Operating manual

Thermophant T TTR31, TTR35
Temperature switch

BA201R/09/en/02.05
51009833
Release 01.01.00

TTR31, TTR35

Table of contents
1

Safety instructions. . . . . . . . . . . 3

1. 1
1. 2
1. 3
1. 4

Designated use . . . . . . . . . . . . . . . . . . . . . . 3
Installation, commissioning and operation . . 3
Operational safety . . . . . . . . . . . . . . . . . . . . 3
Return . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3

2

Device identification . . . . . . . . . 4

9Dangerous good sheet . . . . . . .27

2. 1 Nameplate . . . . . . . . . . . . . . . . . . . . . . . . . 4

3Installation . . . . . . . . . . . . . . . . 5

3. 1
3. 2
3. 3
3. 4

Incoming acceptance, storage . . . . . . . . . . . 5
Dimensions . . . . . . . . . . . . . . . . . . . . . . . . 5
Process connection . . . . . . . . . . . . . . . . . . . 6
Installation instructions . . . . . . . . . . . . . . . 7

4

Wiring . . . . . . . . . . . . . . . . . . . . 8

4. 1 DC voltage version with M12 connector . . . 8
4. 2 DC voltage version with valve connector . . 8

5Operation . . . . . . . . . . . . . . . . . 9

5. 1 On-site operation . . . . . . . . . . . . . . . . . . . . 9
5. 2 Operation with PC and Readwin® 2000 . 17

6

Accessories . . . . . . . . . . . . . . . 18

6. 1 Adapter concept for TTR35 . . . . . . . . . . . 18
6. 2 Welding bosses and coupling . . . . . . . . . 20
6. 3 Electrical connection . . . . . . . . . . . . . . . . 21
6. 4 Configuration kit . . . . . . . . . . . . . . . . . . . 22

7

Trouble-shooting. . . . . . . . . . . 23

7. 1
7. 2
7. 3
7. 4
7. 5

Errors and warnings . . . . . . . . . . . . . . . . . 23
Repair . . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Disposal . . . . . . . . . . . . . . . . . . . . . . . . . . 24
Change status (release) . . . . . . . . . . . . . . . 24
Release history . . . . . . . . . . . . . . . . . . . . 24

8

The most important
technical data . . . . . . . . . . . . . 25

8. 1 Power supply . . . . . . . . . . . . . . . . . . . . . . 25
8. 2 Output . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
8. 3 Operating conditions . . . . . . . . . . . . . . . . 25

2

Endress+Hauser

TTR31, TTR35

1

Safety instructions

Safety instructions

1. 1 Designated use
The Thermophant T is a temperature switch for monitoring, displaying and regulating processtemperatures. The device has been safely built with state-of-the-art technology and meets theapplicable requirements and EC Directives. It can, however, be a source of danger if used incorrectly or for anything other than the designated use.

1. 2 Installation, commissioning and operation
Installation, electrical connection, commissioning, operation and maintenance of the measuringsystem must be carried out by trained, qualified specialists authorised to perform such work by
the facility's owner-operator. The specialist must have read and understood these Operating Instructions and must follow the instructions they contain. The device may only be modified andrepair work carried out if this is explicitly permitted in the Operating Instructions. Damaged devices which could be a source of danger may not be commissioned and must be labelled andidentified as defective.

1. 3 Operational safety• Functional safety
The Thermophant T temperature switches were developed according to the standards IEC
61508 and IEC 61511-1 (FDIS). The device version with PNP switch output and additional
analog output is equipped with fault detection and fault prevention facilities within theelectronics and software. This device version can therefore be used to monitor temperature
up to SIL 2 (Safety Integrity Level). The attainable SIL value is determined by the safety
technical characteristics of probability of failure, hardware fault tolerance and the safe failure
fraction. Details on this may be found in the Functional Safety Manual (in development).
• Ex-area
The Thermophant T is not approved for use in Ex-areas.

1. 4 Return
The following procedures must be carried out before a device is returned to Endress+Hauser:• Always enclose a fully completed “Declaration of Contamination” form with the device. Only
then can Endress+Hauser transport and examine a returned device. A copy of the “Declaration of Contamination” can be found on the second last page of these Operating Instructions.
• Remove all fluid residues. This is particularly important if the fluid is hazardous to health, e.g.
flammable, toxic, caustic, carcinogenic, etc.

# Warning!
Do not return a measuring device if you are not absolutely certain that all traces of hazardous
substances have been removed, e.g. substances which have penetrated crevices or diffusedthrough plastic.

Endress+Hauser

3

Device identification

2

TTR31, TTR35

Device identification

2. 1 Nameplate
To identify your device, compare the complete order code and the version information on thedelivery papers with the data on the nameplate.

T09-TTR31xxx-18-xx-xx-xx-000

Fig. 1: Nameplate for device identification (as example)mOrder coder

Connection diagramnSerial numbersMeasuring rangeo

TAG numbertAmbient temperaturepRelease number (change status)u

Degree of protectionq

Connection valuesv

Approvals! Note!
The release number indicates the change status of the device. A change in the last two figures
does not have any affect on the compatibility - see also → Chap. 7.

4

Endress+Hauser

TTR31, TTR35

3

Installation

Installation

3. 1 Incoming acceptance, storage• Incoming acceptance:
Check the packaging and the device for damage. Check that the goods delivered are complete
and nothing is missing.
• Storage:
Storage temperature −40 °C to +85 °C (-40 °F to +185 °F).

3. 2 Dimensions

T09-TTR31xxx-06-xx-xx-en-000

Fig. 2: Dimensions in mm (inches)
Version L in 100 and 200 mm (3.94 and 7.87"), version L* = 50 mm (1.97") with reduced sensor tip
M 12x1 connector as per IEC 60947-5-2
M 16x1.5 or ½ NPT valve plug as per DIN 43650A/ISO 4400

Endress+Hauser

5Installation

TTR31, TTR35

3. 3 Process connection
The following table illustrates the versions of Thermophant T.
TTR31

TTR35

T09-TTR31xxx-17-xx-xx-en-000

Field of application

Item B
Version with thread
process connection
ANSI ¼" NPT(m = AF14) and½" NPT (m = AF27)

Item C
Version with thread
process connection
G ¼A (n = AF14) andG ½A (n = AF27) as
per ISO 228

Monitoring, display and control of process temperaturesin hygienic processes

Process connection

Item A
Version without process connection (’w’).
Suitable welding
bosses and coupling(see Section 6)

Sensor length L

Version L in 100 and 200 mm (3.94 and 7.87"), version L = 50 mm (1.97") only with reduced sensor tip

Measuring range

6

Monitoring, display and control of process temperatures

Item D
Adapter concept - version
with M24x1.5 thread foradapters with process connection for hygienic processes(see Section 6.1.2)-50 °C to +150 °C (-58 °F to 302 °F)

Endress+Hauser

TTR31, TTR35

Installation

3. 4 Installation instructions

T09-TTR31xxx-11-00-xx-xx-000

Fig. 3: Possible installation options for temperature monitoring in pipes
m TTR31n TTR35 for use in hygienic processes

Mounting instructions:• Installation at angle pieces, against the direction of flow (Fig. 3, Item A)• Installation in smaller pipes, inclined against the direction of flow (Fig. 3, Item B)• Installation vertical to the direction of flow (Fig. 3, Item C)• The on-site display can be rotated electronically 180° − see Section 5. 1 "On-site operation"• The housing can be rotated up to 310°

Endress+Hauser

7

Wiring

TTR31, TTR35

4

Wiring4. 1 DC voltage version with M12 connector

P01-PTx3xxxx-04-xx-xx-xx-002

Fig. 4: Thermophant T with M12x1 connector
A1: 1x PNP switch output
A2: PNP switch outputs R1 and m(R2)A2’: PNP switch outputs R1 and m (diagnosis/break contact with adjustment "DESINA")
A3: PNP switch output with additional analog output
A3’: PNP switch output with additional analog output (PIN assignment with "DESINA" setting)! Note!
DESINA (→ Chap. 5.1.3 Basic settings):
R2 = Diagnosis/break contact (more informations about DESINA see www.desina.de)

4. 2 DC voltage version with valve connector

P01-PTx3xxxx-04-xx-xx-xx-003

Fig. 5: Thermophant T with M 16x1.5 or ½ NPT valve plug
B: 1x PNP switch output

8

Endress+Hauser

TTR31, TTR35

5

Operation

Operation

5. 1 On-site operation
The Thermophant T is operated by means of three keys. The digital display and the light emitting
diodes (LED) support navigation in the operating menu.

T09-TTR31xxx-19-xx-xx-en-001

Fig. 6: Position of operating elements and possibilities for display
Background illumination of the digital display:– White = OK status– Red = error status

Endress+Hauser

9Operation

TTR31, TTR35

5.1.1 Navigating in the operating menu

T09-TTR31xxx-19-xx-xx-xx-002

Fig. 7: Navigating in the operating menu
A Function group selection
B Function selection
m Enter the operating menu
– Press the E key for longer than 3 s
n Select the "Function group" with the + or − keyo Select the "Function" with the E key
p Enter or change parameters with the + or − key– Then return to "Function" with the E key Note: If software locking is enabled, it must be disabled before
making entries or changes
q Press the E key several times to return to the "Function group"r Jump back to the measuring position (Home position)– Press the E key for longer than 3 s
s Query to save data (select "YES" or "NO" with the + or − key)
– Confirm with the E key! Note!
Changes to the parameter settings only become effective if you choose s 'YES' when asked tosave data.

10

Endress+Hauser

TTR31, TTR35

Operation

5.1.2 Structure of the operating menu
The structure below shows all the possible fields of the operating menu.

T09-TTR31xxx-19-xx-xx-xx-003

Fig. 8: Operating menu: A function groups, B functions, C settings

Endress+Hauser

11

Operation

TTR31, TTR35

5.1.3 Basic settings
Base

Basic settings

!"#$

%&'(

Technical unit)*)+

Select technical unit:
°C
°F
K

,
-$./

Configure zero
point

010

Position adjustment:
within ±10 °C/K (±19 °F)of the upper range limit

2$(3-Accept zero point010

No settings possible (notavailable in PC software)

4'#5

Display56/++.

PV: measured value
display
PVRO: measured value
display rotated 180°
SP: set switch point
display
SPRO: set switch point
display rotated 180°
OFF: display off
OFFR: display off rotated
180°

56./
#5
#5./
/++

("%

Damping:
display value, output signal010

0...40 s

4$#'

DESINA&/

Connection in accordance with DESINA
guidelines

7$#
#"+*

Safety confirmation

&/
7$#

#"+.

12

Safety release code0

Operation with increased
functional safety (indevelopment). Factory
setting NO should not bechangedEnter the calculated safetycode of the PC Software

Endress+Hauser

TTR31, TTR35

Operation

5.1.4 Output setting• Hysteresis function
The hysteresis function enables two-point control via a hysteresis. Depending on the temperature T, the hysteresis can be set via the switch point SP and the switch-back point
RSP.
• Window function
The window function enables the monitoring of a process temperature range.
• NO contact or NC contact
This switch function is freely selectable.
• Factory setting (if no customer-specific settings have been ordered):
Switch point SP 1: 45 %; Switch-back point RSP 1: 44.5 %Switch point SP 2: 55 %; Switch-back point RSP 2: 54.5 %• Range of adjustment
LRL = Lower Range Limit
URL = Upper Range Limit
LRV = Lower Range Value
URV = Upper Range Value

T09-TTR31xxx-05-xx-xx-xx-001

Fig. 9: m Hysteresis function, n Window function, o NO contact switch status , p NC contact switch status SP Switch point; RSP Switch-back point

Endress+Hauser

13

Operation

TTR31, TTR35

OUT/
OUT2

Output/output 2 (optional)

/%(
/%(8

+%&*
+&*8

Switching characteristic

9'&*
:7&*
9'&/
:7&/
;<80

WINC:
window/NC contact
HYNC:
hysteresis/NC contact
WINO:
window/NO contact
HYNO:
hysteresis/NO contact
4 - 20:
Analog output (only if
available)#5#58

Switch point value010

Switch point-49.5 to 150 °C(-57.1 to 302 °F)
in increments of 0.1 °C(0.18 °F)

.#5
.#58

Switch-back point
value

010

Switch-back point-50 to 149 °C(-58 to 300.2 °F)
in increments of 0.1 °C(0.18 °F)

(#5
(#58

Switch point delay

010

Delay time
0...99 s
in increments of 0.1 s

(.#5
(.#8

Switch-back point
delay

010

Delay time 0...99 s
in increments of 0.1 s

Min. distance between SP and RSP: 0.5 °C/K (0.9 °F)

14Endress+Hauser

TTR31, TTR35

Operation

OUT/
OUT2

Output/output 2 (optional)

;<80

#$(=

Value for 4 mA
(LRV)

010

-50 to 130 °C(-58 to 266 °F)
Lower range value inincrements of 0.1 °C(0.18 °F)#$(%

Value for 20 mA
(URV)

010

-30 to 150 °C(-22 to 302 °F)
Enter upper range value inincrements of 0.1 °C(0.18 °F)

2$(3=Temperatureapplied for 4 mA
(LRV)

010

Take temperature value aslower range value (not via
PC software)

2$(3%Temperatureapplied for 20 mA
(URV)

010

Take temperature value as
upper range value (not vial
PC software)+*%.

Error current

>'&
>"?
:/=4

Current value in event oferror:
MIN = ≤ 3.6 mA
MAX = ≥ 21.0 mA
HOLD = last value

Min. distance between SETL and SETU: 20 °C/K (36 °F)

Endress+Hauser

15

Operation

TTR31, TTR35

5.1.5 Service function setting
SERV

Service functions

#$.6

=/*,

Locking code0Enter the locking code forenabling the device.

*/4$

Change locking
code0Freely selectable code
1...9999.
0 = no locking;
A locking code alreadyassigned can only bechanged by first entering
the old code for enabling
the device.

5.$#

Reset

&/
7$#

Reset all entries to the factory setting.$63*

Revision counter0

Increases by 1 with eachconfiguration=#(3"

Last device status0

Displays the last device
status to occur ≠ 0

#'>
#'>8

Simulation
output 1 or 2

/++

OFF:
No simulation
OPEN:
Switch output open
CLOS:
Switch output closed
3.5:
Simulation values for analog output in mA(3.5/4.0/8.0/12.0/16.0/20.0/21.7)/5$&(if output 2
available)

*=/#
@1A
(if analog output available)

16

>"?3

Max. indicator010

Display of max. measured
process value>'&3

Min. indicator010

Display of min. measured
process value

Endress+Hauser

TTR31, TTR35

Operation

5. 2 Operation with PC and Readwin® 2000

T09-TTR31xxx-04-xx-xx-en-004

Fig. 10: Operation with PC

5.2.1 Additional operating options
In addition to the operating options listed in the previous "On-site operation" section, the ReadWin® 2000 configuration software provides further information on the Thermophant T:
Function group

Description

SERV

Number of switch changes for output 1
Number of switch changes for output 2
Device status

INFO

Tag number
Order code
Limit switch serial number
Sensor serial number
Electronics serial number
Device release(change status)
Hardware version

Endress+Hauser

17

Accessories

TTR31, TTR35

Function group

Description

INFO

Software version

5.2.2 Operating Instructions for Readwin® 2000
Comprehensive information on the ReadWin ® 2000 configuration software may be found in
the Operating Instructions BA 137R/09/en.

6

Accessories

All dimensions in the drawings are given in mm (inches).

6. 1 Adapter concept for TTR35
The process connection is an adapter and the sensor module has an adapter thread(see Section 3. 3, process connection). As a result, the process connection can easily be changed
at a later stage.
6.1.1 Adapter change
The adapter can be changed on TTR35.

T09-TTR31xxx-17-xx-xx-xx-000

Fig. 11: Changing the adapter
m Sensor module with adapter threadn Standard O-ringo Adapter

18

Endress+Hauser

TTR31, TTR35

Accessories

Please note the following when changing the adapter:• Use a new O-ring. Diameter 15.54 x 2.62 mm (0.612" x 0.103").
EPDM 70 Shore FDA 3-A approved.
• The device (sensor module) can be fixed in place with an open-ended wrench AF 27.
• The adapter can be screwed on with an open-ended wrench AF 27 or AF 32 (depending on
the process connection, (see Section 6.1.2 adapter versions)).
The maximum torque is 80 Nm. The thread can become loose if exposed to severe strain
through pressure and temperature. For this reason, the air-tightness must be checked
regularly and the thread tightened if necessary.
• When changing the adapter, make sure that the sensor tube of the sensor is not damaged.

! Note!
We recommend to change the O-ring in the same time frame as of all other sealings in yourprocess.
6.1.2 Adapter versions
TTR35: order numbers for clamp adapter
versions.
Version DB: order no. 52023994
Version DL: order no. 52023995

P01-PTx3xxxx-06-xx-xx-en-009

Endress+Hauser

19

Accessories

TTR31, TTR35

TTR35: order numbers for hygiene adapter
versions.
Version LB: order no. 52023996
Version LL: order no. 52023997
Version PH: order no. 52023999
Version PL: order no. 52023998
Version HL: order no. 52024000

P01-PTx3xxxx-06-xx-xx-en-010

6.1.3 O-ring for adapter change
O-ring 15.54 x 2.62 mm (0.612" x 0.103"), EPDM 70 Shore FDA, order number 52024267

6. 2

Welding bosses and coupling

6.2.1 Welding boss with sealing taper
Collar welding boss
Seal, moveable coupling, material of parts incontact with process: 316L, PEEK
Order number: 51004751

T09-TSM470AX-06-09-00-en-000

20Endress+Hauser

TTR31, TTR35

Accessories

6.2.2 Collar welding boss
Material of parts in contact with process: 316L
Order no. 51004752

T09-TSM470BX-06-09-00-en-000

6.2.3 Coupling with sealing taper
G ½" process connection
Seal, moveable coupling, material ofparts in contact with process: 316L
Order no. 51004753

T09-TSM470AX-06-09-00-en-001

6. 3 Electrical connection
6.3.1 Plug-in jackM 12x1 plug-in jackSelf-made connection to
M 12x1 housing connector
Order number: 52006263
P01-PMP13xxx-00-xx-00-xx-003

6.3.2 Elbow plug
Elbow plug
4-pole M12 connector for customised cable
construction, elbowed, IP67, PG7
Order number: 51006327

T09-TTR3xxxx-06-09-xx-en-000

Endress+Hauser

21

Accessories

TTR31, TTR35

6.3.3 Connecting cable• Cable, 4 x 0.34 mm2 (22 AWG) with M12
socket, elbowed, screw plug, length 5 m(16.4 ft), PVC cable - Order number:
52010285• Cable, 4 x 0.34 mm2 (22 AWG) with M12
socket, with LED, elbowed, 316L screw plug,
length 5 m (16.4 ft), PVC cable, special forhygiene applications,order number: 52018763
Display:− gn: device operational− ye1: switch status 1− ye2: switch status 2Core colours:– 1 = BN brown– 2 = WH white– 3 = BU blue
– 4 = BK black

T09-TTR31xxx-00-00-xx-xx-001

6. 4 Configuration kit• Configuration kit for PC-programmabletransmitters - ReadWin® 2000 setup programand interface cable for PCs with USB port;
Adapter for transmitters with 4-pole postconnector
Order code: TXU10-AA• ReadWin® 2000 can be downloaded free ofcharge directly from the internet at the following address:www.endress.com/readwin

22

T09-TTR31xxx-00-00-xx-xx-000

Endress+Hauser

TTR31, TTR35

7

Trouble-shooting

Trouble-shooting

7. 1 Errors and warnings
If an error in the device occurs, the colour of the status LED changes from green to red and thebackground illumination of the digital display changes from white to red. The display shows:• E-code for errors
In the event of an error message, the measured value is uncertain.
• W-code for warnings
In the event of a warning, the measured value is reliable.

Code

Explanation

E011

Device configuration faulty

E012

Error in measurement or underreach/overreach in SIL mode

E015

Error in EEPROM

E019

Power supply has undervoltage/overvoltage

E020

Error in Flash

E021

Error in RAM

E022

USB supply voltage

E025

Switching contact 1 is not open although it should be

E026

Switching contact 2 is not open although it should be

E040

VCC (Controller voltage) is out of working area

E042

Output current cannot be generated

E044

Output current drifts too much (± 0.5 mA)

CodeExplanation

W107

Simulation active

W202

Temperature outside the sensor range (not SIL mode)Endress+Hauser

23Trouble-shooting

TTR31, TTR35

CodeExplanation

W209

Device starts

W210

Configuration modified

W212

Sensor signal outside the permitted range (not SIL mode)

W250

Number of switch cycles exceeded

W270

Short-circuit and overload at output 1

W280

Short-circuit and overload at output 2

7. 2 Repair
A repair is not planned.

7. 3 Disposal
When disposing, ensure that the materials of the device components are separated and processed
accordingly.

7. 4 Change status (release)
The release number on the nameplate and in the Operating Instructions indicates the change
status of the device: XX.YY.ZZ (example 01.02.01).
XX
YY
ZZ

Change in the main version.
Compatibility no longer provided. Device and Operating Instructions change.
Change in functionality and operation.
Compatibility provided. Operating Instructions change.
Trouble-shooting and internal modifications.
Operating Instructions do not change.

7. 5 Release history

24

Date

Release no.
deviceDevice and software-No.
(Firmware/Software)

06.2004

1.00.00

1.00.00

12.2004

1.01.00

1.00.00

Changes

OperatingmanualKA174r/09/en(51008032)

New analog
electronics

BA201r/09/en(51009833)

Endress+Hauser

TTR31, TTR35

8

The most important technical data

The most important technical data

8. 1 Power supply
Supply voltage• DC voltage version 12...30 V DC
Current consumption• Without load < 60 mA, with reverse polarity protection
Power supply failure• Behaviour in case of overvoltage (> 30 V)
The device works continuously up to 34 V DC without any damage. No damage is caused tothe device in case of a short-term overvoltage up to 1 kV (as per IEC 61000-4-5). If the supply
voltage is exceeded, the properties specified are no longer guaranteed.
• Behaviour in case of undervoltage
If the supply voltage drops below the minimum value, the device switches off (status as if not
supplied with power = switch open).

8. 2 Output
Switching capacity• Switch status ON: Ia ≤ 250 mA• Switch status OFF: Ia ≤ 1 mA
• Switching cycles: > 10,000,000• Voltage drop PNP: ≤ 2 V• Overload protection
Automatic load testing of switching current; output is switched off in case of overcurrent, theswitching current is tested again every 0.5 s; max. capacitance load: 14 µF for max. supply
voltage (without resistive load).
Load (analog output)• Max. (Vsupply - 6.5 V) / 0.022 A
Signal on alarm• Analog output: ≤ 3.6 mA or ≥ 21.0 mA adjustable(if setting ≥ 21.0 mA the output is ≥ 21.5 mA)• Switch outputs: in safe state (switch normally open)

8. 3 Operating conditions• Any orientation• Any position-dependent zero shift can be corrected
Offset: ±20 % URL
Operating conditions: Environment• Ambient temperature range–40...+85 °C (-40...+ 185 °F)• Storage temperature–40...+85 °C (-40...+ 185 °F)
Endress+Hauser

25

The most important technical data

TTR31, TTR35

Operating conditions: Process• Process temperature limits-50 to 150 °C (-58 to 302 °F)"

Caution!
Restrictions depending on process connection and ambient temperature:– No restriction with coupling (see Accessories, → Chap. 6.2.1, → Chap. 6.2.3, order no.
51004751, 51004753) and neck tube length min. 20 mm.
– with process connection:
max. ambient temperature

max. process temperature

up to 25 °C (77 °F)

no restriction

up to 40 °C (104 °F)

135 °C (275 °F)

up to 60 °C (140 °F)

120 °C (248 °F)

up to 85 °C (185 °F)

100 °C (212 °F)• Process pressure limits
p/T load diagram as per DIN 43763 or Dittrich

T09-TSM470ZZ-05-00-xx-en-000

Fig. 12: p/T load diagram
L = insertion length
vL = medium velocity air
vW = medium velocity water

26

Endress+Hauser

TTR31, TTR35

Dangerous good sheet

9Dangerous good sheet

Declaration of Contamination
Erklärung zur Kontamination
Because of legal regulations and for the safety of our employees and operating equipment, we need the "declaration ofcontamination", with your signature, before your order can be handled. Please make absolutely sure to include it withthe shipping documents, or - even better - attach it to the outside of the packaging.
Aufgrund der gesetzlichen Vorschriften und zum Schutz unserer Mitarbeiter und Betriebseinrichtungen, benötigen wir die
unterschriebene "Erklärung zur Kontamination", bevor Ihr Auftrag bearbeitet werden kann. Legen Sie diese unbedingt den
Versandpapieren bei oder bringen Sie sie idealerweise außen an der Verpackung an.
Type of instrument / sensor
Geräte-/Sensortyp_______________________________________

Process data/Prozessdaten

Temperature / Temperatur

Serial number
Seriennummer ____________________________ [°C] Pressure / Druck__________ [ Pa ]Conductivity / Leitfähigkeit _________ [ S ] Viscosity / Viskosität __________ [mm2/s]
Medium and warnings
Warnhinweise zum Medium

Medium /concentration Identification flammable
CAS No.
Medium /Konzentration
entzündlich

toxic
giftigcorrosiveätzendharmful/irritant
gesundheitsschädlich/
reizendother *
harmlesssonstiges* unbedenklich

Process
mediumMedium imProzess
Medium forprocess cleaning
Medium zur
Prozessreinigung
Returned partcleaned with
Medium zur
Endreinigung* explosive; oxidising; dangerous for the environment; biological risk; radioactive* explosiv; brandfördernd; umweltgefährlich; biogefährlich; radioaktiv
Please tick should one of the above be applicable, include security sheet and, if necessary, special handling instructions.
Zutreffendes ankreuzen; trifft einer der Warnhinweise zu, Sicherheitsdatenblatt und ggf. spezielle Handhabungsvorschriften beilegen.

Reason for return / Grund zur Rücksendung _________________________________________________________________________________________________________________________________________________________________________________________________________________________________________________________________
Company data /Angaben zum AbsenderCompany /Firma ________________________________

Contact person / Ansprechpartner ____________________________________________________________________

Department / Abteilung ______________________________

Address / Adresse

Phone number/ Telefon ____________________________________________________________________________

Fax / E-Mail _____________________________________________________________________________________Your order No. / Ihre Auftragsnr. _______________________

P/SF/Konta VIII

We hereby certify that the returned parts have been carefully cleaned. To the best of our knowledge they are free from any residues indangerous quantities.
Hiermit bestätigen wir, dass die zurückgesandten Teile sorgfältig gereinigt wurden, und nach unserem Wissen frei von Rückständen in
gefahrbringender Menge sind.

___________________________________________________________________________________________(place, date / Ort, Datum)(Company stamp and legally binding signature)(Firmenstempel und rechtsverbindliche Unterschrift)

Endress+Hauser27www.endress.com/worldwide

BA201R/09/en/02.05
51009833
CCS/FM6.0+SGML