PRODUCTS
PRODUCTS

1.8
1.8

LabM-I-6-01
LabM-I-6-01

1.8.1 	2.5 – 7.0 µm and over 3 MHz HgCdTe programmable, laboratory IR detection module with
1.8.1 2.5– 7.0 µm
and over3 MHz HgCdTe
programmable, laboratory IR detection module
optically
immersed
photovoltaicdetector

withoptically immersed photovoltaic detector
LabM-I-6-01 is a laboratory IR detection module with optically immersed photovoltaic detector based on HgCdTe heterostructure, inte-grated with transimpedance, programmable preamplifier. 3° wedged zinc selenide anti-reflection coated window prevents unwanted inLabM-I-6-01 is a laboratory IR detection module with optically immersed photovoltaic detector based on HgCdTe heterostructure,terference effects. For proper operation programmable „smart” VIGO thermoelectric cooler controller PTCC-01 (sold separately) and Smart
integrated with transimpedance, programmable preamplifier. 3° wedged zinc selenide
anti-reflection coated window prevents
Manager Software (freeware) are required. LabM-I-6-01 module comes complete with PTCC-01 and Smart Manager is the best solution for
unwanted interference effects. For proper operation programmable „smart” VIGO thermoelctric cooler controller PTCC-01 (soldprototyping and R&D stage in a variety of MWIR applications. This set provides flexible approach to different needs of system designers.
separately) and Smart Manager Software (freeware) are required. LabM-I-6-01 module comes complete with PTCC-01 and SmartManagerthe best(Tsolutionapprto= 20°C)for prototyping and R&D stage in a variety of MWIR applications. This set provides flexibleoach
Spectralisresponse
adifferent needs of system designers.

Spectral response (Ta = 20°C)

D*, cm·Hz1/2/W

1E+10

1E+09

1E+08
2,53,03,54,0 4,5 5,0 5,5Wavelength, µm

6,06,57,0

Exemplary spectraldetectivity, the spectral response of delivered devices may differ.

Exemplary spectral detectivity, the spectral response of delivered devices may differ.

www.vigophotonics.com
www.vigophotonics.com

Page | 25 z 160
Page | 24 z 156

PRODUCTS

Specification (Ta = 20°C, default module settings)
Parameter

Typical value
Optical parameters

Cut-on wavelength λcut-on (10%), µm

3.0±1.0

Peak wavelength λpeak, µm

5.2±0.5Optimum wavelength λopt, µm

6.0

Cut-off wavelength λcut-off (10%), µm

6.7±0.3

Detectivity D*(λpeak, 25 kV/A), cm·Hz1/2/W≥1.0×1010

Detectivity D*(λopt, 25 kV/A), cm·Hz1/2/W≥7.0×109

Output noise density vn (10 MHz), nV/Hz1/2≤500Electrical parameters
Voltage responsivity Rv (λpeak, 25 kV/A), V/W≥5.0×104

Voltage responsivity Rv (λopt, 25 kV/A), V/W≥3.5×104

Low cut-off frequency flo, Hz

DC/10 (adjustable)

High cut-off frequency fhi, Hz≥3M (adjustable)

Output impedance Rout, Ω50

Output voltage swing Vout, V

1 (RLoad = 50 Ω)

Output voltage offset Voff, mVmax ±20FeaturesH igh performance and reliabilityD C offset compensationC ompatible with optical accessories
V ersatility and flexibilityQ uantity discounted priceF ast delivery
Parameters configurable by the userO
 utput voltage offsetG ain (in 40 dB range)B andwidth (1.5 MHz / 3 MHz)C oupling AC/DCD
 etector’s parameters (temperature, reverse bias etc.
Applications
M WIR gas detection, monitoring and analysis
F lue gas denitrificationF uel combustion monitoring at power plants and otherindustrial facilities
B reath analysis
E xplosion preventionE mission control (exhaust fumes, greenhouse gases)C ontactless temperature measurements

Other information
Active element material

epitaxial HgCdTe heterostructure

Optical area AO, mm×mm

1×1

Window
Acceptance angle ΦAmbient operating temperature Ta, °C
Signal output socket
Power supply and TEC control socket

wZnSeAR~36°
10 to 30SMA
LEMO (female) ECG.0B.309.CLN

Mounting hole

M4

Fanyes

www.vigophotonics.com

Page | 26 z 160

PRODUCTS

Mechanical layout, mm

Power supply and TEC control socket LEMO (female)Powersupply and TEC control socket LEMO (female)
ECG.0B.309.CLN
ECG.0B.309.CLN

ut, mmnd TEC control
socket LEMO(female)
Schematicdiagram
Schematic diagram
N

Schematic diagram

Function
Function
Fan and programmable preampinternal logic auxiliary supply
Fan and programmable preamp
Thermistor
output(2)internal logic
auxiliary
supplySymbol
SymbolFAN+

Pin number
Pin number
1FAN+
TH2

21

TEC supply input (–)
Thermistor output (2)

TEC–
TH2

32

Power supply input (–)–Vsup

4TEC supply input (–)
GroundTEC–GND

35Powersupplyinput(+)(–)
Powersupplyinput+V
–Vsup
sup64

TEC supply input (+)
GroundThermistor output (1)

TEC+
GND
TH1

758

Power supply input (+)
Biderictional data pin

+Vsup
DATA

69TEC supply input (+)

TEC+

7Thermistor output (1)TH18Biderictional data pin

DATA

9Included accessories•SMA-BNC, LEMO-DB9 cables

Dedicated accessories

ionSymbol

Pin numberble preamp
yersupply

FAN+1)

TH22TEC–3–)+)

–Vsup

4

GND5+Vsup

6TEC+7)

TH18nDATA

9•Included
PTCC-01-BAS
TEC controller + USB: TypeA-MicroB cable + ACaccessories
adaptor

MA-BNC, LEMO-DB9cables + USB: TypeA-MicroB cable + AC
• S PTCC-01-ADV
TEC controller

adaptor
Dedicated accessories
PTCC-01-OEM TEC controller + USB: TypeA-MicroB, KK2POWER
cablesP
 TCC-01-BAS
TEC controller + USB: TypeA-MicroB cable +•• ACOTA
optical threaded adapter
adaptor• P DRB-2
base mountingsystem
TCC-01-ADV
TEC controller+ USB: TypeA-MicroB cable +AC adaptor
P TCC-01-OEM TEC controller + USB: TypeA-MicroB,
KK2- POWER cables
www.vigophotonics.com
O TA optical threaded adapter
D RB-2 base mounting systemsories

EMO-DB9 cables

ssories

S TEC controller + USB: TypeA-MicroB cable + ACV TEC controller + USB: TypeA-MicroB cable + AC

eM+TECAC controller + USB: TypeA-MicroB, KK2-esthreaded
e+ AC adapter

mounting system

2-com

Page | 26 z 156

www.vigophotonics.com

Page | 27 z 160
Page | 26 z 156