EA01060D/06/A2/03.18
71414161
2018-10-01

Products

Solutions

Services

Installation Instructions
Replacing the DSC-Sensor andthe S-DAT
Prowirl 200

EN Installation Instructions: EA01060D →  3
DE Einbauanleitung: EA01060D →  25

EA01060D

Replacement of DSC sensor and S-DAT
Prowirl 200
Table of contents
1

Overview of spare part sets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

2

Designated use . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4

3

Personnel authorized to carry out repairs . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

4

Safety instructions . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 5

5Symbols used . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6

6Tools list . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

7Important information regarding the replacement of the compact and remote sensor
version . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

8

Removing the transmitter housing Prowirl 200 compact version, replacing S-DAT . . . . 99Removing the sensor connection housing Prowirl 200 remote version . . . . . . . . . . . . . . 11

10

Replacing the DSC sensor, DSC sensor seal Prowirl 200 . . . . . . . . . . . . . . . . . . . . . . . . . . . . 12

11

Assembling the DSC sensor, DSC sensor seal Prowirl 200 . . . . . . . . . . . . . . . . . . . . . . . . . . . 15

12

Installing the Prowirl 200 transmitter housing, compact version, inserting the
S-DAT . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 18

13

Installing the sensor connection housing, Prowirl 200 remote version . . . . . . . . . . . . . . 20

14Important setting following DSC sensor replacement . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 21

15

Recommended torques for Prowirl 200 . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 22

Endress+Hauser

3

EA01060D

1

Overview of spare part sets

The Installation Instructions apply to the following spare part sets:
Order number

Original spare part set

Contents

XPD0012–Set, DSC sensor Prowirl 200 standard

1 × sensor, 1 × seal, 1 × S-DAT

XPD0013–Set, S-DAT

1 × S-DAT• The order number of the spare parts set (on the product label on the package) candiffer from the production number (on the label directly on the spare part)!
• You can find the order number of the relevant spare parts set by entering the
production number of the spare part in the spare parts search tool.
• We recommend that you keep the Installation Instructions and packaging together at
all times.

2

Designated use

The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type. Only original parts from Endress+Hauser may be used. Only
spare parts kits designed by Endress+Hauser for the measuring device can be used at any
time.
Inspection is performed using the W@M Device Viewer. The procedure involved is describedas follows.
A spare parts overview is located in the interior of some measuring devices. If the spareparts kit is specified on this list, no inspection is required.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Enter serial number (ser. no.), search for productdata and click Spare Parts.

3.

All spare parts for the measuring device aredisplayed.

4.

Determine the order number of the spare parts kit.

spare parts kit may be used only if the order Thenumber of the spare parts kit corresponds to anorder number on the spare parts list.

4

Endress+Hauser

EA01060D

3

Personnel authorized to carry out repairs

Authorization to carry out repairs depends on the measuring device's approval type. The tablebelow shows the authorized group of people in each case.
Whoever carries out the repairs has full responsibility to ensure that work is carried out
safely and to the required quality standard. He/she must also guarantee the safety of thedevice following repair.

1)

4

Measuring device approval

Personnel authorized to perform repairs 1)

Without approval

2, 3With approval (e.g. IECEx)

2, 3For custody transfer

4

1 = Qualified specialist on customer side, 2 = Service technician authorized by Endress+Hauser,
3 = Endress+Hauser (return measuring device to manufacturer)
4 = Check with local approval center if installation/modification must be performed under supervision.

Safety instructions• Check whether the spare part matches the labeling on the measuring device as described on
the cover page.
• The spare part sets and Installation Instructions are used to replace a faulty unit with afunctioning unit of the same type.
Only use original parts from Endress+Hauser.
• Comply with national regulations governing mounting, electrical installation,commissioning, maintenance and repair procedures.
• The following requirements must be met with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance and repair of the measuring
devices:– Specialized technical staff must be trained in instrument safety.
– They must be familiar with the individual operating conditions of the devices.
– In the case of Ex-certified measuring devices, they must also be trained in explosion
protection.
• The measuring device is energized! Risk of fatal injury from electric shock. Open themeasuring device only when the device is de-energized.
• When using Ex-certified measuring devices: Open device only when in a de-energized state(allow 10 minutes to elapse after switching off the power supply) or in environments which
do not have a potentially explosive atmosphere.
• In the case of measuring devices in safety-related applications in accordance withIEC 61508 or IEC 61511: Commission in accordance with Operating Instructions after
repair. Document the repair procedure.
• Before removing the device: set the process to a safe state and purge the pipe of dangerous
process substances.
• Hot surfaces! Risk of injury! Before commencing work: allow the system and measuring
device to cool down to a touchable temperature.
Endress+Hauser

5

EA01060D• In the case of devices in custody transfer, the custody transfer status no longer applies oncethe seal has been removed.
• The Operating Instructions for the device must be followed.
• Risk of damaging the electronic components! Ensure you have a working environment
protected from electrostatic discharge.
• After removing the electronics compartment cover: risk of electrical shock due to missing
touch protection!
Turn the measuring device off before removing internal covers.
• Modifications to the measuring device are not permitted.
• Only open the housing for a brief period. Avoid the penetration of foreign bodies, moisture
or contaminants.
• Replace defective seals only with original seals from Endress+Hauser.
• If threads are damaged or defective, the measuring device must be repaired.
• Threads (e.g. of the electronics compartment cover and connection compartment cover)must be lubricated if an abrasion-proof dry lubricant is not available. Use acid-free, nonhardening lubricant.
• If spacing is reduced or the dielectric strength of the measuring device cannot be
guaranteed during repair work, perform a test on completion of the work (e.g. high-voltage
test in accordance with the manufacturer's instructions).
• Service plug:– Do not connect in explosive atmospheres.
– Only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device outlined in the Operating
Instructions.
If you have any questions, please contact your Endress+Hauser service organization.

5

Symbols used

5.1

Safety symbols

6

SymbolMeaning

DANGERDANGER!
This symbol alerts you to a dangerous situation. Failure to avoid this situation will result inserious or fatal injury.

WARNINGWARNING!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inserious or fatal injury.

CAUTIONCAUTION!
This symbol alerts you to a dangerous situation. Failure to avoid this situation can result inminor or medium injury.

NOTICENOTE!
This symbol contains information on procedures and other facts which do not result inpersonal injury.
Endress+Hauser

EA01060D

5.2

Symbols for certain types of information

SymbolMeaning
Permitted
Procedures, processes or actions that are permitted.
Forbidden
Procedures, processes or actions that are forbidden.
Tip
Indicates additional information.

1. , 2. , 3. …

6

Series of steps

Tools list

3 mm, 4 mm,

8 mm

5 mm, 6 mm

17 mm

7

T10, T20

0.5 × 3.5 mm

Important information regarding the replacement of
the compact and remote sensor version

NOTICE‣ 1. The sealing surface may not be scratched.
‣ 2. Remove the sealing washer with a suitable tool without scratching the sealing surface.
‣ 3. Clean the sealing surface on the meter body with a suitable solvent and a lint-free cloth.
‣ 4. The bore must be perfectly clean.
‣ 5. Place the new sensor seal on the sealing surface with the printed surface (if present)
facing upwards (see graphic below).

1

Example: sensor seal with printed surface

6. Grease the threads and the head contact areas of the sensor screws.
Endress+Hauser

7

EA01060D

7. Provide each thread and contact area of the screws with one drop of lubricant that issuitable for the application temperature range. The high-temperature paste HTP (50048898)
is recommended.
8. See the graphic below for the installation direction of the DSC sensor:A2

A = The marking on the sensor must point in the flow direction

LDANGERThe sensor can accelerate to high speeds. If the sensor is exposed to the full processpressure, strong internal forces act on the sensor. The following measures must be
taken:‣ To replace the sensor, the pipe must be completely free from pressure. Residual pressure
on the pipe can cause the sensor to pop out as soon as the retaining screws are released!
‣ In the case of toxic, explosive or flammable fluids, the pipe in which the measuring device
is installed must be rinsed or cleaned before the sensor is replaced!
‣ Before starting work, allow the pipe to cool down to a safe temperature!

8

Endress+Hauser

EA01060D

8

Removing the transmitter housing Prowirl 200
compact version, replacing S-DATLCAUTION
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is deenergized.
Replacing S-DAT: When replacing the S-DAT, the DSC sensor serial number must be
specified. Menu Expert → Diagnostics → Sensor information → Serial number
1→

2→

2.

3 mm (0.12 in)

1.

3 4

1 2

3 mm
3→

4→

5→

2.
–

Esc

+
E

1.
3 mm
6→

Endress+Hauser

7→

8→

9

EA01060D

9→

10 →

11 →

T10

1.

8 mm
2.
12 →

13

~15°

10

Endress+Hauser

EA01060D

9

Removing the sensor connection housing Prowirl 200
remote version

LCAUTION
The measuring device is energized!
Risk of fatal injury from electric shock.
‣ Open the measuring device only when the device is deenergized.
1→

2→

2.

3.
2.
1.
1.

3 4

1 2

3 mm
0.5 × 3.5
3→

4→

5→

T10

6→

7

T10
1.

Endress+Hauser

2.

11

EA01060D

10

Replacing the DSC sensor, DSC sensor seal Prowirl 200

10.1

Removing the DSC sensor, DSC sensor seal Prowirl D, F, R (volume flow basis)

LDANGER
Risk of injury!
‣ To replace the DSC sensor, the pipe must be completely free from pressure. Residualpressure on the pipe can cause the DSC sensor to pop out as soon as the
DSC sensor retaining screws are released!
‣ In the case of toxic, explosive or flammable fluids, the pipe in which the measuring device
is installed must be rinsed or cleaned before the DSC sensor is replaced!
‣ Before starting work, allow the pipe to cool down to a safe temperature.
Proceed as described in Section 8 →  9or Section 9 →  11 and as illustrated in the
diagrams below.
1→

2

2.

2.
4 mm
4 mm

1.
1.

3.

10.2

Removing the DSC sensor, DSC sensor seal Prowirl D, F, R (volume flow, high/low temperature or mass flow (integrated temperature measurement)

LDANGER
Risk of injury!
‣ To replace the DSC sensor, the pipe must be completely free from pressure. Residualpressure on the pipe can cause the DSC sensor to pop out as soon as the
DSC sensor retaining screws are released!
‣ In the case of toxic, explosive or flammable fluids, the pipe in which the measuring device
is installed must be rinsed or cleaned before the DSC sensor is replaced!
‣ Before starting work, allow the pipe to cool down to a safe temperature.

12

Endress+Hauser

EA01060D

Proceed as described in Section 8 →  9or Section 9 →  11 and as illustrated in the
diagrams below.
1→

2

2.

4 mm

4 mm

1.

3.

10.3

Removing the DSC sensor, DSC sensor seal, Prowirl O, C (volume
flow, Alloy 718 or mass flow, Alloy 718)

LDANGER
Risk of injury!
‣ To replace the DSC sensor, the pipe must be completely free from pressure. Residualpressure on the pipe can cause the DSC sensor to pop out as soon as the
DSC sensor retaining screws are released!
‣ In the case of toxic, explosive or flammable fluids, the pipe in which the measuring device
is installed must be rinsed or cleaned before the DSC sensor is replaced!
‣ Before starting work, allow the pipe to cool down to a safe temperature.
Proceed as described in Section 7 →  9or Section 8 →  11 and as illustrated in the
diagrams below.
1→

2

2.
1.
5 mm

6 mm

3.

Endress+Hauser

13

EA01060D

10.4

Removing the DSC sensor, DSC sensor seal Prowirl O (volume
flow, titanium)

LDANGER
Risk of injury!
‣ To replace the DSC sensor, the pipe must be completely free from pressure. Residualpressure on the pipe can cause the DSC sensor to pop out as soon as the
DSC sensor retaining screws are released!
‣ In the case of toxic, explosive or flammable fluids, the pipe in which the measuring device
is installed must be rinsed or cleaned before the DSC sensor is replaced!
‣ Before starting work, allow the pipe to cool down to a safe temperature.
Proceed as described in Section 7 →  9or Section 8 →  11 and as illustrated in the
diagrams below.
1→

2

17 mm

2.
6 mm

1.

1.
A

C

D

14

2.

B

3.

Endress+Hauser

EA01060D

11

Assembling the DSC sensor, DSC sensor seal Prowirl
200

LDANGER
Risk of injury!
‣ When reassembling, comply with the torques indicated in the tables in Section 15!
→  22

11.1

Important information on replacing the DSC sensor seal, the DSC
sensor, compact and remote version

1. The sealing surface may not be scratched.
2. Remove the sealing washer with a suitable tool without scratching the sealing surface.
3. Clean the sealing surface on the meter body with a suitable solvent and a lint-free cloth.
4. The bore must be perfectly clean.
5. Place the new DSC sensor seal on the sealing surface with the printed surface (if present)
facing upwards (see example in the graphic below).
1‣ Example: DSC sensor seal withprinted surface

Endress+Hauser

15

EA01060D

6. See the graphic below for the installation direction of the DSC sensor:
1

A‣ 7D2C: Marking (A) on the DSC sensor must point in the direction of
flow.

11.2

Reassembling the DSC sensor, DSC sensor seal Prowirl D, F, R
(volume flow, basis)

LDANGER
Risk of injury!
‣ When reassembling, comply with the torques indicated in the tables in Section 15!
→  22
1→

2

2.

3.

1.
2.
4 mm
4 mm

Pos. S

Pos. H1.

16

Endress+Hauser

EA01060D

11.3

Reassembling the DSC sensor, DSC sensor seal Prowirl D, F, R
(volume flow, high/low temp. or mass flow (integrated temperature measurement))

LDANGER
Risk of injury!
‣ When reassembling, comply with the torques indicated in the tables in Section 15!
→  22
1→

2

2.

1.

3.

2.

4 mm

4 mm

Pos. H
Pos. S1.

11.4

Reassembling the DSC sensor, DSC sensor seal, Prowirl O, C (volume flow, Alloy 718 or mass flow, Alloy 718)

LDANGER
Risk of injury!
‣ When reassembling, comply with the torques indicated in the tables in Section 15!
→  231→

2

2.

Pos. S

T10

2.

1.

3.
Sensor O =

6 mm

5 mm

Sensor C =1/4"

Pos. H1.

Endress+Hauser

17

EA01060D

11.5

Reassembling the DSC sensor, DSC sensor seal Prowirl O (volume
flow, titanium)

LDANGER
Risk of injury!
‣ When reassembling, comply with the torques indicated in the tables in Section 15!
→  23
1→

2

17 mm

3.

2.

1.
2.

Pos. S
A

C

6 mm

B
Pos. HD

12

1.

Installing the Prowirl 200 transmitter housing,compact version, inserting the S-DAT
Keep the S-DAT in a safe place and insert into the transmitter, see Fig. 5 + 6 below.

1→

2→

3→

~15°

8 mm

18

Endress+Hauser

EA01060D

4→

5→

T10

6→

1.

2.
7→

8→

9→

1.

1.
–

Esc

+
E

2.

2.

1.

3 mm
10 →

11 →

3 mm (0.12 in)

1.

2.
3 4

1 2

12

1.

2.
3 mm

Endress+Hauser

19

EA01060D

For detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

13

Installing the sensor connection housing, Prowirl 200
remote version

1→

2→

3→

T10
1.
2.

4→

5→

T10

20

Endress+Hauser

EA01060D

6→

7

1.

2.
2.
3.
1.
3 4

1 2

3 mm
0.5 × 3.5

For detailed information on the electrical connection, see the "Electrical connection"section of the Operating Instructions for the device.
For detailed information on commissioning, see the "Commissioning" section of the
Operating Instructions for the device.

14

Important setting following DSC sensor replacement

After replacing the DSC sensor, the service technician must take the following action: Expert→ Sensor → Sensor adjustm. → Sensor adjustm. (7734) Start the DSC sensor adjustment forsaving the reference measuring data.
For more information, see the "Sensor adjustment" section• HART: SH01019D• FOUNDATION Fieldbus: SH01039D• Profibus PA: SH01030D

Endress+Hauser

21

EA01060D

15

Recommended torques for Prowirl 200

Position H = housing support screws
Position S = DSC sensor screws

1

4

3

2

3

Tightening sequence for the sensor screws on the sensor holder for Prowirl D, R, F, O, C

DSC sensor D, F, R (volume flow basis) and housing support
See Section 11.2, pos. S and pos. H in the diagrams →  16
Pos. S

Pos. H

Step

Torque[Nm (lbft)]

Tightening sequence1

2.0 (1.5)12

5.0 (3.7)

2, 1, 3, 4

3

7.0 (5.2)

2, 1, 3, 4

Step

Torque[Nm (lbft)]

Tightening sequence1

7.0 (5.2)

1, 3, 2, 4

Provide each screw with one drop of lubricant that is suitable for the application temperature range. The hightemperature paste HTP (50048898) is recommended.

DSC sensor D, F, R (high/low temperature) and housing support
See Section 11.3, pos. S and pos. H in the diagrams →  17
Pos. S

Pos. H

Step

Torque[Nm (lbft)]

Tightening sequence1

2.0 (1.5)12

5.0 (3.7)

2, 1, 3, 4

3

7.0 (5.2)

2, 1, 3, 4

Step

Torque[Nm (lbft)]

Tightening sequence1

7.0 (5.2)

1, 3, 2, 4

Provide each screw with one drop of lubricant that is suitable for the application temperature range. The hightemperature paste HTP (50048898) is recommended.

22

Endress+Hauser

EA01060D

DSC sensor O, C (volume flow, Alloy 718) and housing support
See Section 11.4, pos. S and pos. H in the diagrams →  17
Pos. S

Pos. H

Step

Torque[Nm (lbft)]

Tightening sequence1

2.0 (1.5)12

5.0 (3.7)

2, 1, 3, 4

3

10.0 (7.4)

2, 1, 3, 4

4

15.0 (11.0)

2, 1, 3, 4

Step

Torque[Nm (lbft)]

Tightening sequence1

7.0 (5.2)

1, 3, 2, 4

Provide each screw with one drop of lubricant that is suitable for the application temperature range. The hightemperature paste HTP (50048898) is recommended.

DSC sensor O (titanium) and housing support
See Section 11.5, pos. S and pos. H in the diagrams →  18
Pos. S

Pos. H

Step

Torque[Nm (lbft)]

Tightening sequence1

5.0 (3.7)

12

10.0 (7.4)

1, 2, 1, 2, 3, 4

3

20.0 (14.8)

2, 1, 3, 4

4

26.0 (19.2)

1, 2, 3, 4

Step

Torque[Nm (lbft)]

Tightening sequence1

14.0 (10.3)

1, 3, 2, 4

Provide each screw with one drop of lubricant that is suitable for the application temperature range. The hightemperature paste HTP (50048898) is recommended.

Endress+Hauser

23

EA01060D

Austausch DSC-Sensor und S-DAT
Prowirl 200
Inhaltsverzeichnis
1Übersicht Ersatzteilsets . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26

2

Bestimmungsgemäße Verwendung . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 26

3

Reparaturberechtigte Personen . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27

4

Sicherheitshinweise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 27

5

Verwendete Symbole . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 28

6

Werkzeugliste . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 29

7

Wichtige Hinweise zum Austausch des Sensor Kompakt- und Getrennt Version . . . . . 29

8

Ausbau Messumformergehäuse Prowirl 200 Kompaktausführung, Austausch
S-DAT . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 31

9

Ausbau Sensor Anschlussgehäuse Prowirl 200 Getrenntausführung . . . . . . . . . . . . . . . . 33

10Austausch DSC-Sensor, DSC-Sensordichtung Prowirl 200 . . . . . . . . . . . . . . . . . . . . . . . . . . 34

11

Zusammenbau DSC-Sensor, DSC-Sensordichtung Prowirl 200 . . . . . . . . . . . . . . . . . . . . . . 37

12Einbau Messumformergehäuse Prowirl 200 Kompaktausführung, einsetzen
S-DAT . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 40

13Einbau Sensor Anschlussgehäuse Prowirl 200 Getrenntausführung . . . . . . . . . . . . . . . . . 42

14

Wichtige Einstellung nach dem DSC-Sensoraustausch . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 43

15

Empfohlene Drehmomente für Prowirl 200 . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 44

Endress+Hauser

25

EA01060D

1Übersicht Ersatzteilsets

Die Einbauanleitung ist für folgende Ersatzteilsets gültig:
Bestellnummer

Original Ersatzteilset

Inhalt

XPD0012–

Set DSC-Sensor Prowirl 200 Standard

1 × Sensor, 1 × Dichtung, 1 × S-DAT

XPD0013–

Set S-DAT

1 × S-DAT• Die Bestellnummer des Ersatzteilsets (auf dem Produktaufkleber der Verpackung)kann sich von der Produktionsnummer (auf dem Aufkleber direkt auf dem Ersatzteil)unterscheiden!
• Durch Eingabe der Produktionsnummer des Ersatzteiles im Ersatzteilfindetool kanndie Bestellnummer des ensprechenden Ersatzteilsets ermittelt werden.
• Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

2

Bestimmungsgemäße Verwendung

Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile vonEndress+Hauser verwendet werden. Grundsätzlich dürfen nur Ersatzteilsets verwendet
werden, die von Endress+Hauser für das Messgerät vorgesehen sind.
Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist
nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Inneren des Gerätes eine Ersatzteilübersicht.
Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

2.

www.endress.com/deviceviewer

Ser. No.: 12345

3.

4.

26

Order code=

Description?

1.

www.endress.com/deviceviewer

2.

Seriennummer (Ser. No.) eingeben, Produktdaten
suchen und auf Ersatzteile klicken.

3.

Anzeige aller Ersatzteile zum Messgerät.

4.

Die Bestellnummer des Ersatzteilsets ermitteln.

Nur wenn die Bestellnummer des Ersatzteilsets mit
 einer
Bestellnummer in der Ersatzteillisteübereinstimmt, darf das Ersatzteilset verwendet
werden.

Endress+Hauser

EA01060D

3

Reparaturberechtigte Personen

Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts
abhängig. Die Tabelle zeigt den jeweils berechtigten Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die
Sicherheit während der Arbeiten, die Qualität der Ausführung und die Sicherheit desGeräts nach der Reparatur.

1)

4

Zulassung des Messgeräts

Reparaturberechtigter Personenkreis 1)

Ohne Zulassung

2, 3

Mit Zulassung (z.B. IECEx)

2, 3

Bei eichfähigem Verkehr

4

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker,
3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)
4 = Mit der lokalen Zulassungsstelle prüfen, ob ein Ein-/Umbau unter Aufsicht erfolgen muss.

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem Messgerät passt, wie auf
der Titelseite beschrieben.
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen einefunktionierende Einheit des gleichen Typs zu ersetzen.
Nur Originalteile von Endress+Hauser verwenden.
• Nationale Vorschriften bezüglich der Montage, elektrischen Installation, Inbetriebnahme,
Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage, elektrische Installation,
Inbetriebnahme, Wartung und Reparatur der Messgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet.
– Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
– Bei Ex-zertifizierten Messgeräten: zusätzlich im Explosionsschutz ausgebildet.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag. Messgerät nur im
spannungslosen Zustand öffnen.
• Bei Ex-zertifizierten Messgeräten: Nur in spannungslosem Zustand (nach Berücksichtigung
einer Wartezeit von 10 Minuten nach Abschalten der Energiezufuhr) oder in Umgebungenöffnen, die keine explosionsfähige Atmosphäre enthalten.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw.
IEC 61511: Nach Reparatur Neuinbetriebnahme gemäß Betriebsanleitung durchführen.
Reparatur dokumentieren.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen und Leitung von gefährlichen
Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn: Anlage und Messgerät
auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der
geeichte Zustand aufgehoben.
Endress+Hauser

27

EA01060D• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte Arbeitsumgebung
herstellen.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durch aufgehobenen
Berührungsschutz!
Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern, Feuchtigkeit oder
Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen von Endress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel) müssen geschmiert sein,
sofern keine abriebfeste Trockenschmierung vorhanden ist. Säurefreies, nicht härtendes
Fett verwenden.
• Wenn bei den Reparaturarbeiten Abstände reduziert oder die Spannungsfestigkeit desMessgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten
durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– Nicht in explosionsfähiger Atmosphäre anschließen.
– Nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport und zur Rücksendung
beachten.
Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser Serviceorganisation.

5

Verwendete Symbole

5.1

Warnhinweissymbole

28

SymbolBedeutung

GEFAHRGEFAHR!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen wird.

WARNUNGWARNUNG!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu Tod oder schwerer Körperverletzung führen kann.

VORSICHTVORSICHT!
Dieser Hinweis macht auf eine gefährliche Situation aufmerksam, die, wenn sie nicht
vermieden wird, zu leichter oder mittelschwerer Körperverletzung führen kann.

HINWEISHINWEIS!
Dieser Hinweis enthält Informationen zu Vorgehensweisen und weiterführenden
Sachverhalten, die keine Körperverletzung nach sich ziehen.

Endress+Hauser

EA01060D

5.2

Symbole für Informationstypen

SymbolBedeutung
Erlaubt
Abläufe, Prozesse oder Handlungen, die erlaubt sind.
Verboten
Abläufe, Prozesse oder Handlungen, die verboten sind.
Tipp
Kennzeichnet zusätzliche Informationen.

1. , 2. , 3. …

6

Handlungsschritte

Werkzeugliste

3 mm, 4 mm,

8 mm

5 mm, 6 mm

17 mm

7

T10, T20

0,5 × 3,5 mm

Wichtige Hinweise zum Austausch des Sensor
Kompakt- und Getrennt Version

HINWEIS‣ 1. Es dürfen keine Kratzer auf der Dichtfläche entstehen.
‣ 2. Dichtscheibe mit einem geeigneten Gegenstand entfernen ohne die Dichtfläche zu
zerkratzen.
‣ 3. Die Dichtfläche auf dem Grundkörper mit einem geeigneten Lösungsmittel und einem
fuselfreiem Tuch reinigen.
‣ 4. Die Bohrung zur Aufnahme muss absolut sauber sein.
‣ 5. Die neue Sensordichtung mit dem Aufdruck (falls vorhanden) nach oben auf die
Dichtfläche legen siehe Grafik unten).

4

Beispiel Sensordichtung mit AufdruckEndress+Hauser

29

EA01060D

6. Die Gewinde und Kopfauflagen der Sensorschrauben fetten.
7. Die Gewinde und Auflageflächen der Schrauben mit je einem Tropfen eines Fettesversehen, das für den Anwendungstemperatur-bereich geeignet ist. Empfohlen wird die
Hochtemperatur-Paste HTP (50048898).
8. Einbaurichtung des DSC-Sensors siehe Grafik unten:A5

A = Markierung auf dem Sensor muss in Durchflussrichtung zeigen

LGEFAHR
Der Messaufnehmer kann auf hohe Geschwindigkeiten beschleunigen. Wird der
Messaufnehmer dem vollen Prozessdruck ausgesetzt, wirken auf ihn hohe innere Kräfte.
Folgende Maßnahmen sind zu ergreifen:‣ Zum Austausch des Sensors muss die Rohrleitung absolut druckfrei sein. Restdruck auf derRohrleitung kann den Sensor herraussprengen, sobald die Halteschrauben gelöst werden!
‣ Bei giftigen, explosiven oder brennbaren Messstoffen muss die Rohrleitung in der das
Messgerät eingebaut ist, vor dem Austausch des Sensors, gespült bzw. gereinigt werden!
‣ Vor Arbeitsbeginn die Rohrleitung auf eine ungefährliche Temperatur abkühlen lassen!

30

Endress+Hauser

EA01060D

8

Ausbau Messumformergehäuse Prowirl 200
Kompaktausführung, Austausch S-DATLVORSICHT
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
Austausch S-DAT: Beim Austausch des S-DAT muss die DSC-Sensor-Seriennummer
angegeben werden. Menu Expert → Diagnostics → Sensorinformation → Serialnumber
1→

2→

2.

3 mm (0.12 in)

1.

3 4

1 2

3 mm
3→

4→

5→

2.
–

Esc

+
E

1.
3 mm
6→

Endress+Hauser

7→

8→

31

EA01060D

9→

10 →

11 →

T10

1.

8 mm
2.
12 →

13

~15°

32

Endress+Hauser

EA01060D

9

Ausbau Sensor Anschlussgehäuse Prowirl 200
Getrenntausführung

LVORSICHT
Messgerät unter Spannung!
Lebensgefahr durch Stromschlag.
‣ Messgerät nur im spannungslosen Zustand öffnen.
1→

2→

2.

3.
2.
1.
1.

3 4

1 2

3 mm
0.5 × 3.5
3→

4→

5→

T10

6→

7

T10
1.

Endress+Hauser

2.

33

EA01060D

10Austausch DSC-Sensor, DSC-Sensordichtung
Prowirl 200

10.1

Ausbau DSC-Sensor, DSC-Sensordichtung Prowirl D, F, R (Volumenfluss Basis)

LGEFAHR
Verletzungsgefahr!
‣ Zum Austausch des DSC-Sensors muss die Rohrleitung absolut druckfrei sein. Restdruck auf
der Rohrleitung kann den DSC-Sensor herraussprengen, sobald die
DSC-Sensorhalteschrauben gelöst werden!
‣ Bei giftigen, explosiven oder brennbaren Messstoffen muss die Rohrleitung in der das
Messgerät eingebaut ist, vor dem Austausch des DSC-Sensors, gespült bzw. gereinigt
werden!
‣ Vor Arbeitsbeginn die Rohrleitung auf eine ungefährliche Temperatur abkühlen lassen.
Vorgehen wie in Kapitel 8 →  31 oder wie in Kapitel 9 →  33 und wie in den Bildern
unten.
1→

2

2.

2.
4 mm
4 mm

1.
1.

3.

10.2

Ausbau DSC-Sensor, DSC-Sensordichtung Prowirl D, F, R (Volumenfluss Hoch-/Tieftemperatur oder Massefluss (integrierte
Temperaturmessung)

LGEFAHR
Verletzungsgefahr!
‣ Zum Austausch des DSC-Sensors muss die Rohrleitung absolut druckfrei sein. Restdruck auf
der Rohrleitung kann den DSC-Sensor herraussprengen, sobald die
DSC-Sensorhalteschrauben gelöst werden!
‣ Bei giftigen, explosiven oder brennbaren Messstoffen muss die Rohrleitung in der das
Messgerät eingebaut ist, vor dem Austausch des DSC-Sensors, gespült bzw. gereinigt
werden!
‣ Vor Arbeitsbeginn die Rohrleitung auf eine ungefährliche Temperatur abkühlen lassen.

34

Endress+Hauser

EA01060D

Vorgehen wie in Kapitel 8 →  31 oder wie in Kapitel 9 →  33 und wie in den Bildern
unten.
1→

2

2.

4 mm

4 mm

1.

3.

10.3

Ausbau DSC-Sensor, DSC-Sensordichtung Prowirl O, C (Volumenfluss Alloy 718 oder Massefluss Alloy 718)

LGEFAHR
Verletzungsgefahr!
‣ Zum Austausch des DSC-Sensors muss die Rohrleitung absolut druckfrei sein. Restdruck auf
der Rohrleitung kann den DSC-Sensor herraussprengen, sobald die
DSC-Sensorhalteschrauben gelöst werden!
‣ Bei giftigen, explosiven oder brennbaren Messstoffen muss die Rohrleitung in der das
Messgerät eingebaut ist, vor dem Austausch des DSC-Sensors, gespült bzw. gereinigt
werden!
‣ Vor Arbeitsbeginn die Rohrleitung auf eine ungefährliche Temperatur abkühlen lassen.
Vorgehen wie in Kapitel 7 →  31 oder wie in Kapitel 8 →  33 und wie in den Bildern
unten.
1→

2

2.
1.
5 mm

6 mm

3.

Endress+Hauser

35

EA01060D

10.4

Ausbau DSC-Sensor, DSC-Sensordichtung Prowirl O (Volumenfluss Titan)

LGEFAHR
Verletzungsgefahr!
‣ Zum Austausch des DSC-Sensors muss die Rohrleitung absolut druckfrei sein. Restdruck auf
der Rohrleitung kann den DSC-Sensor herraussprengen, sobald die
DSC-Sensorhalteschrauben gelöst werden!
‣ Bei giftigen, explosiven oder brennbaren Messstoffen muss die Rohrleitung in der das
Messgerät eingebaut ist, vor dem Austausch des DSC-Sensors, gespült bzw. gereinigt
werden!
‣ Vor Arbeitsbeginn die Rohrleitung auf eine ungefährliche Temperatur abkühlen lassen.
Vorgehen wie in Kapitel 7 →  31 oder wie in Kapitel 8 →  33 und wie in den Bildern
unten.
1→

2

17 mm

2.
6 mm

1.

1.
A

C

D

36

2.

B

3.

Endress+Hauser

EA01060D

11

Zusammenbau DSC-Sensor, DSC-Sensordichtung
Prowirl 200

LGEFAHR
Verletzungsgefahr!
‣ Beim Zusammenbau Drehmomente in den Tabellen in Kap. 15 →  44 beachten!

11.1

Wichtige Hinweise zum Austausch der DSC-Sensordichtung, desDSC-Sensors Kompakt- und Getrenntausführung

1. Es dürfen keine Kratzer auf der Dichtfläche entstehen.
2. Dichtscheibe mit einem geeigneten Gegenstand entfernen ohne die Dichtfläche zu
zerkratzen.
3. Die Dichtfläche auf dem Grundkörper mit einem geeigneten Lösungsmittel und einem
fuselfreiem Tuch reinigen.
4. Die Bohrung zur Aufnahme muss absolut sauber sein.
5. Die neue DSC-Sensordichtung mit dem Aufdruck (falls vorhanden) nach oben auf die
Dichtfläche legen (siehe Beispiel in der Grafik unten).
1‣ Beispiel: DSC-Sensordichtung mit
Aufdruck

Endress+Hauser

37

EA01060D

6. Einbaurichtung des DSC-Sensors siehe Grafik unten:
1A‣ 7D2C: Markierung (A) auf dem DSC-Sensor muss in
Durchflussrichtung zeigen.

11.2

Zusammenbau DSC-Sensor, DSC-Sensordichtung Prowirl D, F, R
(Volumefluss Basis)

LGEFAHR
Verletzungsgefahr!
‣ Beim Zusammenbau Drehmomente in den Tabellen in Kap. 15 →  44 beachten!
1→

2

2.

3.

1.
2.
4 mm
4 mm

Pos. S

Pos. H

1.

38

Endress+Hauser

EA01060D

11.3

Zusammenbau DSC-Sensor, DSC-Sensordichtung Prowirl D, F, R
(Volumenfluss Hoch -/Tieftemp. oder Massefluss (integrierte
Temperaturmessung))

LGEFAHR
Verletzungsgefahr!
‣ Beim Zusammenbau Drehmomente in den Tabellen in Kap. 15 →  44 beachten!
1→

2

2.

1.

3.

2.

4 mm

4 mm

Pos. H
Pos. S1.

11.4

Zusammenbau DSC-Sensor, DSC-Sensordichtung Prowirl O, C
(Volumemfluss Alloy 718 oder Massefluss Alloy 718)

LGEFAHR
Verletzungsgefahr!
‣ Beim Zusammenbau Drehmomente in den Tabellen in Kap. 15 →  45 beachten!
1→

2

2.

Pos. S

T10

2.

1.

3.
Sensor O =

6 mm

5 mm

Sensor C =1/4"

Pos. H1.

Endress+Hauser

39

EA01060D

11.5

Zusammenbau DSC-Sensor, DSC-Sensordichtung Prowirl O (Volumenfluss Titan)

LGEFAHR
Verletzungsgefahr!
‣ Beim Zusammenbau Drehmomente in den Tabellen in Kap. 15 →  45 beachten!
1→

2

17 mm

3.

2.

1.
2.

Pos. S
A

C

6 mm

B
Pos. HD

121.

Einbau Messumformergehäuse Prowirl 200
Kompaktausführung, einsetzen S-DATDen S-DAT an einem sicheren Ort aufbewahren und in den Messumformer einsetzen
siehe Bild 5 + 6 unten.

1→

2→

3→

~15°

8 mm

40

Endress+Hauser

EA01060D

4→

5→

T10

6→

1.

2.
7→

8→

9→

1.

1.
–

Esc

+
E

2.

2.

1.

3 mm
10 →

11 →

3 mm (0.12 in)

1.

2.
3 4

1 2

12

1.

2.
3 mm

Endress+Hauser

41

EA01060D

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

13

Einbau Sensor Anschlussgehäuse Prowirl 200
Getrenntausführung

1→

2→

3→

T10
1.
2.

4→

5→

T10

42

Endress+Hauser

EA01060D

6→

7

1.

2.
2.
3.
1.
3 4

1 2

3 mm
0.5 × 3.5

Detaillierte Angaben zum elektrischen Anschluss: Kapitel "Elektrischer Anschluss",
Betriebsanleitung zum Gerät.
Detaillierte Angaben zur Inbetriebnahme: Kapitel "Inbetriebnahme", Betriebsanleitungzum Gerät.

14

Wichtige Einstellung nach dem DSC-Sensoraustausch

Nach dem DSC-Sensortausch ist folgende Interaktion durch den Servicetechniker auszuführen:
Expert → Sensor → Sensor adjustm. → Sensor adjustm. (7734) Starten des
DSC-Sensorabgleichs für die Speicherung der Referenzmessdaten.
Weitere Informationen: Kapitel "Sensorabgleich"• HART: SH01019D• FOUNDATION Fieldbus: SH01039D• Profibus PA: SH01030D

Endress+Hauser

43

EA01060D

15

Empfohlene Drehmomente für Prowirl 200

Position H = Schrauben Gehäusestütze
Position S = Schrauben DSC-Sensor

1

4

3

2

6

Anzugssequenz der Sensorschrauben am Sensorhalter für Prowirl D, R, F, O, C

DSC-Sensor D, F, R (Volumenfluss Basis) und Gehäusestütze
siehe Kap. 11.2, Pos. S und Pos. H in den Bildern →  38
Pos. S

Pos. H

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz

1

2.0 (1.5)12

5.0 (3.7)

2, 1, 3, 4

3

7.0 (5.2)

2, 1, 3, 4

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz1

7.0 (5.2)

1, 3, 2, 4

Die Schrauben mit je einem Tropfen eines Fettes versehen, das für den Anwendungstemperaturbereich geeignet ist.
Empfohlen wird die Hochtemperatur-Paste HTP (50048898).

DSC-Sensor D, F, R (Hoch/ Tieftemperatur)und Gehäusestütze
siehe Kap. 11.3, Pos. S und Pos. H in den Bildern →  39
Pos. S

Pos. H

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz

1

2.0 (1.5)12

5.0 (3.7)

2, 1, 3, 4

3

7.0 (5.2)

2, 1, 3, 4

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz1

7.0 (5.2)

1, 3, 2, 4

Die Schrauben mit je einem Tropfen eines Fettes versehen, das für den Anwendungstemperaturbereich geeignet ist.
Empfohlen wird die Hochtemperatur-Paste HTP (50048898).

44

Endress+Hauser

EA01060D

DSC-Sensor O, C (Volumenfluss Alloy 718) und Gehäusestütze
siehe Kap. 11.4, Pos. S und Pos. H in den Bildern →  39
Pos. S

Pos. H

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz

1

2.0 (1.5)12

5.0 (3.7)

2, 1, 3, 4

3

10.0 (7.4)

2, 1, 3, 4

4

15.0 (11.0)

2, 1, 3, 4

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz1

7.0 (5.2)

1, 3, 2, 4

Die Schrauben mit je einem Tropfen eines Fettes versehen, das für den Anwendungstemperaturbereich geeignet ist.
Empfohlen wird die Hochtemperatur-Paste HTP (50048898).

DSC-Sensor O (Titan) und Gehäusestütze
siehe Kap. 11.5, Pos. S und Pos. H in den Bildern →  40
Pos. S

Pos. H

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz

1

5.0 (3.7)

12

10.0 (7.4)

1, 2, 1, 2, 3, 4

3

20.0 (14.8)

2, 1, 3, 4

4

26.0 (19.2)

1, 2, 3, 4

Schritt

Drehmoment[Nm (lbft)]

Anzugssequenz1

14.0 (10.3)

1, 3, 2, 4

Die Schrauben mit je einem Tropfen eines Fettes versehen, das für den Anwendungstemperaturbereich geeignet ist.
Empfohlen wird die Hochtemperatur-Paste HTP (50048898).

Endress+Hauser

45

*71414161*
71414161

www.addresses.endress.com