SPEC SHEET

SGT

Thermocouple Transmitter (1-output)
Features:

SGT (1-input 1-output) converts thermocouple signal into isolated
DC voltage/current. Indication displays are equipped.

Functions:Sensor correction (Input value correction)
Normal/Reverse mode
Filter time constantManual mode
Input and output types can be changed.
Output high and low limits






Model: SGT - ① ② - ③ - ④

Model name
Input 1Output 1
Power supply
Option③ Power supply

Thermocouple① INPUT 1
K0
K1
K2
J0
J1J2RS
BET0T1N
PL
W5
W30K

1

100 to 240V AC 50/60Hz
24V AC/DC 50/60Hz
For SGTL (with serial communication function),
only [0: 100 to 240V AC] is selectable.

J④ Option

RSB
E
T

0No option needed

1

Multi-rotation trimmer

2Moisture-proof treatment

3Multi-rotation trimmer + Moisture-proof treatment

N
PL-Ⅱ
W5Re/W26Re
W3Re/W25Re

DC Voltage

DC Current② OUTPUT 1
12
3
45A
B
C
DEF
G

4 to 20mA (allowable load resistance 750Ω max.)
0 to 20mA (allowable load resistance 750Ω max.)※10 to 16mA (allowable load resistance 900Ω max.)※1
2 to 10mA (allowable load resistance 1500Ω max.)
0 to 10mA (allowable load resistance 1500Ω max.)※1
0 to 10mV (allowable load resistance 10kΩ min.)※2
0 to 100mV (allowable load resistance 100kΩ min.)※2
0 to 1V (allowable load resistance 1000Ω min.)※2
0 to 5V (allowable load resistance 5000Ω min.)※2
1 to 5V (allowable load resistance 5000Ω min.)0 to 10V (allowable load resistance 10kΩ min.)※2-5 to 5V (allowable load resistance 10kΩ min.)※1: 0mA or less: Out of base accuracy.
※2: 0V or less: Out of base accuracy.

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

SPEC SHEETPerformance
Base accuracy (at 25℃)

±0.1% of each input span
When input is 0℃ or less: Base accuracy ±0.1% of each input span.
When input has a decimal point: Base accuracy ±0.05% of each input span
R, S input, -50 to 200℃ (-58 to 392˚F): ±0.3% of each input span,B input, 0 to 300℃ (32 to 572˚F): Accuracy is not guaranteed.

Display accuracy
Temperature coefficient
Cold junction compensation accuracy
Response time
Insulation resistanceDielectric strength

Base accuracy ±1 digit±0.015%/℃ (0 to 10mV output: ±0.02%/℃)±0.5℃ (1.0˚F) at 20±10℃
0.5 sec. max. (0→90%)
100MΩ minimum, at 500V DC
2.0kV AC for 1 minute

General specifications
Input
Zero adjustment range
Power supply
Power consumption
Operating temperature and humidity
Storage temperature
Material
Mounting method
External dimensions
WeightK, J, R, S, B, E, T, N, PL-Ⅱ, W5Re/W26Re, W3Re/W25Re
External resistance: 100Ω max. (However, B: 40Ω max.)
Span adjustment range
95 to 105% (Adjustable from front panel)-5 to 5% (Adjustable from front panel)
100 to 240V AC 50/60Hz
Allowable voltage
85 to 264V AC
24V AC/DC 50/60Hzrange
20 to 28V AC/DC
100 to 240 V AC: Approx. 9 VA max. (SGTL: Approx. 10 VA max.)
24 V AC: Approx. 6 VA max.
24 V DC: Approx. 3 W max.
-10 to 55℃ (Non-condensing, No icing), 35 to 85%RH (Non-condensing)-10 to 60℃
Case: Flame-resistant resin, Black color, Panel: Polycarbonate
DIN rail22.5(W) x 89(H) x 70(D)mm (without socket)
Approx. 76g (without socket)

Measurement range
Inputcode
K0
K1
K2J0
J1
J2RSB
ET0T1N
PLW5
W3

Terminal arrangement
Measurement range ※1-200 to 1370 ℃-200 to 200 ℃ ※2
0 to 400 ℃-200 to 1000℃-200 to 200 ※2
0 to 400 ℃-50 to 1760 ℃-50 to 1760 ℃
0 to 1820 ℃-200 to 800 ℃-200 to 400 ℃-100 to 100 ℃-200 to 1300℃
0 to 1390 ℃
0 to 2315 ℃
0 to 2315 ℃-328 to 2498 ˚F-328 to 392 ˚F※232 to 752 ˚F-328 to 1832 ˚F-328 to 392 ˚F※232 to 752 ˚F-58 to 3200 ˚F-58 to 3200 ˚F
32 to 3308 ˚F-328 to 1472 ˚F-328 to 752 ˚F-148 to 212 ˚F-328 to 2372 ˚F
32 to 2534 ˚F
32 to 4199 ˚F
32 to 4199 ˚F

Resolution

PWR ⑬ - ⑭1℃(˚F)1℃(˚F)※31℃(˚F)※31℃(˚F)
1℃(˚F)※31℃(˚F)※31℃(˚F)
1℃(˚F)1℃(˚F)1℃(˚F)1℃(˚F)1℃(˚F)※31℃(˚F)
1℃(˚F)1℃(˚F)1℃(˚F)OUT ⑨ - ⑫
TC ① - ④
CJA ④ - ⑤
RS-485

Power supply voltage

100 to 240V AC, 24V AC/DCOutput
Thermocouple input
Cold junction compensation input
Serial communication RS-485(For SGTL)※1: ℃ or ˚F can be selected in [Input unit].
Please specify the input span. Minimum span is 50℃ or 100˚F.
※2: ‘No decimal point’ and ‘1 digit after decimal point’ can be selected in [Decimalpoint place]. If ‘1 digit after decimal point’ is selected, the input low limit valueis -199.9.
※3: If ‘1 digit after decimal point’ is selected in [Decimal point place], it becomes
0.1.

Circuit configuration(8P Socket)+-For SGTLSHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20190507