QUICK START GUIDE
itnetworks.softing.com

The NetXpert XG network cabling qualifier offers a complete solution to test copper andfiber network cabling. In a quick and easy way, the device tests and documents, whethera cabling link is able to support data rates of up to 10 Gigabit Ethernet in compliance withIEEE 802.3 by testing the data carrying capabilities of Ethernet network cabling up to10Gbit/s, determining the signal-to-noise ratio, and detecting faults in the cable wiring.
This ensures that the cabling is able to support the speed capabilities of active equipmentconnected to the network.
In this quick start guide you find the most important information for carrying out cable
tests with the NetXpert XG. For detailed descriptions please use the user manual on ourwebsite: itnetworks.softing.com/XG

SAFETY INSTRUCTIONSTo ensure safe operations of the NetXpert XG, follow the instructions carefully and observe
caution and warning messages in this manual. Failure to observe caution and warning
messages can damage the equipment and can result in severe injury or death.
Icon

Definition!
Voltage

Warning/caution: potential for personal injury or damage to equipment
Warning: hazards could cause harm or death
Conformité Européenne. Conforms to EU directives
Disposal information!

Warning! To avoid possible fire, electric shock, or personal injury:• Do not open the case; no user serviceable parts are inside.
• Do not modify the tester.
• Do not use the tester if it is damaged. Perform a visual inspection of the tester beforeuse�• The tester is not intended to be connected to active telephone inputs, systems, orequipment, including ISDN devices. Prolonged exposure to the voltages applied by theseinterfaces may damage the tester.
• The Lithium-ion battery pack is to be serviced by qualified personnel only. Only theoriginal battery pack type should be used with this product.

itnetworks.softing.com/XG

15

BATTERYThe NetXpert XG is powered by a rechargeable Lithium-ion battery.
• To charge the battery, plug the charger into the charging jack of the tester and thenplug the charger into the power source. The battery of the NetXpert XG is not user
serviceable.
• In the event that you need to remove the battery, remove the screws on the back panelof the NetXpert XG and unplug the battery. Be sure to remove all the leads connecting
the battery to the device. Please make sure that the battery is charged overnight or atleast for 5 hours. This will increase the accuracy of the battery status indication in thelong run.
Note: The high-quality Lithium-ion battery is designed to last for the life of the testerand should not need replacement. Lithium-ion batteries are hazardous waste and should
be disposed of in compliance with the applicable waste regulations.

STORAGE
When not in use, store the NetXpert XG in the included protective case.
Do not expose the NetXpert XG to high temperatures (above 80°C) or humidity.

SCOPE OF DELIVERY• 1x NetXpert XG main unit
• 1x NetXpert XG Active Remote unit
• 2x Power supply• 1x Micro USB adapter• 1x Hard carrying case• 2x CAT 6A Patch cords

16

DEVICE DESCRIPTIONACTIVE REMOTE
UNIT

MAIN UNIT
Display

Power on/off

Ventilation opening

Micro USB port

Charger

RJ 45 port

Micro USB port

Charger

RJ 45 port

external SMA Wi-Fi
antenna port

SFP and SFP+ port
Status LEDs

itnetworks.softing.com/XG17GENERAL OPERATIONS
For detailed descriptions please consult the user manual on our website.
Except for the power on/off function, the main unit is exclusively operated via thetouch-sensitive screen.

ICONS
Display/hide the menu at the left-hand side

Back to the home screen

This icon indicates that the battery of the NetXpert XG is being charged overan external power supply

Displayed results will be exported in a CSV file

Displayed results will be exported in a PDF file18

BUTTON FUNCTIONSThe NetXpert XG is turned on by pressing the power button. The button indicates theoperating status by light signals:

OFFGREEN LIGHT(CONTINUOUS)

GREEN LIGHT(FLASHING)

ORANGE LIGHT(FLASHING)

The NetXpert XG
is turned off

The NetXpert XG
runs on battery
and is sufficientlycharged

The NetXpert XG
is being charged

The NetXpert XG
is turned on andbattery status islower than 20%• A short press of the power button will put the NetXpert XG into sleep mode.
• The NetXpert XG automatically enters sleep mode to conserve battery power if no input
or activity is performed on the device.
• A long press of the power button will turn off the NetXpert XG.
• Touch the Status icon to check the battery status of the Active Remote and to display
the latest test result.

• Touch the Test button to start a qualification test from the Active Remote.

itnetworks.softing.com/XG

19

PASSIVE NETWORK TESTS –TEST RESULTS IN 10 STEPS

1. Set language
Touch the Settings icon to configure the NetXpert XG.
Touch Language Icon and select the desired language for the user interface.
When changing the language, the keypad will be changed accordingly.
2. Set Date and TimeSet the current time and date. These have to be entered in the defined format.
You can choose the time format displayed – a 12-hour or 24-hour format.

3. Set Standards
Choose between Meters and Feet for the display of the length.
The NetXpert XG supports various wiring standards. Choose between TIA568A/B,PROFINET, and USOC. This affects primarily the display of the wiremap.

4.Personalize your Reports

Must select More Options to access Reports screen. If you want to personalize yourreports, you can enter the name of your company, of the technician doing the job,and a company logo.
To include a company logo, touch Import logo. Connect a USB flash drive containing
your company logo to the NetXpert XG. A file browser will open, and you cannavigate to your logo file on the USB flash drive and select the logo.

20

5. Cable Test Settings
From the home screen, touch Cable Test to display the cable test menu. Further project specific settings can be made here. Moreover, a new test can be started from here.
On the left-handside, the various
menus are listed
and, if alreadycreated, the list ofthe tested cables.

On the righthand side, thetest result of thecable selectedon the left isdisplayed.

A qualification test can directly be started by touching the Test button.
6. Project Options
Touch the Project + Reports icon to make general project settings for yourtest project.

7. Cable Type
Touch the Cable Type icon on the left to select or configure different cable types.

Important: Before running a test, it is necessary to select the correct type of cable,
otherwise the result could be affected.

8. Cable LabelingDefine the labeling of the individual measurements, to be able to retrieve andallocate the measurements more easily later on.

itnetworks.softing.com/XG

21

9. Set Reference
To exclude test cables that are used to connect the NetXpert XG and the Active
Remote, from the length measurement, you can have their length determined
and automatically deducted from the total length of the cabling tested by theNetXpert XG�
10. Cable Test
When touching the Cable Test icon, you will always return to the test screen. The
test screen displays an overview over the cabling links to be tested/already tested.
Moreover, you can view the results of the tests that are already completed andstart a new test�
Touching the Test button will launch a complete cable test on the cable selectedin Cable Type. The next free cable will be selected from the list or a new one isadded below.

All test results can be managed in the file manager.
For detailed descriptions please consult the user manual on our website.

Add new cable
Delete cable and test
Edit cable – Labels and attributes such as building, floor, room, etc.
can be adjusted
Start test22

ACTIVE NETWORKS TESTSThe NetXpert XG offers different possibilities for analyzing troubleshootingin active networks.

An overview over the connections to the network
DHCP Test
Network Discovery

Ping Lists
Traceroute
LinkLayer Discovery Protocol
LLDP

Cisco Discovery Protocol
CDP

VLAN

For detailed descriptions please consult the user manual on our website.

itnetworks.softing.com/XG

23

ADDITIONAL TEST OPTIONSTOOLSNetXpert XG offers various individual cable tests for targeted troubleshooting
or cable localisation.

Wiremap and signal-to-noise ratio
Port Discovery
PoE Test

Only Bit Error Rate Test (BERT)
Tone Generator
Link LightFor detailed descriptions please consult the user manual on our website.

Softing SARL
87, Rue du Général Leclerc
Creteil, Île-de-France
94000 (Paris)
Phone: +33 1 45 17 28 05
E-mail: info.france@softing.com
itnetworks.softing.com

24