Check Weighing Indicator

AD4412-CW

7” high-visibility touch panel color LCD
Up to 1,000 registered items
4000 times/second high-speed sampling
A/D resolution 16,000,000 counts

http://www.aandd.jp

AD4412-CW

Check Weighing Indicator

Feedback control

Three types of weighing modes: Passing, Stopping, and Static Weighing
Reports for operational history, weighing history, histogram, control charts, andsummary reports.
Display various data on smartphones and tablets over wireless LAN during measurement.
Item photo import from a USB Flash Drive
Easy setting restoration using a USB Flash Drive.
Save data to a USB Flash Drive and import it to personal computer.
Dust/water protection for the IP65 display when mounted on a panel with OP-10.
Wireless LAN*

7” LCD touch panel

USB interface Type-A

Start/Stop switch

RS-485, photo sensor power supply

Panel mounting
bracket (OP-10)

Power switch

Load cell terminal

Power supply connection terminal
LAN connector

RS-232C

Option slot

Control I/O*Wireless LAN can be used depending on the regulations of your region.

Data output via wireless LAN*

TabletiPhone or AndroidPC

USB Flash Drive*Backup
Product photo registration

Register all products,
scale settings,
system settings, andoperation history

Weighing history,
histograms,
control charts, andaggregated data

*Wireless LAN can be used depending on the regulations of your region.
*A USB Flash Drive not included as a standard accessory.

Modbus RTU

Modbus RTU

PLC

HMI

RS-485 communication

RS-485PC

Photo sensor (power supply 12V and signal input)
Modbus TCP

Modbus TCP

PLC

HMI

LAN connector
RJ45

Control I/O

RS-232C

Postscript laser printerFeedback control
EMERGENCY

STOP

Barcode reader
item settingPC

AD8927

Rotating light

Rejector

Filling andpacking machineWarning light

Emergency stopbutton

Data Output via Wireless LAN*

The measurement results, setting conditions, and statistical data can be checked at any time without stopping the checkweigher.
The X control chart and R control chart can be used to understand the trends of measurement values, and check the eﬀect of feedback control.
*Wireless LAN can be used depending on the regulations of your region.

Operational History
The history of conﬁguration changes can be recorded and displayed.
This is suitable for use in HACCP programs along with the weighing history function.
Users can be registered and the scope of their permitted operations
can be controlled according to 4 management levels.
[Operator][Supervisor][Quality Manager][Administrator]
By assigning each user to the appropriate level of access,inadvertent operations can be avoided.*2* 2 "Operator" is set when the power is turned on.

Weighing History
Weighing results are automatically recorded to a USB Flash Drive during weighing.
Output data: date, time, group product number, weight data, and judgments.
Only USB Flash Drive formatted in FAT32 can be used.The weighing history
that is output to the USB Flash Drive is saved in the memory root in CSV
format.
USB Flash Drive

Approx. 6MB of memory is required for 8 hours of operation at the maximum
throughput (320pcs/min.)Approx. 7GB of memory is required for 24-hour/365-day operation.
*A USB Flash Drive not included as a standard accessory.

Weighing history output example
2016/10/3, 19:45:52, 40, 01–001, 100.05, 0K
2016/10/3, 19:45:56, 54, 01–001, 150.1, Over
2016/10/3, 19:46:00, 58, 01–001, 70.5, Under
2016/10/3, 19:46:04, 52, 01–001, 0.0, Detect
2016/10/3, 19:46:08, 56, 01–001, 0.0, Unsp l i t
2016/10/3, 19:46:12, 60, 01–001, 100.5, Metal
2016/10/3, 19:46:16, 54, 01–001, 105.1, Ext 1
2016/10/3, 19:46:20, 78, 01–001, 95.5, Ext 2

Histograms, Control Charts, and Summary Reports
After weighing, press the PDF key on the summary display.
A PDF report is output to a USB Flash Drive.
The same reports can be printed out by pressing

Histogram printing example

the PRINT key when a PostScript printer isPrint date and timeconnected to the checkweigher.

Settings

Frequency distribution

Frequency distribution

Control chart printing exampleSummary results printing example
Print date and time

Print date and time

Settings

Settings

X control chart

R control chart

Dump Print Printing Sample

Compact printer AD8127

Character speciﬁcations 7 ✕ 9 dots, 1.6 (W) ✕ 3.2 (H) mm
Printing speed
Approx. 1.6 lines/second
Paper feeding speed
Approx. 4 lines/second
Number of characters printed 24 characters/line
Printer paper width
57.5 mm
Operating environment

0 to 40˚C/32 to 104˚F, 35 to 85%RH(no condensation)

Storage environment–20 to 60˚C/–4 to 140˚F, 10 to 90%RH(no condensation)

Interface
Power supply
Size / weight

RS-232C (D-Sub 9P, male) / Current loop (DIN)
AC adapter
174 (W) ✕ 141 (D) ✕ 87 (H) mm / approx. 500 g

Summary data
Statistical data

Applications
RS-232C (item setting)

Postscript printerLAN

Histograms, control chars, andsummary reports

Reject signal (control I/O)
Barcodereader

Rejector
Photoelectricsensor

Load cell

Load cell● feedback control
Filling machine

HMI
Target 1000g
High limit 10g
Low limit 5g

1002g

Modbus communication (LAN or RS-485)

Feedback signal output (control I/O)

● Conveyor speed control
The conveyor speed can be changed

OP-07 + OP-17 ✕ 2

automatically for each item
4-20mA or 0-10V

4-20mA or 0-10V
4-20mA or 0-10V

Load cell● Item is set in the barcode reader
RS-232C (item setting)
Barcodereader
Settings change to the item type of the number read by the barcode reader.

Speciﬁcations

Temperature coeﬃcients
Nonlinearity
Sampling rate
A/D resolution■ Digital unit

Number of digits
Display■ Registered items
Registered items■ General speciﬁcations
Power source voltage
Operating temperature
Operating humidity
Dust/water protection■ Standard Interface

Communication interface
Control I/OPower supply for photo sensor

Over 0.15µV/d (d=minimum division)–35mV to +35mV (–7mV/V to +7mV/V)
DC 5V 60mA up to 4 load cells (350Ω)Remote sensing function included
Zero: ±0.02µV/˚C (Typ.)Span: ±3ppm/˚C (Typ.)0.005% of F.S.
4000 times per second
Approx. 16,000,000■ Options
OP-02
OP-05
OP-07

OP-17
OP-10
OP-11

Relay output 9 points DC40V 50mA
Parallel I/O DI 16 points DO 16 points DC40V 50mA
Analog output 1ch 4-20mA or 0-10V
add OP-17 ✕ 3 then Maximum 4ch
Output weighing value, conveyer speed control
Analog output module 4-20mA or 0-10V
Panel mount bracket
Bracket for display stand

999,999 d7” color TFT LCD with touch panel

1,000

AC100V - 240V (+10%, -15%, 50/60Hz, 30VA)−10˚C to 50˚C
85% RH or less (no condensation)
The IP65 display when mounted on a panel (requires OP-10).

RS-232C, RS-485, LAN, USB2.0, Wireless LAN*
Input 11 points (non-voltage connection point or open collector)
Output 11 points (open collector)
DC40V 50mA12V 250mA*Wireless LAN can be used depending on the regulations of your region.

External dimensions
2147 21

Unit: mm
19.3

198

*Wireless LAN receiving sensitivity depends on the radio waveenvironment.
Caution is required when performing continuous monitoring.

(191.8)
144.5

STOP

180

Input sensitivity
Zero adjustment range
Load cell excitation

START

190 +0.3
0Panel cut

5R14-

182 +0.30■ Analog unit

Appearance and/or speciﬁcations are subject to change without notice.

A&D Company, Ltd.

3-23-14 Higashi-Ikebukuro, Toshima-Ku, Tokyo, 170-0013, Japan Tel: +81 3-5391-6132 Fax: +81 3-5391-1566 http://www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131, U.S.A. Tel: +1 408-263-5333 Fax: +1 408-263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031, Australia Tel: +61 8-8301-8100 Fax: +61 8-8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire, OX14 1DY, United Kingdom Tel: +44 1235-550420 Fax: +44 1235-550485<German Sales Office>Hamburger Straße 30, D-22926, Ahrensburg, Germany Tel: +49 4102-459230 Fax: +49 4102-459231

A&D Korea Ltd.

8F Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331, Korea Tel: +82 2-780-4101 Fax: +82 2-782-4280

A&D Rus Co., Ltd.

Vereyskaya Str. 17, 121357, Moscow, Russia Tel: +7 495-937-33-44 Fax: +7 495-937-55-66

A&D Instruments India (P) Ltd.

509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India Tel: +91 (124) 471-5555 Fax: +91 (124) 471-5599* AD4412CW-ADCC-01-ZW1-19903GP