Ultra-high-speed chart measurements
for calibration, profiling, and quality control.

Fast
MYIRO-9 can read 1,500 patches in about 4 minutes and 2,500 patches in only about 7 minutes.
Plus, a single scan can provide data for M0, M1 and M2, conforming to ISO13655:2009 and offering improved certainty of colors and OBA.

Optional Auto Sheet Feeder for automatic
reading of large numbers of sheets
Users who need to measure dozens of sheets every day
can benefit from using the optional Auto Sheet Feeder.
This optional accessory enables up to 100 sheets tobe preloaded and measured with minimal user input.
When used with the QR code reader function, themeasurement results can be automatically linked tothe printed sheet, so even multi-page charts canbe handled easily and automatically.

VersatileFree-format function
The free-format function of the MYIRO-9 enables it to measure virtually every chart thatmight be used to calibrate printers and create profiles. Even quality control wedges, used
for conformance testing of proof prints, etc., can be measured with the MYIRO-9 without
cutting them from the print. Color unevenness of the printed surface can be evaluated up
to the edge of the paper.

Virtual Fluorescent Standard
MYIRO-9's Virtual Fluorescent Standard (VFS) allows the creation of precision custom profiles to compensate for the environment where the print will be viewed, so users can obtain
amazing results and achieve the best possible neutral grey balance and flawless skintones ineven non-standard viewing environments.

USB and Ethernet connections
MYIRO-9 has both USB and Ethernet connections, so one MYIRO-9 can be easily shared among
multiple PCs and multiple printing machines.

AccurateMYIRO-9's built-in wavelength calibration and temperaturedrift compensation maximize day-to-day accuracy and reliability.

Detailed specifications of MYIRO-9 are available on MYIRO's website.
https://www.myiro.com/MYIRO-9

Worldwide standardized service for repair, certification and adjustments to absolute accuracy lets users offer their customers true local or
global consistency.
Users who want to monitor and audit the absolute accuracy of the system on aregular basis will benefit from the MYIRO-9's self-check function, which uses certified tiles specifically designed for MYIRO-9 to monitor the accuracy at any time

Color tiles (Order-made)Main Specifications of MYIRO-9
Illumination/viewingsystem
Wavelength range
Wavelength pitch
Half bandwidth
Measurement area
Light source
Measurement range
Repeatability
Inter-instrument agreement
Measurement time
Measurement conditions*2
Interface
Size (W × D × H)

45°a: 0°(annular illumination)*1
Conforms to CIE No. 15, ISO 7724/1, DIN5033 Teil 7, ASTM E 1164, and JIS Z 8722 Condition a for reflectance measurements.
380 to 730 nm
10 nm
Approx. 10 nm
Approx. ø3 mm
LEDSpectral reflectance: 0 to 150%
Colorimetric: Within ΔE00 0.05
(When white plate is measured 30 times at 10-second intervals after white calibration has been performed)
Within ΔE00 0.3 (Average for 12 BCRA Series II color tiles compared to values measured with a master body)
Approx. 4 min per 1,500 patches (* Manufacturer's standard conditions*3)
M0 (A), M1 (D50) , M2 (A + UV-cut filter), C, ID50, D65, ID65, F2, F6, F7, F8, F9, F10, F11, F12, User-defined illuminant
USB2.0, 100Base-TX
MYIRO-9 only :716 x 251 x 159 mm
Auto Sheet Feeder :601 x 1,158 (when tray is open) x 245 mm
MYIRO-9 mounted on Auto Sheet Feeder : 716 x 1,158 x 256 mm

Measurable Chart Specifications *4
WidthLength
Thickness
Smallest patch size
Maximum patches
per sheet
Margins (minimum)
*1
*2
*3
*4

45 to 330 mm (When using the Auto Sheet Feeder: 90 to 320 mm)
170 to 1,500 mm (When using the Auto Sheet Feeder: 170 to 450 mm)
0.05 to 0.6 mm (When using the Auto Sheet Feeder: 0.08 to 0.25 mm)
6×6 mm
1,394 (A4)2,928 (A3)
Leading: 23 mm; Trailing: 33 mm; Left/right sides: 4 mm each

Illumination for wavelengths under 400 nm is unidirectional.
M0, M1 and M2 are lighting conditions contained in ISO13655 4.2.2 Illumination requirements and measurement conditions.
Chart size: A3, Patch size: 6 x 6 mm, Patch layout: 32 rows x 47 columns
Even if paper size is within the ranges specified, some charts may not be measured.

Dimensions

System Diagram

Auto Scan Spectrophtometer MYIRO-9

Auto Sheet Feeder FD-A09

Auto Scan Spectrophtometer

Auto Sheet Feeder

MYIRO-9

FD-A09USB Cable (3m)457

Measurement
Utility Software

IF-A181158
461

FD-S2w
235

AC Adapter
173
200
245

339
225

156
121
108

159

251

39

Color tiles

175
337

601
343
205

199

716

Computer(commercially available)
Optional accessories
Order-made accessories

SAFETY PRECAUTIONS
For correct use and for your safety, be sure to read the instruction manual before using the instrument.

• Always connect the instrument to the specified power supply voltage.
Improper connection may cause a fire or electric shock.

• Displays shown are for illustration purpose only.
• The specifications and drawings given here are subje ct to change without
prior notice.

For additional information, please visit:https://www.myiro.com•Sales contact:

9242-A8AN-42 CCBDK 2