Publication: INFICON AB – nine69e4-b (1111) - All information can be modified without prior notice

TGF10

T RACER G AS F ILLER

User‘s manual
Manuel de l’utilisateur
Benutzerhandbuch
Manual del usario

TGF10

T RACER G AS F ILLER

User‘s manual

EN

Contents2

1.1.

General .......................................................................................................................................... 3

1.2.

Installation ..................................................................................................................................... 4

1.3.

Operation....................................................................................................................................... 5

1.4.

Pneumatic diagram ........................................................................................................................ 6

1.5.

Specifications ................................................................................................................................ 7

INFICON – User’s manual TGF10

EN

TRACER GAS FILLER TGF10
1.1.

GeneralTracer Gas Filler TGF10 evacuates and fills an
object subject to leak detection.TGF10 is anaccessory to Hydrogen Leak Detector Sensistor
ISH2000 and can be used together with most of
INFICON probes.
Evacuation and filling is time controlled by the
APC system in the Sensistor ISH2000 Leak
Detector.
TGF10 feature:
The air in the test object is pumped out by abuilt in venturi pump before tracer gas is
injected. This is done to make sure that thetracer gas completely fills the test object. Preevacuation is particularly important on very
long objects e.g. pipes or heat exchangers.

Sequence:1) Press the start button.
a) Test object is evacuated.
The “Pre Evac Time” is set and adjusted
by a timer in Sensistor ISH2000.
NOTE! Setting a timer equal to zero(0,0) means that entire line is skipped,and outputs are not changed.
b) Evacuation ends.
c) Test object is filled with tracer gas.
Filling pressure is controlled by thegasbottle regulator.
(Test object is ready for leak detection).
2) Press stop button.
a) Test object is evacuated.
The “Post Evac Time” is set by a timer in
Sensistor ISH2000.
b) Evacuation stops and test object isrefilled with air to atmospheric pressure

INFICON – User’s manual TGF103

EN
1.2. Installation
Independent leak test station:
1.

Connect TGF10 to Leak Detector
Sensistor ISH2000.

2.

Connect compressed air to compressed
air port.

3.

Connect Tracer gas port to gas bottle.

4. Connect a large hose (∅ 12 mm. or more) to the exhaust port. Remote end of hose should be placed ina ventilation stud carrying out of the building.
NOTE: Maximum length for a 12 mm hose is 5m. Use bigger diameter if longer hose is needed. Toohigh resistance in the exhaust hose will severely reduce capacity of venturi pump.
5.

Connect Test port to test object or fixture.

Adding leak detection equipment to an existing pressure decay station:
1 - 4 As above.
5

Connect test port to exhaust port of pressure decay tester.

NOTE. Test port of TGF10 is open to exhaust in standby mode. This means that the pressure decay
instrument can operate normally with the TGF10 instrument connected.

4

INFICON – User’s manual TGF10

EN

1.3.

Operation

Leak Detection Sequence:
1) Press the start button.
a) The “Evac” diode comes on and the test object is evacuated.
Pre fill evacuating time is set and adjusted by a timer in Sensistor
ISH2000.
b) The “Fill” diode comes on and the “Evac” diode goes off.
Note: The “Evac” diode goes off shortly after the “Fill” diodehas come on.
Test object is now filled with tracer gas. Filling pressure iscontrolled by the gas bottle regulator.

2) Search and locate leak(s).

3) Press stop button.
a) The “Evac” diode comes on and the test object is evacuated.
b) After test evacuation time is set by a timer in Sensistor
ISH2000.
c) Evacuation stops and Test object is refilled with air to
atmospheric pressure

INFICON – User’s manual TGF10

5

EN

1.4.

6

Pneumatic diagram

INFICON – User’s manual TGF10

EN
1.5.

Specifications

Connections:

4 ea. ISO G1/4”

Manometer:- 1 to + 9 bar

Supplies:
Tracer Gas:

5% Hydrogen in 95% Nitrogen 5% H2 / 95% N2

Tracer Gas Pressure:0 – 900 kPa

Compressed air supply:

400 – 800 kPa (4 – 8 bar, 58-116 PSI)

Compressed Air

Filtered to 40 mm(0 – 9 bar, 0-130 PSI)

Capacity:
Evacuation:

70 std l / min free flow

Ultimate vacuum:- 0.85 bar (-12,3 PSI)

Evacuation time:

0.7 s / l to - 0.5 bar (-7,2 PSI)
1.6 s / l to - 0.7 bar (-10,2 PSI)
3.0 s / l to - 0.8 bar (-11,6 PSI)

Filling:

600 std l / min free flow
Typically 1.0 s / l

Dimensions:
Weight:4,7 kgDementions:

100 x 275 x 205 mm

INFICON – User’s manual TGF10

7

EN

Disposal of product when taken out of service.
According to EU legislation, this product must be recovered forseparation of materials and may not be disposed of as unsorted
municipal waste.
If you wish you can return this INFICON product to the
manufacturer for recovery.
The manufacturer has the right to refuse taking back products thatare inadequately packaged and thereby presents safety and/or healthrisks to the staff. The manufacturer will not reimburse you for the
shipping cost.
Shipping address:
INFICON AB
Westmansgatan 49
582 16 Linköping
Sweden8

INFICON – User’s manual TGF10

TGF10D ISPOSITIF DE
R EMPLISSAGE DE G AZ DE
T RAÇAGE

Manuel de l’utilisateur

FR

Table des matières

2

1.1.

Généralités .................................................................................................................................... 3

1.2.

Installation ..................................................................................................................................... 4

1.3.

Fonctionnement ............................................................................................................................. 5

1.4.

Schéma ......................................................................................................................................... 6

1.5.

Caractéristiques ............................................................................................................................. 7

INFICON – Manuel de l’utilisateur TGF10FRDISPOSITIF DE REMPLISSAGE
DE GAZ DE TRAÇAGE TGF10
1.1.

Généralités

Le dispositif de remplissage de gaz de traçage
TGF10 évacue et remplit de gaz de traçage un
objet à tester afin qu’il soit ensuite possible d’ylocaliser des fuites. TGF10 est un accessoire dudétecteur de fuites d’hydrogène Sensistor
ISH2000 et peut être utilisé avec la plupart dessondes à hydrogène INFICON.

Séquence:
1) Appuyes sur le bouton “Start” (démarrer).
a) L’objet à tester est evacuee.
Le temps de pré-évacuation “PreEvac Time” est predefine et réglé parune minuterie dans le Sensistor
ISH2000.

Les temps d’évacuation et de remplissage sontprédéfinis par le système APC du détecteur defuites Sensistor ISH2000.
Fonction du TGF10:
L’objet du test est vidé de son air au moyen
d’une pompe venturi intégrée avant l’injection
du gaz de traçage. Ceci garantit une distribution
rapide du gaz de traçage dans toutes les parties
de l’objet du test. La pré-évacuation est
particulièrement importante sur les objets très
longs tels que les tuyaux ou les échangeurs dechaleur.
Lisez attentivement ce guide et tous les guides
connexes avant toute utilisation de l’appareil.

REMARQUE ! Lorsqu’une minuterie
est réglée sur une valeur nulle (0,0),
la totalité de la ligne est ignorée et lessorties ne sont pas modifiées.
b) L’évacuation est terminée.
c) L’objet du test est rempli de gaz detraçage. La pression de remplissage est
réglée sur le régulateur de la bouteille degaz.
(L’objet à tester est prêt pour la détection
de fuites).
2) Appuyez sur le bouton “Stop”
a)L’objet du test est évacué. Le temps depost-évacuation, “Post Evac Time”,
est réglé par une minuterie dansSensistor ISH2000.
L’évacuation est terminée et l’objet du test est ànouveau rempli d’air à la pression
atmosphérique.

INFICON – Manuel de l’utilisateur TGF103FR
1.2. Installation
Station d’essai d’étanchéité autonome:
1.

Connectez le TGF10 au détecteur defuites Sensistor ISH2000 (connecteur
25 pts).

2.

Raccordez l’air comprimé à l’orifice“Compressed Air” du TGF10

3.

Connectez la bouteille de gaz de traçage à l'orifice “Tracer gas” du TGF10.

4.

Raccordez un grand gros diamètre (∅ 12 mm ou plus) à l’orifice d’échappement “Exhaust” duTGF10. L’autre extrémité du tuyau doit être placée dans un circuit d’aération donnant sur l’extérieurdu bâtiment.

5.

REMARQUE : Le tuyau de 12 mm de diamètre ne doit pas dépasser 5 m de long. Utilisez undiamètre plus grand si un tuyau plus long est requis. Une résistance trop élevée dans le tuyau
d’échappement réduit considérablement les performances de la pompe venturi.

6.

Connectez la prise d’essai “Out” à l’objet ou à la fixation du test.

Ajout d’un appareil de détection de fuites à un équipement de chute de pression :1- 4 Voir la procédure ci-dessus.
5

Connectez la prise d’essai à l’orifice d’échappement de l’équipement de chute de pression.

REMARQUE. La prise d’essai « Out » du TGF10 est ouverte sur l’échappement en mode veille. Cela
signifie que l’équipement de chute de pression peut fonctionner normalement lorsque l’appareil TGF10
est connecté
Air comprimé

L’objetdu test

Test de chute

Air compriméde pression

ÉchappementÉ

Gaz de traçage

4

INFICON – Manuel de l’utilisateur TGF10

FR
1.3.

Fonctionnement

Séquence de détection des fuites:

1) Appuyez sur le bouton “Start”a) Le voyant d’évacuation ”Evac” s’allume et l’objet du test est
èvacué.
Le temps d’évacuation de pré-remplissage est predefine et réglé
par une minuterie dans le Sensistor ISH2000.
b) Le voyant de remplissage “Fill” s’allume et le voyantd’évacuation “Evac” s’éteint. (Remarque : Le voyantd’évacuation “Evac” s’éteint peu après l’allumage du voyant deremplissage “Fill”).
L’objet du test est à présent rempli de gaz de traçage.
La pression de remplissage est réglée sur le détendeur de labouteille de gaz.

2) Recherchex et localisez les fuites éventuelles.

3) Appuyez sur le bouton “Stop”a) Le voyant d’évacuation “Evac” a’allume et l’objet du test est
evacuee.
Le temps d’évacuation après essai est predefine par une minuteriedans le Sensistor ISH2000.
b) L’évacuation est terminée et l’objet du test est à nouveau remplid’air à la pression atmosphérique.

INFICON – Manuel de l’utilisateur TGF10

5

FR
1.4.

6

SchémaINFICON – Manuel de l’utilisateur TGF10

FR

1.5. Caractéristiques
Raccordements :4 ISO G de 1/4”

Manomètre :- 1 à + 9 bar

Fournitures :Gaz de traçage :5% d’hydrogène dans 95% d’azote (5% H2 / 95% N2)

Pression du gaz de traçage:0 – 900 kPa (0 – 9 bar, 0-130 PSI)

Air comprimé :400 – 800 kPa (4 – 8 bar, 58-116 PSI)

Air comprimé :filtré à 40 microns

Capacité :Évacuation :70 l/min std. flux libre

Vide final :- 0,85 bar (-12,3 PSI)

Temps d’évacuation :0,7 s/l à -0,5 bar (-7,2 PSI)
1,6 s/l à -0,7 bar (-10,2 PSI)
3,0 s/l à -0,8 bar (-11,6 PSI)

Remplissage :600 l/min std flux libre ;type 1,0 s/l

Remarque : La capacité est fonction du raccordement à l’objet du test. Les valeurs ci-dessus sont basées
sur un tuyau d’une longueur de 1,5 m et d’un diamètre intérieur de 5,5 mm.

Autres caractéristiques :Poids :

4,7 kgDimensions :100 x 275 x 205 mm

INFICON – Manuel de l’utilisateur TGF10

7

FR

Recyclage d’un produit à mettre au rebut
Conformément à la Directive EU relative aux déchets électriques etélectroniques, ce produit doit être recyclé en fin de service et ne peut
donc pas être jeté comme déchet domestique.
Vous pouvez, si vous le désirez, retourner ce produit INFICON au
fabricant qui assumera la responsabilité de son recyclage.
Le fabricant retient le droit de refuser de reprendre un produit dontl’emballage pourrait représenter un risque pour la santé et/ou lasécurité de son personnel.
Le fabricant ne compensera pas vos frais d’expédition.
INFICON AB
Westmansgatan 49
582 16 Linköping
Suède

8

INFICON – Manuel de l’utilisateur TGF10

TGF10S PURENGASFÜLLEINHEIT

Benutzerhandbuch

DE

Inhalt2

1.1.

Allgemeines .................................................................................................................................. 3

1.2.

Installation ..................................................................................................................................... 4

1.3.

Betrieb ........................................................................................................................................... 5

1.4.

Schaltplan ...................................................................................................................................... 6

1.5.

Technische Daten .......................................................................................................................... 7

INFICON – Benutzerhandbuch TGF10DE

SPURENGASFÜLLEINHEIT TGF10
1.1.

Allgemeines

Die Spurengasfülleinheit TGF10 evakuiert undbefüllt ein Objekt, an dem eine Lecksuche
durchgeführt werden soll. Der TGF10 ist ein
Zusatzgerät zum Wasserstoff-Lecksuchgerät
Sensistor ISH2000. Er kann zusammen mit
verschiedenen INFICON Messköpfen eingesetzt
werden.
Das Evakuieren und Befüllen erfolgt
zeitgesteuert durch das APC-System im
Lecksuchgerät Sensistor ISH2000.
Leistungsmerkmale des TGF10:
Die Luft im Testobjekt wird von einer
integrierten Venturi-Pumpe abgepumt, bevor
Spurengas eingefüllt wird. Dadurch wird
sichergestellt, dass das Testobjekt vollständig
mit Spurengas befüllt ist. Bei sehr langen
Objekten, beispielsweise Rohrleitungen oderWärmetauschern, ist eine Vorevakuierung
besonders wichtig.
Lesen Sie dieses Handbuch sowie alle weiteren
zugehörigen Handbücher vor dem Arbeiten mit
dem System aufmerksam durch.

Ablauf:1) Drücken Sie die Starttaste.
a.

Das Testobjekt wird evakuiert. Die„Vorevakuierungszeit“ wird von einem
Timer im Sensistor ISH2000 eingestellt.
HINWEIS! Wenn Sie einen Timer auf
Null (0,0) setzen, wird der betreffende
Schritt übersprungen, und die Ausgänge
werden nicht verändert.

b.

Die Evakuierung wird beendet.

c.

Das Testobjekt wird mit Spurengas
befüllt. Der Befüllungsdruck wird vom
Druckregler an der Flasche geregelt.
(Das Testobjekt ist bereit für die
Lecksuche.)2) Drücken Sie die Stopptaste.
a.

Das Testobjekt wird evakuiert. Die„Nachevakuierungszeit“ wird von einem
Timer im Sensistor ISH2000 eingestellt.

b.

Die Evakuierung wird gestoppt, und das
Testobjekt wird mit Luft unteratmosphärischem Druck befüllt.

INFICON – Benutzerhandbuch TGF103

DE
1.2. Installation
Als eigenständige Lecksuchstation:
1.

Schließen Sie den TGF10 an den
Lecksucher Sensistor ISH2000 an.

2.

Schließen Sie die Druckluft an den
Druckluftanschluss an.

3.

Schließen Sie den
Spurengasanschluss an die Gasflasche an.

4.

Schließen Sie einen Schlauch (∅ mindestens 12 mm) an den Abluftanschluss an. Das andere
Schlauchende sollte in einen Abluftkanal geführt werden, der aus dem Gebäude herausführt.

HINWEIS! Ein Schlauch mit einem Durchmesser von 12 mm darf höchstens 5 m lang sein. Falls Sieeinen längeren Schlauch benötigen, muss ein Schlauch mit einem größeren Durchmesser verwendet
werden. Bei zu hohem Widerstand im Abluftschlauch verringert sich die Leistung der Venturi-Pumpe
beträchtlich.

5.

Schließen Sie den Testanschluss an das Testobjekt oder die Montagevorrichtung an.

Verbindung mit einem vorhandenen Druckabfallmessgerät:
1-4

Wie oben.

5

Schließen Sie den Testanschluss des TGF10 an den Abluftanschluss des Druckabfall-messgerätsan.
HINWEIS. Der Testanschluss des TGF10 ist im Bereitschaftsbetrieb zum Abluftanschluss hingeöffnet. Das bedeutet, dass das Druckabfallmessgerät normal arbeitet, wenn der TGF10
angeschlossen ist.
Luftversorgung

Testobjekt

Druckabfallmessgerät

Druckluft

AbluftSpurengas

4

INFICON – Benutzerhandbuch TGF10

DE
1.3.

Betrieb

Lecksuche:
1)

Drücken Sie die Starttaste.
a. Die LED „Evac“ leuchtet, und das Testobjekt wird
evakuiert. Die „Evakuierungsdauer vor derBefüllung“ wird von einem Timer im Sensistor
ISH2000 eingestellt.
b. Die LED „Fill“ leuchtet, und die LED „Evac“erlischt.
(Hinweis: Die LED „Evac“ erlischt kurz nach
Aufleuchten der LED „Fill“.)Das Testobjekt wird jetzt mit Spurengas befüllt. Der
Befüllungsdruck wird vom Druckregler an der
Gasflasche geregelt.

2) Suchen und lokalisieren Sie Lecks

3) Drücken Sie die Stopptaste.
a. Die LED „Evac“ leuchtet, und das Testobjekt wird
evakuiert. Die Evakuierungsdauer nach der Prüfung
wird von einem Timer im Sensistor ISH2000
eingestellt.
b. Die Evakuierung wird gestoppt, und das Testobjekt
wird unter Luft mit atmosphärischem Druck befüllt.

INFICON – Benutzerhandbuch TGF10

5

DE
1.4.

6

Schaltplan

INFICON – Benutzerhandbuch TGF10

DE

1.5. Technische Daten
Anschlüsse:4 Stck. ISO G1/4“Manometer:-1 bis +9 bar

Versorgungsleitungen:Spurengas:

5 % Wasserstoff in 95 % Stickstoff 5% H2 / 95% N2

Spurengasdruck:0 - 900 kPa (0 - 9 bar, 0-130 PSI)

Druckluftzufuhr:

400 - 800 kPa (4 - 8 bar, 58-116 PSI)

Druckluft

Gefiltert auf 40 μm

Leistung:
Evakuierung:

70 Nl/min bei freiem Durchfluss

Endvakuum:-0,85 bar (-12,3 PSI)

Evakuierungszeit:

0,7 s/l bis auf -0,5 bar (-7,2 PSI)
1,6 s/l bis auf -0,7 bar (-10,2 PSI)
3,0 s/l bis auf -0,8 bar (-11,6 PSI)

Befüllung:

600 Nl/min bei freiem Durchfluss typisch 1,0 s/l

Hinweis: Die Leistungswerte hängen vom Anschluss an das Testobjekt ab. Die vorstehenden Werte
beziehen sich auf einen Schlauch mit einer Länge von 1,5 m und einem Innendurchmesser von 5,5 mm.

Abmessungen:
Gewicht:

4,7 kgAbmessungen:

100 x 275 x 205 mm

INFICON – Benutzerhandbuch TGF10

7

DE

Entsorgung des Gerätes
Gemäß geltender EU Gesetzgebung muss dieses Produkt einem
Materialrecycling zugeführt werden und darf nicht in den
kommunalen, unsortierten Müll entsorgt werden.
Zur fachgerechten Entsorgung können Sie dieses INFICON Produkt
an den Hersteller zurück schicken.
Dieser hat das Recht die Annahme von Produkten, die nicht
ordnungsgemäß verpackt sind und somit ein Sicherheits- bzw.
Gesundheitsrisiko für die Mitarbeiter darstellen, zu verweigern.
In diesem Fall wird Ihnen der Hersteller die Frachtkosten nicht
erstatten.
Lieferadresse:
INFICON AB
Westmansgatan 49
582 16 Linköping
Schweden

8

INFICON – Benutzerhandbuch TGF10

TGF10

A LIMENTADOR DE G AS
I NDICADORManual del usarioSPÍndice2

1.1.

General .......................................................................................................................................... 3

1.2.

Instalación ..................................................................................................................................... 4

1.3.

Functionamiento............................................................................................................................ 5

1.4.

Esquema ........................................................................................................................................ 6

1.5.

Especificaciones ............................................................................................................................ 7

INFICON – Manual del usario TGF10SP

ALIMENTADOR DE GASINDICADOR TGF10
1.1. General
El alimentador de gas indicador TGF10 evacua
y llena el objeto sometido a la detección defugas. El TGF10 es un accesorio del detector defugas por hidrógeno Sensistor ISH2000 y puede
usarse junto con la mayoría de las sondas dehidrógeno de Adixen. El tiempo de evacuacióny llenado se controla mediante el sistema APC
del detector de fugas Sensistor ISH2000.
Característica del TGF10:
Mediante una bomba Venturi incorporada se
extrae el aire del objeto de comprobación antes
de inyectar el gas indicador. Esto se hace para
asegurarse de que el gas indicador llena
completamente el objeto de comprobación. La
evacuación previa es particularmente importanteen objetos muy largos, por ejemplo, tuberías otermointercambiadores.
Lea esta guía y todos los manuales relevantes
antes de usar el equipo.

Secuencia:
1)

Pulse el botón de arranque.
a. Elobjeto de comprobación es evacuado. El
tiempo de evacuación previa (“Pre Evac
Time”) se fija y ajusta en el Sensistor
ISH200 mediante un temporizador.
¡NOTA! El ajuste del temporizador a cero(0,0) significa que se saltará la línea entera
y las salidas no habrán cambiado.
b. La evacuación finaliza.
c. El objeto de comprobación se llena de gas
indicador. La presión de llenado se controla
mediante el regulador de la botella de gas.
(El objeto de comprobación está listo para la
detección de fugas).

2) Pulse el botón de parada.
a.

El objeto de comprobación es evacuado.
El tiempo posterior a la evacuación(“Post Evac Time”) se ajusta en el
Sensistor ISH2000 mediante un
temporizador.
La evacuación se detiene y el objeto decomprobación se rellena con aire a presión
atmosférica.

INFICON – Manual del usario TGF10

3

SP
1.2.

Instalación

Estación de prueba de fugas independiente:
1. Conecte el TGF10 al detector de fugas
Sensistor ISH2000.
2. Conecte aire comprimido al puerto de airecomprimido.
3. Conecte el puerto del gas indicador a labotella de gas.
4. Conecte una manguera grande (∅ de 12 mm
o más) al puerto de exhaustación. El extremo
remoto de la manguera debe colocarse en un
soporte de ventilación que la lleve fuera deledificio.
NOTA: La longitud máxima de una manguera de
12 mm. es de 5 m. Use un diámetro superior si
necesita una manguera más larga. Una
excesiva resistencia en la manguera deexhaustación reducirá gravemente lacapacidad de la bomba Venturi. .
5. Conecte el puerto de prueba al objeto oaccesorio de comprobación.
Cómo agregar un equipo de detección de fugas a
una estación de disminución de preción
existente:
1-4
5Igual al anterior.

Conecte el puerto de prueba al puerto de
exhaustación del analizador de disminución depresión.
NOTA. El puerto de prueba del TGF10 estáabierto a la exhaustación en el modo deespera. Esto significa que el instrumento dedisminución de presión puede funcionar
normalmente con el instrumento TGF10
conectado.

Objetodecompro-Controlador dedisminución de lapresión

Aire comprimido

Airecomprimido
Exhaustación

Gas indicador

4

INFICON – Manual del usario TGF10

SP
1.3.

Functionamiento

Secuencia de detección de una fuga:
1)

Pulse el boutón de arranque.
a. El diodo „Evac“ se enciende y elobjeto de comprobación es
evacuado.
El tiempo de evacuación delprellenado se fija y ajusta en el
Sensistor ISH2000 mediante un
temporizador.
b. El diodo „Fill“ (llenar) se
enciende y el „el diodo evac“ seapaga.
(Nota: El diodo „Evac“ se apaga
brevemente después de que eldiodo « Fill » se haya encendido).
A continuación, el objeto decomprobación se llena de gas
indicador. La presión de llenado
se controla mediante el regulador
de la botella de gas.

2) Busque y localice la(s) fuga(s)3) Pulse el botón de parada.
a. El diodo “Evac” se enciende y elobjeto de comprobación es
evacuado. Después de la prueba,
el tiempo de evacuación se ajusta
en el Sensistor ISH2000 mediante
un temporizador.
b. La evacuación se detiene y elobjeto de comprobación se rellena
con aire a presión atmosférica.

INFICON – Manual del usario TGF10

5

SP
1.4.

6

Esquema

INFICON – Manual del usario TGF10

SP

1.5. Especificaciones

Conexiones:

4 ea. ISO G1/4”

Manómetro:- 1 a + 9 bar

Suministros:
Gas indicador:

5% hidrógeno en 95% de nitrógeno 5% H2 / 95% N2

Presión del gas indicador:

0 – 900 kPa (0 – 9 bar, 0-130 PSI)

Suministro de aire comprimido: 400 -800 kPa (4 - 8 bar, 58-116 PSI)
Aire comprimido:

filtrado a 40 mm

Capacidad:
Evacuación:

70 std l / flujo libre mínimo.

Vacío último:- 0,85 bar (-12,3 PSI)

Tiempo de evacuación:

0,7 s / l a –0,5 bar (-7,2 PSI)
1,6 s / l a - 0,7 bar (-10,2 PSI)
3,0 s / l a -0,8 bar (-11,6 PSI)

Llenado:600 std l / flujo libre mínimo

Típicamente

1,0 s / lNota: La capacidad depende de la conexión al objeto de comprobación. Las cifras dadas son para una
manguera de 1,5 m.con un diám. interior de 5,5 mm.

Dimensiones:Peso:

4,7 kgDimensiones:

100 x 275 x 205 mm

INFICON – Manual del usario TGF107SPDesecho del producto cuando esté fuera deservicio.
De acuerdo a la legislación de la UE, este producto debe serrecuperado para la separación de materiales y no debe ser desechadocomo un desperdicio municipal sin clasificar.
Si lo desea, puede devolver este producto INFICON al fabricantepara que realice su recuperación.
El fabricante tiene el derecho de rechazar productos que estén
empacados en forma inadecuada y que por lo tanto representen
riesgos de seguridad o salud para el personal.
El fabricante no le reembolsará los gastos de transporte.
Dirección de envío:
INFICON AB
Westmansgatan 49
582 16 Linköping
Suecia8

INFICON – Manual del usario TGF10INFICON AB, Box 76, SE-581 02 Linköping, Sweden
Phone: +46 (0) 13 35 59 00 Fax: +46 (0) 13 35 59 01
www.inficon.com E-mail: reach.sweden@inficon.com