B/W CCD Camera
Model CS8560BD
Specifications

Contents
1. Product Description ························· 7
2. Features·············································· 7
3. Configuration ···································· 8
4. Option Unit········································ 8
5. Operation Mode································ 8
6. Specifications····································· 25
7. Timing Chart····································· 33
8. Guarantee ·········································· 41
9. Repair················································· 41
10. External View Drawing···················· 42

D4190543A

Important Safety Precautions
BEFORE USE - GENERAL SAFETY INSTRUCTIONSThis specifications contains important information for the operator (user) and/or people in thevicinity to avoid personal injury, or property damage.
• Prior to use, read this specifications carefully to fully understand its instructions for correct use.
• After reading, keep operation manual near the equipment for future reference.
WARNINGS & CAUTIONS
Indication
Meaning
This indicates the existence of a hazard that death or catastrophic bodily
WARNING injury(*1) may result from improper use.

CAUTION

This indicates the existence of a hazard that bodily injury(*2) or property
damage(*3) may result from improper use.

Notes
*1 Catastrophic bodily injury means loss of eyesight, burns (high and low temperatured), shock,fracture, poisoning, etc. which leaves a sequela and repuire hospitalization or prolonged
treatment.
*2 Bodily injury means injuries, burns and electric shock which does not require hospitalizationor prolonged treatment.
*3 Property damage means extended harm to home, household effects, domesticated animals, andpets.
Graphic symbol definitions
Indication
Meaning
This mark indicates a prohibited action that must not be carried out. The actualprohibited action is indicated in the symbol or nearby graphically or described in
text.
This mark indicates a mandatory action that must not be carried out. The actual
instruction is indicated in the symbol or nearby graphically or described in text.

●Handling Precautions

WARNING
Stop operation immediately when any abnormality or defect occurs.
Use during an abnormal condition; such as emitting smoke, burning odors,
damage from dropping invasion of foreign objects, etc. may cause fire and/or
electric shock. Be always sure to disconnect the power plug from the electricaloutlet (socket) at once and contact your dealer.
Do not operate in places with possibility of becoming wet.
This may cause fire and/or electric shock.
Do not repair, disassemble and/or modify by yourself.
This may cause fire and/or electric shock. Be always sure to contact your dealer
for internal repair, check and cleaning of the product.

2

D4190543A

Don’t place things or materials on the unit.
Ingress of foreign materials such as metallic things and liquid into the unit may
cause a fire or an electric shock.
Do not put the product in an unstable, slanting and/or vibrated place.
Drop and/or fail of the product may cause injury.
Do not touch the power cord or other connection cables during athunderstorm.
This might cause electric shock.
Use the specified power supply.
Use of an unspecified power supply may result in fire or electric shock.
Do not be handled roughly, damaged, fabricated, bent forcefully, pulled,
twisted, bundled, placed under heavy objects or heated the power cord ,connection cable.
Otherwise, fire or electric shock may result.

CAUTION
Note the following instructions when installing.
-Do not wrap the product in an inflammable material, such as cloth.
-Do not put the product in a narrow space, since the heat generated from theproduct may be difficult to emanate.
If you do not follow the above, the heat generated by the product may cause
fire.
Avoid setting in humid, smoky, vaporized or dusty places. A fire or anelectric shock may occur in such places.
This may cause fire and/or electric shock.
Do not put the product in direct sunshine and/or high temperature.
The temperature inside the product may cause fire.
Use the specified DC power cable or connection cable.
Otherwise, a fire or an electric shock may occur.
Turn OFF the power in the case of connection.
Turn OFF the power in the case of connection of power cable or connection
cable.
Otherwise, an electric shock or a malfunction may occur.
Do not expose its camera head to any intensive light (such as direct
sunlight).
Otherwise, its inner image pickup device might get damaged.
Avoid short-circuiting signal output.
Otherwise, a malfunction may occur.
Avoid giving a strong shock against the camera body.
It might cause a breakdown or damage.
If your camera is used in a system where its camera connector is subjected tostrong repetitive shocks, its camera connector is possible to break down. If
you intend to use your camera in such a situation, if possible, bundle and fix
a camera cable in the place near the camera, and do not transmit a shock to
the camera connector.
Ask your dealer to perform a periodical check and internal cleaning(approx. once every five years).
Dust inside the product may cause fire and/or trouble. For check and cleaning
cost, please consult your dealer.
3

D4190543ADISCLAIMER (LIMITED WARRANTY)
We disclaim any responsibility and shall be held harmless for any damages or losses incurred by
the user in any of the following cases;z Fire, earthquake or any other act of God; acts by third parties; misuse by the user, whether
intentional or accidental; use under extreme operating conditions.
z Malfunction or non-function resulting in indirect, additional or consequential damages,
including but not limited to loss of expected income and suspension of business activities.
z Incorrect use not in compliance with instructions in this instruction specifications and manual.
z Malfunctions resulting from misconnection to other equipment.
z Repairs or modifications made by the user or caused to be made by the user and carried out by
an unauthorized third party.
z Notwithstanding the foregoing, Teli’s liabilities shall not, in any circumstances, exceed thepurchase price of the product.
z About the item which does not have a publication in the specifications and manual of thisproduct, it considers as the outside for a guarantee.

RESTRICTION FOR USE
z Should the equipment be used in the following conditions or environments, give consideration
to safely measures and inform us of such usage:
1. Use of the equipment in the conditions or environment contrary to those specified, or use
outdoors.
2. Use of the equipment in applications expected to cause potential hazard to people orproperty, which require special safety measures to be adopted.
z This product can be used under diverse operating conditions. Determination of applicability ofequipment or devices concerned shall be determined after analysis or testing as necessary by
the designer of such equipment or devices, or personnel related to the specifications. Suchdesigner or personnel shall assure the performance and safety of the equipment or devices.
z This product is not designed or manufactured to be used for control of equipment directly
concerned with human life (*1) or equipment relating to maintenance of public
services/functions involving factors of safety (*2). Therefore, the product shall not be used forsuch applications.
(*1): Equipment directly concerned with human life refers to.
- Medical equipment such as life-support systems, equipment for operating rooms.
- Exhaust control equipment for exhaust gases such as toxic fumes or smoke.
- Equipment mandatory to be installed by various laws and regulations such as the FireAct or Building Standard Law- Equipment related to the above.
(*2) : Equipment relating to maintenance of public services/functions involving factors ofsafety refers to.
- Traffic control systems for air transportations, railways, roads, or marinetransportation- Equipment for nuclear power generation- Equipment related to the above

4

D4190543A

CAUTIONS ON USEz Carefully handle the units.
Do not drop, or give a strong shock or vibration to the camera. This may cause problems. Treat
the camera cables carefully to prevent cable problems, such as cable breakdown and loosened
connections.
z Operating ambient temperature and humidity.
Do not use the camera in places where temperature and humidity exceed the specifications.
Picture quality will lower and internal parts may be damaged.
Be particularly careful when using in places exposed to direct sunlight. When shooting in hot
places, depending on the conditions of the object and the camera (for example when the gain is
increased), noise in the form of vertical strips or white dots may occur. This is not a
malfunction.
z Combination of C-mount lens
Depending on the lens you use, the performance of the camera may not be brought out fully
due to the deterioration in resolution and brightness in the peripheral area, occurrence of aghost, aberration and others. When you check the combination between the lens and camera, besure to use the lens you actually use.
As for the C-mount lens used combining this product, the projection distance from bottom of
the screw should use 7.9mm or less.
C-mount lensBottom ofthe screw

7.9mm or lessz Restriction for the lens combination
This camera might form a ghost to image area depending on the combination of a lens and an
illumination with this camera. But this is not a failure of this camera. Therefore, please checkthe combination of the lens and the illumination with this camera when use.
When mounting a lens, take extra caution so that the lens is not tilted, nor does flaw exist at
the lens-mount-screw part. Also check to confirm that no dirt nor other foreign object is put
insideImproper mounting might cause the parts to become locked.
z Do not shoot under intense light.
Avoid intense light such as spot light on part of the screen because it may cause blooming or
smears. If intense light falls on the screen, vertical stripes may appear on the screen, but this is
not a malfunction.
z Do not expose the camera’s image-pickup-plane to sunlight or other intense light directly.
Its inner CCD (charge-coupled device) might be damaged.
z MoireWhen thin stripe patterns are shot, stripe patterns that are not actually there (moire) may
appears as interference stripes. This is not a malfunction.
z Undesirable noise
If the camera or the cables are located near something which emit strong magnetism or nearsomething which emit strong electric wave, undesirable noise may appear on the screen. In
such a case, try to change the location of the camera or the cable wiring.
z Handling of the protection cap
When the camera is not in use, put a lens-cap onto the camera head for protection of theimage-pickup-plane.

5

D4190543A

CAUTIONS ON USE
z When cleaning the camera
Always turn off the power and clean with a piece of soft dry cloth.
To remove stubborn stains, use a soft cloth soaked in diluted acid-free detergent. Do not use
alcohol, benzine, thinner, etc. If used, coating and printed letters may be discolored.
In case the image-pickup-plane should be settled with fine dust, dirt, or scratched, ask yourdealer for technical advice.
z Wastes of this product should be separated and discarded in compliance with the various
national and local ordinances.

“This symbol is applicable for EU member states only”

6

D4190543A

1. Product Description
Model CS8560BD is an integrated type B/W CCD camera with a VGA format all-pixel-data
readout CCD. This model has twice greater driving frequency of conventional cameras to achieve
fast-speed data-processing. The model is suited for high-speed, high-resolution image processinguse. Its compact, light-weight body is ideal for system integration.

2. Features(1) Double-speed scanThis model reads out image-data twice as fast as conventional cameras do.
(2) All pixel's data readout
With its built-in all-pixel-data-readout CCD, this model can read out image-data just in
approximately 1/60 sec. A frame-shutter reads out all data even under RTS (Random Trigger
Shutter) mode.
(3) High vertical resolution
As all pixel's data are read out even under RTS mode (in 1/60 sec.), images with nodeterioration in vertical resolution are obtained.
(4) Square grid pattern CCD
Pixel's in CCD are aligned in square grid pattern. This makes it easier to perform computation
correctly for image processing use.
(5) External Sync.
The camera is switched over to external synchronization operation automatically when external
HD signal is input.
(6) Random trigger shutter function
With a built-in RTS, the camera’s CCD starts light-exposure in synchronization with external
trigger signals. This function enables the camera to capture fast-moving subjects at constantposition for precise image processing.
(7) Restart / Reset
Under the restart / reset mode, this model can capture images at an arbitrary timing cued byexternal VD signal.
(8) Multiple shutterWith this shutter, this model capture images at an arbitrary timing cued by external trigger
signal, and then outputs video at an arbitrary timing cued by external VD signal.
(9) Partial-scan
Under the partial scan mode, only 1/2 or 1/4 screen center portion of image information is readout, resulting in a faster operation.
(10) Ultra-compact & light-weight camera head
The model features its ultra-compact and light-weight camera head, freeing you from yourintegration-space-problem. In addition, it has an excellent shock and vibration resistance.

7

D4190543A

3. Configuration(1) Camera body··································································· 1(2) Accessory
Manual ············································································ 1

4. Option Unit
(1) DC SYNC IN cable ············ Model name : CPRC3700 [2m,3m,5m,10m](2) Camera adapter ·················· Model name : CA170(3) Camera-mounting kit ··········· Model name : CPT8560(4) Camera-connector fixing hardware*Contact your dealer / distributor for details of option units.
*Conformity of an option part and EMC conditions
About the conformity of EMC standard of this machine, it has guaranteed in theconditions combined with the above-mentioned option part.
When used combined parts other than specification of our company, I ask you to havefinal EMC conformity checked of a visitor with a machine and the whole equipment.

5. Operation Mode
(1) GAIN selection (Camera rear-panel SW)
Switches sensitivity setting(1-1) FIX ------------------- Factory-prefixed gain
(1-2) MANU--------------- Gain is adjustable via the manual gain potentiometer (M.GAIN)(2) Video output mode selection (Camera rear-panel DIP SW)
Switches video format(2-1) 1/60: 1/60s ---------- Non-interlace mode
As all pixels are read out in 1/60s, you will get images with thehigher V resolution.
(2-2) 1/120: 1/120s ------- 2:1 interlace MIX mode
As vertical pixels are added in readout, the sensitivity is same asthat of 1/60s non-interlace mode during electronic shutter OFF.
Twice greater sensitivity is obtained under shutter-speed range of
1/200 – 1/20000.
(3) RTS (Random Trigger Shutter) exposure selection (Camera rear-panel DIP SW)
Switches light exposure mode under RTS mode(3-1) FIX mode ----------- Rear DIP SW
Exposure-time control via rear-panel DIP switch(3-2) PULSE W mode---- TRIG signal pulse width control
Exposure-time control via TRIG signal pulse width8

D4190543A

(4) Shutter mode selection (Camera rear-panel DIP SW or TRIG signal IN [Automatic])
Switches shutter mode(4-1) NOR mode ---------- Normal electronic shutter
Exposure control via internal sync signal
High-speed shutter: From 1/20,000s through OFF (8 position)(4-2) RTS mode----------- Random trigger shutter
Exposure control via ext. trigger or ext. sync input
Timing charts are shown below. (TRIG timing: Positive)
Notes: * RTS selection is automatic with TRIG status** Neither under FIX nor PULSE W mode, RTS doesn’t
work if Electronic shutter speed SW is set in OFF
position.
(a)Non-reset mode (Under internal sync / external sync --- Consecutive VD IN)
Exposure starts at the timing of TRIG signal IN. After each exposure is completed, thecamera outputs video at each next VD IN timing.
□1/60s Non-interlace
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)
INT/EXT VD
CCD Exposure

Exposure time
(E-shutter speed orpulse width)

VIDEO OUT

Outputs VD,V-SYNC
during standby

Definite

Outputs video at the next VD
timing after exposure

WEN OUT(Standard)
WEN OUT[Option]Normal shutter

Random trigger shutter

Normal shutter□1/120s 2:1 Interlace
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)
INT/EXT VD

EVEN

ODD

EVEN

ODD

EVEN

ODD

EVEN

ODD

EVEN

ODD

EVEN

ODD

EVEN

CCD Exposure
Exposure time

EVEN
image

VIDEO OUTEVENODD (E-shutter speed or
image
ODD

EVEN

Outputs VD,V-SYNC
during standby

WEN OUT(Standard)

ODD
image

pulse width)
ODD

EVEN

ODD

EVEN
image
EVEN

Outputs video at the next VD
timing after exposure

ODD

EVEN

DefiniteWEN OUT[Option]Random trigger shutter

Normal shutter

9

Normal shutter

D4190543A

(b) Non-reset mode (Under external sync --- Single VD IN)
After TRIG IN and exposure, the camera goes into standby until next ext. VD IN.
□1/60s Non-interlace
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)
EXT VD
CCD Exposure

Exposure time
(E-shutter speed orpulse width)

VIDEO OUT

In sync with ext.VD during standby(No video output)

Definite

Outputs video at the next VD
timing after exposure

WEN OUT(Standard)
WEN OUT[Option]Random trigger shutter

Normal shutter

Normal shutter*Don’t provide ext. VD IN during exposure.
** After automatic return, fix ext. VD IN at Hi.
□1/120s 2:1 Interlace
Video output field (ODD/EVEN) is determined by ext. VD falling edge and ext. HD
phase.
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)

ODD
phase

EXT VD

ODD

EVEN
phase

ODD
phase

EVEN

EVEN

ODD

EVEN

ODD

CCD Exposure

VIDEO OUT

Exposure time
(E-shutter speed orpulse width)ODD

WEN OUT(Standard)

ODD

EVEN

In sync with ext.VD during standby(No video output)

EVEN
image
EVEN

Outputs video at the next VD
timing after exposure

ODD
image

ODD

DefiniteWEN OUT[Option]Normal shutter

Random trigger shutter

Normal shutter*Don’t provide ext. VD IN during exposure.
** After automatic return, fix ext. VD IN at Hi.

10

D4190543A

(c)V-reset mode (Under internal sync / external sync --- No VD IN)
Exposure starts at the timing of TRIG signal IN. After each exposure is completed, thecamera outputs video immediately by resetting VD. (HD is not reset)□1/60s Non-interlace
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)
CCD Exposure

Exposure time
(E-shutter speed orpulse width)

VIDEO OUT

VD,V-SYNC pause during standby

Definite
VD,V-SYNC output for 1V only

WEN OUT(Standard)
WEN OUT[Option]Normal shutter

Random trigger shutter

Normal shutter□1/120s 2:1 Interlace
Irrespective of TRIG IN phase, the camera always outputs ODD field image.
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)
CCD Exposure
ODD
image

VIDEO OUT

ODD

EVEN
image

Exposure time
(E-shutter speed orpulse width)

EVEN

ODD
image
ODD

ODD
image
ODD

Definite

VD,V-SYNC pause during standby

VD,V-SYNC output for 1V only

WEN OUT(Standard)
WEN OUT[Option]Normal shutter

Random trigger shutter

Normal shutter(d) SYNC reset mode (Under internal sync)
Exposure starts at TRIG signal input timing, resets HD, and outputs video immediately
after exposure by resetting VD.
* Available under FIX mode only.
□1/60s Non-interlace
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)
CCD Exposure

Exposure time
(E-shutter speed)

VIDEO OUT
Definite

VD,V-SYNC pause during standby

VD,V-SYNC output for 1V only

WEN OUT(Standard)
WEN OUT[Option]Normal shutter

Random trigger shutter

11

Normal shutter

D4190543A

□1/120s 2:1 Interlace
Irrespective of TRIG IN phase, the camera always outputs ODD field image.
Automatically returns to normal shutter
if fixed H level (after approx. 1/6s)
TRIG IN not acceptable for
1V after mode chenge

TRIG IN(Positive)
CCD Exposure
ODD
image

VIDEO OUT

ODD

EVEN
image

Exposure time
(E-shutter speed)

EVEN

ODD
image
ODD

VD,V-SYNC pause during standby

ODD
image
ODD

Definite
VD,V-SYNC output for 1V only

WEN OUT(Standard)
WEN OUT[Option]Normal shutter

Random trigger shutter

Normal shutter(4-3) MULTIPLE SHUTTER mode
Multiple shutter operation is available by providing TRIG IN more than one time before
ext. VD IN. (Non-reset mode, single VD, consecutive VD IN)□1/60s Non-interlace
TRIG IN(Positive)
EXT VD
CCD Exposure
Exposure time
(E-shutter speed orpulse width)

VIDEO OUT

Three exposures are mixed and output
here

WEN OUT(Standard)
WEN OUT[Option]□1/120s 2:1 Interlace
Video output field (ODD/EVEN) is determined by ext. VD falling edge and ext. HD
phase.
TRIG IN(Positive)
ODD
phase

EVEN
phase

EXT VD
CCD Exposure
ODD
image

VIDEO OUT

Exposure time
(E-shutter speed orpulse width)

EVEN
image

Three exposures are mixed and output here

WEN OUT(Standard)
WEN OUT[Option]12

D4190543A

 RTS TRIG IN notes
1/60s Non-interlace
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

510
511
512
513
514
515
516
517
518
519
520
521

9H

INT VD
INT HDOB

1 2 3 4 5490 492 494 OB

VIDEO OUTWEN OUT(Standard)

WEN OUT(Standard):The rising edge is effectiv.
Video output time
TRIG invalid period

1/120s Interlace ODD Field
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

INT HDOB8
OB 2 4 6 9
1 3 5 7

12
13

16
17

259
260
261
262
263
264
265
266
267
268
269
270

9H

INT VD484 488 492
485 489 493

OB

VIDEO OUTWEN OUT(Standard)

WEN OUT(Standard):The rising edge is effectiv.
TRIG invalid period

Video output time

1/120s Interlace EVEN Field

INT HDOB7
1 3 5 8
2 4 6

11
12

15
16

259
260
261
262
263
264
265
266
267
268
269
270

9H
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19
20
21
22
23
24
25

INT VD483 487 491
484 488 492

OB

VIDEO OUTWEN OUT(Standard)

WEN OUT(Standard):The rising edge is effectiv.
TRIG invalid period

Video output time

* TRIG becomes invalid even if TRIG (exposure start) is inputted during
TRIG invalid period.
* An image may be affected if TRIG (exposure start) is inputted during
the video output time.
* An imege may not be outputted normally if exposure is terminated during
the video output time.

 Exposure time delay under RTS
When the RTS is active, both in FIX mode and PULSE W mode, there is a time delay of
approximately 1.4 micro s until the start of exposure after the rising edge of TRIG signal(positive).
 Exposure time under pulse width mode
Under RTS pulse mode, the exposure time is determined by the pulse width. More exactly,
the actual time is the pulse width plus approximately 8.0 micro s.
TRIG IN
Pulse width

CCD
Exposure
Exposure delay time = Approximately 1.4 micro s
Exposure time = Pulse width + Approx. 8.0 micro s

13

D4190543A

(5-4) Restart / Reset
The restart / reset function is available with the ext.VD signal. You can get an arbitraryslower shutter speed than normal shutter and random trigger shutter.
Here are some notes;*
The shutter speed (exposure time) is determined by ext. VD signal interval.
** This function is enabled when the rear-panel shutter speed DIP SW is OFF.
*** Supply consecutive HD.
□1/60s Non-interlace
EXT VD IN
CCD Exposure

VIDEO OUT

The exposure time isdetermined by VD interval

WEN OUT(Standard)
WEN OUT[Option]□1/120s 2:1 Interlace
Video output field (ODD/EVEN) is determined by ext. VD falling edge and ext. HD
phase.
EVEN
phase

ODD
phase

EXT VD IN
CCD Exposure
ODD
image

EVEN
image

VIDEO OUTODDThe exposure time isdetermined by VD interval

EVEN

WEN OUT(Standard)
WEN OUT[Option]14

D4190543A

(5) Partial-scan mode selection (Camera rear-panel DIP SW)
Switches partial-scan mode
Note: Sometimes phenomenon called as “whiteout” occurs at the top of the screen when
there is strong incident light entering in the wide area of a CCD, however, this is nota malfunction. If this occurs, reduce the amount of incoming rays.
(5-1)1/2 Partial-scan (Rear-panel SW: #8-OFF, #9-ON) --- Screen center 1/2 readout□1/60s Non-interlace
Under 1/60s non-interlace mode, only the center portion of 227H out of the total
effective lines 494H (excluding BLK time) is read out. Available both under external /internal mode.
130H

High-speedtransfer

227H

Normalspeedtransfer

137H

High-speedtransfer

227H

Normalspeedtransfer■Under normal shutter (Electronic shutter OFF)
Notes: * Under ext. sync, the ext. VD should be 1V = 262H.
** Under normal shutter, set the rear-panel DIP SW #6, #7 in OFF.
131

357

131

357

<V-reset>

VIDEO OUT
227H

14H

Video output

High
speed
transfer

21H
V BLK(IncludeHigh-speedtransfer)

262H

WEN OUT(Standard)

262H

WEN OUT(Standard):
The rising edge is effective.

4H

4H

4H

4H

4H

4H

WEN OUT[OPTION]■Under other shutter modes
131

357

<Non-reset>

VIDEO OUT
227H

14H

(263H)

Video output

High
speed
transfer※1

21H

WEN OUT(Standard)

V BLK(IncludeHigh-speedtransfer)

WEN OUT(Standard):The rising edge is effective.

4H

4H
WEN OUT[OPTION]*2

4H*1:Arbitrary under ex-sync*2:See "7.TIMING CHART(4)WEN timing".

15

D4190543A

□1/120s 2:1 Interlace
Under 1/120s interlace mode, only the center portion of 207H out of the total effectivelines 485H (excluding BLK time) is read out. Available both under external / internalmode.
ODD Field
73H

103H

High-speed
trasfer

103H

Normalspeedtransfer

104H

Normalspeedtransfer

Normalspeedtransfer

66.5H

High-speedtransfer

72.5H

High-speedtransfer

104H

Normalspeedtransfer

66H

High-speedtransfer

EVEN Field■Under normal shutter (Electronic shutter OFF)
Notes: * Under ext. sync, the ext. VD should be 1V = 131.5H.
** Under normal shutter, set the rear-panel DIP SW #6, #7 in OFF.

154
155

ODD image

153
154

358
359

EVEN image

359
360

<V reset>

VIDEO OUTWEN OUT(Standard)

10H

WEN OUT[OPTION]

10H

7H

20.5H

Video output

High
speed
transfer

131.5H

0.5H

V BLK(IncludeHigh-speedtransfer)

103H

21H
V BLK
(IncludeHigh-speedtransfer)

WEN OUT(Standard):
The rising edge is effective.

16

104H

7H

Video output

High
speed
transfer

131.5H

9.5H

10H

9.5H

10H

D4190543A

■Under other shutter modes
ODD image

154
155

358
359

<Non-reset>

VIDEO OUT
103H

7H

(131.5H)

Video output

High
speed
transfer

※1

21H
V BLK
(IncludeHigh-speedtransfer)

WEN OUT(Standard)

WEN OUT[OPTION]10H

WEN OUT(Standard):The rsing edge is effective.

10H

9.5H

*2

EVEN image

153
154

359
360

VIDEO OUT
104H

7H

(132H)

Video output

High
speed
transfer

※1

20.5H
V BLK
(IncludeHigh-speedtransfer)

WEN OUT(Standard)

9.5H

WEN OUT[OPTION]

9.5H

WEN OUT(Standard):The rsing edge is effective.

10H*2*1:Arbitrary under ex-sync*2:See" 7.TIMING CHART (4) WEN timing".

17

D4190543A

(5-2)1/4 Partial-scan (Rear-panel SW: #8-ON, #9-ON) --- Screen center 1/4 readout□1/60s Non-interlace
Under 1/60s non-interlace mode, only the center portion of 81H out of the total effectivelines 494H (excluding BLK time) is read out. Available both under external / internal
mode.
200H

81H

High-speedtransfer

Normalspeedtransfer

Normalspeedtransfer

81H

High-speedtransfer

213H■Under normal shutter (Electronic shutter OFF)
Notes: * Under ext. sync, the ext. VD should be 1V = 131H.
** Under normal shutter, set the rear-panel DIP SW #6, #7 in OFF.
201

281

201

281

<V-reset>

VIDEO OUT
81H

22H

Video output

High
speed
transfer

28H
V BLK(IncludeHigh-speedtransfer)

131H
WEN OUT(Standard)

131H

WEN OUT(Standard):
The rising edge is effective.

4H

4H

4H

4H

4H

4H

WEN OUT[OPTION]■Under other shutter modes
201

281

<Non-reset>

VIDEO OUT
81H

22H

(394H)

Video output

High
speed
transfer※1

28H

WEN OUT(Standard)

V BLK(IncludeHigh-speedtransfer)

WEN OUT(Standard):The rising edge is effective.

4H

4H

WEN OUT[OPTION]*2*1:Arbitrary under ex-sync*2:See "7.TIMING CHART (4) WEN timing".

4H

18

D4190543A

□1/120s 2:1 Interlace
Under 1/120s interlace mode, only the center portion of 61H out of the total effectivelines 485H (excluding BLK time) is read out. Available both under external / internalmode.
ODD Field

103H

High-speedtransfer

30H

Normalspeedtransfer

109.5H

30H

Normalspeedtransfer

High-speedtransfer

EVEN Field
102.5H

High-speedtransfer

31H

Normalspeedtransfer

31H

Normalspeedtransfer

High-speedtransfer

109H■Under normal shutter (Electronic shutter OFF)
Notes: * Under ext. sync, the ext. VD should be 1V = 65.5H.
** Under normal shutter, set the rear-panel DIP SW #6, #7 in OFF.
214
215

ODD image

272
273

213
214

EVEN image

273
274

<V reset>

VIDEO OUT
24H
V BLK
(IncludeHigh-speedtransfer)

WEN OUT(Standard)

10H

WEN OUT[OPTION]

10H

30H

11H

23.5H

Video output

High
speed
transfer

65.5H

0.5H

V BLK
(IncludeHigh-speedtransfer)

WEN OUT(Standard):
The rising edge is effective.

19

31H

11H

Video output

High
speed
transfer

65.5H

9.5H

10H

9.5H

10H

D4190543A

■Under other shutter modes

ODD image

214
215

272
273

<Non-reset>

VIDEO OUT
30H

11H

(197.5H)

Video output

High
speed
transfer

※1

24H
V BLK
(IncludeHigh-speedtransfer)

WEN OUT(Standard)

10H

WEN OUT[OPTION]10H

WEN OUT(Standard):The rising edge is effective.

9.5H

*2

EVEN image

213
214

273
274

VIDEO OUT
31H

11H

(197H)

Video output

High
speed
transfer

※1

23.5H
V BLK
(IncludeHigh-speedtransfer)

WEN OUT
Standard

9.5H

WEN OUT[OPTION]

9.5H

WEN OUT(Standard):The rising edge is effective.

10H*2*1:Arbitrary under ex-sync*2:See "7.TIMING CHART (4) WEN timing".

20

D4190543A

(5-3) Programmable partial [Option]
By designating the high-speed transfer portion with external PARTIAL signal input, thecamera read out only the portion of CCD area necessary for your application. This isavailable under ext. sync.
EXT VD
EXT HD1(H)10 Line2(H)PARTIAL PULSEa(H)[ *1 ]VIDEO OUTn(H) Total line = n(H) x 10 line
Start

FinishHigh-speed trans. n(H)

[*1]
a(H)

1/60s non-interlace
6.0

BLK trans. 2(H)

Normal-speed

1/120s Interlace
1st field

2nd field

12.0

11.5

(Conditions)z The starting point of external partial signal is [*1] from the falling edge of ext. VD.
z The external partial signal is controlled at each ext. HD falling edge. Set the start / finish of
the external partial signal in 1H increments.
z The number of 1H high-speed transfer line is 10 lines. The actual lines are determined by
the external partial signal “hi” period. (Minimum: 2H = 20 lines)z Set video output time more than 30H. And high-speed transfer time (the external partial
signal “hi” period) before video output time is is as follows;from 2H through 22H under 1/60s non-interlace mode
from 2H through 11H under 1/120s interlace modez After high-speed transfer, 2H is allocated to blank transfer period. Normal transfer starts atthe next line.
z VIDEO OUT vertical blanking is;
V. blanking = [*1](H) + n(H) + BLK transfer [2(H)] – 1H
Example follows below.
(High-speed 2H = 20 lines (minimum), Normal-speed 70 lines + BLK 2H)
Note: Items shown as [Option] in this document is not included in your purchase as
standard components. Contact our dealer / distributor for details.

21

D4190543A

□1/60s Non-interlace
10H

High-speedtransfer

68H

Normalspeedtransfer

416H

High-speedtransfer

ＯＢ ２ライン

68H

Normalspeedtransfer

The timing is as follows;
EXT.VD
1H
494

1

VIDEO OUT
(Normal scan)
20H

494H

V BLK

WEN OUT(Standard)

WEN OUT(Standard) :The rising edge is effective.

4H

70H

2H

*2

Partial Pulse
6H
11

78

VIDEO OUT
(Partial scan)
9H

68H
VIDEO OUTV BLK

WEN OUT(Standard)(448H)*1WEN OUT(Standard) :The rising edge is effective.

4H
WEN OUT(Normal shutter mode)[Option]3H

WEN OUT(Other shutter mode)[Option]*3

4H*1:Arbitrary under ext.sync.
*2:Partial over actual line is OK.
*3:See "7.TIMING CHART (4)WEN Timing".

22

D4190543A

□1/120s 2:1 Interlace, ODD field
ＯＢ ２ライン13H

High-speedtransfer

68H

Normalspeedtransfer

161.5H

High-speedtransfer

Normalspeedtransfer

68H

The timing is as follows;
ODD Feild
EXT.VD
1H

8
9

492
493

VIDEO OUT
(Normal scan)
20H

242.5HV BLK

WEN OUT(Standard)

WEN OUT(Standard) :The rising edge is effective.

10H

*2

70H

2H
Partial Pulse
12H
34
35

168
169

VIDEO OUT
(Partil scan)
15H

68H
VIDEO OUTV BLK

WEN OUT(Standard)(179.5H)*1

WEN OUT(Standard) :The rising edge is effective.

10H
WEN OUT(Normal shutter mode)[Option]9H

WEN OUT(Other shutter mode)[Option]*3

10H*1:Arbitrary under ext.sync.
*2:Partial over actual line is OK.
*3:See "7.TIMING CHART (4)WEN Timing" .

23

D4190543A

□1/120s 2:1 Interlace, EVEN field
ＯＢ ２ライン

12.5H

High-speedtransfer

68H

Normalspeedtransfer

162H

High-speedtransfer

68H

Normalspeedtransfer

The timing is as follows;
EVEN Feild
EXT.VD
1H

7
8

491
492

VIDEO OUT
(Normal scan)
20H

242.5HV BLK

WEN OUT(Standard)

WEN OUT(Standard) :The rising edge is effective.

9.5H

*2

70H

2H
Partial Pulse
11.5H
33
34

167
168

VIDEO OUT
(Partial scan)
14.5H

68H
VIDEO OUTV BLK

WEN OUT(Standard)(180H)*1WEN OUT(Standard) :The rising edge is effective.

9.5H
WEN OUT(Normal shutter mode)[Option]

8.5H

WEN OUT(Other shutter mode)[Option]*3

9.5H*1:Arbitrary under ext.sync.
*2:Partial over actual line is OK.
*3:See "7.TIMING CHART (4)WEN Timing".

Note: Items shown as [Option] in this document is not included in your purchase as
standard components. Contact our dealer / distributor for details.

24

D4190543A

6. Specifications[Basic spec](1) Image sensor
Total pixels
Active pixelVideo output pixels
Scanning area
Unit cell size
(2) TV system(3) Scanning lines(4) Interlace(5) Sync system(6) Aspect ratio(7) Video output(8) Resolution(9) S/N(10) Illumination(11) Gain(12) Gamma correction(13) White-clip level(14) Power source(15) Power consumption[Internal sync spec](1) Base clock frequency(2) H sync frequency(3) V sync frequency

All Pixel’s Data Read-out Interline CCD
692(H) x 504(V)
659(H) x 494(V)
648(H) x 494(V) (Under non-interlace)
4.88(H) x 3.66(V) mm (=Equivalent to 1/3 type CCD size)
7.4(H) x 7.4(V) micro m (Square-grid array)
Special format (Non-conforming to EIA)
525 lines
1/60s Non-interlace mode
1/120s 2:1 Interlace mode
Switching via rear-panel DIP SW
Internal/External automatic switch-over
4:3
VS 1.0V(p-p) / 75Ω, DC coupled, 1 line (AC as [Option])
485 TV lines(H)485 lines (350 TV lines)(V)Standard: 52dB(p-p)/rms (Initial factory setting)Standard 400 lx (F5.6)
Minimum 4 lx (F1.4) (GAIN MAX, Approx. 50% video output)
FIX (Fixed) gain: Factory-shipped preset level
MANU (Manual) gain: Setting through GAIN VR
FIX / MANU switching via rear-panel DIP SW
Gamma = 1 (Fixed)
Approx. 840mV(p-p) (Excluding SYNC)
DC12V ±10%
Ripple voltage: 50mV(p-p) or lessApprox. 1.5W

24.545MHz (1CLK) ±200ppm
31.468kHz (1H = 780CLK)
59.94Hz (Under non-interlace)
119.88Hz (Under 2:1 interlace)

Note: Items shown as [Option] in this document is not included in your purchase as standard
components. Contact our dealer / distributor for details.

25

D4190543A

[External sync spec](1) Ext. sync input signal(2) Input impedance(3) Input level(4) Interlace(5) Polarity(6) Pulse width(7) Repeating frequency(8) Phase difference[Shutter trigger spec](1) Input level(2) Input impedance(3) Capture timing(4) Pulse width[Sync signal spec](1) Readout signal(2) Polarity(3) Output circuit

HD/VD
75-ohm / High impedance 10k-ohm (switching via rear-panel SW)(Initial factory setting: High)
From 2 through 5V (p-p) under high impedance input
From 2 through 4V (p-p) under 75-ohm input
1/60s non-interlace or 1/120s 2:1 interlace
Negative
HD: 3.2 +/- 1 micro s (LOW)VD: From 125 through 400 micro s (LOW)
fH = 31.468kHz +/- 1%
fV = fH/262.5 or fH/525
HD/VD: 0 +/- 5.0 micro s, 1/fH/2 +/- 5.0 micro s
Exposure-starting-cue signal in random trigger shutter mode
LOW level: From 0 through 0.5V(p-p)
HIGH level: From 4 through 5V(p-p)
High impedance (10k-ohm)
Rising edge detection (Positive)(Falling edge detection (Negative) as [Option])
Minimum 2 micro s
Maximum 1/8s

WEN readout timing signal
Rising edge (Positive under 1V mode [Option])* Set WEN termination to be high impedance.
5V

WEN

WEN OUT
50

100

Note: Items shown as [Option] in this document is not included in your purchase as standard
components. Contact our dealer / distributor for details.

26

D4190543A

[Electronic shutter spec](1)Normal shutter

Shutter-speed setting via rear-panel SW (Initial: OFF)8 steps switch-able (= OFF, 1/200s, 1/500s, 1/1000s, 1/2000s, 1/4000s,
1/8000s, 1/20000s)(2)RTS(a)Operation mode
No.

Reset

1

Rear SW(FIX mode)2
3
4

Non-reset
TRIG pulse width(PULSE width mode)

5
6
7
8
9
10
11

Exposure

V-resetSYNC reset
V resetRear SW(FIX mode)
TRIG pulse width(PULSE width mode)

Sync
Internal
Consecutive HD / Consecutive VD IN
Consecutive HD / Single VD IN
Internal
Consecutive HD / Consecutive VD IN
Consecutive HD / Single VD IN
Internal
Consecutive HD IN
Internal
Internal
Consecutive HD IN

Notes : * RTS mode automatically switches over through TRIG IN**RTS disabled under Electronic shutter OFF(b)Multiple shutter
Multiple shutter via ext. trigger signal and ext. VD signal
Notes : * Operation like No.3, 6 above(3)Restart / Reset

Restart / reset available via ext. VD signal(Switching via rear panel DIP SW, Initial OFF)
Notes : * The exposure-time (shutter-speed) is determined by ext.
VD interval.
** Enabled when rear-panel DIP SW OFF.
***Provide Consecutive HD.

[Partial scan](1)Operation mode
No
1
2
3
4
5
6
7
8
9

Scan mode

Sync
Internal

1/2 partial

Consecutive HD VD IN
Consecutive HD (VD) IN
Internal

1/4 partial
Programmable

Consecutive HD VD IN
Consecutive HD (VD) IN
Consecutive HD VD IN

Reset
Non-reset
V-resetNon-reset
V-resetNon-reset
V-resetNon-reset
V-resetNon-reset

E-shutter Normal
Enabled [Option]
Disabled
Enabled [Option]
Disabled
Enabled [Option]
Disabled
Enabled [Option]
Disabled
Enabled [Option]

E-shutter RTS
EnabledEnabledEnabledEnabledEnabled [Option]

Note: Items shown as [Option] in this document is not included in your purchase as
standard components. Contact our dealer / distributor for details.

27

D4190543A

(2) Reset mode
As shown in (1) above, non-reset and V-reset is available.
([Option]: Doesn’t come as standard. Contact our dealer / distributor for details)(a) non-reset (Electronic shutter enabled)
VD doesn’t get reset after video readout. The interval of VD signal is as follows.
1/60s non-interlace
1/120s interlace

1/2 partial scan 1/4 partial scan
525H
525H
262.5H
262.5H

Partial A

Partial B

Partial C

VIDEO OUTVD Interval*Note: Under normal shutter mode, when non-reset is selected on partial scan,electronic shutter is valid.
Please note that the exposure time is shortened than the setting value
when the external VD is input at shorter than the above VD interval.
(b) V-reset (Electronic shutter disabled)
VD does get reset after video readout. Under internal sync, the interval of VD signal is asfollows.
1/60s non-interlace
1/120s interlace
Partial A

1/2 partial scan 1/4 partial scan
262H
131H
131.5H
65.5H
Partial B

Partial C

Partial D

Partial E

VIDEO OUT
VD interval(3) Partial signal [Option]
Programmable mode input signal(a) Input level
LOW level: From 0 through 0.5V
HIGH level: From 4 through 5V(b) Input impedanceHigh impedance (10kΩ)(c) Polarity
Positive (Hi: High-speed transfer)

Note: Items shown as [Option] in this document is not included in your purchase as standard
components. Contact our dealer / distributor for details.

28

D4190543A

[Mechanical spec](1) External dimension(2) Weight(3) Lens mount(4) GND / insulation

29 x 29 x 39.5(D) mm (Not including protrusion)
Refer to the attached external view drawing
Approximately 41g
C mount
Circuit GND - Chassis electrically conducted[Ambient condition](1)Environment condition
Performance guaranteed Temperature: From 0 through 40 °C
Humidity: From 30 through 90 % (No condensing)Operation guaranteed
Temperature: From -5 through 45 °C
Humidity: From 10 through 90 % (No condensing)
Storage
Temperature: From -20 through 60 °C
Humidity: From 10 through 90 % (No condensing)(2)EMC conditions (Electro-Magnetic Compatibility)
EMI (Electro-Magnetic Interference)
EN61000-6-4 Conformity
EMS (Electro-Magnetic Susceptibility)
EN61000-6-2 Conformity*Conformity of EMC conditions
About the conformity of the EMC standard of this machines, it has guaranteed in theconditions combined with the option part of 4th clause.
When used combining parts other than specification of our company, I ask you to havefinal EMC conformity checked of a visitor with a machine and the whole equipment.

29

D4190543A

[Connector pin assignment](1) Compatible connector(2) Pin assignment

HR10A-10P-12S (Supplied by HIROSE ELEC.)
Connector pin layout

①
②

⑨

⑪

③

⑧

⑩
⑫

④

⑦
⑥

⑤

Picture Rear-panel camera connector(Rear-view)

12 pin male

Pin
No.
1
2
3
4
5
6
7
8
9
10
11
12

Signal(Standard)
DC12V GND
DC12V
VIDEO GNDVIDEO OUTHD GNDHD IN
VD IN
TRIG GNDNC
WEN OUT
TRIG INVD GND

Signal
[Option]NC
TRIG IN
GND
DC12V
PARTIALNC

Notes : *Before connecting / disconnecting the connector, make sure the camera power is OFF.
**For board connection, check compatibility.
Note: Items shown as [Option] in this document is not included in your purchase as standard
components. Contact our dealer / distributor for details.

30

D4190543A

[Switch setting](1) CCU rear-panel DIP SW
No.

FunctionOFF

1

GAIN selection(GAIN)

Factory-set
GAIN

2
3
4

E-shutter-speed(SHUT)

See shutter-speed table (Table 1)

5
6
7
8
9
0

Video output
mode (VIDEO)
Shutter mode
(SMODE)
Partial scan
(PART)
RTS Exposure(EXP)

Notes:

1/60s non-interlace

ON
Manual GAIN
adjustable
via GAINGAIN 1
2
SHUT 3
4
VIDEO 5
6
SMODE
7
8
PART
9
EXP 0

1/120s interlace

See shutter-mode table (Table 3)
See partial-scan table (Table 2)
FIX mode*

PULSE W mode

Initial factory setting: All OFF(Table 1) Electronic shutter-speed(Table 2) Partial-scan

Shutter-speedOFF
1/200s1/500s1/1,000s
1/2,000s
1/4,000s
1/8,000s
1/20,000s

Partial scan No.8 No.9
OFF
OFF OFFNot acceptable ON OFF
1/2 partial
OFF ON
1/4 partial
ON ON

No.2
OFF
ON
OFF
ON
OFF
ON
OFF
ON

O
N

No.3
OFF
OFF
ON
ON
OFF
OFF
ON
ON

No.4
OFF
OFF
OFF
OFF
ON
ON
ON
ON

Notes : *Don’t set Electronic shutter-speed in OFF under RTS mode.
(Table 3) Shutter-mode
Shutter mode
V resetRandomSYNC reset
trigger
Non-reset
Not acceptable
Non-reset
Random(Multiple shutter)
trigger
Non-reset
V resetRestart / Reset

SYNC

No.6 No.7
OFF OFFON OFFOFF ON
ON ON

Internal sync

OFF OFF

Single VD

ON OFF Consecutive VD
OFF ONNo VD
ON ONSingle VD

Ext. sync
HD IN

Notes : * Under normal shutter mode partial-scan, set No.6,7 in OFF.
**Under PULSE W mode, SYNC reset is disabled.
(2)CCU rear-panel SW
FunctionSW

Selected Function

Ext. SYNC IN impedance

HIGH

HIGH impedance (Initial factory setting)(HD/VD)

75Ω

75Ω

31

D4190543A

[Relative Spectrum Response]*Including lens characteristics, Excluding light source characteristics[Optical black characteristics]
2

Effective
Image Area

V

494

504

8
2

659

31

692
H
Device structure
Total pixels : 692(H) x 504(V)
Effective pixels : 659(H) x 494(V)
Optical black
Horizontal : 2pixels --- 31pixels
Vertical : 8pixels --- 2pixels

32

D4190543A

33

WEN
OUT

CAM
VIDEO
OUT

78CLK
(3.18µs)

INT HD

EXT HD

6 6
5 5
5 6

6
5
9 O
B
1

18CLK
(0.73µs)

4CLK±0.5µsO
B
3
1

H. trans.
pause

72CLK

60CLK
(2.44µs)

D O O 1
1 B B
6 1 2

2CLK

54CLK
(2.20µs)

Dummy Opt.
pixels Blk.

D
1

16CLK

7CLK

Horizontal scan (1H)
780CLK(31.78µs)

H. blanking
132CLK(5.38µs)

H.SYNC

Opt.Blk.

31CLK

7 8

648CLK(26.40µs)

V-OUT Effective pixels

Effective pixels = 659CLK

648CLK
6 6
5 5
5 6

4CLK
6
5
9

1CLK=40.74ns

7. Timing Chart(1)H rate timing

D4190543A

(2) 1/60s Non-interlace mode
EXT VD IN
1H delay

6

5

4

3

2

492 493 494

1

3

525

2

517

1

516

515

512

514

23

513

22

20

19

18

17

16

15

14

13

OB OB OB OB OB OB OB OB

21

VIDEO
OUT

12

11

9

10

8

7

6

5

3

2

1

525

524

523

INT HD4

9H

INT VD

OB OB

V.Blanking
20H

WEN OUT(Standard)

WEN OUT(Standard):The rising edge is effective.

Ext. VD – Ext. HD phase difference
EXT VD IN

EXT HD IN
TP2TP1

INT VDTP1 : 10.0 us
TP2 : 5.0 us

34

D4190543A

(3) 1/120s 2:1 Interlace modeODD (1st field)
EXT VD IN
1H delay35

10 12 14 16 18 20 22 24 26 28 30 32 34 36
11 13 15 17 19 21 23 25 27 29 31 33 35 37

34

32

8
9

33

31

30

29

28

27

26

25

24

6
7

23

4
5

22

20

2
3

21

18

OB OB OB OB
OB
OB OB OB 1

19

17

16

15

14

13

12

11

9

10

8

7

6

4

3

2

1

525

524

523

5

9H

INT VD

INT HD487 489 491
488 490 492

VIDEO OUT

493 OB494 OB

V Blanking
20H
WEN OUT(Standard)

WEN OUT(Standard):The rising edge is effective.

EVEN (2nd field)
EXT VD IN
1H delay

297

296

295

294

293

292

291

290

289

288

287

5
6

286

282

3
4

285

281

1
2

284

280

OB OB OB OB
OB OB OB OB

283

279

278

277

276

275

274

273

272

271

270

269

268

266

265

264

263

262

261

260

267

9H

INT VD78

9 11 13 15 17 19 21 23 25 27 29 31 33 35
10 12 14 16 18 20 22 24 26 28 30 32 34 36

INT HD486 488 490 492
487 489 491 493

VIDEO OUT

494
OB
OB

V Blanking
20H

WEN OUT(Standard)

WEN OUT(Standard):The rising edge is effective.

Ext. VD – Ext. HD phase difference
EXT VD IN

(1)

(2)

EXT HD IN
TP3

TP3

TP1

(1)

INT VD

ODD

TP2(2) EVEN

INT VDTP1 : ODD reset range
TP2 : EVEN reset range
TP3 : 5.0 us

35

10.0 us10.0 usD4190543A

(4) WEN timing(a)Normal shutter mode
<1/60s Non interlace mode>
Under normal scan mode
9H

INT VDOB

VIDEO OUT
(Normal scan)

494

1 2

OB

V.Blanking

WEN OUT(Standard)

4H

WEN OUT[Option]

4H

V.Blanking

WEN OUT(Standard):The rising edge is effective.
521H

3H

Under partial scan V-reset mode (Standard)
9H

INT VD
VIDEO OUT
(Partial scan)

High-speedtransfer

OB

V.Blanking

WEN OUT(Standard)

4H

WEN OUT[Option]

4H

V.Blanking

WEN OUT(Standard):The rising edge is effective.
3H** 1/2 partial scan : 258H
1/4 partial scan : 127H

Under partial scan Non-reset mode [Option]
9H

INT VD
VIDEO OUT
(Partial scan)

High-speedtransfer

OB

V.Blanking

WEN OUT(Standard)

4H

WEN OUT[Option]

4H

V.Blanking

WEN OUT(Standard):The rising edge is effective.
521H

36

3H

D4190543A

<1/120s Interlace modeODD Field>

Under normal scan mode
INT VD

9H

VIDEO OUT
(Normal scan)

8
9

492
493

OB

V.Blanking

WEN OUT(Standard)

10H

WEN OUT[Option]

10H

V.Blanking
WEN OUT(Standard):The rising edge is effective.
252.5H

8.5H

Under partial scan V-reset mode (Standard)
INT VD
VIDEO OUT
(Partial scan)

9H

High-speedtransfer

OB

494
OB

V.Blanking

V.Blanking

WEN OUT(Standard)

10H

WEN OUT[Option]10H

WEN OUT(Standard):
The rising edge is effective.
8.5H** 1/2 partial scan : 121.5H
1/4 partial scan : 55.5H

Under partial scan Non-reset mode [Option]
INT VD
VIDEO OUT
(Partial scan)

9H

High-speedtransfer

OB

494
OB

V.Blanking

WEN OUT(Standard)

10H

WEN OUT[Option]10H<1/120s Interlace mode

V.Blanking
WEN OUT(Standard):
The rising edge is effective.
252.5H

8.5H

EVEN Field>

Under normal scan mode
INT VD
VIDEO OUT
(Normal scan)

9H

7
8

491
492

V.Blanking

WEN OUT(Standard)

9.5H

WEN OUT[Option]

9.5H

OB
OB

V.Blanking
WEN OUT(Standard):The rising edge is effective.
253H

9H

Under partial scan V-reset mode (Standard)
INT VD

9H

VIDEO OUT
(Partial scan)

V.Blanking

WEN OUT(Standard)

9.5H

WEN OUT[Option]

9.5H

High-speedtransfer

OB
OB

V.Blanking
WEN OUT(Standard):
The rising edge is effective.
9H** 1/2 partial scan : 122H
1/4 partial scan : 56H

Under partial scan Non-reset mode [Option]
INT VD

9H

VIDEO OUT
(Partial scan)

V.Blanking

WEN OUT(Standard)

9.5H

WEN OUT[Option]

9.5H

High-speedtransfer

OB
OB

V.Blanking
WEN OUT(Standard):
The rising edge is effective.
253H

37

9H

D4190543A

(b)RTS
Non-reset mode (Consecutive VD)<1/60s Non interlace mode>
Under normal scan mode
9H

INT VDOB

VIDEO OUT
(Normal scan)
WEN OUT(Standard)

494

1 2

OB

V.Blanking
4H

WEN OUT(Standard):The rising edge is effective.

WEN OUT[Option]

521H

3H

4H

Under partial scan mode
9H

INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)

High-speedtransfer

OB

V.Blanking

V.Blanking
WEN OUT(Standard):The rising edge is effective.

4H

WEN OUT[Option]

521H

3H

4H

<1/120s Interlace modeODD Field>

Under normal scan mode
INT VD
VIDEO OUT
(Normal scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

8
9

492
493

OB

V.Blanking

V.Blanking
WEN OUT(Standard):The rising edge is effective.

10H

8.5H

252.5H
10H

Under partial scan mode
INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

High-speedtransfer

OB

494
OB

V.Blanking

V.Blanking
WEN OUT(Standard):
The rising edge is effective.

10H

252.5H

8.5H

10H<1/120s Interlace mode

EVEN Field>

Under normal scan mode
INT VD
VIDEO OUT
(Normal scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

7
8

491
492

OB
OB

V.Blanking
9.5H

V.Blanking
WEN OUT(Standard):The rising edge is effective.
253H

9H

9.5H

Under partial scan mode
INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

High-speedtransfer
OB
OB

V.Blanking

V.Blanking
9.5H

WEN OUT(Standard):
The rising edge is effective.
253H

9H

9.5H

38

D4190543A

(c)RTS
Non-reset mode (Single VD) / V-reset mode / SYNC-reset mode
<1/60s Non interlace mode>
Under normal scan mode
9H

INT VDOB

VIDEO OUT
(Normal scan)
WEN OUT(Standard)

494

1 2

OB

V.Blanking
WEN OUT(Standard):The rising edge is effective.

4H

WEN OUT[Option]

521H

3H

4H

Under partial scan mode
9H

INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)

High-speedtransfer

OB

V.Blanking

V.Blanking

WEN OUT(Standard):The rising edge is effective.

4H

WEN OUT[Option]

3H

*

4H

* 1/2 partial scan : 258H
1/4 partial scan : 127H
Programable partial scan : Arbitrary [Option]<1/120s Interlace modeODD Field>

Under normal scan mode
INT VD
VIDEO OUT
(Normal scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

492
493

8
9

OB

V.Blanking
10H

WEN OUT(Standard):The rising edge is effective.
252.5H

8.5H

10H

Under partial scan mode
INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

High-speedtransfer 494
OB OB

V.Blanking

V.Blanking
10H

WEN OUT(Standard):
The rising edge is effective.
8.5H

*

10H

* 1/2 partial scan : 121.5H
1/4 partial scan : 55.5H
Programable partial scan : Arbitrary [Option]<1/120s Interlace mode EVEN Field>*Under Non-reset mode (Single VD) only.
Under normal scan mode
INT VD
VIDEO OUT
(Normal scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

7
8

491
492

OB
OB

V.Blanking
9.5H

WEN OUT(Standard):The rising edge is effective.
253H

9H

9.5H

Under partial scan mode
INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

High-speedtransfer

OB
OB

V.Blanking

V.Blanking

9.5H
9.5H

WEN OUT(Standard):
The rising edge is effective.
9H** 1/2 partial scan : 122H
1/4 partial scan : 56H
Programable partial scan : Arbitrary [Option]

39

D4190543A

(d)Restart / reset mode
<1/60s Non interlace mode>
Under normal scan mode
9H

INT VDOB

VIDEO OUT
(Normal scan)
WEN OUT(Standard)

494

1 2

OB

V.Blanking
WEN OUT(Standard):The rising edge is effective.

4H

WEN OUT[Option]

521H

3H

4H

Under partial scan mode
9H

INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)

High-speedtransfer

OB

V.Blanking

V.Blanking
WEN OUT(Standard):The rising edge is effective

4H

WEN OUT[Option]

521H

3H

4H

<1/120s Interlace modeODD Field>

Under normal scan mode
INT VD
VIDEO OUT
(Normal scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

492
493

8
9

OB

V.Blanking
WEN OUT(Standard):The rising edge is effective.

10H

252.5H

8.5H

10H

Under partial scan mode
INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

High-speedtransfer

OB

494
OB

V.Blanking

V.Blanking
WEN OUT(Standard):The rising edge is effective

10H

252.5H

8.5H

10H<1/120s Interlace mode

EVEN Field>

Under normal scan mode
INT VD
VIDEO OUT
(Normal scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

7
8

491
492

OB
OB

V.Blanking

9.5H

WEN OUT(Standard):The rising edge is effective.
253H

9H

9.5H

Under partial scan mode
INT VD
VIDEO OUT
(Partial scan)
WEN OUT(Standard)
WEN OUT[Option]

9H

High-speedtransfer

OB
OB

V.Blanking

V.Blanking

9.5H

WEN OUT(Standard):The rising edge is effective
253H

9H

9.5H

40

D4190543A

8. Guarantee
The term of guarantee is one year after the product delivery.
If by any chance trouble by responsibility of our company occurs before an above period, TELI
repairs it free of charge.
-During terms of a guarantee, when the trouble cause is the case of below, TELI charges the repair
costs.
(1) Troubles and the damages that causes by misuse, unsuitable repair of remodeling.
(2) Distribution hazards like drops and vibrations after purchase. Troubles and damages bytransportation.
(3) Troubles and damages by fire, natural calamity (earthquake, storm and flood damage,thunderbolt), damages from salty breeze, gas harm, abnormal voltage.

9. Repair(1) Condition for repair
Basically, has to return it to our company when the user requests us to repair product.
Beside that, customer should pay these expenses (travel expenses, camera disassembly technology
costs) of both customer and end user. Also customer should pay in themselves costs for returncamera to us.
(2) The period of repairing product- Repair free of charge
Refer to Clause 8.
-Charged repair
Basically, repair period is 7 years after the last production end of
products.

41

D4190543A

10. External-View Drawing

42

D4190543A

Head Office : 7-1, 4 chome, Asahigaoka, Hino-shi, Tokyo, 191-0065, Japan(International Business Department)
Phone : +81-42-589-8771
Fax: +81-42-589-8774
URL: http://www.toshiba-teli.co.jp/

Distributor• This product must be classified for disposal according to the laws of each country and municipal laws.
• Information contained in this document is subject to change without prior notice.

43

D4190543A