Tierrett Stomach Tube for Large Animals
Instruction for Use[Shape, structure, principle][2] Pay attention to damage of the tracheal wall and incorrect insertion andPolyvinyl chloride (plasticizer: di (2-ethylhexyl) phthalate) is used for thisplacement in the trachea or lungs. If resistance is felt when inserting theproduct.

catheter, there is a risk of erroneous insertion into the lungs. Do not insert thecatheter forcibly, remove it first, then insert again.
[3] Never block the tube lumen with lubricant.

<Shape>[Precautions]

Effective length<Important basic caution>

Size description

Outer diameter[1] Perform flushing before and after administration.
O.D.**ｍｍ - I.D.**ｍｍ[2] In removing the blockage of the catheter, use an injector and never perform a
Inner diameter

Connector tube

Body tube

forcible pressurizing operation.
[3] Do not pinch the device with forceps too strongly.
Connector（Milky white）[4] The milky white connector is for keeping the connector tube shape and out ofuse this product.

SizeOuterdiameter

Innerdiameter

Effective length

OD10-ID7

10.0 mm

7.0 mm2,100 mm

OD13-ID10

13.0 mm10.0 mm

OD15-ID11

15.0 mm11.0 mm

OD18-ID14

18.0 mm14.0 mm

OD20-ID16

20.0 mm

16.0 mm<Failures・Adverse events>
Failures[1] Occlusion of the catheter.
[The lumen of the catheter may be occluded by the adhesion of drug solution,
stomach constituents or etc.][2] Cut of the catheter.

2,700 mm[Cut due to the following causes]- Damage caused by forceps, scissors, knife or other apparatuses.

3,000 mm

- Sudden load on the product such as self (accidental) removal.

3,300 mm- Other combined causes due to the above events.

<Raw Materials>

Adverse events
In using this product, the following adverse events are expected:

Polyvinyl chloride- Nasal bleeding, middle ear inflammation, erosion and ulcer of the ala of nose,<Principles>damage and ulcer of airway mucosa and gastric mucosa, intratracheal invasion,

Insert a catheter nasally or orally to the stomach to clean or aspirate the esophagus

aspiration pneumonia, vomiting, gastroesophageal reflux, etc.

and stomach and to administrate drug solution. Injected solution is administered- Remains in the body due to cutting of the tube.

through a lumen and side holes into the body.
[Storage conditions and duration of use]
<Storage conditions>Store the product hygienically, avoiding the direct sun light, high humidity andultraviolet rays such as a sterilizing lamp and taking care of wetting.

[Intended purpose, efficacy or effect]
Insert a catheter nasally or orally to the stomach to clean or aspirate the esophagus
and stomach and to administrate drug solution.
[Operating or using method]
The below are general procedures.

<Expiration date>
See the expiration date given on each package provided that the device is stored
appropriately.

[1] Have a patient take a proper body position for a situation and performmechanical retention appropriately.

[By self-authentication (our data).][2] Apply a lubricant or a surface anesthetic to the nostrils if necessary.
[3] Remove the milky white connector at the catheter end.

Manufacturer[4] Apply a lubricant or a surface anesthetic to the catheter tip and insert it throughthe nostrils or mouth.

CREATE MEDIC CO.,LTD.
2-5-25 Chigasakiminami, Tsuzuki-Ku, Yokohama, Japan[5] When the catheter reaches the pharynx, guide it to the esophagus. Afterchecking that it has entered the esophagus, insert it into the stomach.
[6] When the catheter tip reaches the stomach, perform the necessary treatment

In-house administration number: DC67348

via the catheter.
[7] In removing the catheter, slowly remove it from the nasal or oral cavity.
<Precautions for the using method>[1] In sterilizing this product, never perform autoclave sterilization.
[The tube and connector may be damaged.]1／1

R0-0 2021.12.18