Product Information

PIX-RAY TUBE
D-054
D-054S
D-054SB
Stationary Anode X-ray Tube


Especially designed for dental radiography unit.

Low target angle adaptive for dental tomography.

Provided with an insulation cylinder (D-054S) andlead cylinder. (D-054SB)



These tubes have 0.5 focus, and are available for
maximum tube voltage 100 kV.

Installed in the same enclosure with the highvoltage transformer.

General Data
Electrical:
Circuit:High Voltage Generator .......................................................... Two-peak High-voltage Generator
Constant Potential High-voltage Generator
Grounding .......................................................................................................... Center-grounded
Nominal X-ray Tube Voltage ................................................................................................... 100 kV
Nominal Focal Spot Value ............................................................................................................ 0.5
Nominal Anode Input Power (at 1.0s):
Two-peak High-voltage Generator ...................................................................................... 840 W
Constant Potential High-voltage Generator ....................................................................... 1750 W
Nominal Radiographic Anode Input Power:
Two-peak High-voltage Generator ...................................................................................... 840 W
Constant Potential High-voltage Generator ....................................................................... 1750 W
Exposure Duty Cycle ..................................................................................................... 1:60 or More
(Exposure Time : Interval Time)★The information contained herein is presented only as a guide for the application of our products. No responsibility is assumed by Canon Electron Tubes& Devices Co., Ltd. (CETD) for any infringements of patents or other rights of the third parties which may result from its use.
No license is granted by implication or otherwise under any patent or patent rights of CETD or others.
★The information contained herein may be changed without prior notice. It is therefore, advisable to contact to CETD before processing with the design ofequipment incorporating this product.

D-054 / D-054S / D-054SB

Mechanical:
Dimensions:Overall Length ........................................................................................ See Dimensional Outline
Maximum Diameter ................................................................................ See Dimensional Outline
Target:Anode Angle ................................................................................................................. 5 degrees
Material .......................................................................................................................... Tungsten
Inherent Filtration .................................................................................... At least 0.8 mm Al at 50 kV
X-ray Coverage .................................................................................... 75 × 380 mm at SID 550 mm
Weight:D-054 ...................................................................................................................... Approx. 320 g
D-054S .................................................................................................................... Approx. 460 g
D-054SB ................................................................................................................. Approx. 780 g
Cooling Method .............................................. Oil immersed (60℃ Max.) and convection oil cooling.
Tube Holding:D-054 ......................................... Holding the glass envelope of the anode end and cathode end,or the screw of the anode shank.
D-054S ........................................................................................... Holding the insulation cylinder
D-054SB ........................................................ Holding the insulation cylinder or the lead cylinder-2-D-054 / D-054S / D-054SB

Absolute Maximum and Minimum Ratings(At any time, these values must not be exceeded.)Maximum X-ray Tube Voltage ................................................................................................ 100 kV
Between Anode (or Cathode) and Ground ................................................................................. 50 kV
Minimum X-ray Tube Voltage ..................................................................................................... 50 kV
Maximum X-ray Tube Current: (See Rating Charts)
Two-peak High-voltage Generator ...................................................................................... 24 mA
Constant Potential High-voltage Generator ......................................................................... 22 mA
Maximum Filament Current ........................................................................................................ 3.5 A
Maximum Filament Voltage ....................................................................................................... 4.2 V
Filament Frequency Limits ........................................................... DC or AC (Sine Wave) 0 ~ 20 kHz
Thermal Characteristics:
Anode Heat Content ............................................................................................................ 35 kJ
Maximum Anode Heat Dissipation ...................................................................................... 250 W
Maximum Radiographic Exposure Time ..................................................................................... 20 s

Environmental Limits
Operating Limits (In Dielectric Oil):
Oil Temperature ........................................................................................................... 10 ~ 60 ℃
Oil Pressure ............................................................................................................. 70 ~ 140 kPa
Shipping and Storage Limits:
Temperature .............................................................................................................. -40 ~ 70 ℃
Humidity ........................................................................................................................ 10 ~ 90 %(No Condensation)
Atmospheric Pressure .................................................................................................50 ~ 106 kPa-3-D-054 / D-054S / D-054SB

Maximum Rating Charts(Absolute maximum rating charts)
Two-peak High-voltage Generator
Nominal Focal Spot Value: 0.550kV

2460kVTUBE CURRENT [mA]

2070kV
80kV

1590kV100kV10512357

1020EXPOSURE TIME [s]Constant Potential High-voltage Generator
Nominal Focal Spot Value: 0.580kV

2270kV60kV90kV

2050kVTUBE CURRENT [mA]

100kV

1510512357

EXPOSURE TIME [s]-4-1020

D-054 / D-054S / D-054SB

Emission & Filament Characteristics
Two-peak High-voltage Generator
Nominal Focal Spot Value: 0.525TUBE CURRENT [mA]

20

650kV

155100kV

104Ef

5

30
2.9

3.0

3.1

3.2

3.3

3.4

FILAMENT VOLTAGE [V]72
3.5

FILAMENT CURRENT [A]This graph indicates typical characteristics.

Constant Potential High-voltage Generator
Nominal Focal Spot Value: 0.5625TUBE CURRENT [mA]

205100kV
Ef

15410350kV

5202.9

3.0

3.1

3.2

3.3

3.4

FILAMENT CURRENT [A]This graph indicates typical characteristics.

-5-13.5

FILAMENT VOLTAGE [V]80kV

D-054 / D-054S / D-054SB

Anode Heating / Cooling Curve35225W

HEAT STORAGE [kJ]30175W

25HEATING
201510COOLING
500246

TIME [min]-6-810D-054 / D-054S / D-054SB

Dimensional Outline of D-054
Unit: mm

φ１５．７

７３±１（８）Ｍ６ ８ＤＥＥＰ

１３８±３５°ＣＥＮＴＲＡＬ
Ｘ－ＲＡＹ

φ４５ＭＡＸ

１４ＭＡＸ

ＦＯＣＡＬ ＳＰＯＴ

Ｃ２ＦＮ

３０．
φ２

ＣＥＮＴＲＡＬ Ｘ－ＲＡＹ
Ｃ１Ｆ

ＳＨＯＲＴ ＰＩＮ

Ｆ：ＦＩＬＡＭＥＮＴ
Ｃ１，Ｃ２：ＣＡＴＨＯＤＥ
Ｎ：ＮＯＴ ＦＯＲ ＵＳＥ-7-D-054 / D-054S / D-054SB

Dimensional Outline of D-054S
Unit: mm

φ５４
ＩＮＳＵＬＡＴＩＯＮ ＣＹＬＩＮＤＥＲ

Ｍ６ ８ＤＥＥＰ

φ２５７３±１

φ１５．７

１４３

１４６±３５°ＣＥＮＴＲＡＬ
Ｘ－ＲＡＹ

６ＭＡＸ

ＦＯＣＡＬ ＳＰＯＴ

３０．
φ２Ｃ２Ｆ
ＣＥＮＴＲＡＬ Ｘ－ＲＡＹ

ＮＣ１Ｆ

ＳＨＯＲＴ ＰＩＮ
Ｆ：ＦＩＬＡＭＥＮＴ
Ｃ１，Ｃ２：ＣＡＴＨＯＤＥ
Ｎ：ＮＯＴ ＦＯＲ ＵＳＥ-8-D-054 / D-054S / D-054SB

Dimensional Outline of D-054SB
Unit: mm

φ５８

φ５４
ＩＮＳＵＬＡＴＩＯＮ ＣＹＬＩＮＤＥＲ

Ｍ６ ８ＤＥＥＰ

φ３０６５１４３

１４６±３５°４０

φ２５７３±１３０

φ１５．７

ＣＥＮＴＲＡＬ
Ｘ－ＲＡＹ

ＦＯＣＡＬ ＳＰＯＴ

６ＭＡＸ

ＬＥＡＤ ＳＹＬＩＮＤＥＲ

３０．
φ２ＦＣ２

ＣＥＮＴＲＡＬ Ｘ－ＲＡＹ

ＮＦＣ１ＳＨＯＲＴ ＰＩＮ
Ｆ：ＦＩＬＡＭＥＮＴ
Ｃ１，Ｃ２：ＣＡＴＨＯＤＥＮ：ＮＯＴ ＦＯＲ ＵＳＥ-9-Marketing Engineering Group, Sales Department
1385, Shimoishigami, Otawara-shi, Tochigi 324-8550, Japan
Tel: +81-287-26-6666
Fax: +81-287-26-6060
https://etd.canon･The head office of Canon Electron Tubes & Devices Co., Ltd. has been certified to meet all the requirements of Environmental Management System ISO14001.
･Canon Electron Tubes & Devices Co., Ltd. has been certified to meet all the requirements of Quality Management Systems ISO9001 and ISO13485.
Product scope is referred to the following URL. https://etd.canon/eng/company/quality.htm.

PE-D-054/S/SB 2018-11