EL3000 SERIES
100mm CHART ANALOG RECORDER (PEN WRITING TYPE)

EL 3000 series are analog recorders sized 144 x
144mm with 100mm width chart which have 3 typesof 1-pen, 2-pen and 3-pen.
The unit starts recording as soon as the power
supply and input are connected and it is also easyto operate.
Scale plate, input range and function of therecorder can be selected for various purpose andapplications as many kinds of options are prepared.
 FEATURES
y Universal power supply
Universal power supply with voltage range of 100 to240V AC is applied. (50/60Hz switchable)y Linear temperature scale
Temperature scale of thermocouple and resistancethermometer input is a linear scale that is excellent inreading indication value.
y Standard 6 chart speeds
6 chart speeds (5,10,20,40,80,160mm/h) areswitchable as standard. 5 chart speed and hour/minutechange are prepared as option.

y Unit structure and light-weightLight-weight (50% of the previous unit weight) was
realized by easy maintenance unit structure.
y Alarm setting (individual input alarm) as standard
Higher and lower limit alarm can be programmed foreach channel. Alarm value is easy to be programmedby pointer location. You can check the alarm by front
LED lighting. Alarm output is prepared as option.

 MODELSy 1-pen typey 2-pen type, 3-pen type

EL3P1 -EL3
Input signals
5 : Thermocouple/DC voltage
7 : Resistance thermometer
Thermocouple with burnout/DC voltage
Built-in voltage divider input (option)*1
Input and scale plate (option)
0 : Standard input + standard scale plate
1 : Non-standard input(Including current input, and built-involtage divider) + standard scale plate2 : Standard input + non-standard scale plate
3 : Non-standard input(Including current input, and built-in
voltage divider) + Non-standard scale plate
Alarm output (option)
0 : None
1 : 2 alarm outputs
Chart speed and burnout (option)
0 : 6-speed+ burnout disabled
1 : 6-speed + up-scale burnout
2 : 6-speed + down-scale burnout
A : 5-speed hour/minute change + burnout disabled
B : 5-speed hour/minute change + up-scale burnout
C : 5-speed hour/minute change+ down-scale burnout)*1: Optional built-in voltage divider and thermocouple/resistance thermometer burnout input is only type "7".
*2: Input and scale selection are needed for non-standard input and non-standard scale plate. 1st pen must be
selected as thermocouple input.
*3 Burnout are programmed together for all channels for thermocouple/ resistance thermometer input.

1Input points
F : 2 penG : 3 pen
1st pen input and scale plate*2
0 : Standard input + standard scale plate
1 : Non-standard input (Including current input
and built-in voltage divider) +standard scale plate
2 : Standard input + non-standard scale plate
3 : Non-standard input (Including current input
and built-in voltage divider) +non-standard scale plate
2nd pen input and scale plate*2
0 : Standard input + standard scale plate
1 : Non-standard input (Including current input
and built-in voltage divider) +standard scale plate
2 : Standard input + non-standard scale plate
3 : Non-standard input (Including current input
and built-in voltage divider) +non-standard scale plate
3rd pen input and scale plate*2
N : None (for 2 pen)
0 : Standard input + standard scale plate
1 : Non-standard input (Including current input
and built-in voltage divider)+ standard scale plate
2 : Standard input + non-standard scale plate
3 : Non-standard input (Including current input
and built-in voltage divider) +non-standard scale plate
Alarm output (option)
0 : None
2 : Alarm output 4points/6points(2-pen type : 4 points, 3-pen type : 6 points)
Chart speed and burnout (option)*3
0 : 6-speed + burnout disabled
1 : 6-speed + up-scale burnout
2 : 6-speed + down-scale burnout
A : 5-speed hour/minute change + burnout disabled
B : 5-speed hour/minute change + up-scale burnout
C : 5-speed hour/minute change + down-scale burnout

PSE-361 INPUT SPECIFICATIONS

 OPERATION / PROGRAMMING SPECIFICATIONS

Measuring channels :Reference range and types :Switches :1, 2, 3 channels
DC voltage --- ±13.8mV, ±27.6mV, ±69mV,±200mV, ±500mV, ±2V, ±5V
Built-in voltage divider (option) ; ±10V, ±25V,±50V
DC current --- External installation of shunt
resister (250Ω) is applied (option)
Thermocouples --- K, E, J, T, R, B (option)
Resistance thermometer --- Pt100 (1997)(Measured current ; 1mA)
*Linear scale for thermocouple and resistancethermometer
Input designation :Refer to above reference range
Single scale
Accuracy rating :±0.5% of input span (except for some input
under standard operating condition)
Refer to the table of standard range andminimum width of scale for non-standard input
Indicating deadband :
0.3% of input span
Reference junction compensation accuracy :K, E, J, T --- ±1.0°C or less (23°C ±10°C)±2.0°C or less (0 to 50°C)(For internal reference junction compensation,
the errors above are added to the accuracyrating)Temperature drift :±0.02%/°C (Converted into reference ranges)
Sampling rate :125ms
Indicating resolution :Approximately 1/2000
Burnout (option) :On thermocouple or resistance thermometer
input, disconnection of signal can be detected.
(Specify up-scale or down-scale)Burnout detection --- Voltage application
method (approximately 8V, 1mA)
Allowable signal source resistance :Thermocouple inputs, DC voltage inputs (±5Vor less)--- 1KΩ (burnout disabled) or less
DC voltage inputs (input more than ±5V)--- 100Ω or less
Resistance thermometer inputs--- per wire 10Ω or less (Same resistance for 3wires)
Input resistance :Thermocouple inputs, DC voltage inputs (±5Vor less) --- Approximately 8MΩ
DC voltage inputs (more than ±5V)--- Approximately 1MΩ
Maximum allowable input voltage :Thermocouple inputs, DC voltage inputs--- ±10V DC or less
Applied voltage :DC voltage inputs (Voltage divider built-in)--- ±60V DC or less
Resistance thermometer --- ±6V DC or lessMaximum common mode voltage:30V AC
Common mode rejection ratio : 120dB or more (50/60Hz ±0.1%)
Normal mode rejection ratio :50dB or more (50/60Hz ±0.1%)

 RECORDING SPECIFICATIONS
Recording type :Balancing time :Recording color :Chart paper :Chart speed :Chart speed accuracy :Pen lift :Disposable fiber tipped pen
Input span movement--- approximately 2 seconds
1 Red, 2 Green, 3 Blue
Fan-fold type : total width of 114mm,
total length of 10m
effective chart width of100mm
6-speed change, 5, 10, 20, 40, 80, 160mm/h(standard)±0.1% or less (It is based on the chart scale)
Manual operation (up or down) INDICATING SPECIFICATIONS
Analog indication:
Scale plate:

Scale plate and pointerSingle scale (minimum scale division: 80)

 ALARM SPECIFICATIONS
Alarm display:

Alarm types:
Alarm programming:

Alarm deadband:
Alarm output (option):

Pointer and alarm-point sticker pasted onscale.
Alarm LED lamp lightens for alarming(All channels OR output)
Higher and lower-limit alarm
Individual setting for higher and lower-limitvalue(Programming percentage of input span byindicating pointer, input resolution 0.5%)
0.4% of input span
Individual for each channels, a contact and 2outputs (common)
Maximum contact capacity:
2A (resistive load), 0.5A (inductive load)

Indication :POWER --- ON/OFF the recorder power supply
INDICATE --- Normal operation / stop indication andrecording
CHART SPEED --- Selecting chart speed(Chart feed stops when all switches are OFF)SET-RUN --- Switching alarm setup/normal operationmode--- Moves pointer for alarms setup and calibration
Hz --- Power frequency 50/60Hz switchable
SELECT --- Pen selection for programming and adjusting(2-pen type and 3-pen type only)
LED (green) --- Power ON monitor
LED (red) --- Alarm monitor

 GENERAL SPECIFICATIONS
Rated power voltage : 100 to 240V AC, 50/60Hz (switchable)with power supply switch
Power consumption : 1- pen type --- Maximum 16VA (100V AC),
Maximum 22VA (240V AC)
2- pen type --- Maximum 24VA (100V AC),
Maximum 34VA (240V AC)
3- pen type --- Maximum 30VA (100V AC),
Maximum 40VA (240V AC)
Environmental conditions :Reference operation condition--- Ambient temperature range : 21 to 25 °C
Ambient humidity range : 45 to 65% RH
Power voltage : 100V AC ±1%
Power frequency : 50/60 Hz ±0.5%
Attitude : left/right 0°, forward tilting 0°,
backward tilting 0°
Warm-up time : longer than 30 minutes
Normal operation condition--- Ambient temperature range : 0 to 50 °C
Ambient humidity range : 20 to 80% RH
Power voltage : 90 to 264V AC
Power frequency : 50/60 Hz ±2%.
Attitude : left/right 0 to 10°, forward tilting 0°,
backward tilting 0 to 20°
Transportation condition (at the packed condition on
shipment from our factory)--- Ambient temperature range : -20 to 60°C
Ambient humidity range : 5 to 90% RH (No dewcondensation)
2Vibration : 10 to 60 Hz, 4.9m/s , (0.5G) or less
Impact : 392m/s2 (40G) or less
Storage condition--- Ambient temperature range : -20 to 60°C
Ambient humidity range : 5 to 90% RH(No dew condensation)
Insulation resistance : Secondary terminals and protective conductor terminals--- 20MΩ or more at 500V DC
Primary terminals and protective conductor terminals--- 20MΩ or more at 500V DC
Primary and secondary terminals--- 20MΩ or more at 500V DC
Note : Primary terminals--- Power (L.N), Alarm terminals (mechanical relay)
Secondary terminals --- Measurement input terminals
Dielectric strength : Secondary terminals and protective conductor terminals--- 1 minute at 500V AC
Primary terminals and protective conductor terminals--- 1 minute at 1500V AC
Primary and secondary terminals--- 1 minute at 1500V AC
Note : Primary terminals--- Power (L.N), Alarm terminals (mechanical relay)
Secondary terminals --- Measurement input terminals
Case :
Door (frame) --- ABS resin
Window --- glass
Case --- 1-pen type --- ABS resin
2-pen type and 3-pen-type --- SteelColor :
Door (frame) --- Black (equivalent to Mussel N1.5),
Window --- Transparent
Case --- Black (equivalent to Mussel N1.5)
Mounting :Panel mounting
Weight :1-pen type --- Approximately 1.6kg (full option)
2-pen type and 3-pen type --- Approximately 2.6kg (fulloption)
Power voltage fluctuation :Indication fluctuation 0.2% or less (converted into reference
ranges at 90 to 264V AC) STANDARDS (Conformity pending)CE marking:

EMC directive, low voltage directive conformity
EN61326+A1+A2+A3, EN61010-1* Under EMC directive test condition, indication equivalent
to maximum 500µV fluctuates in case

 MAINTENANCE
Input correction :Memory reset :2

Zero/span correction for individual input
Initializes indication adjustment value(User maintenance area) OPTION SPECIFICATIONS
Options

Alarm output

DC current input
Built-in voltage
divider

Non-standard input

Non-standard
scale plate Standard range and minimum width of scale

Contents
Alarm contact output is available
Alarm relay --- Individual mechanical relay acontact, 2 outputs (common)
Maximum contact rating --250V AC 2A, 30V DC 2A(resistive load)
250V AC 0.5A, 30V DC 0.5A (inductive load))250Ω of shunt resistor is applied to measure
voltage input
Built-in voltage divider(1/1000) measures input
in the range of ±5V to ±50V (input type "7" onlyfor 1-pen type)
Refer to the table of standard range andprogrammable minimum width of scale
Minimum width of scale --DC voltage: 10mV DC width or more
Thermocouple:
K; 250°C width or more
E,J,T; 200°C width or more
R; 800°C width or more
Resistance thermometer: 100°C width or more

Input typeDC voltage

DC current
KE

Scale plate for non-standard input

Burnout

Chart speed
Hour/minute change
16m chart paper

T/CJ

Function for detecting disconnection for sensor
with thermocouple or resistance thermometer
input.
Specify up-scale or down-scale(Input type "7" only for 1-pen type), parallel
operation is not possible
5-speed change, 5,10,20,40,80mm/minute,hour change
Maximum length 15.6m

TRBRTDDC voltage

KE

T/C

J
T
R

RTD

Scales000-5-1010000000000000-50
0000000-20-50-50

totototototototototototototototototototototototototototototo

10mV
20mV
50mV
5mV
10mV
5V
250ºC
300ºC
400ºC
600ºC
800ºC
1000ºC
1200ºC
200ºC
300ºC
300ºC
400ºC
200ºC
300ºC
150ºC
1400 ºC
1600ºC
100ºC
150ºC
200ºC
300ºC
500ºC
80ºC
50ºC
150ºC

Chart
paper
Nos.
EM008
EM519
EL42003
EL42056
EL42057
EL42010EL05096EL05010EL05009
EL05081EL05121EL05157
EL05116EL05047EL05010EL05010EL05009
EL05047EL05010EL05007
EL05137EL05147EL05052
EL05034
EL05047EL05010EL05048EL05035
EL05006
EL05007

Minimum
scalesInput
signals

0.2
0.5
10.2
0.5
0.05
5510101020205551055520
20225510
225

M1
M8
M9
M6
M7
V6K2
K3
K4
K6K8
KA
KC
E2
E3
J3
J4
T2
T3
T5R4
R6313A
32333538
3E
3B

Minimum width of scale
10mV17mV35mV100mV250mV
1V
2.5V
5V
13V
25V
10mA
250ºC
400ºC
700ºC
200ºC
250ºC
380ºC
720ºC
200ºC
300ºC
500ºC
200ºC
300ºC
800ºC
1480ºC
900ºC
150ºC
200ºC
400ºC

K,E,J,T,R : IEC584,JIS C 1602-1995
Pt100 : IEC751,JIS C 1604-1997
Programmable minimum width of scale: DC voltage --- 10mV DC width or more
Thermocouple ---K:
250℃ width or more
E,J,T: 200℃ width or more
R:
800℃ width or more
Resistance thermometer --- 100℃ width or more

zStandard scale and chart paper Nos.
Input type

Standard range-13.8 to 13.8mV-27.6 to 27.6mV-69 to 69mV
-200 to 200mV-500 to 500mV-2 to 2V
-5 to 5V
-10 to 10V-25 to 25V-50 to 50V
4 to 20mA-200 to 330ºC-200 to 660ºC-200 to 1370ºC-200 to 200ºC-200 to 380ºC-200 to 720ºC-200 to 900ºC-200 to 250ºC-200 to 500ºC-200 to 1200ºC-200 to 280ºC-200 to 400ºC
0 to 1240 ºC
0 to 1760ºC
0 to 1820ºC-140 to 150ºC-200 to 300ºC-200 to 650ºC

zExceptions of accuracy ratings
Input types
K,E,J,T

Measuring range-200to-50ºCB

0to400ºC

R0to400ºCAccuracy ratings±1.0% of measuring
range
None±1.0% of measuring
rangeNote) The accuracy ratings are converted into the measuring range

K,E,J,T,R : IEC584,JIS C 1602-1995
Pt100 : IEC751,JIS C 1604-1997

3 TERMINAL BOARD
y 1-Pen typey 2-Pen type, 3-Pen type

 DIMENSIONS
y 1-Pen typezPanel cutout

y 2-Pen type, 3-Pen typezMinimum clearance for plural installation

Unit: mm·Mounting bracket

mm PaperSpecifications subject to change without notice. Printed in Japan (I) 2008. Unit:
11. Recycled

32-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
PHONE: +81-3-3956-2171
FAX: +81-3-3956-0915
E-mail: inter@chino.co.jp

Website: http://www.chino.co.jp

4