8X5BXX-*********A• Proline 500 transmitter:
Order number: 8X5BXX-*********B
500 transmitter for replacement: Proline
It is essential to specify the serial number of the current transmitter whenordering. On the basis of the serial number, the device-specific data (e.g.
calibration factors) of the replaced device can be used for the newtransmitter.
500 – digital transmitter: Installation Instructions EA01151D
 •• Proline
Proline 500 transmitter: Installation Instructions EA01152D

External WLAN antenna

External WLAN antenna with 1.5 m (59.1 in) connecting cable and two angle
brackets. Order code for "Accessory enclosed", option P8 "Wireless antenna widearea".
The external WLAN antenna is not suitable for use in hygienic • applications.
• Additional information regarding the WLAN interface →  113.

 Order number: 71351317
 Installation Instructions EA01238D
Pipe mounting set

Pipe mounting set for transmitter.
500 – digital transmitter Proline
Order number: 71346427
 Installation Instructions EA01195D
500 transmitter Proline
Order number: 71346428

Weather protection coverTransmitter• Proline 500 – digital• Proline 500

Is used to protect the measuring device from the effects of the weather: e.g.
rainwater, excess heating from direct sunlight.
500 – digital transmitter • Proline
Order number: 71343504• Proline 500 transmitter
Order number: 71343505

 Installation Instructions EA01191D
Display guard
Proline 500 – digital

Is used to protect the display against impact or scoring, for example from sand indesert areas.

 Order number: 71228792
 Installation Instructions EA01093D

Endress+Hauser

127

Proline Promass Q 500

Connecting cable
Proline 500 – digital
Sensor –Transmitter

The connecting cable can be ordered directly with the measuring device (order codefor "Cable, sensor connection) or as an accessory (order number DK8012).
The following cable lengths are available: order code for "Cable, sensor connection"• Option B: 20 m (65 ft)• Option E: User-configurable up to max. 50 m
• Option F: User-configurable up to max. 165 ft
possible cable length for a Proline 500 – digital connecting cable:
 Maximum
300 m (1 000 ft)

Connecting cable
Proline 500
Sensor –Transmitter

The connecting cable can be ordered directly with the measuring device (order codefor "Cable, sensor connection") or as an accessory (order number DK8012).
The following cable lengths are available: order code for "Cable, sensor connection"• Option 1: 5 m (16 ft)• Option 2: 10 m (32 ft)• Option 3: 20 m (65 ft)

 Possible cable length for a Proline 500 connecting cable: max. 20 m (65 ft)
For the sensor
Accessories

Description

Heating jacket

Is used to stabilize the temperature of the fluids in the sensor. Water, water vaporand other non-corrosive liquids are permitted for use as fluids.

 If using oil as a heating medium, please consult with Endress+Hauser.
Use the order code with the product root DK8003.

 Special Documentation SD02161D
Communication-specificaccessories

Accessories

Description

Commubox FXA195
HART

For intrinsically safe HART communication with FieldCare via the USB interface.

HART Loop Converter
HMX50

Is used to evaluate and convert dynamic HART process variables to analog current
signals or limit values.

 Technical Information TI00404F
Information TI00429F
 •• Technical
Operating Instructions BA00371F

Fieldgate FXA42

Is used to transmit the measured values of connected 4 to 20 mA analog
measuring devices, as well as digital measuring devices
Technical Information TI01297S
 •• Operating
Instructions BA01778S• Product page: www.endress.com/fxa42

Field Xpert SMT50

The Field Xpert SMT70 tablet PC for device configuration enables mobile plant
asset management in non-hazardous areas. It is suitable for commissioning andmaintenance staff to manage field instruments with a digital communicationinterface and to record progress.
This tablet PC is designed as an all-in-one solution with a preinstalled driver libraryand is an easy-to-use, touch-sensitive tool which can be used to manage field
instruments throughout their entire life cycle.
Technical Information TI01342S
 •• Operating
Instructions BA01709S• Product page: www.endress.com/smt50

128Endress+Hauser

Proline Promass Q 500

Field Xpert SMT70

The Field Xpert SMT70 tablet PC for device configuration enables mobile plant
asset management in hazardous and non-hazardous areas. It is suitable forcommissioning and maintenance staff to manage field instruments with a digitalcommunication interface and to record progress.
This tablet PC is designed as an all-in-one solution with a preinstalled driver libraryand is an easy-to-use, touch-sensitive tool which can be used to manage field
instruments throughout their entire life cycle.
Technical Information TI01342S
 •• Operating
Instructions BA01709S• Product page: www.endress.com/smt70

Field Xpert SMT77

The Field Xpert SMT77 tablet PC for device configuration enables mobile plant
asset management in areas categorized as Ex Zone 1.
Technical Information TI01418S
 •• Operating
Instructions BA01923S• Product page: www.endress.com/smt77

Service-specific accessories

Accessory

Description

Applicator

Software for selecting and sizing Endress+Hauser measuring devices:• Choice of measuring devices with industrial requirements• Calculation of all the necessary data for identifying the optimum flowmeter:
e.g. nominal diameter, pressure loss, flow velocity and accuracy.
• Graphic illustration of the calculation results• Determination of the partial order code, administration, documentation andaccess to all project-related data and parameters over the entire life cycle ofa project.
Applicator is available:• Via the Internet: https://portal.endress.com/webapp/applicator• As a downloadable DVD for local PC installation.

W@M

W@M Life Cycle Management
Improved productivity with information at your fingertips. Data relevant to aplant and its components is generated from the first stages of planning andduring the asset’s complete life cycle.
W@M Life Cycle Management is an open and flexible information platform
with online and on-site tools. Instant access for your staff to current, in-depth
data shortens your plant’s engineering time, speeds up procurement processes
and increases plant uptime.
Combined with the right services, W@M Life Cycle Management boosts
productivity in every phase. For more information, see:www.endress.com/lifecyclemanagement

FieldCare

FDT-based plant asset management tool from Endress+Hauser.
It can configure all smart field units in your system and helps you manage
them. By using the status information, it is also a simple but effective way ofchecking their status and condition.

 Operating Instructions BA00027S and BA00059S
DeviceCare

Tool for connecting and configuring Endress+Hauser field devices.

 Innovation brochure IN01047S
System components

Accessories

Description

Memograph M graphic
data manager

The Memograph M graphic data manager provides information on all the relevantmeasured variables. Measured values are recorded correctly, limit values aremonitored and measuring points analyzed. The data are stored in the 256 MB
internal memory and also on a SD card or USB stick.
Information TI00133R
 •• Technical
Operating Instructions BA00247R

Cerabar M

The pressure transmitter for measuring the absolute and gauge pressure of gases,
steam and liquids. It can be used to read in the operating pressure value.
Information TI00426P and TI00436P
 •• Technical
Operating Instructions BA00200P and BA00382P

Endress+Hauser

129

Proline Promass Q 500

Accessories

Description

Cerabar S

The pressure transmitter for measuring the absolute and gauge pressure of gases,
steam and liquids. It can be used to read in the operating pressure value.
Technical Information TI00383P
 •• Operating
Instructions BA00271P

iTEMP

The temperature transmitters can be used in all applications and are suitable forthe measurement of gases, steam and liquids. They can be used to read in themedium temperature.

 "Fields of Activity" document FA00006T
Documentation
For an overview of the scope of the associated Technical Documentation, refer to the following:• Device Viewer (www.endress.com/deviceviewer): Enter serial number from nameplate.
• Endress+Hauser Operations app: Enter serial number from nameplate or scan matrix code on
nameplate.
Standard documentation

Brief Operating Instructions

Brief Operating Instructions for the sensor
Measuring device

Documentation code

Proline Promass Q

KA01262D

Brief Operating Instructions for the transmitter
Documentation code

Measuring device

HART

FOUNDATION Fieldbus

PROFIBUSPA

PROFIBUSDP

Modbus
RS485

EtherNet/IP

PROFINETPROFINET
with
EthernetAPL

Proline 500 – digital

KA01315D

KA01233D

KA01392D

KA01390D

KA01319D

KA01346D

KA01351D

KA01521D

Proline 500

KA01314D

KA01291D

KA01391D

KA01389D

KA01318D

KA01347D

KA01350D

KA01520D

Operating Instructions
Measuring device

Promass Q 500

Documentation code
HART

FOUNDATION Fieldbus

PROFIBUSPA

PROFIBUSDP

Modbus
RS485

EtherNet/IP

PROFINETPROFINET
with
EthernetAPL

BA01534D

BA01567D

BA01556D

BA01878D

BA01545D

BA01755D

BA01766D

BA02129D

Description of Device Parameters
Documentation code

Measuring device

HART

FOUNDATION Fieldbus

PROFIBUSPA

PROFIBUSDP

Modbus
RS485

EtherNet/IP

PROFINETPROFINET
with
EthernetAPL

Promass 500

GP01060D

GP01096D

GP01061D

GP01137D

GP01062D

GP01120D

GP01121D

GP01173D

130

Endress+Hauser

Proline Promass Q 500

Supplementary devicedependent documentation

Safety instructions
Safety instructions for electrical equipment for hazardous areas.

Contents

Documentation code
Measuring device

ATEX/IECEx Ex i

XA01473D

ATEX/IECEx Ex ec

XA01474D

cCSAus IS

XA01475D

cCSAus Ex i

XA01509D

cCSAus Ex nA

XA01510D

INMETRO Ex i

XA01476D

INMETRO Ex ec

XA01477D

NEPSI Ex i

XA01478D

NEPSI Ex nA

XA01479D

NEPSI Ex i

XA01658D

NEPSI Ex nA

XA01659DJPN

XA01780D

Functional Safety ManualContents

Documentation code

Proline Promass 500

SD01729D

Special Documentation
Contents

Documentation code

Information on the Pressure Equipment Directive

SD01614D

Radio approvals for WLAN interface for A309/A310 display module

SD01793D

OPC-UA server 1)

SD02040D

Overrun measurementSD02342D

1)This Special Documentation is only available for device versions with a HART output.

Contents

Documentation code
HART

FOUNDATION Fieldbus

PROFIBUSPA

PROFIBUSDP

ModbusRS485

PROFINET

EtherNet/IP

PROFINET
with
EthernetAPL

Web server

SD01666D

SD01669D

SD01668D

SD02232D

SD01667D

SD01971D

SD01970D

SD02769DHeartbeat Technology

SD01643D

SD01608D

SD01705D

SD02203D

SD01704D

SD01989D

SD01983D

SD02732D

Concentration
measurementSD01645D

SD01709D

SD01711D

SD02213D

SD01710D

SD02007D

SD02006D

SD02736D

PetroleumSD02013D–

SD02292D

SD02217D

SD02014D

SD02015D

SD02012D

SD02740D

Petroleum & locking
function

SD02499D–––

SD02500D–––

Advanced density
function

SD02354D–––

SD02354D––

SD02354D

Gas Fraction Handler

SD02584D–––

SD02584D

SD02584D–

SD02584D

Endress+Hauser

131

Proline Promass Q 500

Contents

Documentation code
HART

FOUNDATION Fieldbus

PROFIBUSPA

PROFIBUSDP

ModbusRS485

PROFINET

EtherNet/IP

PROFINET
with
EthernetAPL

Custody transfer(counter for liquids
other than water)

SD01690D–––

SD01691D–––

Custody transfer(counter for gas)SD02464D–––

SD02465D–––

Custody transfer(counter for gas, inaccordance with theGerman Measurement
and Calibration
Ordinance (Mess- undEichverordnung))

SD02582D–––

SD02583D–––

Installation Instructions
Contents

Comment

Installation instructions for spare part sets andaccessories

Documentation code: specified for each individual accessory →  127.

Registered trademarks
HART®
Registered trademark of the FieldComm Group, Austin, Texas, USA
PROFIBUS®
Registered trademark of the PROFIBUS Nutzerorganisation e.V. (PROFIBUS User Organization),
Karlsruhe, Germany
FOUNDATION™ Fieldbus
Registration-pending trademark of the FieldComm Group, Austin, Texas, USA
Modbus®
Registered trademark of SCHNEIDER AUTOMATION, INC.
EtherNet/IP™
Trademark of ODVA, Inc.
Ethernet-APL™
Registered trademark of the PROFIBUS Nutzerorganisation e.V. (PROFIBUS User Organization),
Karlsruhe, Germany
PROFINET®
Registered trademark of the PROFIBUS Nutzerorganisation e.V. (PROFIBUS User Organization),
Karlsruhe, Germany

132

Endress+Hauser

*71574742*
71574742

www.addresses.endress.com