HANDHAND
TACHOMETERTACHOMETER

TM-5013/TM-5014

TM-5000/TM-5010

In contact adapter
Rubber tip
Surface speed wheel

Non-contact and contact
measurement using an adapter
Measurement for hard to reachareas using the remote sensor
6.0~99999.9 rev/min Measuring
range with multi-unit selection
Easy to measure
In conformance to CE

TM-4015
Remote sensor

RoHS
TM-5000

0.1 rpm resolution

TM-5010/TM-5010E

8 items memory,
MIN/MAX memory,
rpm/rpsunit of peripheral speed selection

TM-5000K/TM5000EK
TM-5000 Kit models
TM-5010K/TM-5010EK
TM-5010 Kit models
TM-5010E Kit models
All kit models includes the main unit,reflective tape,contact adapter, rubber tip, surface speed wheel, AAAbatteries and carring case.

ModelsSpecifications (Main unit)
TM-5000

Model

TM-5010/5010E
6.0 - 99999.9 rpm
0.10 - 1666.66 rps
0.60 - 9999.99 m/min
1.00 -16666.6 cm/s
[0.39 - 6561.65 inch/s]*[1.9 - 32808.3 f/min]*MeasuringRange

6.0 - 99999.9 r/min

Resolution

0.1 r/minAccuracy±0.01%±1digit r/min, m/min(for other units, the conversion accuracy is ±0.05%±1digit)

1 count, 0.1 rpm fpm m f,others 0.01

Sampling Time
Display
Battery alarm
Reflective light: 6 digits, 7 segment LCD:mark:mark
Display unit : r/min, r/s, m/min, cm/s[y/min, inch/s, f/min]*, count, m, [y, inch, f]*Memory display : MAX,MIN,MEM1-8

Display unit : r/min

Auto Power-off

After 3 mins. from last measurement or key operation
Measurement data: until the next data is defined

Data Hold Time
Measuring
MethodMeasuring
Distance

TM-5000K/
TM-5000EK
TM-5010K/
TM-5010EK

TM-5010/TM-5010E Contents
In-contact Adapter unit
Rubber Tip, 3 pieces
Surface speed wheel (1/10m)
Carrying Case (C-4000)

TM-5000

TM-5000 main unit
AAA alkaline battery, 4 pieces
Reflective tape, 10 pieces
Instruction manual

TM-5010/TM-5010E

TM-5010 / TM-5010E main unit
AAA alkaline battery, 4 pieces
Reflective tape, 10 pieces
Instruction manual

TM-5013

In-contact adaptor
Rubber tip, 3 pieces
Surface speed wheel (1/10 m)

TM-5014

In-contact adaptor
Rubber tip, 3 pieces
Surface speed wheel (1/10 y)

TM-4015

Remote sensor(21 mm diameter, maximum
cable length is 1 meter)

1.0 - 10.0 seconds

Display

Memory data: same as battery life

[Non-contact measurement using the main unit or with remote sensor (use with reflective tape)Contact measurement using the in-contact adapter (use with rubber tip, surface speed wheel)]* This item is for TM-5000EK only.

50 - 300mm (using reflective tape)Power Supply

Accessories (Sold separately)

4 pieces of AAA alkaline battery (continuous measurement of 20 hrs.)

Storage
Temperature
OperatingTemperature
Operating
Humidity
Dimension/
Weight[[0.65 - 10936.1 y/min]*
0 - 999999
count
0.0 - 99999.9 m[0.00 - 99999.9 y]*[0.00 - 99996.1 inch]*[0.0 - 99999.7 f]*

TM-5000 Contents
In-contact Adapter unit
Rubber Tip, 3 pieces
Surface speed wheel (1/10m)[Surface speed wheel (1/10y)]*
Carrying Case (C-4000)-10 - 60°C (Non-Freezing)5 - 40°C (Non-Freezing)
35 - 85%RH (Non-Condensing)122(H) x 58(46)(W) x 28(D)mmapprox. 130g (including batteries)

Reflective tape

10 pieces / set

Rubber tip

3 pieces / set

Surface speed wheel (m)Circumference 1/10 m

Surface speed wheel (y)Circumference 1/10 y

C-4000Carrying Case]* These measuring units are only available in TM-5010E model.

PartsSurface speed wheel (circumference 1/10m or 1/10y)

Rubber Tip

Combination of adapter with TM-5000 or TM-5010 / TM-5010E is available.
For TM-5000, displayed value is 1/10m/min.
For TM-5010 / TM-5010E, calculations are done internally to display
the units m/min, cm/sec, f/min, inch/sec and y/min,
which can be read directly.

Remote Sensor
In-contact adapter

Contact measurement or surface speed measurement is possible using the rubber tip or thesurface speed wheel in combination with TM-5000 or TM-5010 / TM-5010E.

Attention!
When using at high speed, be very careful that the portion being measured don't get dragged on.

Memory number display

rotational speed mode(r/min r/s)

Photoelectric part
Unit display
r/min

Photoelectricdetector
indication

Photoelectric detector indication

cm/s m/min r/s r/min

MAX MEM
MIN

rotation count and length
measurement mode
(count m y *inch f* )*Rotational / Linear speed
Units Group selection key

Measurement display
Measurement button

While it is pressed, measurement is performed.

Connector for remote sensor probe
Measurement mode switch

Connector for remote sensor probe

The remote sensor can be used by turning
the slide switch inside the battery box
located at the back of this unit.

MEAS

By pressing this button, display changes as follows:
Measurement value MAX value MIN value

TM-4015
TM-5000

LINE SEIKI

TM-5000

8-point memory data
input key
8-point memory data
read/clear key

linear speed mode**
(m/min cm/s y/min inch/s
f/min)*MAX MIN.../min.../sm/min

TM-5010f/min *CALLCLR

Unit selection key
r/minr/s

LINE SEIKI

TM-5010 / TM-5010E

countf*cm/sy/min *inch/s *my *inch ** TM-5010E Only

Address: 37-7 Chuo-cho, 2-Chome Meguro-ku,Tokyo 152-0001 Japan
Please visit our company website for contact details
E-mail : webtrade@line.co.jp
Web: http: //www.lineseiki.com
This catalog was last revised as of Nov 24, 2022.
*Specifications and printed contents are subject to change without prior notice.
All Rights Reserved, Copyright2022, LINE SEIKI CO., LTD.