SPEC SHEET

SGU

Universal Transmitter (1-output)
Features:

SGU (1-input 1-output) converts DC current, DC voltage,
thermocouple or RTD signal into isolated DC voltage/current.
Indication displays are equipped.

Functions:Sensor correction (Input value correction)
Normal/Reverse modeFilter time constantManual mode
Input and output types can be changed.
Output high and low limits

Model: SGU - ① ② - ③ - ④

Model name
Input 1
Output 1
Power supply
Option

RTD

Thermocouple

DC voltage② OUTPUT 1
4 to 20mA (Built-in 50Ω shunt resistor)
4 to 20mA (250Ω shunt resistor)
4 to 20mA (50Ω shunt resistor)
0 to 20mA (250Ω shunt resistor)
0 to 16mA (62.5Ω shunt resistor)
2 to 10mA (250Ω shunt resistor)
0 to 10mA (100Ω shunt resistor)
1 to 5mA (100Ω shunt resistor)
0 to 1mA (1000Ω shunt resistor)
10 to 50mA (10Ω shunt resistor)
0 to 10mV (Input resistance 1MΩ)
0 to 50mV (Input resistance 1MΩ)
0 to 60mV (Input resistance 1MΩ)
0 to 100mV (Input resistance 1MΩ)
0 to 1V (Input resistance 1MΩ)
0 to 5V (Input resistance 1MΩ)
1 to 5V (Input resistance 1MΩ)
-5 to 5V (Input resistance 1MΩ)
0 to 10V (Input resistance 1MΩ)
-10 to 10V (Input resistance 1MΩ)K

12345ABCDEFGDC Current

A0A1A2
A3A4A5A6A7A8A9V0
V1V2
V3V4V5V6V7V8V9K0
K1
K2
J0
J1
J2
R
S
B
E
T0T1N
PL
W5W3P0P1P2P3

DC Voltage

DC current① INPUT 1

4 to 20mA (allowable load resistance 750Ω max.)
0 to 20mA (allowable load resistance 750Ωmax.)※10 to 16mA (allowable load resistance 900Ωmax.)※1
2 to 10mA (allowable load resistance 1500Ω max.)
0 to 10mA (allowable load resistance 1500Ω max.)※1
0 to 10mV (allowable load resistance 10kΩ min.)※2
0 to 100mV (allowable load resistance 100kΩ min.)※2
0 to 1V (allowable load resistance 1000Ω min.)※2
0 to 5V (allowable load resistance 5000Ω min.)※2
1 to 5V (allowable load resistance 5000Ω min.)0 to 10V (allowable load resistance 10kΩ min.)※2-5 to 5V (allowable load resistance 10kΩ min.)※1: 0mA or less: Out of base accuracy.
※2: 0V or less: Out of base accuracy.

③ Power supply
01100 to 240V AC 50/60Hz
24V AC/DC 50/60Hz
For SGUL (with serial communication function),
only [0: 100 to 240V AC] is selectable.

④ Option

JRS
B
E

0

No option needed

1

Multi-rotation trimmer

2

Moisture-proof treatment

3

Multi-rotation trimmer + Moisture-proof treatment

T
N
PL-Ⅱ
W5Re/W26Re
W3Re/W25Re
Pt100
JPt100SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

SPEC SHEETPerformance
Base accuracy (at 25℃)

±0.1% of each input span
Thermocouple input: When input is 0℃ or less: Base accuracy ± 0.1% of each input span
When input has a decimal point: Base accuracy ± 0.05% of each input span
R, S input, -50 to 200℃ (-58 to 392˚F): ±0.3% of each input span,
B input, 0 to 300℃ (32 to 572˚F): Accuracy is not guaranteed.

Display accuracy
Temperature coefficient
Cold junction compensation accuracy
Effect of allowable lead wire resistance
Response time
Insulation resistanceDielectric strength

Base accuracy ±1 digit± 0.015%/℃ (0 to 10mV output:±0.02%/℃)±0.5℃ (1.0˚F) at 20±10℃
RTD input: Less than 20Ω per wire: Base accuracy, 20Ω or more per wire: Base accuracy + 0.005 %/Ω
0.5 sec. max. (0→90%)
100MΩ minimum, at 500V DC
2.0kV AC for 1 minute

General specifications

K, J, R, S, B, E, T, N, PL-Ⅱ, W5Re/W26Re, W3Re/W25Re
External resistance: 100Ω max. (However, B: 40Ω max.)
Pt100, JPt100
RTDInput detection current: Approx. 200 μA,Allowable lead wire resistance: 200Ω or less per wire
4 to 20mA, 0 to 20mA, 0 to 16mA, 2 to 10mA, 0 to 10mA, 1 to 5mA, 0 to 1mA,DC current
10 to 50mA DC
0 to 10mV, 0 to 50mV, 0 to 60mV, 0 to 100mV, 0 to 1V, 0 to 5V, 1 to 5V, -5 to 5V,
DC voltage
0 to 10V, -10 to 10V DC-5 to 5% (Adjustable from front panel)
Span adjustment range
95 to 105% (Adjustable from front panel)
100 to 240V AC 50/60Hz
85 to 264V AC
Allowable voltage range
24V AC/DC 50/60Hz
20 to 28V AC/DC
100 to 240 V AC: Approx. 9 VA max. (SGUL: Approx. 10 VA max.)
24 V AC: Approx. 6 VA max.
24 V DC: Approx. 3 W max.
-10 to 55℃ (Non-condensing, No icing), 35 to 85 %RH (Non-condensing)-10 to 60℃
Case: Flame-resistant resin, Black color, Panel: Polycarbonate
DIN rail22.5(W) x 89(H) x 70(D)mm (without socket)
Approx. 78g (without socket)
Thermocouple

Input

Zero adjustment range
Power supply
Power consumption
Operating temperature, humidity
Storage temperature
MaterialMounting method
External dimensions
WeightMeasurement range
Input code

Terminal arrangement
Measurement range

Resolution-1999 to 9999 ※4

1A0 to A9V0 to V9K0-200 to 1370 ℃※2-328 to 2498 ˚F※21℃(˚F)

K1
K2J0J1J2RSBET0T1N
PLW5W3P0P1P2
P3-200 to 200 ℃ ※1※20 to 400 ℃※2-200 to 1000℃※2-200 to 200 ※1※20 to 400 ℃※2-50 to 1760 ℃※2-50 to 1760 ℃※20 to 1820 ℃※2-200 to 800 ℃※2-200 to 400 ℃※2-100 to 100 ℃※2-200 to 1300℃※20 to 1390 ℃※20 to 2315 ℃※20 to 2315 ℃※2-200 to 650 ℃※2-100 to 100 ℃※2-200 to 500 ℃※2-100 to 100 ℃※2-328 to 392 ˚F※1※232 to 752 ˚F※2-328 to 1832 ˚F※2-328 to 392 ˚F※1※232 to 752 ˚F※2-58 to 3200 ˚F※2-58 to 3200 ˚F※232 to 3308 ˚F※2-328 to 1472 ˚F※2-328 to 752 ˚F※2-148 to 212 ˚F※2-328 to 2372 ˚F※232 to 2534 ˚F※232 to 4199 ˚F※232 to 4199 ˚F※2-328 to 1202 ˚F※2-148 to 212 ˚F※2-328 to 932 ˚F※2-148 to 212 ˚F※21℃(˚F)※31℃(˚F)※31℃(˚F)
1℃(˚F)※31℃(˚F)※31℃(˚F)
1℃(˚F)1℃(˚F)1℃(˚F)1℃(˚F)1℃(˚F)※31℃(˚F)
1℃(˚F)1℃(˚F)1℃(˚F)1℃(˚F)1℃(˚F)※31℃(˚F)
1℃(˚F)※3※1: ‘No decimal point’ and ‘1 digit after decimal point’ can be selected in [Decimalpoint place]. If ‘1 digit after decimal point’ is selected, the input low limit valueis -199.9.
※2: In the case of thermocouple or RTD input, ℃ or ˚F can be selected in [Input unit].
Please specify the input span. Minimum span is 50℃ or 100˚F.
※3: If ‘1 digit after decimal point’ is selected in [Decimal point place], it becomes 0.1.
※4: Decimal point position change and scaling are possible.

PWR ⑬ - ⑭
OUT ⑨ - ⑫
TC ① - ④
RTD ① - ④ - ⑤
A ①-④
V ①-④
CJA ④ - ⑤
RS-485

Power supply voltage

100 to 240V AC, 24V AC/DCOutput
Thermocouple input
RTD input
DC current input
DC voltage input
Cold junction compensator input
Serial communication RS-485(For SGUL)(8P Socket)

Circuit configuration+-SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20190507