Product Information

PIX-RAY TUBE
D-0510D-0510S
D-0510SB
Stationary Anode X-ray Tube


Especially designed for dental radiography unit.

Provided with an insulation cylinder (D-0510S)
and lead cylinder. (D-0510SB)

These tubes have 0.5 focus, and are available for
maximum tube voltage 100 kV.

Installed in the same enclosure with the highvoltage transformer.

General Data
Electrical:
Circuit:High Voltage Generator ............................................. Constant Potential High-voltage Generator
Grounding .......................................................................................................... Center-grounded
Nominal X-ray Tube Voltage ................................................................................................... 100 kV
Nominal Focal Spot Value ............................................................................................................ 0.5
Nominal Anode Input Power (at 1.0s) .................................................................................... 1270 W
Nominal Radiographic Anode Input Power ............................................................................ 1270 W
Exposure Duty Cycle ..................................................................................................... 1:60 or More
(Exposure Time : Interval Time)

Mechanical:
Dimensions:Overall Length ........................................................................................ See Dimensional Outline
Maximum Diameter ................................................................................ See Dimensional Outline
Target:Anode Angle ............................................................................................................... 10 degrees
Material .......................................................................................................................... Tungsten
Inherent Filtration .................................................................................... At least 0.8 mm Al at 75 kV
X-ray Coverage .................................................................................. 160 × 340 mm at SID 550 mm★The information contained herein is presented only as a guide for the application of our products. No responsibility is assumed by Canon Electron Tubes& Devices Co., Ltd. (CETD) for any infringements of patents or other rights of the third parties which may result from its use.
No license is granted by implication or otherwise under any patent or patent rights of CETD or others.
★The information contained herein may be changed without prior notice. It is therefore, advisable to contact to CETD before processing with the design ofequipment incorporating this product.

D-0510 / D-0510S / D-0510SBWeight:
D-0510 .................................................................................................................... Approx. 320 g
D-0510S .................................................................................................................. Approx. 460 g
D-0510SB ............................................................................................................... Approx. 780 g
Cooling Method .............................................. Oil immersed (60℃ Max.) and convection oil cooling.
Tube Holding:D-0510 ....................................... Holding the glass envelope of the anode end and cathode end,or the screw of the anode shank.
D-0510S ......................................................................................... Holding the insulation cylinder
D-0510SB ...................................................... Holding the insulation cylinder or the lead cylinder

Absolute Maximum and Minimum Ratings(At any time, these values must not be exceeded.)Maximum X-ray Tube Voltage ................................................................................................ 100 kV
Between Anode (or Cathode) and Ground ................................................................................ 50 kV
Minimum X-ray Tube Voltage ................................................................................................... 50 kV
Maximum X-ray Tube Current ............................................................................... See Rating Charts
Constant Potential High-voltage Generator ......................................................................... 22 mA
Maximum Filament Current ........................................................................................................ 3.5 A
Maximum Filament Voltage ....................................................................................................... 4.2 V
Filament Frequency Limits ........................................................... DC or AC (Sine Wave) 0 ~ 20 kHz
Thermal Characteristics:
Anode Heat Content ............................................................................................................ 35 kJ
Maximum Anode Heat Dissipation ...................................................................................... 250 W
Maximum Radiographic Exposure Time ..................................................................................... 20 s

Environmental Limits
Operating Limits (In Dielectric Oil):
Oil Temperature ........................................................................................................... 10 ~ 60 ℃
Oil Pressure ............................................................................................................. 70 ~ 140 kPa
Shipping and Storage Limits:
Temperature .............................................................................................................. -40 ~ 70 ℃
Humidity ........................................................................................................................ 10 ~ 90 %(No Condensation)
Atmospheric Pressure .............................................................................................. 50 ~ 106 kPa-2-

D-0510 / D-0510S / D-0510SB

Maximum Rating Charts(Absolute maximum rating charts)
Constant Potential High-voltage Generator
Nominal Focal Spot Value: 0.525TUBE CURRENT [mA]

2050kV
1560kV
70kV
80kV
90kV
100kV

1050213572010

EXPOSURE TIME [s]Emission & Filament Characteristics
Constant Potential High-voltage Generator
Nominal Focal Spot Value: 0.5256TUBE CURRENT [mA]

580kV
100kV

154Ef

1035201
2.9

3.0

3.1

3.2

3.3

3.4

FILAMENT CURRENT [A]This graph indicates typical characteristics.
-3-3.5

FILAMENT VOLTAGE [V]50kV
20D-0510 / D-0510S / D-0510SBAnode Heating / Cooling Curve35225W

HEAT STORAGE [kJ]30175W

25201510
COOLING5HEATING00246TIME [min]-4-810D-0510 / D-0510S / D-0510SB

Dimensional Outline of D-0510
Unit: mm
M6 8DEEP

73 ±18O 15.7

138 ±310°

CENTRAL X-RAY

φ45 max.

14 max.

FOCAL SPOT

FC2.3)
20(φN

CENTRAL X-RAY
SHORT PIN

C1F-5-F: FILAMENTC1,C2 : CATHODE
N: NOT FOR USE

D-0510 / D-0510S / D-0510SB

Dimensional Outline of D-0510S
Unit: mm

M6 8DEEP

O 54
O 15.714310°

146 ±3

O 25

73 ±1

INSULATION CYLINDERCENTRAL X-RAY

6 max.

FOCAL SPOT

FC2.3)
20(φN

CENTRAL X-RAY
SHORT PIN

C1F-6-F: FILAMENTC1,C2 : CATHODE
N: NOT FOR USE

D-0510 / D-0510S / D-0510SB

Dimensional Outline of D-0510SB
Unit: mm

M6 8DEEP

O 58
O 54
O 15.714310°

146 ±340

O 25
O 30

73 ±1

INSULATION CYLINDERCENTRAL X-RAY

65

FOCAL SPOT

6 max.

LEAD CYLINDERFC2.3)
20φ(NCENTRAL X-RAY
SHORT PIN

C1F-7-F: FILAMENTC1,C2 : CATHODE
N: NOT FOR USEMarketing Engineering Group, Sales Department
1385, Shimoishigami, Otawara-shi, Tochigi 324-8550, Japan
Tel: +81-287-26-6666
Fax: +81-287-26-6060
https://etd.canon･The head office of Canon Electron Tubes & Devices Co., Ltd. has been certified to meet all the requirements of Environmental Management System ISO14001.
･Canon Electron Tubes & Devices Co., Ltd. has been certified to meet all the requirements of Quality Management Systems ISO9001 and ISO13485.
Product scope is referred to the following URL. https://etd.canon/eng/company/quality.htm.

PE-D-0510/S/SB 2018-11