型式：RS02T-R03-K300

75Ø1

RS02T-R03-K300

Transmitter

179.50Ø 100 - 1

101.5

0...9mm± 8mm

2 signals

Operating distanceCenter oﬀset

Input channel

8.5L

Signaltransmission

Output sensor
RS02E

RS02E-R03E-PU-̲̲

Output Sensor

velee（Sindner<= 0.5 sec.

◆ The inzone signal is a preliminary signal for conﬁrming that the output signal is established within the speciﬁcation range. Please note that it does notguarantee signals output outside the speciﬁcation range.

2017.03.22

No.T315B02B◆ Please note that the signal may become unstable (false signal or chattering) when the transmission distance and the center oﬀset are outside thespeciﬁcation range.

Transmitter and Output sensor
Operating Temperature 0...+50 degree CMaterial
PUR<= +/- 0.8 % full schale range

PUR、 φ 5mm/4x0.25mm2

Responce speedLinearity≦ 400 Ω<= 0.04 % full schale range

Resolution

Compensated cold junction ≦± 0.5℃

Cable

4 ... 20 mA × 2 signals

Load resistance

Measuring temperature range 0...300℃

Compensation lead wire(JIS) phi 0.9 mm x 2
All heat-resistant vinyl (90 deg. C)

INZONE (data valid)

Current consumption ≦ 150mA

L=Cable length
The notation in meters tothe end of the model･･･PU-01 ⇒ 1mter）

eiamA p p l i c a b l e
RS02T-R03-K300
Transmitter
Supply voltage
24V DC +/- 5 % (include ripple)

Type code

Output sensor：RS02E-R03E-PU-̲ ̲

Output Sensor: Change the temperature data to analog signal (4...20mA)
and output to external unit and supplies power for
operation of Transmitter at the same time.

: (1) Detects the voltage of thermocouples which changes
depending on temperature.
(2)The internal CPU converts the temperature data of (1)into digital signals and transmits the signals to the
Output Sensor.

Transmitter

Hostdevice（24 VDC): Two thermocouples type K are used as a detector
and it detects temperature.

OutputCable

PowerDetector

4...20mA

LED± 15mm0...15mm

Non metal shaft

28.5

4×4× Ø
1ve
lee付ス（S取Ø 9（r dブneーinリete径r））
iam内Detectedsignal

Transmitter
RS02T【Function of each component】

Thernocouple

Thernocouple

Power supply

Power supply[Detector]㻼㻸㻯㻌㻌㼛㼞㻌㻼㼍㼚㼑㼘㻌㼙㼑㼠㼑㼞㻌㻻㼁㼀㻝㼍㼚㼐㻌㻞㻌㼍㼞㼑㻌㻠䈈㻞㻜㼙㻭㻌㼏㼡㼞㼞㼑㼚㼠㻌㼟㼛㼡㼞㼏㼑㻚㻻㼡㼠㼜㼡㼠㻌㼟㼑㼚㼟㼛㼞㻌㻮㼞㼛㼣㼚㻔㻗㻞㻠㼂㻕㻌㻮㼘㼍㼏㼗㻔㻻㼁㼀㻝㻕㻌㼃㼔㼕㼠㼑㻔㻻㼁㼀㻞㻕㻌㻌㻮㼘㼡㼑㻔㻳㻺㻰㻕㻌[Output Sensor]

5101520-40 -30 -20 -10 0Y10 20 30 40mm

XX

YSetting metal shaft

Non metal shaft

X:Center offset(mm)Y:Operating distance(mm)RS02T-R03-K300 / RS02E-R03E-PU-̲ ̲

Typical Transmitting Diagram (Supply voltage at 24V /non-ﬂush mount）- Output is current source , therefore please connect the load
between output and GND.

- For a detector, please use a thermocouple K that meets JIS.
- The measurement temperature should be lower than upper
limit of the temperature shown at the code end.
example : RS02T-R03-K300 ： 0...300 degree C【Causion】- Measure to static electricity
When using RS02 in environment with the electrostatic
discharge, take the ground to prevent infuluence by the staticelectricity on RS02.

Cable length
Transmitter (RS02T-R03-K300) ：max.3m
Output sensor (RS02E-R03E-PU)：max.10m

㻵㻺㻞㻌㻔㻗㻕㻌 㼞㼑㼐㻌
Thermocoupl 㻔㻙㻕㻌 㼣㼔㼕㼠㼑㻌㻵㻺㻝㻌㼀㼞㼍㼚㼟㼙㼕㼠㼠㼑㼞㻌㼀㼔㼑㼞㼙㼛㼏㼛㼡㼜㼘㼑㻌 㻔㻗㻕㻌 㼞㼑㼐㻌㻔㻙㻕㻌 㼣㼔㼕㼠㼑㻌[Transmitter]

Wiring diagram[Output sensor][Detector]

Internalcircuit[Transmitter]

Construction of the system

Applicable thermal sensor Thermocouple per K(1CH, 2CH)

Setting metal shaft

Installation

Applicable Output sensor RS02E-R03E-PU-̲̲

Type code

Speciﬁcation of the System

35

4×90°

Transmitter： RS02T-R03-K300

Dimension◆ To avoid malfunction caused by induction noise, cableshould be kept apart from motor or other power cable.

◆ Ensure correct connections by reference to the wiring diagram.

◆ Use a regulated power supply, e.g. switch-model type.
Simpler power supplies, such as a full-wave rectiﬁcation type,will cause the permissible ripple rating to be exceed and may
cause malfunction.

◆ Ensure the power is switched oﬀ during installation or
maintenance operations.

Before using the Remote Sensor, read this manual carefully.
During installation and operation, pay close attention to the safety aspect.

Attention for Installation(Read this section thoroughly before installation.)

Output Sensor : RS02E-R03E-PU
Transmitter:
RS02T-R03-K300

Remote Sensor system /Analog signal
Thermocouple / 2 signal transmission

10Internal
circuitRemote System
User’s Guide

86.5

56

13.5

Ø150mmAB
35C400(mm)Web：http://www.b-plus-kk.com(Former NIHON BALLUFF co., Ltd.)

200

C■ Parallel installation

Mail ：b-plus-usa@b-plus-kk.com

Wireless Power Supply byType code
RS02T-R03-K300
RS02E-R03E-PU-̲ ̲BA■ Surrounding metal

In order to avoid inﬂuence of surrounding metal, or to avoid mutualinﬂuence between parallel-mounted sensors, keep the minimum freezone as described below.

Installation notes＊ Never pull the cable strongin installing

The minimum
bendingradius forthesensors
are 50mm.

Bending radius of Cable

Output current
0 mA
４ ... 20 mA
Receiving the temperature（at 0 deg. C : 4mA, at the hightest temperature
data from Transmitterof the measurement range : 20 mA)
Thermocouple is notconnected, though the21 mA
transmitter is in thetransmitting range.

statusWithout Transmitter

Output sensor RS02E outputs the electric current as described below.

Current output