YMC USER'S MANUAL

Column Care and Use Instructions

Alcyon SFC CSP Amylose-C, Amylose-C Neo, Cellulose-C
Supercritical Fluid Chromatography / Optical Isomer Separation Column (Coated type)1.

Introduction
Thank you for purchasing an Alcyon SFC Column for Supercritical Fluid Chromatography (SFC).
The Alcyon SFC CSP Amylose-C/Amylose-C Neo/Cellulose-C column is designed for separating optical isomers. The chiralselector, a polysaccharide derivative, is coated onto silica gel. With its superior separation and selectivity, a Alcyon SFC CSP
Amylose-C/Amylose-C Neo/Cellulose-C column is suitable for separating a wide range of chiral compounds.
Alcyon SFC Columns, which are manufactured under highly controlled conditions, must pass a series of strict tests before being
accepted for shipment (Please refer to the column inspection report). To ensure optimal performance and durability of the column,
please read these instructions carefully before using this column.

2.

Specifications
Item

Alcyon SFC CSP Amylose-C
Alcyon SFC CSP Amylose-C Neo

Particle size
Chiral selector

3, 5 µm
Amylose tris(3,5-dimethylphenylcarbamate)Type

Cellulose tris(3,5-dimethylphenylcarbamate)

Coated type

Shipping solvent 1n-hexane/2-propanol (90/10)

Usable temperature range

0 – 40℃
2.1 , 3.0 , 4.6 mmI.D.
10 , 20 mmI.D.

Pressure limit 2
Recommended flow rate 3Alcyon SFC CSP Cellulose-C

2.1 mmI.D.
3.0 mmI.D.
4.6 mm I.D.

： 0.2～0.6 mL/min： 0.4～1.2 mL/min： 1.0～3.0 mL/min： 30 MPa： 20 MPa
10 mm I.D.
20 mm I.D.

： 5～15 mL/min： 20～60 mL/min

1: If you intend to store the column for a long time, replace the mobile phase in the column with shipping solvent or 100%2-propanol.
2: Avoid using a column repeatedly near the pressure limit or abrupt change in the pressure in order to prevent from shortening thecolumn lifetime.
3: Adjust flow rate according to the recommendation in the table above to obtain the optimum results under the application.
2, 3: Pressure changes depending on column length, temperature, types of organic solvent etc. If pressure exceeds the upper limit,

reduce flow rate to below the lower rate of recommended range.

3.

Precautions for use・ The column endfitting is Waters style connection.
・ Tubing must have flat ends and must bottom out in the column endfitting. Tubing must be connected to the column correctly to
avoid creating a void between the column frit and tubing, which can cause a leak and result in poor column performance (e.g.
peak tailing, loss of theoretical plate number).
・ The correct direction of the solvent flow is indicated by an arrow on the column identification label.
・ Do not disconnect a column from the SFC system before the pressure drops to zero.

(UM210617CE) 1/2

YMC USER'S MANUAL

4.

Mobile phase and sample solvent
・ When a target compound is ionic, addition of additives listed below can improve peak shape and/or separation reproducibility.
High concentrations of additives can result in reducing column lifetime. Add/reduce the additives according to the notes in thetable. Additive concentration below is concentration for the entire mobile phase.
・ When possible, the sample should be dissolved in the same modifier as the mobile phase. Using a stronger solvent than mobilephase for sample dissolution might result in distorted peak symmetry and degraded resolution.
・ In order to avoid blockage which can cause pressure increase, the sample solution should be filtered through a membrane filter(0.2 μm or smaller porosity).
【Recommended solvents】
Acidic compounds
Modifiers
Additives

Basic compounds

alcohols (methanol, ethanol, 2-propanol, etc.)
0.1% (Upper limit 0.5%)
trifluoroacetic acid (TFA),
acetic acid, formic acid, etc.

Composition ratio

5.

Non-Ionic compounds

0.1% (Upper limit 0.5%)
diethylamine (DEA), butylamine,ethanolamine, etc.

NoneCO2/modifiers (99/1 – 40/60)

Column cleaning (general method)・ Flush the column with solution containing a higher ratio of modifiers (for example, for CO2/methanol mobile phase,
concentration of methanol should be increased) for washing out the compounds that have a great capacity for retention in thecolumn. When further cleaning is required, flush with 100% ethanol is effective.
・ When a mobile phase containing acid or amine is used, replace with CO2/modifiers containing neither of them (at the same ratio
as the mobile phase), then wash as above procedure. Storing a column with a mobile phase containing additive is notrecommended even for a short period of time.
・ The column needs to be replaced when these cleaning methods do not regenerate the column performance. To extend thecolumn lifetime, especially for samples containing large amount of impurities, we recommend a sample pretreatment conductedcarefully prior to introducing the sample to the column.

YMC Karasuma-Gojo Bldg., 284 Daigo-cho, Karasuma Nishiiru Gojo-dori, Shimogyo-ku,
Kyoto 600-8106, Japan
TEL +81-75-342-4515 FAX +81-75-342-4550URL:www.ymc.co.jp
E-Mail:sales@ymc.co.jp(UM210617CE) 2/2