Eclipse 5®

Smart technology to ensure saturation

O2 is what we do.

All devices are variable in delivering consistent volumes of oxygen. The volume, timing of delivery in the firsthalf of inspiration, and purity all contribute to effective oxygenation. Eclipse 5, the most clinically-advanced
transportable oxygen concentrator, offers continuous and pulse dose oxygen delivery to address a wide varietyof patients’ needs. With powerful air separation technology to deliver up to 95% oxygen purity, the Eclipse 5 alsooffers proprietary features to ensure saturation:••autoSAT automatically adjusts the delivery of oxygen to the patient’s breathing rate
UltraSense breath detection and adjustable rise time ensure consistent timing of oxygen delivery

autoSAT®: Clinically Proven Oxygen Delivery.
Most portable oxygen concentrators (POCs) deliver a decreased bolus size as the patient’s breath rate increases.
To compensate with those devices, the user would need to manually increase the setting on the POC to ensure
they receive enough oxygen during activity.
The Eclipse 5’s unique design includes CAIRE’s proprietary autoSAT technology, which maintains a consistentpulse dose volume for every breath as the patient’s respiratory rate changes.

••

Designed to ensure the prescribed dosing ismaintained throughout delivery

O2 mL per bolus at setting 47060

Proven to maintain a patient’s mean oxygen
saturation (SpO2) over 90% during exertion at any
setting, enhancing the patient’s ability to sustain
an active lifestyle for improved overall physical andmental health150403020
10015202530

3540

Breaths per minuteEclipse 5 w/ autoSAT®

Traditional POC Pulse

The Eclipse 5 delivers the largest oxygen bolus compared to competitive devices.
The Eclipse 5 delivers the largest bolus compared to competitive devices.
200180160bolus size

140120100
806040200Setting 1

Setting 2

Setting 3
TMGCE® Zen-OTM

Setting 4
Respironics® Simply Go®

Setting 5

Setting 6Setting 7

O2 Concepts® Oxlife Independence®

Setting 8Eclipse 5®

Setting 9Adjustable Rise Time.
Adjustable rise time allows for longer or shorter oxygen delivery time, providing the same bolus of oxygenvolume and accommodating a greater patient population. The Eclipse 5 offers three different settings to provide
increased patient comfort and address the oxygen needs of different disease states.

Adjustable Rise Time Options
FASTMEDIUMSLOWTIME•••

Fast setting: offers quick delivery
Medium setting: offers slower delivery ofthe same bolus of oxygen
Slow setting: mimics continuous flowby delivering the bolus of oxygen over alonger period of time

UltraSense® with Adjustable Pulse Sensitivity.
UltraSense is CAIRE’s proprietary technology for sensitive breath detection—helping to ensure patients receive
oxygen when needed.

••

Adjustable Pulse Sensitivity allows pulse mode to be used by a wider variety of patients, including
patients with weak inspiratory effort who can use a more sensitive setting for more comfortable use
Provides quick and reliable oxygen delivery in the “golden third” of inhalation with the most advanced trigger
sensitivity on the market today

Why is sensitivity important?
The first phase of a patient’s inhalation is essential
for the gaseous exchange in the alveoli, known as
the “golden third” of inhalation. High sensitivity
is key to proper oxygen delivery within that earlyphase of inhalation. A delay in oxygen delivery
impedes oxygen absorption and gaseous
exchange in the lungs, which can cause thepatient to desaturate. 2Patients may experience an 11% drop in SpO2
with a Pulse Dose Oxygen Concentrator
Device (PDOCD) due to inadequate triggering
sensitivity settings. 3

1. A Comparative Study of 3 Portable Oxygen Concentrators During a 6-Minute Walk Test in Patients With Chronic Lung Disease Respir Care. LeBlanc C, Lavallee L, King J, TaylorSussex R, Woolnough A and McKim D. 2013;58(10):1598 –1605.
2. Effect of the Anatomic Reservoir on Low-Flow Oxygen Delivery Via Nasal Cannula: Constant Flow Versus Pulse Flow With Portable Oxygen Concentrator. Steven Zhou andRobert Chatburn. Respir Care 2014;59(8):1199 –1209.
3. Nocturnal Oxygenation Using a Pulsed-Dose Oxygen-Conserving Device Compared to Continuous Flow. Chatburn R, Lewarski J and McCoy R. Respir Care 2006;51(3):252–256.
Precision Medical® EasyPulse is a registred trademark of Precision Medical, Inc., a Pennsylvania corporation. Inogen One® G5/G3TM is a registered trademark of Inogen.
O2 Concepts® Oxlife Freedom® are registered trademarks of O2 Concepts. SimplyGo® Mini is a registered trademark of Koninklijke Philips N.V. DeVilbiss® and iGO®2 areregistered trademarks of DeVilbiss Healthcare. Data collected from product user manuals published on company websites, December 2020. The productcomparison isdesigned to demonstrate the technical advancements of the most common commercially available portable oxygen concentrators.

Find out more about Eclipse 5
and accessory products ateclipseoxygen.com

SEE PRODUCT WARRANTY STATEMENT FOR COMPLETE INFORMATION.
Please consult the applicable product instructions for use for product indications,contraindications, warnings, precautions, and detailed safety information.
© 2021 CAIRE Inc. All Rights Reserved. ML-CONC0215 A