INSTRUCTION MANUAL

Digital Indicating Controller

ACS-13A

和文は裏面をご覧下さい。
No. ACS11JE13 2019.01

For detailed usage and options, refer to the full Instruction Manual for the ACS-13A.
Please download the full Instruction Manual from Shinko website.
http://www.shinko-technos.co.jp/e/
Support & Downloads
Downloads
Manuals
Thank you for purchasing our digital indicating controller ACS-13A. This manual contains instructions for themounting, functions, operations and notes when operating the ACS-13A. To prevent accidents arising from themisuse of this controller, please ensure the operator receives this manual.
Safety Precautions (Be sure to read these precautions before using our products.)
The safety precautions are classified into categories: “Warning” and “Caution”.
Warning: Procedures which may lead to dangerous conditions and cause death or serious injury, if not
carried out properly.
Caution: Procedures which may lead to dangerous conditions and cause superficial to medium injury or
physical damage or may degrade or damage the product, if not carried out properly.

Warning• To prevent an electrical shock or fire, only Shinko or other qualified service personnel may handle the inner assembly.
• To prevent an electrical shock, fire or damage to the instrument, parts replacement may only beundertaken by Shinko or other qualified service personnel.

Safety Precautions• To ensure safe and correct use, thoroughly read and understand this manual before using this instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify
correct usage after purpose-of-use consultation with our agency or main office. (Never use this instrument for
medical purposes with which human lives are involved.)• External protection devices such as protective equipment against excessive temperature rise, etc. must be
installed, as malfunction of this product could result in serious damage to the system or injury to personnel.
Proper periodic maintenance is also required.
• This instrument must be used under the conditions and environment described in this manual. Shinko TechnosCo., Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being
used under conditions not otherwise stated in this manual.

Caution with Respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the manufacture of weaponsof mass destruction (i.e. military applications, military equipment, etc.), please investigate the end users and thefinal use of this instrument. In the case of resale, ensure that this instrument is not illegally exported.

1. ModelACS – 1 3
Control action 3
A1,

Series name: ACS-13A (W48 x H48 x D62 mm)
PID
A
Alarm type can be selected by keypad. *1
R
Relay contact: 1aControl output
S
Non-contact voltage (for SSR drive): 12 V DC 15%(OUT1)A
Direct current: 4 to 20 mA DC
Input
M
Multi-range *2
100 to 240 V AC (standard)
Power supply voltage
1
24 V AC/DC *3
A2
Alarm 2 output (A2) *1
W(20A)CT rated current: 20 A (Single-phase)
HeaterW(50A)CT rated current: 50 A (Single-phase)burnout
OptionW3(20A)CT rated current: 20 A (3-phase)alarm
W3(50A)CT rated current: 50 A (3-phase)(Multiple options are selectable.)DR
Heating/Cooling control, Relay contact: 1a
Non-contact voltage: 12 V DC 15%
Control output (OUT2)DS
C5Serial communication (RS-485)
SM
Set value memory external selection*1: Alarm types (9 types and No alarm action) and Energized/De-energized can be selected by keypad.
*2: Thermocouple, RTD, Direct current and DC voltage can be selected by keypad.
*3: Power supply voltage 100 to 240 V AC is standard. When ordering 24 V AC/DC, enter “1” after the input code.
12. Name and Functions(1) PV indicator:(2) PV display:

(1)
(2)
(3)
(4)
(5)
(6)
(8)
(9)

(7)

(11)
(10)

(12)

Lights when PV is indicated in the PV/SV display mode.

Indicates the PV (process variable) or setting
characters during the setting mode.

(3) SV indicator:
Lights when SV is indicated in the PV/SV display mode.
(4) MEMO indicator: Lights when Set value memory external selection(SM option) is added.
(5) MEMO display: Indicates the set value memory number.
(6) SV display:
Indicates the SV (desired value), MV (manipulated
variable) or each set value during the setting mode.
(7) Action indicators
O1 (OUT1): Lights when control output (OUT1) is ON.
For direct current output type, flashes corresponding
to the MV in 250 ms cycles.
O2 (OUT2): Lights when control output (OUT2) (D option) is ON.
EV1: Lights when Alarm 1 output is ON.
EV2: Lights when Alarm 2 output (A2 option) is ON or when Heaterburnout alarm (W, W3 option) is ON.
AT: Flashes while AT (auto-tuning) or auto-reset is performing.
T/R: Lights during Serial communication (C5 option) (TX output).
LOCK: Lights when Lock 1, Lock 2 or Lock 3 is selected.
(8)
UP key:Increases the numeric value.
(9)
DOWN key: Decreases the numeric value.
(10)
Mode key: Selects the setting mode, or registers the set value.
To register the set (selected) value, press this key.
(11) OUT/OFF key: Switches control output ON/OFF or Auto/Manual control.

(Fig. 2-1)
(12) Console connector: By connecting to the USB communication cable (CMA, sold separately), the following
operations can be conducted from the external computer using the Console software SWS-ACS01M.
1 Reading and setting of SV, PID, various set values, 2 Reading of PV and action status, 3 Function change

3. Mounting to the Control Panel
3.1 Site Selection

CautionUse within the following temperature and humidity ranges.
Temperature: 0 to 50 (32 to 122 ), Humidity: 35 to 85 %RH (No icing and non-condensing)If the ACS-13A is installed through a control panel, the ambient temperature of the ACS-13A must be kept to under
50 . Otherwise the life of electronic parts (especially electrolytic capacitors) of the ACS-13A will be shortened.
This instrument is intended to be used under the following environmental conditions (IEC61010-1):
Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• Few mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of 0 to 50 (32 to 122 ) that does not change
rapidly, and no icing• An ambient non-condensing humidity of 35 to 85 %RH• No large capacity electromagnetic switches or cables through which large current is flowing• No water, oil, chemicals or the vapors of these substances can come into direct contact with the controller
3.2 External Dimensions (Scale: mm)
Gasket A

Mounting frame

Terminal cover (sold separately)(*) When Terminal cover (soldseparately) is used
(Fig. 3.2-1)
2

3.3 Panel Cutout (Scale: mm)

Caution45 +0.5075

45 +0.5
0

If horizontal close mounting is used for the controller, IP66 specification (Drip-proof/Dust-proof)
may be compromised, and all warranties will be invalidated.

45 +0.5
0n x 48-3 +0.5
0Horizontal close mounting
n: Number of mounted units(Fig. 3.3-1)

3.4 Mounting to and Removal from the Control Panel
3.4.1 How to Mount the ACS-13A
Mount the controller vertically to the flat, rigid panel to ensure it adheres to the Drip-proof/Dust-proof specification (IP66).
Mountable panel thickness: 1 to 5 mm(1) Insert the controller from the front side of the panel. (Fig.3.4.1-1)(2) Insert the mounting frame until it comes into contact with the panel, and fasten with the screw.
Tighten screws with one rotation upon the screw tips touching the panel. (Fig.3.4.1-2)
The torque is 0.05 to 0.06 N•m.
3.4.2 How to Remove the Mounting Frame and Unit (Fig. 3.4.2-1)(1) Turn the power to the unit OFF, and disconnect all wires before removing the mounting frame.
(2) Insert a flat blade screwdriver between the screw frame and unit 1 .
(3) Slowly push the frame upward using the screwdriver 2 , while pushing the unit toward the panel 3 .
(4) Repeat step (2) and slowly push the frame downward using the screwdriver for the other side.
The frame can be removed little by little by repeating these steps.
Gasket A(Fig.3.4.1-1) Mounting

4. Wiring

Mounting frame(Fig.3.4.1-2) Mounting(Fig.3.4.2-1) Removal

Warning
Turn the power supply to the instrument off before wiring or checking. Working on or touching theterminal with the power switched on may result in severe injury or death due to electrical shock.

Caution• The terminal block of this instrument is designed to be wired from the left side. The lead wiremust be inserted from the left side of the terminal, and fastened with the terminal screw.
The torque should be 0.63 N•m.
• This instrument does not have a built-in power switch, circuit breaker or fuse. It is necessary to installthem near the controller. (Recommended fuse: Time-lag fuse, rated voltage 250 V AC, rated current 2 A)
• For a 24 V AC/DC power source, do not confuse polarity when using direct current (DC).
• Use a thermocouple and compensating lead wire according to the sensor input
specifications of this controller.
• Use the 3-wire RTD according to the sensor input specifications of this controller.
• (+) side input terminal number of 0 to 5 V DC, 1 to 5 V DC, 0 to 10 V DC differs from that of 0 to 1 V DC.
(+) side input terminal number of 0 to 5 V DC, 1 to 5 V DC, 0 to 10 V DC: 9(+) side input terminal number of 0 to 1 V DC: 10• When using a relay contact output type, externally use a relay according to the capacity of
the load to protect the built-in relay contact.
• When wiring, keep input wires (thermocouple, RTD, etc.) away from AC sources or load wires.
3

4.1 Terminal Arrangement• EV1: Alarm 1 output• EV2: Alarm 2 output (A2 option)
or Heater burnout alarm output (W,W3 option)• O2: Control output (OUT2) (D option)• O1: Control output (OUT1)• DC: DC voltage, current inputs(For DC voltage input, + side terminal
number differs depending on the voltage
input.)• TC: Thermocouple input• RTD: Resistance temperature detector input• CT1: CT input 1 (W, W3 option)• CT2: CT input 2 (W3 option)• DI: Contact input (SM option)• RS-485: Serial communication (C5 option)(Fig. 4.1-1)

4.2 Lead Wire Solderless Terminal
Use a solderless terminal with an insulation sleeve in which an M3 screw fits as shown below. (Fig. 4.2-1)
The torque should be 0.63 N•m.
4.3 When Using a Terminal CoverWhen using a terminal cover (sold separately), pass terminal wires numbered 7 to 12 into the holes of theterminal cover. (Fig. 4.3-1)
Solderlessterminal

Ring-type

Model

Nichifu Terminal Industries CO.,LTD.
Japan Solderless Terminal MFG CO.,LTD.
Nichifu Terminal Industries CO.,LTD.
Japan Solderless Terminal MFG CO.,LTD.

5.8 mm max.

3.2 mm

5.8 mm max.

Y-typeManufacturer

Tightening
torqueTMEV1.25Y-3
VD1.25-B3A
0.63 N•m
TMEV1.25-3
V1.25-3

3.2 mm(Fig. 4.2-1)

Terminal cover(sold separately)(Fig. 4.3-1)

4.4 Heater Burnout Alarm (W, W3 option) Wiring
This alarm is not usable for detecting heater current under phase control.
Use the CT (current transformer) provided, and pass one lead wire of the heater circuit into the holeof the CT. (Fig. 4.4-1). When wiring, keep the CT wire away from AC sources or load wires to avoid
the external interference.
In the case of 3-phase (W3 option), pass any 2 lead wires of R, S, T into the CT, and connect them to
CT1 (13, 14) and CT2 terminals (14, 15). (Fig. 4.4-2)(13)
CT1 input terminals
(14)
Power
supplyCT

RS
T

Heater(Fig. 4.4-1)(Fig. 4.4-2)
4

Pass any 2 wires of R, S and T into CT.

5. Operation Flowchart
Power ONFor 3 seconds after power-on, the PV display indicates the input type, and the SV display
indicates input range high limit value(TC,RTD) or scaling high limit value(DC voltage, current).

PV/SV display mode(Automatic
control)

Control output OFF: (Approx. 1 sec)
Auto/Manual control: key

Main Setting ModeSV

Sub Setting Mode
Auto-tuning/Auto-reset+

Scaling low limit to
Scaling high limit:
AT/Auto-reset Cancel

SV2

/
:
AT/Auto-reset Perform

Control output OFF Manual control+(Approx. 3 sec)
Auxiliary Function Setting ModeSet value lock
: Unlock: Lock 1: Lock 2: Lock 3

Scaling low limit to
Scaling high limit

0 to 1000 (2000 )0.0 to 1000.0 (1999.9 )0.0 to 100.0% (DC input)

Unlock: All set values can be changed.
Lock1: None of the set values can be changed.
Lock2: Only main set mode can be changed.
Lock3: All set values except input type can be changed.
However, changed values revert to their previous
value after power-off because they are not saved in
the non-volatile memory. Do not change any setting
item in Setup mode. If any item in Setup mode ischanged, it will affect other setting items such as
the SV and Alarm value.

SV4

OUT2 proportional band

Sensor correction

Scaling low limit to
Scaling high limit

0.0 to 10.0 times (multipliedvalue of OUT1 P-band)-100.0 to 100.0( )-1000 to 1000 (DC input)

Integral time

Communication protocol

Scaling low limit to
Scaling high limit

SV3

OUT1 proportional band

: Shinko protocol: MODBUS ASCII: MODBUS RTU

0 to 1000 sec(Approx. 3 sec)

Output MV indication

Derivative time
0 to 300 sec

ARW

Instrument number

0 to 100%

0 to 95

OUT1 proportional cycle

Communication speed

Alarm 1, 2 value setting range
High limit alarm (Deviation setting):-(Input span) to Input span( ) *1
Low limit alarm (Deviation setting):-(Input span) to Input span( ) *1
High/Low limits alarm (Deviation setting):
0 to Input span( ) *1
High/Low limit range alarm (Deviation setting):
0 to Input span( ) *1
Process high alarm:
Input range low to Input range high limit *2
Process low alarm:
Input range low to Input range high limit *2High limit with standby alarm (Deviation setting):-(Input span) to Input span( ) *1
Low limit with standby alarm (Deviation setting):-(Input span) to Input span( ) *1
High/Low limits with standby (Deviation setting):
0 to Input span( ) *1

Heater burnout alarm
CT1 current value and characterare indicated alternately.

1 to 120 sec

OUT2 proportional cycle: 2400 bps: 4800 bps: 9600 bps: 19200 bps

1 to 120 sec

Alarm 1 value *3
Setting range: Refer to the left.

Alarm 2 value *3
Setting range: Refer to the left.

Data bit/Parity: 8-bit/No parity: 7-bit/No parity: 8-bit/Even: 7-bit/Even: 8-bit/Odd: 7-bit/Odd

Heater burnout alarm 2CT2 current value and characters
are indicated alternately.

Heater burnout alarm value*4*1: For DC input, the input span is the

0.0 to 20.0 A (Rated current: 20 A)
0.0 to 50.0 A (Rated current: 50 A)

same as the scaling span.
*2: For DC input, input range low (orhigh) limit value is the same as
scaling low (or high) limit value.
*3: Setting the value to 0 or 0.0 disables
the alarm (except Process high alarmand Process low alarm).
*4: Setting to 0.0 disables the alarm.

Heater burnout alarm 2
value *4

0.0 to 20.0 A (Rated current: 20 A)
0.0 to 50.0 A (Rated current: 50 A)

5

Stop bit:1:2

Characters used in this manual:
Indication
Number,

/

-1

0

1

2

3

4

5

6

7

8

9

A

B

C

D

E

F

G

H

I

J

K

L

M

N

O

P

Q

R

S

T

U

V

W

X

Y

Z

Indication
Alphabet
Indication
Alphabet(Table 5-1) Input type selection: [] (Default:
K-200 to 1370
K
-200.0 to 400.0
J-200 to 1000
R
0 to 1760
S
0 to 1760
B
0 to 1820
E-200 to 800
T-200.0 to 400.0N-200 to 1300
PL0 to 1390
C(W/Re5-26)
0 to 2315
Pt100-200.0 to 850.0
JPt100-200.0 to 500.0
Pt100-200 to 850
JPt100-200 to 500
4 to 20 mA-2000 to 10000
0 to 20 mA
0 to 1 V
0 to 5 V-2000 to 10000
1 to 5 V
0 to 10 V(Table 5-2) Alarm 1 type selection [] (Default

High limit alarmON

A 1 hysteresis

ON

OFF

OFF
A1 value+ A1 value

SV

High/ Low limits alarm

A 1 hysteresis

ON

OFF

A1 value

High/ Low limit range alarmONSV

A1 value+ A1 value

A 1 hysteresis

ONOFF
A1 valueSV

OFF

A1 valueA1 valueHigh limit with standby alarmON

A1 value

Low limit with standby alarm

A 1 hysteresis

OFFA 1 hysteresis

ON

OFF

OFF
A1 value+ A1 valueSV

A1 valueSV:Alarm 1 output is in. standby.

“A1” means Alarm 1.

High/Low limits with standby alarm

A 1 hysteresis

ON

Alarm
actionA1 value

Process low alarm

A 1 hysteresis

ON

OFF

SV

Process high alarm

A 1 hysteresis

Alarm
action: No alarm action)

Low limit alarm

A 1 hysteresis

Alarm
actionK, -200 to 1370 )K-320 to 2500
K
-320.0 to 750.0
J-320 to 1800
R
0 to 3200
S
0 to 3200
B
0 to 3300
E-320 to 1500
T
-320.0 to 750.0
N-320 to 2300
PL0 to 2500
C(W/Re5-26)
0 to 4200
Pt100-320.0 to 1500.0
JPt100-320.0 to 900.0
Pt100-320 to 1500
JPt100-320 to 900

For Alarm 2, read “A2” for “A1”.
6+ A1 value

A1 valueSV

A1 valuePV display indicates setting item characters, and SV display indicates default values.
Setting items with dotted lines are optional, and they appear only when the options are ordered.
Key operation (Use theor
key to set or select each setting item.)• If thekey is held down for 3 sec in any setting mode, the unit will revert to the PV/SV display mode.
•: This means that if the
key is pressed, the unit proceeds to the next setting mode.
•+: Press and hold theandkeys (in that order) together.
•+(Approx. 3 sec): Press and hold theandkeys (in that order) together for approx. 3 sec.
•+
(Approx. 3 sec): Press and hold theandkeys (in that order) together for approx. 3 sec.
+
(Approx. 3 sec)
Setup Mode
Input type
Alarm 1 typeContact input function
Selection item: See Table
5-1.

: Set value memory

Selection item: See Table 5-2.

external selection: OUT/OFF external

Scaling high limit

Scaling low limit to
Input range high limit

selection 1 (SV, SV2
switchable): OUT/OFF external
selection 2

Alarm 2 type
Selection item: See Table 5-2.

-2000 to 10000 (DC input)

Scaling low limit

Input range low limit to
Scaling high limit

Alarm1 Energized/De-energized: Energized-2000 to 10000 (DC input): De-energized

Decimal point place

Alarm2 Energized/De-energized: No decimal point: 1 digit after the point: 2-digit after the point: 3-digit after the point: Energized: De-energized

PV filter time constant

Alarm 1 hysteresis

0.0 to 10.0 sec

0.1 to 100.0(: Control output OFF: Auto/Manual control

1 to 1000 (DC input)

Alarm 2 hysteresis

OUT1 low limit to 100%

0.1 to 100.0

OUT1 low limit to 105% (A/M)

1 to 1000 (DC input)

OUT1 low limit

Alarm 1 delay time

0% to OUT1 high limit

0 to 10000 sec(

Backlight selection: All are backlit: PV display is backlit: SV display is backlit: Indicators are backlit: PV+SV display are backlit: PV+ Indicators are backlit: SV+ Indicators are backlit)-5% to OUT1 high limit (A/M)

OUT1 ON/OFF
hysteresis

Alarm 2 delay time

OUT2 action modeSV rise rate

0.1 to 100.0 ( )1 to 1000 (DC input): Air cooling: Oil cooling: Water cooling

OUT2 high limit
OUT2 low limit to 100%

OUT2 low limit

Overlap/Dead band

-100.0 to 100.0( )-1000 to 1000 (DC input)

OUT2 ON/OFF
hysteresis

0.1 to 100.0 ( )1 to 1000 (DC input)

PV color

0 to 10000 sec

0 to 10000 /min.( /min.)
0.0 to 1000.0 /min.( /min.)0 to 10000/min.

SV fall rate

0 to 10000 /min.( /min.)
0.0 to 1000.0 /min.( /min.)0 to 10000/min.

Direct/Reverse control

AT bias0 to 50 (0 to 100 )0.0 to 50.0 (0.0 to 100.0

SVTC bias

PV color range
0.1 to 100.0()

1 to 1000 (DC input)

Backlight time

Indication when output OFF)

Converted value of 20% of
input span
DC input: 20% of scaling span

7

: Green: Red: Orange: Alarm 1 or Alarm 2: ON
Green
Red
: Alarm 1 or Alarm 2: ON
OrangeRed: PV continuous change: PV continuous change+ Alarm 1 or Alarm 2 ON (Red)

0 to 99 minutes(0 minutes: Continuous): Reverse (Heating): Direct (Cooling)

0% to OUT2 high limit: Outputs OFF(4mA) or
OUT1(OUT2) low limit.
: Outputs a value between
OFF(4mA) and ON(20mA) or
between OUT1(OUT2) low limit
and OUT1(OUT2) high limit value
depending on the deviation.

OUT/OFF key function)

OUT1 high limit

Output status when
input errors occur: OFF indication: No indication: PV indication: PV+ Alarm output

OUT1 rate-of-change
0 to 100 %/sec

6. Running

After the unit is mounted to the control panel and wiring is completed, operate the unit following the procedure below.
(1) Turn the power supply to the ACS-13A ON.
(2) Set up the unit. (Refer to “5. Operation Flowchart”.) Setup should be done before using this controller: Setting the
Input type, Alarm type, Direct/Reverse control, etc. in Setup mode. If the user’s specification is the same as the default
value of the ACS-13A, it is not necessary to set up the controller. Proceed to Step (3).
(3) Input each set value. Input each set value. Refer to “5. Operation Flowchart” and “7. Basic Settings”.
(4) Turn the load circuit power ON. Control action starts so as to keep the control target at the SV.

7. Basic Settings

Main setting mode (When setting SV to 100 )Proceed to the Main setting mode.
(1)
Press thekey in the PV/SV
display mode. The unit proceeds
to the Main setting mode.
Set SV.
Set SV with the

(2)
,
(3)

(4)

or

key.

Register the SV.
Register the SV by pressing the
key. The unit reverts to the
PV/SV display mode.
Control starts.
Control starts so as to keep themeasuring temperature at 100 .

Auto-tuning Perform/Cancel (PID control)
Proceed to the Sub setting mode.
(1)
Press and hold the andkeys
in the PV/SV display mode.
The unit proceeds to the+
Sub setting mode.
Select AT Perform/Cancel.
(2)
Select AT Perform with thekey, or select AT Cancel with
the
key.
，
(3)
Confirm AT Perform/Cancel.
Press thekey. The unit
reverts to the PV/SV display
mode.
AT Perform/Cancel(4)
While AT is performing, the AT
indicator flashes, and it turns
off if AT is cancelled.

• In order to decide each P, I, D and ARW value automatically, the autotuning process has been made to fluctuate to get an optimal value.

• Sometimes the auto-tuning process will not fluctuate if auto-tuning is performed

8. Specificationsat or near room temperature. Therefore auto-tuning might not finish normally.

• Auto-reset is available for P or PD action. Auto-reset is cancelled inapprox. 4 minutes. It cannot be released while performing this function.

Power supply voltage: 100 to 240 V AC 50/60 Hz, 24 V AC/DC 50/60 Hz
Allowable voltage fluctuation: 100 to 240 V AC: 85 to 264 V AC, 24 V AC/DC: 20 to 28 V AC/DC
Accuracy (Setting and Indication):
Thermocouple input: Within 0.2% of each input span 1 digit, or within 2 (4 ), whichever is greater
However, R, S inputs, 0 to 200 (0 to 400 ): Within 6 (12 )B input, 0 to 300 (0 to 600 ): Accuracy is not guaranteed.
K, J, E, T, N inputs, less than 0 (32 ): Within 0.4% of input span 1 digit
RTD input: Within 0.1% of each input span 1 digit, or within 1 (2 ), whichever is greater
DC voltage, current inputs: Within 0.2% of each input span 1 digit
Control output (OUT1):
Relay contact: 1a, Control capacity: 3 A 250 V AC (resistive load)
1 A 250 V AC (inductive load cos =0.4),
Electrical life: 100,000 cycles
Non-contact voltage (for SSR drive): 12 V DC 15%, Max. 40 mA DC (short circuit protected)
Direct current: 4 to 20 mA DC, Load resistance, Max. 550
Alarm 1, Alarm 2 (A2 option), Heater burnout alarm output (W, W3 option):
Relay contact 1a, Control capacity: 3 A 250 V AC (resistive load), Electrical life: 100,000 cycles
Control output (OUT2) (Doption):Relay contact 1a, Control capacity: 3 A 250 V AC (resistive load), Electrical life: 100,000 cycles
Non-contact voltage (for SSR drive): 12 V DC 15%, Max. 40 mA DC (short circuit protected)
Contact input: Circuit current when closed: Approx. 12 mA
Power consumption: Approx. 8 VA
Ambient temperature: 0 to 50 (32 to 122 )Ambient humidity: 35 to 85 %RH (No icing and non-condensing)
Weight: Approx. 120 g
Accessories included: Instruction manual: 1 copy, Mounting frame: 1 piece
Gasket A (Front mounted to the ACS-13A): 1 piece
CT (Current transformer):CTL-6S: 1 piece [W (20A) option] CTL-12-S36-10L1U: 1 piece [W (50A) option]
CTL-6S: 2 pieces [W3 (20A) option] CTL-12-S36-10L1U: 2 pieces [W3 (50A) option]
Accessories sold separately: Terminal cover, USB communication cable (CMA)

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION

Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, Japan
URL:
http://www.shinko-technos.co.jp/e/Tel: +81-72-727-6100
E-mail:overseas@shinko-technos.co.jp
Fax: +81-72-727-7006

8