Nonhazardous Locations

Hazardous (Classified) Locations
Class I, Groups ABCDClass II, Groups EFG
Class IIIClass I, Zone 0 Group IIC
Class I, Zone 2 Group IIC

Installation Notes RIA46- FM Approved Apparatus must be installed in accordance with manufacturer’s instructions and the controldrawing.
- Depending on location install per National Electrical Code (NEC) using wiring methods described inarticle 500 through article 510.

- Use supply wires suitable for 5°C above surroundings.

FM approved intrinsicallysafe apparatus

L/-, L/+INTRINSICALLY SAFE CONNECTION TO
Class I, II, III / Div. 1+2 / Groups ABCDEFG- The device is an Associated intrinsically safe equipment and must be installed in Division 2 or nonhazardous locations only.
- Installation should be in accordance with ANSI/ISA RP 12.06.01 “Installation of Intrinsically safe systems
for Hazardous (classified) locations” and the National Electrical Code (ANSI/NFPA 70).
- For entity installations use certified equipment that satisfy the following condition
Uo/Voc £ Vmax/Ui
Io/Isc £ Imax/Ii
Po £ Pi
Co/Ca ³ Ci + Ccable Lo/ La ³ Li + Lcable- The Terminal of the intrinsically safe circuit must be placed at least a distance of 50mm from terminals ofthenon intrinsically safe circuits, or adequate separators (e.g. ground metal partitions) must be used.
NONINCENDIVE Field WIRING CONNNECTION TO Class I, II, III / Div. 2 / Groups ABCDEFG- The device is an Associated Nonincendive safe equipment and must be installed in Division 2 or nonhazardous locations only.
- The Nonincendive Field Wiring Circuit Concept allows interconnection of Nonincendive Field Wiring
Apparatus with Associated Nonincendive Field Wiring Apparatus or Associated Intrinsically Safe
Apparatus or Associated Apparatus not specifically examined in combination as a system using any of thewiring methods permitted for unclassified locations, when Voc £ Vmax, Ca ³ Ci + Ccable, La ³ Li + Lcable.

Note wiring scheme on device!
ANALOG OUT

D11 015 025
D12016 026-++0...20mA
4...20mA
CH1 0...10V
2...10V
0...5VOC 30V/200mA+0...20mA
4...20mA
CH2 0...10V
2...10V
0...5VANALOG INI

U>1Vx4x8MUS 2-Lx1 x2x8+-0/4...20mA

U<=1Vx30...10V
0/1...5V±10V±30Vx1 x2x7 x8=+MUS 4-Lx4+x2x8-+I

4...20mA

0/4...20mA

Pfanzelt

2009-06-03

Pfanzelt

2009-05-11

TCx4x5 x6 x7 x8x7 x8+4-L

I-±100mV...±1V

RTD
x4

3-L

-

Typ J,K,T,N...

2-L

12 04 00 111
RIA46-

1:1
A4--

CONTROL DRAWING
FM approval AIS, ANI-71540266
XA02312R/09/EN/01.20

Serie
1 of 2

Temperature range
Ta4-wire transmitter power supply:
Terminal 11, 12, 21, 22-20°C ... +60°C

AISClass I, II, III, Div. 1+2, Groups ABCDEFG
Cl. I, Zone 0 [AEx ia] IIC
Class I, II, III, Div. 2, Groups ABCDEFG

ANIPower supply
Terminal LI+, LI-, PEU £ 24…230 V AC/DC (-20%/+10%) 50/60 Hz

Output circuit limit relays
Terminal R12, R11, R13 orR22, R21, R23

Umax £ 250 VAC
Umax £ 30 DC

Group A, B resp. IIC
Group C, D resp. IIB, IIA
4-wire transmitter power supply:
Terminal 14, 18, 24, 28

Imax £ 3A
Imax £ 3A

Group A, B resp. IIC
Group C, D resp. IIB, IIA
temperature input:
Terminal 15, 16, 17, 18

CDI interface for device configuration
Impulse or Current output
Terminal O15, O16 or O25, O26

0/4…20 mA
Um £ 250 V

Output collector
Terminal D11, D12Imax £ 200 mA
Um £ 30 V

2-wire transmitter power supply:
Terminal 11, 14, 12, 18
21, 24, 22, 28
Group A, B resp. IIC
Group C, D resp. IIB, IIA

Voc £ 27.3 V
Isc £ 96.5 mA
Po = 659 mW
Ci = 8nF
Li = 75µH
Ca = 88 nF
Ca = 683 nF

Group A, B resp. IIC
Group C, D resp. IIB, IIA
Current input:
Terminal 14, 18, 24, 28

Group A, B resp. IIC
Group C, D resp. IIB, IIA

La = 4.2 mH
La = 17.1mH

Voltage input:
Terminal 17, 18, 13, 18
27, 28, 23, 28

Group A, B resp. IIC
Group C, D resp. IIB, IIA

Pfanzelt

2009-06-03

Pfanzelt

2009-05-11

12 04 00 111
RIA46-

1:1
A4-

Voc £ 27.3 V
Isc £ 91.1 mA
Po = 622 mW
Ci = 8nF
Li = 75µH
Ca = 88 nF
Ca = 683 nF

La = 4.7 mH
La = 19.2 mH

Voc £ 27.3 V
Isc £ 5 mA
Po = 34.2 mW
Ci = 8nF
Li = 75µH
Ca = 88 nF
Ca = 683 nF

La = 1.6 HLa = 6.4 H

Voc £ 27.3 V
Isc £ 22.1 mA
Po = 151 mW
Ci = 8nF
Li = 75µH
Ca = 88 nF
Ca = 683 nF

La = 81.8 mH
La = 327.5 mH

Voc £ 27.3 V
Isc £ 5 mA
Po = 34.2 mW
Ci = 8nF
Li = 75µH
Ca = 88 nF
Ca = 683 nF

La = 1.6 H
La = 6.4 H

Voc £ 27.3 V
Isc £ 5 mA
Po = 34.2 mW
Ci = 8nF
Li = 75µH
Ca = 88 nF
Ca = 683 nF

La = 1.6 H
La = 6.4 H-

CONTROL DRAWING
FM approval AIS, ANI-71540266
XA02312R/09/EN/01.20

Serie
2 of 2