Rev 1.1

High-Resolution CCD B/W Camera
CleverDragon series
CSCQS15BC23
CSCQS15BC23-01
Specification
Contents
RESTRICTION FOR USE ·····································································1
CASES FOR INDEMNITY (LIMITED WARRANTY) ···························2
NOTES ON USING THIS PRODUCT ····················································3
1. Overview ····························································································5
2. Features ······························································································5
3. Configuration ······················································································6
4. Optional parts ·····················································································65. Functions ····························································································7
6. Serial communication ········································································· 12
7. Specifications ····················································································· 13
8. Timing chart ······················································································· 20
9. Outline drawing ················································································· 22
10. Guarantee ························································································· 23
11. Repair ······························································································ 23

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A

Rev 1.1

RESTRICTION FOR USE
 Should the equipment be used in the following conditions or environments, give consideration to safetymeasures and inform us of such usage:

1. Use of the equipment in the conditions or environment contrary to those specified, or use outdoors.
2. Use of the equipment in applications expected to cause potential hazard to people or property, which
require special safety measures to be adopted.

 This product can be used under diverse operating conditions. Determination of applicability ofequipment or devices concerned shall be determined after analysis or testing as necessary by thedesigner of such equipment or devices, or personnel related to the specifications. Such designer or
personnel shall assure the performance and safety of the equipment or devices.

 This product is not designed or manufactured to be used for control of equipment directly concernedwith human life (*1) or equipment relating to maintenance of public services/functions involving
factors of safety (*2). Therefore, the product shall not be used for such applications.
(*1): Equipment directly concerned with human life refers to.
· Medical equipment such as life-support systems, equipment for operating theaters.
· Exhaust control equipment for exhaust gases such as toxic fumes or smoke.
· Equipment mandatory to be installed by various laws and regulations such as the Fire Act or
Building Standard Law· Equipment related to the above(*2): Equipment relating to maintenance of public services/functions involving factors of safety refers to.
· Traffic control systems for air transportation, railways, roads, or marine transportation· Equipment for nuclear power generation· Equipment related to the above

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-1-

Rev 1.1

CASES FOR INDEMNITY (LIMITED WARRANTY)
We shall be exempted from taking responsibility and held harmless for damage or losses incurred by the user
in the following cases.

 In the case damage or losses are caused by fire, earthquake, or other acts of God, acts by a third party,
deliberate or accidental misuse by the user, or use under extreme operating conditions.

 In the case of indirect, additional, consequential damages (loss of business interests, suspension ofbusiness activities) are incurred as result of malfunction or non-function of the equipment, we shall be
exempted from responsibility for such damages.

 In the case damage or losses are caused by failure to observe the information contained in the
instructions in this instruction manual and specifications.

 In the case damage or losses are caused by use contrary to the instructions in this instruction manual andspecifications.

 In the case damage or losses are caused by malfunction or other problems resulting from use of
equipment or software that is not specified.

 In the case damage or losses are caused by repair or modification conducted by the customer or any
unauthorized third party (such as an unauthorized service representative).

 Expenses we bear on this product shall be limited to the individual price of the product.
 About the item which does not have a publication in the specifications and manual of this product, it considers
as the outside for a guarantee.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-2-

Rev 1.1

NOTES ON USING THIS PRODUCT Handle carefully
Do not drop the equipment or allow it to be subject to strong impact or vibration, as such action may cause
malfunctions. Further, do not damage the connection cable, since this may cause wire breakage.

 Environmental operating conditions
Do not use the product in locations where the ambient temperature or humidity exceeds the specifications.
Otherwise, image quality may be degraded or internal components may be adversely affected. In
particular, do not use the product in areas exposed to direct sunlight. Moreover, during shooting underhigh temperatures, vertical stripes or white spots (noise) may be produced, depending on the subject orcamera conditions (such as increased gain). However, such phenomena are not malfunctions.

 About lens mountAs for the Special-mount lens used combining this camera, the projection distance from bottom of the
Bottom ofscrew should use 9.5mm or less.
Special-mount lens

the screw

9.5mm or less

 Check a combination with the lens
Depending on the lens and lighting you use, an image is reflected as a ghost in the imaging area.
However, this is not because of a fault the camera.
In addition, depending on the lens you use, the performance of the camera may not be brought out fully
due to deterioration in resolution and brightness in the peripheral area, aberration and others.
Be sure to check a combination with the camera by using the lens and lighting you actually use.
When installing a lens in the camera, make sure carefully that it is not tilted.
In addition, use a mounting screw free from defects and dirt. Otherwise, the camera may be unable to be
removed.
 About the difference between the right and on the left of an image output screen
As for CCD used with this camera, the left half and the right half of the screen are independently output(2ch output method). Therefore, It is not failure, although a level difference may be visible to an image
on either side or a boundary line may be visible to middle of the screen bordering on middle of thescreen with the setting mode of a camera.

 Defect of CCD
In transportation and keeping, it is not a breakdown though the defect might occur by accident in CCD.

 Do not shoot under intense light.
Avoid intense light such as spot lights on part of the screen because it may cause blooming or smears. If
intense light falls on the screen, vertical stripes may appear on the screen, but this is not a malfunction.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-3-

Rev 1.1

 Occurrence of moiré
If you shoot thin stripe patterns, moiré patterns (interference fringes) may appear. This is not a
malfunction.

 Occurrence of noise on the screen
If an intense magnetic or electromagnetic field is generated near the camera or connection cable, noise
may be generated on the screen. If this occurs, move the camera or the cable.

 Handling of the protective cap
If the camera is not in use, attach the lens cap to the camera to protect the image pickup surface.

 Turn OFF the power in the case of connection
Turn OFF the power in the case of connection of connection camera cable.
Otherwise, an electric shock or a malfunction may occur.

 Disposal
When disposing of the camera, it may be necessary to disassemble it into separate parts, in accordancewith the laws and regulations of your country and/or municipality concerning environmental
contamination.
The use of the symbol indicates that this product may not be treated as household waste. By ensuring thisproduct is disposed of correctly, you will help prevent potential negative consequences for theenvironment and human health, which could otherwise be caused by inappropriate waste handling of thisproduct. For more detailed information about the take-back and recycling of this product, please contact
your supplier where you purchased the product.

”This symbol is applicable for EU member states only”

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A-4-

Rev 1.1

1. Overview
This CCD camera is a high-resolution camera that features all pixel readout mode 2/3 CCD.
Model
CSCQS15BC23
CSCQS15BC23-01

Power supply
I/O connector
Camera Link Connector

Interface
Camera Link
Power over Camera Link

2. Features(1) High resolution
High pixel density CCD (number of effective pixels 5.05M, number of total pixels 5.24M) is used.
(2) Square grids
The CCD pixels arrayed in square grids facilitates computation for image processing.
(3) Full-frame shutter
Since all pixels are output even by shutter operation, high resolution can be achieved,
without deteriorating the vertical resolution.
(4) Camera link interface
The interface of image output and the camera control adopts the Camera Link standard.
When you use frame grabber board for the camera link, high-speed image transfer and various
controls to PC are possible.
Non-PoCL model: CSCQS15BC23
PoCL model:

CSCQS15BC23-01(5) All-pixel readout mode (normal mode)
All pixel signals (in the effective area) are output in approximately 1/15 second.
(6) Programming partial scan mode
Partial scan within the range arbitrary from 100 lines to 2456 lines is possible.
(7) High-speed draft readout mode
By reading 4 lines from every 16 lines, all signals in the effective area are output in approximately in
1/37.1 second.
(8) Random trigger shutter
By external trigger signal input, the shot image can be grabbed at an arbitrary timing.
(9) Multiple-shutter
By external trigger signal input, the shot image can be grabbed at an arbitrary timing and theaccumulated shot images can be output at an arbitrary timing.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-5-

Rev 1.1

3. Configuration(1) Camera body·········································································· 1(2) Accessories
Instruction Manual (Japanese) ·················································· 1
Instruction Manual (English) ···················································· 1

4. Optional parts(1) I/O cable

Model name: CPRC3700-**(2) Camera Link cable

Model name: 14B26-SZLB-**-0LC(3) Camera adapter

Model name: CA170(4) Camera mounting kit

Model name: CPT4000CL*NOTE: Application software is not supplied as a standard item.
*NOTE: If you need PoCL cable, contact your dealer / distributor.

Notes on optional parts and compliance with safety standard conditions:
We assure the compliance of this camera with the safety standard when it is used in combination withthe optional parts listed above.
If you use the camera in combination with parts other than specified by our company, you are responsible
for finally confirming the compliance with the safety standard by using the entire machine/equipment.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-6-

Rev 1.1

5. Functions
By accessing the camera register published on the camera link I/F, you can control/set each function.
Since access to the camera register is performed via the frame grabber board, the controlling and setting
methods differ depending on the frame grabber board you use. For details, refer to the instruction
manual of the relevant frame grabber board or contact our sales representative.

5-1. Readout mode
Video is output from the camera link connector. The output video can be grabber by the frame
grabber board. The frame rate and resolution of output images that this model supports are as
follows:
1) All pixel readout: Approximately 15 fps / 2456(H) x 2058(V)2) High-speed draft readout: Approximately 37.1 fps / 2456(H) x 257(V)3) Partial scan

: Approximately 52 to 15 fps / 2456(H) x 100(V) to 2058(V)* Number of minimum line is 100 lines.

1) All pixel readout
Reads out all pixels in about 1/15 second.
2064H
FVAL
LVAL

1
2
3
4
5
6
7
8

2055
2056
2057
2058

DVAL

DATA OUT

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-7-

Rev 1.1
2) High-speed draft readout
By reading 4 lines out of every 16 lines, reads out the whole valid area in approximately 1/37.1
seconds. Control and setting of functions can be done by accessing the cam. When the frame rateof a partial scanning exceeds it when the shutter mode is switched from normality (internal
synchronization) or the partial scanning mode to the draft mode, more high-speed than the settingof the speed of the shutter it is changed at the speed of the shutter matched to the frame rate.
259.5H
FVAL
LVAL

1+5
8+12
17+21
24+28
33+37
40+44

2041+2044
2049+2053

DVAL

DATA OUT

Reading 4 lines out of every 16 lines in
the inside of a total of 2058 lines

Effective lines : 257 lines

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-8-

Rev 1.1
3) Programmable partial scan
A range arbitrary from 100 lines to 2058 lines can be read. The frame rate can be raised to 52 fps
or less by skipping it at high speed excluding an effective area. When the frame rate of a partial
scanning exceeds it when the shutter mode is switched from normality (internal synchronization)
to the partial scanning mode, more high-speed than the setting of the speed of the shutter it ischanged at the speed of the shutter matched to the frame rate.
- Starting position (Partial V Start1, 2) of an effective line can be set.
- Number (Partial Height1, 2) of effective line can be set.

Example: (Partial V Start1, Partial Height1) = 100 effective lines and start 980 lines eyes(center partial)(Partial V Start2, Partial Height2) = 0 effective lines (60fps) and start 0 lines eyes(center partial)
595H
FVAL
LVAL

980
981
982
983
984
985
986

1077
1078
1079

DVAL

DATA OUT

Skipped lines ： 979 line

High-speed transfer ： 245 line＋
Nomal-speed transfer ： 2 line

Effective line : 100 line

Effective line : 100 line
High-speed transfer ： 245 line

Skipped lines ： 979 line

Example: (Partial V Start1, Partial Height1) = 100 effective lines and start 1 lines eyes(center partial)(Partial V Start2, Partial Height2) =100 effective lines (60fps) and start 1959 lines eyes(center partial)
672H
FVAL
LVAL

2056
2057
2058

1959
1960
1961

96
97
98
99
100

1
2
3
4
5

DVAL

DATA OUT
Skipped lines ： 100 line
Effective line ： 100 line
Effective line ： 1858 line

High-speed transfer ： 463 line＋
Nomal-speed transfer ： 2 line
Effective line ： 100 line

Skipped lines ： 100 line

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A
-9-

Rev 1.1
5-2. Random trigger shutter
In the random trigger shutter mode, you can shoot and grab an image at an arbitrary timing bytrigger signal input from the external.
- External trigger signals can be input either from the camera link I/F CC1 or I/O connector.
- If polarity is set to negative polarity, exposure starts at the falling edge of the trigger.
- The random trigger shutter of this camera can be operated in two types of mode: fixed mode andpulse width mode. How to determine the exposure time differs depending on the mode.

1) Fixed mode
- The exposure time is determined by the setting value for the shutter speed.
exposure time

※ H ･･･ It differs depending on the image output mode.

TRIG
approx. 1.58μs

CCD※ 5.2H

FVAL
LVAL
DVAL
DATA OUT

2) Pulse width mode
- The exposure time is determined by the pulse width.
(exposure time = pulse width + approximately 14μs)- Set a pulse width of 1H (approximately 30.1μs) or more.
exposure time

※ H ･･･ It differs depending on the image output mode.

TRIG
approx. 1.58μs

approx. 15.2μs

CCD※ 5.2H

FVAL
LVAL
DVAL
DATA OUT

Notes of trigger mode:
When the interval of the input trigger signal is extremely short, or when the trigger signal is noisy, there
is a possibility of causing the malfunction. In this case, please input a proper trigger signal.
This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 10 -

Rev 1.1
5-3. Multiple-shutter mode
In the multiple-shutter mode, video is output in sync with a MULTI signal from the external after
the end of exposure time.
- Valid only when the random trigger shutter mode is ON.
- MULTI signals can be input from the camera link I/F CC2.
- If exposure is executed several times before MULTI signal input, the images are input, the images
are output superposed.
- The exposure time is determined by the random trigger shutter mode setting and its dtermination
method.
- The pulse width must be set to negative polarity and 30.1μs to 10 ms.

※ H ･･･

It differs depending on the image output mode.

TRIG
MULTI
A

B

CCD

※4.1H

FVAL
LVAL
DVAL
DATA OUT
A+B

Notes on multiple-shutter:
An image has been stored to CCD device until a MULTI signal is inputted and it begins to read out
image after CCD is exposed. Therefore, an image may degenerate step by step when an image is stored
to CCD device for a long time.
And, Electric charge is superimposed in CCD device when the multiplex exposure. Therefore, CCD willflood with electric charge when electric charge is over superimposed in CCD device. Consequently
vertical stripes noise will appear. At that time please stop using spotlight and dimmer for example close
the iris.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 11 -

Rev 1.1

6. Serial communication
By Camera link serial communication interface, it is possible to control the following functions.

(1) Memory control- Store- Readout- Reset(2) Setup

0 - 528 LSB(3) Gain

0 - +12 dB(4) Shutter speed

2 - 1/20,000 s(5) Random shutter trigger

ON / OFF- Trigger polarity

Positive / Negative- Random trigger mode

Pulse width mode / FIX mode

(6) Multiple shutter

ON / OFF(7) High-speed draft readout

ON / OFF(8) Programmable partial scan

ON / OFF* High-speed draft readout mode must be disabled.

- Start position of effective line 0 – 1957 line (1line / 1step)- Number of effective line

100 – 2058 line (1line / 1step)(9) TAP correction of
the left half and the right half

AUTO / ONE PUSH / MANUALThis specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 12 -

Rev 1.1

7. Specifications[Electrical specification](1) Image sensor

Interline CCD·

Number of total pixels

2536 (H) x 2068 (V)·Number of effective pixels

2456 (H) x 2058 (V)·Pixel size

3.45μm (H) x 3.45μm (V)Optical size

2/3 type·(2) Scan method

Progressive(3) Aspect ratio

6:5(4) Synchronization method

Internal synchronization(5) Standard subject illumination

400 lx, F5.6, 3000 K(6) Minimum subject illumination

6.25 lx (F1.4, Gain max, all pixel readout, video level 50%)

(7) Video output

Compliant with the camera link standard. Base configuration 2tap.

Data

12 / 10 / 8 bit switching

Readout mode
All pixel readout

Approximately 15 fps / 2456(H) x 2058(V)High-speed draft mode

Approximately 37.2 fps / 2456(H) x 257(V)Partial scan

Approximately 15 fps to 52fps / 2456(H) x 100 to 2058(V)* About an image output:
As for CCD used with this camera, the left half and the right half of the screen are
independently output (2ch output method). Therefore, it is not failure, although a level
difference may be visible to an image on either side or a boundary line may be visible to middleof the screen boarding on middle of the screen with the setting mode of a camera.
* About a right and left level difference between TAP of the image output
In the test environment of the Teli standard, we measure the level difference by using
the gray scale chart of ITE-Ⅱ type. We measure a center white part of the chart.
- Shipment standard (difference of the right part and left part ): 105 [digit] or less (12bit)
In the following cases, the camera might not be able to correct the level difference
automatically.
- When the image level of a right and left center part is low- When the image level of a right and left center part is high- When the level difference of each pixel (right and left part of center) is large
When you cannot correct the level difference, please correct the level difference by thefollowing method.
- After the source of light and object are changed, one push correction is done.
- The level difference is corrected with manual.
In the shipment setting, the calibration mode is AUTO.
This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 13 -

Rev 1.1(8) Gain0 to +12 dB (1211 levels)(initial factory setting: 0 dB)(9) Setup

0 to 528 LSB (529 levels)(initial factory setting: 264 LSB (calculated value))(10) Gamma correction

OFF(γ=1.0) fixed(11) TAP correction of the left part and the right part: AUTO / ONE PUSH / MANUAL(12) Power supply voltage

DC+12V (10.8 to 13.2 V)(ripple 50 mV(p-p) or less)(13) Power consumption

Approximately 4.0 W[Electronic shutter signal specification](1) Shutter speed
Readout mode
All pixel readout

2 to 1/20,000 s

High-speed draft readout 2 to 1/20,000 s
Programmable partial scan 2 to 1/20,000 s(2) Random shutter trigger

ON/OFF switching (initial factory setting: OFF)

Fixed mode

The exposure time depends on the shutter speed setting.

Pulse width mode

The exposure time depends on the pulse width.

(3) Multiple-shutter

ON/OFF switching (initial factory setting: OFF)
Exposure by TRIG input, readout by MULTI input* Enabled when random trigger shutter is ON.

[Internal sync signal specification](1) Driving frequency

60.000MHz(2) Scanning frequency
Readout mode
All pixel readout

Horizontal:31.185 kHz

Vertical:15.104 Hz

High-speed draft readout Horizontal

Partial scan

:9.632 kHz

Vertical:37.098 Hz

Horizontal:31.185 kHz

Vertical:Arbitrariness[Input signal specification](1) TRIG

Camera link I/F and I/O connector input

Signal level (I/O input)

TTL level

Polarity

Positive/Negative polarity switching possible(initial factory setting: Negative)

Pulse width(2) MULTI

30.1μs or more
Camera link I/F input

Polarity

Negative polarity

Pulse width

30.1μs to 10 ms
This specification subjects to change without notice.

Oct. 07. 2009

Documents No. D4167119A- 14 -

Rev 1.1[Output signal specification](1) WEN

I/O connector output

Signal level

4 V (p-p)

Polarity

Positive polarity

Pulse width

Approximately 32.1μs[Dimensions](1) Lens mountC-mount

Notes on combination of C-mount lens:
Depending on the lens you use, the performance of the camera may not be brought out fully due tothe deterioration and brightness in the peripheral area, occurrence of a ghost, aberration and others.
When you check the combination between the lens and camera, be sure to use the lens you actuallyuse.
As for the C-mount lens used combining this camera, the projection distance from bottom of the
Bottom ofthe screw

screw should use 9.5mm or less.
Special-mount lens

9.5mm or less(2) Flange back17.526 mm(3) Dimensions

54 mm (W) x 43 mm (H) x 59 mm (D)(4) Mass

Approximately 180 g(5) Camera body grounding: Insulation status
Conductive between circuit GND and camera body

[Operating ambient conditions](1) Performance assurance(2) Operating guaranteed(3) Storage environment

Temperature

0 to 40 °C

Humidity

10 to 90 % (no condensation)

Temperature-5 to 45 °C

Humidity

90% or less (no condensation)

Temperature-20 to 60 °C

Humidity

10 to 90% (no condensation)

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 15 -

Rev 1.1[Typical spectral response]

The lens characteristics and light source characteristics is not reflected table.

[Applicable safety standards](1) EMC conditions
EMI (Electro-Magnetic Interference)

EN61000-6-4 / 2007

EMS (Electro-Magnetic Susceptibility)

EN61000-6-2 / 2005

(2) FCCFCC Part 15 Subpart B class A* About the conformity of EMC standard of this machine, it has guaranteed in the conditions combinedwith our system condition. When used combined parts other than specification of our company, I ask
you to have final EMC conformity checked of a visitor with a machine and the whole equipment.

[Communication specification](1) Communication speed

9600 / 19200 / 38400 / 57600 bps(2) Start bit

1 bit(3) Data bit

8 bit(4) Start bit

1 bit(5) Parity bit

None(6) Handshake

None

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 16 -

Rev 1.1[Connector pin assignment](1) Video output/controlling connector (Camera Link Base Configuration) CAMERA LINK
Output video signals and VALID, based on the camera link standard LVDS.
This connector is connected to the frame grabber board, image processing device and others.
Connector model: MDR26-PIN connector 10226-2210PE (manufactured by 3M)
Pin No.

I/O

Signal namePin No.

I/O

Signal name1-

GND (+12V)

14

-

GND

2

O

Tx OUT0-

15

O

Tx OUT0+

3

O

Tx OUT1-

16O

Tx OUT1+

4O

Tx OUT2-

17

O

Tx OUT2+

5O

Tx CLK OUT-

18

O

Tx CLK OUT+

6O

Tx OUT3-

19

O

Tx OUT3+

7

I

Ser TC（RxD）+

20

I

Ser TC（RxD）-

8

O

Ser TFG（TxD）-

21

O

Ser TFG（TxD）+

9

I

CC1（TRIG）-

22

I

CC1（TRIG）+

10

I

CC2（MULTI）+

23

I

CC2（MULTI）-

11

I

CC3-

24

I

CC3+

12

I

CC4+

25

I

CC4-

13

-

GND

26

-

GND (+12V)

NOTE: For model CSCQS15BC23, 1-pin and 26-pin are GND.
Please combine and use the cable and the grabber board for Camera Link.
NOTE: For model CSCQS15BC23-01, 1-pin and 26-pin are POWER (+12V).
Please combine and use the cable and the grabber board for PoCL.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 17 -

Rev 1.1(2) Connector for power supply and sync signal input/output I/OThis is a terminal used for power supply to the camera. This connector is connected to the power
supply unit. In addition, this connector is used for sync signal (WEN signal) output and external
trigger signal input.
Connector (Camera side): HR10A-10R-12PB (Manufactured by HIROSE DENKI)

Plug (Cable side): HR10A-10P-12S (Manufactured by HIROSE DENKI)

Pin No.

I/O

Signal name

1

-

GND

2

I

+12V (N.C.)

3

-

N.C.

4

-

N.C.

5

-

GND

6

-

N.C.

7

O

VD

8

-

GND

9

-

N.C.

10

O

WEN.

11

I

TRIG

12

-

GND

NOTE: For model CSCQS15BC23, 2-pin is POWER (+12V).
NOTE: For model CSCQS15BC23-01, 2-pin is OPEN.

[Switch setting]
By using the DIP switches on the back surface of the camera body, you can set serial transmission speed
and memory readout for when the power supply is turned on. If you change the switch setting after the
power supply is turned on, the change is not reflected.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 18 -

Rev 1.1(1) Transmission speed setting
You can set the speed of serial transmission by camera link.

SW2

SW3

Transmission speed

OFF

OFF

9600 bps

ON

OFF

19200 bps

OFFON

38400 bpsONON

57600 bps(2) Memory readout setting
You can set the number of setting the value saving memory bank to be called when the power
supply is turned on. The memory consists of 8 banks.

SW4

SW5

SW6

Memory number

OFF

OFF

OFF

1

ON

OFF

OFF

2

OFF

ON

OFF

3

ON

ON

OFF

4

OFF

OFF

ON

5

ON

OFF

ON

6

OFF

ON

ON

7

ON

ON

ON

8

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 19 -

Rev 1.1

8. Timing chart(1) Horizontal timing
1) All pixel readout, Partial scan
FVAL
LVAL
DVAL
DATA OUT1DATA OUT2
a

c
d

b
e

a = 41CLK

b = 1228CLK

c = 40CLK

d = 615CLK

e = 1924CLK

2) High-speed draft readout
FVAL
LVAL
DVAL
DATA OUT1DATA OUT2
a

c

b

d

e

a = 41CLK

b = 1228CLK

c = 40CLK

d = 4920CLK

e = 6229CLK3) CLK late

DATA CLOCKDVALDATA OUT1

1

2

3

4

1225

1226

1227

1228

DATA OUT2

2456

2455

2454

2453

1232

1231

1230

1229

a
b
a = 16.67ns（60.000MHz)b = 20.47μs（48.86kHz）
This specification subjects to change without notice.

Oct. 07. 2009

Documents No. D4167119A- 20 -

Rev 1.1(2) Vertical timing
1) All pixel readout*ex) shutter mode: OFF

The period of A/C depend on the shutter speed. B period is fixed.

2) High-speed draft readout*ex) shutter mode: OFF

The period of A/C depend on the shutter speed. B period is fixed.

3) Programmable partial scanThis specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 21 -

Rev 1.1

9. Outline Drawing
CSCQS15BC23 (Non-PoCL):
There is NOT "PoCL" display.
CSCQS15BC23-01 (PoCL):
There is "PoCL" display.

4-40 UNC
depth3C-mount

4-M3 depth6

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 22 -

Rev 1.1

10. Guarantee
The term of a guarantee is one year after the product delivery. If by any chance trouble by responsibilityof our company occurs before an above period, TELI repairs it free of charge. During terms of aguarantee, when the trouble cause is the case of below, TELI charges the repair costs.
(1) Troubles and the damages that causes by misuse, unsuitable repair or remodeling.
(2) Distribution hazards like drops and vibrations after purchase. Troubles and damages bytransportation.
(3) Troubles and damages by fire, natural calamity (earthquake, storm and flood damage, thunderbolt),
damages from salty breeze, gas harm, abnormal voltage.

11. Repair(1) Condition for repair
Basically, has to return it to our company when the user requests us to repair product.
Beside that, customer should pay these expenses (travel expenses, camera disassemblyTechnology costs) of both customer and end user. Also customer should pay in themselves costs
for return camera to us.
(2) The period of repairing product- Repair free of charge … Refer to Clause 10.
- Charged repair ……….. Basically, repair period is 7 years after the last production end of
products.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 23 -

Rev 1.1

Head Office: 7-1, 4 chome, Asahigaoka, Hino-shi, Tokyo, 191-0065, Japan(International business department)

Phone : +81-42-589-8771
Fax: +81-42-589-8774

URL: http://www.toshiba-teli.co.jp/

Distributor•This product must be classified for disposal according to the laws of each country and municipal laws.
•Information contained in this document is subject to change without prior notice.

This specification subjects to change without notice.
Oct. 07. 2009

Documents No. D4167119A- 24 -