SureFlow™
Oxygen Flow Station

Create Your Multi-Patient Oxygen Flow Station with Ease
SureFlow™ is a uniquely engineered flow station that economicallyadministers medical-grade oxygen to those in need — from a single oxygen
source or oxygen concentrator.

For Diverse Applications
Serve up to five individuals per SureFlow unit and join clinicians and medics
around the world who rely on a single oxygen source to service multiple
patients at a time.
Two or more SureFlow units can be connected together to serve even moreindividuals simultaneously. Each flowmeter is adjusted separately to ensure
precise control with a visual indication of flow for safety and comfort.
Dual flow models of NewLife® Intensity 10 LPM concentrators provide evengreater flexibility in delivering oxygen by allowing SureFlow to operateconnected to one of the concentrator’s flowmeters while the secondary flowmetercan supply oxygen for a nebulizer treatment or a higher flow patient.
SureFlow units can even serve the low flow oxygen needs of pediatrics in acriticalcare nursery in underdeveloped areas. SureFlow is configured with 5 low flow
flowmeters, which display in 1/10 LPM increments (on 1 LPM flowmeters) and canbe used within a wide range of settings from 1/10 LPM to 1 LPM. SureFlow canbe ordered with either: five 1 LPM flowmeters (with 1/10 LPM increments), five 2
LPM flowmeters (with 1/8 LPM increments), or five 5 LPM flowmeters (with 1/2
LPM increments).

SureFlow Provides Oxygen for up to Five Patients per
Flowmeter Station from a Single Oxygen Source
Combined flows at the SureFlow station(s) cannot exceed the maximum
output capability of the oxygen source used as the supply to the station(s).
Additionally, if oxygen flows greater than 1 LPM per flowmeter are required, theSureFlow station can be ordered with either 1 LPM, 2 LPM, or 5 LPM flowmeters toaccommodate a variety of low flow and higher flow applications. This enables adults
and children alike to benefit in an emergency or incident management situation
where oxygen is necessary.

Features• Ideal for pediatric-to-adult use in hospitals, clinics, emergency shelters, and incident facilities• Can easily connect multiple flow stations together, resulting in the lowest oxygen cost per 			patient• Use ideally with NewLife Intensity 10-series units, VisionAire™ units, or any up-to-20 psig 			
(138 kPa) O2 source

www.caireinc.com

SEE PRODUCT WARRANTY STATEMENT FOR COMPLETE INFORMATION.
Please consult the applicable product instructions for use for productindications, contraindications, warnings, precautions, and detailed
safety information.
© 2024 CAIRE Inc. All Rights Reserved.

ML-CONC0001 D

SureFlowTM
Oxygen Flow Station

Positioning

The SureFlow oxygen flow station can be used on a table or desktop, or wallmounted. Pediatric and adult cannulas with up to 50 feet (15.2 m) of safety
tubing connect to any or all of the standard oxygen outlets on the flow station.
Longer lengths of tubing can be accommodated when SureFlow is being
supplied by a NewLife Intensity-series oxygen concentrator.
Mount or place the SureFlow station at proper eye level to easily adjust
SureFlow’s flowmeters accurately; back lines on the flowmeter ensure a proper
viewing angle. If two or more SureFlow units are connected together to serve
even more individuals simultaneously, be certain to allow adequate space toconnect the standard hose between the DISS oxygen fittings on either side ofthe unit. This is the same type of hose used to connect the SureFlow unit (fromthe opposite side) to the oxygen outlet on the oxygen concentrator.

Product Specifications
Flowmeters

Low flow flowmeters can be used from 1/10 LPM to 1 LPM with .1/10 LPM
increments with back lines for proper viewing angle.

Flowmeter Accuracy

0-1 LPM flowmeter ± 3% of full scale
0-2 LPM flowmeter ± 5% of full scale
0-5 LPM flowmeter ± 5% of full scale

Maximum Inlet Pressure

20 psig (138 kPa)

Dimensions (HxWxD)

5.8 in., 9.7 in., 7.2 in. (14.7 cm, 24.6 cm, 18.3 cm)

Weight3.28 lb (1.49 kg)
5.6 lb (2.5 kg) - shipping weight

Operating Temperature

5˚-40˚C (40˚-140˚F)

Storage Temperature-5˚-60˚C (30˚-140˚F)

WARRANTY: Three-year warranty
INTERNATIONAL APPROVALS: CE 0459 (FM069-2 only); ISO 13485:2016-certified.

Ordering Information
Description

SureFlow

SureFlow (1 LPM flowmeters) oxygen flow station with 7 ft (2.1 m) connection hose, 2outlet adapters, and user manual

FM069-1

SureFlow (2 LPM flowmeters) oxygen flow station with 7 ft (2.1 m) connection hose, 2outlet adapters, and user manual.

FM069-2

SureFlow (5 LPM flowmeters) oxygen flow station with 7 ft (2.1 m) connection hose, 2
outlet adapters, and user manual

FM069-5

Oxygen flow stations also can be specified with various flowmeter configurations at time of order.
Catalog Numbers reflect US English Part Numbers, for language-specific part numbers please contact Customer Service.

www.caireinc.com

SEE PRODUCT WARRANTY STATEMENT FOR COMPLETE INFORMATION.
Please consult the applicable product instructions for use for productindications, contraindications, warnings, precautions, and detailed
safety information.
© 2024 CAIRE Inc. All Rights Reserved.

ML-CONC0001 D