Spectrophotometer

Advanced color management with simple operation.
Introducing a next-generation color management tool.

Wireless connection for greater flexibility
Wireless connection untethers you from the computer for improved handling and stress-freemeasurements and data transfer.

Data under multiple conditions from just one scanA single scan can provide data under M0, M1, and M2 illumination conditions or other user-definedconditions* to improve printer operating efficiency.
* Separate software required.

Helps improve printing quality
Helps improve color reproduction quality to contribute to acquiring high added value printing work.

Innovative, user-friendly & accurateIncreases the efficiency of day-to-day printer color adjustment by allowing you to take measurements
where you want to, such as right where the printed materials come out.
12Print

3

ScanThe calibration cap can be stored under theinstrument when taking measurements.

Main specifications

Data transfer

Data can be transferred wirelessly to acomputer.

System Diagram

Spectrophotometer MYIRO-1
ModelMY-1
45°a: 0°(annular illumination)*1
Illumination/Conforms to CIE No. 15, ISO 13655, DIN5033 Teil 7, ASTM E 1164,
viewing system
and JIS Z 8722 Condition a for reflectance measurements.
Spectral separation device Concave grating
Spectral reflectance: 380 to 730 nm;
Wavelength range
Spectral irradiance: 360 to 730 nm
Wavelength pitch
10 nm
Half bandwidth
Approx. 10 nm
Measurement area
3.5 mmLight source
LEDMeasurement range
Density: 0.0D to 2.5D; Reflectance: 0 to 150%
Colorimetric: Within σ ΔE00 0.05
Short-term repeatability(When white plate is measured 30 times at 10-second intervals after white

Spectrophotometer
MY-1

Calibration Cap
MY-A01

Soft CaseMY-A03
USB Cable(2m)IF-A41

Ruler
MY-A02
Spectrophotometer Configuration Tool
MY-CT1*Avallable for download on the web

calibration has been performed)

Within ΔE00 0.3(Average of 12 BCRA Series II color tiles compared to values measured
with a master body under manufacturer's standard conditions)

Measurement
conditions *2
Observers
Status indicatorInterface
Scanning measurements
PowerDimensions (W×D×H)
WeightOperating temperature/ humidity range
Storage temperature/ humidity range

Ambient Light adaptor
MY-A04

Approx. 1s
M0 (CIE Illuminant A), M1 (CIE Illuminant D50),
M2 (illumination with UV-cut),User-defined illuminant
2° Standard Observer, 10° Standard observer
LED to indicate instrument status
Wireless LAN (802.11 b/g/n); USB2.0
Scanning measurement of a color chart can be performed.

Dimensions(unit : mm)

73

Measurement time(single-point measurement)

Optional accesories(Values under all illumination conditions can be obtained with single scan)
171

USB bus power; Rechargeable internal battery
171 mm × 73 mm × 71 mm
Approx. 340 g
10 to 35°C, 30 to 85% relative humidity with no condensation

71Inter-instrumentagreement

0 to 45°C, 0 to 85% relative humidity with no condensation*1 Illumination for wavelengths under 400 nm is unidirectional.
*2 M0, M1, M2: Illumination conditions defined in ISO 13655 4.2.2 Illumination requirements and measurement.

SAFETY PRECAUTIONS
For correct use and for your safety, be sure to read the instruction manual before using the instrument.

• Displays shown are for illustration purpose only.
• The specifications and drawings given here are subject to change without priornotice.

For additional information, please visit:https://www.myiro.com• Always connect the instrument to the specified power supply voltage.
Improper connection may cause a fire or electric shock.

• Sales contact:

9242-ACJ1-41

BJHPK 1