GP01094D/06/EN/02.18
71399896
2018-08-01

Products

Solutions

Valid as of version
01.00.zz (Device firmware)

Description of Device Parameters
Proline Promass 300
FOUNDATION Fieldbus
Coriolis flowmeter

Menu/Variable
P

Proline Device
Access status tooling:
Operation
Setup
Diagnostics
ExpertP

Locking status:

P

Access status tooling:
Enter access code:
SystemP

Sensor
Input
OutputCommunication
Application
Diagnostics

Services

Proline Promass 300 FOUNDATION Fieldbus

Table of contents

Table of contents
1

About this document . . . . . . . . . . . . . . . . 4

1.1
1.2
1.3

Document function . . . . . . . . . . . . . . . . . . . . . 4
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using this document . . . . . . . . . . . . . . . . . . . . 4
1.3.1 Information on the document
structure . . . . . . . . . . . . . . . . . . . . . . . 4
1.3.2 Structure of a parameter
description . . . . . . . . . . . . . . . . . . . . . 6
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . 6
1.4.1 Symbols for certain types of
information . . . . . . . . . . . . . . . . . . . . 6
1.4.2 Symbols in graphics . . . . . . . . . . . . . . . 7
Documentation . . . . . . . . . . . . . . . . . . . . . . . . 7
1.5.1 Standard documentation . . . . . . . . . . . 7
1.5.2 Supplementary device-dependent
documentation . . . . . . . . . . . . . . . . . . 7

1.4

1.5

2

Overview of the Expert operating
menu . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 8

3

Description of device parameters . . . 11

3.1

"System" submenu . . . . . . . . . . . . . . . . . . . . . 13
3.1.1 "Display" submenu . . . . . . . . . . . . . . . 14
3.1.2 "Configuration backup" submenu . . . . 27
3.1.3 "Diagnostic handling" submenu . . . . . . 30
3.1.4 "Administration" submenu . . . . . . . . . 45
"Sensor" submenu . . . . . . . . . . . . . . . . . . . . . . 50
3.2.1 "Measured val." submenu . . . . . . . . . . 51
3.2.2 "System units" submenu . . . . . . . . . . . 66
3.2.3 "Process param." submenu . . . . . . . . . 74
3.2.4 "Measurement mode" submenu . . . . . 82
3.2.5 "External comp." submenu . . . . . . . . . 84
3.2.6 "Calculated value" submenu . . . . . . . . 87
3.2.7 "Sensor adjustment" submenu . . . . . . . 90
3.2.8 "Calibration" submenu . . . . . . . . . . . . 97
"I/O configuration" submenu . . . . . . . . . . . . . . 99
"Input" submenu . . . . . . . . . . . . . . . . . . . . . . 101
3.4.1 "Current input 1 to n" submenu . . . . . 101
3.4.2 "Status input 1 to n" submenu . . . . . . 104"Output" submenu . . . . . . . . . . . . . . . . . . . . . 106
3.5.1 "Current output 1 to n" submenu . . . . 107
3.5.2 "Pulse/frequency/switch output
1 to n" submenu . . . . . . . . . . . . . . . 121
3.5.3 "Relay output 1 to n" submenu . . . . . 142"Communication" submenu . . . . . . . . . . . . . . 148
3.6.1 "Resource block" submenu . . . . . . . . 149
3.6.2 "WLAN settings" submenu . . . . . . . . 170
3.6.3 "Web server" submenu . . . . . . . . . . . 176"Analog inputs" submenu . . . . . . . . . . . . . . . 179
3.7.1 "Analog input 1 to n" submenu . . . . . 180"Discrete inputs" submenu . . . . . . . . . . . . . . . 228
3.8.1 "Discrete input 1 to n" submenu . . . . 229"Analog outputs" submenu . . . . . . . . . . . . . . 257
3.9.1 "Multiple AO" submenu . . . . . . . . . . 257

3.2

3.3
3.4
3.5

3.6

3.7
3.8
3.9

Endress+Hauser

3.10"Discrete outputs" submenu . . . . . . . . . . . . . . 266
3.10.1 "Multiple DO" submenu . . . . . . . . . . 266
3.11 "Application" submenu . . . . . . . . . . . . . . . . . 275
3.11.1 "Totalizer 1 to n" submenu . . . . . . . . 275
3.11.2 "Viscosity" submenu . . . . . . . . . . . . . 279
3.11.3 "Concentration" submenu . . . . . . . . . 280
3.12 "Diagnostics" submenu . . . . . . . . . . . . . . . . . 280
3.12.1 "Diagnostic list" submenu . . . . . . . . . 283
3.12.2 "Event logbook" submenu . . . . . . . . . 287
3.12.3 "Device information" submenu . . . . . 289
3.12.4 "Mainboard module" submenu . . . . . 294
3.12.5 "Sens. electronic" submenu . . . . . . . . 295
3.12.6 "I/O module 1" submenu . . . . . . . . . . 296
3.12.7 "I/O module 2" submenu . . . . . . . . . . 297
3.12.8 "Display module" submenu . . . . . . . . 298
3.12.9 "Min/max val." submenu . . . . . . . . . 299
3.12.10 "Data logging" submenu . . . . . . . . . . 309
3.12.11 "Heartbeat" submenu . . . . . . . . . . . . 318
3.12.12 "Simulation" submenu . . . . . . . . . . . 318

4Country-specific factory settings . . 328

4.1

SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . .
4.1.1 System units . . . . . . . . . . . . . . . . . .
4.1.2 Full scale values . . . . . . . . . . . . . . .
4.1.3 Output current span . . . . . . . . . . . .
4.1.4 Pulse value . . . . . . . . . . . . . . . . . . .
4.1.5 On value low flow cut off . . . . . . . . .
US units . . . . . . . . . . . . . . . . . . . . . . . . . . .
4.2.1 System units . . . . . . . . . . . . . . . . . .
4.2.2 Full scale values . . . . . . . . . . . . . . .
4.2.3 Output current span . . . . . . . . . . . .
4.2.4 Pulse value . . . . . . . . . . . . . . . . . . .
4.2.5 On value low flow cut off . . . . . . . .

4.2

328
328
328
329
329
329
330
330
331
331
331
332

5

Explanation of abbreviated units . . 334

5.1
5.2
5.3

SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 334
US units . . . . . . . . . . . . . . . . . . . . . . . . . . . 334
Imperial units . . . . . . . . . . . . . . . . . . . . . . . 336

Index . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 337

3About this document

Proline Promass 300 FOUNDATION Fieldbus

1

About this document

1.1

Document function

The document is part of the Operating Instructions and serves as a reference forparameters, providing a detailed explanation of each individual parameter of the Expert
operating menu.
It is used to perform tasks that require detailed knowledge of the function of the device:• Commissioning measurements under difficult conditions• Optimal adaptation of the measurement to difficult conditions• Detailed configuration of the communication interface• Error diagnostics in difficult cases

1.2

Target group

The document is aimed at specialists who work with the device over the entire life cycle
and perform specific configurations.

1.3

Using this document

1.3.1

Information on the document structure

The document lists the submenus and their parameters according to the structure from theExpert menu (→  8), which is displayed when the "Maintenance" user role isenabled.

4

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

About this document

Operating menu for operators and maintenances
Language
Operatation!
Language

Operator

Parameter 1
Parameter n
Submenu 1
Submenu n
Setup

Device tag

Wizard n / Parameter nAdvanced setup

Enter access code
Parameter 1

Task-oriented

Maintenance

Wizard 1 / Parameter 1

Parameter n
Submenu 1
Submenu n
Diagnostics

Parameter 1
Parameter n
Submenu 1
Submenu nOperating menu for experts
ExpertAccess status display
Parameter n

Sensor
Input
OutputCommunication

Function-oriented

ExpertSystemApplication
Diagnostics

A0029160-EN

1

Sample graphic for the schematic layout of the operating menu

Additional information regarding:• The arrangement of the parameters according to the menu structure of the
Operation menu, Setup menu, Diagnostics menu with a brief description:
Operating Instructions →  7• Operating concept of the operating menus: Operating Instructions →  7

Endress+Hauser

5About this document

Proline Promass 300 FOUNDATION Fieldbus

1.3.2

Structure of a parameter description

The individual parts of a parameter description are described in the following section:
Complete parameter nameWrite-protected parameter =Navigation

Navigation path to the parameter via the local display (direct access code) or web browser
Navigation path to the parameter via the operating tool
The names of the menus, submenus and parameters are abbreviated to the form in which they appear on
the display and in the operating tool.

Prerequisite

The parameter is only available under these specific conditions

Description

Description of the parameter function

Selection

List of the individual options for the parameter• Option 1• Option 2

User entry

Input range for the parameter

User interface

Display value/data for the parameter

Factory setting

Default setting ex works

Additional information

Additional explanations (e.g. in examples):• On individual options• On display values/data• On the input range• On the factory setting• On the parameter function

1.4

Symbols used

1.4.1

Symbols for certain types of information

SymbolMeaning
Tip
Indicates additional information.
Reference to documentationA

Reference to page
Reference to graphic
Operation via local display
Operation via operating tool
Write-protected parameter6

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

1.4.2

About this document

Symbols in graphics

SymbolMeaning

SymbolMeaning

1, 2, 3 ...

Item numbers

A, B, C, ...

Views

A-A, B-B, C-C, ...

Sections

1.5

Documentation

1.5.1

Standard documentation

Operating Instructions
Measuring device

Documentation code

Promass A 300 (8A3B**-…)

BA01515D

Promass A 300 (8A3C**-…)

BA01843D

Promass E 300

BA01517D

Promass F 300

BA01518D

Promass H 300

BA01519D

Promass I 300

BA01520D

Promass O 300

BA01521D

Promass P 300

BA01522D

Promass Q 300

BA01523D

Promass S 300

BA01524D

Promass X 300

BA01525D

1.5.2

Supplementary device-dependent documentation

Special documentation
Contents

Documentation code

Information on the Pressure Equipment Directive

SD01614D

Remote display and operating module DKX001

SD01763D

Radio approvals for WLAN interface for A309/A310 display module

SD01793D

Web server

SD01665D

Heartbeat Technology

SD01696D

Concentration measurementSD01706D

Petroleum–

Viscosity measurement Promass I

SD01720D

Viscosity measurement Promass Q

SD01720D

Endress+Hauser

7

Overview of the Expert operating menu

2

Proline Promass 300 FOUNDATION Fieldbus

Overview of the Expert operating menu

The following table provides an overview of the menu structure of the expert operating
menu and its parameters. The page reference indicates where the associated description of
the submenu or parameter can be found.

 Expert
Direct access (0106)→  11

Locking status (0004)→  12

Access status (0005)→  13

Ent. access code (0003)→  13‣ System→  13‣ Display→  14‣ Config. backup→  27‣ Diagn. handling→  30‣ Administration→  45‣ Sensor→  50‣ Measured val.

→  51‣ System units→  66‣ Process param.

→  74‣ Measurement mode→  82‣ External comp.

→  84‣ Calculated value→  87‣ Sensor adjustm.

→  90‣ Calibration→  97‣ I/O config.

8→  99I/O 1 to n terminals (3902–1 to n)→  99I/O 1 to n info (3906–1 to n)→  99Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Overview of the Expert operating menu

I/O 1 to n type (3901–1 to n)→  100

Apply I/O config (3907)→  100

Conversion code (2762)→  101‣ Input→  101‣ Current input 1 to n

→  101‣ Status input 1 to n

→  104‣ Output→  106‣ Curr.output 1 to n

→  107‣ PFS output 1 to n

→  121‣ Relay output 1 to n

→  142‣ Communication→  148
Device address (11061)→  149‣ Resource block→  149‣ WLAN settings→  170‣ Web server→  176‣ Analog inputs→  179‣ Analog input 1 to n
‣ Discrete inputs→  228‣ Discrete input 1 to n
‣ Analog outputs→  229→  257‣ Multiple AO‣ Discrete outputs→  257→  266‣ Multiple DO‣ Application→  266→  275

Reset all tot. (2806)

Endress+Hauser→  180→  2759Overview of the Expert operating menu

Proline Promass 300 FOUNDATION Fieldbus‣ Totalizer 1 to n

→  275‣ Viscosity→  279‣ Concentration→  280‣ Diagnostics→  280
Actual diagnos. (0691)→  281Prev.diagnostics (0690)→  282

Time fr. restart (0653)→  282

Operating time (0652)→  283‣ Diagnostic list

→  283‣ Event logbook→  287‣ Device info

→  289‣ Mainboard module

10‣ Sens. electronic→  295‣ I/O module 1→  296‣ I/O module 2→  297‣ I/O module 3→  296‣ I/O module 4→  296‣ Display module→  298‣ Min/max val.

→  299‣ Data logging→  309‣ Heartbeat→  318‣ Simulation→  318

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

3

Description of device parameters

Description of device parameters

In the following section, the parameters are listed according to the menu structure of thelocal display. Specific parameters for the operating tools are included at the appropriate
points in the menu structure.

 Expert
Direct access (0106)→  11

Locking status (0004)→  12

Access status (0005)→  13

Ent. access code (0003)→  13‣ System→  13‣ Sensor→  50‣ I/O config.

→  99‣ Input→  101‣ Output→  106‣ Communication→  148‣ Analog inputs→  179‣ Discrete inputs→  228‣ Analog outputs→  257‣ Discrete outputs→  266‣ Application→  275‣ Diagnostics→  280

Direct access

NavigationDescription

Use this function to enter the access code to enable direct access to the desired parametervia the local display. A parameter number is assigned to each parameter for this purpose.

User entry

0 to 65 535

Endress+Hauser

Expert → Direct access (0106)

11

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

User entry
The direct access code consists of a 5-digit number (at maximum) and the channel
number, which identifies the channel of a process variable: e.g. 00914-2. In the navigation
view, this appears on the right-hand side in the header of the selected parameter.
00914-2

1

A0029414

1

Direct access code

Note the following when entering the direct access code:• The leading zeros in the direct access code do not have to be entered.
Example: Enter "914" instead of "00914"• If no channel number is entered, channel 1 is accessed automatically.
Example: Enter 00914 → Assign variable parameter• If a different channel is accessed: Enter the direct access code with the corresponding
channel number.
Example: Enter 00914-2 → Assign variable parameter

Locking status
Navigation

 Expert → Locking status (0004)

Description

Displays the active write protection.

User interface• Hardware locked• Temp. locked

Additional information

Display
If two or more types of write protection are active, the write protection with the highest
priority is shown on the local display. In the operating tool all active types of write
protection are displayed.
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device →  7
Selection

12

Options

Description

None

The access status displayed in the Access status parameter (→  13) applies .
Only appears on local display.

Hardware locked(priority 1)

The DIP switch for hardware locking is activated on the PCB board. This locks write
access to the parameters (e.g. via local display or operating tool) .

Temp. locked

Write access to the parameters is temporarily locked on account of internal
processes running in the device (e.g. data upload/download, reset etc.). Once theinternal processing has been completed, the parameters can be changed once
again.

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Access status
Navigation

 Expert → Access status (0005)

Description

Displays the access authorization to the parameters via the local display, Web browser or
operating tool.

User interface• Operator• MaintenanceFactory setting

Maintenance

Additional information

Description
Access authorization can be modified via the Ent. access code parameter(→  13).
If additional write protection is active, this restricts the current access authorizationeven further.
Display
Detailed information on access authorization is provided in the "User roles andassociated access authorization" and "Operating concept" sections of the Operations
Instructions for the device →  7

Ent. access code
Navigation

 Expert → Ent. access code (0003)

Description

Use this function to enter the user-specific release code to remove parameter write
protection.

User entry

0 to 9 999

3.1
Navigation"System" submenu
 Expert → System‣ SystemEndress+Hauser‣ Display→  14‣ Config. backup→  27‣ Diagn. handling→  30‣ Administration→  4513

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

3.1.1
Navigation"Display" submenu
 Expert → System → Display‣ Display

14

Display language (0104)→  15

Format display (0098)→  15

Value 1 display (0107)→  180% bargraph 1 (0123)→  19100% bargraph 1 (0125)→  20Decimal places 1 (0095)→  20

Value 2 display (0108)→  20Decimal places 2 (0117)→  21

Value 3 display (0110)→  210% bargraph 3 (0124)→  22

100% bargraph 3 (0126)→  22

Decimal places 3 (0118)→  23

Value 4 display (0109)→  23

Decimal places 4 (0119)→  24

Display interval (0096)→  24

Display damping (0094)→  25Header (0097)→  25Header text (0112)→  26

Separator (0101)→  26

Contrast display (0105)→  27

Backlight (0111)→  27

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Display language
Navigation

 Expert → System → Display → Display language (0104)

Prerequisite

A local display is provided.

Description

Use this function to select the configured language on the local display.

Selection• English• Deutsch *• Français *• Español *• Italiano *• Nederlands *• Portuguesa *• Polski *• русский язык(Ru) *• Svenska *• Türkçe *• 中文 (Chinese) *• 日本語 (Japanese) *• 한국어 (Korean) *• Bahasa Indonesia *• tiếng Việt (Vit) *• čeština (Czech) *Factory setting

English (alternatively, the ordered language is preset in the device)

Format display
Navigation

 Expert → System → Display → Format display (0098)

Prerequisite

A local display is provided.

Description

Use this function to select how the measured value is shown on the local display.

Selection• 1 value, max.
• Bargr. + 1 value• 2 values• Val. large+2val.
• 4 values

Factory setting

1 value, max.

*

Visibility depends on order options or device settings

Endress+Hauser

15Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
The display format (size, bar graph etc.) and number of measured values displayed
simultaneously (1 to 4) can be configured. This setting only applies to normal operation.
• The Value 1 display parameter (→  18) to Value 4 display parameter(→  23) are used to specify which measured values are shown on the local
display and in what order.
• If more measured values are specified than the display mode selected permits, thenthe values alternate on the device display. The display time until the next change isconfigured via the Display interval parameter (→  24).

16

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Possible measured values shown on the local display:"1 value, max." option
XXXXXXXXX

900.00kg/h
A0013099"Bargr. + 1 value" option
XXXXXXXXX
900.00 kg/h

900.00l/h
A0013098"2 values" option
XXXXXXXXX

900.00kg/h

900.00l/h
A0013100"Val. large+2val." option
XXXXXXXXX

900.00kg/h
900.00 l/h
1.00 kg/l
A0013102"4 values" option
XXXXXXXXX
900.00 kg/h
900.00 l/h
1.0 kg/l
213.94 kg
A0013103

Endress+Hauser

17Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Value 1 display

Navigation

 Expert → System → Display → Value 1 display (0107)

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values to be shown on the local display.

Selection• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *• HBSI *• Totalizer 1• Totalizer 2• Totalizer 3• Curr.output 1

Factory setting

Mass flow*

18

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
If several measured values are displayed at once, the measured value selected here will be
the first value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  15) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  66).
Selection• Oscil. frequency option
Displays the current oscillation frequency of the measuring tubes. This frequency
depends on the density of the medium.
• Oscil. amplitude option
Displays the relative oscillation amplitude of the measuring tubes in relation to thepreset value. This value is 100 % under optimum conditions.
• Oscil. damping option
Displays the current oscillation damping. Oscillation damping is an indicator of thesensor's current need for excitation power.
• Signal asymmetry option
Displays the relative difference between the oscillation amplitude at the inlet and outletof the sensor. The measured value is the result of production tolerances of the sensor
coils and should remain constant over the life time of a sensor.

0% bargraph 1

Navigation

 Expert → System → Display → 0% bargraph 1 (0123)

Prerequisite

A local display is provided.

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 1.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 kg/h• 0 lb/min

Additional information

Description
The Format display parameter (→  15) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  66).

Endress+Hauser

19Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

100% bargraph 1

Navigation

 Expert → System → Display → 100% bargraph 1 (0125)

Prerequisite

A local display is provided.

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 1.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter →  328

Additional information

Description
The Format display parameter (→  15) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  66).

Decimal places 1Navigation

 Expert → System → Display → Decimal places 1 (0095)

Prerequisite

A measured value is specified in the Value 1 display parameter (→  18).

Description

Use this function to select the number of decimal places for measured value 1.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

Value 2 display

Navigation

 Expert → System → Display → Value 2 display (0108)

Prerequisite

A local display is provided.

20

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to select one of the measured values to be shown on the local display.

Selection

For the picklist, see the Value 1 display parameter (→  18)

Factory setting

None

Additional information

Description
If several measured values are displayed at once, the measured value selected here will be
the second value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  15) is used to specify how many measured
values are displayed simultaneously and how.
Dependency
The unit of the displayed measured value is taken from the System units submenu(→  66).

Decimal places 2Navigation

 Expert → System → Display → Decimal places 2 (0117)

Prerequisite

A measured value is specified in the Value 2 display parameter (→  20).

Description

Use this function to select the number of decimal places for measured value 2.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

Value 3 display

Navigation

 Expert → System → Display → Value 3 display (0110)

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values to be shown on the local display.

Selection

For the picklist, see the Value 1 display parameter (→  18)

Factory setting

None

Endress+Hauser

21Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
If several measured values are displayed at once, the measured value selected here will be
the third value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  15) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu(→  66).

0% bargraph 3

Navigation

 Expert → System → Display → 0% bargraph 3 (0124)

Prerequisite

A selection was made in the Value 3 display parameter (→  21).

Description

Use this function to enter the 0% bar graph value to be shown on the display for themeasured value 3.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 kg/h• 0 lb/min

Additional information

Description
The Format display parameter (→  15) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  66).

100% bargraph 3

Navigation

 Expert → System → Display → 100% bargraph 3 (0126)

Prerequisite

A selection was made in the Value 3 display parameter (→  21).

Description

Use this function to enter the 100% bar graph value to be shown on the display for themeasured value 3.

User entry

Signed floating-point number

Factory setting

0

22

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
The Format display parameter (→  15) is used to specify that the measured value
is to be displayed as a bar graph.
User entry
The unit of the displayed measured value is taken from the System units submenu(→  66).

Decimal places 3Navigation

 Expert → System → Display → Decimal places 3 (0118)

Prerequisite

A measured value is specified in the Value 3 display parameter (→  21).

Description

Use this function to select the number of decimal places for measured value 3.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

Value 4 display

Navigation

 Expert → System → Display → Value 4 display (0109)

Prerequisite

A local display is provided.

Description

Use this function to select one of the measured values to be shown on the local display.

Selection

For the picklist, see the Value 1 display parameter (→  18)

Factory setting

None

Endress+Hauser

23Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
If several measured values are displayed at once, the measured value selected here will be
the fourth value to be displayed. The value is only displayed during normal operation.
The Format display parameter (→  15) is used to specify how many measured
values are displayed simultaneously and how.
Selection
The unit of the displayed measured value is taken from the System units submenu(→  66).

Decimal places 4Navigation

 Expert → System → Display → Decimal places 4 (0119)

Prerequisite

A measured value is specified in the Value 4 display parameter (→  23).

Description

Use this function to select the number of decimal places for measured value 4.

Selection•x• x.x• x.xx• x.xxx• x.xxxx

Factory setting

x.xx

Additional information

Description
This setting does not affect the measuring or computational accuracy of the device.
The arrow displayed between the measured value and the unit indicates that thedevice computes with more digits than are shown on the local display.

Display interval
Navigation

 Expert → System → Display → Display interval (0096)

Prerequisite

A local display is provided.

Description

Use this function to enter the length of time the measured values are displayed if thevalues alternate on the display.

User entry

1 to 10 s

Factory setting

5s

24

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
This type of alternating display only occurs automatically if the number of measured
values defined exceeds the number of values the selected display format can display
simultaneously.
• The Value 1 display parameter (→  18) to Value 4 display parameter (→  23)
are used to specify which measured values are shown on the local display.
• The display format of the displayed measured values is specified using the Format
display parameter (→  15).

Display damping

Navigation

 Expert → System → Display → Display damping (0094)

Prerequisite

A local display is provided.

Description

Use this function to enter a time constant for the reaction time of the local display tofluctuations in the measured value caused by process conditions.

User entry

0.0 to 999.9 s

Factory setting

0.0 s

Additional information

User entry
Use this function to enter a time constant (PT1 element 1)) for display damping:• If a low time constant is entered, the display reacts particularly quickly to fluctuating
measured variables.
• On the other hand, the display reacts more slowly if a high time constant is entered.
Damping is switched off if 0 is entered (factory setting).

Header

Navigation

 Expert → System → Display → Header (0097)

Prerequisite

A local display is provided.

Description

Use this function to select the contents of the header of the local display.

Selection• Device tag• Free text

Factory setting

Device tag

Additional information

Description
The header text only appears during normal operation.

1)

proportional transmission behavior with first order delay

Endress+Hauser

25Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

1

XXXXXXXXX

A0029422

1

Position of the header text on the display

Selection• Device tag
Is defined in the Device tag parameter (→  290).
• Free text
Is defined in the Header text parameter (→  26).

Header text

Navigation

 Expert → System → Display → Header text (0112)

Prerequisite

In the Header parameter (→  25), the Free text option is selected.

Description

Use this function to enter a customer-specific text for the header of the local display.

User entry

Max. 12 characters such as letters, numbers or special characters (e.g. @, %, /)Factory setting------------

Additional information

Description
The header text only appears during normal operation.
1

XXXXXXXXX

A0029422

1

Position of the header text on the display

User entry
The number of characters displayed depends on the characters used.

Separator



Navigation

 Expert → System → Display → Separator (0101)

Prerequisite

A local display is provided.

26

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to select the decimal separator.

Selection• . (point)• , (comma)

Factory setting. (point)

Contrast display
Navigation

 Expert → System → Display → Contrast display (0105)

Prerequisite

A local display is provided.

Description

Use this function to enter a value to adapt the display contrast to the ambient conditions(e.g. the lighting or viewing angle).

User entry

20 to 80 %Factory setting

Depends on the display

Backlight
Navigation

 Expert → System → Display → Backlight (0111)

Prerequisite

One of the following conditions is met:
• Order code for "Display; operation", option F "4-line, illum.; touch control"• Order code for "Display; operation", option G "4-line, illum.; touch control +WLAN"• Order code for "Display; operation", option O "Separate 4-line display, illum.; 10m/30ft
cable; touch control"

Description

Use this function to switch the backlight of the local display on and off.

Selection• Disable• Enable

Factory setting

Enable3.1.2"Configuration backup" submenu

Navigation

 Expert → System → Config. backup‣ Config. backup
Operating time

Endress+Hauser→  2827

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Last backup→  28Config. managem.

→  28Backup state→  29Compar. result→  29Operating time
Navigation

 Expert → System → Config. backup → Operating time (0652)

Description

Use this function to display the length of time the device has been in operation.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The maximum number of days is 9999, which is equivalent to 27 years.

Last backup
Navigation

 Expert → System → Config. backup → Last backup (2757)

Description

Displays the time since a backup copy of the data was last saved to the device memory.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Config. managem.

Navigation

 Expert → System → Config. backup → Config. managem. (2758)

Description

Use this function to select an action to save the data to the device memory.

Selection• Cancel• Execute backup• Restore• Compare• Clear backup

Factory setting

Cancel28

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

Execute backup

A backup copy of the current device configuration is saved from the HistoROM
backup to the memory of the device. The backup copy includes the transmitter dataof the device.
The following message appears on local display: Backup active, please wait!

RestoreThe last backup copy of the device configuration is restored from the device
memory to the device's HistoROM backup. The backup copy includes thetransmitter data of the device.
The following message appears on local display: Restore active! Do not interrupt
power supply!

CompareThe device configuration saved in the device memory is compared with the current
device configuration of the HistoROM backup.
The following message appears on local display: Comparing files
The result can be viewed in Compar. result parameter.

Clear backup

The backup copy of the device configuration is deleted from the memory of thedevice.
The following message appears on local display: Deleting file

HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

Backup state
Navigation

 Expert → System → Config. backup → Backup state (2759)

Description

Displays the status of the data backup process.

User interface• None• Backup in progr.
• Restore in progr• Delete in progr.
• Comp. in progr.
• Restoring failed• Backup failed

Factory setting

None

Compar. result
Navigation

 Expert → System → Config. backup → Compar. result (2760)

Description

Displays the last result of the comparison of the data records in the device memory and in
the HistoROM.

User interface• Set. identical• Set. not ident.
• No backup

Endress+Hauser

29Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Backup corrupt• Check not done
• Dataset incomp.
Factory setting

Check not done

Additional information

Description
The comparison is started via the Compare option in the Config. managem.
parameter (→  28).
Selection
Options

Description

Set. identical

The current device configuration of the HistoROM is identical to the backup copy inthe device memory.
If the transmitter configuration of another device has been transmitted to thedevice via HistoROM in the Config. managem. parameter, the current device
configuration of the HistoROM is only partially identical to the backup copy in thedevice memory: The settings for the transmitter are not identical.

Set. not ident.

The current device configuration of the HistoROM is not identical to the backup
copy in the device memory.

No backup

There is no backup copy of the device configuration of the HistoROM in the device
memory.

Backup corrupt

The current device configuration of the HistoROM is corrupt or not compatible withthe backup copy in the device memory.

Check not done

The device configuration of the HistoROM has not yet been compared to the
backup copy in the device memory.

Dataset incomp.

The backup copy in the device memory is not compatible with the device.

HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

3.1.3"Diagnostic handling" submenu

Navigation

 Expert → System → Diagn. handling‣ Diagn. handling

30

Alarm delay (0651)→  31‣ Diagn. behavior→  31

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Alarm delay

Navigation

 Expert → System → Diagn. handling → Alarm delay (0651)

Description

Use this function to enter the time interval until the device generates a diagnostic
message.
The diagnostic message is reset without a time delay.

User entry

0 to 60 s

Factory setting

0s

Additional information

ResultThis setting affects the following diagnostic messages:• 046 Sensor limit• 140 Sensor sig.asym.
• 144 MeasErrorTooHigh• 830 Sensor temp.
• 831 Sensor temp.
• 832 Electronic temp.
• 833 Electronic temp.
• 834 Process temp.
• 835 Process temp.
• 843 Process limit• 862 Partly filled• 912 Medium inhomog.
• 913 Medium unsuitab.
• 944 MonitoringFailed"Diagnostic behavior" submenu
Each item of diagnostic information is assigned a specific diagnostic behavior at thefactory. The user can change this assignment for specific diagnostic information in the
Diagn. behavior submenu (→  31).
The following options are available in the Assign behavior of diagnostic no. xxx
parameters:

Endress+Hauser

Options

Description

Alarm

The device stops measurement. The signal outputs and totalizers assume the definedalarm condition. A diagnostic message is generated.
The background lighting changes to red.

Warning

The device continues to measure. The signal outputs and totalizers are not affected. A
diagnostic message is generated.

31Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Options

Description

Logbook only

The device continues to measure. The diagnostic message is displayed only in the Eventlogbook submenu (→  287) (Event list submenu (→  288)) and is not displayed inalternation with the operational display.

Off

The diagnostic event is ignored, and no diagnostic message is generated or entered.

Assignment of the status of a diagnostic event.
The following options are available in the Assign status of diagnostic no. xxx
parameters:• Bad
The status of the input value (PV) is Bad.
• Uncertain
The status of the input value (PV) is Uncertain.
• GoodThe status of the input value (PV) is Good.
For a list of all the diagnostic events, see the Operating Instructions for the device→ 7
Navigation

 Expert → System → Diagn. handling → Diagn. behavior‣ Diagn. behavior

32

Diagnostic no. 046 (0709)→  33

Diagnostic no. 140 (0708)→  34

Diagnostic no. 144 (0731)→  34

Diagnostic no. 374 (0710)→  35

Diagnostic no. 441 (0657)→  35

Diagnostic no. 442 (0658)→  35

Diagnostic no. 443 (0659)→  36

Diagnostic no. 444 (0740)→  36

Diagnostic no. 830 (0800)→  36

Diagnostic no. 831 (0641)→  37

Diagnostic no. 832 (0681)→  37

Diagnostic no. 833 (0682)→  38Diagnostic no. 834 (0700)→  38Diagnostic no. 835 (0702)→  38Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Diagnostic no. 046 (Sensor limit)

Description of device parameters

Diagnostic no. 862 (0679)→  39Diagnostic no. 912 (0703)→  39Diagnostic no. 913 (0712)→  39Diagnostic no. 944 (0732)→  40Diagnostic no. 948 (0744)→  40Status diag. 046 (11042)→  40Status diag. 140 (11041)→  41

Status diag. 144 (11016)→  41

Status diag. 374 (11000)→  41

Status diag. 830 (11006)→  42

Status diag. 831 (11007)→  42

Status diag. 832 (11002)→  42

Status diag. 833 (11003)→  43

Status diag. 834 (11004)→  43

Status diag. 835 (11005)→  43

Status diag. 862 (11008)→  44

Status diag. 912 (11009)→  44

Status diag. 913 (11010)→  44

Status diag. 944 (11014)→  45

Status diag. 948 (11011)→  45



Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 046 (0709)

Description

Option for changing the diagnostic behavior of the diagnostic message 046 Sensor limit.

Endress+Hauser

33

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Alarm

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 140 (Sensor sig.asym.)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 140 (0708)

Description

Option for changing the diagnostic behavior of the diagnostic message 140 Sensor
sig.asym..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Alarm

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 144 (MeasErrorTooHigh)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 144 (0731)

Description

Option for changing the diagnostic behavior of the diagnostic message
144 MeasErrorTooHigh.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Alarm

Additional information

34

Detailed description of the options available for selection: →  31

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Diagnostic no. 374 (Sensor electron.)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 374 (0710)

Description

Option for changing the diagnostic behavior of the diagnostic message 374 Sensor
electron..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 441 (Curr.output 1 to n)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 441 (0657)

Description

Option for changing the diagnostic behavior of the diagnostic message 441 Curr.output
1 to n.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 442 (Freq. output 1 to n)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 442 (0658)

Prerequisite

The measuring device has a pulse/frequency/switch output.

Description

Option for changing the diagnostic behavior of the diagnostic message 442 Freq. output
1 to n.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

WarningEndress+Hauser

35Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Detailed description of the options available for selection: →  31

Diagnostic no. 443 (Pulse output 1 to n)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 443 (0659)

Prerequisite

The measuring device has a pulse/frequency/switch output.

Description

Option for changing the diagnostic behavior of the diagnostic message 443 Pulse output
1 to n.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 444 (Current input 1 to n)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 444 (0740)

Prerequisite

The device has one current input.

Description

Option for changing the diagnostic behavior of the diagnostic message 444 Current input
1 to n.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 830 (Sensor temp.)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 830 (0800)

Description

Option for changing the diagnostic behavior of the diagnostic message 830 Sensor temp..

36

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Description of device parameters

Detailed description of the options available for selection: →  31

Diagnostic no. 831 (Sensor temp.)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 831 (0641)

Description

Option for changing the diagnostic behavior of the diagnostic message 831 Sensor temp..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 832 (Electronic temp.)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 832 (0681)

Description

Option for changing the diagnostic behavior of the diagnostic message 832 Electronic
temp..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Logbook only

Additional information

Endress+Hauser

Detailed description of the options available for selection: →  31

37

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Diagnostic no. 833 (Electronic temp.)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 833 (0682)

Description

Option for changing the diagnostic behavior of the diagnostic message 833 Electronic
temp..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Logbook only

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 834 (Process temp.)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 834 (0700)

Description

Option for changing the diagnostic behavior of the diagnostic message 834 Process temp..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 835 (Process temp.)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 835 (0702)

Description

Option for changing the diagnostic behavior of the diagnostic message 835 Process temp..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

38

Detailed description of the options available for selection: →  31

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Diagnostic no. 862 (Empty pipe)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 862 (0679)

Description

Option for changing the diagnostic behavior of the diagnostic message 862 Empty pipe.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 912 (Medium inhomog.)



Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 912 (0703)

Description

Option for changing the diagnostic behavior of the diagnostic message 912 Medium
inhomog..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 913 (Medium unsuitab.)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 913 (0712)

Description

Option for changing the diagnostic behavior of the diagnostic message 913 Medium
unsuitab..

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Endress+Hauser

Detailed description of the options available for selection: →  31

39

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Diagnostic no. 944 (MonitoringFailed)

Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 944 (0732)

Description

Option for changing the diagnostic behavior of the diagnostic message
944 MonitoringFailed.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Diagnostic no. 948 (Oscill. damping)Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Diagnostic no. 948 (0744)

Description

Option for changing the diagnostic behavior of the diagnostic message 948 Oscill.
damping.

Selection• Off• Alarm• Warning• Logbook only

Factory setting

Warning

Additional information

Detailed description of the options available for selection: →  31

Status diag. 046 (Sensor limit)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 046 (11042)

Description

Use this function to change the status of diagnostic message 046 Sensor limit.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information40

Detailed description of the options available for selection: →  32

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Status diag. 140 (Sensor sig.asym.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 140 (11041)

Description

Use this function to change the status of diagnostic message 140 Sensor sig.asym..

Selection• Bad• Uncertain• GoodFactory setting

Bad

Additional information

Detailed description of the options available for selection: →  32

Status diag. 144 (MeasErrorTooHigh)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 144 (11016)

Description

Use this function to change the status of diagnostic message 144 MeasErrorTooHigh.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 374 (Sensor electron.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 374 (11000)

Description

Use this function to change the status of diagnostic message S374 Sensor electron..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Endress+Hauser

Detailed description of the options available for selection: →  32

41

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Status diag. 830 (Sensor temp.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 830 (11006)

Description

Use this function to change the status of diagnostic message 830 Sensor temp..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 831 (Sensor temp.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 831 (11007)

Description

Use this function to change the status of diagnostic message 831 Sensor temp..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 832 (Electronic temp.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 832 (11002)

Description

Option for changing the status of the diagnostic message 832 Electronic temp..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

42

Detailed description of the options available for selection: →  32

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Status diag. 833 (Electronic temp.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 833 (11003)

Description

Option for changing the status of the diagnostic message 833 Electronic temp..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 834 (Process temp.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 834 (11004)

Description

Use this function to change the status of diagnostic message 834 Process temp..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 835 (Process temp.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 835 (11005)

Description

Use this function to change the status of diagnostic message 835 Process temp..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Endress+Hauser

Detailed description of the options available for selection: →  32

43

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Status diag. 862 (Empty pipe)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 862 (11008)

Description

Use this function to change the status of diagnostic message 862 Empty pipe.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 912 (Medium inhomog.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 912 (11009)

Description

Use this function to change the status of diagnostic message 912 Medium inhomog..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 913 (Medium unsuitab.)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 913 (11010)

Description

Use this function to change the status of diagnostic message 913 Medium unsuitab..

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

44

Detailed description of the options available for selection: →  32

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Status diag. 944 (MonitoringFailed)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 944 (11014)

Description

Use this function to change the status of diagnostic message 944 MonitoringFailed.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

Status diag. 948 (Oscill. damping)
Navigation

 Expert → System → Diagn. handling → Diagn. behavior → Status diag. 948 (11011)

Description

Use this function to change the status of diagnostic message 948 Oscill. damping.

Selection• Bad• Uncertain• GoodFactory setting

Uncertain

Additional information

Detailed description of the options available for selection: →  32

3.1.4

"Administration" submenu

Navigation

 Expert → System → Administration‣ Administration

Endress+Hauser‣ Def. access code→  46‣ Reset acc. code→  47

Device reset (0000)→  48

Activate SW opt. (0029)→  48SW option overv. (0015)→  4945

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus"Def. access code" wizard
The Def. access code wizard (→  46) is only available when operating via the local
display or Web browser.
If operating via the operating tool, the Def. access code parameter can be founddirectly in the Administration submenu. There is no Confirm code parameter if thedevice is operated via the operating tool.
Navigation

Expert → System → Administration → Def. access code

‣ Def. access codeDef. access code→  46

Confirm code→  47

Def. access code

Navigation

 Expert → System → Administration → Def. access code → Def. access code

Description

Use this function to enter a user-specific release code to restrict write-access to theparameters. This protects the device configuration against any inadvertent modifications
via the local display, Web browser, FieldCare or DeviceCare (via CDI-RJ45 service
interface).

User entry

Max. 16-digit character string comprising numbers, letters and special characters

Additional information

Description
The write protection affects all parameters in the document marked with the  symbol.
On the local display, the  symbol in front of a parameter indicates that the parameter iswrite-protected.
The parameters that cannot be write-accessed are grayed out in the Web browser.
Once the access code has been defined, write-protected parameters can only be
modified if the access code is entered in the Ent. access code parameter (→  13).
If you lose the access code, please contact your Endress+Hauser sales organization.
User entry
A message is displayed if the access code is not in the input range.
Factory setting
If the factory setting is not changed or 0 is defined as the access code, the parameters are
not write-protected and the device configuration data can be modified. The user is logged
on in the "Maintenance" role.

46

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Confirm code

Navigation

 Expert → System → Administration → Def. access code → Confirm code

Description

Enter the defined release code a second time to confirm the release code.

User entry

Max. 16-digit character string comprising numbers, letters and special characters"Reset access code" submenu
Navigation

 Expert → System → Administration → Reset acc. code

‣ Reset acc. code
Operating time (0652)→  47

Reset acc. code (0024)→  47

Operating time
Navigation

 Expert → System → Administration → Reset acc. code → Operating time (0652)

Description

Use this function to display the length of time the device has been in operation.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The maximum number of days is 9999, which is equivalent to 27 years.

Reset acc. code
Navigation

 Expert → System → Administration → Reset acc. code → Reset acc. code (0024)

Description

Use this function to enter a reset code to reset the user-specific release code to the factory
setting.

User entry

Character string comprising numbers, letters and special characters

Factory setting

0x00

Endress+Hauser

47

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
For a reset code, contact your Endress+Hauser service organization.
User entry
The reset code can only be entered via:
• Web browser• DeviceCare, FieldCare (via interface CDI RJ45)• Fieldbus

Additional parameters in the "Administration" submenu

Device reset

Navigation

 Expert → System → Administration → Device reset (0000)

Description

Use this function to choose whether to reset the device configuration - either entirely or inpart - to a defined state.

Selection• Cancel• To delivery set.
• Restart device• ENP restart

Factory setting

CancelAdditional information

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

To delivery set.

Every parameter for which a customer-specific default setting was ordered is resetto this customer-specific value. All other parameters are reset to the factory
setting.

Restart device

The restart resets every parameter whose data are in the volatile memory (RAM)to the factory setting (e.g. measured value data). The device configuration remains
unchanged.

ENP restart

The parameters of the electronic name plate are reset.
The device is restarted.

Activate SW opt.



Navigation

 Expert → System → Administration → Activate SW opt. (0029)

Description

Use this function to enter an activation code to enable an additional, ordered software
option.

48

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User entry

Max. 10-digit string consisting of numbers.

Factory setting

Depends on the software option ordered

Additional information

Description

Description of device parameters

If a measuring device was ordered with an additional software option, the activation codeis programmed in the device at the factory.
User entry
To activate a software option subsequently, please contact your Endress+Hauser salesorganization.
NOTE!
The activation code is linked to the serial number of the measuring device and varies
according to the device and software option.
If an incorrect or invalid code is entered, this results in the loss of software options thathave already been activated.
▸ Before you enter a new activation code, make a note of the current activation code .
▸ Enter the new activation code provided by Endress+Hauser when the new software
option was ordered.
▸ Once the activation code has been entered, check if the new software option is displayedin the SW option overv. parameter (→  49).
 The new software option is active if it is displayed.
 If the new software option is not displayed or all software options have been deleted,the code entered was either incorrect or invalid.
▸ If the code entered is incorrect or invalid, enter the old activation code .
▸ Have your Endress+Hauser sales organization check the new activation code
remembering to specify the serial number or ask for the code again.
Example for a software option
Order code for "Application package", option EA "Extended HistoROM"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).
Web browser
Once a software option has been activated, the page must be loaded again in the Webbrowser.

SW option overv.
Navigation

 Expert → System → Administration → SW option overv. (0015)

Description

Displays all the software options that are enabled in the device.

User interface• Extend. HistoROM• HBT Verification• HBT Monitoring• Concentration• Viscosity

Endress+Hauser

49

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
Displays all the options that are available if ordered by the customer.
"Extend. HistoROM" option
Order code for "Application package", option EA "Extended HistoROM""HBT Verification" option and "HBT Monitoring" option
Order code for "Application package", option EB "Heartbeat Verification + Monitoring""Concentration" option
Order code for "Application package", option ED "Concentration" and option EE "Special
density""Viscosity" option
Only available for Promass I.
Order code for "Application package", option EG "Viscosity"

3.2
Navigation"Sensor" submenu
 Expert → Sensor‣ Sensor50‣ Measured val.

→  51‣ System units→  66‣ Process param.

→  74‣ Measurement mode→  82‣ External comp.

→  84‣ Calculated value→  87‣ Sensor adjustm.

→  90‣ Calibration→  97

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

3.2.1

Description of device parameters"Measured val." submenu

Navigation

 Expert → Sensor → Measured val.

‣ Measured val.
‣ Process variab.

→  51‣ Totalizer→  59‣ Input values→  60‣ Output values→  62"Process variab." submenu
Navigation

 Expert → Sensor → Measured val. → Process variab.

‣ Process variab.

Endress+Hauser

Mass flow (1838)→  52

Volume flow (1847)→  52

Correct.vol.flow (1851)→  52

Density (1850)→  53

Ref.density (1852)→  53Temperature (1853)→  53Pressure value (6129)→  54

Dynam. viscosity (1854)→  54

Kinematic visc. (1857)→  54

TempCompDynVisc (1872)→  55

TempCompKinVisc (1863)→  55

Concentration (1887)→  55

Target mass flow (1864)→  56

Carrier mass fl. (1865)→  56

Targ.corr.vol.fl (1893)→  5751

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Carr.corr.vol.fl (1894)→  57

Target vol. flow (1895)→  58Carrier vol. fl. (1896)→  58Mass flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Mass flow (1838)

Description

Displays the mass flow that is currently measured.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Mass flow unit parameter (→  66)

Volume flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Volume flow (1847)

Description

Displays the volume flow currently calculated.

User interface

Signed floating-point number

Additional information

Description
The volume flow is calculated from the mass flow currently measured and the density
currently measured.
Dependency
The unit is taken from the Volume flow unit parameter (→  68)

Correct.vol.flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Correct.vol.flow (1851)

Description

Displays the corrected volume flow currently measured.

User interface

Signed floating-point number

52

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Dependency
The unit is taken from the Cor.volflow unit parameter (→  69)

Density
Navigation

 Expert → Sensor → Measured val. → Process variab. → Density (1850)

Description

Displays the density currently measured.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Density unit parameter (→  71)

Ref.density
Navigation

 Expert → Sensor → Measured val. → Process variab. → Ref.density (1852)

Description

Displays the reference density currently calculated.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Ref. dens. unit parameter (→  71)

Temperature
Navigation

 Expert → Sensor → Measured val. → Process variab. → Temperature (1853)

Description

Displays the medium temperature currently measured.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)

Endress+Hauser

53Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Pressure value
Navigation

 Expert → Sensor → Measured val. → Process variab. → Pressure value (6129)

Description

Displays the fixed or external pressure value.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Pressure unit parameter (→  73)

Dynam. viscosity
Navigation

 Expert → Sensor → Measured val. → Process variab. → Dynam. viscosity (1854)

Prerequisite

For the following order code:"Application package", option EG "Viscosity"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the dynamic viscosity currently calculated.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Dyn. visc. unit parameter.

Kinematic visc.
Navigation

 Expert → Sensor → Measured val. → Process variab. → Kinematic visc. (1857)

Prerequisite

For the following order code:"Application package", option EG "Viscosity"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the kinematic viscosity currently calculated.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Kin. visc. unit parameter (0578).

54

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

TempCompDynVisc
Navigation

 Expert → Sensor → Measured val. → Process variab. → TempCompDynVisc (1872)

Prerequisite

For the following order code:"Application package", option EG "Viscosity"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the temperature compensation currently calculated for the viscosity.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Dyn. visc. unit parameter.

TempCompKinVisc
Navigation

 Expert → Sensor → Measured val. → Process variab. → TempCompKinVisc (1863)

Prerequisite

For the following order code:"Application package", option EG "Viscosity"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the temperature compensation currently calculated for the kinetic viscosity.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Kin. visc. unit parameter (0578).

Concentration
Navigation

 Expert → Sensor → Measured val. → Process variab. → Concentration (1887)

Prerequisite

For the following order code:
Order code for "Application package", option ED "Concentration"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the concentration currently calculated.

User interface

Signed floating-point number

Endress+Hauser

55Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Dependency
The unit is taken from the Concentr. unit parameter (0613).

Target mass flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Target mass flow (1864)

Prerequisite

With the following conditions:
Order code for "Application package", option ED "Concentration"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the mass flow currently measured for the target medium.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Mass flow unit parameter (→  66)

Carrier mass fl.
Navigation

 Expert → Sensor → Measured val. → Process variab. → Carrier mass fl. (1865)

Prerequisite

With the following conditions:
Order code for "Application package", option ED "Concentration"
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the mass flow currently measured for the carrier medium.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Mass flow unit parameter (→  66)

56

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Targ.corr.vol.fl
Navigation

 Expert → Sensor → Measured val. → Process variab. → Targ.corr.vol.fl (1893)

Prerequisite

With the following conditions:• Order code for "Application package", option ED "Concentration"• In the Liquid type parameter, the Ethanol in water option or %mass / %volume option
is selected.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the corrected volume flow currently measured for the target fluid.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Volume flow unit parameter (→  68)

Carr.corr.vol.fl
Navigation

 Expert → Sensor → Measured val. → Process variab. → Carr.corr.vol.fl (1894)

Prerequisite

With the following conditions:• Order code for "Application package", option ED "Concentration"• In the Liquid type parameter, the Ethanol in water option or %mass / %volume option
is selected.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the corrected volume flow currently measured for the carrier fluid.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Volume flow unit parameter (→  68)

Endress+Hauser

57Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Target vol. flow
Navigation

 Expert → Sensor → Measured val. → Process variab. → Target vol. flow (1895)

Prerequisite

With the following conditions:• Order code for "Application package", option ED "Concentration"• The Ethanol in water option or %mass / %volume option is selected in the Liquid typeparameter.
• The %vol option is selected in the Concentr. unit parameter.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Displays the volume flow currently measured for the target medium.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Volume flow unit parameter (→  68)

Carrier vol. fl.
Navigation

 Expert → Sensor → Measured val. → Process variab. → Carrier vol. fl. (1896)

Prerequisite

With the following conditions:• Order code for "Application package", option ED "Concentration"• The Ethanol in water option or %mass / %volume option is selected in the Liquid typeparameter.
• The %vol option is selected in the Concentr. unit parameter.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Use this function to display the volume flow currently measured for the carrier medium.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Volume flow unit parameter (→  68)

58

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters"Totalizer" submenu
Navigation

 Expert → Sensor → Measured val. → Totalizer‣ Totalizer
Totalizer val. 1 to n (0911–1 to n)→  59Tot. overflow 1 to n (0910–1 to n)→  59Totalizer val. 1 to n

Navigation

 Expert → Sensor → Measured val. → Totalizer → Totalizer val. 1 to n (0911–1 to n)

Prerequisite

A process variable is selected in the Assign variable parameter (→  276) of the
Totalizer 1 to n submenu.

Description

Displays the current totalizer reading.

User interface

Signed floating-point number

Additional information

Description
As it is only possible to display a maximum of 7 digits in the operating tool, the current
counter value is the sum of the totalizer value and the overflow value from the Tot.
overflow 1 to n parameter if the display range is exceeded.
In the event of an error, the totalizer adopts the mode defined in the Failure modeparameter (→  279).
User interface
The value of the process variable totalized since measuring began can be positive or
negative. This depends on the settings in the Operation mode parameter (→  277).
The unit of the selected process variable is specified for the totalizer in the Unit
totalizer parameter (→  276).
ExampleCalculation of the current totalizer reading when the value exceeds the 7-digit display
range of the operating tool:• Value in the Totalizer val. 1 parameter: 1 968 457 m3• Value in the Tot. overflow 1 parameter: 1 ⋅ 107 (1 overflow) = 10 000 000 [m3]• Current totalizer reading: 11 968 457 m3

Tot. overflow 1 to n

Navigation

 Expert → Sensor → Measured val. → Totalizer → Tot. overflow 1 to n (0910–1 to n)

Prerequisite

A process variable is selected in the Assign variable parameter (→  276) of the
Totalizer 1 to n submenu.

Endress+Hauser

59Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Description

Displays the current totalizer overflow.

User interface

Integer with sign

Additional information

Description
If the current totalizer reading exceeds 7 digits, which is the maximum value range thatcan be displayed by the operating tool, the value above this range is output as an overflow.
The current totalizer value is therefore the sum of the overflow value and the totalizer
value from the Totalizer val. 1 to n parameter.
User interface
The unit of the selected process variable is specified for the totalizer in the Unit
totalizer parameter (→  276).
ExampleCalculation of the current totalizer reading when the value exceeds the 7-digit display
range of the operating tool:• Value in the Totalizer val. 1 parameter: 1 968 457 m3• Value in the Tot. overflow 1 parameter: 2 ⋅ 107 (2 overflows) = 20 000 000 [m3]• Current totalizer reading: 21 968 457 m3"Input values" submenu
Navigation

 Expert → Sensor → Measured val. → Input values‣ Input values‣ Current input 1 to n

→  60‣ Val.stat.inp. 1 to n

→  61"Current input 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Input values → Current
input 1 to n

‣ Current input 1 to n

60

Measured val. 1 to n (1603–1 to n)→  61

Measur. curr. 1 to n (1604–1 to n)→  61

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Measured val.
Navigation

Expert → Sensor → Measured val. → Input values → Current input 1 to n
→ Measured val. 1 to n (1603–1 to n)

Description

Displays the current input value.

User interface

Signed floating-point number

Measur. curr.
Navigation

Expert → Sensor → Measured val. → Input values → Current input 1 to n → Measur.
curr. 1 to n (1604–1 to n)

Description

Displays the current value of the current input.

User interface

0 to 22.5 mA"Value status input 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Input values → Val.stat.inp.
1 to n‣ Val.stat.inp. 1 to nVal.stat.inp. (1353–1 to n)→  61Val.stat.inp.
NavigationDescription

Displays the current input signal level.

User interface• High• Low

Endress+Hauser

Expert → Sensor → Measured val. → Input values → Val.stat.inp. 1 to n
→ Val.stat.inp. (1353–1 to n)

61

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus"Output values" submenu
Navigation

 Expert → Sensor → Measured val. → Output values‣ Output values‣ Value curr.out 1 to n

→  62‣ PFS output 1 to n

→  63‣ Relay output 1 to n

→  65"Value current output 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Output values → Valuecurr.out 1 to n

‣ Value curr.out 1 to n
Output curr. 1 to n (0361–1 to n)→  62

Measur. curr. 1 to n (0366–1 to n)→  62

Output curr.
Navigation

Expert → Sensor → Measured val. → Output values → Value curr.out 1 to n → Output
curr. 1 to n (0361–1 to n)

Expert → Output → Curr.output 1 to n → Output curr. 1 to n (0361–1 to n)Diagnostics → Measured val. → Output values → Curr.output 1 to n → Output curr.
1 to n (0361–1 to n)

Description

Displays the current value currently calculated for the current output.

User interface

0 to 22.5 mA

Measur. curr.
Navigation

Description

62



Expert → Sensor → Measured val. → Output values → Value curr.out 1 to n
→ Measur. curr. 1 to n (0366–1 to n)

Expert → Output → Curr.output 1 to n → Measur. curr. 1 to n (0366–1 to n)Diagnostics → Measured val. → Output values → Curr.output 1 to n → Measur. curr.
1 to n (0366–1 to n)

Use this function to display the actual measured value of the output current.

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User interface

Description of device parameters

0 to 30 mA"Pulse/frequency/switch output 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Output values → PFS output
1 to n‣ PFS output 1 to n
Output freq. 1 to n (0471–1 to n)→  63

Pulse output 1 to n (0456–1 to n)→  63

Switch status 1 to n (0461–1 to n)→  64

Output freq.
Navigation

Expert → Sensor → Measured val. → Output values → PFS output 1 to n → Output
freq. 1 to n (0471–1 to n)

Expert → Output → PFS output 1 to n → Output freq. 1 to n (0471–1 to n)Diagnostics → Measured val. → Output values → PFS output 1 to n → Output freq.
1 to n (0471–1 to n)

Prerequisite

In the Operating mode parameter (→  123), the Frequency option is selected.

Description

Displays the actual value of the output frequency which is currently measured.

User interface

0.0 to 12 500.0 Hz

Pulse output
Navigation

Expert → Sensor → Measured val. → Output values → PFS output 1 to n → Pulse
output 1 to n (0456–1 to n)

Expert → Output → PFS output 1 to n → Pulse output 1 to n (0456–1 to n)Diagnostics → Measured val. → Output values → PFS output 1 to n → Pulse output
1 to n (0456–1 to n)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  123) parameter.

Description

Displays the pulse frequency currently output.

User interface

Positive floating-point number

Endress+Hauser

63Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description• The pulse output is an open collector output.
• This is configured at the factory in such a way that the transistor is conductive for theduration of the pulse (NO contact) and is safety-oriented.

1

1

t

0

t

0

NC

NO
A0028726

0
1
NC
NO

Non-conductive
Conductive
NC contact (normally closed)
NO contact (normally open)

The output behavior can be reversed via the Invert outp.sig. parameter (→  142) i.e.
the transistor does not conduct for the duration of the pulse.
In addition, the behavior of the output in the event of a device alarm (Failure modeparameter (→  127)) can be configured.

Switch status
Navigation

Expert → Sensor → Measured val. → Output values → PFS output 1 to n → Switch
status 1 to n (0461–1 to n)

Expert → Output → PFS output 1 to n → Switch status 1 to n (0461–1 to n)Diagnostics → Measured val. → Output values → PFS output 1 to n → Switch status
1 to n (0461–1 to n)

Prerequisite

The Switch option is selected in the Operating mode parameter (→  123).

Description

Displays the current switch status of the status output.

User interface• Open• Closed

Additional information

User interface• OpenThe switch output is not conductive.
• Closed
The switch output is conductive.

64

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters"Relay output 1 to n" submenu
Navigation

 Expert → Sensor → Measured val. → Output values → Relay
output 1 to n

‣ Relay output 1 to n
Switch status (0801–1 to n)→  65

Switch cycles (0815–1 to n)→  65Max. cycles no. (0817–1 to n)→  66

Switch status
Navigation

Expert → Sensor → Measured val. → Output values → Relay output 1 to n → Switch
status (0801–1 to n)

Expert → Output → Relay output 1 to n → Switch status (0801–1 to n)Diagnostics → Measured val. → Output values → Relay output 1 to n → Switch status(0801–1 to n)

Description

Displays the current status of the relay output.

User interface• Open• Closed

Additional information

User interface• OpenThe relay output is not conductive.
• Closed
The relay output is conductive.

Switch cycles
Navigation

Expert → Sensor → Measured val. → Output values → Relay output 1 to n → Switch
cycles (0815–1 to n)Diagnostics → Measured val. → Output values → Relay output 1 to n → Switch cycles(0815–1 to n)

Description

Displays all the switch cycles performed.

User interface

Positive integer

Endress+Hauser

65Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Max. cycles no.
Navigation

Expert → Sensor → Measured val. → Output values → Relay output 1 to n → Max.
cycles no. (0817–1 to n)Diagnostics → Measured val. → Output values → Relay output 1 to n → Max. cycles
no. (0817–1 to n)

Description

Displays the maximum number of guaranteed switch cycles.

User interface

Positive integer

3.2.2"System units" submenu

Navigation

 Expert → Sensor → System units‣ System units
Mass flow unit (0554)→  66Mass unit (0574)→  67

Volume flow unit (0553)→  68Volume unit (0563)→  69Cor.volflow unit (0558)→  69Corr. vol. unit (0575)→  70Density unit (0555)→  71

Ref. dens. unit (0556)→  71

Temperature unit (0557)→  72Pressure unit (0564)→  73Date/time format (2812)→  73Mass flow unitNavigation

 Expert → Sensor → System units → Mass flow unit (0554)

Description

Use this function to select the unit for the mass flow.

66

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d

Factory setting

Country-specific:• kg/h (DN > 150 (6"): t/h)
• lb/min

Additional information

ResultDescription of device parameters

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d

The selected unit applies for:• Target mass flow parameter (→  56)• Carrier mass fl. parameter (→  56)• Mass flow parameter (→  52)
Selection
For an explanation of the abbreviated units: →  334

Mass unitNavigation

Expert → Sensor → System units → Mass unit (0574)

Description

Use this function to select the unit for the mass.

Selection

SI units
•g
• kg
•t

Factory setting

Country-specific:• kg (DN > 150 (6"): t)• lb

Additional information

Selection

US units• oz• lb• STonFor an explanation of the abbreviated units: →  334

Endress+Hauser

67Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Volume flow unitNavigation

 Expert → Sensor → System units → Volume flow unit (0553)

Description

Use this function to select the unit for the volume flow.

Selection

SI units• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• m³/s• m³/min• m³/h• m³/d• ml/s• ml/min• ml/h• ml/d• l/s• l/min• l/h• l/d• hl/s• hl/min• hl/h• hl/d• Ml/s• Ml/min• Ml/h• Ml/dFactory setting

Country-specific:• l/h (DN > 150 (6"): m³/h)• gal/min (us)68

US units• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)

Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

ResultThe selected unit applies for:Volume flow parameter (→  52)
Selection
For an explanation of the abbreviated units: →  334

Volume unitNavigation

Expert → Sensor → System units → Volume unit (0563)

Description

Use this function to select the unit for the volume.

Selection

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega

Factory setting

Country-specific:• l (DN > 150 (6"): m³)• gal (us)Additional information

Selection

US units• af• ft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;beer)• bbl (us;tank)

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

For an explanation of the abbreviated units: →  334

Cor.volflow unitNavigation

 Expert → Sensor → System units → Cor.volflow unit (0558)

Description

Use this function to select the unit for the corrected volume flow.

Endress+Hauser

69Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection

SI units• Nl/s• Nl/min• Nl/h• Nl/d• Nm³/s• Nm³/min• Nm³/h• Nm³/d• Sm³/s• Sm³/min• Sm³/h• Sm³/dUS units• Sft³/s• Sft³/min• Sft³/h• Sft³/d• Sgal/s (us)• Sgal/min (us)• Sgal/h (us)• Sgal/d (us)• Sbbl/s (us;liq.)• Sbbl/min (us;liq.)• Sbbl/h (us;liq.)• Sbbl/d (us;liq.)

Factory setting

Country-specific:• Nl/h (DN > 150 (6"): Nm³/h)• Sft³/min

Additional information

ResultImperial units• Sgal/s (imp)• Sgal/min (imp)• Sgal/h (imp)• Sgal/d (imp)

The selected unit applies for:
Correct.vol.flow parameter (→  52)
Selection
For an explanation of the abbreviated units: →  334

Corr. vol. unitNavigation

Expert → Sensor → System units → Corr. vol. unit (0575)

Description

Use this function to select the unit for the corrected volume.

Selection

SI units
• Nl
• Nm³
• Sm³

Factory setting

Country-specific:• Nl (DN > 150 (6"): Nm³)
• Sft³

Additional information

Selection

US units• Sft³• Sgal (us)• Sbbl (us;liq.)

Imperial units
Sgal (imp)

For an explanation of the abbreviated units: →  334

70

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Density unitNavigation

 Expert → Sensor → System units → Density unit (0555)

Description

Use this function to select the unit for the density.

Selection

SI units• g/cm³• g/m³• kg/dm³• kg/l• kg/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG15°C• SG20°C

Factory setting

Country-specific:• kg/l• lb/ft³

Additional information

ResultUS units• lb/ft³• lb/gal (us)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/bbl (us;tank)

Imperial units• lb/gal (imp)• lb/bbl (imp;beer)• lb/bbl (imp;oil)The selected unit applies for:• Density setpt 1 parameter• Density setpt 2 parameter• Density parameter (→  53)
Selection• SD = specific density
The specific density is the ratio of the medium density to the water density at a watertemperature of +4 °C (+39 °F), +15 °C (+59 °F), +20 °C (+68 °F).
• SG = specific gravity
The specific gravity is the ratio of the medium density to the water density at a watertemperature of +4 °C (+39 °F), +15 °C (+59 °F), +20 °C (+68 °F).
For an explanation of the abbreviated units: →  334

Ref. dens. unitNavigation

 Expert → Sensor → System units → Ref. dens. unit (0556)

Description

Use this function to select the unit for the reference density.

Selection

SI units• kg/Nm³• kg/Nl• g/Scm³• kg/Sm³

Endress+Hauser

US units
lb/Sft³71

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Country-dependent• kg/Nl• lb/Sft³

Additional information

ResultThe selected unit applies for:• Ext. ref.density parameter (→  88)• Fix ref.density parameter (→  88)• Ref.density parameter (→  53)
Selection
For an explanation of the abbreviated units: →  334

Temperature unitNavigation

 Expert → Sensor → System units → Temperature unit (0557)

Description

Use this function to select the unit for the temperature.

Selection

SI units• °C•KFactory setting

Country-specific:
• °C
• °F

Additional information

ResultUS units• °F• °R

The selected unit applies for:• Maximum value parameter (→  300)
• Minimum value parameter (→  300)
• Maximum value parameter (→  301)
• Minimum value parameter (→  301)
• Maximum value parameter (→  302)
• Minimum value parameter (→  302)
• External temp. parameter (→  86)• Ref. temperature parameter (6222)• Temperature parameter (→  53)• Ref. temperature parameter (→  89)
Selection
For an explanation of the abbreviated units: →  334

72

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Pressure unitNavigation

 Expert → Sensor → System units → Pressure unit (0564)

Description

Use this function to select the unit for the pipe pressure.

Selection

SI units• Pa a• kPa a• MPa a• bar
• Pa g• kPa g• MPa g• bar gFactory setting

Country-specific:• bar a• psi aAdditional information

ResultUS units• psi a• psi g

The unit is taken from:• Pressure value parameter (→  85)• External press. parameter (→  85)• Pressure value parameter (→  54)
Selection
For an explanation of the abbreviated units: →  334

Date/time format



Navigation

 Expert → Sensor → System units → Date/time format (2812)

Description

Use this function to select the desired time format for calibration history.

Selection• dd.mm.yy hh:mm• dd.mm.yy am/pm• mm/dd/yy hh:mm• mm/dd/yy am/pm

Factory setting

dd.mm.yy hh:mm

Additional information

Selection
For an explanation of the abbreviated units: →  334

Endress+Hauser

73Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

3.2.3"Process param." submenu

Navigation

 Expert → Sensor → Process param.

‣ Process param.
Flow damping (1802)→  74

Density damping (1803)→  75

Temp. damping (1822)→  75

Flow override (1839)→  76‣ Low flow cut off→  76‣ Partial pipe det→  79Flow damping

Navigation

 Expert → Sensor → Process param. → Flow damping (1802)

Description

Use this function to enter a time constant for flow damping (PT1 element). Reduction of
the variability of the flow measured value (in relation to interference). For this purpose,the depth of the flow filter is adjusted: when the filter setting increases, the reaction timeof the device also increases.

User entry

0 to 100.0 s

Factory setting

0s

Additional information

Description
The damping is performed by a PT1 element 2).
User entry• Value = 0: no damping• Value > 0: damping is increased
Damping is switched off if 0 is entered (factory setting).
Result
The damping affects the following variables of the device:• Outputs• Low flow cut off →  76
• Totalizers →  275

2)

74

Proportional behavior with first-order lag

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Density damping

Navigation

 Expert → Sensor → Process param. → Density damping (1803)

Description

Use this function to enter a time constant for the damping (PT1 element) of the density
measured value.

User entry

0 to 999.9 s

Factory setting

0s

Additional information

Description
The damping is performed by a PT1 element 3).
User entry• Value = 0: no damping• Value > 0: damping is increased
Damping is switched off if 0 is entered (factory setting).

Temp. damping

Navigation

 Expert → Sensor → Process param. → Temp. damping (1822)

Description

Use this function to enter a time constant for the damping (PT1 element) of thetemperature measured value.

User entry

0 to 999.9 s

Factory setting

0s

Additional information

Description
The damping is performed by a PT1 element 4).
User entry• Value = 0: no damping• Value > 0: damping is increased
Damping is switched off if 0 is entered (factory setting).

3)
4)

Proportional behavior with first-order lag
Proportional behavior with first-order lag

Endress+Hauser

75Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Flow override

Navigation

 Expert → Sensor → Process param. → Flow override (1839)

Description

Use this function to select whether to interrupt the evaluation of measured values. This isuseful for the cleaning processes of a pipeline, for example.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
Flow override is active• The diagnostic message diagnostic message C453 Flow override is displayed.
• Output values– Temperature: continues to be output– Totalizers 1-3: stop being totalized
The Flow override option can also be activated in the Status input submenu: Assign
stat.inp. parameter (→  105).

"Low flow cut off" submenu
Navigation

 Expert → Sensor → Process param. → Low flow cut off‣ Low flow cut off
Assign variable (1837)→  76

On value (1805)→  77

Off value (1804)→  77

Pres. shock sup. (1806)→  78Assign variable

Navigation

 Expert → Sensor → Process param. → Low flow cut off → Assign variable (1837)

Description

Use this function to select the process variable for low flow cutoff detection.

Selection• Off• Mass flow
• Volume flow
• Correct.vol.flow

76

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Description of device parameters

Mass flow

On value

Navigation

 Expert → Sensor → Process param. → Low flow cut off → On value (1805)

Prerequisite

A process variable is selected in the Assign variable parameter (→  76).

Description

Use this function to enter a switch-on value for low flow cut off. Low flow cut off isactivated if the value entered is not equal to 0 →  77.

User entry

Positive floating-point number

Factory setting

Depends on country and nominal diameter→  329

Additional information

Dependency
The unit depends on the process variable selected in the Assign variable parameter(→  76).

Off valueNavigation

 Expert → Sensor → Process param. → Low flow cut off → Off value (1804)

Prerequisite

A process variable is selected in the Assign variable parameter (→  76).

Description

Use this function to enter a switch-off value for low flow cut off. The off value is entered asa positive hysteresis from the on value→  77.

User entry

0 to 100.0 %Factory setting

50 %Additional information

Example
Q

2

2

4
H

3

t
1

A

1

A
A0012887

Q
t
H
A
1
2
3
4

Endress+Hauser

Flow
Time
Hysteresis
Low flow cut off active
Low flow cut off is activated
Low flow cut off is deactivatedOn value entered
Off value entered77

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Pres. shock sup.



Navigation

 Expert → Sensor → Process param. → Low flow cut off → Pres. shock sup. (1806)

Prerequisite

A process variable is selected in the Assign variable parameter (→  76).

Description

Use this function to enter the time interval for signal suppression (= active pressure shock
suppression).

User entry

0 to 100 s

Factory setting

0s

Additional information

Description
Pressure shock suppression is enabled• Prerequisite:– Flow rate < on-value of low flow cut offor– Changing the flow direction• Output values– Flow displayed: 0– Totalizer: the totalizers are pegged at the last correct value
Pressure shock suppression is disabled• Prerequisite: the time interval set in this function has elapsed.
• If the flow also exceeds the switch-off value for low flow cut off, the device starts
processing the current flow value again and displays it.
ExampleWhen closing a valve, momentarily strong fluid movements may occur in the pipeline,
which are registered by the measuring system. These totalized flow values lead to a false
totalizer status, particularly during batching processes.

78

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

C

D

Q

2

1

D

3

6
5

4

t

A

B
A0012888

Q
t
A
B
C
D
1
2
3
4
5
6

Flow
Time
Drip
Pressure shock
Pressure shock suppression active as specified by the time entered
Pressure shock suppression inactive
Valve closes
Flow falls below the on-value of the low flow cut off: pressure shock suppression is activated
The time entered has elapsed: pressure shock suppression is deactivated
The actual flow value is now displayed and output
On-value for low flow cut off
Off-value for low flow cut off"Partial pipe det" submenu
Navigation

 Expert → Sensor → Process param. → Partial pipe det‣ Partial pipe det

Endress+Hauser

Assign variable (1860)→  80Low value (1861)→  80High value (1858)→  80

Response time (1859)→  81Max. damping (6040)→  8179

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Assign variable

Navigation

 Expert → Sensor → Process param. → Partial pipe det → Assign variable (1860)

Description

Use this function to select a process variable to detect empty or partially filled measuring
tubes.
For gas measurement: Deactivate monitoring due to low gas density.

Selection• Off• Density• Ref.density

Factory settingOff

Low valueNavigation

 Expert → Sensor → Process param. → Partial pipe det → Low value (1861)

Prerequisite

A process variable is selected in the Assign variable parameter (→  80).

Description

Use this function to enter a lower limit value to enable detection of empty or partially filled
measuring tubes. If the measured density falls below this value, monitoring is enabled.

User entry

Signed floating-point number

Factory setting

200

Additional information

User entry
The lower limit value must be less than the upper limit value defined in the High value
parameter (→  80).
The unit depends on the process variable selected in the Assign variable parameter(→  80).
Limit value
If the displayed value is outside the limit value, the measuring device displays thediagnostic message S862 Partly filled.

High value

Navigation

 Expert → Sensor → Process param. → Partial pipe det → High value (1858)

Prerequisite

A process variable is selected in the Assign variable parameter (→  80).

Description

Use this function to enter an upper limit value to enable detection of empty or partiallyfilled measuring tubes. If the measured density exceeds this value, detection is enabled.

User entry

Signed floating-point number

80

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

6 000

Additional information

User entry

Description of device parameters

The upper limit value must be greater than the lower limit value defined in the Low value
parameter (→  80).
The unit depends on the process variable selected in the Assign variable parameter(→  80).
Limit value
If the displayed value is outside the limit value, the measuring device displays thediagnostic message S862 Partly filled.

Response timeNavigation

 Expert → Sensor → Process param. → Partial pipe det → Response time (1859)

Prerequisite

A process variable is selected in the Assign variable parameter (→  80).

Description

Use this function to enter the minimum length of time (debouncing time) the signal must
be present for the diagnostic message S862 Partly filled to be triggered if themeasuring pipe is empty or partially full.

User entry

0 to 100 s

Factory setting

1sMax. damping

Navigation

 Expert → Sensor → Process param. → Partial pipe det → Max. damping (6040)

Description

Use this function to enter a damping value to enable detection of empty or partially filled
measuring tubes.

User entry

Positive floating-point number

Factory setting0

Additional information

Description
If oscillation damping exceeds the specified value, the measuring device presumes that thepipe is partially filled and the flow signal is set to 0. The measuring device displays theEndress+Hauser

81Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

diagnostic message S862 Partly filled. In the case of non-homogeneous media or air
pockets, the damping of the measuring tubes increases.
User entry• Damping is disabled if 0 is entered (factory setting).
• Damping is enabled if the value entered is greater than 0.
• The value entered depends on application-specific influence variables, such as themedium, nominal diameter, sensor etc.
Example• If the pipe is filled normally the value of the oscillation damping is 500.
• If the pipe is partially filled the value of the oscillation damping is > 5000.
• A practical damping value would then be 2000: enter 2000 as the value.

3.2.4
Navigation"Measurement mode" submenu
 Expert → Sensor → Measurement mode

‣ Measurement mode
Select medium (6062)→  82

Select gas type (6074)→  83

Sound velocity (6147)→  83

Temp. coeff. SV (6181)→  84Multi-freq.activ

Navigation

 Expert → Sensor → Measurement mode → Multi-freq.activ (6242)

Description

Switching the dual mode of the sensor on and off.

Selection

• No
• Yes

Factory settingNo

Select medium



Navigation

 Expert → Sensor → Measurement mode → Select medium (6062)

Description

Use this function to select the type of medium.

82

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection

LiquidFactory setting

LiquidDescription of device parameters

Select gas type

Navigation

 Expert → Sensor → Measurement mode → Select gas type (6074)

Prerequisite

The Gas option is selected in the Select medium parameter (→  82).

Description

Use this function to select the type of gas for the measuring application.

Selection• Air• Ammonia NH3• Argon Ar• Sulf. hex.fl.SF6• Oxygen O2• Ozone O3• Nitrog. ox. NOx• Nitrogen N2• Nitrous ox. N2O• Methane CH4• Hydrogen H2• Helium He• Hydrog.chlor.HCl• Hydrog.sulf. H2S• Ethylene C2H4
• Carbon diox. CO2• Carbon monox. CO• Chlorine Cl2• Butane C4H10• Propane C3H8
• Propylene C3H6
• Ethane C2H6
• Others

Factory setting

Methane CH4

Additional information

Description
The gas type needs to be selected so that it is possible to comply with accuracy
specifications in gas applications.

Sound velocity

Navigation

 Expert → Sensor → Measurement mode → Sound velocity (6147)

Prerequisite

In the Select gas type parameter (→  83), the Others option is selected.

Description

Use this function to enter the sound velocity of the gas at 0 °C (+32 °F).

Endress+Hauser

83Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

User entry

1 to 99 999.9999 m/s

Factory setting

415.0 m/s

Temp. coeff. SVNavigation

 Expert → Sensor → Measurement mode → Temp. coeff. SV (6181)

Prerequisite

The Others option is selected in the Select gas type parameter (→  83).

Description

Use this function to enter a temperature coefficient for the sound velocity of the gas.

User entry

Positive floating-point number

Factory setting

0 (m/s)/K

3.2.5"External comp." submenu

Navigation

 Expert → Sensor → External comp.

‣ External comp.
Pressure compen. (6130)→  84Pressure value (6059)→  85External press. (6209)→  85Temperature mode (6184)→  86External temp. (6080)→  86

Pressure compen.



Navigation

 Expert → Sensor → External comp. → Pressure compen. (6130)

Description

Use this function to select the type of pressure compensation.

84

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection• Off• Fixed value• External value• Current input 1 *Factory setting

Off

Additional information

Selection

Description of device parameters• Fixed value
A fixed pressure value is used for compensation: Pressure value parameter (→  85)• External value
The pressure value read in via FOUNDATION Fieldbus is used for compensation.
• Current input 1
The pressure value read in via the current input is used for compensation.
For more information, see the "Cyclic data transmission" section of the Operating
Instructions

Pressure value

Navigation

 Expert → Sensor → External comp. → Pressure value (6059)

Prerequisite

The Fixed value option or the Current input 1…n option is selected in the Pressurecompen. parameter (→  84).

Description

Use this function to enter a value for the process pressure that is used for pressure
correction.

User entry

Positive floating-point number

Factory setting

0 bar

Additional information

User entry
The unit is taken from the Pressure unit parameter (→  73)

External press.
Navigation

 Expert → Sensor → External comp. → External press. (6209)

Prerequisite

The Fixed value option or the Current input 1…n option is selected in the Pressurecompen. parameter (→  84).

Description

Use this function to enter an external pressure value.

User interface

Positive floating-point number

Factory setting

0 bar

*

Visibility depends on order options or device settings

Endress+Hauser

85Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

User entry
The unit is taken from the Pressure unit parameter (→  73)

Temperature modeNavigation

 Expert → Sensor → External comp. → Temperature mode (6184)

Description

Use this function to select the temperature mode.

Selection• Internal value• External value• Current input 1 *Factory setting

Internal value

Additional information

Description
Use this function to select the type of temperature compensation.
Selection
All the options available for selection are used for measured value compensation.
• Internal value
The temperature value measured internally (temperature sensor of the measuring
sensor) is used for compensation.
• External value
The temperature value read in via FOUNDATION Filedbus is used for compensation.
• Current input 1 optionVisibility depends on order options or device settings
The temperature value read in via the current input is used for compensation.
For more information, see the "Cyclic data transmission" section of the Operating
Instructions

External temp.
Navigation

 Expert → Sensor → External comp. → External temp. (6080)

Prerequisite

The External value option or the Current input 1…n option is selected in theTemperature mode parameter (→  86).

Description

Use this function to enter the external temperature.

User interface–273.15 to 99 999 °C

Factory setting

Country-specific:• 0 °C• +32 °F*

86

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
The unit is taken from the Temperature unit parameter (→  72)

3.2.6"Calculated value" submenu

Navigation

 Expert → Sensor → Calculated value‣ Calculated value‣ Corr. vol.flow.

→  87"Corr. vol.flow." submenu
Navigation

 Expert → Sensor → Calculated value → Corr. vol.flow.

‣ Corr. vol.flow.
Corr. vol.flow. (1812)→  87Ext. ref.density (6198)→  88

Fix ref.density (1814)→  88

Ref. temperature (1816)→  89Linear exp coeff (1817)→  89Square exp coeff (1818)→  90Corr. vol.flow.

Navigation

 Expert → Sensor → Calculated value → Corr. vol.flow. → Corr. vol.flow. (1812)

Description

Use this function to select the reference density for calculating the corrected volume flow.

Selection• Fix ref.density• Calc ref density• Ref. dens API 53• Ext. ref.density• Current input 1 **Visibility depends on order options or device settings

Endress+Hauser

87Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Calc ref density

Additional information

Selection
The Ref. dens API 53 option is suitable only for applications involving LPG 5), where theflow rate is measured on the basis of the corrected volume flow.
Selecting this option means that the reference density is used, taking into account thevalues in table 53 E of API MPMS section 11.2. Temperature measurement (measured
internally or read into the device from an external source→  84→  84) and density
measurement take place during operation while the medium is flowing. The mass flow isdivided by the reference density to give the corrected volume flow and is issued as an
output signal.

Ext. ref.density
Navigation

 Expert → Sensor → Calculated value → Corr. vol.flow. → Ext. ref.density (6198)

Description

Displays the reference density which is read in externally, e.g. via the current input.

User interface

Floating point number with sign

Additional information

Dependency
The unit is taken from the Ref. dens. unit parameter (→  71)

Fix ref.density

Navigation

 Expert → Sensor → Calculated value → Corr. vol.flow. → Fix ref.density (1814)

Prerequisite

The Fix ref.density option is selected in the Corr. vol.flow. parameter (→  87)
parameter.

Description

Use this function to enter a fixed value for the reference density.

User entry

Positive floating-point number

Factory setting

1 kg/Nl

Additional information

Dependency
The unit is taken from the Ref. dens. unit parameter (→  71)

5)88

liquefied petroleum gas

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Ref. temperature

Navigation

 Expert → Sensor → Calculated value → Corr. vol.flow. → Ref. temperature (1816)

Prerequisite

The Calc ref density option is selected in the Corr. vol.flow. parameter (→  87)
parameter.

Description

Use this function to enter a reference temperature for calculating the reference density.

User entry–273.15 to 99 999 °C

Factory setting

Country-specific:• +20 °C
• +68 °F

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)
Reference density calculation
ρn = ρ ⋅ (1 + α ⋅ ∆t + β ⋅ ∆t2)
A0023403• ρN: reference density• ρ: fluid density currently measured• t: fluid temperature currently measured• tN: reference temperature at which the reference density is calculated (e.g. 20 °C)• ∆t: t - tN
• α: linear expansion coefficient of the fluid, unit = [1/K]; K = Kelvin• β: square expansion coefficient of the fluid, unit = [1/K2]

Linear exp coeff

Navigation

 Expert → Sensor → Calculated value → Corr. vol.flow. → Linear exp coeff (1817)

Prerequisite

The Calc ref density option is selected in the Corr. vol.flow. parameter (→  87)
parameter.

Description

Use this function to enter a linear, fluid-specific expansion coefficient for calculating thereference density.

User entry

Signed floating-point number

Factory setting

0.0

Endress+Hauser

89Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Square exp coeff

Navigation

 Expert → Sensor → Calculated value → Corr. vol.flow. → Square exp coeff (1818)

Prerequisite

The Calc ref density option is selected in the Corr. vol.flow. parameter (→  87)
parameter.

Description

For fluid with a non-linear expansion pattern: use this function to enter a quadratic, fluidspecific expansion coefficient for calculating the reference density.

User entry

Signed floating-point number

Factory setting

0.0

3.2.7

"Sensor adjustment" submenu

Navigation

 Expert → Sensor → Sensor adjustm.

‣ Sensor adjustm.
Install. direct. (1809)→  90Inst. angle roll (6282)→  91

Inst.angle pitch (6236)→  91‣ Zero point adj.

→  92‣ Variable adjust→  93

Install. direct.



Navigation

 Expert → Sensor → Sensor adjustm. → Install. direct. (1809)

Description

Use this function to change the sign of the medium flow direction.

Selection• In arrow direct.
• Against arrow

Factory setting

In arrow direct.

Additional information

Description
Before changing the sign: ascertain the actual direction of fluid flow with reference tothe direction indicated by the arrow on the sensor nameplate.

90

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Inst. angle roll

Navigation

 Expert → Sensor → Sensor adjustm. → Inst. angle roll (6282)

Prerequisite

Available only with Promass Q.

Description

Use this function to enter the roll installation angle in degrees.

User entry–180 to 180 °Factory setting0°

Additional information

FIGURE: ROLL ANGLE
Roll angle• The roll angle is the angle β from the vertical V to align the central axis Z of thetransmitter.
• The roll angle can be between –180 to +180 °.

Inst.angle pitchNavigation

 Expert → Sensor → Sensor adjustm. → Inst.angle pitch (6236)

Prerequisite

Available only with Promass Q.

Description

Use this function to enter the installation angle pitch in degrees.

User entry–180 to 180 °Factory setting0°

Additional information

FIGURE: PITCH ANGLE
Pitch angle• The pitch angle is the angle α from the horizontal H to align the central axis Z of themeasuring device.
• The pitch angle can be between –90 to +90 °.

Endress+Hauser

91Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus"Zero point adj." submenu• It is generally not necessary to perform zero point adjustment.
• However, this function may be needed in some applications with low flow and strict
accuracy requirements.
• A zero point adjustment cannot increase repeatability.
• The following conditions should be met to perform a zero point adjustment
successfully without the adjustment finishing in an error:– The real flow must be 0.
– The pressure must be at least 15 psi g.
• The adjustment takes a maximum of 60 s. The more stable the conditions, thefaster the adjustment is completed.
• This function can also be used to check the health of the measuring device.
A healthy measuring device has a maximum zero point deviation of ±100 comparedto the factory setting of the measuring device (calibration report).
Navigation

 Expert → Sensor → Sensor adjustm. → Zero point adj.

‣ Zero point adj.
Zero point adj. (6196)→  92

Progress (2808)→  93

Zero point adj.
Navigation

Description

Expert → Sensor → Sensor adjustm. → Zero point adj. → Zero point adj. (6196)

Expert → Sensor → Sensor adjustm. → Zero point adj. → Zero point adj. (6196)

Use this function to select the start of the zero point adjustment.
Observe conditions →  92.

Selection• Cancel• Busy• Zero adjust fail
• StartFactory setting

CancelAdditional information

Description• CancelIf zero point adjustment has failed, select this option to cancel zero point adjustment.
• BusyIs displayed during zero point adjustment.
• Zero adjust failIs displayed if zero point adjustment has failed.
• StartSelect this option to start zero point adjustment.

92

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Progress
Navigation

 Expert → Sensor → Sensor adjustm. → Zero point adj. → Progress (2808)

Description

The progress of the process is indicated.

User interface

0 to 100 %"Variable adjust" submenu
Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust‣ Variable adjust
Mass flow offset (1831)→  93Mass flow factor (1832)→  94Vol. flow offset (1841)→  94Vol. flow factor (1846)→  94

Density offset (1848)→  95

Density factor (1849)→  95

Corr. vol offset (1866)→  95

Corr. vol factor (1867)→  96

Ref.dens. offset (1868)→  96

Ref.dens. factor (1869)→  96

Temp. offset (1870)→  97

Temp. factor (1871)→  97Mass flow offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Mass flow offset (1831)

Description

Use this function to enter the zero point shift for the mass flow trim. The mass flow unit
on which the shift is based is kg/s.

Endress+Hauser

93Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

User entry

Signed floating-point number

Factory setting

0 kg/sAdditional information

Description
Corrected value = (factor × value) + offset

Mass flow factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Mass flow factor (1832)

Description

Use this function to enter a quantity factor (without time) for the mass flow. This
multiplication factor is applied over the mass flow range.

User entry

Positive floating-point number

Factory setting

1

Additional information

Description
Corrected value = (factor × value) + offset

Vol. flow offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Vol. flow offset (1841)

Description

Use this function to enter the zero point shift for the volume flow trim. The volume flow
unit on which the shift is based is m³/s.

User entry

Signed floating-point number

Factory setting

0 m³/sAdditional information

Description
Corrected value = (factor × value) + offset

Vol. flow factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Vol. flow factor (1846)

Description

Use this function to enter a quantity factor (without time) for the volume flow. This
multiplication factor is applied over the volume flow range.

User entry

Positive floating-point number

94

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

1

Additional information

Description

Description of device parameters

Corrected value = (factor × value) + offset

Density offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Density offset (1848)

Description

Use this function to enter the zero point shift for the density trim. The density unit onwhich the shift is based is kg/m³.

User entry

Signed floating-point number

Factory setting

0 kg/m³

Additional information

Description
Corrected value = (factor × value) + offset

Density factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Density factor (1849)

Description

Use this function to enter a quantity factor for the density. This multiplication factor isapplied over the density range.

User entry

Positive floating-point number

Factory setting

1

Additional information

Description
Corrected value = (factor × value) + offset

Corr. vol offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Corr. vol offset (1866)

Description

Use this function to enter the zero point shift for the corrected volume flow trim. The
corrected volume flow unit on which the shift is based is 1 Nm³/s.

User entry

Signed floating-point number

Factory setting

0 Nm³/sEndress+Hauser

95Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
Corrected value = (factor × value) + offset

Corr. vol factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Corr. vol factor (1867)

Description

Use this function to enter a quantity factor (without time) for the corrected volume flow.
This multiplication factor is applied over the corrected volume flow range.

User entry

Positive floating-point number

Factory setting

1

Additional information

Description
Corrected value = (factor × value) + offset

Ref.dens. offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Ref.dens. offset (1868)

Description

Use this parameter to enter the zero point shift for the reference density trim. The
reference density unit on which the shift is based is 1 kg/Nm³.

User entry

Signed floating-point number

Factory setting

0 kg/Nm³

Additional information

Description
Corrected value = (factor × value) + offset

Ref.dens. factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Ref.dens. factor (1869)

Description

Use this function to enter a quantity factor (without time) for the reference density. This
multiplication factor is applied over the reference density range.

User entry

Positive floating-point number

Factory setting

1

96

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
Corrected value = (factor × value) + offset

Temp. offset

Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Temp. offset (1870)

Description

Use this function to enter the zero point shift for the temperature trim. The temperatureunit on which the shift is based is K.

User entry

Signed floating-point number

Factory setting

0K

Additional information

Description
Corrected value = (factor × value) + offset

Temp. factor



Navigation

 Expert → Sensor → Sensor adjustm. → Variable adjust → Temp. factor (1871)

Description

Use this function to enter a quantity factor for the temperature. In each case, this factor
refers to the temperature in K.

User entry

Positive floating-point number

Factory setting

1

Additional information

Description
Corrected value = (factor × value) + offset

3.2.8"Calibration" submenu

Navigation

 Expert → Sensor → Calibration‣ Calibration

Endress+Hauser

Cal. factor (6025)→  98Zero point (6195)→  9897

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Nominal diameter (2807)→  98

C0 to 5 (6022)→  99Cal. factor
Navigation

 Expert → Sensor → Calibration → Cal. factor (6025)

Description

Displays the current calibration factor for the sensor.

User interface

Signed floating-point number

Factory setting

Depends on nominal diameter and calibration.

Zero pointNavigation

 Expert → Sensor → Calibration → Zero point (6195)

Description

Use this function to enter the zero point correction value for the sensor.

User entry

Signed floating-point number

Factory setting

Depends on nominal diameter and calibration.

Nominal diameter
Navigation

 Expert → Sensor → Calibration → Nominal diameter (2807)

Description

Displays the nominal diameter of the sensor.

User interface

DNxx / x"Factory setting

Depends on the size of the sensor

Additional information

Description
The value is also specified on the sensor nameplate.

98

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

C0 to 5
Navigation

 Expert → Sensor → Calibration → C0 to 5 (6022)

Description

Displays the current density coefficients C0 to 5 of the sensor.

User interface

Signed floating-point number

Factory setting

0

3.3"I/O configuration" submenu

Navigation

 Expert → I/O config.

‣ I/O config.
I/O 1 to n terminals (3902–1 to n)→  99I/O 1 to n info (3906–1 to n)→  99I/O 1 to n type (3901–1 to n)→  100

Apply I/O config (3907)→  100

Conversion code (2762)→  101

I/O terminals
Navigation

Expert → I/O config. → I/O 1 to n terminals (3902–1 to n)

Description

Displays the terminal numbers used by the I/O module.

User interface• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)

I/O info
NavigationDescription

Displays information about the plugged in I/O module.

Endress+Hauser

Expert → I/O config. → I/O 1 to n info (3906–1 to n)

99

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

User interface• Not plugged• Invalid• Not configurable• Configurable• Fieldbus

Additional information"Not plugged" option
The I/O module is not plugged in.
"Invalid" option
The I/O module is not plugged correctly.
"Not configurable" option
The I/O module is not configurable.
"Configurable" option
The I/O module is configurable.
"Fieldbus" option
The I/O module is configured for the fieldbus.

I/O type

Navigation

Expert → I/O config. → I/O 1 to n type (3901–1 to n)

Prerequisite

For the following order code:"Output; input 2", option D "Configurable I/O initial setting off"

Description

Use this function to select the I/O module type for the configuration of the I/O module.

Selection• Off• Curr.output *• Current input *• Status input *• PFS output *Factory setting

Off

Apply I/O config

NavigationDescription

Use this function to activate the newly configured I/O module type.

Selection

• No
• Yes

*

100

Expert → I/O config. → Apply I/O config (3907)

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Description of device parameters

No

Conversion code

Navigation

Expert → I/O config. → Conversion code (2762)

Description

Use this function to enter the ordered activation code to activate the I/O configuration
change.

User entry

Positive integer

Factory setting0

Additional information

Description
The I/O configuration is changed in the I/O type parameter (→  100).

3.4
Navigation"Input" submenu
 Expert → Input‣ Input

3.4.1
Navigation‣ Current input 1 to n

→  101‣ Status input 1 to n

→  104"Current input 1 to n" submenu
 Expert → Input → Current input 1 to n

‣ Current input 1 to n

Endress+Hauser

Terminal no. (1611–1 to n)→  102

Signal mode (1610–1 to n)→  102

Current span (1605–1 to n)→  102

0/4 mA value (1606–1 to n)→  103

20 mA value (1607–1 to n)→  103101

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Failure mode (1601–1 to n)→  103

Failure value (1602–1 to n)→  104

Terminal no.
Navigation

Expert → Input → Current input 1 to n → Terminal no. (1611–1 to n)

Description

Displays the terminal numbers used by the current input module.

User interface• Not used
• 24-25 (I/O 2)

Additional information"Not used" option
The current input module does not use any terminal numbers.

Signal modeNavigation

Expert → Input → Current input 1 to n → Signal mode (1610–1 to n)

Prerequisite

The measuring device is not approved for use in the hazardous area with type ofprotection Ex-i.

Description

Use this function to select the signal mode for the current input.

Selection• Passive• Active

Factory setting

ActiveCurrent spanNavigationDescription

Use this function to select the current range for the process value output and the upperand lower level for signal on alarm.

Selection• 4...20 mA• 4...20 mA NAMUR• 4...20 mA US• 0...20 mA

Factory setting

Country-specific:• 4...20 mA NAMUR• 4...20 mA US

102

Expert → Input → Current input 1 to n → Current span (1605–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

ExamplesSample values for the current range: Current span parameter (→  109)

0/4 mA value

Navigation

Expert → Input → Current input 1 to n → 0/4 mA value (1606–1 to n)

Description

Use this function to enter a value for the 4 mA current.

User entry

Signed floating-point number

Factory setting0

Additional information

Current input behavior
The current input behaves differently depending on the settings configured in thefollowing parameters:• Current span (→  102)• Failure mode (→  103)
Configuration examples
Pay attention to the configuration examples for 4 mA value parameter (→  110).

20 mA value

Navigation

Expert → Input → Current input 1 to n → 20 mA value (1607–1 to n)

Description

Use this function to enter a value for the 20 mA current.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter

Additional information

Configuration examples
Pay attention to the configuration examples for 4 mA value parameter (→  110).

Failure modeNavigationDescription

Use this function to select the input behavior when measuring a current outside theconfigured Current span parameter (→  102).

Endress+Hauser

Expert → Input → Current input 1 to n → Failure mode (1601–1 to n)

103

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• Alarm• Last valid value• Defined value

Factory setting

Alarm

Additional information

Options• Alarm
An error message is set.
• Last valid value
The last valid measured value is used.
• Defined value
A user-defined measured value is used (Failure value parameter (→  104)).

Failure value

Navigation

Expert → Input → Current input 1 to n → Failure value (1602–1 to n)

Prerequisite

In the Failure mode parameter (→  103), the Defined value option is selected.

Description

Use this function to enter the value that the device uses if it does not receive an input
signal from the external device, or if the input signal is invalid.

User entry

Signed floating-point number

Factory setting

0

3.4.2

"Status input 1 to n" submenu

Navigation

 Expert → Input → Status input 1 to n

‣ Status input 1 to n

104

Terminal no. (1358–1 to n)→  105

Assign stat.inp. (1352–1 to n)→  105

Val.stat.inp. (1353–1 to n)→  105

Active level (1351–1 to n)→  106

Response time (1354–1 to n)→  106

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Terminal no.
Navigation

Expert → Input → Status input 1 to n → Terminal no. (1358–1 to n)

Description

Displays the terminal numbers used by the status input module.

User interface• Not used
• 24-25 (I/O 2)

Additional information"Not used" option
The status input module does not use any terminal numbers.

Assign stat.inp.

Navigation

Expert → Input → Status input 1 to n → Assign stat.inp. (1352–1 to n)

Description

Use this function to select the function for the status input.

Selection• Off• Reset totaliz. 1• Reset totaliz. 2• Reset totaliz. 3• Reset all tot.
• Flow override

Factory setting

Off

Additional information

Selection• Off
The status input is switched off.
• Reset totaliz. 1…3
The individual totalizers are reset.
• Reset all tot.
All totalizers are reset.
• Flow override
The Flow override (→  76) is activated.
Note on the Flow override (→  76):
• The Flow override (→  76) is enabled as long as the level is at the status input(continuous signal).
• All other assignments react to a change in level (pulse) at the status input.

Val.stat.inp.
NavigationDescription

Displays the current input signal level.

Endress+Hauser

Expert → Input → Status input 1 to n → Val.stat.inp. (1353–1 to n)

105

Description of device parameters

User interface

Proline Promass 300 FOUNDATION Fieldbus• High• Low

Active level

Navigation

Expert → Input → Status input 1 to n → Active level (1351–1 to n)

Description

Use this function to determine the input signal level at which the assigned function is
activated.

Selection• High• Low

Factory settingHigh

Response timeNavigation

Expert → Input → Status input 1 to n → Response time (1354–1 to n)

Description

Use this function to enter the minimum time period for which the input signal level must
be present before the selected function is activated.

User entry

5 to 200 ms

Factory setting

50 ms

3.5
Navigation"Output" submenu
 Expert → Output‣ Output106‣ Curr.output 1 to n

→  107‣ PFS output 1 to n

→  121‣ Relay output 1 to n

→  142Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

3.5.1

Description of device parameters"Current output 1 to n" submenu

Navigation

 Expert → Output → Curr.output 1 to n

‣ Curr.output 1 to n
Terminal no. (0379–1 to n)→  107

Signal mode (0377–1 to n)→  108

Assign curr. 1 to n (0359–1 to n)→  108

Current span (0353–1 to n)→  109

Fixed current (0365–1 to n)→  1100/4 mA value (0367–1 to n)→  11020 mA value (0372–1 to n)→  112

Measuring mode (0351–1 to n)→  113

Damping out. 1 to n (0363–1 to n)→  117

Response time (0378–1 to n)→  118

Failure mode (0364–1 to n)→  119

Failure current (0352–1 to n)→  120

Output curr. 1 to n (0361–1 to n)→  120

Measur. curr. 1 to n (0366–1 to n)→  121

Terminal no.
Navigation

Expert → Output → Curr.output 1 to n → Terminal no. (0379–1 to n)

Description

Displays the terminal numbers used by the current output module.

User interface• Not used
• 24-25 (I/O 2)

Additional information"Not used" option
The current output module does not use any terminal numbers.

Endress+Hauser

107Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Signal modeNavigation

Expert → Output → Curr.output 1 to n → Signal mode (0377–1 to n)

Description

Use this function to select the signal mode for the current output.

Selection• Passive• Active

Factory setting

ActiveAssign curr.



Navigation

Expert → Output → Curr.output 1 to n → Assign curr. 1 to n (0359–1 to n)

Description

Use this function to select a process variable for the current output.
Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Value 1 display parameter (→  18)

Selection

*

108

• Off
• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *• HBSI *Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Description of device parameters

Mass flow

Current spanNavigation

Expert → Output → Curr.output 1 to n → Current span (0353–1 to n)

Description

Use this function to select the current range for the process value output and the upperand lower level for signal on alarm.

Selection• 4...20 mA NAMUR• 4...20 mA US• 4...20 mA• 0...20 mA• Fixed current

Factory setting

Country-specific:• 4...20 mA NAMUR• 4...20 mA US

Additional information

Description• In the event of a device alarm, the current output adopts the value specified in the
Failure mode parameter (→  119).
• If the measured value is outside the measuring range, the diagnostic message
S441 Curr.output 1 to n is displayed.
• The measuring range is specified via the 0/4 mA value parameter (→  110) and
20 mA value parameter (→  112).
"Fixed current" option
The current value is set via the Fixed current parameter (→  110).
ExampleShows the relationship between the current span for the output of the process variable andthe lower and upper alarm levels:
2

1

3

I [mA]
A0034351

1
2
3

Current span for process value
Lower level for signal on alarm
Upper level for signal on alarm

Selection

Endress+Hauser

Options

1

2

3

4...20 mA NAMUR

3.8 to 20.5 mA< 3.6 mA

> 21.95 mA

4...20 mA US

3.9 to 20.8 mA US< 3.6 mA

> 21.95 mA

109

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Options

1

2

3

4...20 mA

4 to 20.5 mA< 3.6 mA

> 21.95 mA

0...20 mA

0 to 20.5 mA< 0 mA> 21.95 mA

If the flow exceeds or falls below the upper or lower signal on alarm level, thediagnostic message S441 Curr.output 1 to n is displayed.

Fixed current

Navigation

Expert → Output → Curr.output 1 to n → Fixed current (0365–1 to n)

Prerequisite

The Fixed current option is selected in the Current span parameter (→  109).

Description

Use this function to enter a constant current value for the current output.

User entry

0 to 22.5 mA

Factory setting

22.5 mA

0/4 mA value

Navigation

Expert → Output → Curr.output 1 to n → 0/4 mA value (0367–1 to n)

Prerequisite

One of the following options is selected in the Current span parameter (→  109):• 4...20 mA NAMUR• 4...20 mA US• 4...20 mA• 0...20 mA

Description

Use this function to enter a value for the 0/4 mA current.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 kg/h• 0 lb/min

Additional information

Description
Positive and negative values are permitted depending on the process variable assigned in
the Assign curr. parameter (→  108). In addition, the value can be greater than or

110

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

smaller than the value assigned for the 20 mA current in the 20 mA value parameter(→  112).
Dependency
The unit depends on the process variable selected in the Assign curr. parameter(→  108).
Current output behavior
The current output behaves differently depending on the settings configured in thefollowing parameters:• Current span (→  109)
• Failure mode (→  119)
Configuration examples
Some examples of parameter settings and their effect on the current output are given in
the following section.
Configuration example A
Measuring mode with Forward flow option• 0/4 mA value parameter (→  110) = not equal to zero flow (e.g. –250 m3/h)• 20 mA value parameter (→  112) = not equal to zero flow (e.g. +750 m3/h)• Calculated current value = 8 mA at zero flow
I [mA]
20
84Q-250 01+750
1A0013757

Q
I
1

Flow
Current
Measuring range is exceeded or undershot

The operational range of the measuring device is defined by the values entered for the 0/4
mA value parameter (→  110) and 20 mA value parameter (→  112). If theeffective flow exceeds or falls below this operational range, the diagnostic message
S441 Curr.output 1 to n is displayed.
Configuration example B
Measuring mode with Forward/Reverse option
I [mA]
20

4
0

Q

0
3

1

2
A0013758

I
Q
1
2
3

Endress+Hauser

Current
Flow
Value assigned to the 0/4 mA current
Forward flow
Reverse flow

111

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

The current output signal is independent of the direction of flow (absolute amount of themeasured variable). The values for the 0/4 mA value parameter (→  110) and 20 mA
value parameter (→  112) must have the same sign. The value for the 20 mA value
parameter (→  112) (e.g. reverse flow) corresponds to the mirrored value for the 20
mA value parameter (→  112) (e.g. forward flow).
Configuration example C
Measuring mode with Rev. flow comp. option
If flow is characterized by severe fluctuations (e.g. when using reciprocating pumps), flow
components outside the measuring range are buffered, balanced and output after amaximum delay of 60 s →  113.

20 mA value

Navigation

Expert → Output → Curr.output 1 to n → 20 mA value (0372–1 to n)

Prerequisite

One of the following options is selected in the Current span parameter (→  109):• 4...20 mA NAMUR• 4...20 mA US• 4...20 mA• 0...20 mA

Description

Use this function to enter a value for the 20 mA current.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter→  328

Additional information

Description
Positive and negative values are permitted depending on the process variable assigned in
the Assign curr. parameter (→  108). In addition, the value can be greater than orsmaller than the value assigned for the 0/4 mA current in the 0/4 mA value parameter(→  110).
Dependency
The unit depends on the process variable selected in the Assign curr. parameter(→  108).
Example• Value assigned to 0/4 mA = –250 m3/h
• Value assigned to 20 mA = +750 m3/h
• Calculated current value = 8 mA (at zero flow)
If the Forward/Reverse option is selected in the Measuring mode parameter(→  113), different signs cannot be entered for the values of the 0/4 mA value
parameter (→  110) and 20 mA value parameter (→  112). The diagnostic message
S441 Curr.output 1 to n is displayed.
Configuration examples
Observe the configuration examples for the 0/4 mA value parameter (→  110).

112

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Measuring modeNavigation

Expert → Output → Curr.output 1 to n → Measuring mode (0351–1 to n)

Prerequisite

In the Assign curr. parameter (→  108), one of the following options is selected:• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *• HBSI *Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Value 1 display parameter (→  18)
In the Current span parameter (→  109), one of the following options is selected:• 4...20 mA NAMUR• 4...20 mA US• 4...20 mA• 0...20 mA

Description

Use this function to select the measuring mode for the current output.

Selection• Forward flow
• Forward/Reverse• Rev. flow comp.

Factory setting

Forward flow*

Visibility depends on order options or device settings

Endress+Hauser

113

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
The process variable that is assigned to the current output via the Assign curr.
parameter (→  108) is displayed below the parameter.
"Forward flow" option
The current output signal is proportional to the process variable assigned. The measuring
range is defined by the values that are assigned to the 0/4 mA and 20 mA current value.
The flow components outside the scaled measuring range are taken into account for signal
output as follows:• Both values are defined such that they are not equal to zero flow e.g.:– 0/4 mA current value = –5 m3/h
– 20 mA current value = 10 m3/h
• If the effective flow exceeds or falls below this measuring range, the diagnostic message
S441 Curr.output 1 to n is displayed.
"Forward/Reverse" option
I [mA]
20

4
0

Q

0
3

1

2
A0013758

I
Q
1
2
3

Current
Flow
Value assigned to the 0/4 mA current
Forward flow
Reverse flow

• The current output signal is independent of the direction of flow (absolute amount of themeasured variable). The values for the 0/4 mA value parameter (→  110) and 20 mA
value parameter (→  112) must have the same sign.
• The value for the 20 mA value parameter (→  112) (e.g. reverse flow) corresponds tothe mirrored value for the 20 mA value parameter (→  112) (e.g. forward flow).
"Rev. flow comp." option
The Rev. flow comp. option is primarily used to compensate for abrupt reverse flow which
can occur in connection with positive displacement pumps as a result of wear or highviscosity. The reverse flows are recorded in a buffer and balanced against forward flow thenext time flow is in the forward direction.
If buffering cannot be processed within approx. 60 s, the diagnostic message
S441 Curr.output 1 to n is displayed.
Flow values can aggregate in the buffer in the event of prolonged and unwanted fluid
reverse flow. However, these flows are not taken into consideration by the current output
configuration, i.e. the reverse flow is not compensated.
If this option is set, the measuring device does not attenuate the flow signal. The flow
signal is not attenuated.
Examples of how the current output behaves
Example 1
Defined measuring range: lower range value and upper range value with the same sign

114

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

I [mA]
20

4
0

1

2

Q
A0028084

2
I
Q
1
2

Measuring range
Current
Flow
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)With the following flow response:Qt
A0028091

3
Qt

Flow response
Flow
Time

With Forward flow option
The current output signal is proportional to the process variable assigned. The flow
components outside the scaled measuring range are not taken into account for signal
output:.
I [mA]

t
A0028092

It

Current
Time

With Forward/Reverse option
The current output signal is independent of the direction of flow.
I [mA]

t
A0028093

It

Current
Time

With Rev. flow comp. option

Endress+Hauser

115

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Flow components outside the span are buffered, balanced and output after a maximum
delay of 60 s.
I [mA]At

S
S=A

A0028094

It
S
A

Current
Time
Flow components saved
Balancing of saved flow components

Example 2
Defined measuring range: lower range value and upper range value with different signs
I [mA]
20
4
0

Q

0
1

2
A0028095

4
I
Q
1
2

Measuring range
Current
Flow
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)With flow a (—) outside, b (- -) inside the measuring range
Q
1
+

t

0
2

-

a=
b=
A0028098

Q
t
1
2

Flow
Time
Lower range value (value assigned to 0/4 mA current)
Upper range value (value assigned to 20 mA current)With Forward flow option• a (—): The flow components outside the scaled measuring range cannot be taken into
account for signal output.
The diagnostic message S441 Curr.output 1 to n is displayed.
• b (- -): The current output signal is proportional to the process variable assigned.

116

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

I [mA]ta=b=
A0028100It

Current
Time

With Forward/Reverse option
This option is not possible in this case as the values for the 0/4 mA value parameter(→  110) and 20 mA value parameter (→  112) have different signs.
With Rev. flow comp. option
Flow components outside the span are buffered, balanced and output after a maximum
delay of 60 s.
I [mA]At

S
S=A

A0028101It
S
A

Current
Time
Flow components saved
Balancing of saved flow components

Damping out.



Navigation

Prerequisite

A process variable is selected in the Assign curr. parameter (→  108) and one of thefollowing options is selected in the Current span parameter (→  109):• 4...20 mA NAMUR• 4...20 mA US• 4...20 mA• 0...20 mA

Description

Use this function to enter a time constant for the reaction time of the current output signal
to fluctuations in the measured value caused by process conditions.

User entry

0.0 to 999.9 s

Factory setting

1.0 s

Endress+Hauser

Expert → Output → Curr.output 1 to n → Damping out. 1 to n (0363–1 to n)

117

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

User entry
Use this function to enter a time constant (PT1 element 6)) for current output damping:• If a low time constant is entered, the current output reacts particularly quickly tofluctuating measured variables.
• On the other hand, the current output reacts more slowly if a high time constant isentered.
Damping is switched off if 0 is entered (factory setting).

Response time
Navigation

Expert → Output → Curr.output 1 to n → Response time (0378–1 to n)

Prerequisite

One of the following options is selected in the Assign curr. parameter (→  108):• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *• HBSI *Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Value 1 display parameter (→  18)
One of the following options is selected in the Current span parameter (→  109):• 4...20 mA NAMUR• 4...20 mA US• 4...20 mA• 0...20 mA

6)
*

118

proportional transmission behavior with first order delay
Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Description

Displays the response time. This specifies how quickly the current output reaches themeasured value change of 63 % of 100 % of the measured value change.

User interface

Positive floating-point number

Additional information

Description
The response time is made up of the time specified for the following dampings:• Current output damping →  117
and• Depending on the measured variable assigned to the output.
– Flow dampingor– Density dampingor– Temperature damping

Failure modeNavigation

Prerequisite

A process variable is selected in the Assign curr. parameter (→  108) and one of thefollowing options is selected in the Current span parameter (→  109):• 4...20 mA NAMUR• 4...20 mA US• 4...20 mA• 0...20 mA

Description

Use this function to select the value of the current output in the event of a device alarm.

Selection• Min.
• Max.
• Last valid value• Actual value• Defined value

Factory settingMax.

Endress+Hauser

Expert → Output → Curr.output 1 to n → Failure mode (0364–1 to n)

119Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
This setting does not affect the failsafe mode of other outputs and totalizers. This isspecified in separate parameters.
"Min." option
The current output adopts the value of the lower level for signal on alarm.
The signal on alarm level is defined via the Current span parameter (→  109).
"Max." option
The current output adopts the value of the upper level for signal on alarm.
The signal on alarm level is defined via the Current span parameter (→  109).
"Last valid value" option
The current output adopts the last measured value that was valid before the device alarm
occurred.
"Actual value" option
The current output adopts the measured value on the basis of the current flow
measurement; the device alarm is ignored.
"Defined value" option
The current output adopts a defined measured value.
The measured value is defined via the Failure current parameter (→  120).

Failure current

Navigation

Expert → Output → Curr.output 1 to n → Failure current (0352–1 to n)

Prerequisite

The Defined value option is selected in the Failure mode parameter (→  119).

Description

Use this function to enter a fixed value that the current output adopts in the event of adevice alarm.

User entry

0 to 22.5 mA

Factory setting

22.5 mA

Output curr.
NavigationDescription

Displays the current value currently calculated for the current output.

User interface

3.59 to 22.5 mA

120

Expert → Output → Curr.output 1 to n → Output curr. 1 to n (0361–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Measur. curr.
Navigation

Expert → Output → Curr.output 1 to n → Measur. curr. 1 to n (0366–1 to n)

Description

Use this function to display the actual measured value of the output current.

User interface

0 to 30 mA

3.5.2"Pulse/frequency/switch output 1 to n" submenu

Navigation

 Expert → Output → PFS output 1 to n

‣ PFS output 1 to n

Endress+Hauser

Terminal no. (0492–1 to n)→  122Signal mode (0490–1 to n)→  123

Operating mode (0469–1 to n)→  123

Assign pulse 1 to n (0460–1 to n)→  125

Value per pulse (0455–1 to n)→  125

Pulse width (0452–1 to n)→  126

Measuring mode (0457–1 to n)→  126

Failure mode (0480–1 to n)→  127

Pulse output 1 to n (0456–1 to n)→  128

Assign freq. (0478–1 to n)→  128

Min. freq. value (0453–1 to n)→  129

Max. freq. value (0454–1 to n)→  130

Val. at min.freq (0476–1 to n)→  130

Val. at max.freq (0475–1 to n)→  130

Measuring mode (0479–1 to n)→  131

Damping out. 1 to n (0477–1 to n)→  132

121

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Response time (0491–1 to n)→  133

Failure mode (0451–1 to n)→  134

Failure freq. (0474–1 to n)→  135

Output freq. 1 to n (0471–1 to n)→  135

Switch out funct (0481–1 to n)→  135

Assign diag. beh (0482–1 to n)→  136

Assign limit (0483–1 to n)→  137

Switch-on value (0466–1 to n)→  138

Switch-off value (0464–1 to n)→  139

Assign dir.check (0484–1 to n)→  139

Assign status (0485–1 to n)→  140

Switch-on delay (0467–1 to n)→  140

Switch-off delay (0465–1 to n)→  140

Failure mode (0486–1 to n)→  141

Switch status 1 to n (0461–1 to n)→  141

Invert outp.sig. (0470–1 to n)→  142

Terminal no.
Navigation

Expert → Output → PFS output 1 to n → Terminal no. (0492–1 to n)

Description

Displays the terminal numbers used by the pulse/frequency/switch output module.

User interface• Not used
• 24-25 (I/O 2)

Additional information"Not used" option
The pulse/frequency/switch output module does not use any terminal numbers.

122

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Signal modeNavigation

Expert → Output → PFS output 1 to n → Signal mode (0490–1 to n)

Description

Use this function to select the signal mode for the pulse/frequency/switch output.

Selection• Passive• Active

Factory setting

Passive

Operating modeNavigation

Expert → Output → PFS output 1 to n → Operating mode (0469–1 to n)

Description

Use this function to select the operating mode of the output as a pulse, frequency or switch
output.

Selection• Pulse• Frequency• Switch

Factory setting

Pulse

Additional information"Pulse" option
Quantity-dependent pulse with configurable pulse width• Whenever a specific mass, volume, corrected volume, target mass or carrier mass isreached (pulse value), a pulse is output, the duration of which was set previously (pulse
width).
• The pulses are never shorter than the set duration.
Example• Flow rate approx. 100 g/s• Pulse value 0.1 g• Pulse width 0.05 ms• Pulse rate 1 000 Impuls/s

Endress+Hauser

123Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

U[V]

B<P
B

24
t

0
P

A0026883

5
BP

Quantity-proportional pulse (pulse value) with pulse width to be configured
Pulse width entered
Pauses between the individual pulses"Frequency" option
Flow-proportional frequency output with 1:1 on/off ratio
An output frequency is output that is proportional to the value of a process variable, suchas mass flow, volume flow, corrected volume flow, target mass flow, carrier mass flow,
density, reference density, concentration, dynamic viscosity, kinematic viscosity,
temperature-compensated dynamic viscosity, temperature-compensated kinematic
viscosity, temperature, carrier tube temperature, electronic temperature, vibration
frequency, frequency fluctuation, oscillation amplitude, oscillation damping, oscillation
damping fluctuation, signal asymmetry or excitation current.
Example• Flow rate approx. 100 g/s• Max. frequency 10 kHz• Flow rate at max. frequency 1 000 g/s• Output frequency approx. 1 000 Hz
U[V]
24
t

0

A0026886

6

Flow-proportional frequency output"Switch" option
Contact for displaying a condition (e.g. alarm or warning if a limit value is reached)
ExampleAlarm response without alarm
U[V]
24
0

t
A0026884

7

No alarm, high level

ExampleAlarm response in case of alarm

124

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

U[V]
24
0

t
A0026885

8

Alarm, low level

Assign pulseNavigation

Expert → Output → PFS output 1 to n → Assign pulse 1 to n (0460–1 to n)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  123) parameter.

Description

Use this function to select the process variable for the pulse output.

Selection• Off• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *Factory setting

Off

Value per pulseNavigation

Expert → Output → PFS output 1 to n → Value per pulse (0455–1 to n)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  123) and a process
variable is selected in the Assign pulse parameter (→  125).

Description

Use this function to enter the value for the measured value that a pulse is equivalent to.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter→  329

Additional information

User entry
Weighting of the pulse output with a quantity.
The lower the pulse value, the• better the resolution.
• the higher the frequency of the pulse response.

*

Visibility depends on order options or device settings

Endress+Hauser

125Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Pulse widthNavigation

Expert → Output → PFS output 1 to n → Pulse width (0452–1 to n)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  123) and a process
variable is selected in the Assign pulse parameter (→  125).

Description

Use this function to enter the duration of the output pulse.

User entry

0.05 to 2 000 ms

Factory setting

100 msAdditional information

Description• Define how long a pulse is (duration).
• The maximum pulse rate is defined by fmax = 1 / (2 × pulse width).
• The interval between two pulses lasts at least as long as the set pulse width.
• The maximum flow is defined by Qmax = fmax × pulse value.
• If the flow exceeds these limit values, the measuring device displays the diagnostic
message S443 Pulse output 1 to n.
U[V]

U[V]

B<P

B=P
B

B

t
P

t
P
A0026882

B
P

Pulse width entered
Pauses between the individual pulses

Example• Pulse value: 0.1 g• Pulse width: 0.1 ms• fmax: 1 / (2 × 0.1 ms) = 5 kHz• Qmax: 5 kHz × 0.1 g = 0.5 kg/s

Measuring modeNavigation

Prerequisite

In the Operating mode parameter (→  123), the Pulse option is selected, and one of
the following options is selected in the Assign pulse parameter (→  125):• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. **126

Expert → Output → PFS output 1 to n → Measuring mode (0457–1 to n)

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to select the measuring mode for the pulse output.

Selection• Forward flow
• Forward/Reverse• Reverse flow
• Rev. flow comp.

Factory setting

Forward flow

Additional information

Selection• Forward flow
Positive flow is output, negative flow is not output.
• Forward/Reverse
Positive and negative flow are output (absolute value), but a distinction is not madebetween positive and negative flow.
• Reverse flow
Negative flow is output, positive flow is not output.
• Rev. flow comp.
The flow components outside the span are buffered, balanced and output after amaximum delay of 60 s.
For a detailed description of the options available, see the Measuring modeparameter (→  113)
ExamplesFor a detailed description of the configuration examples, see the Measuring modeparameter (→  113)

Failure modeNavigation

Expert → Output → PFS output 1 to n → Failure mode (0480–1 to n)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  123) and a process
variable is selected in the Assign pulse parameter (→  125).

Description

Use this function to select the failure mode of the pulse output in the event of a device
alarm.

Selection• Actual value• No pulses

Factory setting

No pulses

Additional information

Description
The dictates of safety render it advisable to ensure that the pulse output shows apredefined behavior in the event of a device alarm.
Selection• Actual value
In the event of a device alarm, the pulse output continues on the basis of the current
flow measurement. The fault is ignored.
• No pulses
In the event of a device alarm, the pulse output is "switched off".

Endress+Hauser

127

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

NOTICE! A device alarm is a measuring device error that must be taken seriously. It canaffect the measurement quality such that the quality can no longer be guaranteed. The
Actual value option is only recommended if it can be guaranteed that all possible alarm
conditions will not affect the measurement quality.

Pulse output
Navigation

Expert → Output → PFS output 1 to n → Pulse output 1 to n (0456–1 to n)

Prerequisite

The Pulse option is selected in the Operating mode parameter (→  123) parameter.

Description

Displays the pulse frequency currently output.

User interface

Positive floating-point number

Additional information

Description• The pulse output is an open collector output.
• This is configured at the factory in such a way that the transistor is conductive for theduration of the pulse (NO contact) and is safety-oriented.

1

1

t

0
NC

t

0
NO

A0028726

0
1
NC
NO

Non-conductive
Conductive
NC contact (normally closed)
NO contact (normally open)

The output behavior can be reversed via the Invert outp.sig. parameter (→  142) i.e.
the transistor does not conduct for the duration of the pulse.
In addition, the behavior of the output in the event of a device alarm (Failure modeparameter (→  127)) can be configured.

Assign freq.

Navigation

Expert → Output → PFS output 1 to n → Assign freq. (0478–1 to n)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  123)
parameter.

Description

Use this function to select the process variable for the frequency output.
Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Value 1 display parameter (→  18)

128

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection• Off• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *• HBSI

Factory setting

Off

Description of device parameters

Min. freq. value

Navigation

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  123) and aprocess variable is selected in the Assign freq. parameter (→  128).

Description

Use this function to enter the start value frequency.

User entry

0.0 to 10 000.0 Hz

Factory setting

0.0 Hz*Expert → Output → PFS output 1 to n → Min. freq. value (0453–1 to n)

Visibility depends on order options or device settings

Endress+Hauser

129

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Max. freq. value

Navigation

Expert → Output → PFS output 1 to n → Max. freq. value (0454–1 to n)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  123) and aprocess variable is selected in the Assign freq. parameter (→  128).

Description

Use this function to enter the end value frequency.

User entry

0.0 to 10 000.0 Hz

Factory setting

10 000.0 Hz

Val. at min.freq

Navigation

Expert → Output → PFS output 1 to n → Val. at min.freq (0476–1 to n)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  123) and aprocess variable is selected in the Assign freq. parameter (→  128).

Description

Use this function to enter the measured value for the start value frequency.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter

Additional information

Dependency
The entry depends on the process variable selected in the Assign freq. parameter(→  128).

Val. at max.freq

Navigation

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  123) and aprocess variable is selected in the Assign freq. parameter (→  128).

Description

Use this function to enter the measured value for the end value frequency.

User entry

Signed floating-point number

Factory setting

Depends on country and nominal diameter

130

Expert → Output → PFS output 1 to n → Val. at max.freq (0475–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
Use this function to enter the maximum measured value at the maximum frequency. The
selected process variable is output as a proportional frequency.
Dependency
The entry depends on the process variable selected in the Assign freq. parameter(→  128).

Measuring modeNavigation

Expert → Output → PFS output 1 to n → Measuring mode (0479–1 to n)

Prerequisite

In the Operating mode parameter (→  123), the Frequency option is selected, and oneof the following options is selected in the Assign freq. parameter (→  128):• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Value 1 display parameter (→  18)

Description

Use this function to select the measuring mode for the frequency output.

Selection• Forward flow
• Forward/Reverse• Rev. flow comp.

*

Visibility depends on order options or device settings

Endress+Hauser

131Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Forward flow

Additional information

Selection
For a detailed description of the options available, see the Measuring modeparameter (→  113)
ExamplesFor a detailed description of the configuration examples, see the Measuring modeparameter (→  113)

Damping out.



Navigation

Expert → Output → PFS output 1 to n → Damping out. 1 to n (0477–1 to n)

Prerequisite

In the Operating mode parameter (→  123), the Frequency option is selected, and oneof the following options is selected in the Assign freq. parameter (→  128):• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Value 1 display parameter (→  18)

Description

Use this function to enter a time constant for the reaction time of the output signal tofluctuations in the measured value.

User entry

0 to 999.9 s*

132

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

0.0 s

Additional information

User entry

Description of device parameters

Use this function to enter a time constant (PT1 element 7)) for frequency output damping:• If a low time constant is entered, the current output reacts particularly quickly tofluctuating measured variables.
• On the other hand, the current output reacts more slowly if a high time constant isentered.
Damping is switched off if 0 is entered (factory setting).
The frequency output is subject to separate damping that is independent of all precedingtime constants.

Response time
Navigation

Expert → Output → PFS output 1 to n → Response time (0491–1 to n)

Prerequisite

In the Operating mode parameter (→  123), the Frequency option is selected, and oneof the following options is selected in the Assign freq. parameter (→  128):• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Osc. ampl. 0 *• Osc. ampl. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Value 1 display parameter (→  18)

7)*

proportional transmission behavior with first order delay
Visibility depends on order options or device settings

Endress+Hauser

133

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Description

Displays the response time. This specifies how quickly the pulse/frequency/switch output
reaches the measured value change of 63 % of 100 % of the measured value change.

User interface

Positive floating-point number

Additional information

Description
The response time is made up of the time specified for the following dampings:• Damping of pulse/frequency/switch output →  117
and• Depending on the measured variable assigned to the output.
– Flow dampingor– Density dampingor– Temperature damping

Failure modeNavigation

Expert → Output → PFS output 1 to n → Failure mode (0451–1 to n)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  123) and aprocess variable is selected in the Assign freq. parameter (→  128).

Description

Use this function to select the failure mode of the frequency output in the event of a device
alarm.

Selection• Actual value• Defined value• 0 HzFactory setting

0 Hz

Additional information

Selection• Actual value
In the event of a device alarm, the frequency output continues on the basis of the current
flow measurement. The device alarm is ignored.
• Defined value
In the event of a device alarm, the frequency output continues on the basis of apredefined value. The Failure freq. (→  135) replaces the current measured value,
making it possible to bypass the device alarm. The actual measurement is switched off
for the duration of the device alarm.
• 0 HzIn the event of a device alarm, the frequency output is "switched off".
NOTICE! A device alarm is a measuring device error that must be taken seriously. It canaffect the measurement quality such that the quality can no longer be guaranteed. The
Actual value option is only recommended if it can be guaranteed that all possible alarm
conditions will not affect the measurement quality.

134

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Failure freq.

Navigation

Expert → Output → PFS output 1 to n → Failure freq. (0474–1 to n)

Prerequisite

The Frequency option is selected in the Operating mode parameter (→  123) and aprocess variable is selected in the Assign freq. parameter (→  128).

Description

Use this function to enter the value for the frequency output in the event of a device alarmin order to bypass the alarm.

User entry

0.0 to 12 500.0 Hz

Factory setting

0.0 HzOutput freq.
Navigation

Expert → Output → PFS output 1 to n → Output freq. 1 to n (0471–1 to n)

Prerequisite

In the Operating mode parameter (→  123), the Frequency option is selected.

Description

Displays the actual value of the output frequency which is currently measured.

User interface

0.0 to 12 500.0 Hz

Switch out funct

Navigation

Prerequisite

The Switch option is selected in the Operating mode parameter (→  123).

Description

Use this function to select a function for the switch output.

Selection

• Off
• On
• Diag. behavior• Limit• Fl. direct.check• Status

Factory setting

Off

Endress+Hauser

Expert → Output → PFS output 1 to n → Switch out funct (0481–1 to n)

135

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Selection• Off
The switch output is permanently switched off (open, non-conductive).
• On
The switch output is permanently switched on (closed, conductive).
• Diag. behavior
Indicates if the diagnostic event is present or not. Is used to output diagnostic
information and to react to it appropriately at the system level.
• Limit
Indicates if a specified limit value has been reached for the process variable. Is used to
output diagnostic information relating to the process and to react to it appropriately at
the system level.
• Fl. direct.check
Indicates the flow direction (forward or reverse flow).
• Status
Indicates the device status depending on whether empty pipe detection or low flow cut
off is selected.

Assign diag. behNavigation

Expert → Output → PFS output 1 to n → Assign diag. beh (0482–1 to n)

Prerequisite• In the Operating mode parameter (→  123), the Switch option is selected.
• In the Switch out funct parameter (→  135), the Diag. behavior option is selected.

Description

Use this function to select the diagnostic event category that is displayed for the switch
output.

Selection• Alarm• Alarm or warning• Warning

Factory setting

Alarm

Additional information

Description
If no diagnostic event is pending, the switch output is closed and conductive.
Selection• Alarm
The switch output signals only diagnostic events in the alarm category.
• Alarm or warning
The switch output signals diagnostic events in the alarm and warning category.
• Warning
The switch output signals only diagnostic events in the warning category.

136

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Assign limitNavigation

Expert → Output → PFS output 1 to n → Assign limit (0483–1 to n)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  123) parameter.
• The Limit option is selected in the Switch out funct parameter (→  135) parameter.

Description

Use this function to select a process variable for the limit function.

Selection• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Dynam. viscosity *• Concentration *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Totalizer 1• Totalizer 2• Totalizer 3• Oscil. damping

Factory setting

Mass flow

Additional information

Description
Behavior of status output when Switch-on value > Switch-off value:• Process variable > Switch-on value: transistor is conductive• Process variable < Switch-off value: transistor is non-conductive
A
1
2
t
B
3
4

t
A0026891

1
2
3
4
A
B

*

Switch-on value
Switch-off value
Conductive
Non-conductive
Process variable
Status output

Visibility depends on order options or device settings

Endress+Hauser

137

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Behavior of status output when Switch-on value < Switch-off value:• Process variable < Switch-on value: transistor is conductive• Process variable > Switch-off value: transistor is non-conductive
A
1
2
t
B
3
4

t
A0026892

1
2
3
4
A
B

Switch-off value
Switch-on value
Conductive
Non-conductive
Process variable
Status output

Behavior of status output when Switch-on value = Switch-off value:• Process variable > Switch-on value: transistor is conductive• Process variable < Switch-off value: transistor is non-conductive
A
1
t
B
2
3

t
A0026893

1
2
3
A
B

Switch-on value = Switch-off value
Conductive
Non-conductive
Process variable
Status output

Switch-on value

Navigation

Prerequisite• In the Operating mode parameter (→  123), the Switch option is selected.
• In the Switch out funct parameter (→  135), the Limit option is selected.

Description

Use this function to enter the measured value for the switch-on point.

User entry

Signed floating-point number

138

Expert → Output → PFS output 1 to n → Switch-on value (0466–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Country-specific:• 0 kg/h• 0 lb/min

Additional information

Description

Description of device parameters

Use this function to enter the limit value for the switch-on value (process variable >switch-on value = closed, conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit depends on the process variable selected in the Assign limit parameter(→  137).

Switch-off value

Navigation

Expert → Output → PFS output 1 to n → Switch-off value (0464–1 to n)

Prerequisite• In the Operating mode parameter (→  123), the Switch option is selected.
• In the Switch out funct parameter (→  135), the Limit option is selected.

Description

Use this function to enter the measured value for the switch-off point.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 kg/h• 0 lb/min

Additional information

Description
Use this function to enter the limit value for the switch-off value (process variable <switch-off value = open, non-conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit depends on the process variable selected in the Assign limit parameter(→  137).

Assign dir.checkNavigation

Prerequisite• The Switch option is selected in the Operating mode parameter (→  123).
• The Fl. direct.check option is selected in the Switch out funct parameter (→  135).

Description

Use this function to select a process variable for monitoring the flow direction.

Endress+Hauser

Expert → Output → PFS output 1 to n → Assign dir.check (0484–1 to n)

139Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• Off• Volume flow
• Mass flow
• Correct.vol.flow

Factory setting

Mass flow

Assign status



Navigation

Expert → Output → PFS output 1 to n → Assign status (0485–1 to n)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  123).
• The Status option is selected in the Switch out funct parameter (→  135).

Description

Use this function to select a device status for the switch output.

Selection• Partial pipe det• Low flow cut off• Digital outp. 6

Factory setting

Partial pipe det

Additional information

Options
If empty pipe detection or low flow cut off are enabled, the output is conductive.
Otherwise, the switch output is non-conductive.

Switch-on delay

Navigation

Expert → Output → PFS output 1 to n → Switch-on delay (0467–1 to n)

Prerequisite• The Switch option is selected in the Operating mode parameter (→  123).
• The Limit option is selected in the Switch out funct parameter (→  135).

Description

Use this function to enter a delay time for switching on the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Switch-off delay

Navigation

Prerequisite• The Switch option is selected in the Operating mode parameter (→  123).
• The Limit option is selected in the Switch out funct parameter (→  135).

140

Expert → Output → PFS output 1 to n → Switch-off delay (0465–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to enter a delay time for switching off the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Failure modeNavigation

Expert → Output → PFS output 1 to n → Failure mode (0486–1 to n)

Description

Use this function to select a failsafe mode for the switch output in the event of a device
alarm.

Selection• Actual status• Open• Closed

Factory setting

Open

Additional information

Options• Actual status
In the event of a device alarm, faults are ignored and the current behavior of the input
value is output by the switch output. The Actual status option behaves in the same way
as the current input value.
• OpenIn the event of a device alarm, the switch output's transistor is set to non-conductive.
• ClosedIn the event of a device alarm, the switch output's transistor is set to conductive.

Switch status
Navigation

Expert → Output → PFS output 1 to n → Switch status 1 to n (0461–1 to n)

Prerequisite

The Switch option is selected in the Operating mode parameter (→  123).

Description

Displays the current switch status of the status output.

User interface• Open• Closed

Additional information

User interface• OpenThe switch output is not conductive.
• Closed
The switch output is conductive.

Endress+Hauser

141

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Invert outp.sig.

Navigation

Expert → Output → PFS output 1 to n → Invert outp.sig. (0470–1 to n)

Description

Use this function to select whether to invert the output signal.

Selection

• No
• Yes

Factory setting

No

Additional information

Selection
No option (passive - negative)

1
0
A0026693

Yes option (passive - positive)

1
0
A0026692

3.5.3
Navigation"Relay output 1 to n" submenu
 Expert → Output → Relay output 1 to n

‣ Relay output 1 to n

142

Terminal no.

→  143

Relay outp.func.

→  143

Assign dir.check→  144

Assign limit→  144

Assign diag. beh→  145

Assign status→  145

Switch-off value→  146

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Switch-off delay→  146

Switch-on value→  146

Switch-on delay→  147

Failure mode→  147

Switch status→  148

Powerless relay→  148

Terminal no.
Navigation

Expert → Output → Relay output 1 to n → Terminal no. (0812–1 to n)

Description

Displays the terminal numbers used by the relay output module.

User interface• Not used
• 24-25 (I/O 2)

Additional information"Not used" option
The relay output module does not use any terminal numbers.

Relay outp.func.

Navigation

Expert → Output → Relay output 1 to n → Relay outp.func. (0804–1 to n)

Description

Use this function to select an output function for the relay output.

Selection• Closed• Open• Diag. behavior• Limit• Fl. direct.check• Digital Output

Factory setting

ClosedAdditional information

Selection• Closed
The relay output is permanently switched on (closed, conductive).
• OpenThe relay output is permanently switched off (open, non-conductive).
• Diag. behavior
Indicates if the diagnostic event is present or not. Is used to output diagnostic
information and to react to it appropriately at the system level.

Endress+Hauser

143

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Limit
Indicates if a specified limit value has been reached for the process variable. Is used to
output diagnostic information relating to the process and to react to it appropriately at
the system level.
• Fl. direct.check
Indicates the flow direction (forward or reverse flow).
• Digital Output
Indicates the device status depending on whether empty pipe detection or low flow cut
off is selected.

Assign dir.checkNavigation

Expert → Output → Relay output 1 to n → Assign dir.check (0808–1 to n)

Prerequisite

In the Relay outp.func. parameter (→  143), the Fl. direct.check option is selected.

Description

Use this function to select a process variable for monitoring the flow direction.

Selection• Off• Volume flow
• Mass flow
• Correct.vol.flow

Factory setting

Mass flow

Assign limitNavigation

Prerequisite

The Limit option is selected in the Relay outp.func. parameter (→  143) parameter.

Description

Use this function to select a process variable for the limit value function.

Selection• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Dynam. viscosity *• Concentration *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Totalizer 1*

144

Expert → Output → Relay output 1 to n → Assign limit (0807–1 to n)

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Totalizer 2• Totalizer 3• Oscil. damping
Factory setting

Mass flow

Assign diag. behNavigation

Expert → Output → Relay output 1 to n → Assign diag. beh (0806–1 to n)

Prerequisite

In the Relay outp.func. parameter (→  143), the Diag. behavior option is selected.

Description

Use this function to select the category of the diagnostic events that are displayed for therelay output.

Selection• Alarm• Alarm or warning• Warning

Factory setting

Alarm

Additional information

Description
If no diagnostic event is pending, the relay output is closed and conductive.
Selection• Alarm
The relay output signals only diagnostic events in the alarm category.
• Alarm or warning
The relay output signals diagnostic events in the alarm and warning category.
• Warning
The relay output signals only diagnostic events in the warning category.

Assign status



Navigation

Prerequisite

In the Relay outp.func. parameter (→  143), the Digital Output option is selected.

Description

Use this function to select the device status for the relay output.

Selection• Partial pipe det• Low flow cut off• Digital outp. 6

Factory setting

Partial pipe det

Endress+Hauser

Expert → Output → Relay output 1 to n → Assign status (0805–1 to n)

145Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Switch-off value

Navigation

Expert → Output → Relay output 1 to n → Switch-off value (0809–1 to n)

Prerequisite

In the Relay outp.func. parameter (→  143), the Limit option is selected.

Description

Use this function to enter the measured value for the switch-off point.

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 kg/h• 0 lb/min

Additional information

Description
Use this function to enter the limit value for the switch-off value (process variable <switch-off value = open, non-conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit is dependent on the process variable selected in the Assign limit parameter(→  144).

Switch-off delay

Navigation

Expert → Output → Relay output 1 to n → Switch-off delay (0813–1 to n)

Prerequisite

In the Relay outp.func. parameter (→  143), the Limit option is selected.

Description

Use this function to enter a delay time for switching off the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Switch-on value

Navigation

Prerequisite

In the Relay outp.func. parameter (→  143), the Limit option is selected.

Description

Use this function to enter the measured value for the switch-on point.

User entry

Signed floating-point number

146

Expert → Output → Relay output 1 to n → Switch-on value (0810–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Country-specific:• 0 kg/h• 0 lb/min

Additional information

Description

Description of device parameters

Use this function to enter the limit value for the switch-on value (process variable >switch-on value = closed, conductive).
When using a hysteresis: Switch-on value > Switch-off value.
Dependency
The unit is dependent on the process variable selected in the Assign limit parameter(→  144).

Switch-on delay

Navigation

Expert → Output → Relay output 1 to n → Switch-on delay (0814–1 to n)

Prerequisite

In the Relay outp.func. parameter (→  143), the Limit option is selected.

Description

Use this function to enter a delay time for switching on the switch output.

User entry

0.0 to 100.0 s

Factory setting

0.0 s

Failure modeNavigation

Expert → Output → Relay output 1 to n → Failure mode (0811–1 to n)

Description

Use this function to select the failure mode of the relay output in the event of a device
alarm.

Selection• Actual status• Open• Closed

Factory setting

Open

Additional information

Selection• Actual status
In the event of a device alarm, faults are ignored and the current behavior of the input
value is output by the relay output. The Actual status option behaves in the same way as
the current input value.
• OpenIn the event of a device alarm, the relay output's transistor is set to non-conductive.
• ClosedIn the event of a device alarm, the relay output's transistor is set to conductive.

Endress+Hauser

147

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Switch status
Navigation

Expert → Output → Relay output 1 to n → Switch status (0801–1 to n)

Description

Displays the current status of the relay output.

User interface• Open• Closed

Additional information

User interface• OpenThe relay output is not conductive.
• Closed
The relay output is conductive.

Powerless relay

Navigation

Expert → Output → Relay output 1 to n → Powerless relay (0816–1 to n)

Description

Use this function to select the quiescent state for the relay output.

Selection• Open• Closed

Factory setting

Open

Additional information

Selection• OpenThe relay output is not conductive.
• Closed
The relay output is conductive.

3.6"Communication" submenu

Navigation

Expert → Communication‣ Communication

148

Device address (11061)→  149‣ Resource block→  149‣ WLAN settings→  170‣ Web server→  176Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Device address
Navigation

Expert → Communication → Device address (11061)

Description

Displays the device address.

User interface

1 to 255

3.6.1"Resource block" submenu

Navigation

 Expert → Communication → Resource block‣ Resource block
Block tag (10702)→  149

Target mode (10728)→  151

Actual mode (10725)→  152

Manufacturer Id (10721)→  155

Device type (10711)→  155

Device revision (10710)→  156

DD Revision (10709)→  156

Restart (10800)→  157

Write Lock (10747)→  164

ITK Version (10794)→  170

Block tag
Navigation

 Expert → Communication → Resource block → Block tag (10702)

Description

Use this function to enter the Block tag: Used for specifying a "label" for identifying thefunction block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Endress+Hauser

149

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Static Revision
Navigation

Expert → Communication → Resource block → Static Revision (10735)

Description

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Communication → Resource block → Tag Description (10736)

Description

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
Navigation

Expert → Communication → Resource block → Strategy (10734)

Description

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0

Alert Key
NavigationDescription

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

150

Expert → Communication → Resource block → Alert Key (10696)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Target mode
Navigation

 Expert → Communication → Resource block → Target mode (10728)

Description

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Options• ROutIn the ROut operating mode, the set point (SP) for the function block is controlled by thefield bus host system, which runs over an interface, via the RIn parameter. The set point(SP) is used as an output signal after internal calculations. The output value and thestatus of the function block is communicated to the field bus host system as feedback via
the ROut parameter. The set point can be initiated or retained for the value of theprocess variables.
• RCasIn the RCas operating mode (external cascade mode), the set point (SP) for the function
block is controlled by the field bus host system, which runs over an interface, via theRCas-In parameter. The set point (SP) is used as an output signal after internal
calculations. The output value and status of the function block is communicated to thefield bus host system as feedback via the RCas-Out parameter.
• Cas
In the Cas (cascade mode) operating mode, the function block receives a discrete signal
via the function block input, which is provided by the Cas-In parameter of an upstream
function block. This signal controls the set point (SP) of the function block and is used as
an output signal after internal calculations. The output is used to communicate theoutput value and status to the upstream function block.
• AutoThe Auto operating mode is the normal operating mode of the function block. The set
point (SP) is specified locally in the function block and is used as an output signal after
internal calculations. This set point can be specified by the user via an interface.
• Man
In the Man operating mode, the output value can be specified directly in the function
block. This is specified by the user via an interface. There is no internal calculation. The
algorithm is initiated in such a way that there is no interruption when the operating
mode is changed. The set point can be retained or initiated for the value of the process
variables or for the set point of the previous operating mode if the system is switchedback to it.

Endress+Hauser

151

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• LO
The LO operating mode is used in control and output blocks that support a track input
parameter. The manufacturer can also provide a local lock switch on the device to enable
the LO operating mode. Tracking must be supported in the group of control parameters
and is initiated by a discrete track-in parameter.
In local bridge mode, the output value of the function block is set in order to track thevalue of the track-input parameter. The algorithm is initiated in such a way that there isno interruption when the operating mode switches from LO back to the previous
operating mode. The set point can be initiated or retained for the value of the process
variables.
• IManIn the IMan operating mode, the output value of the function block is performed as areaction to the status of the back-calculation-input parameter. When this statusindicates that there is no signal for the final output element, the control blocks ensurethat there is a smooth transition. The back-calculation-input parameter is supported by
all control and output blocks. The set point can be initiated or retained for the value ofthe process variables.
• OOS
In the OOS operating mode, implementation of the function block is blocked. Either thelast valid value or, in the case of an output block, the last valid set point is retained as theoutput value. This operating mode is used during configuration of the device.

Actual mode
Navigation

 Expert → Communication → Resource block → Actual mode (10725)

Description

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modethat the function block is currently operating in. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  151).

User interface• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

User interface
Detailed description of the options displayed: Target mode parameter (→  151)

Permitted mode
NavigationDescription

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  151) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

152

Expert → Communication → Resource block → Permitted mode (10727)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Options

Description of device parameters

Detailed description of the options available for selection: Target mode parameter(→  151)

Normal mode
Navigation

Expert → Communication → Resource block → Normal mode (10726)

Description

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Block Error
NavigationDescription

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten

Endress+Hauser

Expert → Communication → Resource block → Block Error (10703)

153Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Resource State
Navigation

Expert → Communication → Resource block → Resource State (10730)

Description

Displays the Resource State: Displays the current operating mode of the resource block.

User interface• Uninitialized• StartRestart• Initialization• Online Linking• Online• Standby• Failure

Additional information

User interface• Uninitialized
The resource block is in an invalid operating mode.
• StartRestart
The resource block is in the start-up or restore phase. The necessary memory andhardware tests are carried out for the current operation. If these tests are successful, theresource block switches to the Initialization operating mode. If these tests are notsuccessful, the resource block switches to the Failure operating mode.
• Initialization
The resource block is in this operating mode if the block successfully passes the
hardware tests from the StartRestart or Failure status. In this operating mode, all alarmsof the function block that are not displayed are automatically confirmed andacknowledged. When system management is operational, the implementation of theblock can be planned and the resource block switches to the Online Linking operatingmode.
• Online Linking
The resource block is in this operating mode when the block switches from theInitialization or Online operating mode. The configured connections between thefunction blocks are not yet established. When all configured connections have been
established, the resource block switches to the Online operating mode.

154

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Online
Normal operating mode, the resource block is in the Auto operating mode. The
configured connections between the function blocks have been established. If one of theconnections could not be established, the resource block switches back to the Online
Linking operating mode.
• Standby
The Resource Block is in the OOS option operating mode. It is not possible to implement
the remaining blocks. The operating mode of the transducer block is not necessarily
affected by this. Switching the resource block to the Auto mode causes the resource
block to switch back to the StartRestart operating mode.
• Failure
The resource block is in the fault state. It switches to this operating mode if a memory or
hardware fault that would hinder current operation has been detected in the block. The
fault can affect the block or the entire device. When this operating mode is active, blocks
with an output function are also in the fault state. The hardware test is carried out again.
If the error does not reoccur, the resource block switches back to the Initialization
operating mode.

DD Resource
Navigation

Expert → Communication → Resource block → DD Resource (10708)

Description

Displays the DD Resource: Indicates the reference source for the device description (DD).

User interface

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Additional information

User interface
Zero: No device description is stored on the device.

Manufacturer Id
Navigation

 Expert → Communication → Resource block → Manufacturer Id (10721)

Description

Manufacturer Id display: is used by the interface to assign the right DD file for the
resource.

User interface

Endress+Hauser

Device type
Navigation

 Expert → Communication → Resource block → Device type (10711)

Description

Displays the device type with which the measuring device is registered with Fieldbus
Foundation.

User interface

Promass 300/500

Endress+Hauser

155

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Device revision
Navigation

 Expert → Communication → Resource block → Device revision (10710)

Description

Use this function to view the device revision with which the measuring device is registeredwith Fieldbus Foundation.

User interface

1

DD Revision
Navigation

 Expert → Communication → Resource block → DD Revision (10709)

Description

Displays the revision number of the device description (DD).

User interface

1

Additional information

Description
This display can be used to ensure that the right system files (DD = device description)
are used for integration into the host system. The system files can be downloaded freeof charge online at: www.endress.com.

Grant
Navigation

Expert → Communication → Resource block → Grant (10718)

Description

Option for releasing certain access authorizations of the field bus host system on thedevice.

Selection• Program• Tune• Alarm• Local• Operate• Service• Diagnostic

Deny
NavigationDescription

Option for restricting certain access authorizations of the field bus host system on thedevice.

156

Expert → Communication → Resource block → Deny (10717)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection

Description of device parameters• Program Denied• Tune Denied• Alarm Denied• Local• Operate Denied

Hard Types
Navigation

Expert → Communication → Resource block → Hard Types (10719)

Description

Displays the input signal type for the Analog input block.

User interface• Scalar Input• Scalar Output• Discrete Input• Discrete Output

RestartNavigation

 Expert → Communication → Resource block → Restart (10800)

Description

Use this function to select a manual restart or a manual device reset.

Selection• Uninitialized• Run• Resource• Defaults• Processor• To delivery set.

Factory setting

Uninitialized

Service reset
NavigationDescription

Extended selection for a manual restart or a manual device reset.

Selection• Uninitialized• DeliverySett+MIB• ENP restart

Factory setting

Uninitialized

Endress+Hauser

Expert → Communication → Resource block → Service reset (10749)

157

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Selection• Uninitialized
Factory setting• DeliverySett+MIB
Reset the device to the as-delivered state. Important communication settings are reset tothe factory default settings here.
• ENP restart
Reset the parameters for the electronic name plate (ENP).

Features
Navigation

Expert → Communication → Resource block → Features (10713)

Description

Displays the additional options that are supported by the measuring device.

User interface• Reports• Faultstate• Soft W Lock
• Hard W Lock
• Chng Bypass Auto
• MVCReporDistrsup• Multibit AlmSupp• InterParamWrChk

Feature SelectNavigation

Expert → Communication → Resource block → Feature Select (10714)

Description

Use this function to select additional options that are supported by the measuring device.

Selection• Reports• Faultstate• Soft W Lock
• Hard W Lock
• Chng Bypass Auto
• MVCReporDistrsup• Multibit AlmSupp• InterParamWrChk

Cycle Type
NavigationDescription

Displays the implementation methods for the block that are supported by the measuring
device.

158

Expert → Communication → Resource block → Cycle Type (10707)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User interface

Description of device parameters• Scheduled• Block Execution

Cycle Selection
Navigation

Expert → Communication → Resource block → Cycle Selection (10706)

Description

Use this function to select the implementation method for the block that is used by thefield bus host system. This implementation method is selected by the field bus hostsystem.

Selection• Scheduled• Block Execution

Minumum Cyc.Time
Navigation

Expert → Communication → Resource block → Minumum Cyc.Time (10724)

Description

Displays the implementation time of all function blocks that are available in the measuring
device.

User interface

Positive integer

Memory Size
Navigation

Expert → Communication → Resource block → Memory Size (10723)

Description

Displays the available configuration memory in kilobytes.

User interface

0 to 65 535 Kbytes

Nonvolat CycTime
NavigationDescription

Displays the time interval during which the dynamic device parameters are stored in the
non-volatile memory.

User interface

Positive integer

Endress+Hauser

Expert → Communication → Resource block → Nonvolat CycTime (10729)

159

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Free Space
Navigation

Expert → Communication → Resource block → Free Space (10715)

Description

Displays the free system memory volume available for implementing further function
blocks as a percentage.

User interface

0.000000 to 100.000 %Free Time
Navigation

Expert → Communication → Resource block → Free Time (10716)

Description

Displays the free system time available for implementing further function blocks as apercentage.

User interface

0.000000 to 100.000 %Clear Fault Stat
Navigation

Expert → Communication → Resource block → Clear Fault Stat (10704)

Description

Use this function to select the fault state for the Discrete outputs (→  266) block.

Selection• Uninitialized• Off• Clear

Factory setting

Uninitialized

Additional information

Options
The fault state can be disabled with Clear option.

Confirm Time
NavigationDescription

Use this function to enter a time interval for confirming the event report. If the measuring
device does not receive a confirmation within this interval, the event report is sent to thefield bus host system again.

User entry

Positive integer

Factory setting

0 1/32 ms

160

Expert → Communication → Resource block → Confirm Time (10705)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Fault State
Navigation

Expert → Communication → Resource block → Fault State (10712)

Description

Displays the current status of the fault state for the Discrete outputs (→  266) block.

User interface• Uninitialized• Clear• Active

Additional information

User interface• Uninitialized• Clear
The fault state is disabled.
• Active
The fault state is enabled.

Limit Notify
Navigation

Expert → Communication → Resource block → Limit Notify (10720)

Description

Use this function to enter the maximum number of event reports that may be pending at
the same time without confirmation.

User entry

0 to 255

Factory setting0

Max Notify
Navigation

Expert → Communication → Resource block → Max Notify (10722)

Description

Displays the maximum number of event reports that are supported by the measuring
device and may be pending at the same time without confirmation.

User interface

0 to 255

Set Fault State
NavigationDescription

Option for manually enabling or disabling the fault state for the Discrete outputs(→  266) function block.

Endress+Hauser

Expert → Communication → Resource block → Set Fault State (10731)

161

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection

• OFF
• SET

Factory setting

OFF

Additional information

Options• OFF
The fault state is disabled.
• SET
The fault state is enabled.

Shed Remote Casc
Navigation

Expert → Communication → Resource block → Shed Remote Casc (10732)

Description

Use this function to enter the time interval after which the computer stops writing tofunction blocks with RCas points.

User entry

Positive integer

Factory setting

0 1/32 ms

Shed Remote Out
Navigation

Expert → Communication → Resource block → Shed Remote Out (10733)

Description

Use this function to enter the time interval after which the computer stops writing tofunction blocks with ROut points.

User entry

Positive integer

Factory setting

0 1/32 ms

Unacknowledged
NavigationDescription

Option for manually acknowledging an update of the static block parameter.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

162

Expert → Communication → Resource block → Unacknowledged (10740)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description• If a new diagnostic event occurs, the measuring device sets Unacknowledged
option.
• If the diagnostic event has been acknowledged, the user can set Acknowledged
option.

Update State
Navigation

Expert → Communication → Resource block → Update State (10741)

Description

Displays the status of an update of the static block parameter. The status indicates
whether the update was communicated or not.

User interface• Uninitialized• Reported• Not Reported

Time Stamp
Navigation

Expert → Communication → Resource block → Time Stamp (10739)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of an update to the static block parameter that has not yet beencommunicated was identified. The time stamp is retained until the update confirmation isreceived.

User interface

Days (d), hours (h), minutes (m), seconds (s)Static revision
Navigation

Expert → Communication → Resource block → Static revision (10738)

Description

Displays the Static revision: Each instance of a static block parameter being accessed withwrite and report access due to an update is counted (event counter).

User interface

0 to 65 535

Relative Index
NavigationDescription

Displays the relative index of the static block parameter that triggered the alarm (OD index
minus field bus start index).

Endress+Hauser

Expert → Communication → Resource block → Relative Index (10737)

163

Description of device parameters

User interface

Proline Promass 300 FOUNDATION Fieldbus

0 to 65 535

Write Lock
Navigation

 Expert → Communication → Resource block → Write Lock (10747)

Description

Option for enabling and disabling the hardware write protection.

Selection• Not Locked• Locked

Factory setting

Not Locked

Additional information

Description
Option for enabling and disabling write access to the measuring device via the
FOUNDATION Fieldbus (acyclic data transmission, e.g. via the "FieldCare" operating
program).
For detailed information on hardware write protection, see the "Write protection via
write protection switch" section of the Operating Instructions.
Options• Not Locked
Device data can be changed via the FOUNDATION Fieldbus interface.
• Locked
Device data cannot be changed via the FOUNDATION Fieldbus interface.

Unacknowledged
Navigation

Expert → Communication → Resource block → Unacknowledged (10700)

Description

Option for manually acknowledging a block alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

164

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Alarm State
Navigation

Expert → Communication → Resource block → Alarm State (10697)

Description

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

Time Stamp
Navigation

Expert → Communication → Resource block → Time Stamp (10699)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of the block alarm that has not yet been communicated to the field bus hostsystem was detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigationDescription

Displays the specific cause of the block alarm.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Endress+Hauser

Expert → Communication → Resource block → Subcode (10698)

165

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Value
Navigation

Expert → Communication → Resource block → Value (10701)

Description

Displays the value of the affected parameter at the time at which the block alarm was
detected.

User interface

0 to 255

Current
Navigation

Expert → Communication → Resource block → Current (10692)

Description

Use this function to view the current status of the process alarms.

User interface• Discrete Alarm• Block Alarm• Fail Alarm• Off Spec Alarm• Maint. Alarm• Check Alarm

Unacknowledged
Navigation

Expert → Communication → Resource block → Unacknowledged (10694)

Description

Displays an unacknowledged process alarm.

User interface• Disc Alm Unack• Block Alm Unack• Fail Alm Unack• Off SpecAlmUnack• Maint Alm Unack• Check Alm Unack

Unreported
NavigationDescription

Displays a process alarm that has not been communicated.

User interface• Disc Alm Unrep• Block Alm Unrep• Fail Alm Unrep

166

Expert → Communication → Resource block → Unreported (10695)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Off SpecAlmUnrep• Maint Alm Unrep• Check Alm Unrep

Disabled
Navigation

Expert → Communication → Resource block → Disabled (10693)

Description

Option for disabling a process alarm category.

Selection• Disc Alm Disabl• Block Alm Disabl• Fail Alm Disabl• OffSpecAlmDisabl• Maint Alm Disabl• Check Alm Disab.

Ack. Option
Navigation

Expert → Communication → Resource block → Ack. Option (10691)

Description

Option for automatic acknowledgment of process alarms in a specific category.

Selection• Disc Alm Aut Ack• Blk Alm Auto Ack• Fail Alm Aut Ack• OffSpecAlmAutAck• Maint Alm AutAck• Check Alm AutAck

Additional information

Description
Use this function to determine whether an alarm must be acknowledged via the field bus
host system.
If the process alarm option has not been enabled in this parameter, this process alarm
must only be acknowledged in Unacknowledged parameter (→  164).
Current parameter (→  166) indicates the current status of all process alarms.

Write Priority
NavigationDescription

Use this function to enter the priority for the write protection alarm.

User entry

0 to 15

Factory setting0

Endress+Hauser

Expert → Communication → Resource block → Write Priority (10748)

167

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
If write protection is disabled on the FOUNDATION Fieldbus I/O board, the alarm priority
entered here is checked before the status change is transferred to the field bus hostsystem. The alarm priority determines the behavior in the event of an active write
protection alarm.
The alarm is triggered when the write protection is disabled.

Unacknowledged
Navigation

Expert → Communication → Resource block → Unacknowledged (10745)

Description

Option for manually acknowledging a write protection alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Communication → Resource block → Alarm State (10742)

Description

Displays the status of the write protection alarm. The status indicates whether the write
protection alarm is active and whether it has already been communicated to the field bus
host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

Time Stamp
NavigationDescription

Displays the time stamp indicating when the analysis of the block was started and when
the status change of a write protection alarm that has not yet been communicated was
identified. The time stamp is retained until the alarm confirmation is received.

168

Expert → Communication → Resource block → Time Stamp (10744)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User interface

Description of device parameters

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigation

Expert → Communication → Resource block → Subcode (10743)

Description

Displays the specific cause of the write protection alarm, which is to be communicated to
the field bus host system.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Discrete Value
NavigationDescription

Displays the discrete value of the affected parameter at the time at which the write
protection alarm was detected.

User interface• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

Endress+Hauser

Expert → Communication → Resource block → Discrete Value (10746)

169

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

ITK Version
Navigation

 Expert → Communication → Resource block → ITK Version (10794)

Description

Displays the revision status of the Interoperability Test Kits (ITK).

User interface

6

3.6.2

"WLAN settings" submenu

Navigation

 Expert → Communication → WLAN settings‣ WLAN settings

170

WLAN (2702)→  171

WLAN mode (2717)→  171

SSID name (2714)→  171

Network security (2705)→  172

Sec. identific. (2718)→  172

User name (2715)→  172

WLAN password (2716)→  172

WLAN IP address (2711)→  173

WLAN MAC address (2703)→  173

WLAN subnet mask (2709)→  173

WLAN MAC address (2703)→  173

WLAN passphrase (2706)→  174

Assign SSID name (2708)→  174

SSID name (2707)→  174

WLAN channel (2704)→  175

Select antenna (2713)→  175

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Connection state (2722)→  175

Rec.sig.strength (2721)→  175

WLAN IP address (2711)→  173

Gateway IP addr. (2719)→  176

IP address DNS (2720)→  176

WLAN

Navigation

 Expert → Communication → WLAN settings → WLAN (2702)

Description

Use this function to enable and disable the WLAN connection.

Selection• Disable• Enable

Factory setting

EnableWLAN modeNavigation

 Expert → Communication → WLAN settings → WLAN mode (2717)

Description

Use this function to select the WLAN mode.

Selection• Access point• WLAN Client

Factory setting

Access point

SSID name

Navigation

 Expert → Communication → WLAN settings → SSID name (2714)

Prerequisite

The client is activated.

Description

Use this function to enter the user-defined SSID name (max. 32 characters).

User entry–

Factory setting–

Endress+Hauser

171

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Security type
Navigation

 Expert → Communication → WLAN settings → Security type (2705)
 Expert → Communication → WLAN settings → Network security (2705)

Description

Use this function to select the type of security for the WLAN interface.

Selection• Unsecured• WPA2-PSK

Factory setting

WPA2-PSK

Additional information

Selection• Unsecured
Access the WLAN connection without identification.
• WPA2-PSK
Access the WLAN connection with a network key.

Sec. identific.
Navigation

 Expert → Communication → WLAN settings → Sec. identific. (2718)

Description

Use this function to select the security settings (download via the menu: DataManagement > Security > Download WLAN).

User interface• Root certificate• Device certific.
• Dev. private key

User name

Navigation

 Expert → Communication → WLAN settings → User name (2715)

Description

Use this function to enter the user name.

User entry–

Factory setting–

WLAN passwordNavigation

 Expert → Communication → WLAN settings → WLAN password (2716)

Description

Use this function to enter the WLAN password.

172

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User entry–

Factory setting–

Description of device parameters

WLAN IP addressNavigation

 Expert → Communication → WLAN settings → WLAN IP address (2711)

Description

Use this function to enter the IP address of the measuring device's WLAN connection.

User entry

4 octet: 0 to 255 (in the particular octet)

Factory setting

192.168.1.212

WLAN MAC address
Navigation

 Expert → Communication → WLAN settings → WLAN MAC address (2703)

Description

Displays the MAC 8) address of the measuring device.

User interface

Unique 12-digit character string comprising letters and numbers

Factory setting

Each measuring device is given an individual address.

Additional information

ExampleFor the display format
00:07:05:10:01:5F

WLAN subnet mask

Navigation

 Expert → Communication → WLAN settings → WLAN subnet mask (2709)

Description

Use this function to enter the subnet mask.

User entry

4 octet: 0 to 255 (in the particular octet)

Factory setting

255.255.255.0

8)Media Access Control

Endress+Hauser

173

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

WLAN passphraseNavigation

 Expert → Communication → WLAN settings → WLAN passphrase (2706)

Prerequisite

The WPA2-PSK option is selected in the Security type parameter (→  172).

Description

Use this function to enter the network key.

User entry

8 to 32-digit character string comprising numbers, letters and special characters

Factory setting

Serial number of the measuring device (e.g. L100A802000)

Assign SSID name

Navigation

 Expert → Communication → WLAN settings → Assign SSID name (2708)

Description

Use this function to select which name is used for the SSID 9).

Selection• Device tag• User-defined

Factory settingUser-defined

Additional information

Selection• Device tag
The device tag name is used as the SSID.
• User-defined
A user-defined name is used as the SSID.

SSID name

Navigation

 Expert → Communication → WLAN settings → SSID name (2707)

Prerequisite• The User-defined option is selected in the Assign SSID name parameter (→  174).
• The Access point option is selected in the WLAN mode parameter (→  171).

Description

Use this function to enter a user-defined SSID name.

User entry

Max. 32-digit character string comprising numbers, letters and special characters

Factory setting

EH_device designation_last 7 digits of the serial number (e.g.
EH_Promass_300_A802000)

9)

174

Service Set Identifier

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

WLAN channel

Navigation

 Expert → Communication → WLAN settings → WLAN channel (2704)

Description

Use this function to enter the WLAN channel.

User entry

1 to 11

Factory setting

6

Additional information

Description• It is only necessary to enter a WLAN channel if multiple WLAN devices are in use.
• If just one measuring device is in use, it is recommended to keep the factory setting.

Select antennaNavigation

 Expert → Communication → WLAN settings → Select antenna (2713)

Description

Use this function to select whether the external or internal antenna is used for reception.

Selection• External antenna• Internal antenna

Factory setting

Internal antenna

Connection state
Navigation

Expert → Communication → WLAN settings → Connection state (2722)

Description

The connection status is displayed.

User interface• Connected• Not connected

Factory setting

Not connected

Rec.sig.strength
Navigation

 Expert → Communication → WLAN settings → Rec.sig.strength (2721)

Description

Displays the signal strength received.

User interface• Low• Medium• HighEndress+Hauser

175

Description of device parameters

Factory setting

Proline Promass 300 FOUNDATION Fieldbus

High

Gateway IP addr.
Navigation

 Expert → Communication → WLAN settings → Gateway IP addr. (2719)

Description

Use this function to enter the IP address of the gateway.

Factory setting

192.168.1.212

IP address DNSNavigation

Expert → Communication → WLAN settings → IP address DNS (2720)



Expert → Communication → WLAN settings → IP address DNS (2720)

Description

Use this function to enter the IP address of the domain name server.

Factory setting

192.168.1.212

3.6.3"Web server" submenu

Navigation

 Expert → Communication → Web server‣ Web server

176

Webserv.language (7221)→  177

MAC Address (7214)→  177

IP address (7209)→  178

Subnet mask (7211)→  178

Default gateway (7210)→  178

Webserver funct. (7222)→  179

Login page (7273)→  179

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Webserv.language
Navigation

 Expert → Communication → Web server → Webserv.language (7221)

Description

Use this function to select the Web server language setting.

Selection• English• Deutsch *• Français *• Español *• Italiano *• Nederlands *• Portuguesa *• Polski *• русский язык(Ru) *• Svenska *• Türkçe *• 中文 (Chinese) *• 日本語 (Japanese) *• 한국어 (Korean) *• Bahasa Indonesia *• tiếng Việt (Vit) *• čeština (Czech) *Factory setting

EnglishMAC Address
Navigation

Expert → Communication → Web server → MAC Address (7214)

Description

Displays the MAC 10) address of the measuring device.

User interface

Unique 12-digit character string comprising letters and numbers

Factory setting

Each measuring device is given an individual address.

Additional information

ExampleFor the display format
00:07:05:10:01:5F*
10)

Visibility depends on order options or device settings
Media Access Control

Endress+Hauser

177

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

DHCP client
Navigation

 Expert → Communication → Configuration → DHCP client (7212)


Setup → Communication → DHCP client (7212)

Description

Use this function to activate and deactivate the DHCP client functionality.

Selection

• Off
• On

Factory setting

Off

Additional information

ResultIf the DHCP client functionality of the Web server is activated, the IP address (→  178),
Subnet mask (→  178) and Default gateway (→  178) are set automatically.
Identification is via the MAC address of the measuring device.

IP addressNavigation

 Expert → Communication → Web server → IP address (7209)

Description

Displays the IP address of the Web server of the measuring device.

User interface

4 octet: 0 to 255 (in the particular octet)

Factory setting

192.168.1.212

Subnet mask

Navigation

 Expert → Communication → Web server → Subnet mask (7211)

Description

Displays the subnet mask.

User interface

4 octet: 0 to 255 (in the particular octet)

Factory setting

255.255.255.0

Default gateway

Navigation

 Expert → Communication → Web server → Default gateway (7210)

Description

Displays the default gateway.

178

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User interface

4 octet: 0 to 255 (in the particular octet)

Factory setting

0.0.0.0

Description of device parameters

Webserver funct.



Navigation

 Expert → Communication → Web server → Webserver funct. (7222)

Description

Use this function to switch the Web server on and off.

Selection• Off• HTML Off• On

Factory setting

On

Additional information

Description
Once disabled, the Webserver funct. can only be re-enabled via or the operating tool
FieldCare.
Options
OptionDescription

Off• The web server is completely disabled.
• Port 80 is locked.

On• The complete functionality of the web server is available.
• JavaScript is used.
• The password is transferred in an encrypted state.
• Any change to the password is also transferred in an encrypted state.

Login page

Navigation

 Expert → Communication → Web server → Login page (7273)

Description

Use this function to select the format of the login page.

Selection• Without header• With header

Factory settingWith header

3.7"Analog inputs" submenu

In the analog input function block (AI function block), the process variables from thetransducer block are prepared in terms of control for the subsequent automation functions

Endress+Hauser

179

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus(e.g. scaling, limit value processing). The automation function is defined byinterconnecting the outputs.
Navigation

 Expert → Analog inputs‣ Analog inputs‣ Analog input 1 to n

3.7.1

→  180"Analog input 1 to n" submenu

Navigation

 Expert → Analog inputs → Analog input 1 to n

‣ Analog input 1 to n
Block tag (6901–1 to n)→  180

Status (6906–1 to n)→  186

Value (6907–1 to n)→  189

Channel (6902–1 to n)→  208

Lin Type (6905–1 to n)→  209

PV Filter Time (6909–1 to n)→  214Block tag
Navigation

 Expert → Analog inputs → Analog input 1 to n → Block tag (6901–1 to n)

Description

Use this function to enter the Block tag: specify a "label" for identifying the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e. g. @, %, /).

Static Revision
NavigationDescription

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

180

Expert → Analog inputs → Analog input 1 to n → Static Revision (6973–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Analog inputs → Analog input 1 to n → Tag Description (6974–1 to n)

Description

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
Navigation

Expert → Analog inputs → Analog input 1 to n → Strategy (6972–1 to n)

Description

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0

Alert Key
Navigation

Expert → Analog inputs → Analog input 1 to n → Alert Key (6916–1 to n)

Description

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

Target mode
NavigationDescription

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Target mode (6960–1 to n)

181

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Actual mode
Navigation

Expert → Analog inputs → Analog input 1 to n → Actual mode (6957–1 to n)

Description

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modein which the function block is currently operating. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  181).

User interface• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

User interface
Detailed description of the options displayed: Target mode parameter (→  151)

Permitted mode
NavigationDescription

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  181) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man

182

Expert → Analog inputs → Analog input 1 to n → Permitted mode (6959–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• LO• IMan• OOS
Factory setting• Auto• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Normal mode
Navigation

Expert → Analog inputs → Analog input 1 to n → Normal mode (6958–1 to n)

Description

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

Auto

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Block Error
NavigationDescription

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Block Error (6922–1 to n)

183

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

StatusNavigationDescription

Displays the status of the input value (PV).

User interface• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)

184

Expert → Analog inputs → Analog input 1 to n → Status (6964–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)
Endress+Hauser

185

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)

Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Value (6965–1 to n)

Description

Displays the input value (PV).

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

StatusNavigation

 Expert → Analog inputs → Analog input 1 to n → Status (6906–1 to n)

Description

Displays the status of the output value (OUT).

User interface• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)

186

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)
Endress+Hauser

187

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)

188

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Value
Navigation

 Expert → Analog inputs → Analog input 1 to n → Value (6907–1 to n)

Description

Input for the output value (OUT).

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

Simulate Status
NavigationDescription

Use this function to select the status that is used for the transducer block simulation.

Selection• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Simulate Status (6967–1 to n)

189

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)
190

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)
Factory setting

Bad (0x00)

Simulate Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Simulate Value (6968–1 to n)

Description

Use this function to enter the value that is used for the transducer block simulation.

User entry

Signed floating-point number

Factory setting0%

Additional information

Dependency
The unit is taken from the: Units index parameter (→  195)

Endress+Hauser

191

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Transducer Stat
NavigationDescription

Displays the current status of the transducer block.

User interface• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)

192

Expert → Analog inputs → Analog input 1 to n → Transducer Stat (6969–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)
Endress+Hauser

193

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)

Transducer Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Transducer Value (6970–1 to n)

Description

Displays the current value of the transducer block.

User interface

Signed floating-point number

Factory setting0%

Additional information

Dependency
The unit is taken from the: Units index parameter (→  195)

Sim. En/Disable
Navigation

Expert → Analog inputs → Analog input 1 to n → Sim. En/Disable (6966–1 to n)

Description

Use this function to enable or disable the function block simulation.

Selection• Uninitialized• Disabled• Active

Factory setting

Uninitialized

EU at 100%
NavigationDescription

Use this function to enter the upper limit value of the output value measuring range.

User entry

Signed floating-point number

194

Expert → Analog inputs → Analog input 1 to n → EU at 100% (6982–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting0%

Additional information

Dependency

Description of device parameters

The unit is taken from the: Units index parameter (→  195)

EU at 0%Navigation

Expert → Analog inputs → Analog input 1 to n → EU at 0% (6981–1 to n)

Description

Use this function to enter the lower limit value of the output value measuring range.

User entry

Signed floating-point number

Factory setting0%

Additional information

Dependency
The unit is taken from the: Units index parameter (→  195)

Units index
NavigationDescription

Use this function to select the unit for the output value.

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Units index (6983–1 to n)

195Description of device parameters

Selection

196

SI units• g/s• g/min• g/h• g/d• kg/s• kg/min• kg/h• kg/d• t/s• t/min
• t/h
• t/d
• l/s
• l/min• l/d• l/h• m³/s• m³/min• m³/h• m³/d• cm³/s• cm³/min• cm³/h• cm³/d• dm³/s• dm³/min• dm³/h• dm³/d• ml/s• ml/min• ml/h• ml/d• Nl/s• Nl/d• Nl/h• Nl/min• Nm³/s• Nm³/min• Nm³/h• Nm³/d• Sm³/s• Sm³/min• Sm³/h• Sm³/d• MSft³/d• kg/l• kg/m³• kg/dm³• g/cm³• g/m³• SD4°C• SD15°C• SD20°C• SG4°C• SG20°C• SG15°C• kg/Nl• g/Scm³

Proline Promass 300 FOUNDATION Fieldbus

US units• oz/s• oz/min• oz/h• oz/d• lb/s• lb/min• lb/h• lb/d• STon/s• STon/min• STon/h• STon/d• af/s• af/min• af/h• af/d• ft³/s• ft³/min• ft³/h• ft³/d• fl oz/s (us)• fl oz/min (us)• fl oz/h (us)• fl oz/d (us)• gal/s (us)• gal/min (us)• gal/h (us)• gal/d (us)• Mgal/s (us)• Mgal/min (us)• Mgal/h (us)• Mgal/d (us)• bbl/s (us;oil)• bbl/min (us;oil)• bbl/h (us;oil)• bbl/d (us;oil)• bbl/s (us;liq.)• bbl/min (us;liq.)• bbl/h (us;liq.)• bbl/d (us;liq.)• bbl/s (us;beer)• bbl/min (us;beer)• bbl/h (us;beer)• bbl/d (us;beer)• bbl/s (us;tank)• bbl/min (us;tank)• bbl/h (us;tank)• bbl/d (us;tank)• kgal/s (us)• kgal/min (us)• kgal/h (us)• kgal/d (us)• Sft³/s• Sft³/min• Sft³/h• Sft³/d• Sgal/s (us)• Sgal/min (us)Imperial units• gal/s (imp)• gal/min (imp)• gal/h (imp)• gal/d (imp)• Mgal/s (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• bbl/s (imp;oil)• bbl/min (imp;oil)• bbl/h (imp;oil)• bbl/d (imp;oil)• bbl/s (imp;beer)• bbl/min (imp;beer)• bbl/h (imp;beer)• bbl/d (imp;beer)• Sgal/s (imp)• Sgal/min (imp)• Sgal/h (imp)• Sgal/d (imp)• lb/gal (imp)• lb/bbl (imp;oil)• Mgal (imp)• bbl (imp;beer)• Sgal (imp)• bbl (imp;oil)• gal (imp)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus• kg/Sm³• kg/Nm³
•K
• °C
•%
•g
• Nl
• Nm³
• Sl
• Sm³
• ml
• kg
•l
•t
• m³
• dm³
• cm³
• mA

Factory setting%

Additional information

ResultDescription of device parameters• Sgal/d (us)• Sgal/h (us)• Sbbl/s (us;liq.)• Sbbl/min (us;liq.)• Sbbl/h (us;liq.)• Sbbl/d (us;liq.)• MSft³/D• lb/ft³• lb/gal (us)• lb/bbl (us;tank)• lb/bbl (us;liq.)• lb/bbl (us;beer)• lb/bbl (us;oil)• lb/Sft³
• °F
• °R
• oz
• af
• fl oz (us)• Mgal (us)• bbl (us;oil)• bbl (us;liq.)• bbl (us;tank)• Sgal (us)• Sbbl (us;liq.)• Sft³• kgal (us)
• lb
• ft³
• gal (us)• bbl (us;beer)• STonThe selected unit applies for:• Simulate Value parameter (→  191)
• Transducer Value parameter (→  194)
• EU at 0% parameter (→  195)
• EU at 100% parameter (→  194)

DecimalNavigationDescription

Use this function to enter the number of decimal places for the output value.

User entry–128 to 127

Factory setting0

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Decimal (6980–1 to n)

197

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

EU at 100%
Navigation

Expert → Analog inputs → Analog input 1 to n → EU at 100% (6963–1 to n)

Description

Use this function to enter the upper limit value of the input value measuring range fromthe transducer block (input value).

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

EU at 0%Navigation

Expert → Analog inputs → Analog input 1 to n → EU at 0% (6962–1 to n)

Description

Function for entering the lower limit value of the input value measuring range from thetransducer block (input value).

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

Units index
NavigationDescription

Use this function to select the unit for the input value from the transducer block (input
value).

198

Expert → Analog inputs → Analog input 1 to n → Units index (6908–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection

Endress+Hauser

SI units•K•m• m³• Pa s• m²/s
•P
• cP
• St
• cSt
• Wbm
• Ns/m
• 1/Jm³• e/Vm³• m³/C• (1/32 millisec)/min
• °C
•'
•'
• rad
•°
• gon
• µm
• mm
• cm³
• dm³
• hl
•l
• ml
•s
• min
•d
•h
• ks
• µs
• ms
• m/h
• m/s
• mm/s
• Hz
•g
• kg
• GWh
•J
• kWh
• MWh
• kcal• Mcal• kW• MW•W• MJ/h
• mV
• Ohm
• pF
•V
• Ml Mega• mbar a• bar

Description of device parameters

US units• mils
• °F
• °R
• ft
• in
• bbl (us;liq.)• Sft³• in/min• oz• STon• lb/d• lb/h• lb/min• lb/s• STon/d• STon/h• STon/min• STon/s• mile
• yd
• ft²
• in²
• mile²• yd²• ft³• gal (us)• quart• pint• yd³• mile³• in³• bushel• bbl (us;oil)• ft/s• ft/h• lb• ft/min• in/h• in/s• yd/h• yd/min• yd/s• lb/in³• lb/gal (us)
• STon/yd³• psi• psi a• psi g• ftlbf• hp• lb/ft³• MPH• ft³/d• ft³/h• ft³/min• ft³/s• Sft³/h• Sft³/min

Imperial units• lbf/in• gal (imp)• Btu• LTon• datherm• Btu/h• Btu/day• Btu/min• Btu/s• LTon/d• LTon/h• LTon/min• LTon/s• gal/d (imp)• gal/h (imp)• gal/min (imp)• gal/s (imp)• lb/gal (imp)• Mgal/min (imp)• Mgal/h (imp)• Mgal/d (imp)• kgal/d (imp)• kgal/h (imp)• kgal/min (imp)• kgal/s (imp)• µgal/d (imp)• µgal/h (imp)• µgal/min (imp)• µgal/s (imp)• mgal/d (imp)• mgal/h (imp)• mgal/min (imp)• mgal/s (imp)• Mgal/s (imp)• µbbl/d (us;oil)• µbbl/h (us;oil)• µbbl/min (us;oil)• µbbl/s (us;oil)• kImpGal• Btu/lb• oz/ft199

Description of device parameters• dB• kPa a• kPa g• MPa a• MPa g• Pa a• Pa g• g/d• g/h• g/min• g/s• kg/d• kg/h• kg/min• kg/s
• cm
• km
• nm
• pm
•a
• ha
• cm²
• dm²
• km²
• m²
• mm²
• mm³
• cl
•t
• kg/m³• kg/dm³• g/cm³• g/m³• kg/l• kgf/cm²
• GJ
• kJ
• MJ
• km/h• kt• m/s²
• GHz
• kHz
• MHz
• 1/min• 1/s• THz• rad/s• 1/s²
• Mg
• mg
• g/l
• g/ml• Mg/m³• t/m³• mg/m• tex• kg/m• kgm/s
200

Proline Promass 300 FOUNDATION Fieldbus• gal/d (us)• gal/h (us)• gal/min (us)• gal/s (us)• Mgal/d (us)• bbl/d (us;oil)• bbl/h (us;oil)• bbl/min (us;oil)• bbl/s (us;oil)• Mgal/h (us)• Mgal/min (us)• Mgal/s (us)• Mgal (us)• af
• af/d• af/h• af/min• af/s• bbl/d (us;beer)• bbl/h (us;beer)• bbl/min (us;beer)• bbl/s (us;beer)• kgal/d (us)• kgal/h (us)• kgal/min (us)• kgal/s (us)• µgal/d (us)• µgal/h (us)• µgal/min (us)• µgal/s (us)• mgal/d (us)• mgal/h (us)• mgal/min (us)• mgal/s (us)• Mbbl/d (us;oil)• Mbbl/h (us;oil)• Mbbl/min (us;oil)• Mbbl/s (us;oil)• mbbl/d (us;oil)• mbbl/h (us;oil)• mbbl/min (us;oil)• mbbl/s (us;oil)• kft³/d• kft³/h• kft³/min• kft³/s• mft³/d• mft³/h• mft³/min• mft³/s• kbbl(US Beer)/d• kbbl(US Beer)/h• kbbl(US Beer)/min• ubbl(US Beer)/min• ubbl(US Beer)/s• mbbl(US Beer)/d• mbbl(US Beer)/h• mbbl(US Beer)/min• mbbl(US Beer)/s
Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus• kgm²• kgm²/s
• kNm
• MNm
• mNm
• Nm
• kN
• MN
• µN
• mN
•N
• mN/m
• N/m
• atm
• GPa
• hPa
• kPa
• MPa
• µPa
• mPa
• Pa
• torr• gf/cm²
• cal
• EJ
• mJ
• PJ
• TJ
• TWh
• Wh
• GW
• µW
• mW
• nW
• pW
• TW
• Mcal/h• kcal/d• kcal/h• kcal/min• kcal/s• Mcal/d• Mcal/min• Mcal/s• kJ/d• kJ/h• kJ/min• kJ/s
•A
• mA
• kA
• µA
• nA
• pA
•C
• kC
• MC
• µC
• nC
Endress+Hauser

Description of device parameters• kgal (us)• ac-in/d• ac-in/h• ac-in/m• ac-in/s• Mft³/d• ac-in• Mft³• inH2Oa• inH2Og• inH2O a(4°C)• inH2Og(4°C)• inH2O a(68°F)• inH2Og(68°F)• ftH2Oa• ftH2Og• ftH2O a(4°C)• ftH2Og(4°C)• ftH2O a(68°F)• ftH2Og(68°F)• inHga• inHgg• inHg a(0°C)• inHgg(0°C)• klb(US)/d• klb(US)/h• klb(US)/min• klb(US)/s• MSft³/D• mils/yr• ft/s²• MLB/H• lbf-in/deg

201

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

• pC
• Ah
• W/mK• m2K/W• W/m²K• J/K• kJ/K• J/(kgK)• kJ/(kgK)• J/kg• kJ/kg• MJ/kg• C/cm³• C/m³• C/mm³• kC/m³• µC/m³• mC/m³• C/cm²• C/m²• C/mm²• kC/m²• µC/m²• mC/m²• kV/m• MV/m
• µV/m• mV/m
• V/cm
• V/m
• kV
• MV
• µV
•F
• µF
• mF
• nF
• F/m
• A/cm²• kA/m²• MA/m²• A/cm• A/m
• kA/m
• µT
• mT
• nT
•T
• mWb
• Wb
• kWb/m• Wb/m
•H
• µH
• mH
• nH
• pH
• H/m
• µH/m202

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• nH/m• Am²• GOhm
• kOhm• MOhm• µOhm• mOhm• kS• µS• µS/cm
• mS
•S
• t/d
• t/h
• t/min• t/s•%• m³/d• m³/h• m³/min• m³/s• GOhmm• kOhmm• MOhmm• µOhmm• mOhmm• nOhmm• Ohmcm• Ohmm
• kS/m• MS/m• µS/mm• mS/cm
• S/m
• sr
• l/d
• l/h
• l/min• l/s• Ml/d• kW/m²• W/(sr-m²)• cd• cd/m²• lm• lm/m²• lm/W
• lmh
• lms
• lx
• lxs
• µW/m²• mW/m²• pW/m²• Pas/m³• Pas/m• ppm• MJ/d• MJ/minEndress+Hauser

203

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• MJ/s• cm³/d• cm³/h• cm³/min• cm³/s• Nm³• Nm³/d• Nm³/h• Nm³/min• Nm³/s• Sm³• Sm³/d• Sm³/h• Sm³/min• Sm³/s• Nl• Nl/d• Nl/h• Nl/min• Nl/s• Sl• ml/min•B• ppb• ppth• °Balling• km³/d• km³/h• km³/min• km³/s• Mm³/d• Mm³/h• Mm³/min• Mm³/s• µm³/d• µm³/h• µm³/min• µm³/s• mm³/d• mm³/h• mm³/min• mm³/s• kl/d• kl/h• kl/min• kl• Sl/d• Sl/h• Sl/min• Sl/s• kL/s• Ml/h• Ml/min• Mm³/d• Mm³• GPa a• GPa g• mPa a• mPa g204

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• µPa a• µPa g• hPa a• hPa g• gf/cm² a• gf/cm² g• kgf/cm² a• kgf/cm² g• mBarg• µbar• Gy• kcal/kg
• mGy
• mSv
• rad
• rem
• Sv
• Bq
• kBq
• MBq
• cnt/s• MSft³/d• SCCM• dm• mm/yr• g/m• µg/m³• µg/l• mg/m³• kmol• µmol• mmol• mol• mol/dm³• mol/m³• mol/l• cm³/mol• dm³/mol• m³/mol• g/mol• kg/mol• l/mol• mmol/kg• mol/kg• mg/l• µS/m• mS/m• nS/cm• S/cm
• kOhmcm• MOhmcm• l/m³• L/m• µL/L• ml/m³• ml/l
• %Sat• % sol/vol• % sol/mass
Endress+Hauser

205Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• %vol
• WT-%• J/mol• kJ/mol• J/(molK)• Bq/kg• kBq/kg• MBq/kg• mV/K• V/K• J/g• mV/pH• pH• pH/°C• mV/%• %/s• %/V• nA/ppm• 1/32 ms
• 1/H
• /cm
• 1/K
• 1/m
• 1/mm
• A/hPa• A/Pa• Nm²/A• Pa/A• pA/hPa• C/kg• mC/kg• dyne-cm/deg• newton-m/deg• bar a• bar gCustom-specific units
PV/SecFactory setting

K

Additional information

ResultThe selected unit applies for:• Value parameter (→  186)
• Value parameter (→  189)
• EU at 100% parameter (→  194)
• EU at 0% parameter (→  195)
• Low Cutoff parameter (→  211)• High High Limit parameter (→  219)
• Float Value parameter (→  223)
• High Limit parameter (→  220)
• Float Value parameter (→  225)
• Low Limit parameter (→  221)
• Float Value parameter (→  226)
• Low Low Limit parameter (→  221)
• Float Value parameter (→  228)

206

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

DecimalNavigation

Expert → Analog inputs → Analog input 1 to n → Decimal (6961–1 to n)

Description

Use this function to select the number of decimal places for the input value from thetransducer block (input value).

User entry–128 to 127

Factory setting

0

Grant
Navigation

Expert → Analog inputs → Analog input 1 to n → Grant (6926–1 to n)

Description

Option for releasing certain access authorizations of the field bus host system on thedevice.

Selection• Program• Tune• Alarm• Local• Operate• Service• Diagnostic

Deny
Navigation

Expert → Analog inputs → Analog input 1 to n → Deny (6925–1 to n)

Description

Option for restricting certain access authorizations of the field bus host system on thedevice.

Selection• Program Denied• Tune Denied• Alarm Denied• Local• Operate Denied

I/O Options
NavigationDescription

Option for activating the low flow cut off.

Selection

Low Cutoff

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → I/O Options (6941–1 to n)

207

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
The limit value for the low flow cut off is defined in Low Cutoff parameter(→  211).

ChannelNavigation

 Expert → Analog inputs → Analog input 1 to n → Channel (6902–1 to n)

Description

Use this function to select the input value that should be processed in the Analog Input
function block.

Selection• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. ampl. 0• Freq. fluct. 0• Osc. damping 0• Osc.damp.fluct 0• Signal asymmetry• Exc. current 0• HBSI *• Totalizer 1• Totalizer 2• Totalizer 3• Current input 1 *• Uninitialized

Factory setting

Uninitialized

Status Options
Navigation

Prerequisite

The measuring device must be in the OOS operating mode so that the parameter can beedited.

*208

Expert → Analog inputs → Analog input 1 to n → Status Options (6971–1 to n)

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to select an option for the status of the output value that is supported by
the analog input block.

Selection• Propag Fault Fwd• Uncertain if Lim• Bad if Limited• Uncertain if Man

Additional information

Options• Propag Fault Fwd
If the measuring device has the status Bad DeviceFailure or Bad SensorFailure, thedevice continues to measure and no alarm is triggered. The use of this substatus in the
output value (OUT) is defined by Propag Fault Fwd option. With the aid of this option,
the user/operator can specify whether the alarm generation (sending an alarm) is
triggered by the block or is forwarded downstream.
• Uncertain if Lim
If the measured or calculated value reaches a limit value, Uncertain if Lim option is used
for the output status.
• Bad if Limited
If the measured value exceeds or falls below an upper or lower limit value, Bad if
Limited option is used as the output status.
• Uncertain if Man
If the Actual mode of the function block is in the Man option operating mode, Uncertain
if Man option is used as the output status.

Lin Type
Navigation

 Expert → Analog inputs → Analog input 1 to n → Lin Type (6905–1 to n)

Description

Use this function to select the type of linearization of the input value or simulation value.

Selection• Uninitialized• Direct• Indirect• Indirect Sq Root

Factory setting

Uninitialized

Additional information"Uninitialized" option
The function block does not switch to Auto operating mode.
"Direct" option
The measured value from the transducer block (input value) bypasses the linearization
function and is routed unchanged through the analog input function block (Xd_Scale =Out_Scale). This is selected if the input value already has the required physical units.
PV = Input value

Endress+Hauser

209

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

The units in Units index (→  195) (Xd_Scale) and Units index (→  198) (Out_Scale)must be the same. Otherwise, the function block will remain in the OOS operating modeand the BlockConfigurat block error is displayed in Block Error (→  183).
"Indirect" option
The measured value from the transducer block (input value) is linearly rescaled via the
Xd_Scale input scaling to the required Out_Scale output range.

PV =X⋅ (Y - Z) -Z
100
A0024820

PV
X
YZ

Primary value
Value (→  214) (Field_Val)
EU at 100% (→  198) (Out_Scale)
EU at 0% (→  198) (Out_Scale)"Indirect Sq Root" option
The measured value from the transducer block (input value ) is rescaled via the Xd_Scaleparameter group and recalculated using a root function. It is then rescaled again to therequired output range via the Out_Scale parameter group.
PV = √X⋅ (Y - Z) -Z
100
A0024847PVX
YZ

Primary value
Value (→  214) (Field_Val)
EU at 100% (→  198) (Out_Scale)
EU at 0% (→  198) (Out_Scale)Example• System unit in transducer block: kg/h
• Sensor measuring range: 0 to 30 kg/h
• Output range for the automation system: 0 to 100 %The analog input function block must be configured as follows:1.

In the Channel parameter (→  208), select the Mass flow option.

2.

In the Lin Type parameter (→  209), select the Indirect option.
 The "Volume Flow" process variable of the transducer block "Flow" is linearly
rescaled in the AI block to the required Out_Scale output range via the Xd_Scale
input scaling.

3.

In the Xd_Scale parameter group: EU at 0% parameter (→  195), enter the value 0.
In EU at 100% parameter (→  194), enter the value 30.
In the Units index parameter (→  195), select the kg/h option.

4.

In the Out_Scale parameter group: EU at 0% parameter (→  198), enter the value 0.
In EU at 100% parameter (→  198), enter the value 100.
In the Units index parameter (→  198), select the % option.

The result is that an input value of 15 kg/h, for example, outputs a value of 50 % via the
Value parameter (→  189).

210

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

AI
A
30

B %
100

TB
50 %

15
0

100 %

0

%
100

0…30
A0027100

9Engineering unit in kg/h

Navigation

Expert → Analog inputs → Analog input 1 to n → Low Cutoff (6956–1 to n)

Description

Use this function to enter a limit value for low flow cut off.

User entry

Positive floating-point number

Factory setting0

Additional information

Description

Low Cutoff

If the converted input value (PV) falls below this limit value, then it is set to zero.
The low flow cut off is enabled via I/O Options parameter (→  207).
Dependency
The unit is taken from the: Units index parameter (→  198)

StatusNavigationDescription

Displays the status of the unprocessed measured value from the device, which reflects thestatus of the transducer before signal transmission.

User interface• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Status (6923–1 to n)

211

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)
212

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)

Endress+Hauser

213

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Value (6924–1 to n)

Description

Displays the unprocessed measured value from the device as a % of the primary value (PV).

User interface

Signed floating-point number

PV Filter Time
Navigation

 Expert → Analog inputs → Analog input 1 to n → PV Filter Time (6909–1 to n)

Description

Use this function to enter the filter time specification for the filtering of the unconverted
input value (PV).

User entry

Positive floating-point number

Factory setting

0s

Additional information

Factory setting
If the value 0 s is entered, filtering will not be performed.

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to n → Unacknowledged (6978–1 to n)

Description

Option for manually acknowledging an update of the static block parameter.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new diagnostic event occurs, the measuring device sets Unacknowledged
option.
• If the diagnostic event has been acknowledged, the user can set Acknowledged
option.

214

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Update State
Navigation

Expert → Analog inputs → Analog input 1 to n → Update State (6979–1 to n)

Description

Displays the status of an update of the static block parameter. The status indicates
whether the update was communicated or not.

User interface• Uninitialized• Reported• Not Reported

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to n → Time Stamp (6977–1 to n)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of an update to the static block parameter that has not yet beencommunicated was identified. The time stamp is retained until the update confirmation isreceived.

User interface

Days (d), hours (h), minutes (m), seconds (s)Static revision
Navigation

Expert → Analog inputs → Analog input 1 to n → Static revision (6976–1 to n)

Description

Displays the Static revision: Each instance of a static block parameter being accessed withwrite and communicate access due to an update is counted (event counter).

User interface

0 to 65 535

Relative Index
NavigationDescription

Displays the relative index of the static block parameter that triggered the alarm (OD index
minus field bus start index).

User interface

0 to 65 535

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Relative Index (6975–1 to n)

215

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to n → Unacknowledged (6920–1 to n)

Description

Option for manually acknowledging a block alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Blk Alm Auto Ack option is not enabled in Ack. Option parameter (→  218), theprocess alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to n → Alarm State (6917–1 to n)

Description

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

Time Stamp
NavigationDescription

Displays the time stamp indicating when the analysis of the block was started and when astatus change of the block alarm that has not yet been communicated to the field bus hostsystem was detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)216Expert → Analog inputs → Analog input 1 to n → Time Stamp (6919–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

SubcodeNavigation

Expert → Analog inputs → Analog input 1 to n → Subcode (6918–1 to n)

Description

Displays the specific cause of the block alarm.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Value (6921–1 to n)

Description

Displays the value of the affected parameter at the time at which the block alarm was
detected.

User interface

0 to 255

Current
NavigationDescription

Use this function to view the current status of the process alarms.

User interface• HiHi Alarm• Hi Alarm• LoLo Alarm• Lo Alarm• Block Alarm

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Current (6912–1 to n)

217

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to n → Unacknowledged (6914–1 to n)

Description

Displays an unacknowledged process alarm.

User interface• HiHi Alarm Unack• Hi Alm Unack• LoLo Alm Unack• Lo Alm Unack• Block Alm Unack

Unreported
Navigation

Expert → Analog inputs → Analog input 1 to n → Unreported (6915–1 to n)

Description

Displays a process alarm that has not been communicated.

User interface• HiHi Alm Unrep• Hi Alm Unrep• LoLo Alm Unrep• Lo Alm Unrep• Block Alm Unrep

Disabled
Navigation

Expert → Analog inputs → Analog input 1 to n → Disabled (6913–1 to n)

Description

Option for disabling a process alarm category.

Selection• HiHi Alm Disabl• Hi Alm Disabled• LoLo Alm Disabl• Lo Alm Disabled• Block Alm Disabl

Ack. Option
NavigationDescription

Option for automatic acknowledgment of process alarms in a specific category.

Selection• HiHi Alm Aut Ack• Hi Alm Auto Ack• LoLo Alm Aut Ack• Lo Alm Auto Ack

218

Expert → Analog inputs → Analog input 1 to n → Ack. Option (6910–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Blk Alm Auto Ack• Fail Alm Aut Ack• OffSpecAlmAutAck• Maint Alm AutAck• Check Alm AutAck
Additional information

Description
Use this function to determine whether an alarm must be acknowledged via the field bus
host system.
If the process alarm option has not been enabled in this parameter, this process alarm
must only be acknowledged in Unacknowledged parameter (→  218).
Current parameter (→  217) indicates the current status of all process alarms.

Alarm Hysteresis
Navigation

Expert → Analog inputs → Analog input 1 to n → Alarm Hysteresis (6911–1 to n)

Description

Use this function to enter the hysteresis value for the upper and lower warning or alarm
limit values.

User entry

0.000000 to 50.0000 %Factory setting0%

Hi Hi Priority
Navigation

Expert → Analog inputs → Analog input 1 to n → Hi Hi Priority (6938–1 to n)

Description

Use this function to enter the priority for the upper alarm limit, which determines the
behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting0

High High Limit
NavigationDescription

Use this function to enter the value for the upper alarm limit.

User entry

Signed floating-point number

Factory setting0

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → High High Limit (6937–1 to n)

219

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Dependency
The unit is taken from the: Units index parameter (→  198)

High Priority
Navigation

Expert → Analog inputs → Analog input 1 to n → High Priority (6940–1 to n)

Description

Use this function to enter the priority for the upper early warning limit, which determines
the behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting0

High Limit
Navigation

Expert → Analog inputs → Analog input 1 to n → High Limit (6939–1 to n)

Description

Use this function to enter the value for the upper early warning limit.

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

Low Priority
NavigationDescription

Use this function to enter the priority for the lower early warning limit, which determines
the behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting

0220

Expert → Analog inputs → Analog input 1 to n → Low Priority (6955–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Low Limit
Navigation

Expert → Analog inputs → Analog input 1 to n → Low Limit (6947–1 to n)

Description

Use this function to enter the value for the lower early warning limit.

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

Low Low Priority
Navigation

Expert → Analog inputs → Analog input 1 to n → Low Low Priority (6954–1 to n)

Description

Use this function to enter the priority for the lower alarm limit, which determines the
behavior in the event of an active limit value violation.

User entry

0 to 15

Factory setting0

Low Low Limit
Navigation

Expert → Analog inputs → Analog input 1 to n → Low Low Limit (6953–1 to n)

Description

Use this function to enter the value for the lower alarm limit.

User entry

Signed floating-point number

Factory setting0

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

Unacknowledged
NavigationDescription

Option for manually acknowledging a process alarm that has exceeded the upper alarm
limit (High High Limit (→  219)).

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Unacknowledged (6935–1 to n)

221Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If HiHi Alm Aut Ack option is not enabled in Ack. Option parameter (→  218), theprocess alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to n → Alarm State (6932–1 to n)

Description

Displays the status of the process alarm for the upper alarm limit. The status indicates
whether the process alarm is active and whether it has already been communicated to thefield bus host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to n → Time Stamp (6934–1 to n)

Description

Displays the time stamp of the process alarm for the upper alarm limit. This records thetime at which analysis of the block was started and at which a status change of the process
alarm that has not yet been communicated to the field bus host system was detected. The
time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigationDescription

Displays the specific cause of the process alarm for the upper alarm limit.

User interface• Other• BlockConfigurat• LinkConfigurat

222

Expert → Analog inputs → Analog input 1 to n → Subcode (6933–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Float Value (6936–1 to n)

Description

Displays the value of the affected parameter at the time at which the process alarm for theupper alarm limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to n → Unacknowledged (6930–1 to n)

Description

Option for manually acknowledging a process alarm that has exceeded the upper early
warning limit (High Limit (→  220)).

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Hi Alm Auto Ack option is not enabled in Ack. Option parameter (→  218), theprocess alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Endress+Hauser

223

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to n → Alarm State (6927–1 to n)

Description

Displays the status of the process alarm for the upper early warning limit. The statusindicates whether the process alarm is active and whether it has already beencommunicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to n → Time Stamp (6929–1 to n)

Description

Displays the time stamp of the process alarm for the upper early warning limit. This
records the time at which analysis of the block was started and at which a status change ofthe process alarm that has not yet been communicated to the field bus host system was
detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigationDescription

Displays the specific cause of the process alarm for the upper early warning limit.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

224

Expert → Analog inputs → Analog input 1 to n → Subcode (6928–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Float Value (6931–1 to n)

Description

Displays the value of the affected parameter at the time at which the process alarm for theupper early warning limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to n → Unacknowledged (6945–1 to n)

Description

Option for manually acknowledging a process alarm that has exceeded the lower early
warning limit (Low Limit (→  221)).

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Lo Alm Auto Ack option is not enabled in Ack. Option parameter (→  218), theprocess alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
NavigationDescription

Displays the status of the process alarm for the lower early warning limit. The statusindicates whether the process alarm is active and whether it has already beencommunicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Alarm State (6942–1 to n)

225

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Time Stamp
Navigation

Expert → Analog inputs → Analog input 1 to n → Time Stamp (6944–1 to n)

Description

Displays the time stamp of the process alarm for the lower early warning limit. This
records the time at which analysis of the block was started and at which a status change ofthe process alarm that has not yet been communicated to the field bus host system was
detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)SubcodeNavigation

Expert → Analog inputs → Analog input 1 to n → Subcode (6943–1 to n)

Description

Displays the specific cause of the process alarm for the lower early warning limit.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Float Value (6946–1 to n)

Description

Displays the value of the affected parameter at the time at which the process alarm for the
lower early warning limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

226

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Unacknowledged
Navigation

Expert → Analog inputs → Analog input 1 to n → Unacknowledged (6951–1 to n)

Description

Option for manually acknowledging a process alarm that has exceeded the lower alarm
limit (Low Low Limit (→  221)).

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If LoLo Alm Aut Ack option is not enabled in Ack. Option parameter (→  218), theprocess alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Alarm State
Navigation

Expert → Analog inputs → Analog input 1 to n → Alarm State (6948–1 to n)

Description

Displays the status of the process alarm for the lower alarm limit. The status indicates
whether the process alarm is active and whether it has already been communicated to thefield bus host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

Time Stamp
NavigationDescription

Displays the time stamp of the process alarm for the lower alarm limit. This records thetime at which analysis of the block was started and at which a status change of the process
alarm that has not yet been communicated to the field bus host system was detected. The
time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)Endress+Hauser

Expert → Analog inputs → Analog input 1 to n → Time Stamp (6950–1 to n)

227

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

SubcodeNavigation

Expert → Analog inputs → Analog input 1 to n → Subcode (6949–1 to n)

Description

Displays the specific cause of the process alarm for the lower alarm limit.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Float Value
Navigation

Expert → Analog inputs → Analog input 1 to n → Float Value (6952–1 to n)

Description

Displays the value of the affected parameter at the time at which the process alarm for the
lower alarm limit was triggered.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the: Units index parameter (→  198)

3.8

"Discrete inputs" submenu

Navigation

 Expert → Discrete inputs‣ Discrete inputs‣ Discrete input 1 to n

228

→  229

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

3.8.1

Description of device parameters"Discrete input 1 to n" submenu

Navigation

 Expert → Discrete inputs → Discrete input 1 to n

‣ Discrete input 1 to n
Block tag (6851–1 to n)→  229

Status (6853–1 to n)→  235

Value (6854–1 to n)→  238

Channel (6852–1 to n)→  246PV Filter Time (6855–1 to n)→  247

Block tag
Navigation

 Expert → Discrete inputs → Discrete input 1 to n → Block tag (6851–1 to n)

Description

Use this function to enter the Block tag: Used for specifying a "label" for identifying thefunction block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Static Revision
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Static Revision (6884–1 to n)

Description

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Tag Description
NavigationDescription

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Tag Description (6885–1 to n)

229

Description of device parameters

User entry

Proline Promass 300 FOUNDATION Fieldbus

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Strategy (6883–1 to n)

Description

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0

Alert Key
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Alert Key (6846–1 to n)

Description

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

Target mode
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Target mode (6873–1 to n)

Description

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

230

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Actual mode
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Actual mode (6870–1 to n)

Description

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modein which the function block is currently operating. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  230).

User interface• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

User interface
Detailed description of the options available for selection: Target mode parameter(→  151)

Permitted mode
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Permitted mode (6872–1 to n)

Description

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  230) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Endress+Hauser

231

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Normal mode
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Normal mode (6871–1 to n)

Description

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Block Error
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Block Error (6857–1 to n)

Description

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

StatusNavigationDescription

Displays the status of the discrete input value (PV).

232

Expert → Discrete inputs → Discrete input 1 to n → Status (6875–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User interface

Endress+Hauser

Description of device parameters• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)
233

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)
234

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)

Value
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Value (6876–1 to n)

Description

Displays the discrete process variable that is used for the block implementation.

User interface• Low flow or PFS Stat.=Not active• Low flow or PFS Stat.=Active or Verif. stat.=Check not done
• Verification status=failed• Verification status=busy• Verification status=ready• Verificaton overall result=failed• Status=Check not done/Result=failed• Status=failed/Result=failed• Status=busy/Result=failed• Status=ready/Result=failed• Verificaton overall result=passed• Status=Check not done/Result=passed• Status=failed/Result=passed• Status=busy/Result=passed• Status=ready/Result=passed• Verificaton overall result=Check not done
• Status=Check not done/Result=Check not done
• Status=failed/Result=Check not done
• Status=busy/Result=Check not done
• Status=ready/Result=Check not done

StatusNavigation

 Expert → Discrete inputs → Discrete input 1 to n → Status (6853–1 to n)

Description

Displays the status of the discrete output value.

User interface• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)

Endress+Hauser

235

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)
236

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)

Endress+Hauser

237

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Value
Navigation

 Expert → Discrete inputs → Discrete input 1 to n → Value (6854–1 to n)

Description

Use this function to select the discrete output value.

Selection• Low flow or PFS Stat.=Not active• Low flow or PFS Stat.=Active or Verif. stat.=Check not done
• Verification status=failed• Verification status=busy• Verification status=ready• State 15• Verificaton overall result=failed• Status=Check not done/Result=failed• Status=failed/Result=failed• Status=busy/Result=failed• Status=ready/Result=failed• Verificaton overall result=passed• Status=Check not done/Result=passed• Status=failed/Result=passed• Status=busy/Result=passed• Status=ready/Result=passed• Verificaton overall result=Check not done
• Status=Check not done/Result=Check not done
• Status=failed/Result=Check not done
• Status=busy/Result=Check not done
• Status=ready/Result=Check not done

Factory setting

Low flow or PFS Stat.=Not active

Simulate Status
NavigationDescription

Use this function to select the status that is used for the transducer block simulation.

Selection• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)

238

Expert → Discrete inputs → Discrete input 1 to n → Simulate Status (6878–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)
Endress+Hauser

239

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)
Factory setting240

Bad (0x00)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Simulate Value
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Simulate Value (6879–1 to n)

Description

Use this function to select the simulation value that is used for the transducer block
simulation.

Selection• Low flow or PFS Stat.=Not active• Low flow or PFS Stat.=Active or Verif. stat.=Check not done
• Verification status=failed• Verification status=busy• Verification status=ready• Verificaton overall result=failed• Status=Check not done/Result=failed• Status=failed/Result=failed• Status=busy/Result=failed• Status=ready/Result=failed• Verificaton overall result=passed• Status=Check not done/Result=passed• Status=failed/Result=passed• Status=busy/Result=passed• Status=ready/Result=passed• Verificaton overall result=Check not done
• Status=Check not done/Result=Check not done
• Status=failed/Result=Check not done
• Status=busy/Result=Check not done
• Status=ready/Result=Check not done

Factory setting

Low flow or PFS Stat.=Not active

Transducer Stat
NavigationDescription

Displays the current status of the transducer block.

User interface• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Transducer Stat (6880–1 to n)

241

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)
242

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)

Transducer Value
NavigationDescription

Displays the current value of the transducer block.

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Transducer Value (6881–1 to n)

243

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

User interface• Low flow or PFS Stat.=Not active• Low flow or PFS Stat.=Active or Verif. stat.=Check not done
• Verification status=failed• Verification status=busy• Verification status=ready• Verificaton overall result=failed• Status=Check not done/Result=failed• Status=failed/Result=failed• Status=busy/Result=failed• Status=ready/Result=failed• Verificaton overall result=passed• Status=Check not done/Result=passed• Status=failed/Result=passed• Status=busy/Result=passed• Status=ready/Result=passed• Verificaton overall result=Check not done
• Status=Check not done/Result=Check not done
• Status=failed/Result=Check not done
• Status=busy/Result=Check not done
• Status=ready/Result=Check not done

Factory setting

Low flow or PFS Stat.=Not active

Sim. En/Disable
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Sim. En/Disable (6877–1 to n)

Description

Use this function to enable or disable the function block simulation.

Selection• Uninitialized• Disabled• Active

Factory setting

Uninitialized

Transducer State
NavigationDescription

Use this function to enter the transducer state: this is required by FieldValD (Statusparameter (→  247), Value parameter (→  249)) to display the actual on/off statusof the hardware.

User entry

0 to 65 535

Factory setting

0244

Expert → Discrete inputs → Discrete input 1 to n → Transducer State (6891–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Output State
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Output State (6874–1 to n)

Description

Use this function to enter the output state: This is required for scaling the discrete input
value (PV).

User entry

0 to 65 535

Factory setting0

Deny
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Deny (6867–1 to n)

Description

Option for restricting certain access authorizations of the field bus host system on thedevice.

Selection• Program Denied• Tune Denied• Alarm Denied• Local• Operate Denied

Grant
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Grant (6868–1 to n)

Description

Option for releasing certain access authorizations of the field bus host system on thedevice.

Selection• Program• Tune• Alarm• Local• Operate• Service• Diagnostic

I/O Options
NavigationDescription

Option for activating the inversion of the signal.

Selection

InvertEndress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → I/O Options (6869–1 to n)

245

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
This selection determines whether the discrete input is inverted before it is stored as aprocess variable. Normally a discrete value of 0 is regarded as a logical value of 0 and adiscrete value that is not equal to 0 is regarded as a logical value of 1. If inversion isenabled, this correlation is reversed. A field device input value that is not equal to 0 results
in a discrete output value of 0 and an input value of 0 results in a discrete output value of1.

Status Options
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Status Options (6882–1 to n)

Prerequisite

The measuring device must be in the OOS operating mode so that the parameter can beedited.

Description

For selecting an option for the status of the output value that is supported by the Discrete
input block.

Selection• Propag Fault Fwd• Uncertain if Man

Additional information

Options• Propag Fault Fwd
If the measuring device indicates the status Bad DeviceFailure or Bad SensorFailure,
the device continues to measure and no alarm is triggered. The use of this substatus in
the output value (OUT) is defined by Propag Fault Fwd option. With the aid of thisoption, the user/operator can specify whether the alarm generation (sending an alarm)is triggered by the block or is forwarded downstream.
• Uncertain if Man
If the Actual mode of the function block is in the Man option operating mode, Uncertain
if Man option is used as the output status.

ChannelNavigation

 Expert → Discrete inputs → Discrete input 1 to n → Channel (6852–1 to n)

Description

Use this function to select the input value that should be processed in the discrete input
function block.

Selection• Uninitialized• Empty pipe det.
• Low flow cut off• Switch out.stat.
• Verific. status

Factory setting

Empty pipe det.

246

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

PV Filter Time
Navigation

 Expert → Discrete inputs → Discrete input 1 to n → PV Filter Time (6855–1 to n)

Description

Use this function to enter the filter time specification for the filtering of the unconverted
input value (PV).

User entry

Positive floating-point number

Factory setting

0s

Additional information

Factory setting
If the value 0 s is entered, filtering will not be performed.

StatusNavigationDescription

Displays the status of the discrete input value from a measuring device in the field.

User interface• Bad (0x00)• Bad (0x01)• Bad (0x02)• Bad (0x03)• Bad (0x04)• Bad (0x05)• Bad (0x06)• Bad (0x07)• Bad (0x08)• Bad (0x09)• Bad (0x0A)• Bad (0x0B)• Bad (0x0C)• Bad (0x0D)• Bad (0x0E)• Bad (0x0F)• Bad (0x10)• Bad (0x11)• Bad (0x12)• Bad (0x13)• Bad (0x14)• Bad (0x15)• Bad (0x16)• Bad (0x17)• Bad (0x18)• Bad (0x19)• Bad (0x1A)• Bad (0x1B)• Bad (0x1C)• Bad (0x1D)• Bad (0x1E)• Bad (0x1F)

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Status (6865–1 to n)

247

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Uncertain (0x40)• Uncertain (0x41)• Uncertain (0x42)• Uncertain (0x43)• Uncertain (0x44)• Uncertain (0x45)• Uncertain (0x46)• Uncertain (0x47)• Uncertain (0x48)• Uncertain (0x49)• Uncertain (0x4A)• Uncertain (0x4B)• Uncertain (0x4C)• Uncertain (0x4D)• Uncertain (0x4E)• Uncertain (0x4F)• Uncertain (0x50)• Uncertain (0x51)• Uncertain (0x52)• Uncertain (0x53)• Uncertain (0x54)• Uncertain (0x55)• Uncertain (0x56)• Uncertain (0x57)• Uncertain (0x58)• Uncertain (0x59)• Uncertain (0x5A)• Uncertain (0x5B)• Good (0x80)• Good (0x81)• Good (0x82)• Good (0x83)• Good (0x84)• Good (0x85)• Good (0x86)• Good (0x87)• Good (0x88)• Good (0x89)• Good (0x8A)• Good (0x8B)• Good (0x8C)• Good (0x8D)• Good (0x8E)• Good (0x8F)• Good (0x90)• Good (0x91)• Good (0x92)• Good (0x93)• Good (0x94)• Good (0x95)• Good (0x96)• Good (0x97)• Good (0x98)• Good (0x99)• Good (0x9A)• Good (0x9B)• Good (0xC0)• Good (0xC1)• Good (0xC2)
248

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Good (0xC3)• Good (0xC4)• Good (0xC5)• Good (0xC6)• Good (0xC7)• Good (0xC8)• Good (0xC9)• Good (0xCA)• Good (0xCB)• Good (0xCC)• Good (0xCD)• Good (0xCE)• Good (0xCF)• Good (0xD0)• Good (0xD1)• Good (0xD2)• Good (0xD3)• Good (0xD4)• Good (0xD5)• Good (0xD6)• Good (0xD7)• Good (0xD8)• Good (0xD9)• Good (0xDA)• Good (0xDB)• Good (0xDC)• Good (0xDD)• Good (0xDE)• Good (0xDF)• Good (0xE0)• Good (0xE1)• Good (0xE2)• Good (0xE3)
Factory setting

Bad (0x00)

Additional information

Description
An output parameter can be linked to an input parameter from another function block.
Both the input parameter and the output parameter have a field value and status. The
status of the input parameter is taken from the linked output parameter.

Value
NavigationDescription

Displays the discrete input value from a measuring device in the field.

User interface• Low flow or PFS Stat.=Not active• Low flow or PFS Stat.=Active or Verif. stat.=Check not done
• Verification status=failed• Verification status=busy• Verification status=ready• Verificaton overall result=failed• Status=Check not done/Result=failed

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Value (6866–1 to n)

249

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Status=failed/Result=failed• Status=busy/Result=failed• Status=ready/Result=failed• Verificaton overall result=passed• Status=Check not done/Result=passed• Status=failed/Result=passed• Status=busy/Result=passed• Status=ready/Result=passed• Verificaton overall result=Check not done
• Status=Check not done/Result=Check not done
• Status=failed/Result=Check not done
• Status=busy/Result=Check not done
• Status=ready/Result=Check not done
Factory setting

Low flow or PFS Stat.=Not active

Unacknowledged
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Unacknowledged (6889–1 to n)

Description

Option for manually acknowledging an update of the static block parameter.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description• If a new diagnostic event occurs, the measuring device sets Unacknowledged
option.
• If the diagnostic event has been acknowledged, the user can set Acknowledged
option.

Update State
NavigationDescription

Displays the status of an update of the static block parameter. The status indicates
whether the update was communicated or not.

User interface• Uninitialized• Reported• Not Reported

250

Expert → Discrete inputs → Discrete input 1 to n → Update State (6890–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Time Stamp
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Time Stamp (6888–1 to n)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of an update to the static block parameter that has not yet beencommunicated was identified. The time stamp is retained until the update confirmation isreceived.

User interface

Days (d), hours (h), minutes (m), seconds (s)Static revision
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Static revision (6887–1 to n)

Description

Displays the Static revision: Each instance of a static block parameter being accessed withwrite and communicate access due to an update is counted (event counter).

User interface

0 to 65 535

Relative Index
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Relative Index (6886–1 to n)

Description

Displays the relative index of the static block parameter that triggered the alarm (OD index
minus field bus start index).

User interface

0 to 65 535

Ack. Option
NavigationDescription

Option for automatic acknowledgment of process alarms in a specific category.

Selection• Disc Alm Aut Ack• Blk Alm Auto Ack• Fail Alm Aut Ack• OffSpecAlmAutAck• Maint Alm AutAck• Check Alm AutAck

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Ack. Option (6841–1 to n)

251Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
Use this function to determine whether an alarm must be acknowledged via the field bus
host system.
If the process alarm option has not been enabled in this parameter, this process alarm
must only be acknowledged in Unacknowledged parameter (→  218).
Current parameter (→  217) indicates the current status of all process alarms.

Current
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Current (6842–1 to n)

Description

Use this function to view the current status of the process alarms.

User interface• Discrete Alarm• Block Alarm• Fail Alarm• Off Spec Alarm• Maint. Alarm• Check Alarm

Disabled
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Disabled (6843–1 to n)

Description

Option for disabling a process alarm category.

Selection• Disc Alm Disabl• Block Alm Disabl• Fail Alm Disabl• OffSpecAlmDisabl• Maint Alm Disabl• Check Alm Disab.

Unacknowledged
NavigationDescription

Displays an unacknowledged process alarm.

User interface• Disc Alm Unack• Block Alm Unack• Fail Alm Unack• Off SpecAlmUnack• Maint Alm Unack• Check Alm Unack

252

Expert → Discrete inputs → Discrete input 1 to n → Unacknowledged (6844–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Unreported
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Unreported (6845–1 to n)

Description

Displays a process alarm that has not been communicated.

User interface• Disc Alm Unrep• Block Alm Unrep• Fail Alm Unrep• Off SpecAlmUnrep• Maint Alm Unrep• Check Alm Unrep

Alarm State
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Alarm State (6847–1 to n)

Description

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

User interface• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

SubcodeNavigationDescription

Displays the specific cause of the block alarm.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Subcode (6848–1 to n)

253

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Time Stamp
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Time Stamp (6849–1 to n)

Description

Displays the time stamp indicating when the analysis of the block was started and when astatus change of the block alarm that has not yet been communicated to the field bus hostsystem was detected. The time stamp is retained until the alarm confirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)Unacknowledged
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Unacknowledged (6850–1 to n)

Description

Option for manually acknowledging a block alarm.

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Blk Alm Auto Ack option is not enabled in Ack. Option parameter (→  251), theprocess alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Value
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Value (6856–1 to n)

Description

Displays the value of the affected parameter at the time at which the block alarm was
detected.

User interface

0 to 255

Alarm State
NavigationDescription

Displays the status of the block alarm. The status indicates whether the block alarm isactive and whether it has already been communicated to the field bus host system.

254

Expert → Discrete inputs → Discrete input 1 to n → Alarm State (6858–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User interface

Description of device parameters• Uninitialized• Clear-Reported• ClearNotReported• Active-Reported• ActiveNotRep

SubcodeNavigation

Expert → Discrete inputs → Discrete input 1 to n → Subcode (6859–1 to n)

Description

Displays the specific cause of the discrete alarm.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

Time Stamp
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Time Stamp (6860–1 to n)

Description

Displays the time stamp indicating when the analysis of the function block was started andwhen a status change of the discrete alarm that has not yet been communicated to thefield bus host system was detected. The time stamp is retained until the alarmconfirmation is received.

User interface

Days (d), hours (h), minutes (m), seconds (s)Unacknowledged
NavigationDescription

Option for manually acknowledging a discrete alarm.

Endress+Hauser

Expert → Discrete inputs → Discrete input 1 to n → Unacknowledged (6861–1 to n)

255

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• Uninitialized• Acknowledged• Unacknowledged

Factory setting

Uninitialized

Additional information

Description
If Disc Alm Aut Ack option is not enabled in Ack. Option parameter (→  251), theprocess alarm must be manually acknowledged in this parameter.
• If a new alarm occurs, the measuring device sets Unacknowledged option.
• If the alarm has been acknowledged, the user can set Acknowledged option.

Discrete Value
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Discrete Value (6862–1 to n)

Description

Displays the value of the associated parameter at the time at which the alarm was
detected.

User interface• State 0• State 1• State 2• State 3• State 4• State 5• State 6• State 7• State 8• State 9• State 10• State 11• State 12• State 13• State 14• State 15• State 16

Discrete Limit
NavigationDescription

Use this to enter the status of the discrete input value that triggers an alarm.

User entry

0 to 255

Factory setting

0256

Expert → Discrete inputs → Discrete input 1 to n → Discrete Limit (6863–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Discrete Prio
Navigation

Expert → Discrete inputs → Discrete input 1 to n → Discrete Prio (6864–1 to n)

Description

Use this to enter the priority of a discrete alarm.

User entry

0 to 15

Factory setting

0

3.9"Analog outputs" submenu

Navigation

 Expert → Analog outputs‣ Analog outputs‣ Multiple AO

3.9.1→  257"Multiple AO" submenu

Navigation

 Expert → Analog outputs → Multiple AO‣ Multiple AO

Endress+Hauser

Block tag (11351)→  258

Status Options (11363)→  261

Fault State Time (11354)→  262

FaultState Val 1 (11355)→  263

FaultState Val 2 (11356)→  263

FaultState Val 3 (11357)→  263

FaultState Val 4 (11358)→  264

FaultState Val 5 (11359)→  264

FaultState Val 6 (11360)→  264

FaultState Val 7 (11361)→  265

257

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

FaultState Val 8 (11362)→  265

FaultStateStatus (11353)→  265

Block tag
Navigation

 Expert → Analog outputs → Multiple AO → Block tag (11351)

Description

Use this function to enter the Block tag: specify a "label" for identifying the function block.

User entry

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Static Revision
Navigation

Expert → Analog outputs → Multiple AO → Static Revision (11371)

Description

Displays the Static Revision: every write access to a static block parameter is counted(event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Analog outputs → Multiple AO → Tag Description (11372)

Description

Use this function to enter the Tag Description: define a user-specific text for the detailed
description of the function block.

User entry

Max. 32 characters, such as letters, numbers or special characters (e.g. @, %, /).

Strategy
NavigationDescription

Use this function to enter the Strategy: makes it possible to group blocks by entering
identical numbers.

User entry

0 to FFFF

258

Expert → Analog outputs → Multiple AO → Strategy (11370)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Description of device parameters

0Alert Key
Navigation

Expert → Analog outputs → Multiple AO → Alert Key (11365)

Description

Use this function to enter the Alert Key: identifies the section of the plant where thetransmitter is located. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

Target mode
Navigation

Expert → Analog outputs → Multiple AO → Target mode (11369)

Description

Use this function to select the Target mode: the selection specifies which operating modeis used for this block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Selection
Detailed description of the options available for selection: Target mode parameter(→  151)

Actual mode
NavigationDescription

Displays the Actual mode: under certain conditions a block may not work in the operating
mode to be used. In this case, the Actual mode represents the valid mode in which theblock is currently operating. A comparison of the Actual mode with the Target modeindicates whether the Target mode (→  259) could be reached.

User interface• ROut• RCas• Cas

Endress+Hauser

Expert → Analog outputs → Multiple AO → Actual mode (11366)

259

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Auto• Man• LO• IMan• OOS
Additional information

Selection
Detailed description of the options available for selection: Target mode parameter(→  151)

Permitted mode
Navigation

Expert → Analog outputs → Multiple AO → Permitted mode (11368)

Description

Use this function to select the Permitted mode: the selection defines which operating
modes are available for the function block in the Target mode (→  259). The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Selection
Detailed description of the options available for selection: Target mode parameter(→  151)

Normal mode
NavigationDescription

Use this function to select the Normal mode: this mode is provided to enable the user toselect the Normal mode among the operating modes available. This can be set using an
operating tool in order to help the user configure the operating mode of a block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

260

Expert → Analog outputs → Multiple AO → Normal mode (11367)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Factory setting

Auto

Additional information

Selection

Description of device parameters

Detailed description of the options available for selection: Target mode parameter(→  151)

Block Error
Navigation

Expert → Analog outputs → Multiple AO → Block Error (11364)

Description

Displays the short text for the Block Error that occurred in the function block.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

ChannelNavigation

Expert → Analog outputs → Multiple AO → Channel (11352)

Description

Use this function to select the assignment or connection between the Analog Output
function block and the Transducer Block.

Selection• Uninitialized• Channel_0

Factory setting

Channel_0

Status Options
Navigation

 Expert → Analog outputs → Multiple AO → Status Options (11363)

Description

Option for determining the fault state of the function block.

Endress+Hauser

261

Description of device parameters

Selection• Fstate to val 1• Fstate to val 2• Fstate to val 3• Fstate to val 4• Fstate to val 5• Fstate to val 6• Fstate to val 7• Fstate to val 8• Fstate restart 1• Fstate restart 2• Fstate restart 3• Fstate restart 4• Fstate restart 5• Fstate restart 6• Fstate restart 7• Fstate restart 8

Additional information

Description

Proline Promass 300 FOUNDATION Fieldbus

This behavior is activated if an error condition of the set point that applies is present for
longer than defined in the Fault State Time parameter (→  262) or if the Set Fault
State parameter (→  161) is activated in the Resource block.
The fault state is defined via the following parameters:• Fault State Time (→  262)
• FaultState Val 1 to n
Selection• Fstate to val 1…8
The value predefined in the FaultState Val 1 to n parameter is used instead of theanalog set point. The fault state is enabled if there is an error condition.
• Fstate restart 1…8The value predefined in the FaultState Val 1 to n parameter is used if the device isrestarted. The non-volatile value is used otherwise. The fault state is not enabled andonly the predefined value is used.

Fault State Time
Navigation

 Expert → Analog outputs → Multiple AO → Fault State Time (11354)

Description

Use this function to enter a time range during which an error condition (of the currently
valid set point) must be met without interruption before an error message is generated.

User entry

Positive floating-point number

Factory setting

0s

262

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

FaultState Val 1
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 1 (11355)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 1 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 1 option is not selected in the Status Options
parameter (→  261).

FaultState Val 2
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 2 (11356)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 2 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 2 option is not selected in the Status Options
parameter (→  261).

FaultState Val 3
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 3 (11357)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 3 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 3 option is not selected in the Status Options
parameter (→  261).

Endress+Hauser

263

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

FaultState Val 4
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 4 (11358)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 4 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 4 option is not selected in the Status Options
parameter (→  261).

FaultState Val 5
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 5 (11359)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 5 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 5 option is not selected in the Status Options
parameter (→  261).

FaultState Val 6
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 6 (11360)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 6 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 6 option is not selected in the Status Options
parameter (→  261).

264

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

FaultState Val 7
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 7 (11361)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 7 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 7 option is not selected in the Status Options
parameter (→  261).

FaultState Val 8
Navigation

 Expert → Analog outputs → Multiple AO → FaultState Val 8 (11362)

Description

Use this function to enter a predefined analog value to be used if an error condition of theanalog set point 8 is present.

User entry

Signed floating-point number

Factory setting0

Additional information

Description
This value is ignored if the Fstate to val 8 option is not selected in the Status Options
parameter (→  261).

FaultStateStatus
Navigation

 Expert → Analog outputs → Multiple AO → FaultStateStatus (11353)

Description

Displays the values for which the fault state is enabled.

User interface• Val 1 in FState• Val 2 in FState• Val 3 in FState• Val 4 in FState• Val 5 in FState• Val 6 in FState• Val 7 in FState• Val 8 in FState

Endress+Hauser

265Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

3.10"Discrete outputs" submenu

The Discrete outputs function block (DO, discrete output) processes a discrete set point
received from an upstream function block or a higher-level process control system thatenables various device functions (e.g. zero point adjustment or reset of the totalizer) to be
triggered in the downstream transducer block.
Navigation

 Expert → Discrete outputs‣ Discrete outputs‣ Multiple DO

3.10.1→  266"Multiple DO" submenu

Navigation

 Expert → Discrete outputs → Multiple DO‣ Multiple DO

266

Block tag (11252)→  267

Status Options (11268)→  270

Fault State Time (11255)→  271

FStateValue DO 1 (11256)→  272

FStateValue DO 2 (11257)→  272

FStateValue DO 3 (11258)→  272

FStateValue DO 4 (11259)→  273

FStateValue DO 5 (11260)→  273

FStateValue DO 6 (11261)→  273

FStateValue DO 7 (11262)→  274

FStateValue DO 8 (11263)→  274

FaultStateStatus (11254)→  274

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Block tag
Navigation

 Expert → Discrete outputs → Multiple DO → Block tag (11252)

Description

Use this function to enter the Block tag: Used for specifying a "label" for identifying thefunction block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Static Revision
Navigation

Expert → Discrete outputs → Multiple DO → Static Revision (11270)

Description

Displays the Static Revision: Each instance of a static block parameter being accessed withwrite access is counted (event counter).

User interface

0 to FFFF

Additional information

Description
Static parameters are parameters that are not changed by the process.

Tag Description
Navigation

Expert → Discrete outputs → Multiple DO → Tag Description (11271)

Description

Use this function to enter the Tag Description: Used for defining a user-specific text for
detailed description of the function block.

User entry

Max. 32 characters such as letters, numbers or special characters (e.g. @, %, /).

Strategy
NavigationDescription

Use this function to enter the Strategy: Enables blocks to be grouped by entering identicalnumbers.

User entry

0 to FFFF

Factory setting0

Endress+Hauser

Expert → Discrete outputs → Multiple DO → Strategy (11269)

267

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Alert Key
Navigation

Expert → Discrete outputs → Multiple DO → Alert Key (11251)

Description

Use this function to enter the Alert Key: Identifies the plant unit where the transmitter islocated. This helps in pinpointing events.

User entry

0 to 0xFF

Factory setting

1

Target mode
Navigation

Expert → Discrete outputs → Multiple DO → Target mode (11267)

Description

Use this function to select the Target mode: The selection indicates which operating modeis used for this function block. This mode is generally set by a control application.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Actual mode
NavigationDescription

Displays the Actual mode: Under certain conditions, a function block may not work in theoperating mode to be used. In this case, the Actual mode shows the actual operating modethat the function block is currently operating in. By comparing the Actual mode with theTarget mode, users can see whether it was possible to reach the Target mode (→  268).

User interface• ROut• RCas• Cas• Auto• Man

268

Expert → Discrete outputs → Multiple DO → Actual mode (11264)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• LO• IMan• OOS
Additional information

User interface
Detailed description of the options available for selection: Target mode parameter(→  151)

Permitted mode
Navigation

Expert → Discrete outputs → Multiple DO → Permitted mode (11266)

Description

Use this function to select the Permitted mode: The selection defines which operating
modes are available in Target mode (→  268) for the function block. The operating
modes that are supported vary depending on the type and function of the block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting• Auto• OOS

Additional information

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Normal mode
NavigationDescription

Use this function to select the Normal mode: This is available to enable the user to selectthe Normal mode from the available operating modes. This can be set using an operating
tool in order to help the user configure the operating mode of a function block.

Selection• ROut• RCas• Cas• Auto• Man• LO• IMan• OOS

Factory setting

Auto

Endress+Hauser

Expert → Discrete outputs → Multiple DO → Normal mode (11265)

269

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Options
Detailed description of the options available for selection: Target mode parameter(→  151)

Block Error
Navigation

Expert → Discrete outputs → Multiple DO → Block Error (11272)

Description

Displays the short text for the Block Error that has occurred in the function block.

User interface• Other• BlockConfigurat• LinkConfigurat• SimulationActive• LocalOverride• DeviceFaultState• DeviceMainten• SensorFailure• OutputFailure• MemoryFailure• LostStaticData• LostNVData• ReadbackCheck• MaintenanceNeed• PowerUp• OutOfService

ChannelNavigation

Expert → Discrete outputs → Multiple DO → Channel (11253)

Description

Option for the assignment or connection between the discrete output function block andthe transducer block.

Selection• Uninitialized• Channel_D0

Factory setting

Channel_D0

Status Options
Navigation

 Expert → Discrete outputs → Multiple DO → Status Options (11268)

Description

Option for determining the fault state of the function block.

270

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Selection• Fstate to val 1• Fstate to val 2• Fstate to val 3• Fstate to val 4• Fstate to val 5• Fstate to val 6• Fstate to val 7• Fstate to val 8• Fstate restart 1• Fstate restart 2• Fstate restart 3• Fstate restart 4• Fstate restart 5• Fstate restart 6• Fstate restart 7• Fstate restart 8

Additional information

Description

Description of device parameters

This behavior is enabled if an error condition of the valid set point persists for longer than
the time specified in Fault State Time parameter (→  271) or if Set Fault Stateparameter (→  161) is enabled in Resource block.
The fault state is defined via the following parameters:• Fault State Time (→  271)
• FStateValue DO 1 to n
Options• Fstate to val 1…8
The value predefined in FStateValue DO 1 to n parameter is used in place of the discrete
set point. The fault state is enabled if there is an error condition.
• Fstate restart 1…8The value predefined in FStateValue DO 1 to n parameter is used if the device isrestarted. The non-volatile value is used otherwise. The fault state is not enabled andonly the predefined value is used.

Fault State Time
Navigation

 Expert → Discrete outputs → Multiple DO → Fault State Time (11255)

Description

Use this function to enter a time range during which an error condition (of the currently
valid set point) must be met without interruption before an error message is generated.

User entry

Positive floating-point number

Factory setting

0sEndress+Hauser

271

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

FStateValue DO 1
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 1 (11256)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 1 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 1 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

FStateValue DO 2
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 2 (11257)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 2 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 2 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

FStateValue DO 3
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 3 (11258)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 3 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 3 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

272

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

FStateValue DO 4
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 4 (11259)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 4 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 4 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

FStateValue DO 5
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 5 (11260)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 5 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 5 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

FStateValue DO 6
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 6 (11261)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 6 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 6 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

Endress+Hauser

273

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

FStateValue DO 7
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 7 (11262)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 7 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 7 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

FStateValue DO 8
Navigation

 Expert → Discrete outputs → Multiple DO → FStateValue DO 8 (11263)

Description

Use this function to enter a predefined discrete value to be used if an error condition of thediscrete set point 8 is present.

User entry

0 to 255

Factory setting0

Additional information

Description
If Fstate to val 8 option is not selected in Status Options parameter (→  270), thisvalue is ignored.

FaultStateStatus
Navigation

 Expert → Discrete outputs → Multiple DO → FaultStateStatus (11254)

Description

Displays the values for which the fault state is enabled.

User interface• Val 1 in FState• Val 2 in FState• Val 3 in FState• Val 4 in FState• Val 5 in FState• Val 6 in FState• Val 7 in FState• Val 8 in FState

274

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

3.11

Description of device parameters"Application" submenu

Navigation

 Expert → Application‣ Application
Reset all tot. (2806)→  275‣ Totalizer 1 to n

→  275‣ Viscosity→  279‣ Concentration→  280

Reset all tot.
Navigation

 Expert → Application → Reset all tot. (2806)

Description

Use this function to reset all totalizers to the value 0 and restart the totaling process. Thisdeletes all the flow values previously totalized.

Selection• Cancel• Reset + totalize

Factory setting

CancelAdditional information

Selection
Options

Description

CancelNo action is executed and the user exits the parameter.

Reset + totalize

Resets all totalizers to 0 and restarts the totaling process. This deletes all the flow
values previously totalized.

3.11.1"Totalizer 1 to n" submenu

Navigation

 Expert → Application → Totalizer 1 to n

‣ Totalizer 1 to n

Endress+Hauser

Assign variable (0914–1 to n)→  276

Unit totalizer 1 to n (0915–1 to n)→  276

Operation mode (0908–1 to n)→  277275

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Control Tot. 1 to n (0912–1 to n)→  278

Preset value 1 to n (0913–1 to n)→  278

Failure mode (0901–1 to n)→  279

Assign variable

Navigation

 Expert → Application → Totalizer 1 to n → Assign variable (0914–1 to n)

Description

Use this function to select a process variable for the Totalizer 1 to n.

Selection• Off• Volume flow
• Mass flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *Factory setting

Mass flow

Additional information

Description
If the option selected is changed, the device resets the totalizer to 0.
Selection
If the Off option is selected, only Assign variable parameter (→  276) is still displayedin the Totalizer 1 to n submenu. All other parameters in the submenu are hidden.

Unit totalizer 1 to n

Navigation

 Expert → Application → Totalizer 1 to n → Unit totalizer 1 to n (0915–1 to n)

Prerequisite

A process variable is selected in the Assign variable parameter (→  276) of the
Totalizer 1 to n submenu.

Description

Use this function to select the process variable unit for the Totalizer 1 to n (→  275).

Selection

SI units
•g
• kg
•t

US units• oz• lb• STonor*276Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

SI units
• cm³
• dm³
• m³
• ml
•l
• hl
• Ml Mega

Description of device parameters

US units• af• ft³• fl oz (us)• gal (us)• kgal (us)• Mgal (us)• bbl (us;liq.)• bbl (us;beer)• bbl (us;oil)• bbl (us;tank)

Imperial units• gal (imp)• Mgal (imp)• bbl (imp;beer)• bbl (imp;oil)

US units• Sft³• Sgal (us)• Sbbl (us;liq.)

Imperial units
Sgal (imp)orSI units
• Nl
• Nm³
• Sl
• Sm³
Factory setting

Country-specific:
• kg
• lb

Additional information

Description
The unit is selected separately for each totalizer. It is independent of the selection
made in the System units submenu (→  66).
Selection
The selection is dependent on the process variable selected in the Assign variable
parameter (→  276).

Operation modeNavigation

 Expert → Application → Totalizer 1 to n → Operation mode (0908–1 to n)

Prerequisite

A process variable is selected in the Assign variable parameter (→  276) of the
Totalizer 1 to n submenu.

Description

Use this function to select how the totalizer summates the flow.

Selection• Net flow total• Forward total• Reverse total

Factory setting

Net flow total

Endress+Hauser

277

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Selection• Net flow total
Flow values in the forward and reverse flow direction are totalized and balanced against
one another. Net flow is registered in the flow direction.
• Forward total
Only the flow in the forward flow direction is totalized.
• Reverse total
Only the flow in the reverse flow direction is totalized (= reverse flow quantity).

Control Tot. 1 to n
Navigation

 Expert → Application → Totalizer 1 to n → Control Tot. 1 to n (0912–1 to n)

Prerequisite

A process variable is selected in the Assign variable parameter (→  276) of the
Totalizer 1 to n submenu.

Description

Use this function to select the control of totalizer value 1-3.

Selection• Totalize• Reset + hold
• Preset + hold
• Reset + totalize• Preset+totalize• HoldFactory setting

Totalize

Additional information

Selection
Options

Description

Totalize

The totalizer is started or continues running.

Reset + hold

The totaling process is stopped and the totalizer is reset to 0.

Preset + hold

The totaling process is stopped and the totalizer is set to its defined start value
from the Preset value parameter.

Reset + totalize

The totalizer is reset to 0 and the totaling process is restarted.

Preset+totalize

The totalizer is set to the defined start value from the Preset value parameter andthe totaling process is restarted.

Hold

Totalizing is stopped.

Preset value 1 to n
Navigation

 Expert → Application → Totalizer 1 to n → Preset value 1 to n (0913–1 to n)

Prerequisite

A process variable is selected in the Assign variable parameter (→  276) of the
Totalizer 1 to n submenu.

Description

Use this function to enter a start value for the Totalizer 1 to n.

278

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User entry

Signed floating-point number

Factory setting

Country-specific:• 0 kg• 0 lbAdditional information

User entry

Description of device parameters

The unit of the selected process variable is specified for the totalizer in the Unit
totalizer parameter (→  276).
ExampleThis configuration is suitable for applications such as iterative filling processes with a fixed
batch quantity.

Failure modeNavigation

 Expert → Application → Totalizer 1 to n → Failure mode (0901–1 to n)

Prerequisite

A process variable is selected in the Assign variable parameter (→  276) of the
Totalizer 1 to n submenu.

Description

Use this function to select how a totalizer behaves in the event of a device alarm.

Selection• Stop• Actual value• Last valid value

Factory setting

Stop

Additional information

Description
This setting does not affect the failsafe mode of other totalizers and the outputs. Thisis specified in separate parameters.
Selection• Stop
The totalizer is stopped in the event of a device alarm.
• Actual value
The totalizer continues to count based on the actual measured value; the device alarm isignored.
• Last valid value
The totalizer continues to count based on the last valid measured value before the device
alarm occurred.

3.11.2"Viscosity" submenu

Only available for Promass I.
For detailed information on the parameter descriptions for the Viscosity application
package,refer to the Special Documentation for the device →  7
Endress+Hauser

279

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Navigation

 Expert → Application → Viscosity‣ Viscosity

3.11.3"Concentration" submenu

For detailed information on the parameter descriptions for the Concentration
application package,refer to the Special Documentation for the device →  7
Navigation

 Expert → Application → Concentration‣ Concentration

3.12"Diagnostics" submenu

Navigation

 Expert → Diagnostics‣ Diagnostics
Actual diagnos. (0691)→  281Prev.diagnostics (0690)→  282

Time fr. restart (0653)→  282

Operating time (0652)→  283‣ Diagnostic list

→  283‣ Event logbook→  287‣ Device info

→  289‣ Mainboard module

280‣ Sens. electronic→  295‣ I/O module 1→  296‣ I/O module 2→  297‣ I/O module 3→  296‣ I/O module 4→  296‣ Display module→  298‣ Min/max val.

→  299

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters‣ Data logging→  309‣ Heartbeat→  318‣ Simulation→  318

Actual diagnos.
Navigation

 Expert → Diagnostics → Actual diagnos. (0691)

Prerequisite

A diagnostic event has occurred.

Description

Displays the current diagnostic message. If two or more messages occur simultaneously,
the message with the highest priority is shown on the display.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Additional pending diagnostic messages can be viewed in the Diagnostic list
submenu (→  283).
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
ExampleFor the display format:
F271 Main electronics

Timestamp
Navigation

Expert → Diagnostics → Timestamp

Description

Displays the operating time when the current diagnostic message occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Actual diagnos. parameter(→  281).
ExampleFor the display format:
24d12h13m00s

Endress+Hauser

281

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Prev.diagnostics
Navigation

 Expert → Diagnostics → Prev.diagnostics (0690)

Prerequisite

Two diagnostic events have already occurred.

Description

Displays the diagnostic message that occurred before the current message.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
ExampleFor the display format:
F271 Main electronics

Timestamp
Navigation

Expert → Diagnostics → Timestamp

Description

Displays the operating time when the last diagnostic message before the current message
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Prev.diagnostics parameter(→  282).
ExampleFor the display format:
24d12h13m00s

Time fr. restart
Navigation

 Expert → Diagnostics → Time fr. restart (0653)

Description

Use this function to display the time the device has been in operation since the last device
restart.

User interface

Days (d), hours (h), minutes (m) and seconds (s)282Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Operating time
Navigation

 Expert → Diagnostics → Operating time (0652)

Description

Use this function to display the length of time the device has been in operation.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

User interface
The maximum number of days is 9999, which is equivalent to 27 years.

3.12.1"Diagnostic list" submenu

Navigation

 Expert → Diagnostics → Diagnostic list

‣ Diagnostic list
Diagnostics 1 (0692)→  283

Diagnostics 2 (0693)→  284

Diagnostics 3 (0694)→  285

Diagnostics 4 (0695)→  286

Diagnostics 5 (0696)→  286

Diagnostics 1
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 1 (0692)

Description

Displays the current diagnostics message with the highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics• F276 I/O module

Endress+Hauser

283

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp

Description

Displays the operating time when the diagnostic message with the highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Diagnostics 1 parameter (→  283).
ExampleFor the display format:
24d12h13m00s

Diagnostics 2
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 2 (0693)

Description

Displays the current diagnostics message with the second-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics• F276 I/O module

Timestamp
NavigationDescription

Displays the operating time when the diagnostic message with the second-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)284Expert → Diagnostics → Diagnostic list → Timestamp

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Display
The diagnostic message can be viewed via the Diagnostics 2 parameter (→  284).
ExampleFor the display format:
24d12h13m00s

Diagnostics 3
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 3 (0694)

Description

Displays the current diagnostics message with the third-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics• F276 I/O module

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp

Description

Displays the operating time when the diagnostic message with the third-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Diagnostics 3 parameter (→  285).
ExampleFor the display format:
24d12h13m00s

Endress+Hauser

285

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Diagnostics 4
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 4 (0695)

Description

Displays the current diagnostics message with the fourth-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

Additional information

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics• F276 I/O module

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp

Description

Displays the operating time when the diagnostic message with the fourth-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Diagnostics 4 parameter (→  286).
ExampleFor the display format:
24d12h13m00s

Diagnostics 5
Navigation

 Expert → Diagnostics → Diagnostic list → Diagnostics 5 (0696)

Description

Displays the current diagnostics message with the fifth-highest priority.

User interface

Symbol for diagnostic behavior, diagnostic code and short message.

286

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Display
Via the local display: the time stamp and corrective measures referring to the cause of
the diagnostic message can be accessed via the  key.
Examples
For the display format:• F271 Main electronics• F276 I/O module

Timestamp
Navigation

Expert → Diagnostics → Diagnostic list → Timestamp

Description

Displays the operating time when the diagnostic message with the fifth-highest priority
occurred.

User interface

Days (d), hours (h), minutes (m) and seconds (s)Additional information

Display
The diagnostic message can be viewed via the Diagnostics 5 parameter (→  286).
ExampleFor the display format:
24d12h13m00s

3.12.2"Event logbook" submenu

Navigation

Expert → Diagnostics → Event logbook‣ Event logbook
Filter options (0705)→  287‣ Event list

→  288

Filter options

NavigationDescription

Use this function to select the category whose event messages are displayed in the event
list of the local display.

Endress+Hauser

Expert → Diagnostics → Event logbook → Filter options (0705)

287

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• All• Failure (F)• Funct. check (C)• Out of spec. (S)• Mainten. req.(M)• Information (I)Factory setting

All

Additional information

Description
The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107:
• F = Failure• C = Function Check• S = Out of Specification• M = Maintenance Required

Filter options

Navigation

Expert → Diagnostics → Event logbook → Filter options

Description

Use this function to select the category whose event messages are displayed in the event
list of the operating tool.

Selection• All• Failure (F)• Funct. check (C)• Out of spec. (S)• Mainten. req.(M)• Information (I)Factory setting

All

Additional information

Description
The status signals are categorized in accordance with VDI/VDE 2650 and NAMUR
Recommendation NE 107:
• F = Failure• C = Function Check• S = Out of Specification• M = Maintenance Required"Event list" submenu
The Event list submenu is only displayed if operating via the local display.
If operating via the FieldCare operating tool, the event list can be read out with aseparate FieldCare module.
If operating via the Web browser, the event messages can be found directly in the
Event logbook submenu.

288

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Navigation

Description of device parameters

 Expert → Diagnostics → Event logbook → Event list

▸ Event list
Event list

→  289

Event list
Navigation

Expert → Diagnostics → Event logbook → Event list

Description

Displays the history of event messages of the category selected in the Filter options
parameter (→  287).

User interface• For a "Category I" event message
Information event, short message, symbol for event recording and operating time whenerror occurred• For a "Category F, C, S, M" event message (status signal)
Diagnostics code, short message, symbol for event recording and operating time whenerror occurred

Additional information

Description
A maximum of 20 event messages are displayed in chronological order.
If the Extended HistoROM application package (order option) is enabled in the device, theevent list can contain up to 100 entries .
The following symbols indicate whether an event has occurred or has ended:• : Occurrence of the event• : End of the event
Examples
For the display format:• I1091 Configuration modified
 24d12h13m00s• F271 Main electronics
 01d04h12min30s
HistoROM
A HistoROM is a "non-volatile" device memory in the form of an EEPROM.

3.12.3"Device information" submenu

Navigation

 Expert → Diagnostics → Device info

‣ Device info
Device tag (10799)

Endress+Hauser→  290289

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Serial number (10798)→  291

Device name (0020)→  291

Device revision (10710)→  291

Device type (10711)→  291

DD Revision (10709)→  292

Firmware version (10792)→  292

Hardware rev. (10793)→  292

ITK Version (10794)→  292

Order code (10795)→  293Ext. order cd. 1 (10796)→  293Ext. order cd. 2 (10797)→  293

ENP version (10791)→  294

Device tag
Navigation

 Expert → Diagnostics → Device info → Device tag (10799)

Description

Use this function to enter a unique name for the measuring point so it can be identified
quickly within the plant. The name is displayed in the header:
1

XXXXXXXXX

A0029422

User entry

Max. 32 characters such as letters, numbers or special characters (e. g. @, %, /)Factory setting

Promass300/500

Additional information

User entry
The number of characters displayed depends on the characters used.

290

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Serial number
Navigation

 Expert → Diagnostics → Device info → Serial number (10798)

Description

Displays the serial number of the measuring device. It can also be found on the nameplateof the sensor and transmitter.
Uses of the serial number• To identify the measuring device quickly, e.g. when contacting Endress+Hauser.
• To obtain specific information on the measuring device using the Device Viewer:
www.endress.com/deviceviewer

User interface

Max. 11-digit character string comprising letters and numbers.

Additional information

Factory settingThis information varies depending on the device. Only an example is given here.

Device name
Navigation

 Expert → Diagnostics → Device info → Device name (0020)

Description

Displays the name of the transmitter. It can also be found on the nameplate of thetransmitter.

User interface

Promass 300/500

Device type
Navigation

 Expert → Diagnostics → Device info → Device type (10711)

Description

Use this function to view the device type with which the measuring device is registeredwith FOUNDATION Fieldbus.

User interface

Promass 300/500

Device revision
Navigation

 Expert → Diagnostics → Device info → Device revision (10710)

Description

Use this function to view the device revision with which the measuring device is registeredwith FOUNDATION Fieldbus.

User interface

0 to 255

Factory setting1

Endress+Hauser

291

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

DD Revision
Navigation

 Expert → Diagnostics → Device info → DD Revision (10709)

Description

Displays the revision number of the device description (DD).

User interface

1

Additional information

Description
This display can be used to ensure that the right system files (DD = device description)
are used for integration into the host system. The system files can be downloaded freeof charge online at: www.endress.com.

Firmware version
Navigation

 Expert → Diagnostics → Device info → Firmware version (10792)

Description

Displays the device firmware version installed.

User interface

Character string with the following format:
xx.yy.zz

Hardware rev.
Navigation

 Expert → Diagnostics → Device info → Hardware rev. (10793)

Description

Displays the Hardware rev..

User interface

Character string in the format xx.yy.zz

ITK Version
Navigation

 Expert → Diagnostics → Device info → ITK Version (10794)

Description

Displays the revision status of the Interoperability Test Kits (ITK).

User interface

6292

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Order code
Navigation

 Expert → Diagnostics → Device info → Order code (10795)

Description

Use this function to enter the device order code.
The order code can be found on the nameplate of the sensor and transmitter in the"Order code" field.

User entry

Character string composed of letters, numbers and certain punctuation marks

Additional information

Description
The order code is generated from the extended order code, which defines all the device
features of the product structure. In contrast, the device features cannot be read directlyfrom the order code.
Uses of the order code
• To order an identical spare device.
• To identify the device quickly and easily, e.g. when contacting Endress+Hauser.

Ext. order cd. 1
Navigation

 Expert → Diagnostics → Device info → Ext. order cd. 1 (10796)

Description

Use this function to enter the first part of the extended order code.
On account of length restrictions, the extended order code is split into a maximum of 2
parameters.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

User entry

Character string

Additional information

Description
The extended order code indicates the version of all the features of the product structure
for the measuring device and thus uniquely identifies the measuring device.
The extended order code can also be found on the nameplate of the sensor andtransmitter in the "Ext. ord. cd." field.

Ext. order cd. 2
Navigation

 Expert → Diagnostics → Device info → Ext. order cd. 2 (10797)

Description

Use this function to enter the second part of the extended order code.
Deletion of the ordered customer-specific parameter set that was preconfigured
before delivery. This parameter set cannot be reset. It can only be restored to thefactory setting.

User entry
Endress+Hauser

Character string293

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

For additional information, see Ext. order cd. 1 parameter

ENP version
Navigation

 Expert → Diagnostics → Device info → ENP version (10791)

Description

Displays the version of the electronic nameplate. This electronic nameplate stores a data
record for device identification that includes more data than the nameplates attached tothe outside of the device.

User interface

Character string in the format xx.yy.zz

3.12.4"Mainboard module" submenu‣ Mainboard module
Software rev.

→  294

Software rev.
Navigation

 Expert → Diagnostics → Mainboard module → Software rev. (0072)
 Expert → Diagnostics → Display module → Software rev. (0072)
 Expert → Diagnostics → Sens. electronic → Software rev. (0072)
Expert → Diagnostics → I/O module 1 → Software rev. (0072)

Expert → Diagnostics → I/O module 2 → Software rev. (0072)

Expert → Diagnostics → I/O module 3 → Software rev. (0072)

Expert → Diagnostics → I/O module 4 → Software rev. (0072)

Description

Use this function to display the software revision of the module.

User interface

Positive integer

294

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Build no. softw.
Navigation

 Expert → Diagnostics → Mainboard module → Build no. softw. (0079)
 Expert → Diagnostics → Display module → Build no. softw. (0079)
 Expert → Diagnostics → Sens. electronic → Build no. softw. (0079)
Expert → Diagnostics → I/O module 1 → Build no. softw. (0079)

Expert → Diagnostics → I/O module 2 → Build no. softw. (0079)

Expert → Diagnostics → I/O module 3 → Build no. softw. (0079)

Expert → Diagnostics → I/O module 4 → Build no. softw. (0079)

Description

Displays the software build number of the module.

User interface

Positive integer

Bootloader rev.
Navigation

 Expert → Diagnostics → Mainboard module → Bootloader rev. (0073)
 Expert → Diagnostics → Display module → Bootloader rev. (0073)
 Expert → Diagnostics → Sens. electronic → Bootloader rev. (0073)
Expert → Diagnostics → I/O module 1 → Bootloader rev. (0073)

Expert → Diagnostics → I/O module 2 → Bootloader rev. (0073)

Expert → Diagnostics → I/O module 3 → Bootloader rev. (0073)

Expert → Diagnostics → I/O module 4 → Bootloader rev. (0073)

Description

Displays the bootloader revision of the software.

User interface

Positive integer

3.12.5"Sens. electronic" submenu

Navigation

 Expert → Diagnostics → Sens. electronic‣ Sens. electronic
Software rev. (0072)

Endress+Hauser→  296295

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Software rev.
Navigation

 Expert → Diagnostics → Sens. electronic → Software rev. (0072)

Description

Use this function to display the software revision of the module.

User interface

Positive integer

Build no. softw.
Navigation

 Expert → Diagnostics → Sens. electronic → Build no. softw. (0079)

Description

Displays the software build number of the module.

User interface

Positive integer

Bootloader rev.
Navigation

 Expert → Diagnostics → Sens. electronic → Bootloader rev. (0073)

Description

Displays the bootloader revision of the software.

User interface

Positive integer

3.12.6"I/O module 1" submenu

Navigation

 Expert → Diagnostics → I/O module 1‣ I/O module 1
I/O 1 terminals (3902–1)→  296

Software rev. (0072)→  297

I/O 1 terminals
Navigation

 Expert → Diagnostics → I/O module 1 → I/O 1 terminals (3902–1)

Description

Displays the terminal numbers used by the I/O module.

296

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

User interface

Description of device parameters• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)

Software rev.
Navigation

Expert → Diagnostics → I/O module 1 → Software rev. (0072)

Description

Use this function to display the software revision of the module.

User interface

Positive integer

3.12.7"I/O module 2" submenu

Navigation

 Expert → Diagnostics → I/O module 2‣ I/O module 2
I/O 2 terminals→  297

Software rev.

→  297

I/O 1 terminals
Navigation

 Expert → Diagnostics → I/O module 1 → I/O 1 terminals (3902–1)

Description

Displays the terminal numbers used by the I/O module.

User interface• Not used
• 26-27 (I/O 1)• 24-25 (I/O 2)

Software rev.
NavigationDescription

Use this function to display the software revision of the module.

User interface

Positive integer

Endress+Hauser

Expert → Diagnostics → I/O module 2 → Software rev. (0072)

297

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Build no. softw.
Navigation

Expert → Diagnostics → I/O module 2 → Build no. softw. (0079)

Description

Displays the software build number of the module.

User interface

Positive integer

Bootloader rev.
Navigation

Expert → Diagnostics → I/O module 2 → Bootloader rev. (0073)

Description

Displays the bootloader revision of the software.

User interface

Positive integer

3.12.8"Display module" submenu

Navigation

 Expert → Diagnostics → Display module‣ Display module
Software rev. (0072)→  298

Software rev.
Navigation

 Expert → Diagnostics → Display module → Software rev. (0072)

Description

Use this function to display the software revision of the module.

User interface

Positive integer

Build no. softw.
Navigation

 Expert → Diagnostics → Display module → Build no. softw. (0079)

Description

Displays the software build number of the module.

User interface

Positive integer

298

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Bootloader rev.
Navigation

 Expert → Diagnostics → Display module → Bootloader rev. (0073)

Description

Displays the bootloader revision of the software.

User interface

Positive integer

3.12.9"Min/max val." submenu

Navigation

 Expert → Diagnostics → Min/max val.

‣ Min/max val.
Reset min/max (6151)→  299‣ Electronic temp.

→  300‣ Medium temp.

→  301‣ Carr. pipe temp.

→  302‣ Oscil. frequency→  303‣ Tors.oscil.freq.

→  304‣ Oscil. amplitude→  305‣ Tor. osc. amp.

→  305‣ Oscil. damping→  306‣ Tors.oscil.damp.

→  307‣ Signal asymmetry→  308‣ Tors.sig.asymm.

→  308

Reset min/max

Navigation

 Expert → Diagnostics → Min/max val. → Reset min/max (6151)

Description

Use this function to select measured variables whose minimum, maximum and averagemeasured values are to be reset.

Endress+Hauser

299

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Selection• Cancel• Oscil. amplitude• Osc. ampl. 1 *• Oscil. damping• Tors.oscil.damp. *• Oscil. frequency• Tors.oscil.freq. *• Signal asymmetry• Tors.sig.asymm. *Factory setting

Cancel"Electronic temp." submenu
Navigation

 Expert → Diagnostics → Min/max val. → Electronic temp.

‣ Electronic temp.
Minimum value→  300

Maximum value→  300

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Electronic temp. → Maximum value (6051)

Description

Displays the highest previously measured temperature value of the main electronics
module.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Electronic temp. → Minimum value (6052)

Description

Displays the lowest previously measured temperature value of the main electronics
module.

*300

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)"Medium temp." submenu
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp.

‣ Medium temp.
Minimum value (6109)→  301

Maximum value (6108)→  301

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp. → Minimum value (6109)

Description

Displays the lowest previously measured medium temperature value.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Medium temp. → Maximum value (6108)

Description

Displays the highest previously measured medium temperature value.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)

Endress+Hauser

301

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus"Carr. pipe temp." submenu
Navigation

 Expert → Diagnostics → Min/max val. → Carr. pipe temp.

‣ Carr. pipe temp.
Minimum value (6030)→  302

Maximum value (6029)→  302

Minimum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Carr. pipe temp. → Minimum value (6030)
Only available for:• Promass A• Promass F• Promass H• Promass I• Promass O• Promass P• PromassQ• Promass S• Promass X
For the following order code
"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the lowest previously measured temperature value of the carrier pipe.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)

Maximum value
Navigation
Prerequisite302

 Expert → Diagnostics → Min/max val. → Carr. pipe temp. → Maximum value (6029)
Only available for:• Promass A• Promass F• Promass H• Promass I• Promass O• Promass P• PromassQ• Promass S• Promass X
Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

For the following order code
"Application package", option EB "Heartbeat Verification + Monitoring"
Description

Displays the highest previously measured temperature value of the carrier pipe.

User interface

Signed floating-point number

Additional information

Dependency
The unit is taken from the Temperature unit parameter (→  72)"Oscil. frequency" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. frequency‣ Oscil. frequency
Minimum value (6071)→  303

Maximum value (6070)→  303

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. frequency → Minimum value (6071)

Description

Displays the lowest previously measured oscillation frequency.

User interface

Signed floating-point number

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. frequency → Maximum value (6070)

Description

Displays the highest previously measured oscillation frequency.

User interface

Signed floating-point number

Endress+Hauser

303

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus"Tors.oscil.freq." submenu
Navigation

 Expert → Diagnostics → Min/max val. → Tors.oscil.freq.

‣ Tors.oscil.freq.
Minimum value (6069)→  304

Maximum value (6068)→  304

Minimum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tors.oscil.freq. → Minimum value (6069)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the lowest previously measured torsion oscillation frequency.

User interface

Signed floating-point number

Maximum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tors.oscil.freq. → Maximum value (6068)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the highest previously measured torsion oscillation frequency.

User interface

Signed floating-point number

304

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters"Oscil. amplitude" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. amplitude‣ Oscil. amplitude
Minimum value (6010)→  305

Maximum value (6009)→  305

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. amplitude → Minimum value (6010)

Description

Displays the lowest previously measured oscillation amplitude.

User interface

Signed floating-point number

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. amplitude → Maximum value (6009)

Description

Displays the highest previously measured oscillation amplitude.

User interface

Signed floating-point number"Tor. osc. amp." submenu
Navigation

 Expert → Diagnostics → Min/max val. → Tor. osc. amp.

‣ Tor. osc. amp.

Endress+Hauser

Minimum value (6008)→  306

Maximum value (6007)→  306305

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Minimum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tor. osc. amp. → Minimum value (6008)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the lowest previously measured torsion oscillation amplitude.

User interface

Signed floating-point number

Maximum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tor. osc. amp. → Maximum value (6007)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the highest previously measured torsion oscillation amplitude.

User interface

Signed floating-point number"Oscil. damping" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. damping‣ Oscil. damping
Minimum value (6122)→  306

Maximum value (6121)→  307

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. damping → Minimum value (6122)

Description

Displays the lowest previously measured oscillation damping.

User interface

Signed floating-point number

306

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Oscil. damping → Maximum value (6121)

Description

Displays the highest previously measured oscillation damping.

User interface

Signed floating-point number"Tors.oscil.damp." submenu
Navigation

 Expert → Diagnostics → Min/max val. → Tors.oscil.damp.

‣ Tors.oscil.damp.
Minimum value (6120)→  307

Maximum value (6119)→  307

Minimum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tors.oscil.damp. → Minimum value (6120)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the lowest previously measured torsion oscillation damping.

User interface

Signed floating-point number

Maximum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tors.oscil.damp. → Maximum value (6119)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Endress+Hauser

Displays the highest previously measured torsion oscillation damping.

307

Description of device parameters

User interface

Proline Promass 300 FOUNDATION Fieldbus

Signed floating-point number"Signal asymmetry" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Signal asymmetry‣ Signal asymmetry
Minimum value (6015)→  308

Maximum value (6014)→  308

Minimum value
Navigation

 Expert → Diagnostics → Min/max val. → Signal asymmetry → Minimum value(6015)

Description

Displays the lowest previously measured signal asymmetry.

User interface

Signed floating-point number

Maximum value
Navigation

 Expert → Diagnostics → Min/max val. → Signal asymmetry → Maximum value(6014)

Description

Displays the highest previously measured signal asymmetry.

User interface

Signed floating-point number"Torsion signal asymmetry" submenu
Navigation

 Expert → Diagnostics → Min/max val. → Tors.sig.asymm.

‣ Tors.sig.asymm.

308

Minimum value (6284)→  309

Maximum value (6283)→  309

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Minimum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tors.sig.asymm. → Minimum value (6284)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the lowest previously measured torsion signal asymmetry.

User interface

Signed floating-point number

Maximum value
Navigation
Prerequisite Expert → Diagnostics → Min/max val. → Tors.sig.asymm. → Maximum value (6283)
Only available for Promass I.
For the following order code:"Application package", option EB "Heartbeat Verification + Monitoring"

Description

Displays the highest previously measured torsion signal asymmetry.

User interface

Signed floating-point number

3.12.10 "Data logging" submenu
Navigation

 Expert → Diagnostics → Data logging‣ Data logging

Endress+Hauser

Assign chan. 1→  310

Assign chan. 2→  311

Assign chan. 3→  312Assign chan. 4→  312

Logging interval→  312

Clear logging→  313

Data logging→  313309

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Logging delay→  314

Data log.control→  314

Data log. status→  315

Logging duration→  315‣ Displ.channel 1→  315‣ Displ.channel 2→  317‣ Displ.channel 3→  317‣ Displ.channel 4→  318

Assign chan. 1Navigation

 Expert → Diagnostics → Data logging → Assign chan. 1 (0851)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Use this function to select a process variable for the data logging channel.

Selection• Off• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Temperature• Carr. pipe temp. *• Electronic temp.
• Osc. freq. 0• Osc. freq. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Oscil. amplitude *• Osc. ampl. 1 **310

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *• Curr.output 1• Curr.output 2 *• Curr.output 3 *• HBSI
Factory setting

Off

Additional information

Description
A total of 1000 measured values can be logged. This means:• 1000 data points if 1 logging channel is used
• 500 data points if 2 logging channels are used
• 333 data points if 3 logging channels are used
• 250 data points if 4 logging channels are used
Once the maximum number of data points is reached, the oldest data points in the data log
are cyclically overwritten in such a way that the last 1000, 500, 333 or 250 measured
values are always in the log (ring memory principle).
The log contents are cleared if the option selected is changed.
Selection
Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Assign freq. parameter (→  128)

Assign chan. 2Navigation

 Expert → Diagnostics → Data logging → Assign chan. 2 (0852)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Options for the assignment of a process variable to the data logging channel.

Selection

Picklist, see Assign channel 1 parameter (→  310)

Factory setting

Off*

Visibility depends on order options or device settings

Endress+Hauser

311

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Assign chan. 3

Navigation

 Expert → Diagnostics → Data logging → Assign chan. 3 (0853)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Options for the assignment of a process variable to the data logging channel.

Selection

Picklist, see Assign channel 1 parameter (→  310)

Factory setting

Off

Assign chan. 4Navigation

 Expert → Diagnostics → Data logging → Assign chan. 4 (0854)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Options for the assignment of a process variable to the data logging channel.

Selection

Picklist, see Assign channel 1 parameter (→  310)

Factory setting

Off

Logging interval

Navigation

 Expert → Diagnostics → Data logging → Logging interval (0856)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Use this function to enter the logging interval Tlog for data logging.

User entry

0.1 to 999.0 s

Factory setting

1.0 s

312

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
This defines the interval between the individual data points in the data log, and thus the
maximum loggable process time Tlog:• If 1 logging channel is used: Tlog = 1000 × tlog
• If 2 logging channels are used: Tlog = 500 × tlog
• If 3 logging channels are used: Tlog = 333 × tlog
• If 4 logging channels are used: Tlog = 250 × tlog
Once this time elapses, the oldest data points in the data log are cyclically overwritten suchthat a time of Tlog always remains in the memory (ring memory principle).
The log contents are cleared if the length of the logging interval is changed.
ExampleIf 1 logging channel is used:• Tlog = 1000 × 1 s = 1 000 s ≈15 min• Tlog = 1000 × 10 s = 10 000 s ≈ 3 h• Tlog = 1000 × 80 s = 80 000 s ≈ 1 d• Tlog = 1000 × 3 600 s = 3 600 000 s ≈ 41 d

Clear logging

Navigation

 Expert → Diagnostics → Data logging → Clear logging (0855)

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).

Description

Use this function to clear the entire logging data.

Selection• Cancel• Clear data

Factory setting

CancelAdditional information

Selection• Cancel
The data is not cleared. All the data is retained.
• Clear data
The logging data is cleared. The logging process starts from the beginning.

Data logging

Navigation

 Expert → Diagnostics → Data logging → Data logging (0860)

Description

Use this function to select the data logging method.

Selection• Overwriting• Not overwriting

Endress+Hauser

313

Description of device parameters

Factory setting

Overwriting

Additional information

Selection

Proline Promass 300 FOUNDATION Fieldbus• Overwriting
The device memory applies the FIFO principle.
• Not overwriting
Data logging is canceled if the measured value memory is full (single shot).

Logging delay

Navigation

 Expert → Diagnostics → Data logging → Logging delay (0859)

Prerequisite

In the Data logging parameter (→  313), the Not overwriting option is selected.

Description

Use this function to enter the time delay for measured value logging.

User entry

0 to 999 h

Factory setting

0hAdditional information

Description
Once measured value logging has been started with the Data log.control parameter(→  314), the device does not save any data for the duration of the time delay entered.

Data log.control



Navigation

 Expert → Diagnostics → Data logging → Data log.control (0857)

Prerequisite

In the Data logging parameter (→  313), the Not overwriting option is selected.

Description

Use this function to start and stop measured value logging.

Selection• None• Delete + start• Stop

Factory setting

None

Additional information

Selection• NoneInitial measured value logging status.
• Delete + start
All the measured values recorded for all the channels are deleted and measured value
logging starts again.
• StopMeasured value logging is stopped.

314

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Data log. status
Navigation

 Expert → Diagnostics → Data logging → Data log. status (0858)

Prerequisite

In the Data logging parameter (→  313), the Not overwriting option is selected.

Description

Displays the measured value logging status.

User interface• Done• Delay active• Active• Stopped

Factory setting

Done

Additional information

Selection• DoneMeasured value logging has been performed and completed successfully.
• Delay active
Measured value logging has been started but the logging interval has not yet elapsed.
• Active
The logging interval has elapsed and measured value logging is active.
• Stopped
Measured value logging is stopped.

Logging duration
Navigation

 Expert → Diagnostics → Data logging → Logging duration (0861)

Prerequisite

In the Data logging parameter (→  313), the Not overwriting option is selected.

Description

Displays the total logging duration.

User interface

Positive floating-point number

Factory setting

0s"Displ.channel 1" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 1▸ Displ.channel 1
Display channel 1

Endress+Hauser→  316315

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Display channel 1
Navigation

Expert → Diagnostics → Data logging → Displ.channel 1

Prerequisite

The Extended HistoROM application package is available.
The software options currently enabled are displayed in the SW option overv.
parameter (→  49).
One of the following options is selected in the Assign chan. 1 parameter (→  310):• Mass flow
• Volume flow
• Correct.vol.flow• Target mass flow *• Carrier mass fl. *• Density• Ref.density• Concentration *• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Carr. pipe temp. *• Electronic temp.
• Curr.output 1• Osc. freq. 0• Osc. freq. 1 *• Freq. fluct. 0• Freq. fluct. 1 *• Oscil. amplitude *• Osc. ampl. 1 *• Osc. damping 0• Osc. damping 1 *• Osc.damp.fluct 0• Osc.damp.fluct 1 *• Signal asymmetry• Exc. current 0• Exc. current 1 *Description

Displays the measured value trend for the logging channel in the form of a chart.

Additional information

Prerequisite
Detailed description of the options Oscil. frequency, Oscil. amplitude, Oscil.
damping and Signal asymmetry: Assign freq. parameter (→  128)
Description/ ../XXXXXXXX
175.7740.69 kg/h
-100s

0A0016357

 10

*

316

Chart of a measured value trend

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters• x-axis: depending on the number of channels selected displays 250 to 1000 measured
values of a process variable.
• y-axis: displays the approximate measured value span and constantly adapts this to theongoing measurement.

"Displ.channel 2" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 2▸ Displ.channel 2
Display channel 2→  317

Display channel 2
Navigation

Expert → Diagnostics → Data logging → Displ.channel 2

Prerequisite

A process variable is defined in the Assign chan. 2 parameter.

Description

See the Display channel 1 parameter →  316"Displ.channel 3" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 3▸ Displ.channel 3
Display channel 3→  317

Display channel 3
Navigation

Prerequisite

A process variable is defined in the Assign chan. 3 parameter.

Description

See the Display channel 1 parameter →  316

Endress+Hauser

Expert → Diagnostics → Data logging → Displ.channel 3317

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus"Displ.channel 4" submenu
Navigation

Expert → Diagnostics → Data logging → Displ.channel 4▸ Displ.channel 4
Display channel 4→  318

Display channel 4
Navigation

Expert → Diagnostics → Data logging → Displ.channel 4

Prerequisite

A process variable is defined in the Assign chan. 4 parameter.

Description

See the Display channel 1 parameter →  316

3.12.11 "Heartbeat" submenu
For detailed information on the parameter descriptions for the Heartbeat
Verification+Monitoringapplication package, refer to the Special Documentation forthe device →  7
Navigation

 Expert → Diagnostics → Heartbeat‣ Heartbeat‣ Base settings‣ Perform.verific.
‣ Verific. results‣ HBT Monitoring‣ Monitor. results

3.12.12 "Simulation" submenu
Navigation

 Expert → Diagnostics → Simulation‣ Simulation
Assign proc.var. (1810)318→  319

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Proc. var. value (1811)→  320

Status inp. sim. (1355)→  320Signal level (1356)→  321Curr.inp 1 to n sim. (1608–1 to n)→  321

Value curr.inp 1 to n (1609–1 to n)→  322

Curr.out. 1 to n sim. (0354–1 to n)→  322

Value curr.out 1 to n (0355–1 to n)→  322

FreqOutputSim 1 to n (0472–1 to n)→  323

Freq value 1 to n (0473–1 to n)→  323

Puls.outp.sim. 1 to n (0458–1 to n)→  324

Pulse value 1 to n (0459–1 to n)→  324

Switch sim. 1 to n (0462–1 to n)→  324

Switch status 1 to n (0463–1 to n)→  325

Relay out. 1 to n sim (0802–1 to n)→  325

Switch status 1 to n (0803–1 to n)→  326

Dev. alarm sim. (0654)→  326

Event category (0738)→  327

Diag. event sim. (0737)→  327

Assign proc.var.

Navigation

 Expert → Diagnostics → Simulation → Assign proc.var. (1810)

Description

Use this function to select a process variable for the simulation process that is activated.
The display alternates between the measured value and a diagnostic message of the"Function check" category (C) while simulation is in progress.

Selection• Off• Mass flow
• Volume flow
• Correct.vol.flow

Endress+Hauser

319

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus• Density• Ref.density• Temperature• Dynam. viscosity *• Kinematic visc. *• TempCompDynVisc *• TempCompKinVisc *• Concentration *• Target mass flow *• Carrier mass fl. *Factory setting

Off

Additional information

Description
The simulation value of the process variable selected is defined in the Proc. var. value
parameter (→  320).

Proc. var. value

Navigation

 Expert → Diagnostics → Simulation → Proc. var. value (1811)

Prerequisite

A process variable is selected in the Assign proc.var. parameter (→  319).

Description

Use this function to enter a simulation value for the selected process variable. Subsequent
measured value processing and the signal output use this simulation value. In this way,users can verify whether the measuring device has been configured correctly.

User entry

Depends on the process variable selected

Factory setting0

Additional information

User entry
The unit of the displayed measured value is taken from the System units submenu(→  66).

Status inp. sim.



NavigationDescription

Use this function to switch simulation of the status input on and off. The display alternates
between the measured value and a diagnostic message of the "Function check" category (C)while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

*

320

Expert → Diagnostics → Simulation → Status inp. sim. (1355)

Visibility depends on order options or device settings

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Description
The desired simulation value is defined in the Signal level parameter (→  321).
Selection• Off
Simulation for the status input is switched off. The device is in normal measuring modeor another process variable is being simulated.
• On
Simulation for the status input is active.

Signal level

Navigation

Expert → Diagnostics → Simulation → Signal level (1356)

Prerequisite

In the Status inp. sim. parameter (→  320), the On option is selected.

Description

Use this function to select the signal level for the simulation of the status input. In thisway, users can verify the correct configuration of the status input and the correct functionof upstream feed-in units.

Selection• High• LowCurr.inp sim.



Navigation

Expert → Diagnostics → Simulation → Curr.inp 1 to n sim. (1608–1 to n)

Description

Option for switching simulation of the current input on and off. The display alternates
between the measured value and a diagnostic message of the "Function check" category (C)while simulation is in progress.
The desired simulation value is defined in the Value curr.inp 1 to n parameter.

Selection

• Off
• On

Factory setting

Off

Additional information

Selection• Off
Current simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Current simulation is active.

Endress+Hauser

321

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Value curr.inp

Navigation

Expert → Diagnostics → Simulation → Value curr.inp 1 to n (1609–1 to n)

Prerequisite

In the Curr.inp 1 to n sim. parameter, the On option is selected.

Description

Use this function to enter the current value for the simulation. In this way, users can verify
the correct configuration of the current input and the correct function of upstream feed-inunits.

User entry

0 to 22.5 mA

Curr.out. sim.



Navigation

Expert → Diagnostics → Simulation → Curr.out. 1 to n sim. (0354–1 to n)

Description

Use this function to switch simulation of the current output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Value curr.out 1 to n parameter.
Selection• Off
Current simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Current simulation is active.

Value curr.out



Navigation

Prerequisite

In the Curr.out. 1 to n sim. parameter, the On option is selected.

Description

Use this function to enter a current value for the simulation. In this way, users can verify
the correct adjustment of the current output and the correct function of downstream
switching units.

User entry

3.59 to 22.5 mA

322

Expert → Diagnostics → Simulation → Value curr.out 1 to n (0355–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Additional information

Description of device parameters

Dependency
The input range is dependent on the option selected in the Current span parameter(→  109).

FreqOutputSim



Navigation

Expert → Diagnostics → Simulation → FreqOutputSim 1 to n (0472–1 to n)

Prerequisite

In the Operating mode parameter (→  123), the Frequency option is selected.

Description

Use this function to switch simulation of the frequency output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Freq value 1 to n parameter.
Selection• Off
Frequency simulation is switched off. The device is in normal measuring mode oranother process variable is being simulated.
• On
Frequency simulation is active.

Freq value

Navigation

Prerequisite

In the FreqOutputSim 1 to n parameter, the On option is selected.

Description

Use this function to enter a frequency value for the simulation. In this way, users can verify
the correct adjustment of the frequency output and the correct function of downstream
switching units.

User entry

0.0 to 12 500.0 Hz

Endress+Hauser

Expert → Diagnostics → Simulation → Freq value 1 to n (0473–1 to n)

323

Description of device parameters

Proline Promass 300 FOUNDATION Fieldbus

Puls.outp.sim.

Navigation

Expert → Diagnostics → Simulation → Puls.outp.sim. 1 to n (0458–1 to n)

Prerequisite

In the Operating mode parameter (→  123), the Pulse option is selected.

Description

Use this function to switch simulation of the pulse output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection• Off• Fixed value• Down-count. val.

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Pulse value 1 to n parameter.
Selection• Off
Pulse simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Fixed value
Pulses are continuously output with the pulse width specified in the Pulse widthparameter (→  126).
• Down-count. val.
The pulses specified in the Pulse value parameter (→  324) are output.

Pulse value

Navigation

Expert → Diagnostics → Simulation → Pulse value 1 to n (0459–1 to n)

Prerequisite

In the Puls.outp.sim. 1 to n parameter, the Down-count. val. option is selected.

Description

Use this function to enter a pulse value for the simulation. In this way, users can verify thecorrect adjustment of the pulse output and the correct function of downstream switchingunits.

User entry

0 to 65 535

Switch sim.



Navigation

Prerequisite

In the Operating mode parameter (→  123), the Switch option is selected.

324

Expert → Diagnostics → Simulation → Switch sim. 1 to n (0462–1 to n)

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Description

Use this function to switch simulation of the switch output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The desired simulation value is defined in the Switch status 1 to n parameter.
Selection• Off
Switch simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Switch simulation is active.

Switch status



Navigation

Expert → Diagnostics → Simulation → Switch status 1 to n (0463–1 to n)

Description

Use this function to select a switch value for the simulation. In this way, users can verify
the correct adjustment of the switch output and the correct function of downstream
switching units.

Selection• Open• Closed

Additional information

Selection• OpenSwitch simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Closed
Switch simulation is active.

Relay out. 1 to n sim

Navigation

 Expert → Diagnostics → Simulation → Relay out. 1 to n sim (0802–1 to n)

Description

Use this function to switch simulation of the relay output on and off. The display
alternates between the measured value and a diagnostic message of the "Function check"
category (C) while simulation is in progress.

Selection

• Off
• On

Factory setting

Off

Endress+Hauser

325

Description of device parameters

Additional information

Proline Promass 300 FOUNDATION Fieldbus

Description
The desired simulation value is defined in the Switch status 1 to n parameter.
Selection• Off
Relay simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• On
Relay simulation is active.

Switch status 1 to n

Navigation

 Expert → Diagnostics → Simulation → Switch status 1 to n (0803–1 to n)

Prerequisite

The On option is selected in the Switch sim. 1 to n parameter parameter.

Description

Use this function to select a relay value for the simulation. In this way, users can verify thecorrect adjustment of the relay output and the correct function of downstream switchingunits.

Selection• Open• Closed

Additional information

Selection• OpenRelay simulation is switched off. The device is in normal measuring mode or anotherprocess variable is being simulated.
• Closed
Relay simulation is active.

Dev. alarm sim.



Navigation

 Expert → Diagnostics → Simulation → Dev. alarm sim. (0654)

Description

Use this function to switch the device alarm on and off.

Selection

• Off
• On

Factory setting

Off

Additional information

Description
The display alternates between the measured value and a diagnostic message of the"Function check" category (C) while simulation is in progress.

326

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Description of device parameters

Event category



Navigation

Expert → Diagnostics → Simulation → Event category (0738)

Description

Use this function to select the category of the diagnostic events that are displayed for thesimulation in the Diag. event sim. parameter (→  327).

Selection• Sensor• Electronics• Configuration• ProcessFactory setting

Process

Diag. event sim.



Navigation

 Expert → Diagnostics → Simulation → Diag. event sim. (0737)

Description

Use this function to select a diagnostic event for the simulation process that is activated.

Selection• Off• Diagnostic event picklist (depends on the category selected)

Factory setting

Off

Additional information

Description
For the simulation, you can choose from the diagnostic events of the category selectedin the Event category parameter (→  327).

Endress+Hauser

327

Country-specific factory settings

Proline Promass 300 FOUNDATION Fieldbus

4Country-specific factory settings

4.1

SI units
Not valid for USA and Canada.

4.1.1

System units

Mass

kg

Mass flow

kg/hVolumelVolume flow

l/hCorrected volume

NlCorrected volume flowNl/hDensitykg/lReference density

kg/Nl

Temperature°C

Pressure

bar a

4.1.2

Full scale values

The factory settings applie to the following parameters:• 20 mA value (full scale value of the current output)• 100% bar graph value 1

328

Nominal diameter[mm][kg/h]

1

4

2

20

4

90

8

400

15

1 300

15 FB

3 600

25

3 600

25 FB

9 000

40

9 000

40 FB

14 00050

14 00050 FB36 00080

36 000100

60 000150

130 t/h

250

360 t/h

350

650 t/hEndress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

4.1.3

Country-specific factory settings

Output current span

Current output 1 to n

4.1.4

4 to 20 mA NAMUR

Pulse value

Nominal diameter[mm][kg/p]

1

0.001

2

0.01

4

0.01

8

0.1

15

0.1

15 FB

1

25

1

25 FB

1

40

1

40 FB

10

50

10

50 FB

10

80

10

100

10

150

100

250

100

350

100

4.1.5

On value low flow cut off

The switch-on point depends on the type of medium and the nominal diameter.

Endress+Hauser

Nominal diameter[mm]

On-value for liquid[kg/h]

1

0.08

2

0.4

4

1.8

8

8

15

26

15 FB

72

25

72

25 FB

180

40

180

40 FB

300

50

300

50 FB

720

80

720

100

1 200

329

Country-specific factory settings

Proline Promass 300 FOUNDATION Fieldbus

Nominal diameter[mm]

On-value for liquid[kg/h]

150

2.6 t/h

250

7.2 t/h

350

13 t/hNominal diameter[mm]

Switch-on value for gas[kg/h]

1

0.02

2

0.1

4

0.45

8

2

15

6.5

15 FB

18

25

18

25 FB

45

40

45

40 FB

75

50

75

50 FB

180

80

180

100

300

150

650

250

1.8 t/h

350

3.25 t/h

4.2

US units
Only valid for USA and Canada.

4.2.1

330

System units

Mass

lb

Mass flow

lb/minVolumegal (us)

Volume flow

gal/min (us)Corrected volume

Sft³

Corrected volume flow

Sft³/minDensitylb/ft³

Reference density

lb/Sft³

Temperature°F

Pressure

psi a

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

4.2.2

Country-specific factory settings

Full scale values

The factory settings applie to the following parameters:• 20 mA value (full scale value of the current output)• 100% bar graph value 1
Nominal diameter[in][lb/min]

¹/₂₄

0.15

¹/₁₂

0.75

¹/₈

3.3

³/₈

15

½

50

½ FB

130

1

130

1 FB

330

1½

330

1½ FB

550

2

550

2 FB

1 300

3

1 300

4

2 200

6

4 800

10

13 00014

23 5004.2.3

Output current span

Current output 1 to n

4.2.4

Endress+Hauser

4 to 20 mA US

Pulse value

Nominal diameter[in][lb/p]

¹/₂₄

0.002

¹/₁₂

0.02

¹/₈

0.02

³/₈

0.2

½

0.2

½ FB

2

1

2

1 FB

2

1½

2

1½ FB

20

2

20

2 FB

20

3

20

331

Country-specific factory settings

Proline Promass 300 FOUNDATION Fieldbus

Nominal diameter[in][lb/p]

4

20

6

200

10

200

14

200

4.2.5

On value low flow cut off

The switch-on point depends on the type of medium and the nominal diameter.

332

Nominal diameter[in]On-value for liquid[lb/min]

¹/₂₄

0.003

¹/₁₂

0.015

¹/₈

0.066

³/₈

0.3

½

1

½ FB

2.6

1

2.6

1 FB

6.6

1½

6.6

1½ FB

11

2

11

2 FB

26

3

26

4

44

6

95

10

260

14

470

Nominal diameter[in]

Switch-on value for gas[lb/min]

¹/₂₄

0.001

¹/₁₂

0.004

¹/₈

0.016

³/₈

0.075

½

0.25

½ FB

0.65

1

0.65

1 FB

1.65

1½

1.65

1½ FB

2.75

2

2.75

2 FB

6.5

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Endress+Hauser

Country-specific factory settings

Nominal diameter[in]

Switch-on value for gas[lb/min]

3

6.5

4

11

6

23.75

10

65

14

117.5

333

Explanation of abbreviated units

Proline Promass 300 FOUNDATION Fieldbus

5

Explanation of abbreviated units

5.1

SI units

Process variable Units

Explanation

Densityg/cm³, g/m³

Gram/volume unit

kg/dm³, kg/l, kg/m³

Kilogram/volume unit

SD4°C, SD15°C, SD20°C

Specific density: The specific density is the ratio of thedensity of the fluid to the density of water at a watertemperature of 4 °C (39 °F), 15 °C (59 °F), 20 °C (68 °F).

SG4°C, SG15°C, SG20°C

Specific gravity: The specific gravity is the ratio of thedensity of the fluid to the density of water at a watertemperature of 4 °C (39 °F), 15 °C (59 °F), 20 °C (68 °F).

Pa a, kPa a, MPa a

Pascal, kilopascal, megapascal (absolute)

bar

Bar

Pa g, kPa g, MPa g

Pascal, kilopascal, megapascal (relative/gauge)bar g

Bar (relative/gauge)Massg, kg, tGram, kilogram, metric ton

Mass flow

g/s, g/min, g/h, g/dGram/time unit

kg/s, kg/min, kg/h, kg/d

Kilogram/time unit

t/s, t/min, t/h, t/dMetric ton/time unit

Ref.density

kg/Nm³, kg/Nl, g/Scm³, kg/Sm³

Kilogram, gram/standard volume unit

Corrected
volumeNl, Nm³, Sm³

Normal liter, normal cubic meter, standard cubic meter

Correct.vol.flow

Nl/s, Nl/min, Nl/h, Nl/d

Normal liter/time unit

Nm³/s, Nm³/min, Nm³/h, Nm³/d

Normal cubic meter/time unit

Sm³/s, Sm³/min, Sm³/h, Sm³/d

Standard cubic meter/time unit

Temperature°C , K

Celsius, Kelvin

Volumecm³, dm³, m³

Cubic centimeter, cubic decimeter, cubic meter

ml, l, hl, Ml Mega

Milliliter, liter, hectoliter, megaliter

cm³/s, cm³/min, cm³/h, cm³/d

Cubic centimeter/time unit

dm³/s, dm³/min, dm³/h, dm³/dCubic decimeter/time unit

m³/s, m³/min, m³/h, m³/d

Cubic meter/time unit

ml/s, ml/min, ml/h, ml/dMilliliter/time unit

l/s, l/min, l/h, l/d

Liter/time unit

hl/s, hl/min, hl/h, hl/d

Hectoliter/time unit

Ml/s, Ml/min, Ml/h, Ml/d

Megaliter/time unitTimes, m, h, d, ySecond, minute, hour, day, year

5.2

US units

Pressure

Volume flow

334

Process variable Units

Explanation

Densitylb/ft³, lb/gal (us)Pound/cubic foot, pound/gallon

lb/bbl (us;liq.), lb/bbl (us;beer),lb/bbl (us;oil), lb/bbl (us;tank)

Pound/volume unit

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Process variable Units

Explanation

Pressurepsi aPounds per square inch (absolute)psi gPounds per square inch (gauge)Massoz, lb, STon

Ounce, pound, standard ton

Mass flow

oz/s, oz/min, oz/h, oz/dOunce/time unit

lb/s, lb/min, lb/h, lb/dPound/time unit

STon/s, STon/min, STon/h, STon/dStandard ton/time unit

Ref.density

lb/Sft³

Weight unit/standard volume unit

Corrected
volumeSft³, Sgal (us), Sbbl (us;liq.)

Standard cubic foot, standard gallon, standard barrel

Correct.vol.flow

Sft³/s, Sft³/min, Sft³/h, Sft³/dStandard cubic foot/time unit

Sgal/s (us), Sgal/min (us), Sgal/h(us), Sgal/d (us)Standard gallon/time unit

Sbbl/s (us;liq.), Sbbl/min (us;liq.),
Sbbl/h (us;liq.), Sbbl/d (us;liq.)

Barrel/time unit (normal liquids)

Temperature°F, °R

Fahrenheit, Rankine

Volumeaf

Acre footft³

Cubic footfl oz (us), gal (us), kgal (us), Mgal
(us)

Fluid ounce, gallon, kilogallon, million gallonbbl (us;liq.), bbl (us;beer), bbl(us;oil), bbl (us;tank)Barrel (normal liquids), barrel (beer), barrel(petrochemicals), barrel (filling tanks)

af/s, af/min, af/h, af/dAcre foot/time unit

ft³/s, ft³/min, ft³/h, ft³/d

Cubic foot/time unit

fl oz/s (us), fl oz/min (us), fl oz/h(us), fl oz/d (us)Fluid ounce/time unit

gal/s (us), gal/min (us), gal/h (us),gal/d (us)Gallon/time unit

kgal/s (us), kgal/min (us), kgal/h(us), kgal/d (us)Kilogallon/time unit

Mgal/s (us), Mgal/min (us),Mgal/h (us), Mgal/d (us)Million gallon/time unit

bbl/s (us;liq.), bbl/min (us;liq.),bbl/h (us;liq.), bbl/d (us;liq.)

Barrel/time unit (normal liquids)
Normal liquids: 31.5 gal/bbl

bbl/s (us;beer), bbl/min (us;beer),bbl/h (us;beer), bbl/d (us;beer)

Barrel /time unit (beer)
Beer: 31.0 gal/bbl

bbl/s (us;oil), bbl/min (us;oil),bbl/h (us;oil), bbl/d (us;oil)Barrel/time unit (petrochemicals)Petrochemicals: 42.0 gal/bbl

bbl/s (us;tank), bbl/min (us;tank),
bbl/h (us;tank), bbl/d (us;tank)Barrel/time unit (filling tank)
Filling tanks: 55.0 gal/bbl

s, m, h, d, ySecond, minute, hour, day, year

am, pmAnte meridiem ( before midday), post meridiem (after
midday)Volume flow

Time

Endress+Hauser

Explanation of abbreviated units

335

Explanation of abbreviated units

5.3

Imperial units

Process variable Units

Explanation

Densitylb/gal (imp), lb/bbl (imp;beer),lb/bbl (imp;oil)Pound/volume unit

Corrected
volumeSgal (imp)Standard gallon

Correct.vol.flow

Sgal/s (imp), Sgal/min (imp),
Sgal/h (imp), Sgal/d (imp)Standard gallon/time unit

Volumegal (imp), Mgal (imp)

Gallon, mega gallonbbl (imp;beer), bbl (imp;oil)Barrel (beer), barrel (petrochemicals)

gal/s (imp), gal/min (imp), gal/h(imp), gal/d (imp)

Gallon/time unit

Mgal/s (imp), Mgal/min (imp),
Mgal/h (imp), Mgal/d (imp)

Mega gallon/time unit

bbl/s (imp;beer), bbl/min(imp;beer), bbl/h (imp;beer), bbl/d(imp;beer)

Barrel /time unit (beer)
Beer: 36.0 gal/bbl

bbl/s (imp;oil), bbl/min (imp;oil),bbl/h (imp;oil), bbl/d (imp;oil)Barrel/time unit (petrochemicals)Petrochemicals: 34.97 gal/bbl

s, m, h, d, ySecond, minute, hour, day, year

am, pmAnte meridiem ( before midday), post meridiem (after
midday)Volume flow

Time

336

Proline Promass 300 FOUNDATION Fieldbus

Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Index

Index
0…9

0/4 mA value (Parameter) . . . . . . . . . . . . . . . . 103, 110
0% bargraph value 1 (Parameter) . . . . . . . . . . . . . . . 19
0% bargraph value 3 (Parameter) . . . . . . . . . . . . . . . 22
2.4 GHz WLAN channel (Parameter) . . . . . . . . . . . . 175
20 mA value (Parameter) . . . . . . . . . . . . . . . . 103, 112
100% bargraph value 1 (Parameter) . . . . . . . . . . . . . 20
100% bargraph value 3 (Parameter) . . . . . . . . . . . . . 22

A

Access status (Parameter) . . . . . . . . . . . . . . . . . . . . . 13
Acknowledge Option (Parameter) . . . . . . 167, 218, 251
Activate SW option (Parameter) . . . . . . . . . . . . . . . . 48
Active level (Parameter) . . . . . . . . . . . . . . . . . . . . . 106
Actual diagnostics (Parameter) . . . . . . . . . . . . . . . . 281
Actual mode (Parameter) . . . . . 152, 182, 231, 259, 268
Administration (Submenu) . . . . . . . . . . . . . . . . . . . . 45
Alarm delay (Parameter) . . . . . . . . . . . . . . . . . . . . . . 31
Alarm Hysteresis (Parameter) . . . . . . . . . . . . . . . . . 219
Alarm State (Parameter) 165, 168, 216, 222, 224,
225,
227,
253,
254
Alert Key (Parameter) . . . . . . . 150, 181, 230, 259, 268
Analog input 1 to n (Submenu) . . . . . . . . . . . . . . . . 180
Analog inputs (Submenu) . . . . . . . . . . . . . . . . . . . . 179
Analog outputs (Submenu) . . . . . . . . . . . . . . . . . . . 257
Application (Submenu) . . . . . . . . . . . . . . . . . . . . . . 275
Apply I/O configuration (Parameter) . . . . . . . . . . . . 100
Assign behavior of diagnostic no. 046 (Parameter) . . . 33
Assign behavior of diagnostic no. 140 (Parameter) . . . 34
Assign behavior of diagnostic no. 144 (Parameter) . . . 34
Assign behavior of diagnostic no. 374 (Parameter) . . . 35
Assign behavior of diagnostic no. 441 (Parameter) . . . 35
Assign behavior of diagnostic no. 442 (Parameter) . . . 35
Assign behavior of diagnostic no. 443 (Parameter) . . . 36
Assign behavior of diagnostic no. 444 (Parameter) . . . 36
Assign behavior of diagnostic no. 830 (Parameter) . . . 36
Assign behavior of diagnostic no. 831 (Parameter) . . . 37
Assign behavior of diagnostic no. 832 (Parameter) . . . 37
Assign behavior of diagnostic no. 833 (Parameter) . . . 38
Assign behavior of diagnostic no. 834 (Parameter) . . . 38
Assign behavior of diagnostic no. 835 (Parameter) . . . 38
Assign behavior of diagnostic no. 862 (Parameter) . . . 39
Assign behavior of diagnostic no. 912 (Parameter) . . . 39
Assign behavior of diagnostic no. 913 (Parameter) . . . 39
Assign behavior of diagnostic no. 944 (Parameter) . . . 40
Assign behavior of diagnostic no. 948 (Parameter) . . . 40
Assign channel 1 (Parameter) . . . . . . . . . . . . . . . . . 310
Assign channel 2 (Parameter) . . . . . . . . . . . . . . . . . 311
Assign channel 3 (Parameter) . . . . . . . . . . . . . . . . . 312
Assign channel 4 (Parameter) . . . . . . . . . . . . . . . . . 312
Assign current output 1 to n (Parameter) . . . . . . . . . 108
Assign diagnostic behavior (Parameter) . . . . . . 136, 145
Assign flow direction check (Parameter) . . . . . . 139, 144
Assign frequency output (Parameter) . . . . . . . . . . . . 128
Assign limit (Parameter) . . . . . . . . . . . . . . . . . 137, 144
Assign process variable (Parameter) . . . . . . . 76, 80, 276
Endress+Hauser

Assign pulse output 1 to n (Parameter) . . . . . . . . . . 125
Assign simulation process variable (Parameter) . . . . 319
Assign SSID name (Parameter) . . . . . . . . . . . . . . . . 174
Assign status (Parameter) . . . . . . . . . . . . . . . . 140, 145
Assign status input (Parameter) . . . . . . . . . . . . . . . . 105
Assign status of diagnostic number 046 (Parameter) . 40
Assign status of diagnostic number 140 (Parameter) . 41
Assign status of diagnostic number 144 (Parameter) . 41
Assign status of diagnostic number 374 (Parameter) . 41
Assign status of diagnostic number 830 (Parameter) . 42
Assign status of diagnostic number 831 (Parameter) . 42
Assign status of diagnostic number 832 (Parameter) . 42
Assign status of diagnostic number 833 (Parameter) . 43
Assign status of diagnostic number 834 (Parameter) . 43
Assign status of diagnostic number 835 (Parameter) . 43
Assign status of diagnostic number 862 (Parameter) . 44
Assign status of diagnostic number 912 (Parameter) . 44
Assign status of diagnostic number 913 (Parameter) . 44
Assign status of diagnostic number 944 (Parameter) . 45
Assign status of diagnostic number 948 (Parameter) . 45

B

Backlight (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 27
Backup state (Parameter) . . . . . . . . . . . . . . . . . . . . . 29
Block Error (Parameter) . . . . . . 153, 183, 232, 261, 270
Block tag (Parameter) . . . . . . . 149, 180, 229, 258, 267
Bootloader revision (Parameter) . . . . . . . . 296, 298, 299
Build no. software (Parameter) . . . . . . . . . . . . 296, 298

C

C0 to 5 (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Calculated values (Submenu) . . . . . . . . . . . . . . . . . . . 87
Calibration (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 97
Calibration factor (Parameter) . . . . . . . . . . . . . . . . . . 98
Carrier corrected volume flow (Parameter) . . . . . . . . . 57
Carrier mass flow (Parameter) . . . . . . . . . . . . . . . . . . 56
Carrier pipe temperature (Submenu) . . . . . . . . . . . . 302
Carrier volume flow (Parameter) . . . . . . . . . . . . . . . . 58
Channel (Parameter) . . . . . . . . . . . . 208, 246, 261, 270
Clear Fault State (Parameter) . . . . . . . . . . . . . . . . . . 160
Clear logging data (Parameter) . . . . . . . . . . . . . . . . 313
Communication (Submenu) . . . . . . . . . . . . . . . . . . . 148
Comparison result (Parameter) . . . . . . . . . . . . . . . . . 29
Concentration (Parameter) . . . . . . . . . . . . . . . . . . . . 55
Concentration (Submenu) . . . . . . . . . . . . . . . . . . . . 280
Configuration backup (Submenu) . . . . . . . . . . . . . . . . 27
Configuration management (Parameter) . . . . . . . . . . 28
Confirm access code (Parameter) . . . . . . . . . . . . . . . . 47
Confirm Time (Parameter) . . . . . . . . . . . . . . . . . . . . 160
Connection state (Parameter) . . . . . . . . . . . . . . . . . 175
Contrast display (Parameter) . . . . . . . . . . . . . . . . . . . 27
Control Totalizer 1 to n (Parameter) . . . . . . . . . . . . . 278
Conversion code (Parameter) . . . . . . . . . . . . . . . . . . 101
Corrected volume flow (Parameter) . . . . . . . . . . . . . . 52
Corrected volume flow calculation (Parameter) . . . . . . 87
Corrected volume flow calculation (Submenu) . . . . . . 87
337

IndexCorrected volume flow factor (Parameter) . . . . . . . . . 96
Corrected volume flow offset (Parameter) . . . . . . . . . 95
Corrected volume flow unit (Parameter) . . . . . . . . . . . 69
Corrected volume unit (Parameter) . . . . . . . . . . . . . . 70
Current (Parameter) . . . . . . . . . . . . . . . . 166, 217, 252
Current input 1 to n (Submenu) . . . . . . . . . . . . . 60, 101
Current input 1 to n simulation (Parameter) . . . . . . . 321
Current output 1 to n (Submenu) . . . . . . . . . . . . . . . 107
Current output 1 to n simulation (Parameter) . . . . . . 322
Current span (Parameter) . . . . . . . . . . . . . . . . 102, 109
Cycle Selection (Parameter) . . . . . . . . . . . . . . . . . . . 159
Cycle Type (Parameter) . . . . . . . . . . . . . . . . . . . . . . 158

D

Damping output 1 to n (Parameter) . . . . . . . . . 117, 132
Data logging (Parameter) . . . . . . . . . . . . . . . . . . . . 313
Data logging (Submenu) . . . . . . . . . . . . . . . . . . . . . 309
Data logging control (Parameter) . . . . . . . . . . . . . . . 314
Data logging status (Parameter) . . . . . . . . . . . . . . . 315
Date/time format (Parameter) . . . . . . . . . . . . . . . . . . 73
DD Resource (Parameter) . . . . . . . . . . . . . . . . . . . . . 155
DD Revision (Parameter) . . . . . . . . . . . . . . . . . 156, 292
Decimal (Parameter) . . . . . . . . . . . . . . . . . . . . 197, 207
Decimal places 1 (Parameter) . . . . . . . . . . . . . . . . . . 20
Decimal places 2 (Parameter) . . . . . . . . . . . . . . . . . . 21
Decimal places 3 (Parameter) . . . . . . . . . . . . . . . . . . 23
Decimal places 4 (Parameter) . . . . . . . . . . . . . . . . . . 24
Default gateway (Parameter) . . . . . . . . . . . . . . . . . . 178
Define access code (Parameter) . . . . . . . . . . . . . . . . . 46
Define access code (Wizard) . . . . . . . . . . . . . . . . . . . 46
Density (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 53
Density damping (Parameter) . . . . . . . . . . . . . . . . . . 75
Density factor (Parameter) . . . . . . . . . . . . . . . . . . . . 95
Density offset (Parameter) . . . . . . . . . . . . . . . . . . . . . 95
Density unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 71
Deny (Parameter) . . . . . . . . . . . . . . . . . . 156, 207, 245
Device address (Parameter) . . . . . . . . . . . . . . . . . . . 149
Device alarm simulation (Parameter) . . . . . . . . . . . . 326
Device information (Submenu) . . . . . . . . . . . . . . . . 289
Device name (Parameter) . . . . . . . . . . . . . . . . . . . . 291
Device reset (Parameter) . . . . . . . . . . . . . . . . . . . . . . 48
Device revision (Parameter) . . . . . . . . . . . . . . . 156, 291
Device tag (Parameter) . . . . . . . . . . . . . . . . . . . . . . 290
Device type (Parameter) . . . . . . . . . . . . . . . . . 155, 291
Diagnostic behavior (Submenu) . . . . . . . . . . . . . . . . . 31
Diagnostic event category (Parameter) . . . . . . . . . . . 327
Diagnostic event simulation (Parameter) . . . . . . . . . 327
Diagnostic handling (Submenu) . . . . . . . . . . . . . . . . . 30
Diagnostic list (Submenu) . . . . . . . . . . . . . . . . . . . . 283
Diagnostics (Submenu) . . . . . . . . . . . . . . . . . . . . . . 280
Diagnostics 1 (Parameter) . . . . . . . . . . . . . . . . . . . . 283
Diagnostics 2 (Parameter) . . . . . . . . . . . . . . . . . . . . 284
Diagnostics 3 (Parameter) . . . . . . . . . . . . . . . . . . . . 285
Diagnostics 4 (Parameter) . . . . . . . . . . . . . . . . . . . . 286
Diagnostics 5 (Parameter) . . . . . . . . . . . . . . . . . . . . 286
Direct access
0/4 mA value
Current input 1 to n (1606–1 to n) . . . . . . . . 103
Current output 1 to n (0367–1 to n) . . . . . . . 110
338

Proline Promass 300 FOUNDATION Fieldbus

0% bargraph value 1 (0123) . . . . . . . . . . . . . . . . . 19
0% bargraph value 3 (0124) . . . . . . . . . . . . . . . . . 22
2.4 GHz WLAN channel (2704) . . . . . . . . . . . . . 175
20 mA value
Current input 1 to n (1607–1 to n) . . . . . . . . 103
Current output 1 to n (0372–1 to n) . . . . . . . 112
100% bargraph value 1 (0125) . . . . . . . . . . . . . . . 20
100% bargraph value 3 (0126) . . . . . . . . . . . . . . . 22
Access status (0005) . . . . . . . . . . . . . . . . . . . . . . 13
Acknowledge Option
Analog input 1 to n (6910–1 to n) . . . . . . . . . 218
Discrete input 1 to n (6841–1 to n) . . . . . . . . 251
Acknowledge Option (10691) . . . . . . . . . . . . . . . 167
Activate SW option (0029) . . . . . . . . . . . . . . . . . . 48
Active level
Status input 1 to n (1351–1 to n) . . . . . . . . . . 106
Actual diagnostics (0691) . . . . . . . . . . . . . . . . . . 281
Actual mode
Analog input 1 to n (6957–1 to n) . . . . . . . . . 182
Discrete input 1 to n (6870–1 to n) . . . . . . . . 231
Actual mode (10725) . . . . . . . . . . . . . . . . . . . . . 152
Actual mode (11264) . . . . . . . . . . . . . . . . . . . . . 268
Actual mode (11366) . . . . . . . . . . . . . . . . . . . . . 259
Alarm delay (0651) . . . . . . . . . . . . . . . . . . . . . . . 31
Alarm Hysteresis
Analog input 1 to n (6911–1 to n) . . . . . . . . . 219
Alarm State
Analog input 1 to n (6917–1 to n) . . . . . . . . . 216
Analog input 1 to n (6927–1 to n) . . . . . . . . . 224
Analog input 1 to n (6932–1 to n) . . . . . . . . . 222
Analog input 1 to n (6942–1 to n) . . . . . . . . . 225
Analog input 1 to n (6948–1 to n) . . . . . . . . . 227
Discrete input 1 to n (6847–1 to n) . . . . . . . . 253
Discrete input 1 to n (6858–1 to n) . . . . . . . . 254
Alarm State (10697) . . . . . . . . . . . . . . . . . . . . . 165
Alarm State (10742) . . . . . . . . . . . . . . . . . . . . . 168
Alert Key
Analog input 1 to n (6916–1 to n) . . . . . . . . . 181
Discrete input 1 to n (6846–1 to n) . . . . . . . . 230
Alert Key (10696) . . . . . . . . . . . . . . . . . . . . . . . 150
Alert Key (11251) . . . . . . . . . . . . . . . . . . . . . . . 268
Alert Key (11365) . . . . . . . . . . . . . . . . . . . . . . . 259
Apply I/O configuration (3907) . . . . . . . . . . . . . 100
Assign behavior of diagnostic no. 046 (0709) . . . . 33
Assign behavior of diagnostic no. 140 (0708) . . . . 34
Assign behavior of diagnostic no. 144 (0731) . . . . 34
Assign behavior of diagnostic no. 374 (0710) . . . . 35
Assign behavior of diagnostic no. 441 (0657) . . . . 35
Assign behavior of diagnostic no. 442 (0658) . . . . 35
Assign behavior of diagnostic no. 443 (0659) . . . . 36
Assign behavior of diagnostic no. 444 (0740) . . . . 36
Assign behavior of diagnostic no. 830 (0800) . . . . 36
Assign behavior of diagnostic no. 831 (0641) . . . . 37
Assign behavior of diagnostic no. 832 (0681) . . . . 37
Assign behavior of diagnostic no. 833 (0682) . . . . 38
Assign behavior of diagnostic no. 834 (0700) . . . . 38
Assign behavior of diagnostic no. 835 (0702) . . . . 38
Assign behavior of diagnostic no. 862 (0679) . . . . 39
Assign behavior of diagnostic no. 912 (0703) . . . . 39
Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Assign behavior of diagnostic no. 913 (0712) . . . . 39
Assign behavior of diagnostic no. 944 (0732) . . . . 40
Assign behavior of diagnostic no. 948 (0744) . . . . 40
Assign channel 1 (0851) . . . . . . . . . . . . . . . . . . 310
Assign channel 2 (0852) . . . . . . . . . . . . . . . . . . 311
Assign channel 3 (0853) . . . . . . . . . . . . . . . . . . 312
Assign channel 4 (0854) . . . . . . . . . . . . . . . . . . 312
Assign current output 1 to n (0359–1 to n) . . . . . 108
Assign diagnostic behavior
Pulse/frequency/switch output 1 to n (0482–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 136
Relay output 1 to n (0806–1 to n) . . . . . . . . . 145
Assign flow direction check
Pulse/frequency/switch output 1 to n (0484–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Relay output 1 to n (0808–1 to n) . . . . . . . . . 144
Assign frequency output
Pulse/frequency/switch output 1 to n (0478–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 128
Assign limit
Pulse/frequency/switch output 1 to n (0483–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 137
Relay output 1 to n (0807–1 to n) . . . . . . . . . 144
Assign process variable
Totalizer 1 to n (0914–1 to n) . . . . . . . . . . . . 276
Assign process variable (1837) . . . . . . . . . . . . . . . 76
Assign process variable (1860) . . . . . . . . . . . . . . . 80
Assign pulse output 1 to n (0460–1 to n) . . . . . . 125
Assign simulation process variable (1810) . . . . . 319
Assign SSID name (2708) . . . . . . . . . . . . . . . . . . 174
Assign status
Pulse/frequency/switch output 1 to n (0485–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Relay output 1 to n (0805–1 to n) . . . . . . . . . 145
Assign status input
Status input 1 to n (1352–1 to n) . . . . . . . . . . 105
Assign status of diagnostic number 046 (11042) . 40
Assign status of diagnostic number 140 (11041) . 41
Assign status of diagnostic number 144 (11016) . 41
Assign status of diagnostic number 374 (11000) . 41
Assign status of diagnostic number 830 (11006) . 42
Assign status of diagnostic number 831 (11007) . 42
Assign status of diagnostic number 832 (11002) . 42
Assign status of diagnostic number 833 (11003) . 43
Assign status of diagnostic number 834 (11004) . 43
Assign status of diagnostic number 835 (11005) . 43
Assign status of diagnostic number 862 (11008) . 44
Assign status of diagnostic number 912 (11009) . 44
Assign status of diagnostic number 913 (11010) . 44
Assign status of diagnostic number 944 (11014) . 45
Assign status of diagnostic number 948 (11011) . 45
Backlight (0111) . . . . . . . . . . . . . . . . . . . . . . . . . 27
Backup state (2759) . . . . . . . . . . . . . . . . . . . . . . . 29
Block Error
Analog input 1 to n (6922–1 to n) . . . . . . . . . 183
Discrete input 1 to n (6857–1 to n) . . . . . . . . 232
Block Error (10703) . . . . . . . . . . . . . . . . . . . . . . 153
Block Error (11272) . . . . . . . . . . . . . . . . . . . . . . 270
Block Error (11364) . . . . . . . . . . . . . . . . . . . . . . 261
Endress+Hauser

Index

Block tag
Analog input 1 to n (6901–1 to n) . . . . . . . . . 180
Discrete input 1 to n (6851–1 to n) . . . . . . . . 229
Block tag (10702) . . . . . . . . . . . . . . . . . . . . . . . 149
Block tag (11252) . . . . . . . . . . . . . . . . . . . . . . . 267
Block tag (11351) . . . . . . . . . . . . . . . . . . . . . . . 258
Bootloader revision
I/O module 1 (0073) . . . . . . . . . . . . . . . . . . . 298
I/O module 2 (0073) . . . . . . . . . . . . . . . . . . . 298
I/O module 3 (0073) . . . . . . . . . . . . . . . . . . . 298
I/O module 4 (0073) . . . . . . . . . . . . . . . . . . . 298
Bootloader revision (0073) . . . . . . . . . . . . . 296, 299
Build no. software
I/O module 1 (0079) . . . . . . . . . . . . . . . . . . . 298
I/O module 2 (0079) . . . . . . . . . . . . . . . . . . . 298
I/O module 3 (0079) . . . . . . . . . . . . . . . . . . . 298
I/O module 4 (0079) . . . . . . . . . . . . . . . . . . . 298
Build no. software (0079) . . . . . . . . . . . . . . 296, 298
C0 to 5 (6022) . . . . . . . . . . . . . . . . . . . . . . . . . . . 99
Calibration factor (6025) . . . . . . . . . . . . . . . . . . . 98
Carrier corrected volume flow (1894) . . . . . . . . . . 57
Carrier mass flow (1865) . . . . . . . . . . . . . . . . . . . 56
Carrier volume flow (1896) . . . . . . . . . . . . . . . . . 58
ChannelAnalog input 1 to n (6902–1 to n) . . . . . . . . . 208
Discrete input 1 to n (6852–1 to n) . . . . . . . . 246
Channel (11253) . . . . . . . . . . . . . . . . . . . . . . . . 270
Channel (11352) . . . . . . . . . . . . . . . . . . . . . . . . 261
Clear Fault State (10704) . . . . . . . . . . . . . . . . . . 160
Clear logging data (0855) . . . . . . . . . . . . . . . . . . 313
Comparison result (2760) . . . . . . . . . . . . . . . . . . . 29
Concentration (1887) . . . . . . . . . . . . . . . . . . . . . . 55
Configuration management (2758) . . . . . . . . . . . 28
Confirm Time (10705) . . . . . . . . . . . . . . . . . . . . 160
Connection state (2722) . . . . . . . . . . . . . . . . . . . 175
Contrast display (0105) . . . . . . . . . . . . . . . . . . . . 27
Control Totalizer 1 to n (0912–1 to n) . . . . . . . . 278
Conversion code (2762) . . . . . . . . . . . . . . . . . . . 101
Corrected volume flow (1851) . . . . . . . . . . . . . . . 52
Corrected volume flow calculation (1812) . . . . . . . 87
Corrected volume flow factor (1867) . . . . . . . . . . . 96
Corrected volume flow offset (1866) . . . . . . . . . . . 95
Corrected volume flow unit (0558) . . . . . . . . . . . . 69
Corrected volume unit (0575) . . . . . . . . . . . . . . . . 70
Current
Analog input 1 to n (6912–1 to n) . . . . . . . . . 217
Discrete input 1 to n (6842–1 to n) . . . . . . . . 252
Current (10692) . . . . . . . . . . . . . . . . . . . . . . . . 166
Current input 1 to n simulation (1608–1 to n) . . . 321
Current output 1 to n simulation (0354–1 to n) . . 322
Current span
Current input 1 to n (1605–1 to n) . . . . . . . . 102
Current output 1 to n (0353–1 to n) . . . . . . . 109
Cycle Selection (10706) . . . . . . . . . . . . . . . . . . . 159
Cycle Type (10707) . . . . . . . . . . . . . . . . . . . . . . 158
Damping output 1 to n (0363–1 to n) . . . . . . . . . 117
Damping output 1 to n (0477–1 to n) . . . . . . . . . 132
Data logging (0860) . . . . . . . . . . . . . . . . . . . . . . 313
Data logging control (0857) . . . . . . . . . . . . . . . . 314
339

Index

Data logging status (0858) . . . . . . . . . . . . . . . . . 315
Date/time format (2812) . . . . . . . . . . . . . . . . . . . 73
DD Resource (10708) . . . . . . . . . . . . . . . . . . . . . 155
DD Revision (10709) . . . . . . . . . . . . . . . . . 156, 292
DecimalAnalog input 1 to n (6961–1 to n) . . . . . . . . . 207
Analog input 1 to n (6980–1 to n) . . . . . . . . . 197
Decimal places 1 (0095) . . . . . . . . . . . . . . . . . . . . 20
Decimal places 2 (0117) . . . . . . . . . . . . . . . . . . . . 21
Decimal places 3 (0118) . . . . . . . . . . . . . . . . . . . . 23
Decimal places 4 (0119) . . . . . . . . . . . . . . . . . . . . 24
Default gateway (7210) . . . . . . . . . . . . . . . . . . . 178
Density (1850) . . . . . . . . . . . . . . . . . . . . . . . . . . . 53
Density damping (1803) . . . . . . . . . . . . . . . . . . . . 75
Density factor (1849) . . . . . . . . . . . . . . . . . . . . . . 95
Density offset (1848) . . . . . . . . . . . . . . . . . . . . . . 95
Density unit (0555) . . . . . . . . . . . . . . . . . . . . . . . 71
Deny
Analog input 1 to n (6925–1 to n) . . . . . . . . . 207
Discrete input 1 to n (6867–1 to n) . . . . . . . . 245
Deny (10717) . . . . . . . . . . . . . . . . . . . . . . . . . . 156
Device address (11061) . . . . . . . . . . . . . . . . . . . 149
Device alarm simulation (0654) . . . . . . . . . . . . . 326
Device name (0020) . . . . . . . . . . . . . . . . . . . . . . 291
Device reset (0000) . . . . . . . . . . . . . . . . . . . . . . . 48
Device revision (10710) . . . . . . . . . . . . . . . 156, 291
Device tag (10799) . . . . . . . . . . . . . . . . . . . . . . . 290
Device type (10711) . . . . . . . . . . . . . . . . . . 155, 291
Diagnostic event category (0738) . . . . . . . . . . . . 327
Diagnostic event simulation (0737) . . . . . . . . . . 327
Diagnostics 1 (0692) . . . . . . . . . . . . . . . . . . . . . 283
Diagnostics 2 (0693) . . . . . . . . . . . . . . . . . . . . . 284
Diagnostics 3 (0694) . . . . . . . . . . . . . . . . . . . . . 285
Diagnostics 4 (0695) . . . . . . . . . . . . . . . . . . . . . 286
Diagnostics 5 (0696) . . . . . . . . . . . . . . . . . . . . . 286
Direct access (0106) . . . . . . . . . . . . . . . . . . . . . . . 11
Disabled
Analog input 1 to n (6913–1 to n) . . . . . . . . . 218
Discrete input 1 to n (6843–1 to n) . . . . . . . . 252
Disabled (10693) . . . . . . . . . . . . . . . . . . . . . . . . 167
Discrete Limit
Discrete input 1 to n (6863–1 to n) . . . . . . . . 256
Discrete Priority
Discrete input 1 to n (6864–1 to n) . . . . . . . . 257
Discrete Value
Discrete input 1 to n (6862–1 to n) . . . . . . . . 256
Discrete Value (10746) . . . . . . . . . . . . . . . . . . . 169
Display damping (0094) . . . . . . . . . . . . . . . . . . . . 25
Display interval (0096) . . . . . . . . . . . . . . . . . . . . . 24
Display language (0104) . . . . . . . . . . . . . . . . . . . 15
Dynamic viscosity (1854) . . . . . . . . . . . . . . . . . . . 54
ENP version (10791) . . . . . . . . . . . . . . . . . . . . . 294
Enter access code (0003) . . . . . . . . . . . . . . . . . . . 13
Entire logging duration (0861) . . . . . . . . . . . . . . 315
EU at 0%
Analog input 1 to n (6962–1 to n) . . . . . . . . . 198
Analog input 1 to n (6981–1 to n) . . . . . . . . . 195
EU at 100%
Analog input 1 to n (6963–1 to n) . . . . . . . . . 198
340

Proline Promass 300 FOUNDATION Fieldbus

Analog input 1 to n (6982–1 to n) . . . . . . . . . 194
Extended order code 1 (10796) . . . . . . . . . . . . . 293
Extended order code 2 (10797) . . . . . . . . . . . . . 293
External pressure (6209) . . . . . . . . . . . . . . . . . . . 85
External reference density (6198) . . . . . . . . . . . . . 88
External temperature (6080) . . . . . . . . . . . . . . . . 86
Failure current
Current output 1 to n (0352–1 to n) . . . . . . . 120
Failure frequency
Pulse/frequency/switch output 1 to n (0474–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Failure mode
Current input 1 to n (1601–1 to n) . . . . . . . . 103
Current output 1 to n (0364–1 to n) . . . . . . . 119
Pulse/frequency/switch output 1 to n (0451–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 134
Pulse/frequency/switch output 1 to n (0480–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 127
Pulse/frequency/switch output 1 to n (0486–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 141
Relay output 1 to n (0811–1 to n) . . . . . . . . . 147
Totalizer 1 to n (0901–1 to n) . . . . . . . . . . . . 279
Failure value
Current input 1 to n (1602–1 to n) . . . . . . . . 104
Fault State (10712) . . . . . . . . . . . . . . . . . . . . . . 161
Fault State Status (11254) . . . . . . . . . . . . . . . . . 274
Fault State Status (11353) . . . . . . . . . . . . . . . . . 265
Fault State Time (11255) . . . . . . . . . . . . . . . . . . 271
Fault State Time (11354) . . . . . . . . . . . . . . . . . . 262
Fault State Val 1 (11355) . . . . . . . . . . . . . . . . . . 263
Fault State Val 2 (11356) . . . . . . . . . . . . . . . . . . 263
Fault State Val 3 (11357) . . . . . . . . . . . . . . . . . . 263
Fault State Val 4 (11358) . . . . . . . . . . . . . . . . . . 264
Fault State Val 5 (11359) . . . . . . . . . . . . . . . . . . 264
Fault State Val 6 (11360) . . . . . . . . . . . . . . . . . . 264
Fault State Val 7 (11361) . . . . . . . . . . . . . . . . . . 265
Fault State Val 8 (11362) . . . . . . . . . . . . . . . . . . 265
Fault State Value Discrete 1 (11256) . . . . . . . . . 272
Fault State Value Discrete 2 (11257) . . . . . . . . . 272
Fault State Value Discrete 3 (11258) . . . . . . . . . 272
Fault State Value Discrete 4 (11259) . . . . . . . . . 273
Fault State Value Discrete 5 (11260) . . . . . . . . . 273
Fault State Value Discrete 6 (11261) . . . . . . . . . 273
Fault State Value Discrete 7 (11262) . . . . . . . . . 274
Fault State Value Discrete 8 (11263) . . . . . . . . . 274
Feature Selection (10714) . . . . . . . . . . . . . . . . . 158
Features (10713) . . . . . . . . . . . . . . . . . . . . . . . . 158
Filter options . . . . . . . . . . . . . . . . . . . . . . . . . . . 288
Filter options (0705) . . . . . . . . . . . . . . . . . . . . . 287
Firmware version (10792) . . . . . . . . . . . . . . . . . 292
Fixed current
Current output 1 to n (0365–1 to n) . . . . . . . 110
Fixed reference density (1814) . . . . . . . . . . . . . . . 88
Float Value
Analog input 1 to n (6931–1 to n) . . . . . . . . . 225
Analog input 1 to n (6936–1 to n) . . . . . . . . . 223
Analog input 1 to n (6946–1 to n) . . . . . . . . . 226
Analog input 1 to n (6952–1 to n) . . . . . . . . . 228
Flow damping (1802) . . . . . . . . . . . . . . . . . . . . . . 74
Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Flow override (1839) . . . . . . . . . . . . . . . . . . . . . . 76
Format display (0098) . . . . . . . . . . . . . . . . . . . . . 15
Free Space (10715) . . . . . . . . . . . . . . . . . . . . . . 160
Free Time (10716) . . . . . . . . . . . . . . . . . . . . . . . 160
Frequency output simulation 1 to n (0472–1 to n). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 323
Frequency value 1 to n (0473–1 to n) . . . . . . . . . 323
Gateway IP address (2719) . . . . . . . . . . . . . . . . . 176
Grant
Analog input 1 to n (6926–1 to n) . . . . . . . . . 207
Discrete input 1 to n (6868–1 to n) . . . . . . . . 245
Grant (10718) . . . . . . . . . . . . . . . . . . . . . . . . . . 156
Hard Types (10719) . . . . . . . . . . . . . . . . . . . . . . 157
Hardware revision (10793) . . . . . . . . . . . . . . . . 292
Header (0097) . . . . . . . . . . . . . . . . . . . . . . . . . . . 25
Header text (0112) . . . . . . . . . . . . . . . . . . . . . . . 26
High High Limit
Analog input 1 to n (6937–1 to n) . . . . . . . . . 219
High High Priority
Analog input 1 to n (6938–1 to n) . . . . . . . . . 219
High Limit
Analog input 1 to n (6939–1 to n) . . . . . . . . . 220
High Priority
Analog input 1 to n (6940–1 to n) . . . . . . . . . 220
High value partial filled pipe detection (1858) . . . . 80
I/O module 1 terminal numbers (3902–1) . . 296, 297
I/O module 1 to n information (3906–1 to n) . . . . 99
I/O module 1 to n terminal numbers (3902–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 99
I/O module 1 to n type (3901–1 to n) . . . . . . . . . 100
I/O Options
Analog input 1 to n (6941–1 to n) . . . . . . . . . 207
Discrete input 1 to n (6869–1 to n) . . . . . . . . 245
Input signal level (1356) . . . . . . . . . . . . . . . . . . 321
Installation angle pitch (6236) . . . . . . . . . . . . . . . 91
Installation angle roll (6282) . . . . . . . . . . . . . . . . 91
Installation direction (1809) . . . . . . . . . . . . . . . . . 90
Invert output signal
Pulse/frequency/switch output 1 to n (0470–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 142
IP address (7209) . . . . . . . . . . . . . . . . . . . . . . . . 178
IP address domain name server (2720) . . . . . . . . 176
ITK Version (10794) . . . . . . . . . . . . . . . . . . 170, 292
Kinematic viscosity (1857) . . . . . . . . . . . . . . . . . . 54
Last backup (2757) . . . . . . . . . . . . . . . . . . . . . . . 28
Limit Notify (10720) . . . . . . . . . . . . . . . . . . . . . 161
Linear expansion coefficient (1817) . . . . . . . . . . . 89
Linearization Type
Analog input 1 to n (6905–1 to n) . . . . . . . . . 209
Locking status (0004) . . . . . . . . . . . . . . . . . . . . . 12
Logging delay (0859) . . . . . . . . . . . . . . . . . . . . . 314
Logging interval (0856) . . . . . . . . . . . . . . . . . . . 312
Login page (7273) . . . . . . . . . . . . . . . . . . . . . . . 179
Low Cutoff
Analog input 1 to n (6956–1 to n) . . . . . . . . . 211
Low Limit
Analog input 1 to n (6947–1 to n) . . . . . . . . . 221
Low Low Limit
Analog input 1 to n (6953–1 to n) . . . . . . . . . 221
Endress+Hauser

Index

Low Low Priority
Analog input 1 to n (6954–1 to n) . . . . . . . . . 221
Low Priority
Analog input 1 to n (6955–1 to n) . . . . . . . . . 220
Low value partial filled pipe detection (1861) . . . . 80
MAC address (7214) . . . . . . . . . . . . . . . . . . . . . 177
Manufacturer Id (10721) . . . . . . . . . . . . . . . . . . 155
Mass flow (1838) . . . . . . . . . . . . . . . . . . . . . . . . . 52
Mass flow factor (1832) . . . . . . . . . . . . . . . . . . . . 94
Mass flow offset (1831) . . . . . . . . . . . . . . . . . . . . 93
Mass flow unit (0554) . . . . . . . . . . . . . . . . . . . . . 66
Mass unit (0574) . . . . . . . . . . . . . . . . . . . . . . . . . 67
Max Notify (10722) . . . . . . . . . . . . . . . . . . . . . . 161
Maximum damping partial filled pipe det. (6040) . 81
Maximum frequency value
Pulse/frequency/switch output 1 to n (0454–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Maximum value (6009) . . . . . . . . . . . . . . . 305, 306
Maximum value (6014) . . . . . . . . . . . . . . . . . . . 308
Maximum value (6029) . . . . . . . . . . . . . . . . . . . 302
Maximum value (6051) . . . . . . . . . . . . . . . . . . . 300
Maximum value (6068) . . . . . . . . . . . . . . . . . . . 304
Maximum value (6070) . . . . . . . . . . . . . . . . . . . 303
Maximum value (6108) . . . . . . . . . . . . . . . . . . . 301
Maximum value (6119) . . . . . . . . . . . . . . . . . . . 307
Maximum value (6121) . . . . . . . . . . . . . . . . . . . 307
Maximum value (6283) . . . . . . . . . . . . . . . . . . . 309
Measured current 1 to n (0366–1 to n) . . . . . . . . 121
Measured current 1 to n (1604–1 to n) . . . . . . . . . 61
Measured values 1 to n (1603–1 to n) . . . . . . . . . 61
Measuring mode
Current output 1 to n (0351–1 to n) . . . . . . . 113
Pulse/frequency/switch output 1 to n (0457–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 126
Pulse/frequency/switch output 1 to n (0479–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 131
Measuring value at maximum frequency
Pulse/frequency/switch output 1 to n (0475–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Measuring value at minimum frequency
Pulse/frequency/switch output 1 to n (0476–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Memory Size (10723) . . . . . . . . . . . . . . . . . . . . 159
Minimum frequency value
Pulse/frequency/switch output 1 to n (0453–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 129
Minimum value (6008) . . . . . . . . . . . . . . . . . . . 306
Minimum value (6010) . . . . . . . . . . . . . . . . . . . 305
Minimum value (6015) . . . . . . . . . . . . . . . . . . . 308
Minimum value (6030) . . . . . . . . . . . . . . . . . . . 302
Minimum value (6052) . . . . . . . . . . . . . . . . . . . 300
Minimum value (6069) . . . . . . . . . . . . . . . . . . . 304
Minimum value (6071) . . . . . . . . . . . . . . . . . . . 303
Minimum value (6109) . . . . . . . . . . . . . . . . . . . 301
Minimum value (6120) . . . . . . . . . . . . . . . . . . . 307
Minimum value (6122) . . . . . . . . . . . . . . . . . . . 306
Minimum value (6284) . . . . . . . . . . . . . . . . . . . 309
Minumum Cycle Time (10724) . . . . . . . . . . . . . . 159
Multi-frequency activation (6242) . . . . . . . . . . . . 82
341

Index

Network security (2705) . . . . . . . . . . . . . . . . . . 172
Nominal diameter (2807) . . . . . . . . . . . . . . . . . . . 98
Nonvolatile Cycle Time (10729) . . . . . . . . . . . . . 159
Normal mode
Analog input 1 to n (6958–1 to n) . . . . . . . . . 183
Discrete input 1 to n (6871–1 to n) . . . . . . . . 232
Normal mode (10726) . . . . . . . . . . . . . . . . . . . . 153
Normal mode (11265) . . . . . . . . . . . . . . . . . . . . 269
Normal mode (11367) . . . . . . . . . . . . . . . . . . . . 260
Off value low flow cutoff (1804) . . . . . . . . . . . . . . 77
On value low flow cutoff (1805) . . . . . . . . . . . . . . 77
Operating mode
Pulse/frequency/switch output 1 to n (0469–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Operating time (0652) . . . . . . . . . . . . . . 28, 47, 283
Operating time from restart (0653) . . . . . . . . . . 282
Order code (10795) . . . . . . . . . . . . . . . . . . . . . . 293
Output current 1 to n (0361–1 to n) . . . . . . . . . . 120
Output frequency 1 to n (0471–1 to n) . . . . . . . . 135
Output State
Discrete input 1 to n (6874–1 to n) . . . . . . . . 245
Permitted mode
Analog input 1 to n (6959–1 to n) . . . . . . . . . 182
Discrete input 1 to n (6872–1 to n) . . . . . . . . 231
Permitted mode (10727) . . . . . . . . . . . . . . . . . . 152
Permitted mode (11266) . . . . . . . . . . . . . . . . . . 269
Permitted mode (11368) . . . . . . . . . . . . . . . . . . 260
Powerless relay status
Relay output 1 to n (0816–1 to n) . . . . . . . . . 148
Preset value 1 to n (0913–1 to n) . . . . . . . . . . . . 278
Pressure compensation (6130) . . . . . . . . . . . . . . . 84
Pressure shock suppression (1806) . . . . . . . . . . . . 78
Pressure unit (0564) . . . . . . . . . . . . . . . . . . . . . . 73
Pressure value (6059) . . . . . . . . . . . . . . . . . . . . . 85
Pressure value (6129) . . . . . . . . . . . . . . . . . . . . . 54
Previous diagnostics (0690) . . . . . . . . . . . . . . . . 282
Process Value Filter Time
Analog input 1 to n (6909–1 to n) . . . . . . . . . 214
Discrete input 1 to n (6855–1 to n) . . . . . . . . 247
Process variable value (1811) . . . . . . . . . . . . . . . 320
Progress (2808) . . . . . . . . . . . . . . . . . . . . . . . . . . 93
Pulse output 1 to n (0456–1 to n) . . . . . . . . . . . . 128
Pulse output simulation 1 to n (0458–1 to n) . . . 324
Pulse value 1 to n (0459–1 to n) . . . . . . . . . . . . . 324
Pulse width
Pulse/frequency/switch output 1 to n (0452–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 126
Received signal strength (2721) . . . . . . . . . . . . . 175
Reference density (1852) . . . . . . . . . . . . . . . . . . . 53
Reference density factor (1869) . . . . . . . . . . . . . . 96
Reference density offset (1868) . . . . . . . . . . . . . . 96
Reference density unit (0556) . . . . . . . . . . . . . . . 71
Reference sound velocity (6147) . . . . . . . . . . . . . . 83
Reference temperature (1816) . . . . . . . . . . . . . . . 89
Relative Index
Analog input 1 to n (6975–1 to n) . . . . . . . . . 215
Discrete input 1 to n (6886–1 to n) . . . . . . . . 251
Relative Index (10737) . . . . . . . . . . . . . . . . . . . . 163
Relay output 1 to n simulation (0802–1 to n) . . . 325
342

Proline Promass 300 FOUNDATION Fieldbus

Relay output function
Relay output 1 to n (0804–1 to n) . . . . . . . . . 143
Reset access code (0024) . . . . . . . . . . . . . . . . . . . 47
Reset all totalizers (2806) . . . . . . . . . . . . . . . . . 275
Reset min/max values (6151) . . . . . . . . . . . . . . . 299
Resource State (10730) . . . . . . . . . . . . . . . . . . . 154
Response time
Current output 1 to n (0378–1 to n) . . . . . . . 118
Pulse/frequency/switch output 1 to n (0491–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 133
Response time part. filled pipe detect. (1859) . . . . 81
Response time status input
Status input 1 to n (1354–1 to n) . . . . . . . . . . 106
Restart (10800) . . . . . . . . . . . . . . . . . . . . . . . . . 157
Security identification (2718) . . . . . . . . . . . . . . . 172
Security type (2705) . . . . . . . . . . . . . . . . . . . . . . 172
Select antenna (2713) . . . . . . . . . . . . . . . . . . . . 175
Select gas type (6074) . . . . . . . . . . . . . . . . . . . . . 83
Select medium (6062) . . . . . . . . . . . . . . . . . . . . . 82
Separator (0101) . . . . . . . . . . . . . . . . . . . . . . . . . 26
Serial number (10798) . . . . . . . . . . . . . . . . . . . . 291
Service reset (10749) . . . . . . . . . . . . . . . . . . . . . 157
Set Fault State (10731) . . . . . . . . . . . . . . . . . . . 161
Shed Remote Cascade (10732) . . . . . . . . . . . . . . 162
Shed Remote Out (10733) . . . . . . . . . . . . . . . . . 162
Signal mode
Current input 1 to n (1610–1 to n) . . . . . . . . 102
Current output 1 to n (0377–1 to n) . . . . . . . 108
Pulse/frequency/switch output 1 to n (0490–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 123
Simulate En/Disable
Analog input 1 to n (6966–1 to n) . . . . . . . . . 194
Discrete input 1 to n (6877–1 to n) . . . . . . . . 244
Simulate Status
Analog input 1 to n (6967–1 to n) . . . . . . . . . 189
Discrete input 1 to n (6878–1 to n) . . . . . . . . 238
Simulate Value
Analog input 1 to n (6968–1 to n) . . . . . . . . . 191
Discrete input 1 to n (6879–1 to n) . . . . . . . . 241
Software option overview (0015) . . . . . . . . . . . . . 49
Software revision
I/O module 1 (0072) . . . . . . . . . . . . . . . . . . . 297
I/O module 2 (0072) . . . . . . . . . . . . . . . . . . . 297
I/O module 3 (0072) . . . . . . . . . . . . . . . . . . . 297
I/O module 4 (0072) . . . . . . . . . . . . . . . . . . . 297
Software revision (0072) . . . . . . . . . . . . . . 296, 298
Square expansion coefficient (1818) . . . . . . . . . . . 90
SSID name (2707) . . . . . . . . . . . . . . . . . . . . . . . 174
SSID name (2714) . . . . . . . . . . . . . . . . . . . . . . . 171
Static revision
Analog input 1 to n (6976–1 to n) . . . . . . . . . 215
Discrete input 1 to n (6887–1 to n) . . . . . . . . 251
Static Revision
Analog input 1 to n (6973–1 to n) . . . . . . . . . 180
Discrete input 1 to n (6884–1 to n) . . . . . . . . 229
Static Revision (10735) . . . . . . . . . . . . . . . . . . . 150
Static revision (10738) . . . . . . . . . . . . . . . . . . . . 163
Static Revision (11270) . . . . . . . . . . . . . . . . . . . 267
Static Revision (11371) . . . . . . . . . . . . . . . . . . . 258
Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

StatusAnalog input 1 to n (6906–1 to n) . . . . . . . . . 186
Analog input 1 to n (6923–1 to n) . . . . . . . . . 211
Analog input 1 to n (6964–1 to n) . . . . . . . . . 184
Discrete input 1 to n (6853–1 to n) . . . . . . . . 235
Discrete input 1 to n (6865–1 to n) . . . . . . . . 247
Discrete input 1 to n (6875–1 to n) . . . . . . . . 232
Status input simulation (1355) . . . . . . . . . . . . . . 320
Status Options
Analog input 1 to n (6971–1 to n) . . . . . . . . . 208
Discrete input 1 to n (6882–1 to n) . . . . . . . . 246
Status Options (11268) . . . . . . . . . . . . . . . . . . . 270
Status Options (11363) . . . . . . . . . . . . . . . . . . . 261
Strategy
Analog input 1 to n (6972–1 to n) . . . . . . . . . 181
Discrete input 1 to n (6883–1 to n) . . . . . . . . 230
Strategy (10734) . . . . . . . . . . . . . . . . . . . . . . . . 150
Strategy (11269) . . . . . . . . . . . . . . . . . . . . . . . . 267
Strategy (11370) . . . . . . . . . . . . . . . . . . . . . . . . 258
SubcodeAnalog input 1 to n (6918–1 to n) . . . . . . . . . 217
Analog input 1 to n (6928–1 to n) . . . . . . . . . 224
Analog input 1 to n (6933–1 to n) . . . . . . . . . 222
Analog input 1 to n (6943–1 to n) . . . . . . . . . 226
Analog input 1 to n (6949–1 to n) . . . . . . . . . 228
Discrete input 1 to n (6848–1 to n) . . . . . . . . 253
Discrete input 1 to n (6859–1 to n) . . . . . . . . 255
Subcode (10698) . . . . . . . . . . . . . . . . . . . . . . . . 165
Subcode (10743) . . . . . . . . . . . . . . . . . . . . . . . . 169
Subnet mask (7211) . . . . . . . . . . . . . . . . . . . . . . 178
Switch output function
Pulse/frequency/switch output 1 to n (0481–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 135
Switch output simulation 1 to n (0462–1 to n) . . 324
Switch status
Relay output 1 to n (0801–1 to n) . . . . . . . . . 148
Switch status 1 to n (0461–1 to n) . . . . . . . . . . . 141
Switch status 1 to n (0463–1 to n) . . . . . . . . . . . 325
Switch status 1 to n (0803–1 to n) . . . . . . . . . . . 326
Switch-off delay
Pulse/frequency/switch output 1 to n (0465–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Relay output 1 to n (0813–1 to n) . . . . . . . . . 146
Switch-off value
Pulse/frequency/switch output 1 to n (0464–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 139
Relay output 1 to n (0809–1 to n) . . . . . . . . . 146
Switch-on delay
Pulse/frequency/switch output 1 to n (0467–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 140
Relay output 1 to n (0814–1 to n) . . . . . . . . . 147
Switch-on value
Pulse/frequency/switch output 1 to n (0466–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 138
Relay output 1 to n (0810–1 to n) . . . . . . . . . 146
Tag Description
Analog input 1 to n (6974–1 to n) . . . . . . . . . 181
Discrete input 1 to n (6885–1 to n) . . . . . . . . 229
Tag Description (10736) . . . . . . . . . . . . . . . . . . 150
Endress+Hauser

Index

Tag Description (11271) . . . . . . . . . . . . . . . . . . 267
Tag Description (11372) . . . . . . . . . . . . . . . . . . 258
Target corrected volume flow (1893) . . . . . . . . . . 57
Target mass flow (1864) . . . . . . . . . . . . . . . . . . . 56
Target mode
Analog input 1 to n (6960–1 to n) . . . . . . . . . 181
Discrete input 1 to n (6873–1 to n) . . . . . . . . 230
Target mode (10728) . . . . . . . . . . . . . . . . . . . . . 151
Target mode (11267) . . . . . . . . . . . . . . . . . . . . . 268
Target mode (11369) . . . . . . . . . . . . . . . . . . . . . 259
Target volume flow (1895) . . . . . . . . . . . . . . . . . . 58
Temp. compensated dynamic viscosity (1872) . . . . 55
Temp. compensated kinematic viscosity (1863) . . . 55
Temperature (1853) . . . . . . . . . . . . . . . . . . . . . . 53
Temperature coefficient sound velocity (6181) . . . 84
Temperature damping (1822) . . . . . . . . . . . . . . . 75
Temperature factor (1871) . . . . . . . . . . . . . . . . . . 97
Temperature mode (6184) . . . . . . . . . . . . . . . . . . 86
Temperature offset (1870) . . . . . . . . . . . . . . . . . . 97
Temperature unit (0557) . . . . . . . . . . . . . . . . . . . 72
Terminal number
Current input 1 to n (1611–1 to n) . . . . . . . . 102
Current output 1 to n (0379–1 to n) . . . . . . . 107
Pulse/frequency/switch output 1 to n (0492–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 122
Relay output 1 to n (0812–1 to n) . . . . . . . . . 143
Status input 1 to n (1358–1 to n) . . . . . . . . . . 105
Time Stamp
Analog input 1 to n (6919–1 to n) . . . . . . . . . 216
Analog input 1 to n (6929–1 to n) . . . . . . . . . 224
Analog input 1 to n (6934–1 to n) . . . . . . . . . 222
Analog input 1 to n (6944–1 to n) . . . . . . . . . 226
Analog input 1 to n (6950–1 to n) . . . . . . . . . 227
Analog input 1 to n (6977–1 to n) . . . . . . . . . 215
Discrete input 1 to n (6849–1 to n) . . . . . . . . 254
Discrete input 1 to n (6860–1 to n) . . . . . . . . 255
Discrete input 1 to n (6888–1 to n) . . . . . . . . 251
Time Stamp (10699) . . . . . . . . . . . . . . . . . . . . . 165
Time Stamp (10739) . . . . . . . . . . . . . . . . . . . . . 163
Time Stamp (10744) . . . . . . . . . . . . . . . . . . . . . 168
Timestamp . . . . . . . . . 281, 282, 284, 285, 286, 287
Totalizer operation mode
Totalizer 1 to n (0908–1 to n) . . . . . . . . . . . . 277
Totalizer overflow 1 to n (0910–1 to n) . . . . . . . . 59
Totalizer value 1 to n (0911–1 to n) . . . . . . . . . . . 59
Transducer State
Discrete input 1 to n (6891–1 to n) . . . . . . . . 244
Transducer Status
Analog input 1 to n (6969–1 to n) . . . . . . . . . 192
Discrete input 1 to n (6880–1 to n) . . . . . . . . 241
Transducer Value
Analog input 1 to n (6970–1 to n) . . . . . . . . . 194
Discrete input 1 to n (6881–1 to n) . . . . . . . . 243
Unacknowledged
Analog input 1 to n (6914–1 to n) . . . . . . . . . 218
Analog input 1 to n (6920–1 to n) . . . . . . . . . 216
Analog input 1 to n (6930–1 to n) . . . . . . . . . 223
Analog input 1 to n (6935–1 to n) . . . . . . . . . 221
Analog input 1 to n (6945–1 to n) . . . . . . . . . 225
343

IndexAnalog input 1 to n (6951–1 to n) . . . . . . . . . 227
Analog input 1 to n (6978–1 to n) . . . . . . . . . 214
Discrete input 1 to n (6844–1 to n) . . . . . . . . 252
Discrete input 1 to n (6850–1 to n) . . . . . . . . 254
Discrete input 1 to n (6861–1 to n) . . . . . . . . 255
Discrete input 1 to n (6889–1 to n) . . . . . . . . 250
Unacknowledged (10694) . . . . . . . . . . . . . . . . . 166
Unacknowledged (10700) . . . . . . . . . . . . . . . . . 164
Unacknowledged (10740) . . . . . . . . . . . . . . . . . 162
Unacknowledged (10745) . . . . . . . . . . . . . . . . . 168
Unit totalizer 1 to n (0915–1 to n) . . . . . . . . . . . 276
Units index
Analog input 1 to n (6908–1 to n) . . . . . . . . . 198
Analog input 1 to n (6983–1 to n) . . . . . . . . . 195
Unreported
Analog input 1 to n (6915–1 to n) . . . . . . . . . 218
Discrete input 1 to n (6845–1 to n) . . . . . . . . 253
Unreported (10695) . . . . . . . . . . . . . . . . . . . . . . 166
Update State
Analog input 1 to n (6979–1 to n) . . . . . . . . . 215
Discrete input 1 to n (6890–1 to n) . . . . . . . . 250
Update State (10741) . . . . . . . . . . . . . . . . . . . . . 163
User name (2715) . . . . . . . . . . . . . . . . . . . . . . . 172
Value
Analog input 1 to n (6907–1 to n) . . . . . . . . . 189
Analog input 1 to n (6921–1 to n) . . . . . . . . . 217
Analog input 1 to n (6924–1 to n) . . . . . . . . . 214
Analog input 1 to n (6965–1 to n) . . . . . . . . . 186
Discrete input 1 to n (6854–1 to n) . . . . . . . . 238
Discrete input 1 to n (6856–1 to n) . . . . . . . . 254
Discrete input 1 to n (6866–1 to n) . . . . . . . . 249
Discrete input 1 to n (6876–1 to n) . . . . . . . . 235
Value (10701) . . . . . . . . . . . . . . . . . . . . . . . . . . 166
Value 1 display (0107) . . . . . . . . . . . . . . . . . . . . . 18
Value 2 display (0108) . . . . . . . . . . . . . . . . . . . . . 20
Value 3 display (0110) . . . . . . . . . . . . . . . . . . . . . 21
Value 4 display (0109) . . . . . . . . . . . . . . . . . . . . . 23
Value current input 1 to n (1609–1 to n) . . . . . . 322
Value current output 1 to n (0355–1 to n) . . . . . 322
Value per pulse
Pulse/frequency/switch output 1 to n (0455–
1 to n) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 125
Value status input
Status input 1 to n (1353–1 to n) . . . . . . . . . . 105
Value status input 1 to n (1353–1 to n) . . . . . . 61
Volume flow (1847) . . . . . . . . . . . . . . . . . . . . . . . 52
Volume flow factor (1846) . . . . . . . . . . . . . . . . . . 94
Volume flow offset (1841) . . . . . . . . . . . . . . . . . . 94
Volume flow unit (0553) . . . . . . . . . . . . . . . . . . . 68
Volume unit (0563) . . . . . . . . . . . . . . . . . . . . . . . 69
Web server functionality (7222) . . . . . . . . . . . . . 179
Web server language (7221) . . . . . . . . . . . . . . . 177
WLAN (2702) . . . . . . . . . . . . . . . . . . . . . . . . . . 171
WLAN IP address (2711) . . . . . . . . . . . . . . . . . . 173
WLAN MAC address (2703) . . . . . . . . . . . . . . . . 173
WLAN mode (2717) . . . . . . . . . . . . . . . . . . . . . 171
WLAN passphrase (2706) . . . . . . . . . . . . . . . . . 174
WLAN password (2716) . . . . . . . . . . . . . . . . . . . 172
WLAN subnet mask (2709) . . . . . . . . . . . . . . . . 173
344

Proline Promass 300 FOUNDATION Fieldbus

Write Lock (10747) . . . . . . . . . . . . . . . . . . . . . . 164
Write Priority (10748) . . . . . . . . . . . . . . . . . . . . 167
Zero point (6195) . . . . . . . . . . . . . . . . . . . . . . . . . 98
Zero point adjustment control (6196) . . . . . . . . . . 92
Direct access (Parameter) . . . . . . . . . . . . . . . . . . . . . 11
Disabled (Parameter) . . . . . . . . . . . . . . . . 167, 218, 252
Discrete input 1 to n (Submenu) . . . . . . . . . . . . . . . . 229
Discrete inputs (Submenu) . . . . . . . . . . . . . . . . . . . . 228
Discrete Limit (Parameter) . . . . . . . . . . . . . . . . . . . . 256
Discrete outputs (Submenu) . . . . . . . . . . . . . . . . . . . 266
Discrete Priority (Parameter) . . . . . . . . . . . . . . . . . . 257
Discrete Value (Parameter) . . . . . . . . . . . . . . . 169, 256
Display (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Display channel 1 (Submenu) . . . . . . . . . . . . . . . . . 315
Display channel 2 (Submenu) . . . . . . . . . . . . . . . . . 317
Display channel 3 (Submenu) . . . . . . . . . . . . . . . . . 317
Display channel 4 (Submenu) . . . . . . . . . . . . . . . . . 318
Display damping (Parameter) . . . . . . . . . . . . . . . . . . 25
Display interval (Parameter) . . . . . . . . . . . . . . . . . . . 24
Display language (Parameter) . . . . . . . . . . . . . . . . . . 15
Display module (Submenu) . . . . . . . . . . . . . . . . . . . 298
Document
Explanation of the structure of a parameter
description . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Function . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Structure . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Using the document . . . . . . . . . . . . . . . . . . . . . . . . 4
Document function . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Dynamic viscosity (Parameter) . . . . . . . . . . . . . . . . . . 54

EElectronic temperature (Submenu) . . . . . . . . . . . . . . 300
ENP version (Parameter) . . . . . . . . . . . . . . . . . . . . . 294
Enter access code (Parameter) . . . . . . . . . . . . . . . . . . 13
Entire logging duration (Parameter) . . . . . . . . . . . . 315
EU at 0% (Parameter) . . . . . . . . . . . . . . . . . . . 195, 198
EU at 100% (Parameter) . . . . . . . . . . . . . . . . . 194, 198
Event list (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 288
Event logbook (Submenu) . . . . . . . . . . . . . . . . . . . . 287
Extended order code 1 (Parameter) . . . . . . . . . . . . . 293
Extended order code 2 (Parameter) . . . . . . . . . . . . . 293
External compensation (Submenu) . . . . . . . . . . . . . . . 84
External pressure (Parameter) . . . . . . . . . . . . . . . . . . 85
External reference density (Parameter) . . . . . . . . . . . 88
External temperature (Parameter) . . . . . . . . . . . . . . . 86

F

Factory settings . . . . . . . . . . . . . . . . . . . . . . . . . . . . 328
SI units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 328
US units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 330
Failure current (Parameter) . . . . . . . . . . . . . . . . . . . 120
Failure frequency (Parameter) . . . . . . . . . . . . . . . . . 135
Failure mode (Parameter) 103, 119, 127, 134, 141,147,
279
Failure value (Parameter) . . . . . . . . . . . . . . . . . . . . 104
Fault State (Parameter) . . . . . . . . . . . . . . . . . . . . . . 161
Fault State Status (Parameter) . . . . . . . . . . . . . 265, 274
Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Fault State Time (Parameter) . . . . . . . . . . . . . . 262, 271
Fault State Val 1 (Parameter) . . . . . . . . . . . . . . . . . 263
Fault State Val 2 (Parameter) . . . . . . . . . . . . . . . . . 263
Fault State Val 3 (Parameter) . . . . . . . . . . . . . . . . . 263
Fault State Val 4 (Parameter) . . . . . . . . . . . . . . . . . 264
Fault State Val 5 (Parameter) . . . . . . . . . . . . . . . . . 264
Fault State Val 6 (Parameter) . . . . . . . . . . . . . . . . . 264
Fault State Val 7 (Parameter) . . . . . . . . . . . . . . . . . 265
Fault State Val 8 (Parameter) . . . . . . . . . . . . . . . . . 265
Fault State Value Discrete 1 (Parameter) . . . . . . . . . 272
Fault State Value Discrete 2 (Parameter) . . . . . . . . . 272
Fault State Value Discrete 3 (Parameter) . . . . . . . . . 272
Fault State Value Discrete 4 (Parameter) . . . . . . . . . 273
Fault State Value Discrete 5 (Parameter) . . . . . . . . . 273
Fault State Value Discrete 6 (Parameter) . . . . . . . . . 273
Fault State Value Discrete 7 (Parameter) . . . . . . . . . 274
Fault State Value Discrete 8 (Parameter) . . . . . . . . . 274
Feature Selection (Parameter) . . . . . . . . . . . . . . . . . 158
Features (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 158
Filter options (Parameter) . . . . . . . . . . . . . . . . 287, 288
Firmware version (Parameter) . . . . . . . . . . . . . . . . . 292
Fixed current (Parameter) . . . . . . . . . . . . . . . . . . . . 110
Fixed reference density (Parameter) . . . . . . . . . . . . . . 88
Float Value (Parameter) . . . . . . . . . . 223, 225, 226, 228
Flow damping (Parameter) . . . . . . . . . . . . . . . . . . . . 74
Flow override (Parameter) . . . . . . . . . . . . . . . . . . . . . 76
Format display (Parameter) . . . . . . . . . . . . . . . . . . . . 15
Free Space (Parameter) . . . . . . . . . . . . . . . . . . . . . . 160
Free Time (Parameter) . . . . . . . . . . . . . . . . . . . . . . 160
Frequency output simulation 1 to n (Parameter) . . . . 323
Frequency value 1 to n (Parameter) . . . . . . . . . . . . . 323
Function
see Parameter

G

Gateway IP address (Parameter) . . . . . . . . . . . . . . . 176
Grant (Parameter) . . . . . . . . . . . . . . . . . . 156, 207, 245

H

Hard Types (Parameter) . . . . . . . . . . . . . . . . . . . . . 157
Hardware revision (Parameter) . . . . . . . . . . . . . . . . 292
Header (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . 25
Header text (Parameter) . . . . . . . . . . . . . . . . . . . . . . 26
Heartbeat (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 318
High High Limit (Parameter) . . . . . . . . . . . . . . . . . . 219
High High Priority (Parameter) . . . . . . . . . . . . . . . . 219
High Limit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 220
High Priority (Parameter) . . . . . . . . . . . . . . . . . . . . 220
High value partial filled pipe detection (Parameter) . . 80

I

I/O configuration (Submenu) . . . . . . . . . . . . . . . . . . . 99
I/O module 1 (Submenu) . . . . . . . . . . . . . . . . . . . . . 296
I/O module 1 terminal numbers (Parameter) . . 296, 297
I/O module 1 to n information (Parameter) . . . . . . . . 99
I/O module 1 to n terminal numbers (Parameter) . . . . 99
I/O module 1 to n type (Parameter) . . . . . . . . . . . . . 100
I/O module 2 (Submenu) . . . . . . . . . . . . . . . . . . . . . 297
I/O Options (Parameter) . . . . . . . . . . . . . . . . . 207, 245
Endress+Hauser

Index

Input (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 101
Input signal level (Parameter) . . . . . . . . . . . . . . . . . 321
Input values (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 60
Installation angle pitch (Parameter) . . . . . . . . . . . . . . 91
Installation angle roll (Parameter) . . . . . . . . . . . . . . . 91
Installation direction (Parameter) . . . . . . . . . . . . . . . 90
Invert output signal (Parameter) . . . . . . . . . . . . . . . 142
IP address (Parameter) . . . . . . . . . . . . . . . . . . . . . . 178
IP address domain name server (Parameter) . . . . . . . 176
ITK Version (Parameter) . . . . . . . . . . . . . . . . . 170, 292

K

Kinematic viscosity (Parameter) . . . . . . . . . . . . . . . . . 54

L

Last backup (Parameter) . . . . . . . . . . . . . . . . . . . . . . 28
Limit Notify (Parameter) . . . . . . . . . . . . . . . . . . . . . 161
Linear expansion coefficient (Parameter) . . . . . . . . . . 89
Linearization Type (Parameter) . . . . . . . . . . . . . . . . 209
Locking status (Parameter) . . . . . . . . . . . . . . . . . . . . 12
Logging delay (Parameter) . . . . . . . . . . . . . . . . . . . 314
Logging interval (Parameter) . . . . . . . . . . . . . . . . . . 312
Login page (Parameter) . . . . . . . . . . . . . . . . . . . . . . 179
Low Cutoff (Parameter) . . . . . . . . . . . . . . . . . . . . . . 211
Low flow cut off (Submenu) . . . . . . . . . . . . . . . . . . . . 76
Low Limit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 221
Low Low Limit (Parameter) . . . . . . . . . . . . . . . . . . . 221
Low Low Priority (Parameter) . . . . . . . . . . . . . . . . . 221
Low Priority (Parameter) . . . . . . . . . . . . . . . . . . . . . 220
Low value partial filled pipe detection (Parameter) . . . 80

M

MAC address (Parameter) . . . . . . . . . . . . . . . . . . . . 177
Manufacturer Id (Parameter) . . . . . . . . . . . . . . . . . . 155
Mass flow (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 52
Mass flow factor (Parameter) . . . . . . . . . . . . . . . . . . 94
Mass flow offset (Parameter) . . . . . . . . . . . . . . . . . . . 93
Mass flow unit (Parameter) . . . . . . . . . . . . . . . . . . . . 66
Mass unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 67
Max Notify (Parameter) . . . . . . . . . . . . . . . . . . . . . 161
Maximum damping partial filled pipe det.
(Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 81
Maximum frequency value (Parameter) . . . . . . . . . . 130
Maximum value (Parameter) 300, 301, 302, 303,
304,
305,
306,
307,
308,
309
Measured current 1 to n (Parameter) . . . . . . . . . 61, 121
Measured values (Submenu) . . . . . . . . . . . . . . . . . . . 51
Measured values 1 to n (Parameter) . . . . . . . . . . . . . . 61
Measurement mode (Submenu) . . . . . . . . . . . . . . . . . 82
Measuring mode (Parameter) . . . . . . . . . 113, 126, 131
Measuring value at maximum frequency (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Measuring value at minimum frequency (Parameter). . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 130
Medium temperature (Submenu) . . . . . . . . . . . . . . . 301
Memory Size (Parameter) . . . . . . . . . . . . . . . . . . . . 159
Min/max values (Submenu) . . . . . . . . . . . . . . . . . . . 299
Minimum frequency value (Parameter) . . . . . . . . . . 129

345

IndexMinimum value (Parameter) 300, 301, 302, 303,
304,
305,
306,
307,
308,
309
Minumum Cycle Time (Parameter) . . . . . . . . . . . . . 159
Multi-frequency activation (Parameter) . . . . . . . . . . . 82
Multiple analog output (Submenu) . . . . . . . . . . . . . . 257
Multiple discrete output (Submenu) . . . . . . . . . . . . . 266

N

Network security (Parameter) . . . . . . . . . . . . . . . . . 172
Nominal diameter (Parameter) . . . . . . . . . . . . . . . . . 98
Nonvolatile Cycle Time (Parameter) . . . . . . . . . . . . . 159
Normal mode (Parameter) . . . . 153, 183, 232, 260, 269

O

Off value low flow cutoff (Parameter) . . . . . . . . . . . . . 77
On value low flow cutoff (Parameter) . . . . . . . . . . . . . 77
Operating mode (Parameter) . . . . . . . . . . . . . . . . . . 123
Operating time (Parameter) . . . . . . . . . . . . . 28, 47, 283
Operating time from restart (Parameter) . . . . . . . . . 282
Order code (Parameter) . . . . . . . . . . . . . . . . . . . . . . 293
Oscillation amplitude (Submenu) . . . . . . . . . . . . . . . 305
Oscillation damping (Submenu) . . . . . . . . . . . . . . . . 306
Oscillation frequency (Submenu) . . . . . . . . . . . . . . . 303
Output (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Output current 1 to n (Parameter) . . . . . . . . . . . . . . 120
Output frequency 1 to n (Parameter) . . . . . . . . . . . . 135
Output State (Parameter) . . . . . . . . . . . . . . . . . . . . 245
Output values (Submenu) . . . . . . . . . . . . . . . . . . . . . 62

P

Parameter
Structure of a parameter description . . . . . . . . . . . . 6
Partially filled pipe detection (Submenu) . . . . . . . . . . 79
Permitted mode (Parameter) . . 152, 182, 231, 260, 269
Powerless relay status (Parameter) . . . . . . . . . . . . . 148
Preset value 1 to n (Parameter) . . . . . . . . . . . . . . . . 278
Pressure compensation (Parameter) . . . . . . . . . . . . . . 84
Pressure shock suppression (Parameter) . . . . . . . . . . 78
Pressure unit (Parameter) . . . . . . . . . . . . . . . . . . . . . 73
Pressure value (Parameter) . . . . . . . . . . . . . . . . . 54, 85
Previous diagnostics (Parameter) . . . . . . . . . . . . . . . 282
Process parameters (Submenu) . . . . . . . . . . . . . . . . . 74
Process Value Filter Time (Parameter) . . . . . . . 214, 247
Process variable adjustment (Submenu) . . . . . . . . . . . 93
Process variable value (Parameter) . . . . . . . . . . . . . . 320
Process variables (Submenu) . . . . . . . . . . . . . . . . . . . 51
Progress (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 93
Pulse output 1 to n (Parameter) . . . . . . . . . . . . . . . . 128
Pulse output simulation 1 to n (Parameter) . . . . . . . 324
Pulse value 1 to n (Parameter) . . . . . . . . . . . . . . . . . 324
Pulse width (Parameter) . . . . . . . . . . . . . . . . . . . . . 126
Pulse/frequency/switch output 1 to n (Submenu) 63,
121

R

Received signal strength (Parameter) . . . . . . . . . . . . 175
Reference density (Parameter) . . . . . . . . . . . . . . . . . . 53
Reference density factor (Parameter) . . . . . . . . . . . . . 96
Reference density offset (Parameter) . . . . . . . . . . . . . 96
Reference density unit (Parameter) . . . . . . . . . . . . . . 71
346

Proline Promass 300 FOUNDATION Fieldbus

Reference sound velocity (Parameter) . . . . . . . . . . . . 83
Reference temperature (Parameter) . . . . . . . . . . . . . . 89
Relative Index (Parameter) . . . . . . . . . . . . 163, 215, 251
Relay output 1 to n (Submenu) . . . . . . . . . . . . . . 65, 142
Relay output 1 to n simulation (Parameter) . . . . . . . 325
Relay output function (Parameter) . . . . . . . . . . . . . . 143
Reset access code (Parameter) . . . . . . . . . . . . . . . . . . 47
Reset access code (Submenu) . . . . . . . . . . . . . . . . . . . 47
Reset all totalizers (Parameter) . . . . . . . . . . . . . . . . 275
Reset min/max values (Parameter) . . . . . . . . . . . . . 299
Resource block (Submenu) . . . . . . . . . . . . . . . . . . . . 149
Resource State (Parameter) . . . . . . . . . . . . . . . . . . . 154
Response time (Parameter) . . . . . . . . . . . . . . . 118, 133
Response time part. filled pipe detect. (Parameter) . . . 81
Response time status input (Parameter) . . . . . . . . . . 106
Restart (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 157

S

Security identification (Parameter) . . . . . . . . . . . . . . 172
Security type (Parameter) . . . . . . . . . . . . . . . . . . . . 172
Select antenna (Parameter) . . . . . . . . . . . . . . . . . . . 175
Select gas type (Parameter) . . . . . . . . . . . . . . . . . . . . 83
Select medium (Parameter) . . . . . . . . . . . . . . . . . . . . 82
Sensor (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Sensor adjustment (Submenu) . . . . . . . . . . . . . . . . . . 90
Sensor electronic module (ISEM) (Submenu) . . . . . . 295
Separator (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . 26
Serial number (Parameter) . . . . . . . . . . . . . . . . . . . 291
Service reset (Parameter) . . . . . . . . . . . . . . . . . . . . . 157
Set Fault State (Parameter) . . . . . . . . . . . . . . . . . . . 161
Shed Remote Cascade (Parameter) . . . . . . . . . . . . . . 162
Shed Remote Out (Parameter) . . . . . . . . . . . . . . . . . 162
Signal asymmetry (Submenu) . . . . . . . . . . . . . . . . . 308
Signal mode (Parameter) . . . . . . . . . . . . . 102, 108, 123
Simulate En/Disable (Parameter) . . . . . . . . . . . 194, 244
Simulate Status (Parameter) . . . . . . . . . . . . . . 189, 238
Simulate Value (Parameter) . . . . . . . . . . . . . . . 191, 241
Simulation (Submenu) . . . . . . . . . . . . . . . . . . . . . . . 318
Software option overview (Parameter) . . . . . . . . . . . . 49
Software revision (Parameter) . . . . . . . . . 296, 297, 298
Square expansion coefficient (Parameter) . . . . . . . . . 90
SSID name (Parameter) . . . . . . . . . . . . . . . . . . 171, 174
Static revision (Parameter) . . . . . . . . . . . . 163, 215, 251
Static Revision (Parameter) . . . 150, 180, 229, 258, 267
Status (Parameter) . . . . . . 184, 186, 211, 232, 235, 247
Status input 1 to n (Submenu) . . . . . . . . . . . . . . . . . 104
Status input simulation (Parameter) . . . . . . . . . . . . 320
Status Options (Parameter) . . . . . . . 208, 246, 261, 270
Strategy (Parameter) . . . . . . . . 150, 181, 230, 258, 267
Subcode (Parameter) 165, 169, 217, 222, 224, 226,
228,
253,
255
Submenu
Administration . . . . . . . . . . . . . . . . . . . . . . . . . . . 45
Analog input 1 to n . . . . . . . . . . . . . . . . . . . . . . 180
Analog inputs . . . . . . . . . . . . . . . . . . . . . . . . . . 179
Analog outputs . . . . . . . . . . . . . . . . . . . . . . . . . 257
Application . . . . . . . . . . . . . . . . . . . . . . . . . . . . 275
Calculated values . . . . . . . . . . . . . . . . . . . . . . . . . 87
Calibration . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 97
Endress+Hauser

Proline Promass 300 FOUNDATION Fieldbus

Carrier pipe temperature . . . . . . . . . . . . . . . . . . 302
Communication . . . . . . . . . . . . . . . . . . . . . . . . . 148
Concentration . . . . . . . . . . . . . . . . . . . . . . . . . . 280
Configuration backup . . . . . . . . . . . . . . . . . . . . . . 27
Corrected volume flow calculation . . . . . . . . . . . . . 87
Current input 1 to n . . . . . . . . . . . . . . . . . . . 60, 101
Current output 1 to n . . . . . . . . . . . . . . . . . . . . . 107
Data logging . . . . . . . . . . . . . . . . . . . . . . . . . . . 309
Device information . . . . . . . . . . . . . . . . . . . . . . . 289
Diagnostic behavior . . . . . . . . . . . . . . . . . . . . . . . 31
Diagnostic handling . . . . . . . . . . . . . . . . . . . . . . . 30
Diagnostic list . . . . . . . . . . . . . . . . . . . . . . . . . . 283
Diagnostics . . . . . . . . . . . . . . . . . . . . . . . . . . . . 280
Discrete input 1 to n . . . . . . . . . . . . . . . . . . . . . . 229
Discrete inputs . . . . . . . . . . . . . . . . . . . . . . . . . . 228
Discrete outputs . . . . . . . . . . . . . . . . . . . . . . . . . 266
Display . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 14
Display channel 1 . . . . . . . . . . . . . . . . . . . . . . . . 315
Display channel 2 . . . . . . . . . . . . . . . . . . . . . . . . 317
Display channel 3 . . . . . . . . . . . . . . . . . . . . . . . . 317
Display channel 4 . . . . . . . . . . . . . . . . . . . . . . . . 318
Display module . . . . . . . . . . . . . . . . . . . . . . . . . . 298
Electronic temperature . . . . . . . . . . . . . . . . . . . . 300
Event list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 288
Event logbook . . . . . . . . . . . . . . . . . . . . . . . . . . 287
External compensation . . . . . . . . . . . . . . . . . . . . . 84
Heartbeat . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 318
I/O configuration . . . . . . . . . . . . . . . . . . . . . . . . . 99
I/O module 1 . . . . . . . . . . . . . . . . . . . . . . . . . . . 296
I/O module 2 . . . . . . . . . . . . . . . . . . . . . . . . . . . 297
Input . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 101
Input values . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 60
Low flow cut off . . . . . . . . . . . . . . . . . . . . . . . . . . 76
Measured values . . . . . . . . . . . . . . . . . . . . . . . . . 51
Measurement mode . . . . . . . . . . . . . . . . . . . . . . . 82
Medium temperature . . . . . . . . . . . . . . . . . . . . . 301
Min/max values . . . . . . . . . . . . . . . . . . . . . . . . . 299
Multiple analog output . . . . . . . . . . . . . . . . . . . . 257
Multiple discrete output . . . . . . . . . . . . . . . . . . . 266
Oscillation amplitude . . . . . . . . . . . . . . . . . . . . . 305
Oscillation damping . . . . . . . . . . . . . . . . . . . . . . 306
Oscillation frequency . . . . . . . . . . . . . . . . . . . . . 303
Output . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 106
Output values . . . . . . . . . . . . . . . . . . . . . . . . . . . . 62
Partially filled pipe detection . . . . . . . . . . . . . . . . . 79
Process parameters . . . . . . . . . . . . . . . . . . . . . . . 74
Process variable adjustment . . . . . . . . . . . . . . . . . 93
Process variables . . . . . . . . . . . . . . . . . . . . . . . . . 51
Pulse/frequency/switch output 1 to n . . . . . . 63, 121
Relay output 1 to n . . . . . . . . . . . . . . . . . . . . 65, 142
Reset access code . . . . . . . . . . . . . . . . . . . . . . . . . 47
Resource block . . . . . . . . . . . . . . . . . . . . . . . . . . 149
Sensor . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 50
Sensor adjustment . . . . . . . . . . . . . . . . . . . . . . . . 90
Sensor electronic module (ISEM) . . . . . . . . . . . . . 295
Signal asymmetry . . . . . . . . . . . . . . . . . . . . . . . . 308
Simulation . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 318
Status input 1 to n . . . . . . . . . . . . . . . . . . . . . . . 104
System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 13
Endress+Hauser

Index

System units . . . . . . . . . . . . . . . . . . . . . . . . . . . . 66
Torsion oscillation amplitude . . . . . . . . . . . . . . . 305
Torsion oscillation damping . . . . . . . . . . . . . . . . 307
Torsion oscillation frequency . . . . . . . . . . . . . . . 304
Torsion signal asymmetry . . . . . . . . . . . . . . . . . . 308
Totalizer . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 59
Totalizer 1 to n . . . . . . . . . . . . . . . . . . . . . . . . . . 275
Value current output 1 to n . . . . . . . . . . . . . . . . . . 62
Value status input 1 to n . . . . . . . . . . . . . . . . . . . . 61
Viscosity . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 279
Web server . . . . . . . . . . . . . . . . . . . . . . . . . . . . 176
WLAN settings . . . . . . . . . . . . . . . . . . . . . . . . . . 170
Zero point adjustment . . . . . . . . . . . . . . . . . . . . . 92
Subnet mask (Parameter) . . . . . . . . . . . . . . . . . . . . 178
Switch output function (Parameter) . . . . . . . . . . . . . 135
Switch output simulation 1 to n (Parameter) . . . . . . 324
Switch status (Parameter) . . . . . . . . . . . . . . . . . . . . 148
Switch status 1 to n (Parameter) . . . . . . . 141, 325, 326
Switch-off delay (Parameter) . . . . . . . . . . . . . . 140, 146
Switch-off value (Parameter) . . . . . . . . . . . . . . 139, 146
Switch-on delay (Parameter) . . . . . . . . . . . . . . 140, 147
Switch-on value (Parameter) . . . . . . . . . . . . . . 138, 146
System (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . . 13
System units (Submenu) . . . . . . . . . . . . . . . . . . . . . . 66

T

Tag Description (Parameter) . . . 150, 181, 229, 258, 267
Target corrected volume flow (Parameter) . . . . . . . . . 57
Target group . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4
Target mass flow (Parameter) . . . . . . . . . . . . . . . . . . 56
Target mode (Parameter) . . . . . 151, 181, 230, 259, 268
Target volume flow (Parameter) . . . . . . . . . . . . . . . . 58
Temp. compensated dynamic viscosity (Parameter) . . 55
Temp. compensated kinematic viscosity (Parameter) . 55
Temperature (Parameter) . . . . . . . . . . . . . . . . . . . . . 53
Temperature coefficient sound velocity (Parameter) . . 84
Temperature damping (Parameter) . . . . . . . . . . . . . . 75
Temperature factor (Parameter) . . . . . . . . . . . . . . . . 97
Temperature mode (Parameter) . . . . . . . . . . . . . . . . . 86
Temperature offset (Parameter) . . . . . . . . . . . . . . . . 97
Temperature unit (Parameter) . . . . . . . . . . . . . . . . . . 72
Terminal number (Parameter) . 102, 105, 107, 122, 143
Time Stamp (Parameter) 163, 165, 168, 215, 216,
222,
224,
226,
227,
251,
254,
255
Timestamp (Parameter) . . 281, 282, 284, 285, 286, 287
Torsion oscillation amplitude (Submenu) . . . . . . . . . 305
Torsion oscillation damping (Submenu) . . . . . . . . . . 307
Torsion oscillation frequency (Submenu) . . . . . . . . . 304
Torsion signal asymmetry (Submenu) . . . . . . . . . . . 308
Totalizer (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . . 59
Totalizer 1 to n (Submenu) . . . . . . . . . . . . . . . . . . . 275
Totalizer operation mode (Parameter) . . . . . . . . . . . 277
Totalizer overflow 1 to n (Parameter) . . . . . . . . . . . . . 59
Totalizer value 1 to n (Parameter) . . . . . . . . . . . . . . . 59
Transducer State (Parameter) . . . . . . . . . . . . . . . . . 244
Transducer Status (Parameter) . . . . . . . . . . . . . 192, 241
Transducer Value (Parameter) . . . . . . . . . . . . . 194, 243347

Index

Proline Promass 300 FOUNDATION Fieldbus

U

Unacknowledged (Parameter) 162, 164, 166, 168,
214, 216, 218, 221, 223, 225, 227, 250, 252, 254, 255
Unit totalizer 1 to n (Parameter) . . . . . . . . . . . . . . . 276
Units index (Parameter) . . . . . . . . . . . . . . . . . 195, 198
Unreported (Parameter) . . . . . . . . . . . . . 166, 218, 253
Update State (Parameter) . . . . . . . . . . . . 163, 215, 250
User name (Parameter) . . . . . . . . . . . . . . . . . . . . . . 172

V

Value (Parameter) 166, 186, 189, 214, 217, 235,
238,
249,
254
Value 1 display (Parameter) . . . . . . . . . . . . . . . . . . . 18
Value 2 display (Parameter) . . . . . . . . . . . . . . . . . . . 20
Value 3 display (Parameter) . . . . . . . . . . . . . . . . . . . 21
Value 4 display (Parameter) . . . . . . . . . . . . . . . . . . . 23
Value current input 1 to n (Parameter) . . . . . . . . . . . 322
Value current output 1 to n (Parameter) . . . . . . . . . . 322
Value current output 1 to n (Submenu) . . . . . . . . . . . 62
Value per pulse (Parameter) . . . . . . . . . . . . . . . . . . 125
Value status input (Parameter) . . . . . . . . . . . . . 61, 105
Value status input 1 to n (Submenu) . . . . . . . . . . . . . 61
Viscosity (Submenu) . . . . . . . . . . . . . . . . . . . . . . . . 279
Volume flow (Parameter) . . . . . . . . . . . . . . . . . . . . . 52
Volume flow factor (Parameter) . . . . . . . . . . . . . . . . . 94
Volume flow offset (Parameter) . . . . . . . . . . . . . . . . . 94
Volume flow unit (Parameter) . . . . . . . . . . . . . . . . . . 68
Volume unit (Parameter) . . . . . . . . . . . . . . . . . . . . . . 69

W

Web server (Submenu) . . . . . . . . . . . . . . . . . . . . . . 176
Web server functionality (Parameter) . . . . . . . . . . . 179
Web server language (Parameter) . . . . . . . . . . . . . . 177
WizardDefine access code . . . . . . . . . . . . . . . . . . . . . . . . 46
WLAN (Parameter) . . . . . . . . . . . . . . . . . . . . . . . . . 171
WLAN IP address (Parameter) . . . . . . . . . . . . . . . . . 173
WLAN MAC address (Parameter) . . . . . . . . . . . . . . 173
WLAN mode (Parameter) . . . . . . . . . . . . . . . . . . . . 171
WLAN passphrase (Parameter) . . . . . . . . . . . . . . . . 174
WLAN password (Parameter) . . . . . . . . . . . . . . . . . 172
WLAN settings (Submenu) . . . . . . . . . . . . . . . . . . . 170
WLAN subnet mask (Parameter) . . . . . . . . . . . . . . . 173
Write Lock (Parameter) . . . . . . . . . . . . . . . . . . . . . . 164
Write Priority (Parameter) . . . . . . . . . . . . . . . . . . . . 167

Z

Zero point (Parameter) . . . . . . . . . . . . . . . . . . . . . . . 98
Zero point adjustment (Submenu) . . . . . . . . . . . . . . . 92
Zero point adjustment control (Parameter) . . . . . . . . . 92

348

Endress+Hauser

www.addresses.endress.com