Installation Instruction

Ground terminal
Cerabar S, Deltabar S, Deltabar, Cerabar M, Deltabar M, Deltapilot M
EA01131P/00/A2/01.15
71293523
71293523

Instruction is valid for the following spare part set:
Order number

Original spare part set

71290825

Ground terminal T14/T15/T17, F30/F31

We recommend that the Installation Instruction be kept with the packaging at all times.

Designated use
The spare part set and Installation Instruction are used to replace a faulty unit with a functioning unit of the same type.
Use genuine parts from Endress+Hauser only.
Only original spare part sets, intended by Endress+Hauser for the measuring device, shall be used. The verification has to be done via W@M Device Viewer this procedure is explained below.
For some devices there is an overview of the correct spare part set inside the connection compartment cover.
If the spare part is listed there, the verification is not required.

1.

www.endress.com/deviceviewer

2.

Ser. No.:12345

www.endress.com/deviceviewer
Enter the serial number (Ser. No.), search for product data and click on"spare parts".
List of all spare parts suitable for the instrument.

3.

4.

Check the order number of the spare part.

=?

Does the order number of spare part correspond to the order number inthe spare part list?
YES, the spare part can be used.
NO, the spare part may not be used.

Authorized personnel to carry out repairs
Authorization to carry out a repair depends on the approval of the measuring device. The table shows the respective group of persons for each.
The person who carries out the repair is responsible for safety during the work, the quality of work completed and safety of the device
after repair.
Approval of the measuring device

Group of persons authorized to carry out repairs

Without approval

1, 2, 3

With approval (for Ex. IECEx, ATEX, FM, CSA, TIIS, NEPSI)

1, 2, 3

1 = Trained customer technician, 2 = Service technician authorized by Endress+Hauser, 3 = Endress+Hauser (send measuring device back to manufacturer)

Ground terminal

Safety instructions• Check whether the spare part matches the identification label on themeasuring device. See chapter "Designated use".
• The spare parts set and Installation Instructions are used to replace a faulty unit with a functioning unit of the same type. Use genuine parts from Endress+Hauser only.
• In the case of Ex-certified measuring devices: Observe theinformation in the documentation "Safety Instructions" (XA, ZD).
The relevant Safety Instructions (XA) are indicated on the nameplate.
• The measuring device is energized. Danger: Risk of electric shock!
Open the measuring device in a de-energized state only.
• Before removing the device: set the process in a safe condition andpurge the pipe of dangerous materials.
• Hot surfaces! Risk of injury! Before commencing work, allow thesystem and measuring device to cool down to a touchable
temperature.
• In the case of measuring devices in custody transfer, the custodytransfer status no longer applies once the lead seal has been removed.
• Comply with national regulations governing mounting, electricalinstallation, commissioning, maintenance and repair procedures.
• Requirements with regard to specialized technical staff for themounting, electrical installation, commissioning, maintenance andrepair of the measuring devices:– trained in instrument safety– familiar with the individual operation conditions of the devices– for Ex-certified measuring devices: also trained in explosion
protection• Follow the Operating Instructions for the device.
• Risk of damaging electronic components! Ensure you have a working
environment protected from electrostatic discharge.
• After removing the electronics cover, there is a risk of electric shock
as shock protection is removed! Switch off the measuring device before removing internal covers.

• Modifications to the measuring device are not permitted.
• In the case of measuring devices in safety-related applications inaccordance with IEC 61508 or IEC 61511: After repair recommissionin accordance with Operating Instructions. Document the repairprocedure.
• Only open housing for a brief period. Avoid the penetration of foreign
bodies, moisture or contaminants.
• Replace defective seal/gaskets with genuine parts fromEndress+Hauser only.
• If threads are damaged or defective, the measuring device must berepaired.
• Threads (e.g. of the cover for the electronics and terminalcompartment), which have been prepared by Endress+Hauser with
an abrasion resistant dry lubrication should not be additionallylubricated.
Threads without lubricant must be lubricated with an acid-free, 
non-hardening grease.
• If spacing is reduced or the dielectric strength of the measuring device
cannot be guaranteed during repair work, perform a test on
completion of the work (e.g. high-voltage test in accordance with the manufacturer'sinstructions).
• Service connector:– do not connect in potentially explosive atmospheres.
– only connect to Endress+Hauser service devices.
• Observe the instructions for transporting and returning the device
outlined in the Operating Instructions.
• If you have any questions, contact your E+H service organization.

Tool symbols
SymbolMeaning
Phillips head screwdriver

A0011219

Mounting the ground terminal
(as example: housing T14)
12

Terminal1

2

Terminal2

Endress + Hauser

Analyse

Einbauanleitung

Erdungsklemme
Cerabar S, Deltabar S, Deltabar, Cerabar M, Deltabar M, Deltapilot M
EA01131P/00/A2/01.15
71293523

Die Einbauanleitung ist für das folgende Ersatzteilset gültig:
Bestell-Nummer

Original Ersatzeilset

71290825

Erdungsklemme T14/T15/T17, F30/F31

Wir empfehlen Einbauanleitung und Verpackung immer zusammen aufzubewahren.

Bestimmungsgemäße Verwendung
Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit gegen eine funktionierende Einheit des gleichen Typs zu ersetzen. Es dürfen nur Originalteile von Endress+Hauser verwendet werden.
Grundsätzlich dürfen nur Ersatzteilsets verwendet werden, die von Endress+Hauser für das Messgerät vorgesehen sind. Die Überprüfung ist via W@M Device Viewer durchzuführen, die Vorgehensweise dazu ist nachfolgend beschrieben.
Bei einigen Messgeräten befindet sich im Anschlussraumdeckel eine Ersatzteilübersicht. Ist das Ersatzteilset dort aufgelistet, entfällt die Überprüfung.

1.

www.endress.com/deviceviewer

www.endress.com/deviceviewer
Seriennummer (Ser. No.) eingeben, Produktdaten suchen
und auf Ersatzteile klicken.

2.

Ser. No.:12345

Anzeige aller Ersatzteile zum Messgerät

3.

Die Bestellnummer des Ersatzteilsets ermitteln.

4.

=?

Stimmt die Bestellnummer des Ersatzteilsets mit einer der Bestellnummern inder Ersatzteilliste überein?
JA, das Ersatzteilset darf für das Messgerät verwendet werden.
NEIN, das Ersatzteilset darf nicht für das Messgerät verwendet werden.

Reparaturberechtigten Personen
Die Berechtigung zur Durchführung einer Reparatur ist von der Zulassung des Messgeräts abhängig. Die Tabelle zeigt den jeweils berechtigten
Personenkreis.
Die Person, die eine Reparatur vornimmt, übernimmt die Verantwortung für die Sicherheit während der Arbeiten, die Qualität derAusführung und die Sicherheit des Geräts nach der Reparatur.
Zulassung des Messgeräts

Reparaturberechtigter Personenkreis

ohne Zulassung

1, 2, 3

mit Zulassung (z.B. IECEx, ATEX, FM, CSA, TIIS, NEPSI)

1, 2, 3

1 = Ausgebildete Fachkraft des Kunden, 2 = Von Endress+Hauser autorisierter Servicetechniker, 3 = Endress+Hauser (Messgerät an Hersteller zurücksenden)Erdungsklemme

Sicherheitshinweise• Prüfen, ob das vorliegende Ersatzteil zur Kennzeichnung auf dem
Messgerät passt, siehe Kapitel "Bestimmungsgemäße Verwendung".
• Ersatzteilset und Einbauanleitung dienen dazu, eine defekte Einheit
gegen eine funktionierende Einheit des gleichen Typs zu ersetzen. Nur Originalteile von Endress+Hauser verwenden.
• Bei Ex-zertifizierten Messgeräten: Sicherheitshinweise im separaten
Dokument "Saftey Instructions" (XA, ZD) beachten.
Auf dem Typenschild ist angegeben, welche zusätzlichen
Sicherheitshinweise für das jeweilige Gerät relevant sind.
• Messgerät unter Spannung! Lebensgefahr durch Stromschlag.
Messgerät nur im spannungslosen Zustand öffnen.
• Vor einem Geräteausbau: Prozess in sicheren Zustand bringen undLeitung von gefährlichen Prozessstoffen befreien.
• Verbrennungsgefahr durch heiße Oberflächen! Vor Arbeitsbeginn:
Anlage und Messgerät auf berührungssichere Temperatur abkühlen.
• Bei Messgeräten im abrechnungspflichtigen Verkehr: Nach Entfernen der Plombe ist der geeichte Zustand aufgehoben.
• Nationale Vorschriften bezüglich der Montage, elektrischen
Installation, Inbetriebnahme, Wartung und Reparatur einhalten.
• Folgende Anforderungen an das Fachpersonal für Montage,elektrische Installation, Inbetriebnahme, Wartung und Reparatur derMessgeräte müssen erfüllt sein:– In Gerätesicherheit ausgebildet– Mit den jeweiligen Einsatzbedingungen der Geräte vertraut.
– Bei Ex-zertifizierten Messgeräten zusätzlich im Explosionsschutzausgebildet.
• Beschädigungsgefahr elektronischer Bauteile! Eine ESD-geschützte
Arbeitsumgebung herstellen.

• Die Betriebsanleitung zum Messgerät ist zu beachten.
• Nach Entfernen der Elektronikabdeckung: Stromschlaggefahr durchaufgehobenen Berührungsschutz! Messgerät ausschalten, bevor interne Abdeckungen entfernt werden.
• Änderungen am Messgerät sind nicht zulässig.
• Bei Messgeräten in sicherheitstechnischen Applikationen gemäß IEC 61508 bzw. IEC 61511: Nach Reparatur Neuinbetriebnahme
gemäß Betriebsanleitung durchführen. Reparatur dokumentieren.
• Gehäuse nur kurzzeitig öffnen. Eindringen von Fremdkörpern,
Feuchtigkeit oder Verunreinigung vermeiden.
• Defekte Dichtungen nur durch Original-Dichtungen vonEndress+Hauser ersetzen.
• Defekte Gewinde erfordern eine Instandsetzung des Messgeräts.
• Gewinde (z.B. von Elektronikraum- und Anschlussraumdeckel)
müssen geschmiert sein, sofern keine abriebfeste Trockenschmierungvorhanden ist. Säurefreies, nicht härtendes Fett verwenden• Wenn bei den Reparaturarbeiten Abstände reduziert oder die
Spannungsfestigkeit des Messgeräts nicht sichergestellt werden kann: Prüfung nach Abschluss der Arbeiten durchführen (z.B. Hochspannungstest gemäß Herstellerangaben).
• Servicestecker:– nicht in explosionsfähiger Atmosphäre anschließen.
– nur an Servicegeräte von Endress+Hauser anschließen.
• Die in der Betriebsanleitung aufgeführten Hinweise zum Transport
und zur Rücksendung beachten.
• Bei Fragen kontaktieren Sie bitte Ihre zuständige Endress+Hauser
Serviceorganisation.

Werkzeugsymbole
SymbolBedeutungKreuzschlitzschraubendreher

A0011219

Montage der Erdungsklemme
(Beispiel: Gehäuse T14)
1

2

Terminal1

2

Terminal2

Endress + Hauser