GP01202F/00/EN/01.23-00
71601142
2023-01-12

Products

Solutions

Description of Device Parameters
Micropilot FMRxxB; Operating
tool
HART
Free space radar

ServicesAbout this document

Micropilot FMRxxB; Operating tool HART

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters. The document provides a detailed explanation of each individual parameter.
Performance of tasks that require detailed knowledge of the functioning of the device:• Commissioning measurements under difficult conditions• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Information on the document structureThis document lists the submenus and parameters that are available when the"Maintenance" option user role is enabled.
For the operating concept of the operating menus, see the Operating Instructions.

1.3.2

Structure of a parameter description

The individual parts of a parameter description are described in the following section:• Navigation: Navigation path to the parameter via the local display• Prerequisite: The parameter is only available under these specific conditions• Description: Description of the parameter function• Selection: List of the individual options for the parameter• User entry: Input range for the parameter• User interface: Display value/data of the parameter• Factory setting: Default setting on leaving the factory• Read and write authorization: Information about parameter access authorization• Additional information:• On individual options• On display value/data• On the input range• On the factory setting• On the parameter function

2

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

About this document

1.4

Symbols

1.4.1

Symbols for certain types of information

Additional information:
Reference to documentation:
Operation via local display:
Operation via operating tool:
Write-protected parameter:

1.5

Documentation

1.5.1

Standard documentation

Operating Instructions
The Operating Instructions are available via the Internet: www.endress.com →Download

1.5.2

Supplementary device-dependent documentation

Special Documentation
The Special Documentation is available via the Internet: www.endress.com →Download

Endress+Hauser

3

Overview of the operating menu

Micropilot FMRxxB; Operating tool HART

2

Overview of the operating menu

Navigation

 Operating tool→  14

Guidance‣ Heartbeat Technology→  28‣ Heartbeat Verification
Verification result→  29Operating time (Verification)→  29Date/time Heartbeat Verification→  28‣ Loop diagnostics→  30Baseline status→  30Tolerated deviation +/-→  30806 Alarm delay→  31Loop diagnostics→  31‣ Foam detection→  32952 Foam detected→  32

Foam index→  33

Foam detec. threshold→  33

Foam detec. threshold value→  33Lower level range limit→  32Upper level range limit→  32‣ Build-up detection4→  28→  34168 Build-up detected→  34

Build-up index→  34

Build-up detection threshold→  34

Build-up detection threshold value→  35Endress+Hauser

Micropilot FMRxxB; Operating tool HART→  17

Diagnostics‣ Active diagnostics

Endress+Hauser

Overview of the operating menu

→  17

Active diagnostics→  17

Timestamp→  17

Previous diagnostics→  18Timestamp→  18Operating time from restart→  18Operating time→  19‣ Diagnostic list

→  19‣ Event logbook→  19‣ Minimum/maximum values→  21

Min. level value→  22Time min. level→  22Max. draining speed→  22

Minimum sensor temperature→  23Time min. sensor temperature→  23

Minimum terminal voltage→  21Minimum electronics temperature→  22Max. level value→  23Time max. level→  24Max. filling speed→  24

Maximum sensor temperature→  24

Time max. sensor temperature→  24

Maximum terminal voltage→  23

Maximum electronics temperature→  23

5

Overview of the operating menu

Micropilot FMRxxB; Operating tool HART

‣ Simulation→  25
Simulation→  25

Value current output→  25

Simulation distance→  25

Build-up index→  26

Foam index→  26

Process variable value→  26

Diagnostic event simulation→  26‣ Echo curve→  20Save reference curve→  21Time reference curve→  21

Reference curve active→  20→  50Application‣ Measured values→  50Level linearized→  50Level→  50

Output current→  50Distance→  51

Unfiltered distance→  51

Terminal voltage 1→  51

Sensor temperature→  52

Electronics temperature→  51‣ Measuring Units→  52
Level unit6

→  52Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Overview of the operating menu

Distance unit

→  52Temperature unit

→  53‣ Sensor→  46‣ Basic settings→  53
Tank type→  53

Bin type→  54

Empty calibration→  54

Full calibration→  54Max. draining speed solid→  55Max. filling speed solid→  55Max. draining speed liquid→  56Max. filling speed liquid→  56

Tank/silo height→  57

Damping output→  57Distance→  57

Record map→  59Mapping end point→  58

Active map→  60‣ Additional settings

Endress+Hauser→  61

Medium type→  61

Medium group→  61

Medium property→  62Upper blank out→  62

Output mode→  63

Level limit mode→  637

Overview of the operating menu

Micropilot FMRxxB; Operating tool HART

High limit→  64

Low limit→  64

Level correction→  64‣ Linearization→  70Linearization type→  70

Unit after linearization→  71

Free text→  71

Level linearized→  71

Maximum value→  72

Diameter→  72

Intermediate height→  72

Table mode→  73

Table number→  73Level→  73Level→  74

Customer value→  74

Activate table→  74‣ Signal information
Signal quality→  75Absolute echo amplitude→  75

Relative echo amplitude→  75‣ Current output

8→  75→  77
Assign PV→  77

Measuring mode current output→  77

Current range output→  77

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Overview of the operating menu

Lower range value output→  78Upper range value output→  78

Failure behavior current output→  78

Failure current→  78Output current→  79Terminal current→  79‣ HART output→  81‣ Configuration→  80
HART address→  80

HART short tag→  80Device tag→  80No. of preambles→  81Loop current mode→  81‣ HART output→  81
Assign PV→  81Primary variable (PV)→  82

Assign SV→  82

Secondary variable (SV)→  82

Assign TV→  83

Tertiary variable (TV)→  83

Assign QV→  83

Quaternary variable (QV)

→  84‣ Burst configuration 1

Endress+Hauser→  84

Burst mode 1→  84

Burst command 1→  859

Overview of the operating menu

Micropilot FMRxxB; Operating tool HART

Burst variable 0→  85Burst variable 1→  86

Burst variable 2→  86

Burst variable 3→  87

Burst variable 4→  87

Burst variable 5→  88Burst variable 6→  89Burst variable 7→  89Burst trigger mode

→  90

Burst trigger level→  90Min. update period→  90Max. update period→  91‣ Information
Device ID→  91

Device type→  91

Device revision→  91

HART short tag→  92

HART revision→  92

HART descriptor→  92

HART message→  92

HART date code→  93→  94

System‣ Device management10→  91→  94

Device tag→  94

Locking status→  94Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Configuration counter→  95

Reset device→  95‣ User management→  95User role→  95

Password→  96Enter access code→  96

Status password entry→  96

New password→  97

Confirm new password→  97

Status password entry→  96

Old password→  97

New password→  97

Confirm new password→  97

Status password entry→  96

Reset password→  97

Status password entry→  96

Old password→  97

Status password entry→  96‣ Bluetooth configuration
Bluetooth activation‣ Display

Endress+Hauser

Overview of the operating menu

→  98→  98→  98

Language→  98

Format display→  99

Value 1 display→  99Decimal places 1→  100

11

Overview of the operating menu

Micropilot FMRxxB; Operating tool HART

Value 2 display→  100

Decimal places 2→  101

Value 3 display→  101

Decimal places 3→  101

Value 4 display→  102

Decimal places 4→  102

Contrast display→  102‣ Geolocation→  103
Process Unit Tag→  103

Location Description→  103

Longitude→  103

Latitude→  104

Altitude→  104

Location method→  104‣ Information12→  105
Device name→  105

Manufacturer→  105

Serial number→  105

Order code→  105

Firmware version→  106

Hardware version→  106

Extended order code 1→  106

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Extended order code 2→  106

Extended order code 3→  107‣ Software configuration

Endress+Hauser

Overview of the operating menu

→  113

CRC device configuration→  113

Stored CRC device configuration→  113

Timestamp stored CRC device config.

→  114

Activate SW option→  114

Software option overview→  11413

Description of device parameters

Micropilot FMRxxB; Operating tool HART

3

Description of device parameters

In the following section, the parameters are listed according to the menu structure of theoperating tool.
The operating menu is dynamic and adapts the choice of parameters to the selected
options.

3.1

Guidance

The Guidance main menu contains the functions that enable users to perform basic tasks
swiftly, e.g. commissioning. These are primarily guided wizards and cross-thematic special
functions.
Navigation

3.1.1

 Guidance

Overview of the operating menu

User navigation• Commissioning• Heartbeat Technology• Heartbeat Verification• Loop diagnostics• Foam detection• Buildup detection• Safety mode
• Proof testing• Import / Export• Compare

3.1.2

Commissioning

Run this wizard to put the device into operation. Enter the appropriate value in eachparameter or select the appropriate option.
If the wizard is canceled before all the necessary parameters have been configured,
any settings already made are saved. For this reason, the device may then be in an
undefined state!
In such situations, it is advisable to reset the device to the factory default settings.

14

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

The following parameters are configured in the Commissioning wizard:• Device identification• Device tag• HART short tag• HART date code
• HART descriptor• HART message• HART address• Measurement settings• Level unit
• Distance unit
• Temperature unit
• Tank type or vessel type
• Media group• Empty calibration• Full calibration• Mapping• Linearization• Output settings• Current range output• Failure behavior current output• Assignment HART variables

3.1.3

Heartbeat Technology

Heartbeat Technology offers diagnostic functionality through continuous self-monitoring,
the transmission of additional measured variables to an external Condition Monitoring
system and the in-situ verification of measuring devices in the application.
Special Documentation "Heartbeat Monitoring + Verification"
SD02953F
Heartbeat Verification
This wizard is used to start automatic verification of the device functionality. The results
can be documented in a verification report.
Loop diagnostics
Using this wizard, changes in the current-voltage loop characteristics (baseline) of thesignal circuit can be used to detect unwanted installation anomalies such as creep currents
caused by corrosion of the terminals or a deteriorating power supply that can lead to anincorrect current output measured value.
Foam detection
This wizard configures automatic foam detection.
Foam detection can be linked to an output variable or status information e.g. to control asprinkler used to dissolve the foam. It is also possible to monitor the foam growth in what
is known as a "foam index". The foam index can also be linked to an output variable andshown on the display.
Preparation: Foam monitoring initialization should only be performed with little or no
foam
Buildup detection
This wizard configures buildup detection.
Buildup detection can, for example, be linked to a compressed air system to clean theantenna. The maintenance cycles can be optimized with buildup monitoring.
Endress+Hauser

15Description of device parameters

Micropilot FMRxxB; Operating tool HART

Preparation: Buildup monitoring initialization should only be performed with little or nobuildup.

3.1.4

Safety mode

Write protection protects device settings from being overwritten. In addition, for safety
applications it is also recommended to confirm the device settings that are relevant forsafety (safety-related settings). This ensures that the correct values have been entered anddownloaded to the device.
This entry can be used instead of manual checklists as the confirmation sequence.
If the safety-related device settings are confirmed, the device is marked with the "Safetylocked" feature. This indicates that the safety-related parameter settings have beenchecked and judged to be correct.
The sequence must be restarted to deactivate the safety lock. The safety lock is disabled if
the unlocking code (= locking code) is entered.

3.1.5

Proof testing

A proof test simulates the current output. Safe operation during the test is not guaranteed.
Manual, alternative measures must be taken to ensure process safety.
Caution: A proof test can only be performed if there is no active device error and if write
protection has not been enabled via the hardware write protection switch.

3.1.6

Import/export

Save/load• Save: The device settings can be saved in a .deh file.
• Load: The device settings saved in a .deh file can be written to the device.
Create documentation• Device documentation can be saved in PDF format under "Create documentation".
• This documentation contains the following general device information:• Information on device parameters• Information on linearization• Echo curves• Event list
• Diagnostic list

3.1.7

CompareCompare datasets
This function can be used to compare the following datasets:• Datasets in the .deh file format from the import/export function (Section 3.1.6)• Datasets with the configuration currently in the device

16

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

3.2
Navigation

3.2.1
Navigation

Description of device parameters

Diagnostics
 Diagnostics

Active diagnostics
 Diagnostics → Active diagnos.

Active diagnostics
Navigation

 Diagnostics → Active diagnos. → Active diagnos.

Description

Displays the currently active diagnostic message.
If there is more than one pending diagnostic event, the message for the diagnostic eventwith the highest priority is displayed.
The information displayed comprises:• Symbol for event level• Code for diagnostic behavior• Operation time of occurrence• Event text

User interface
Factory setting
Additional information

Information on what is causing the message, and remedy measures, can be viewed via
the I-symbol on the display.
Access:• Read access: Operator• Write access: -Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp for the currently active diagnostic message.

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting
Additional information

Endress+Hauser

Access:• Read access: Operator• Write access: -17

Description of device parameters

Micropilot FMRxxB; Operating tool HART

Previous diagnostics
Navigation

 Diagnostics → Active diagnos. → Prev.diagnostics

Description

Displays the diagnostic message for the last diagnostic event that has ended.

User interface
Factory setting
Additional information

The information displayed comprises:• Symbol for event level• Code for diagnostic behavior• Operation time of occurrence• Event text
The diagnostic message displayed can still be valid.
Information on what is causing the message, and remedy measures, can be viewed via theI-symbol on the display.
Access:• Read access: Operator• Write access: -Timestamp
Navigation

 Diagnostics → Active diagnos. → Timestamp

Description

Displays the timestamp of the diagnostic message generated for the last diagnostic eventthat has ended.

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting
Additional information

Access:• Read access: Operator• Write access: -Operating time from restart
Navigation

 Diagnostics → Active diagnos. → Time fr. restart

Description

Indicates how long the device has been in operation since the last time the device wasrestarted.

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting
18

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Additional information

Description of device parameters

Access:• Read access: Operator• Write access: -Operating time
Navigation

 Diagnostics → Active diagnos. → Operating time

Description

Indicates how long the device has been in operation.

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting
Additional information

Access:• Read access: Operator• Write access: -3.2.2

Diagnostic list

Navigation

3.2.3

 Diagnostics → Diagnostic list

Event logbook

Navigation

 Diagnostics → Event logbook

Filter options
Navigation

Diagnostics → Event logbook → Filter options

Description

Select the category of event notification to display in the event list.
Additional information:
The status signals F, C, S and M are categorized in accordance with VDI/VDE 2650 and
NAMUR Recommendation NE 107.

Selection• All• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• Information (I)• Not categorized

Factory setting

All

Endress+Hauser

19Description of device parameters

Additional information

Micropilot FMRxxB; Operating tool HART

Access:• Read access: Operator• Write access: Maintenance

Clear event listNavigation

Diagnostics → Event logbook → Clear event list

Description

Delete all entries of the event list.

Selection• Cancel• Clear data

Factory setting

3.2.4
Navigation

Echo curve Diagnostics → Echo curve

Reference curve active
Navigation

 Diagnostics → Echo curve → Ref.curve active

User interface• Delivery reference curve available• Customer reference curve available

Factory setting
Additional information

The delivery reference curve is recorded at the factory before the device is delivered.
A customer reference curve is usually recorded at the end of Guidance → Commissioning.
Should troubleshooting be necessary, these reference curves can be used for diagnostic
purposes.
Access:• Read access: Operator• Write access: -20

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Save reference curveNavigation

 Diagnostics → Echo curve → Save ref. curve

Selection• Customer reference curve• Not active

Factory setting
Additional information

Access:• Read access: Operator• Write access: MaintenanceTime reference curve
Navigation

 Diagnostics → Echo curve → Time ref. curve

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting
Additional information

Time the reference curve is recorded
Access:• Read access: Operator• Write access: -3.2.5

Minimum/maximum values

Navigation

 Diagnostics → Min/max val.

Minimum terminal voltage
Navigation

 Diagnostics → Min/max val. → Min.term.volt.

Description

Minimum or maximum measured terminal (supply) voltage.

User interface

0.0 to 50.0 V

Factory setting
Additional information

Endress+Hauser

Access:• Read access: Operator• Write access: 21

Description of device parameters

Micropilot FMRxxB; Operating tool HART

Minimum electronics temperature
Navigation

 Diagnostics → Min/max val. → Min.electr.temp.

Description

Minimum or maximum measured main electronics temperature.

User interface

Signed floating-point number

Factory setting
Additional information

Access:• Read access: Operator• Write access: -Min. level value
Navigation

 Diagnostics → Min/max val. → Min. level value

User interface

Signed floating-point number

Factory setting

Time min. level
Navigation

 Diagnostics → Min/max val. → Time min. level

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting

Max. draining speed
Navigation

 Diagnostics → Min/max val. → Max.drain.speed

User interface

Positive floating-point number

Factory setting

22

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Minimum sensor temperature
Navigation

 Diagnostics → Min/max val. → Min. sensor temp

User interface–150 to 200 °C

Factory setting

Time min. sensor temperature
Navigation

 Diagnostics → Min/max val. → Time min s. temp

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting

Maximum terminal voltage
Navigation

 Diagnostics → Min/max val. → Max.term.voltage

Description

Minimum or maximum measured terminal (supply) voltage.

User interface

0.0 to 50.0 V

Factory setting

Maximum electronics temperature
Navigation

 Diagnostics → Min/max val. → Max.electr.temp.

Description

Minimum or maximum measured main electronics temperature.

User interface

Signed floating-point number

Factory setting

Max. level value
Navigation

 Diagnostics → Min/max val. → Max. level value

User interface

Signed floating-point number

Endress+Hauser

23Description of device parameters

Micropilot FMRxxB; Operating tool HART

Factory setting

Time max. level
Navigation

 Diagnostics → Min/max val. → Time max. level

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting

Max. filling speed
Navigation

 Diagnostics → Min/max val. → Max. fill. speed

User interface

Positive floating-point number

Factory setting

Maximum sensor temperature
Navigation

 Diagnostics → Min/max val. → Max. Sensor temp

User interface–150 to 200 °C

Factory setting

Time max. sensor temperature
Navigation

 Diagnostics → Min/max val. → Time max s. temp

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting

24

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

3.2.6

Description of device parameters

Simulation

Navigation

 Diagnostics → Simulation

Simulation



Navigation

 Diagnostics → Simulation → Simulation

Prerequisite

Options marked with *:The corresponding device function must be available and configured.

Description

Simulates one or more process variables and/or events.
Warning:- Output will reflect the simulated value or event.

Selection• Off• Distance• Level• Level linearized *• Current output• Diagnostic event simulation• Foam index *• Build-up index *Factory setting

Off

Value current output

Navigation

 Diagnostics → Simulation → Val. curr.outp

Prerequisite

Simulation = Current output (→  77)

Description

Defines the value of the simulated output current.

User entry

3.59 to 23 mA

Factory setting

Simulation distanceNavigation

 Diagnostics → Simulation → Sim distancePrerequisite

Simulation = Distance (→  51)*

Visibility depends on order options or device settings

Endress+Hauser

25Description of device parameters

User entry

Micropilot FMRxxB; Operating tool HART

–999 900 to 999 900 mm

Factory setting

Build-up index



Navigation

 Diagnostics → Simulation → Build-up index

Prerequisite

Simulation = Build-up index

User entry

0 to 100.0 %Factory setting

Foam index



Navigation

 Diagnostics → Simulation → Foam index

Prerequisite

Simulation = Foam index (→  33)

User entry

0 to 100.0 %Factory setting

Process variable value

Navigation

 Diagnostics → Simulation → Proc. var. value

Prerequisite

Simulation = Level linearized (→  50)

Description

Defines the value of the selected variable.
The outputs assume values or states according to this value.

User entry

Signed floating-point number

Factory setting

Diagnostic event simulation



Navigation

 Diagnostics → Simulation → Diag. event sim.

Prerequisite

Simulation = Diagnostic event simulation

26

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description

Description of device parameters

Select the diagnostic event to be simulated.
Note:
To terminate the simulation, select "Off".

Selection

F062 Sensor connection faulty
F151 Sensor electronic failureM168 Build-up detected
S203 HART Device Malfunction
F204 HART Electronic DefectF242 Firmware incompatible
F252 Module incompatible
F270 Main electronics defectiveF272 Main electronics faulty
F273 Main electronics defectiveF282 Data storage inconsistentF283 Memory content inconsistentM287 Memory content inconsistentF388 Electronics and HistoROM defective
F410 Data transfer failed
C412 Processing download
S420 HART Device Configuration Locked
S421 HART Loop Current fixedC431 Trim required
F435 Linearization faulty
F437 Configuration incompatible
M438 Dataset different
S441 Current output out of range
C484 Failure mode simulation active
C485 Process variable simulation active
C491 Current output simulation active
S495 Diagnostic event simulation activeF538 Configuration Sensor Unit invalid
C585 Simulation distanceC586 Record map
F801 Supply voltage too low
S802 Supply voltage too highF805 Loop current faulty
M806 Loop diagnostics
M807 No Baseline due to insuf. volt. at 20 mA
S825 Electronics temperature
S826 Sensor temperature out of range
S846 HART Non-Primary Variable Out of Limit
S847 HART Primary Variable Out of Limit

Endress+Hauser

27Description of device parameters

Micropilot FMRxxB; Operating tool HART

S848 HART Device Variable Alert
S941 Echo lost
S942 In safety distanceC952 Foam detected
S968 Level limited
Factory setting

Off

3.2.7

Heartbeat Technology

Navigation

 Diagnostics → Heartbeat Techn.

Navigation

Guidance → Heartbeat Techn.

Heartbeat Verification
Navigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif.

Navigation

Guidance → Heartbeat Techn. → Heartbeat Verif.

Date/time Heartbeat Verification
Navigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Date/time Heartbeat
VerificationDescription

Guidance → Heartbeat Techn. → Heartbeat Verif. → Date/time Heartbeat
Verification

Date and time of last Hearbeat Verification.
This value is updated with every Heartbeat verification.
Note:
If time information is not available, e.g. Heartbeat verification is started from display,'--------' is shown.

User interface

Character string comprising numbers, letters and special characters

Factory setting

28

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Operating time (Verification)
Navigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Operating time

Guidance → Heartbeat Techn. → Heartbeat Verif. → Operating time

Description

Value of the operating time counter at the time of verification

User interface

Days (d), hours (h), minutes (m), seconds (s)Factory setting

Verification result
Navigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Verific. resultGuidance → Heartbeat Techn. → Heartbeat Verif. → Verific. result

Description

Result of Heartbeat Verification

User interface• Not done
• Passed• Failed

Factory setting

StatusNavigation

 Diagnostics → Heartbeat Techn. → Heartbeat Verif. → Status

Description

Shows the actual status.

User interface• Done• Busy• Failed• Not done

Factory setting

Endress+Hauser

29Description of device parameters

Micropilot FMRxxB; Operating tool HART

Loop diagnostics
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn.

Rebuild baselineNavigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Reb. baseline

Description

Notice
The current output is simulated.
Bridge the PLC or take other appropriate measures to prevent an erroneous triggering ofalarm messages or changes in the control loop behavior.
The baseline should be rebuilt if planned changes have been made in the loop.

Selection

• No
• Yes

Factory settingNo

Tolerated deviation +/-Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Toler. deviation

Description

A value should be chosen to ensure that normal voltage deviations do not lead tounwanted messages.
Default
1.5 V DC

User entry

0.5 to 3.0 V

Factory setting

1.5 V

Baseline status
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Baseline status

Description'Failed'Means, baseline is not available or creation not possible.
'Success'
Baseline is available.

User interface• Failed• SuccessFactory setting

Failed30

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Loop diagnostics



Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Loop diagn.

Selection• Disable• Enable

Factory setting

DisableTerminal voltage 1
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Terminal volt. 1

Description

Shows the current terminal voltage that is applied at the output

User interface

0.0 to 50.0 V

Factory setting

0V

Clamping voltage lower threshold
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Lower threshold

User interface

0.0 to 50.0 V

Factory setting

0V

Clamping voltage upper threshold
Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → Upper threshold

User interface

0.0 to 50.0 V

Factory setting

0V

806 Alarm delay

Navigation

 Diagnostics → Heartbeat Techn. → Loop diagn. → 806 Alarm delay

User entry

0 to 60 s

Endress+Hauser

31Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

1s

Foam detection
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection

Navigation

Guidance → Heartbeat Techn. → Foam detection

952 Foam detected
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → 952 Foam detected

Selection

• Off
• On

Factory setting

Off

Guidance → Heartbeat Techn. → Foam detection → 952 Foam detected

Lower level range limit
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → LLR limit

Guidance → Heartbeat Techn. → Foam detection → LLR limit

Description

Assign lower limit of foam monitoring area.

Factory setting0%Upper level range limit
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → ULR limit

Guidance → Heartbeat Techn. → Foam detection → ULR limit

Description

Assign upper limit of foam monitoring area.

Factory setting

100.0 %

32

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Foam index
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → Foam index


Guidance → Heartbeat Techn. → Foam detection → Foam index

Description

Foam index 0% means: no foam.
Foam index 100% means: maximum detectable foam.

User interface

0 to 100 %Factory setting

0%

Foam detec. threshold
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → Foam threshold

Guidance → Heartbeat Techn. → Foam detection → Foam threshold

Description

Enter the threshold for the foam detection. As soon as the foam index has reached thepreset switching point, an event is triggered.

Selection• Sensitive (20%)• Middle (40%)• Insensitive (80%)• User defined (xx%)

Factory setting

Middle (40%)

Foam detec. threshold value
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → Foam detect val.
Guidance → Heartbeat Techn. → Foam detection → Foam detect val.

Description

User-defined threshold value for the foam detection.

User entry

0 to 100.0 %Factory setting

40 %Distance at foam zero adjustment
Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → Dist. @zero foam

User entry

Signed floating-point number

Endress+Hauser33Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

0 mm

0% foam value

Navigation

 Diagnostics → Heartbeat Techn. → Foam detection → 0% foam value

User entry–999 999.9 to 999 999.9 dB

Factory setting

0 dB

Buildup detection
Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect.

168 Build-up detected

Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → 168 Build-up det.

Selection

• Off
• On

Factory setting

Off

Build-up index
Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → Build-up index

Description

Build-up index 0% means: no build-up.
Build-up index 100% means: maximum detectable build-up.

User interface

0 to 100 %Factory setting0%

Build-up detection threshold

Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → Build-up detec.

Description

Enter the threshold for the build-up detection. As soon as the build-up index has reached
the preset switching point, an event is triggered.

34

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Selection• Sensitive (20%)• Middle (40%)• Insensitive (80%)• User defined (xx%)

Factory setting

Middle (40%)

Description of device parameters

Build-up detection threshold value
Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → Build-up value

Description

User-defined threshold value for the build-up detection.

User entry

0 to 100.0 %Factory setting

40 %0% build-up value

Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → 0% build-up val

User entry

Positive floating-point number

Factory setting0

Limit offset for build-up detection
Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → Offset build-up

User entry–999 999.9 to 999 999.9 dB

Factory setting

10 dB

Minimal distance for build-up detection
Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → Min dist buildup

User entry–999 900 to 999 900 mm

Factory setting

0 mm

Endress+Hauser







35

Description of device parameters

Micropilot FMRxxB; Operating tool HART

Maximum distance for build-up detection



Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → Max dist buildup

User entry–999 900 to 999 900 mm

Factory setting

1 000 mm

3.2.8

Self-test

Navigation

 Diagnostics → Self test

Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect.

Start self checkNavigation

 Diagnostics → Self test → Start self check

Selection

• No
• Yes

Factory setting

No

Result self check
Navigation

 Diagnostics → Self test → Result selfcheck

Description

A test signal is generated in the sensor module and fed into the analog path of the device.
• Verification Test echo in the time window.
• Verification of the test echo amplitude.

User interface• Unknown• None• Not used
• Valid range• Not ok
• Ok
• Invalid range• Not defined• Not done
• Passed• Done• Failed• In progress• Finished

36

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Factory setting

Description of device parameters

Unknown

Start device checkNavigation

 Diagnostics → Self test → Start dev. check

Selection

• No
• Yes

Factory setting

No

Result device check
Navigation

 Diagnostics → Self test → Result dev.check

Description

The device has a function that detects and reports any interference by unfavorable
installation situation. Here, the amplitudes of measured signals are monitored, which referto interference in the near range.

User interface• Installation ok• Accuracy reduced• Measurement capability reduced• Check not done

Factory setting

Check not done

Last check time
Navigation

 Diagnostics → Self test → Last check time

User interface

Character string comprising numbers, letters and special characters

Factory setting

Near distanceNavigation

 Diagnostics → Self test → Near distanceSelection• Check not done
• Check not OK• Check OK

Factory setting

Check not done

Endress+Hauser

37Description of device parameters

Micropilot FMRxxB; Operating tool HART

Level signal
Navigation

 Diagnostics → Self test → Level signal

User interface• Check not done
• Check not OK• Check OK

Factory setting

Check not done

Area of incoupling
Navigation

 Diagnostics → Heartbeat Techn. → Build-up detect. → Area incoupling

Description

Ringing integral within the detection area.

User interface

Positive floating-point number

Factory setting

0.0

3.2.9

Diagnostic settings

Navigation

 Diagnostics → Self test

Properties
Navigation

Diagnostics → Diag. settings → Properties‣ Properties

38

806 Alarm delay→  39941 Diagnostic behavior→  40941 Diagnostic behavior→  40941 Event category→  40941 Event category→  40

Value echo lost→  40Ramp at echo lost→  41

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Delay time echo lost→  41942 Diagnostic behavior→  42942 Diagnostic behavior→  42942 Event category→  43942 Event category→  43

Safety distance→  43

Acknowledge alarm→  43

Alarm delay

Navigation

 Diagnostics → Diag. settings → Properties → Alarm delay

User entry

0 to 60 s

Factory setting

0s

806 Alarm delay

Navigation

 Diagnostics → Diag. settings → Properties → 806 Alarm delay

User entry

0 to 60 s

Factory setting

1s

Endress+Hauser

39Description of device parameters

Micropilot FMRxxB; Operating tool HART

941 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Properties → 941 Diag. behav.

Description

Defines the behavior of the output in case of an echo loss.
'Last valid value'
Last valid value is kept.
'Ramp at echo lost'
Output value is continuously shifted towards 0% or 100%.
'Value echo lost'
Output assumes a defined value.
'Alarm'Device generates an alarm.

Selection• Last valid value• Ramp at echo lost *• Value echo lost *• Alarm

Factory setting

Last valid value

941 Event category



Navigation

 Diagnostics → Diag. settings → Properties → 941Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Value echo lostNavigation

 Diagnostics → Diag. settings → Properties → Value echo lost

Description

Value of the output in case of an echo loss.

User entry

0 to 200 000.0 %Factory setting

0.0 %

*

40

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Ramp at echo lostNavigation

 Diagnostics → Diag. settings → Properties → Ramp echo lost

Description

Slope of the ramp in the case of an echo loss.
Note:
If the slope is positive (+), the output increases until it reaches 100%.
If the slope is negative (-), the output decreases until it reaches 0%.

User entry

Signed floating-point number

Factory setting

0.0 %/min

Jump delay echo lostNavigation

 Diagnostics → Diag. settings → Properties → Jump del. echol.

Selection

• Off
• On

Factory setting

On

Delay time echo lostNavigation

 Diagnostics → Diag. settings → Properties → Delay echo lost

Description

Time between the echo loss and the reaction defined for the output.

User entry

0 to 99 999.9 s

Factory setting

900 s

Echo lost window right

Navigation

 Diagnostics → Diag. settings → Properties → Echo l.win.rightUser entry

0 to 99 900 mm

Factory setting

4 000 mmEndress+Hauser

41Description of device parameters

Micropilot FMRxxB; Operating tool HART

Echo lost window left

Navigation

 Diagnostics → Diag. settings → Properties → Echo l.win.left

User entry

0 to 99 000 mm

Factory setting

4 000 mm

Draining speedNavigation

 Diagnostics → Diag. settings → Properties → Draining speed

User entry

Signed floating-point number

Factory setting

100 cm/min

Filling speedNavigation

 Diagnostics → Diag. settings → Properties → Filling speed

User entry

Signed floating-point number

Factory setting

100 cm/min

942 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Properties → 942 Diag. behav.

Selection• Off• Alarm• Warning• Self holding

Factory setting

Warning

42

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

942 Event category



Navigation

 Diagnostics → Diag. settings → Properties → 942Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Safety distanceNavigation

 Diagnostics → Diag. settings → Properties → Safety distanceUser entry–200 000 to 125 000 mm

Factory setting

0 mm

Acknowledge alarmNavigation

 Diagnostics → Diag. settings → Properties → Acknowl. alarm

Selection

• No
• Yes

Factory setting

No

Diagnostic settings
Navigation

 Diagnostics → Self test

‣ Diagnostic settings‣ Properties

Endress+Hauser→  38806 Alarm delay→  39941 Diagnostic behavior→  40941 Diagnostic behavior→  4043

Description of device parameters

Micropilot FMRxxB; Operating tool HART

941 Event category→  40941 Event category→  40

Value echo lost→  40Ramp at echo lost→  41

Delay time echo lost→  41942 Diagnostic behavior→  42942 Diagnostic behavior→  42942 Event category→  43942 Event category→  43

Safety distance→  43

Acknowledge alarm→  43‣ Configuration→  80‣ Sensor→  46168 Diagnostic behavior→  46168 Diagnostic behavior→  46168 Diagnostic behavior→  46168 Diagnostic behavior→  46168 Event category→  46168 Event category→  46168 Event category→  46168 Event category→  46‣ Process44→  47806 Diagnostic behavior→  47806 Diagnostic behavior→  47806 Diagnostic behavior→  47

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Endress+Hauser

Description of device parameters

806 Diagnostic behavior→  47806 Diagnostic behavior→  47806 Diagnostic behavior→  47806 Event category→  47806 Event category→  47806 Event category→  47806 Event category→  47806 Event category→  47806 Event category→  47941 Diagnostic behavior→  48941 Diagnostic behavior→  48941 Diagnostic behavior→  48941 Diagnostic behavior→  48941 Event category→  48941 Event category→  48941 Event category→  48941 Event category→  48942 Diagnostic behavior→  48942 Diagnostic behavior→  48942 Diagnostic behavior→  48942 Diagnostic behavior→  48942 Event category→  49942 Event category→  49942 Event category→  49942 Event category→  4945

Description of device parameters

Micropilot FMRxxB; Operating tool HART

952 Diagnostic behavior→  49952 Diagnostic behavior→  49952 Diagnostic behavior→  49952 Diagnostic behavior→  49952 Event category→  49952 Event category→  49952 Event category→  49952 Event category→  49SensorNavigation

 Diagnostics → Diag. settings → Configuration → Sensor

168 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Sensor → 168 Diag. behav.

Description

Select event behavior'Logbook entry only':
no digital or analog transmission of the message'Warning': Current output unchanged. Message is output digitally (default).
'Alarm': Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

168 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Sensor → 168Event category

Description

Select category for diagnostic message.

46

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Maintenance required (M)Process
Navigation

 Diagnostics → Diag. settings → Configuration → Process

806 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 806 Diag. behav.

Description

Select event behavior'Logbook entry only': no digital or analog transmission of the message.
'Warning': Current output unchanged. Message is output digitally (default).
If the permissible conditions are reached again, the warning is no longer available in the
instrument.

Selection• Warning• Logbook entry only

Factory setting

Warning

806 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 806Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Maintenance required (M)Endress+Hauser

47

Description of device parameters

Micropilot FMRxxB; Operating tool HART

941 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 941 Diag. behav.

Description

Defines the behavior of the output in case of an echo loss.
'Last valid value'
Last valid value is kept.
'Ramp at echo lost'
Output value is continuously shifted towards 0% or 100%.
'Value echo lost'
Output assumes a defined value.
'Alarm'Device generates an alarm.

Selection• Last valid value• Ramp at echo lost *• Value echo lost *• Alarm

Factory setting

Last valid value

941 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 941Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)942 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 942 Diag. behav.

Selection• Off• Alarm• Warning• Self holding

Factory setting

Warning*

48

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

942 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 942Event category

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)952 Diagnostic behavior

Navigation

 Diagnostics → Diag. settings → Configuration → Process → 952 Diag. behav.

Description

Select event behavior'Logbook entry only':
no digital or analog transmission of the message'Warning': Current output unchanged. Message is output digitally (default).
'Alarm': Current output assumes the set alarm current.
Regardless of the setting, the message appears on the display. If the permissible conditions
are reached again, the warning is no longer available in the instrument.

Selection• Off• Alarm• Warning• Logbook entry only

Factory setting

Warning

952 Event category



Navigation

 Diagnostics → Diag. settings → Configuration → Process → 952Event category

Description

Select category for diagnostic message.

Selection• Failure (F)• Function check (C)• Out of specification (S)• Maintenance required (M)• No effect (N)Factory setting

Out of specification (S)Endress+Hauser

49

Description of device parameters

3.3

Micropilot FMRxxB; Operating tool HART

Application

Navigation

3.3.1

 Application

Measured values

Navigation

 Application → Measured values

Level linearized
Navigation

 Application → Measured values → Level linearized

Description

Displays the linearized level.

User interface

Signed floating-point number

Factory setting

0%

Level
Navigation

 Application → Measured values → Level

Description

Currently measured level

User interface–99 999.9 to 200 000.0 %Factory setting

0.0 %Output current
Navigation

 Application → Measured values → Output curr.

Description

Shows the value currently calculated for the current output

User interface

3.59 to 23 mA

Factory setting

3.59 mA

50

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Distance
Navigation

 Application → Measured values → DistanceDescription

Distance from lower edge of device flange to product surface.

User interface

Signed floating-point number

Factory setting

0 mm

Unfiltered distanceNavigation

 Application → Measured values → Unfiltered dist.

User interface

Signed floating-point number

Factory setting

0 mm

Terminal voltage 1
Navigation

 Application → Measured values → Terminal volt. 1

Description

Shows the current terminal voltage that is applied at the output

User interface

0.0 to 50.0 V

Factory setting

0V

Electronics temperature
Navigation

 Application → Measured values → Electronics temp

Description

Displays the current temperature of the main electronics.

User interface

Signed floating-point number

Factory setting

0 °C

Endress+Hauser

51

Description of device parameters

Micropilot FMRxxB; Operating tool HART

Sensor temperature
Navigation

 Application → Measured values → Sensor temp.

Description

Displays the current temperature of the sensor electronics.

User interface–150 to 200 °C

Factory setting–150 °C

3.3.2
Navigation

Measuring units Application → Measuring Units

Level unitNavigation

 Application → Measuring Units → Level unit

Description

Used to display the level.

Selection

SI units
•%
•m
• mm

Factory setting%

US units• ft• inDistance unitNavigation

 Application → Measuring Units → Distance unit

Description

Used for the basic calibration (Empty / Full).

Selection

SI units• mm•m

Factory settingmm52

US units• ft• inEndress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Temperature unitNavigation

 Application → Measuring Units → Temperature unit

Description

Used to display the electronics temperature.

Selection

SI units• °C•KFactory setting

°C

3.3.3

US units°F

SensorNavigation

 Application → Measuring Units

Basic settings
Navigation

 Application → Sensor → Basic settings

Tank type

Navigation

 Application → Sensor → Basic settings → Tank type

Description

Optimizes the signal filters for the respective tank type.
Note:'Workbench test' deactivates all filters. This option should exclusively be used for tests.

Selection• Process vessel standard• Process vessel with agitator• Storage vessel• Sphere• Open channel• Stilling well
• Workbench test *Factory setting

Process vessel standard*

Visibility depends on order options or device settings

Endress+Hauser

53Description of device parameters

Micropilot FMRxxB; Operating tool HART

Bin type

Navigation

 Application → Sensor → Basic settings → Bin type

Description

Optimizes the signal filters for the respective bin type.
Note:'Workbench test' deactivates all filters. This option should exclusively be used for tests.

Selection• Silo• Buffer silo (fast) *• Bin / Pile *• Crusher / belt *• Workbench test *Factory setting

Silo

Empty calibration



Navigation

 Application → Sensor → Basic settings → Empty calibr.

Description

Distance between process connection and minimum level (0%).

User entry

0 to 125 000 mm

Factory setting

20 000 mm

Full calibration



Navigation

 Application → Sensor → Basic settings → Full calibr.

Description

Distance between minimum level (0%) and maximum level (100%).

User entry

1 to 125 000 mm

Factory setting

20 000 mm*

54

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Max. draining speed solidNavigation

 Application → Sensor → Basic settings → Max.drain solid

Description

By selecting the maximum expected filling and draining speed the signal evaluation is
automatically optimized for the process.
Note:
The filling and draining speeds can be set separately as the filling and draining procedures
may be different.
Note:With the ‘No filter / test’ option all signal evaluation filters are deactivated. This option
should exclusively be used for tests.

Selection• Very slow < 0.5m (1.6ft) /h• Slow < 1m (3.3ft) /h• Standard < 2m (6,5ft) /h• Medium < 4m (13ft) /h• Fast < 8m (26ft) /h• Very fast > 8m (26ft) /h• No filter / test

Factory setting

No filter / test

Max. filling speed solidNavigation

 Application → Sensor → Basic settings → Max. fill. solid

Description

By selecting the maximum expected filling and draining speed the signal evaluation is
automatically optimized for the process.
Note:
The filling and draining speeds can be set separately as the filling and draining procedures
may be different.
Note:With the ‘No filter / test’ option all signal evaluation filters are deactivated. This option
should exclusively be used for tests.

Selection• Very slow < 0.5m (1.6ft) /h• Slow < 1m (3.3ft) /h• Standard < 2m (6,5ft) /h• Medium < 4m (13ft) /h• Fast < 8m (26ft) /h• Very fast > 8m (26ft) /h• No filter / test

Factory setting

No filter / test

Endress+Hauser

55Description of device parameters

Micropilot FMRxxB; Operating tool HART

Max. draining speed liquidNavigation

 Application → Sensor → Basic settings → Max drain liquid

Description

By selecting the maximum expected filling and draining speed the signal evaluation is
automatically optimized for the process.
Note:
The filling and draining speeds can be set separately as the filling and draining procedures
may be different.
Note:With the ‘No filter / test’ option all signal evaluation filters are deactivated. This option
should exclusively be used for tests.

Selection• Slow < 1cm (0.4in) /min
• Medium < 10cm (4in) /min
• Standard < 1m (40in) /min
• Fast < 2m (80in) /min
• Very fast > 2m (80in) /min
• No filter / test

Factory settingStandard < 1m (40in) /minMax. filling speed liquidNavigation

 Application → Sensor → Basic settings → Max. fill liquid

Description

By selecting the maximum expected filling and draining speed the signal evaluation is
automatically optimized for the process.
Note:
The filling and draining speeds can be set separately as the filling and draining procedures
may be different.
Note:With the ‘No filter / test’ option all signal evaluation filters are deactivated. This option
should exclusively be used for tests.

Selection• Slow < 1cm (0.4in) /min
• Medium < 10cm (4in) /min
• Standard < 1m (40in) /min
• Fast < 2m (80in) /min
• Very fast > 2m (80in) /min
• No filter / test

Factory settingStandard < 1m (40in) /min

56

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Tank/silo height

Navigation

 Application → Sensor → Basic settings → Tank/silo height

Description

If the parametrized measuring range (Empty calibration) differs significantly from thetank or silo height, it is recommended to enter the tank or silo height in this parameter.
Example:Continuous level monitoring in the upper third of a tank or silo.
Note:
For tanks with conical outlet, this parameter should not be changed as in this type of
applications 'Empty calibration' is usually not << the tank or silo height.

User entry

0 to 125 000 mm

Factory setting

20 000 mm

Damping output

Navigation

 Application → Sensor → Basic settings → Damping out.

Description

Reaction time of the output signal on fluctuation in the measured value.

User entry

0.0 to 1 200.0 s

Factory setting

0.0 sDistance
Navigation

 Application → Sensor → Basic settings → DistanceDescription

Distance from lower edge of device flange to product surface.

User interface

Signed floating-point number

Factory setting

0 mm

Endress+Hauser

57Description of device parameters

Micropilot FMRxxB; Operating tool HART

Confirm distanceNavigation

 Application → Sensor → Basic settings → Confirm distanceDescription

Specify, whether the measured distance matches the real distance. Depending on theselection the device automatically sets the range of mapping.
Note:
If the teaching procedure with 'Distance too small' or 'Distance too big' is quit before thedistance has been confirmed, a map is not recorded and the teaching procedure is reset
after 60 s.

Selection• Manual map• Distance ok• Distance unknown• Tank empty

Factory setting

Distance unknown

Mapping start pointNavigation

 Application → Sensor → Basic settings → Map. start point

User entry–999 900 to 999 900 mm

Factory setting–250 mm

Mapping end pointNavigation

 Application → Sensor → Basic settings → Map. end point

Description

Defines up to which distance the new mapping has to be recorded.
Remark: Make sure the level signal is not covered by the mapping.

User entry

0.1 to 125 mm

Factory setting

100 mm

Map gap

Navigation

 Application → Sensor → Basic settings → Map gap

User entry

0 to 100 000 mm

Factory setting

235 mm58

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Mapping overlay timeNavigation

 Application → Sensor → Basic settings → Map overlay time

User entry

0 to 1 200 s

Factory setting

5s

End of mappingNavigation

 Application → Sensor → Basic settings → End of mapping

Description

Defines the behavior of the mapping curve in the tank bottom area.

Selection• Adjustable• Last map value

Factory setting

Adjustable

End map. ampl.

Navigation

 Application → Sensor → Basic settings → End map. ampl.

Description

Amplitude of the mapping curve in the tank bottom area.

User entry–99 999.0 to 99 999.0 dB

Factory setting–100 dB

Record map

Navigation

 Application → Sensor → Basic settings → Record map

Selection• No• Overlay map• Delete cust map

Factory setting

No

Endress+Hauser

59Description of device parameters

Micropilot FMRxxB; Operating tool HART

MappingRecordingStatus
Navigation

 Application → Sensor → Basic settings → MapRecordingStat

User interface• Not active• In progress• Not ok
• Ok
• Not changed• Recording customer map• Recording factory map• Store• Ready for recalculation• Distances Recalculated

Factory setting

Not active

MappingRecordedCtrl (MappingStatus)
Navigation

 Application → Sensor → Basic settings → MapRecordedCtrl

User interface• Invalid range• Valid range• Not active• Passed• Executing• Store• Not ok
• Ok

Factory setting

Not active

Active mapNavigation

 Application → Sensor → Basic settings → Active map

Description

Select the mapping curve that is to be active. Alternatively, the option 'No map' can beselected.

Selection• Factory map• Customer map• No map

Factory setting

No map60

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

MappingSelectionStatus
Navigation

 Application → Sensor → Basic settings → MapSelectionStat

User interface• Factory map• Customer Map• No map• Factory map• Customer Map• No map

Factory setting

No mapAdditional settings
Navigation

 Application → Sensor → Add. settings

Medium type

Navigation

 Application → Sensor → Add. settings → Medium type

Selection• Liquid• Solid

Factory setting

LiquidMedium group



Navigation

 Application → Sensor → Add. settings → Medium group

Description

Rough specification of the dielectric constant (DC).
This parameter presets the 'Medium property' parameter as follows:'Others'->'Medium property' = 'Unknown''Water based (DC >= 4)'-> 'Medium property' = 'DC 4 ... 7'
Note:
If 'Medium property' is changed afterwards, 'Medium group' retains its value. Only 'Medium
property' is relevant for the signal evaluation.
Note:
The measuring range may be reduced for small dielectric constants. For details refer to theTechnical Information (TI) of the respective device.

Endress+Hauser

61Description of device parameters

Selection• Others• Water based (DC >= 4)

Factory setting

OthersMicropilot FMRxxB; Operating tool HART

Medium property

Navigation

 Application → Sensor → Add. settings → Medium property

Description

Specify the dielectric costant (DC) of the medium.
Note:
For multiple-phase systems this value refers to the upper medium.

Selection• Unknown• DC 1.2 ... 1.6• DC 1.6 ... 1.9• DC 1.9 ... 2.5• DC 2.5 ... 4• DC 4 ... 7• DC 7 ... 15• DC > 15

Factory setting

DC 1.9 ... 2.5

Upper blank outNavigation

 Application → Sensor → Add. settings → Upper blank out

Description

This parameter describes a line segment between reference point and close to maximum
level (100%). The value is calculated by the device to blanket potentially disturbing signals
coming from this space. The value can be adapted manually.

User entry

0 to 125 000 mm

Factory setting

50 mmUpper blank out evaluation mode
Navigation

 Application → Sensor → Add. settings → UBO eval. mode

Selection• No measurement in upper blank out• Measurement in upper blank out

Factory setting

Measurement in upper blank out

62

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Output modeNavigation

 Application → Sensor → Add. settings → Output mode

Description

Select output mode between:
Ullage =The remaining space in the tank or silo is indicated.
orLevel linearized =The level is indicated (more precisely: the linearized value if a linearization has been
activated).

Selection• Ullage• Level linearized

Factory setting

Level linearized

L max. drain speedNavigation

 Application → Sensor → Add. settings → L max draining

User entry

0.0 to 50 000.0 %/min

Factory setting

0.0 %/min

L max. fill speedNavigation

 Application → Sensor → Add. settings → L max.fill speed

User entry

0.0 to 50 000.0 %/min

Factory setting

0.0 %/minLevel limit modeNavigation

 Application → Sensor → Add. settings → Level limit mode

Description

Determines whether the output value is limited by an upper or lower limit (or by both).

Selection• Off• Low limit• High limit• Low and High Limit

Endress+Hauser

63Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

Low limit

High limitNavigation

 Application → Sensor → Add. settings → High limit

Description

Defines the upper limit of the output value.

User entry

Signed floating-point number

Factory setting0%

Low limit

Navigation

 Application → Sensor → Add. settings → Low limit

Description

Defines the lower limit of the output value.

User entry–200 000.0 to 200 000.0 %Factory setting

0.0 %Level correction



Navigation

 Application → Sensor → Add. settings → Level correction

Description

The measured level is corrected by this value to compensate for a constant level error.
Level correction > 0:
The level is increased by this value.
Level correction < 0:
The level is decreased by this value.

User entry–200 000.0 to 200 000.0 %Factory setting

0.0 %

64

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Echo evaluation
Navigation

 Application → Sensor → Add. settings → Echo evaluation

Navigation

 Application → Sensor → Basic settings

Echo curve statistic

Navigation

 Application → Sensor → Add. settings → Echo evaluation → Ec. curve stat.

Selection

• Off
• On

Factory settingOn

Echo curve statistics upNavigation

 Application → Sensor → Add. settings → Echo evaluation → EC. stat. up

User entry

0 to 30

Factory setting

3

Echo curve statistic down

Navigation

 Application → Sensor → Add. settings → Echo evaluation → ECS in downUser entry

0 to 30

Factory setting

5

Echo curve smoothing modeNavigation

 Application → Sensor → Add. settings → Echo evaluation → EC. smooth.mode

Selection• Off• SG smoothing• Symmetric smoothing• Asymmetric smoothing

Factory setting

Symmetric smoothing

Endress+Hauser

65Description of device parameters

Micropilot FMRxxB; Operating tool HART

Echo curve smoothing

Navigation

 Application → Sensor → Add. settings → Echo evaluation → EC. smoothing

User entry

0 to 9 900 mm

Factory setting

0 mm

FAC offset

Navigation

 Application → Sensor → Add. settings → Echo evaluation → FAC offset

User entry–9 999.0 to 9 999.0 dB

Factory setting

12 dB

FAC window sizeNavigation

 Application → Sensor → Add. settings → Echo evaluation → FAC window size

User entry

0 to 9 900 mm

Factory setting

1 600 mm

Max Value EWCNavigation

 Application → Sensor → Add. settings → Echo evaluation → Max Value EWC

User entry–9 999.0 to 9 999.0 dB

Factory setting

100 dBFirst echo factor



Navigation

 Application → Sensor → Add. settings → Echo evaluation → First echo fact.

User entry

0.0 to 100.0 dB

Factory setting

10 dB

66

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Parabolic fit window sizeNavigation

 Application → Sensor → Add. settings → Echo evaluation → Parab fit window

User entry

0 to 9 900 mm

Factory setting

120 mmTank bottom range

Navigation

 Application → Sensor → Add. settings → Echo evaluation → TB range

Description

Defines the range in which the tank bottom echo is searched for.
The Tank bottom range extends downward and starts at level 0% (Empty calibration). It
ends at the entered value.
Note: If the level 0% (Empty calibration) is far above the tank or silo bottom, the Tankbottom range starts at the entered Tank/silo height.

User entry

0 to 312 500 mm

Factory setting

15 000 mm

Min. amplitude TBDNavigation

 Application → Sensor → Add. settings → Echo evaluation → Min. ampl. TBD

User entry

0 to 9 999.0 dB

Factory setting

3 dB

Lower level area



Navigation

 Application → Sensor → Add. settings → Echo evaluation → Lower level area

User entry

0 to 125 000 mm

Factory setting

1 000 mmEndress+Hauser

67Description of device parameters

Micropilot FMRxxB; Operating tool HART

Evaluation modeNavigation

 Application → Sensor → Add. settings → Echo evaluation → Evaluation mode

Description

Defines the evaluation mode for the echo tracking.

Selection• FlexTracking• FlexTracking - Weak signals• FixTracking• FixTracking - Weak signals

Factory setting

FlexTracking

Reset evaluation



Navigation

 Application → Sensor → Add. settings → Echo evaluation → Reset evaluation

Description

Restarts level determination.

Selection• Reset done
• Yes

Factory setting

Reset done

Window size tracking

Navigation

 Application → Sensor → Add. settings → Echo evaluation → Wind.size track.

User entry

0 to 20 500 mm

Factory setting

500 mmMaximal track counter

Navigation

 Application → Sensor → Add. settings → Echo evaluation → Max track countUser entry

0 to 100

Factory setting

2

68

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Distance
Navigation

 Application → Sensor → Basic settings → DistanceDescription

Distance from lower edge of device flange to product surface.

User interface

Signed floating-point number

Factory setting

0 mm

Debug block index
Navigation

 Application → Sensor → Add. settings → Echo evaluation → Debug block indx

Selection
Factory setting

System control block

Debug parameter index



Navigation

 Application → Sensor → Add. settings → Echo evaluation → Debug parm. idx

User entry

0 to 65 535

Factory setting

2

Debug array index



Navigation

 Application → Sensor → Add. settings → Echo evaluation → Debug array indx

User entry

0 to 255

Factory setting0

Status

Navigation

 Application → Sensor → Add. settings → Echo evaluation → Status

User entry

0 to 255

Factory setting0

Endress+Hauser

69Description of device parameters

Micropilot FMRxxB; Operating tool HART

Debug value integer32
Navigation

 Application → Sensor → Add. settings → Echo evaluation → Debug val. int32

User interface

Positive integer

Factory setting0

Debug value
Navigation

 Application → Sensor → Add. settings → Echo evaluation → Debug value

User interface

Signed floating-point number

Factory setting

4.0

Linearization
Navigation

 Application → Sensor → Linearization

Linearization type

Navigation

 Application → Sensor → Linearization → Lineariz. type

Description

Select type of linearization.

Selection• None• Linear• Table• Pyramid bottom• Conical bottom• Angled bottom• Horizontal cylinder• Sphere

Factory setting

None

70

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Unit after linearization



Navigation

 Application → Sensor → Linearization → Unit lineariz.

Description

Defines the unit of the linearized value.
Note:
The selected unit is only used to be indicated on the display. The measured value is nottransformed according to the selected unit.
Note:
If 'Free text' is selected, an additional parameter 'Free text' appears in which thedesignation of the unit can be defined.

Selection

SI units• STon
•t
• kg
• cm³
• dm³
• m³
• hl
•l
•%
• mm
•m

US units• lb• UsGal
• ft³
• ft
• in

Imperial units
impGalCustom-specific units
Free text
Factory setting%

Free text

Navigation

 Application → Sensor → Linearization → Free text

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

Free text

Level linearized
Navigation

 Application → Sensor → Linearization → Level linearized

Description

Displays the linearized level.

User interface

Signed floating-point number

Factory setting0%

Endress+Hauser

71Description of device parameters

Micropilot FMRxxB; Operating tool HART

Maximum value

Navigation

 Application → Sensor → Linearization → Maximum value

Description

Linearized value corresponding to a level of 100%.

User entry–200 000 to 200 000.0 %Factory setting

100.0 %

Diameter

Navigation

 Application → Sensor → Linearization → Diameter

Description

Diameter of the spherical tank or horizontal cylinder tank.

User entry

0 to 125 000 mm

Factory setting

20 000 mm

Intermediate height

Navigation

 Application → Sensor → Linearization → Intermed. height

Description

Height of the pyramid, conical or angled bottom

User entry

0 to 125 000 mm

Factory setting

0 mm

72

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Table modeNavigation

 Application → Sensor → Linearization → Table mode

Description

Defines the editing mode of the linearization table.
'Manual'The level and the associated linearized value are entered manually for each linearization
point.
'Semiautomatic'
The level is measured by the device for each linearization point. The associated linearized
value is entered manually.
'Clear table'
Deletes the existing linearization table.
'Sort table'
Rearranges the linerization points into an ascending order.
Note:
DeviceCare and FieldCare contain a graphical tool for the easy creation of a linearizationtable.
Device Care: 'Additional functions' -> 'Linearization table'
FieldCare: 'Device Operation' -> 'Device Functions' -> 'Additional functions' -> 'Linearizationtable'

Selection• Manual• Semiautomatic• Clear table• Sort table

Factory setting

ManualTable number



Navigation

 Application → Sensor → Linearization → Table number

Description

Table point you are going to enter or change.

User entry

1 to 32

Factory setting

1

Level



Navigation

 Application → Sensor → Linearization → Level

Description

Enter level value of the table point (value before linearization).

User entry

Signed floating-point number

Endress+Hauser

73Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

0%

Level
Navigation

 Application → Sensor → Linearization → Level

Description

Displays measured level (value before linearization). This value is transmitted to the table.

User interface

Signed floating-point number

Factory setting

0.0 %Customer value

Navigation

 Application → Sensor → Linearization → Customer value

Description

Enter linearized value for the table point.

User entry

Signed floating-point number

Factory setting0%

Activate table

Navigation

 Application → Sensor → Linearization → Activate table

Description

Activate or deactivate table.
The table can only be activated if the table values- are present in at least 2 value pairs- do not exceed the sensor limits- represent a monotonously increasing or monotonously decreasing function

Selection• Disable• Enable

Factory setting

Disable74

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Signal information
Navigation

 Application → Sensor → Signal inform.

Signal quality
Navigation

 Application → Sensor → Signal inform. → Signal quality

Description

Shows the quality of the evaluated level signal.

User interface• Strong• Medium• Weak• No signal• (Strong)• (Medium)• (Weak)

Factory setting

StrongAbsolute echo amplitude
Navigation

 Application → Sensor → Signal inform. → Abs. echo ampl.

Description

Shows the absolute amplitude of the evaluated level signal.

User interface–999 999.9 to 999 999.9 dB

Factory setting

0.0 dB

Relative echo amplitude
Navigation

 Application → Sensor → Signal inform. → Relat.echo ampl.

Description

Shows the relative amplitude (i.e. the distance to the evaluation curve) of the evaluatedlevel signal.

User interface–999 999.9 to 999 999.9 dB

Factory setting

0.0 dB

Endress+Hauser

75Description of device parameters

Micropilot FMRxxB; Operating tool HART

Sensor cycle time
Navigation

 Application → Sensor → Signal inform. → Sens. cycle time

User interface

0 to 65 535 ms

Factory setting

0 ms

Actual IF gain
Navigation

 Application → Sensor → Signal inform. → Actual IF gain

User interface

0 to 1 000

Factory setting0

Antenna zero distanceNavigation

 Application → Sensor → Signal inform. → Ant. zero dist.

User entry

0 to 10 000 mm

Factory setting

230 mm

PLL configuration selector



Navigation

 Application → Sensor → Signal inform. → PLLconf.select.

Selection• PLL Configuration 01• PLL Configuration 02• PLL Configuration 03• PLL Configuration 04• PLL Configuration 05• PLL Configuration 06• PLL Configuration 07• PLL Configuration 08• PLL Configuration 09• PLL Configuration 10

Factory setting

PLL Configuration 02

76

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

3.3.4

Description of device parameters

Current output

Navigation

 Application → Curr.output

Assign PV

Navigation

 Application → Curr.output → Assign PV

Description

Assign a measured variable to the primary dynamic variable (PV).
Additional information:
The assigned measured variable is also used by the current output.

Selection• Level linearized• Distance

Factory setting

Level linearized

Measuring mode current output
Navigation

 Application → Curr.output → Meas.mode outp

Description

Select curve of current output.

Selection• Standard• Inverse

Factory settingStandard

Current range output



Navigation

 Application → Curr.output → Cur.range outp

Description

Defines the current range used to transmit the measured or calculated value.
In brackets are indicated the “low saturation value” and the “high saturation value”.
If Measured value ≤ “low saturation”, the output current is set to “low saturation”.
If Measured value ≥ “high saturation”, the output current is set to ”high saturation”.
Note:
Currents below 3.6 mA or above 21.5 mA can be used to signal an alarm.

Selection• 4...20 mA (4... 20.5 mA)• 4...20 mA NE (3.8...20.5 mA)• 4...20 mA US (3.9...20.8 mA)

Factory setting

4...20 mA NE (3.8...20.5 mA)

Endress+Hauser

77Description of device parameters

Micropilot FMRxxB; Operating tool HART

Lower range value output

Navigation

 Application → Curr.output → Low.range outp

Description

Depending of which variable has been selected as PV, define the related lower and upper
range values.
Assignment PV value to 4 mA and 20 mA.

User entry

Signed floating-point number

Factory setting

0.0 %Upper range value output

Navigation

 Application → Curr.output → Upp.range outp

Description

Depending of which variable has been selected as PV, define the related lower and upper
range values.
Assignment PV value to 4 mA and 20 mA.

User entry

Signed floating-point number

Factory setting

100.0 %

Failure behavior current output
Navigation

 Application → Curr.output → Fail.behav.out

Description

Defines which current the output assumes in the case of an error.
Min: < 3.6 mA
Max: >21.5 mA

Selection• Min.
• Max.

Factory setting

Min.

Failure current

Navigation

 Application → Curr.output → Failure current

Description

Enter current output value in alarm condition

User entry

21.5 to 23 mA

78

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Factory setting

Description of device parameters22.5 mA

Output current
Navigation

 Application → Curr.output → Output curr.

Description

Shows the value currently calculated for the current output

User interface

3.59 to 23 mA

Factory setting

3.59 mA

Terminal current
Navigation

 Application → Curr.output → Terminal curr.

Description

Shows the current value of the current output which is currently measured

User interface

0 to 30 mA

Factory setting

0 mA

4 mA trim value

Navigation

 Application → Curr.output → 4 mA trim value

User entry

3 to 5 mA

Factory setting

4 mA

20 mA trim value

Navigation

 Application → Curr.output → 20 mA trim value

User entry

18 to 22 mA

Factory setting

20 mA

Endress+Hauser

79Description of device parameters

3.3.5

Micropilot FMRxxB; Operating tool HART

HART output

Navigation

 Application → Curr.output

Configuration
Navigation

 Application → HART output → Configuration

HART addressNavigation

 Application → HART output → Configuration → HART address

Description

Enter the address to exchange data via the HART protocol.

User entry

0 to 63

Factory setting0

HART short tag

Navigation

 Application → HART output → Configuration → HART short tag

Description

Defines the short tag for the measuring point.
Maximum length: 8 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Character string comprising numbers, letters and special characters (8)Factory setting

SHORTTAG

Device tagNavigation

 Application → HART output → Configuration → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

5xB/6xB

80

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

No. of preambles



Navigation

 Application → HART output → Configuration → No. of preambles

Description

Defines the number of preambles in the HART telegram

User entry

5 to 20

Factory setting

5

Loop current modeNavigation

 Application → HART output → Configuration → Loop curr mode

Description

If Loop current mode is disabled, Multi-drop communication mode is activated. Multi-drop
is a HART digital communication mode where multiple devices may share the same pair ofwires for power and communications.
In this mode the output current is fixed.

Selection• Disable• Enable

Factory setting

EnableHART output
Navigation

 Application → HART output → HART output

Assign PV

Navigation

 Application → HART output → HART output → Assign PV

Description

Assign a measured variable to the primary dynamic variable (PV).
Additional information:
The assigned measured variable is also used by the current output.

Selection• Level linearized• Distance

Factory setting

Level linearized

Endress+Hauser

81Description of device parameters

Micropilot FMRxxB; Operating tool HART

Primary variable (PV)Navigation

 Application → HART output → HART output → Primary var (PV)

Description

Shows the current measured value of the primary dynamic variable (PV)User interface

Signed floating-point number

Factory setting

100.0 %

Assign SVNavigation

 Application → HART output → HART output → Assign SV

Description

Assign a measured variable to the second dynamic variable (SV).

Selection• Level linearized• Distance• Terminal voltage *• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Percent of range• Loop current• Terminal current *• Not used

Factory setting

Distance

Secondary variable (SV)Navigation

 Application → HART output → HART output → Second.var(SV)Description

Shows the current measured value of the secondary dynamic variable (SV)User interface

0 to 410.10498687664 mm

Factory setting

0 mm

*

82

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Assign TV

Navigation

 Application → HART output → HART output → Assign TV

Description

Assign a measured variable to the tertiary dynamic variable (TV).

Selection• Level linearized• Distance• Terminal voltage *• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Percent of range• Loop current• Terminal current *• Not used

Factory setting

Absolute echo amplitude

Tertiary variable (TV)
Navigation

 Application → HART output → HART output → Tertiary var(TV)Description

Shows the current measured value of the tertiary (third) dynamic variable (TV)User interface–150.0 to 0.0 deciBel

Factory setting–150.0 deciBel

Assign QVNavigation

 Application → HART output → HART output → Assign QV

Description

Assign a measured variable to the quaternary dynamic variable (QV).

Selection• Level linearized• Distance• Terminal voltage *• Electronics temperature• Sensor temperature• Absolute echo amplitude*

Visibility depends on order options or device settings

Endress+Hauser

83Description of device parameters

Micropilot FMRxxB; Operating tool HART

• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Percent of range• Loop current• Terminal current *• Not used
Factory setting

Relative echo amplitude

Quaternary variable (QV)
Navigation

 Application → HART output → HART output → Quaterna.var(QV)

Description

Shows the current measured value of the quaternary (fourth) dynamic variable (QV)

User interface–150.0 to 0.0 deciBel

Factory setting–150.0 deciBel

Burst configuration
Navigation

 Application → HART output → Burst config. 1

Burst mode 1Navigation

 Application → HART output → Burst config. 1 → Burst mode 1

Description

Switch HART burst mode for burst message on

Selection

• Off
• On

Factory setting

Off

*

84

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Burst command 1

Navigation

 Application → HART output → Burst config. 1 → Burst command 1

Description

Select the HART command that is sent to the HART masterSelection• Primary variable (PV)• Loop Current and Percent of Range• Dynamic Variables• Device variables with status• Device variables• Additional device status

Factory setting

Loop Current and Percent of Range

Burst variable 0Navigation

 Application → HART output → Burst config. 1 → Burst variable 0

Description

For HART command 9 and 33, assign a HART device variable or process variable to burst
variable

Selection• Level linearized• Distance• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range• Loop current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used

Factory setting

Level linearized*

Visibility depends on order options or device settings

Endress+Hauser

85Description of device parameters

Micropilot FMRxxB; Operating tool HART

Burst variable 1Navigation

 Application → HART output → Burst config. 1 → Burst variable 1

Description

For HART command 9 and 33, assign a HART device variable or process variable to burst
variable

Selection• Level linearized• Distance• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range• Loop current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used

Factory setting

Level linearized

Burst variable 2

Navigation

 Application → HART output → Burst config. 1 → Burst variable 2

Description

For HART command 9 and 33, assign a HART device variable or process variable to burst
variable

Selection• Level linearized• Distance• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range*

86

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters• Loop current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used
Factory setting

Level linearized

Burst variable 3

Navigation

 Application → HART output → Burst config. 1 → Burst variable 3

Description

For HART command 9 and 33, assign a HART device variable or process variable to burst
variable

Selection• Level linearized• Distance• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range• Loop current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used

Factory setting

Level linearized

Burst variable 4Navigation

 Application → HART output → Burst config. 1 → Burst variable 4

Description

For HART command 33, assign a HART device variable or process variable to burst variable

Selection• Level linearized• Distance• Electronics temperature*

Visibility depends on order options or device settings

Endress+Hauser

87Description of device parameters

Micropilot FMRxxB; Operating tool HART

• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range• Loop current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used
Factory setting

Not used

Burst variable 5Navigation

 Application → HART output → Burst config. 1 → Burst variable 5

Description

For HART command 33, assign a HART device variable or process variable to burst variable

Selection• Level linearized• Distance• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range• Loop current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used

Factory setting

Not used*88

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Burst variable 6

Navigation

 Application → HART output → Burst config. 1 → Burst variable 6

Description

For HART command 33, assign a HART device variable or process variable to burst variable

Selection• Level linearized• Distance• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range• Loop current• Primary variable (PV)• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used

Factory setting

Not used

Burst variable 7

Navigation

 Application → HART output → Burst config. 1 → Burst variable 7

Description

For HART command 33, assign a HART device variable or process variable to burst variable

Selection• Level linearized• Distance• Electronics temperature• Sensor temperature• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Build-up index *• Build-up detected *• Foam index *• Foam detected *• Terminal voltage *• Terminal current *• Percent of range• Loop current• Primary variable (PV)*Visibility depends on order options or device settings

Endress+Hauser

89Description of device parameters

Micropilot FMRxxB; Operating tool HART

• Secondary variable (SV)• Tertiary variable (TV)• Quaternary variable (QV)
• Not used
Factory setting

Not used

Burst trigger modeNavigation

 Application → HART output → Burst config. 1 → Trigger mode

Description

Select the event that triggers the burst message

Selection• Continuous• Window *• Rising *• Falling *• On change

Factory setting

Continuous

Burst trigger level

Navigation

 Application → HART output → Burst config. 1 → Trigger level

Description

Enter the burst trigger value that determines together with the option selected in 'Burst
trigger mode' parameter the time of burst message

User entry

Signed floating-point number

Factory setting

2.0E-38

Min. update period

Navigation

 Application → HART output → Burst config. 1 → Min. upd. per.

Description

Enter the minimum time span between two burst responses of one burst message

User entry

Positive integer

Factory setting

1 000 ms*

90

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Max. update period

Navigation

 Application → HART output → Burst config. 1 → Max. upd. per.

Description

Enter the maximum time span between two burst responses of one burst message

User entry

Positive integer

Factory setting

2 000 ms

Information
Navigation

 Application → HART output → Information

Device ID
Navigation

 Application → HART output → Information → Device ID

Description

Shows the device ID for identifying the device in a HART network

User interface

Positive integer

Factory setting

123 456

Device type
Navigation

 Application → HART output → Information → Device type

Description

Shows the device type with which the measuring device is registered with the HART
Communication Foundation

User interface

0 to 65 535

Factory setting

4 545

Device revision
Navigation

 Application → HART output → Information → Device revision

Description

Shows the device revision with which the device is registered with the HART
Communication Foundation

Endress+Hauser

91Description of device parameters

User interface

0 to 255

Factory setting1

Micropilot FMRxxB; Operating tool HART

HART short tag

Navigation

 Application → HART output → Information → HART short tag

Description

Defines the short tag for the measuring point.
Maximum length: 8 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Character string comprising numbers, letters and special characters (8)Factory setting

SHORTTAG

HART revision
Navigation

 Application → HART output → Information → HART revision

User interface

5 to 7

Factory setting

7

HART descriptor

Navigation

 Application → HART output → Information → HART descriptor

Description

Enter description for the measuring point

User entry

Character string comprising numbers, letters and special characters (16)

Factory setting

5xB/6xB

HART message

Navigation

 Application → HART output → Information → HART message

Description

Use this function to define a HART message which is sent via the HART protocol whenrequested by the master.
Maximum length: 32 characters
Allowed characters: A-Z, 0-9, certain special characters

92

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

5xB/6xB

HART date code

Navigation

 Application → HART output → Information → HART date code

Description

Enter date of the last configuration change. Use this format yyyy-mm-dd

User entry

Character string comprising numbers, letters and special characters (10)

Factory setting

2009-07-20

Endress+Hauser

93Description of device parameters

3.4
Navigation

3.4.1
Navigation

Micropilot FMRxxB; Operating tool HART

System System

Device management
 System → Device manag.

Device tagNavigation

 System → Device manag. → Device tag

Description

Enter a unique name for the measuring point to identify the device quickly within theplant.

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

5xB/6xB

Locking status
Navigation

 System → Device manag. → Locking status

Description

Indicates the type of locking.
'Hardware locked' (HW)The device is locked by the 'WP' switch on the main electronics module. To unlock, set theswitch into the OFF position.
'Safety locked' (SW)
Unlock the device by entering the appropriate access code in 'Enter safety unlocking code'.
'Temporarily locked' (SW)The device is temporarily locked by processes in the device (e.g. data upload/download,
reset). The device will automatically be unlocked after completion of these processes.

User interface

94• Hardware locked• Safety locked• Temporarily locked

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Configuration counter
Navigation

 System → Device manag. → Config. counter

Description

Displays the counter for changes to the device parameters.
Additional information:- If the value for a static parameter is changed when optimizing or configuring theparameter, the counter is incremented by 1. This is to enable tracking different parameterversions.
- When multiple parameters are changed simultaneously, e.g. when loading parameters
into the device from an external source such as FieldCare, the counter may display a higher
value. The counter cannot be reset, nor is it reset to a default value on performing a device
reset.
- Once the counter has reached the value 65535, it restarts at 0.

User interface

0 to 65 535

Factory setting0

Reset device

Navigation

 System → Device manag. → Reset device

Description

Reset the device configuration - either entirely or in part - to a defined state

Selection• Cancel• To fieldbus defaults **• To factory defaults *• To delivery settings *• Restart device

Factory setting

Cancel3.4.2

User management

Navigation



System → User manag.

User role
Navigation

 System → User manag. → User role

Description

Shows the access authorization to the parameters via the operating tool***

Visibility depends on communication
Visibility depends on order options or device settings

Endress+Hauser

95Description of device parameters

Micropilot FMRxxB; Operating tool HART

User interface• Operator• Maintenance• Expert• Production• Development

Factory setting

Maintenance

Password
Navigation



System → User manag. → Password

Description

Enter the password for the 'Maintenance' user role to get access to the functionality of thisrole.

User entry

Character string comprising numbers, letters and special characters (16)

Factory settingEnter access code

Navigation



System → User manag. → Ent. access code

Description

For authorized service personnel only.

User entry

0 to 9 999

Factory setting0

Status password entry
Navigation

 System → User manag. → Status pw entry

Description

Use this function to display the status of the password verification.

User interface• -------• Wrong password• Password rule violated• Password accepted• Permission denied• Confirm PW mismatch• Reset password accepted• Invalid user role
• Wrong sequence of entry

Factory setting--------

96

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

New passwordNavigation

 System → User manag. → New password

Description

Define the new 'Maintenance' password.
A new password is valid after it has been confirmed within the 'Confirm new password'
parameter.
Any valid password consists of 4 to 16 characters and can contain letters and numbers.

User entry

Character string comprising numbers, letters and special characters (16)

Factory setting

Confirm new passwordNavigation

 System → User manag. → Conf. new passw.

Description

Enter the new password again to confirm.

User entry

Character string comprising numbers, letters and special characters (16)

Factory setting

Old passwordNavigation

 System → User manag. → Old password

Description

Enter the current password, to subsequently change the existing password.

User entry

Character string comprising numbers, letters and special characters (16)

Factory setting

Reset password
Navigation



System → User manag. → Reset password

Description

Enter a code to reset the current 'Maintenance' password.
The code is deliverd by your local support.

User entry

Character string comprising numbers, letters and special characters (16)

Factory setting

Endress+Hauser

97Description of device parameters

3.4.3

Micropilot FMRxxB; Operating tool HART

Bluetooth configuration

Navigation

 System → Bluetooth conf.

Bluetooth activation
Navigation

 System → Bluetooth conf. → Bluetooth active

Description

If Bluetooth is deactivated, it can only be reactivated via the display or the operating tool.
Reactivating via the SmartBlue app is not possible.

Selection• Disable• Enable

Factory setting

Enable3.4.4

Display

Navigation

 System → Display

Language
Navigation

 System → Display → Language

Description

Set display language

Selection• English• Deutsch *• Français *• Español *• Italiano *• Nederlands *• Portuguesa *• Polski *• русский язык (Russian) *• Svenska *• Türkçe *• 中文 (Chinese) *• 日本語 (Japanese) *• 한국어 (Korean) *• ‫( الْعَرَبيّة‬Arabic) *• Bahasa Indonesia **98

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters• ภาษาไทย (Thai) *• tiếng Việt (Vietnamese) *• čeština (Czech) *Factory setting

EnglishFormat display
Navigation

 System → Display → Format display

Description

Select how measured values are shown on the display

Selection• 1 value, max. size
• 1 bargraph + 1 value• 2 values

Factory setting

1 value, max. size

Number format



Navigation

 System → Display → Number format

Description

Choose number format for the display.

Selection• Decimal• ft-in-1/16''

Factory setting

DecimalValue 1 display

Navigation

 System → Display → Value 1 display

Description

Select the measured value that is shown on the local display

Selection• Level linearized• Distance• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Current output• Build-up index *• Foam index *• Terminal voltage*

Visibility depends on order options or device settings

Endress+Hauser

99Description of device parameters

Micropilot FMRxxB; Operating tool HART

• Electronics temperature• Sensor temperature• Unfiltered distance
Factory setting

Level linearized

Decimal places 1Navigation

 System → Display → Decimal places 1

Description

This selection does not affect the measurement and calculation accuracy of the device.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Value 2 display

Navigation

 System → Display → Value 2 display

Description

Select the measured value that is shown on the local display

Selection• None• Level linearized• Distance• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Current output• Build-up index *• Foam index *• Terminal voltage• Electronics temperature• Sensor temperature

Factory setting

None

*

100

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Decimal places 2Navigation

 System → Display → Decimal places 2

Description

This selection does not affect the measurement and calculation accuracy of the device.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Value 3 display

Navigation

 System → Display → Value 3 display

Description

Select the measured value that is shown on the local display

Selection• None• Level linearized• Distance• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Current output• Build-up index *• Foam index *• Terminal voltage• Electronics temperature• Sensor temperature

Factory setting

None

Decimal places 3Navigation

 System → Display → Decimal places 3

Description

This selection does not affect the measurement and calculation accuracy of the device.

Selection•x• x.x• x.xx• x.xxx• x.xxxx*

Visibility depends on order options or device settings

Endress+Hauser

101Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

x.xx

Value 4 display

Navigation

 System → Display → Value 4 display

Description

Select the measured value that is shown on the local display

Selection• None• Level linearized• Distance• Absolute echo amplitude• Relative echo amplitude• Area of incoupling• Current output• Build-up index *• Foam index *• Terminal voltage• Electronics temperature• Sensor temperature

Factory setting

None

Decimal places 4Navigation

 System → Display → Decimal places 4

Description

This selection does not affect the measurement and calculation accuracy of the device.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Contrast display
Navigation

 System → Display → Contrast display

Description

Adjust local display contrast setting to ambient conditions (e.g. lighting or reading angle)

User entry

20 to 80 %*102

Visibility depends on order options or device settings

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Factory setting

Description of device parameters

30 %

3.4.5

Geolocation

Navigation



System → Geolocation

Process Unit Tag

Navigation

 System → Geolocation → Process Unit Tag

Description

Enter the process unit in which the device is installed.

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

Process Unit Tag

Location Description



Navigation

 System → Geolocation → Location Descr.

Description

Use this function to enter a description of the location so that the device can be located in
the plant.

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

somewhere

Longitude

Navigation

 System → Geolocation → Longitude

Description

Use this function to enter the longitude coordinates that describe the device location.

User entry–180 to 180 °Factory setting0°

Endress+Hauser

103Description of device parameters

Micropilot FMRxxB; Operating tool HART

LatitudeNavigation

 System → Geolocation → Latitude

Description

Use this function to enter the latitude coordinates that describe the device location.

User entry–90 to 90 °Factory setting0°

AltitudeNavigation

 System → Geolocation → Altitude

Description

Use this function to enter the altitude data that describe the device location.

User entry

Signed floating-point number

Factory setting

0m

Location methodNavigation

 System → Geolocation → Location method

Description

Use this function to select the data format for specifying the geographic location. The
codes for specifying the location are based on the US National Marine Electronics
Association (NMEA) Standard NMEA 0183.

Selection• No fix• GPS or Standard Positioning Service fix• Differential GPS fix• Precise positioning service (PPS) fix• Real Time Kinetic (RTK) fixed solution• Real Time Kinetic (RTK) float solution• Estimated dead reckoning• Manual input mode
• Simulation Mode

Factory setting

No fix104

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

3.4.6

Description of device parameters

Information

Navigation

 System → Information

Device name
Navigation

 System → Information → Device name

Description

Use this function to display the device name. It can also be found on the nameplate.

User interface

Character string comprising numbers, letters and special characters

Factory setting

5xB/6xB

Manufacturer
Navigation

 System → Information → Manufacturer

User interface

Character string comprising numbers, letters and special characters

Factory setting

Endress+Hauser

Serial number
Navigation

 System → Information → Serial number

Description

The serial number is a unique alphanumerical code identifying the device.
It is printed on the nameplate.
In combination with the Operations app it allows to access all device related
documentation.

User interface

Character string comprising numbers, letters and special characters

Factory setting

AAFFFFAAFFF

Order code

Navigation

 System → Information → Order code

Description

Shows the device order code.

User interface

Character string comprising numbers, letters and special characters

Endress+Hauser

105Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

- none -Firmware version
Navigation

 System → Information → Firmware version

Description

Displays the device firmware version installed.

User interface

Character string comprising numbers, letters and special characters

Factory setting

01.00

Hardware version
Navigation

 System → Information → Hardware version

User interface

Character string comprising numbers, letters and special characters

Factory setting

01.00.00

Extended order code 1Navigation



System → Information → Ext. order cd. 1

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.

User interface

Character string comprising numbers, letters and special characters

Extended order code 2NavigationDescription

The extended order code is an alphanumeric code containing all information to identify
the device and its options.

User interface

Character string comprising numbers, letters and special characters

Factory setting-

106

System → Information → Ext. order cd. 2

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Extended order code 3

Navigation



System → Information → Ext. order cd. 3

Description

The extended order code is an alphanumeric code containing all information to identify
the device and its options.

User interface

Character string comprising numbers, letters and special characters

Factory setting-

XML build number
Navigation

 System → Information → XML build no.

User interface

Positive integer

Factory setting

1 118

Information
Navigation

 Application → HART output → Information

Navigation



System → Information → Information

Device ID
Navigation

 Application → HART output → Information → Device ID

Description

Shows the device ID for identifying the device in a HART network

User interface

Positive integer

Factory setting

123 456

Device type
Navigation

 Application → HART output → Information → Device type

Description

Shows the device type with which the measuring device is registered with the HART
Communication Foundation

Endress+Hauser

107Description of device parameters

User interface

0 to 65 535

Factory setting

4 545

Micropilot FMRxxB; Operating tool HART

Device revision
Navigation

 Application → HART output → Information → Device revision

Description

Shows the device revision with which the device is registered with the HART
Communication Foundation

User interface

0 to 255

Factory setting

1

HART short tag

Navigation

 Application → HART output → Information → HART short tag

Description

Defines the short tag for the measuring point.
Maximum length: 8 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Character string comprising numbers, letters and special characters (8)Factory setting

SHORTTAG

HART revision
Navigation

 Application → HART output → Information → HART revision

User interface

5 to 7

Factory setting

7

HART descriptor

Navigation

 Application → HART output → Information → HART descriptor

Description

Enter description for the measuring point

User entry

Character string comprising numbers, letters and special characters (16)108

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Factory setting

Description of device parameters

5xB/6xB

HART message

Navigation

 Application → HART output → Information → HART message

Description

Use this function to define a HART message which is sent via the HART protocol whenrequested by the master.
Maximum length: 32 characters
Allowed characters: A-Z, 0-9, certain special characters

User entry

Character string comprising numbers, letters and special characters (32)

Factory setting

5xB/6xB

HART date code

Navigation

 Application → HART output → Information → HART date code

Description

Enter date of the last configuration change. Use this format yyyy-mm-dd

User entry

Character string comprising numbers, letters and special characters (10)

Factory setting

2009-07-20

3.4.7
Navigation

Additional information
 System → Bluetooth conf.

SensorNavigation



System → Additional info → Sensor

Serial number
Navigation

 System → Additional info → Sensor → Serial number

Description

Shows the serial number of the module.

User interface

Character string comprising numbers, letters and special characters

Endress+Hauser

109

Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

AAFFFFAAFFF

Firmware version
Navigation

 System → Additional info → Sensor → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

Factory setting0

Build no. software
Navigation

 System → Additional info → Sensor → Build no. softw.

Description

Shows the build number of the module firmware.

User interface

0 to 65 535

Factory setting0

Hardware version
Navigation

 System → Additional info → Sensor → Hardware version

Description

Displays the hardware version of the module.

User interface

Character string comprising numbers, letters and special characters

Factory setting- none -110

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Electronics
Navigation



System → Additional info → Electronics

Serial number
Navigation

 System → Additional info → Electronics → Serial number

Description

Shows the serial number of the module.

User interface

Character string comprising numbers, letters and special characters

Factory setting

AAFFFFAAFFF

Firmware version
Navigation

 System → Additional info → Electronics → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

Factory setting0

Build no. software
Navigation

 System → Additional info → Electronics → Build no. softw.

Description

Shows the build number of the module firmware.

User interface

0 to 65 535

Factory setting0

Hardware version
Navigation

 System → Additional info → Electronics → Hardware version

Description

Displays the hardware version of the module.

User interface

Character string comprising numbers, letters and special characters

Endress+Hauser

111Description of device parameters

Factory setting

Micropilot FMRxxB; Operating tool HART

- none -Display/Bluetooth
Navigation



System → Additional info → Displ./Bluetooth

Serial number
Navigation

 System → Additional info → Displ./Bluetooth → Serial number

Description

Shows the serial number of the module.

User interface

Character string comprising numbers, letters and special characters

Factory setting

AAFFFFAAFFF

Firmware version
Navigation

 System → Additional info → Displ./Bluetooth → Firmware version

Description

Displays the firmware version of the module.

User interface

Positive integer

Factory setting0

Build no. software
Navigation

 System → Additional info → Displ./Bluetooth → Build no. softw.

Description

Shows the build number of the module firmware.

User interface

0 to 65 535

Factory setting

0

112

Endress+Hauser

Micropilot FMRxxB; Operating tool HART

Description of device parameters

Hardware version
Navigation

 System → Additional info → Displ./Bluetooth → Hardware version

Description

Displays the hardware version of the module.

User interface

Character string comprising numbers, letters and special characters

Factory setting- none -3.4.8

Software configuration

Navigation

 System → Softw. config.

CRC device configuration
Navigation

 System → Softw. config. → CRC device conf.

Description

CRC device configuration based on current settings of safety relevant parameters.
The CRC device configuration is unique and can be used to detect changes in safety
relevant parameter settings.

User interface

0 to 65 535

Factory setting

65 535Stored CRC device configuration
Navigation

 System → Softw. config. → Stored CRC conf.

Description

Stored CRC after the last safety lock. Factory delivery is 65535 means that the device has
not yet been safety locked.

User interface

0 to 65 535

Factory setting

65 535Endress+Hauser

113

Description of device parameters

Micropilot FMRxxB; Operating tool HART

Timestamp stored CRC device config.
Navigation

 System → Softw. config. → TS stored CRC

Description

Gives the time stamp when the CRC was last stored following completion of the safety lockwizard.

User interface

Character string comprising numbers, letters and special characters

Factory setting

Activate SW option



Navigation

 System → Softw. config. → Activate SW opt.

Description

Enter the application package code or code of another re-ordered functionality to enable it

User entry

Positive integer

Factory setting0

Software option overview
Navigation

 System → Softw. config. → SW option overv.

Description

Shows all enabled software options

User interface

• SIL
• WHG
• Heartbeat Verification• Heartbeat Monitoring

114

Endress+Hauser

*71601142*
71601142www.addresses.endress.com