CCD Camera
CleverDragon series
CSCS20BC2
Specification

CONTENTS
Restriction For Use ··············································· 2
Exemption Clauses ················································ 3
Notes on using this product ······································ 4
1. Overview ························································ 7
2. Features ·························································· 7
3. Configuration ··················································· 8
4. Option parts ····················································· 8
5. Function ························································· 8
6. Specification ··················································· 13
7. Timing Chart ··················································· 18
8. Notes regarding specifications ······························ 23
9. Guarantee ······················································· 24
10. Repair ························································· 24
11. External-view Drawing ····································· 25( 1 / 25 )D4159008D

Restriction For UseShould the equipment be used in the following conditions or environments, give consideration tosafety measures and inform us of such usage:
1. Use of the equipment in the conditions or environment contrary to those specified, or use outdoors.
2. Use of the equipment in applications expected to cause potential hazard to people or property, which
require special safety measures to be adopted.

This product can be used under diverse operating conditions. Determination of applicability ofequipment or devices concerned shall be determined after analysis or testing as necessary by thedesigner of such equipment or devices, or personal related to the specifications. Such designer or
personal shall assure the performance and safety of the equipment or devices.

This product is not designed or manufactured to be used for control of equipment directly concernedwith human life (*1) or equipment relating to maintenance of public services/functions involving
factors of safety (*2). Therefore, the product shall not be used for such applications.

(*1): Equipment directly concerned with human life refer to:
Medical equipment such as life-support systems, equipment for operating theaters.
Exhaust control equipment for exhaust gases such as toxic fumes or smoke.
Equipment mandatory to be installed by various laws and regulations such as the Fire Act or
Building Standard Law.
Equipment related to the above.
(*2): Equipment relating to maintenance of public service/functions involving factors of safety refer to:
Traffic control systems for air transportation, railways, roads, or marine transportation.
Equipment for nuclear power generation.
Equipment related to the above.

Although sufficient check is performed about translation of these specifications, we will apply aJapanese sentence, if a doubt should occur.

( 2 / 25 )D4159008D

Exemption Clauses


TELI assumes no responsibility or liability for damage arising from fire, earthquake, an act by a third
party or other accidents, or intentional or careless error or misuse by the user, or use under abnormal
conditions.



TELI assumes no responsibility or liability for incidental damages (e.g., loss of business profits orinterruption of business) arising from use of or inability to use the camera equipment.



TELI assumes no responsibility or liability in the case damages or losses are caused by failure toobserve the information contained in the operation manual and specifications.



TELI assumes no responsibility or liability in the case damages or losses are caused by use contrary to
the instructions in this operation manual and specifications.



TELI assumes no responsibility or liability in the case damages or losses are caused by malfunction orother problems resulting from use of equipment or software that is not specified.



TELI assumes no responsibility or liability in the case damages or losses are caused by repair ormodification conducted by the customer or any unauthorized third party (such as an unauthorized
service representative).



Expenses we bear on this product shall be limited to the individual price of the product.



TELI does NOT guarantee the items that are not described in the specification.

( 3 / 25 )D4159008D

Notes on using this product Handle carefully
Do not drop the equipment or allow it to be subject to strong impact or vibration, as such action may
cause malfunctions. Further, do not damage the connection cable, since this may cause wire breakage.
 Environmental operating conditions
Do not use the product in locations where the ambient temperature or humidity exceeds the
specifications.
Otherwise, image quality may be degraded or internal components may be adversely affected. In
particular, do not use the product in areas exposed to direct sunlight. Moreover, during shooting underhigh temperatures, vertical stripes or white spots (noise) may be produced, depending on the subject orcamera conditions (such as increased gain). However, such phenomena are not malfunctions.
 Regarding a lens mount
Install a next lens; the C mount lens, its dimension of protrusion from flange is equal to or less than 7.9mm. If a lens does not stand to this condition, it might not be installed to this camera.
C-mount lensflange7.9mm or less

 Check a combination with the lens
Depending on the lens and lighting you use, an image is reflected as a ghost in the imaging area.
However, this is not because of a fault of the camera.
In addition, depending on the lens you use, the performance of the camera may not be brought out
fully due to deterioration in resolution and brightness in the peripheral area, aberration and others.
Be sure to check a combination with the camera by using the lens and lightning you actually use.
When installing a lens in the camera, make sure carefully that it is not tilted.
In addition, use a mounting screw free from defects and dirt. Otherwise, the camera may be unable tobe removed.

( 4 / 25 )D4159008D

Notes on using this product Avoid intensive light
Do NOT expose the camera's image-pickup-plane to sunlight or other intense light directly. If the partof CCD is exposed to spot-intensive light, you might get a picture problem like blooming and/or smear.
Under the comparison at the same video output level, the shorter the exposure time setting, the moresmear is generated.
 Do not expose the camera’s image-pickup-plane to sunlight or other intense light directly.
Its inner CCD (charge-coupled device) might be damaged.
 Occurrence of moire
If you shoot thin stripe patterns, moire patterns (interference fringes) may appear. This is not a
malfunction.
 Occurrence of noise on the screen
If an intense magnetic or electromagnetic field is generated near the camera or connection cable, noise
may be generated on the screen. If this occurs, move the camera or the cable.
 Handling of the protective cap
If the camera is not in use, attach the lens cap to the camera to protect the image pickup surface.
 If the equipment is not to be used for a long duration
Turn off power to the camera for safety.
 Maintenance
Turn off power to the equipment and wipe it with a dry cloth.
If it becomes severely contaminated, gently wipe the affected areas with a soft cloth dampened withdiluted neutral detergent. Never use alcohol, benzene, thinner, or other chemicals because suchchemicals may damage or discolor the paint and indications.
If the image pickup surface becomes dusty, contaminated, or scratched, consult your sales
representative.

( 5 / 25 )D4159008D

CAUTIONS ON USE

 When disposing of the camera
Wastes of this product should be separated and discarded in compliance with the various national andlocal ordinances.
This camera is showing the following symbol to body due to EU environmental regulation (Waste
Electrical and Electronic Equipment (WEEE)). However this symbol is applied to only a EU member
state.

( 6 / 25 )D4159008D

1. Overview
CleverDragon series CSCS20BC2 is an integrated type B/W CCD camera with a SXGA format
all-pixel-data readout CCD. The model is suited for high-resolution image processing use. Its compact,
light-weight body is ideal for system integration.

2. Features(1) All-pixel reading
The all-pixel reading system allows the CSCS20BC2 to read all pixels in just 1/19.5 second.
CSCS20BC2 is equipped with a full-frame shutter that allows all-pixel reading even during shutter
operations.
(2) Full-frame shutter
Since all pixels are output even by a random trigger shutter operation, high resolution can be achieved,
without deteriorating the vertical resolution.
(3) Tetragonal lattice layout
The tetragonal lattice layout of CCD pixels facilitates computation for image processing.
(4) Camera Link interface (power supply type)
By using a Camera Link-capable frame grabber board to which power can be supplied, high-speedtransfer of captured images to a PC as well as various types of camera control from the PC are allowed.
Power can also be supplied to the camera with only one cable.
(5) Random trigger shutter function
CSCS20BC2 is equipped with a random trigger shutter, which starts exposure synchronized withexternal trigger signals. Fast-moving objects can thus be captured in place, which ensures accurate
image processing.
(6) Restart-Reset
Images can be shot and fetched at arbitrary timing based on external VD signal input.
(7) Partial scan
Further speed-up is possible because ranges except the range of the image output that the user set are notread.
(8) Ultra-compact and lightweight main unit
The space-saving ultra-compact and lightweight camera has excellent resistance against vibration andimpact.
(9) RoHS compliant
CleverDragon series are complied with EU RoHS.

( 7 / 25 )D4159008D

3. Configuration
Camera body ............................................................................................. 1
Manual................................................................................................. 1

4. Option parts(1) Camera mounting kit

CPT8560*NOTE: Contact your dealer / distributor for details of option units.
*NOTE: Application software is not supplied as a standard item.

5. Function
5-1. Serial communication control
By CameraLink serial communication interface, it is possible to control the following functions.
(1) Set-up Addition Value

10bit: +0 to +255 [digit]
8bit: +0 to +63 [digit](2) Gain

0dB to Approx. +12dB(3) Shutter Speed

2[sec] to 1/20,000[sec](4) Random Trigger Shutter

ON / OFF* Restart-Reset mode must be disabled.

・Trigger / VD Polarity

Positive / Negative・Random Trigger ModeFix / Pulse Width(5) Restart-Reset

ON / OFF* Random Trigger Shutter mode must be disabled.

(6) Partial Scan

Video output start line, Video output width
Minimum width: 512H

5-2. Power supply from PoCL cable
Pin 1, 26+12V±10% (ripple level: 50mVP-P or less)( 8 / 25 )D4159008D

5-3. Command Communication Protocol
The command communication protocol is the teli standard method (method in which parameters are set
in the registers in the camera).
In command send/receive operation, hexadecimal address and data are converted to ASCII data.
All ASCII alphabetic characters used are uppercase characters.

(1) Write to a register
To write data in a register, send a command, as follows. (Address' max-length is 2 bytes, and Data's
max-length is 8 bytes)
Address
2nd byte

Address
1st byte‘,’(0x2C)

Data
8th byte

Data
7th byte

Data
6th byte

Data
5th byte

Data
4th byte

Data
3rd byte

Data
2nd byte

Data
1st byte[CR](0x0D)For example, to write data 0x38 to address 0x76, send a command, as follows:‘7’(0x37)‘6’(0x36)‘,’(0x2C)‘3’(0x33)‘8’(0x38)[CR](0x0D)

The camera responds to the write command with No Error (ACK) or Error (NAK), as follows:
No Error (ACK):[ACK](0x06)[CR](0x0D)

Error (NAK):[NAK](0x15)[CR](0x0D)*Because two kinds of data is needed for the setting about the Partial Scan, the register writing for "Setvalue application" is separately needed.

( 9 / 25 )D4159008D

(2) Reading the register
To read data from a register, send ', (comma)', 'R', 'Q' and [CR] code following the address. For example,
to read data in address 0x91, send a command, as follows:‘9’(0x39)‘1’(0x31)‘ ,’(0x2C)‘R’(0x52)‘Q’(0x51)[CR](0x0D)

The camera responds to the read request, as follows (Data's max-length is 8 bytes):Data
8thby teData
7thby teData
6thby teData
5thby teData
4thby teData
3rd by teData
2nd by teData
1st by te[CR](0x0D)

Actually, the camera responds to the read request as minimum data length: For example, to read data
0x10 to address 0x91, the camera responds as follows:‘1’(0x31)‘0’(0x30)( 10 / 25 )[CR]
(0x0D)

D4159008D

5-4. Register Map

Address
0x00
|
0x0F
0x10
|
0x2F
0x30
|
0x3F
0x40
|
0x47
0x48
|
0x4F
0x50
|
0x57
0x58
|
0x5F
0x60
|
0x67
0x68
0x69
0x6A
0x6B
0x6C
0x6D
0x6E
0x6F
0x70
0x71
|
0x75
0x76
0x77
|
0x7F

Read
Write

B&W Camera
CSCS20BC2

Maker Name
R.O. ASCII Format

Maker NameTOSHIBA TELI

Model Name
R.O. ASCII Format

Model Name
CSCS20BC2

Serial Number
R.O. ASCII Format

Serial Numberex) 0100011

Firmware Version
R.O. ASCII Format

Firmware Versionex) 01.01.01

FPGA Version
R.O. ASCII Format

FPGA Versionex) 01.01.01

CPLD1 Version
R.O. ASCII Format

CPLD1 Versionex) 01.01.01

N.A. Reserved||
N.A. Reserved
Register Map Version
R.O. ASCII Format
N.A. Reserved
R.O. Status
R.O. Expanded Status
N.A. Reserved
R.O. Memory Bank Confirmation
W.O. Save to Memory
R/W Load from Memory
W.O. Initialize Memory
R/W Setup (Offset) Addition Value
N.A. Reserved||
N.A. Reserved
R/W Gain
N.A. Reserved||
N.A. Reserved
R/W
R.O.
W.O.
N.A.

Read/Write
Read Only
Write Only
Not Available

Register Map Versionex) 01.01
StatusStatus information after Camera Controlling
Expanded Status
Detail information of the status
Memory Bank Confirmation
The information of memory banks the setting saved.
0x01(Saved), 0x00(Not saved)
Saved to Memory
Write 0x01 : Save the current camera settings.
The camera uses saved settings after rebooting.
Read from Memory
Write 0x01 : Load saved settings.

Initialize Memory
Write 0x01 : Delete saved user settings.
Setup (Offset) Addition Value
0-255
Default : 0
Gain0(0dB)-90(Approx.+12dB)
Default : 0(0dB)( 11 / 25 )D4159008D

Address

Read
Write

B&W Camera
CSCS20BC2

0x80

R.O. Frame Rate

0x82

R.O. Horizontal Resolution

0x84

R.O. Vertical Resolution

0x86
0x87
0x88
|
0x8F
0x90
0x91
0x92
0x93
0x94
|
0x9F

N.A.
R/W
N.A.
|
N.A.
R/W
R/W
R/W
R/W
N.A.
|
N.A.

0xA0
0xA2
0xA3
0xA4
0xA5
|
0xBF
0xC0
0xC1
|
0xC3
0xC4

Reserved
Output Bit
Reserved|
Reserved
Scan Mode
Shutter Mode
Random Trigger Shutter Mode
Trigger Polarity
Reserved|
Reserved
R/W Shutter Speed (denominator)
N.A. Reserved
N.A. Reserved
R/W Shutter Speed (numerator)
N.A. Reserved||
N.A. Reserved
W.O. Partial Scan Update
N.A. Reserved||
N.A. Reserved
R/W Video Start Line @ Partial Scan

0xC8

N.A. Reserved
N.A. Reserved
R/W Video Width @ Partial Scan

0xCA
|
0xFF

N.A. Reserved||
N.A. Reserved

0xC6
0xC7

R/W
R.O.
W.O.
N.A.

Read/Write
Read Only
Write Only
Not Available

Frame Rate (@Normal Shutter)
Normal : 19fps
Partial : The calculation from partial video width.
(Omit it after the decimal point. )Default : 19fps @ Normal Scan
Horizontal Resolution
1360 (fixed)
Vertical Resolution
1024 @ Normal Scan
Video Width @ Partial Scan
Default : 1024 @ Normal Scan
Output Bit
8(8bit),10(10bit)
Default : 8(8bit)
Scan Mode0(Normal),1(Partial)
Default : 0(Normal)
Shutter Mode0(Normal Shutter),1(Random Trigger),2(Restart Reset)
Default : 0(Normal Shutter)
Random Trigger Shutter Mode0(Fix),1(Pulse Width)
Default : 0(Fix)
Trigger Polarity
0(Nega),1(Posi)
Default : 0(Nega)
Shutter Speed (denominator)
1-20000
Default : 19
Shutter Speed (numerator)
1-255
Default : 1
Partial Scan Update
Write 0x01 : Update registers related with Partial Scan
Video Start Line @ Partial Scan
0-512
Default : 0
Video Width @ Partial Scan
512-1024
Default : 1024

( 12 / 25 )D4159008D6. Specification[Electrical specification](1) Imager

all-pixel-data-readout interline transfer CCD

Number of total pixels

1434(H)  1050(V)

Number of effective pixels

1392(H)  1040(V)

Number of Video out pixels

1360(H)  1024(V)Scanning area

6.47mm(H)  4.84mm(V)(= Equivalent to 1/2” type CCD size)
4.65m(H)  4.65m(V) (Square-grid array)

Pixel size(2) Scan method

Non- interlace(3) Synchronization method

Internal synchronization(4) Aspect ratio

4:3(5) Video Output

Compliant with CameraLink standard version 1.2

Data

10 / 8 bit switching (factory default: 8bit）

Readout mode
All pixel readout (factory default)

1360(H)  1024(V) [Approx. 19.5fps]

Partial Scan (ex.1)

1360(H)  768(V) [Approx. 25.0fps](ex.2)

1360(H)  512(V) [Approx. 34.8fps]
minimum lines: 6512, minimum step: 1 line

(6) Sensitivity

500 lx, F5.6

(7) Minimum subject illuminance5lx F1.4 (GAIN: 10dB, video level: 50 %)(8) Gain0 to Approx. +12 dB [1step= Approx. 0.132dB]

Notes on gain setting:*It is possible to set a maximum of +12 dB (Calculation value) but the warranty range for this productis 0 to +10 dB. When using this product, be sure to set a gain value within the warranty range.
And, setting a too high gain value can increase noise. When you adjust the brightness of the shot
image, you are responsible for finally confirming the image quality by using the entire
machine/equipment.
(9) Setup Level(factory default)

40 ± 20 [digit] (10bit)
10 ± 5 [digit] (8bit)(addition value)+0 to +255 [digit] (10bit, 255 steps)+0 to +63 [digit](8bit, 255 steps)(10) Gamma correction

OFF ( = 1.0 fixed)(11) Power supply voltage

DC12V ± 10% (ripple 50mVP-P or less)(12) Power consumption

Approx. 1.8W

( 13 / 25 )D4159008D

[Internal sync signal specification](1) Driving frequency

36.00 MHz (1 CLK) ±100ppm(2) Horizontal sync frequency

20.11 kHz (1H = 1790CLK)(3) Vertical sync frequency

19.49 Hz(maximum frequency on all pixel readout mode)[Electrical shutter specification](1) Shutter Speed

2/1 to 1/20,000 [sec]Setting format: Numerator / Denominator [sec]Numerator: 1 to 255
Denominator: 1 to 20,000*When you set the exposure time longer than approximately 1 second, white spots and theunevenness in highlight portion might occasionally be observed on screen. This phenomenon isdue to the characteristics of the CCD image-pickup device, and do not reflect performance error in
the pickup device or CCD Camera itself.
(2) Random Trigger Shutter

ON / OFF switching (factory default: OFF)・Fixed mode

The exposure time depends on the shutter speed setting

・pulse width mode

The exposure time depends on the pulse width.

(3) Restart-Reset

ON / OFF switching (factory default: OFF)The exposure time depends on the period of Ext. VD.

[Input signal specification](1) TRIG/VD

Camera Link interface input: CC1

・Polarity

Positive/Negative switching (factory default: Negative)

・Pulse width

8s to 2s

Notes of trigger mode:*When the interval of the input trigger signal is extremely shot, or when the trigger signal is noisy,
there is a possibility of causing the malfunction. In this case, please input a proper trigger signal.

( 14 / 25 )D4159008D

[Camera PSU input impedance]
Ci < 57F, Rx＝10kΩ±10%
PoCL
Cable

PSU
Test Circuit

Ia

CSCS20BC2
PSU

Pin#: 1, 26

Vb

Rx

Cx

CameraPSU

Pin#: 13, 14
Ia = 52uA, Vb = 0.52V, Cx < 50uF

[Mechanical spec](1) Lens mountC-mount*Install a next lens; the C mount lens, its dimension of protrusion from flange is equal to or less than
7.9 mm. If a lens does not stand to this condition, it might not be installed to this camera.
*Depending on the lens you use, the performance of the camera may not be brought out fully due tothe deterioration in resolution and brightness in the peripheral area, occurrence of the ghost,aberration and others. When you check the combination between the lens and camera, be sure to use
the lens you actually use.
(2) Flange back17.526 mm(3) Dimensions

29mm(W)  29mm(H)  26.5mm(D)* Not including protrusion(4) Mass

Approx. 40g(5) Camera body grounding: insulation status Conductive between circuit GND and camera body

[Optical axis accuracy]
Position accuracy ofeffective pixels（Ｘ）（Ｙ）
Optical axis accuracy±25um

Rotation accuracy ofeffective pixels(θ)

Flange focal distances(Flange back)( for 17.526mm )±0.12°

±50um

±25um

( 15 / 25 )D4159008D

[Operating ambient conditions](1) Performance assurance

Temperature: 0C to +40C
Humidity: 10% to 90% (no condensation)(2) Operation guaranteed

Temperature: -5C to +45C
Humidity: 90% or less (no condensation)(3) Storage

Temperature: -20C to +60C
Humidity: 95% or less (no condensation)(4) EMC conditions (Electro-Magnetic Compatibility)
EMI (Electro-Magnetic Interference)
EN61000-6-4(Conformity)

EMS (Electro-Magnetic Susceptibility)
EN61000-6-2(Conformity)(5) FCCFCC Part 15 Subpart B class A(Conformity)*About the conformity of EMC standard of this machine, it has guaranteed in the conditions
combined with our system condition. When used combined parts other than specification of ourcompany, I ask you to have final EMC conformity checked of a visitor with a machine and the whole
equipment.

( 16 / 25 )D4159008D

[Communication specification](1) Communication speed

9600 bps (fixed)(2) Start bit

1(3) Data bit

8(4) Parity

None(5) Stop bit

1(6) Handshake

None[Connector pin assignment]
Video output/controlling/power supply connector: (Camera Link Base Configuration) CAMERA LINK
Connector model: HDR-EC26FDTG2+ (Manufactured by Honda Connectors)
Pin # I/O
1
2
O
3
O
4
O
5
O
6
O
7
I
8
O
9
I
10
I
11
I
12
I
13
-

Signal name
+12V
TxOUT0TxOUT1TxOUT2TxCLK OUTTxOUT3SerTC(RxD)+
SerTFG(TxD)CC1(TRIG/VD)CC2+
CC3CC4+
GND

Pin # I/O
14
15
O
16
O
17
O
18
O
19
O
20
I
21
O
22
I
23
I
24
I
25
I
26
-

Signal name
GND
TxOUT0+TxOUT1+
TxOUT2+TxCLK OUT+
TxOUT3+
SerTC(RxD)SerTFG(TxD)+
CC1(TRIG/VD)+
CC2CC3+
CC4+12V*Please confirm the power supply of the camera cuts when the connector is connected or pulls out.
It causes the breakdown etc.

[Typical ambient conditions]*The lens characteristics and light source characteristics are not reflected in table.
1.0
0.9

Relative Response

0.8
0.7
0.6
0.5
0.4
0.3
0.2
0.1
0.0
400

500

600

700
800
Wave Length[nm]( 17 / 25 )

900

1000

D4159008D

7. Timing Chart
7-1. Normal Scan (All pixel readout)
As all pixels are read out in approx. 1/19.5s, you will get images with the higher V resolution(you have to change the shutter speed to 1/19.5s or shorter).

Vertical Timing (maximum speed on all pixel readout mode)A = approx. 271.1us
B = approx. 50.9ms (1024H-336CLK)
1frame = A + B + C = approx. 51.3ms*1H = 1790CLK

C = approx. 136.0us

FVAL
LVAL
DVAL
DATA OUT
A

B
C

Horizontal Timing
Same timing
FVAL
a
LVAL
DVAL
DATA OUT
b

d
e

c

f

a = 336CLK

b = 22CLK

c = 16CLK

d = 1360CLK( 18 / 25 )e = 16CLK

f = 40CLK

D4159008D

7-2. Partial Scan
Ranges except the range of the image output that the user set are not read.

ex.1) Video Start Line =128, Video Width = 768(maximum speed on all pixel readout mode)
FVAL
LVAL
DVAL
DATA OUT
A

B

C

D

A = approx. 996.4us

B = approx. 38.2ms ( 768H-336CLK )C = approx. 861.3us

Video start line:128

D = approx. 40.0ms

Not readVideo width:768 lines
Video output

The rest of line:128

Valid lines:1024

Not readVideo output:768 lines( 19 / 25 )D4159008D

ex. 2) Video Start Line =256, Video Width=512(maximum speed on all pixel readout mode)
FVAL
LVAL
DVAL
DATA OUT
A

B

C

D

A = approx. 1721.8us

B = approx. 25.5ms (512H-336CLK) C = approx. 1586.7us

D = approx. 28.8ms

Video start line:256
Not readVideo width:512 lines
Video output

Valid lines:1024

The rest of line:256
Not readVideo output:512 lines( 20 / 25 )D4159008D

7-3. Random Trigger Shutter
In the random trigger shutter mode, you can shoot and grab an image at an arbitrary timing by trigger
signal input from the external (It is not possible to use the Random Trigger Shutter concurrently with theRestart Reset mode).

· External trigger signals can be input either from the camera link I/F CC1.
· If polarity is set to negative polarity, exposure starts at the falling edge of the trigger.
· The random trigger shutter of this camera can be operated in two types of mode: fixed mode and pulse
width mode. How to determine the exposure time differs depending on the mode.

 Fix mode
The exposure time is determined by the setting value for the shutter speed.
exposure time

*exposure time = pulse width + approx. 1.8usec

TRIG
approx. 3.5usec

approx. 5.3usec

CCD
approx. 260.6usec (all pixel readout)
FVAL
LVAL
DVAL
DATA OUT· Pulse Width mode (Trigger Polution:Low Active)
The exposure time is determined by the pulse width
Set a pulse width of 1H (approximately 49.7μs) or more.
exposure time
TRIG

CCD
exposure*exposure time = set time of E-shutter speed

approx.
3.5usec

approx. 260.6usec (all pixel readout)

FVAL
LVAL
DVAL
DATA OUT( 21 / 25 )D4159008D

7-4. Restart / Reset
The restart / reset function is available with the ext.VD signal. You can get an arbitrary slower shutter
speed than normal shutter and random trigger shutter (It is not possible to use the Restart Reset mode
concurrently with the Random Trigger Shutter mode).

· External VD signals can be input either from the camera link I/F CC1.
· The shutter speed (exposure time) is determined by ext. VD signal interval.

1) Vertical Timing
VD

CCD
exposure

DATA OUT

FVAL2) Horizontal Timing

Ext. VD input
more than 1H

CCD
exposure

DATA OUT

FVAL
when the external VD input s to the camera while v ideo output( 22 / 25 )D4159008D

8. Notes regarding specifications


Always read the corresponding operating instruction manuals and precautions for any devices
connected to this product before use.

With regard to the data saved using a grabber board connected to the camera, neither TELI, the frame
grabber board manufacturer, nor the peripheral device manufacturer will accept any liability for theloss of data or damage resulting from use of the camera, the frame grabber board, or other peripheraldevices. Always back up important data if there is any possibility of loss of data.

Data may be subject to copyright or other usage restrictions. Users are responsible for performing due
diligence regarding the copying or distribution of such data. TELI rejects all liability for problems
arising from improper use of data or printed matter.



Use in conjunction with devices for which compatibility has not been confirmed by TELI may resultin failure, damage, or malfunctions. Fees will be charged for all repairs made necessary by such
failures. For more information on what devices can be connected, please check with your retailer,sales agent, or TELI customer service department.



Do not expose the CCD to bright light for extended periods.

Pixel drop-outs may occur suddenly or randomly on the CCD during transport or storage. This does
not constitute a defect.



Noise may appear on the screen if the camera or cables are located close to devices such as lighting
cables or motors.

Keep the camera and wiring away from noise sources.



Do not use in locations at ambient temperatures or in humid environments exceeding specifications.
Doing so may affect image quality or internal components. Avoid use in locations subject to direct
sunlight.

Do not store in locations subject to direct sunlight. Doing so may degrade components.



Use in systems in which the camera connector is subject to strong impact may damage the camera
connector. Before using the product under such conditions, bundle the camera cable as close as
possible to the camera to minimize the possibility of such impact.

( 23 / 25 )D4159008D

9. Guarantee
The term of a guarantee is one year after the product delivery.
If by any chance trouble by responsibility of our company occurs before an above period, TELI repairs
it free of charge.
- During terms of a guarantee, when the trouble cause is the case of below, TELI charges the repair
costs.
(1) Troubles and the damages that causes by misuse, unsuitable repair or remodeling.
(2) Distribution hazards like drops and vibrations after purchase. Troubles and damages bytransportation.
(3) Troubles and damages by fire, natural calamity (earthquake, storm and flood damage, thunderbolt),
damages from salty breeze, gas harm, abnormal voltage.

10. Repair
10-1. Condition for repair
Basically, has to return it to our company when the user requests us to repair product.
Beside that, customer should pay these expenses (travel expenses, camera disassembly technology costs)of both customer and end user. Also customer should pay in themselves costs for return camera to us.
10-2. The period of repairing product(1) Repair free of charge ... Refer to Clause 8.
(2) Charged repair ............. Basically, repair period is 7 years after the last production end of products.

( 24 / 25 )D4159008D

11. External-view Drawing[ UNIT : mm ]( 25 / 25 )D4159008D