GC2/GH2/GM2/GS2 Series

Dimensions
Please note that misuse of this device
may lead to injury to the user or
damage to the device.

Address:Contact :E-mail :Web:

LINE SEIKI CO., LTD.

24.8

29

Wide Range Voltage Input Type37-7 Chuo-cho, 2-Chome
Meguro-ku, Tokyo 152-0001 Japan
Please visit our company website for contact details
webtrade@line.co.jp
https://www.lineseiki.com

50+0.7+0

3.8

8 Digit

38.3

3.5

11G2

Wiring

C: Totalizer
H: Hourmeter
M: Minutemeter
S: Secondmeter1 2 3 4 5

Panel Cut Dimension

0: Terminal BlockInput

0: Switch/Open Collector Input
1: Wide Range Voltage Input

Reset

0: Remote
1: Remote & Front

Display

6: 6 Digits
8: 8 Digits (GC2 only)

D r y C o n ta ct/Op e n C o l l e cto r /Vo l ta g e ( N PN Tr a n si sto r ) In p u t Typ e
INPUT

1

2

34 5Wi d e - R a n g e Vo l ta g e In p u t Typ e

RESET
INPUT

1GND

2

34 5RESETDISPLAY DIGITINPUT

3Wide-Range Voltage Input

Vcc
2 to 30V

GND

3GC2 TOTALIZER

GH2 HOURMETER6/8 Digits, 7-Segment LCD
10.6mm (H)
0 - 999999 / 0 - 99999999
Switch / Open Collector
Wide Range Voltage

GM2 MINUTEMETER

GS2 SECONDMETER

GNDMounting

6 Digits, 7-Segment LCD 10.6mm (H)0.0 - 99999.9
L: < 0.8V H: < 2 - 30V
Sink Current:8 A
L:DC / AC 0 - 5V
Input Impedance: 300kΩ
H:DC / AC 20 - 250V

1. Please install the device in the correct panel cut-out size.
2. A gasket is included with the device.
Insert the gasket in the device before inserting it in the panel.
3. Insert the mounting clip to the back of the device.
4. Push the mounting clip until it locks the device tightly against the panel.

30 Hz
COUNT SPEEDRESOLUTION

Gasket20Hz(Wide Range Voltage Input Only)
1 count per 1/10 hour
Front Reset
Remote ResetEMC STANDARD
AMBIENT TEMP. : OPERATINGAMBIENT TEMP. : STORAGE
AMBIENT HUMID. : OPERATINGVIBRATION /SHOCK RESISTANCE
ALTITUDE
INSTALLATION ENVIRONMENTPROTECTION
WEIGHT
WIRINGCOMPLIANCE

Mounting ClipPanel

1 count per 1/10 minute1 count per 1/10 second

Battery Replacement
ATTENTION!!!

L: < 0.8V H: < 2 - 30V Sink Current:
8 A
RESET
L:DC/AC 0 - 5V
Input Impedance: 300kΩ
H:DC/AC 20 - 250V
6ms minimum
RESET PULSEWIDTH
3.6V 1/2AA Lithium Battery (replaceable), 7 years min. with continuous input at 25°C *The battery life is not a guaranteed value.
POWER
NOISE IMMUNITY

Dry Contact / Open Collector /Voltage(NPN transistor) input

GND

2 to 30VDISPLAY

T heory on
GX2 Series Counter

Vcc

Specifications
TYPE

49.8
Terminal No.

WiringModels

Type49.8
Terminal No.

24.8

Panel mount(50.0 x 25.0mm)
6 / 8 digits LCD display
Lithium battery included
Battery life 7 years
IP54 Front Panel protection level
In conformance to CE (EMC),RoHS and UL 863

6 Digit25+0.7+0Main Features

6.5

1 2 3 4 5
10.2 12.6

Customer Service

38.3

3.5

10.6

WARNING

Thank you for purchasing this Line Seiki product.
Please read this instruction manual carefully before using to ensure correct
usage of this product.
Please keep this instruction manual for future reference.

4.8

10.6

Inst r u c t i on M a n ual

Switch/Open Collector Input Type5443.2

Digit Dimension

Switch/Open Collector
Wide Range Voltage

Electrostatic Discharge:
IEC-61000-4-2 8kVElectromagnetic Field:
IEC-61000-4-3 10V/m
EFT/BI/O Leads:
IEC-61000-4-4 1.0kV
EMI:EN61000-6-4, EMS:EN61000-6-2-10 - +55°C (Should not be Frozen))-20 - 65°C (Should not be Frozen)
25-- 85% RH (Non-Condensing)Operation : Single Amplitude 0.35mm, Frequency: 10 - 55Hz, X, Y, Z Axis (3 Axis) / 100m/s2
Storage : Single Amplitude 0.75mm, Frequency: 10 - 55Hz, X, Y, Z Axis (3 Axis) / 300m/s2

Precautions when replacing the battery:- Disconnect the wiring before changing the battery. Do not touch areas
subject to high voltages, this could result to an electric shock.
- Make sure that your body is free from static electricity before changingthe battery.
- Make sure to follow carefully the procedures described below.
- Battery replacement resets the count value.

WARNING!

2,000m max.
Over-voltage category II , Pollution degree 2 (IEC61010-1)
IP54
60g (Battery included)
14-22 AWG Leadwires by Terminal Screw
RoHS, UL863, CE

EXPLOSIVE!

1) Remove the terminal block from the main device. (except wide range
voltage input type)2) Use a flat screwdriver or any appropriate tool to lift the groove in order
to disconnect the 'front and back' case hooks.
3) Pull out the back case from the main body.
4) Replace with a new battery making sure that the positive and negative
terminals of the battery are positioned correctly.
5) Replace the back case to the main body.
Make sure that the case hooks click properly in place.
*Please use 3.6V 1/2AA Lithium Battery only for this device.

This device uses Lithium battery, it must be disposed properly after use.
Do not short the positive and negative terminals, recharge, disassemble, deform by the application ofexcessive pressure, or dispose the battery to fire as this may seriously damage it or may cause explosion.
Do not attempt to disassemble or modify this device.
Mount to the front panel when using this device. (Indoor use)
Do not use organic solvents such as thinners etc. to clean the front panel.
Internal circuit may be destroyed if a voltage outside the rated voltage is applied.
This manual was last revised Nov. 22, 2022. 4GX2001I*Specifications and printed contents are subject to change without prior notice.
All Rights Reserved,Copyright © 2022,LINE SEIKI CO.,LTD.