SI-5780
IEC SYSTEM FOR MU f'UAL RECOGNI riON OF TESTCERTIFICATES FOR ELECTRICAL EQUIPMENr (IECEE)CB SCHEMESYSTEME CEI D"ACCEPTA TION MUTUELLE DE
CERTIFICATS D"ESSAIS DES EQlJIPEMENTS
ELECTRIQlJES ( IECEE) METHODE OC

CB TEST CERTIFICATECERTIFICAT D'ESSAI OCProductProduit

DIN Rail power supply for medical application

Name and address of the appli cantNom et adresse du demandeur

PULSGmbH
Arabellastr. 15, D-81925 MUnchen, Germany

Nam e and address of the manufacturer
Nom et adresse du fa bri cant

PULSGmbHElektrastr. 6, D-81925 MUnchen, Germany

Name and address of the factory
Nom et adresse de l' us ine

See page 2

Note: When more than one jact01y , please report on page 2
Note: Lorsque il y plus d'une usine, Feuillez utiliser Ia 2eme page

[8J Additional Information on page 2Ratings and principal characte ri sti cs
Val e urs nomin ales et caracteri stiques prin c ipa lesSee page 2

Trademark (if any)
Marque de fabrique (s i ell e ex iste)

PULS

Type of M anufacturer's Testin g Laboratori es used
Type de programme du laborato ire d'essais constructeur

I

Model I Type Ref.
Ref. De type

CP10.241-M1 and CP10.241-M1-78

Additional informati on (if necessary may also be reported
on page 2)
Les informations compl ementa ires (s i necessaire, peuvent
etre indiques sur Ia 2cmc pageThe risk management requirements of the standard were notaddressed.
[8J Additional Information on page 2

A sampl e of th e product was tested and fo undto be in conformi ty with
Un echantillon de ce produit a ete essaye et a ete
considere conforme a Ia

IEC 60601-1 :2005 (Third Edition)+ A1:2012

As shown in the Test Report Ref. No. whi ch fo rms
part of thi s Certificate
Comme indique dan s le Rapport d ' essa is numero derefere nce qui constitue parti e de ce Certifi cat

T221-0097/16 issued by laboratory SIQ Testing and Certification
GmbH

This CB Test Certificate is issued by the National Certification Body
Ce Certificat d' essai OC est etabli par I' O rganisme National de Certification

Sl~Date: 2016-12-20

2009-03

SIQ Ljubljana
Trtaska cesta 2, Sl-1000 Ljubljana , Slovenia
T +386 1 4778100, F +386 1 4778 444, info@siq.si, www.siq.si

Signature: Bojan Pecavar

1/2~----SI-5780
Ratings:
Input:
AC: 100-240 Vac; 3,3-1,4 A; 50-60Hz
DC: 110-150 Vdc; 3,0-2,2 A
CP10.241-M1 output:
24-28 V de; 12-1 0,3 A (below +45°C)
24-28 Vdc; 10-8,6 A (below +60°C)
24-28 Vdc; 7,5-6,5 A (below +70°C)
CP10.241-M1-78 output:
24-28 Vdc; 10-8,6 A (below +60°C)
24-28 Vdc; 7,5-6,5 A (below +70°C)

Name and address of factories:1) PULS ELECTRONICS (SUZHOU) CO LTD
NO 1 RUI EN LANE XINGPU RD, SIP
SUZHOUJIANGSU CN-215021, CHINA
2) PULS INVESTICNI S R 0
PRAZSKA 5639
CZ-430 01 CHOMUTOV, CZECH REPUBLIC

Additional information (if necessary)
Information complementaire (si necessaire)

Date: 2016-12-20

2009-03

Signature: Bojan Pecavar

2/2(2~--SLOVENSKA

AKREDITACIJASIST EN ISO/ IEC 17065

CP·001

Certificate of Conformity
Number:C251-0129/16

Project file:

Product:

DIN-rail power supply for medical application

Type reference:

CP10.241-M1

Trademark:

PULSApplicant:

PULS GmbH
Arabellastrasse 15, DE-81925 Munchen, Germany

Manufacturer:

PULS GmbH
Arabellastrasse 15, DE-81925 Munchen, Germany

C20161277

Place of manufacture: PULS GmbH
Arabellastrasse 15, DE-81925 Munchen, Germany
This certificate is granted subject to the SIQ's rules on product certification . SIQ certifies theconformity of the products with the requirements of the listed standards.
Ratings:

AC input: 100- 240 V-; 50 - 60 Hz; 288 W max (at< 45°C)
DC input: 110-150 Vdc
DC output: adjustment range 24- 28 Vdc I 12- 10.3 A (at< 45°C)

Standard:

EN 60601-1-2:2015

Test report:

T251-0895/16 (2016-11-04)

Remarks:

This certificate shall apply to the products identical to the tested sample andshall remain valid for the period of 3 years until 2019-12-21 or until thevalidity date of the listed standards, whichever occurs earlier.

Date:

2016-12-21

Authorized signature:

Bojan Pecavar

Only integral publication of this certificate is allowed. This certificate may only be reproduced in its entirety and without any
changes. On request SIQ will give information about the validity of the certificate.

0w(00-.:tzI-SIQ Ljubljana, Trzaska cesta 2, Sl-1 000 Ljubljana, Slovenia
T +386 1 4778 100, F +386 1 4778 444, info@siq.si, www.siq.si