Instruction Manual

PAPERLESS RECORDERCOMMUNICATIONFUNCTIONS(RS-485 MODBUS/Ethernet)
TYPE: GR200

GR2CE1 2008.07

Note: MODBUS® is the trade mark or registered trade mark of AEG Schneider Automation International.
Note: GW-BASIC, Windows 98, MS-DOS and Internet Explorer are the trade mark or registered trademarks ofMicrosoft Corporation.
NOTICE
1. Exemption items from responsibility

The contents of this document may be changed in the future without prior notice.
We paid the utmost care for the accuracy of the contents. However, we are not liable for direct andindirect damages resulting from incorrect descriptions, omission of information, and use of
information in this document.

CONTENTS
1.

COMMUNICATION FUNCTIONS ·································································································· 1
1.1 General············································································································································ 1
1.2 Overview of MODBUS slave communication function (RS-485 interface)····································· 1
1.3 Overview of Ethernet communication functions·············································································· 22.

SPECIFICATIONS ···························································································································· 3
2.1 Communication Specifications········································································································ 33.

CONNECTION ································································································································· 4
3.1 Communication Terminal Allocation ······························································································ 4
3.2 Wiring············································································································································· 54.

SETTING OF COMMUNICATION CONDITION ··················································································· 6
4.1 Set Items ········································································································································· 6
4.2 Setting Operation Method ··············································································································· 75.

MODBUS COMMUNICATION PROTOCOL ·················································································· 8
5.1 General············································································································································ 8
5.2 Composition of Message ················································································································· 9
5.3 Response of Slave Station ············································································································· 10
5.4 Function Code······························································································································· 11
5.5 Calculation of Error Check Code (CRC-16) ·················································································· 12
5.6 Transmission Control Procedure ··································································································· 13
5.7 FIX Processing (Cautions in data write) ························································································ 15

6.

DETAILS OF MESSAGE················································································································ 16
6.1 Read-out of Word Data [Function code:03H]················································································· 16
6.2 Read-out of Read-out Only Word Data [Function code:04H]···························································· 18
6.3 Write-in of Continuous Word Data [Function code:10H] ······························································· 20

7.

ADDRESS MAP AND DATA FORMAT························································································· 22
7.1 Data Format ·································································································································· 22
7.1.1 Transmission data format ····················································································· 22
7.1.2 Control of decimal point ························································································ 23
7.1.3 Data with input error ··························································································· 23
7.1.4 Range of write-in data ·························································································· 23
7.2 Address Map ································································································································· 24
7.2.1 Word data [ read-out / write-in ] : Function code [ 03H, 10H ] ····································· 24
7.2.2 Word data [ read-out only ] : Function code [ 04H ] ··················································· 36
7.3 Additional Explanation of Address Map ······················································································· 38
8.

SAMPLE PROGRAM ····················································································································· 48

9.
9.1
9.2
9.3
9.4
9.5

ETHERNET COMMUNICATION FUNCTIONS ··········································································· 53
LAN port specification·················································································································· 53
Connection to the terminal ············································································································ 53
Connection ···································································································································· 53
Setting Ethernet communicating conditions ·················································································· 54
Ethernet communicating conditions setting operation ··································································· 54

10. FTP SERVER FUNCTION·············································································································· 55
10.1 Description of FTP server function ······························································································· 55－i－10.2 Setting FTP server function ···········································································································5710.3 FTP server function setting operation ····························································································5710.4 FTP server operation······················································································································58
11. Web server function··························································································································59
11.1 Description of web server function ································································································5911.2 Setting web server function ···········································································································59
11.3 Web server function setting operation····························································································59
11.4 Web server operation·····················································································································60
11.5 Web monitor screen·······················································································································60
12. E-MAIL SEND FUNCTION············································································································63
12.1 Description of E-mail send function ······························································································63
12.2 Setting E-mail function··················································································································63
12.3 E-mail function setting operation···································································································64
12.4 E-mail send test operation··············································································································67
12.5 E-mail send contents······················································································································67
13. MODBUS TCP/IP FUNCTION ·······································································································68
13.1 Description of MODBUS TCP/IP function····················································································68
13.2 Setting MODBUS TCP/IP function ·······························································································68
13.3 MODBUS TCP/IP function setting operation ················································································68
13.4 MODBUS TCP/IP communication protocol ··················································································69
14.

TROUBLESHOOTING ···················································································································71－ii－

1. COMMUNICATION FUNCTIONS
1.1 General・ This equipment providesss a communication function (optional) using an RS-485 interface and also a
communication function (optional) using an Ethernet interface.

1.2 Overview of MODBUS slave communication function(RS-485 interface)
See Chapter 2 through Chapter 8 for the method for use of MODBUS slave communication function in detail.
・ The MODBUS slave communication function permits exchange of data with host computer, programmablecontroller, graphic display panel, etc.
・ The communication system consists of master station and slave stations.

Up to 31 slave stations (GR200) can

be connected per master station.
Note that, because the master station can communicate with only one slave station at a time, a party tocommunicate with must be specified by the "MODBUS Station No." set at each slave station.
・ In order that the master station and slave station can communicate, the format of the transmit/receive data mustcoincide.

For the GR200, the format of the communication data is determined by the MODBUS protocol.

・ Please use an RS-232C ⇔ RS-485 converter in case of designating a personal computer or other devices whichhave an RS-232C interface as a master station.
［RS-232C ⇔ RS-485 converter］ (recommended article)
Type: K3SC-10 (isolated type)/ OMRON Corporation.

Personalcomputer

Programmablecontroller

RS-485
RS-232C
RS-232C

RS-485 converter

RS-485

Caution:When using the RS-232C ⇔ RS-485 converter, pay attention to cable connection between the converter andmaster station. If the cable is not connected correctly, the master station and slave station cannot communicate.
In addition, be careful about communication settings such as baud rate and parity set for the converter.

－1－

1.3 Overview of Ethernet communication functions
See Chapter 10 and Chapter 11 for the method for use of Ethernet communication functions in detail.
・ The following functions are available as Ethernet communication functions.
(1) FTP server function
Permits take-out of files from the compact flash of the paperless recorder, using personal computer'sbrowser (Internet Explorer Ver.6) or DOS prompt.
(2) Web server function
Permits check of measured values and event information recorded in the paperless recorder, using personal
computer's browser (Internet Explorer Ver.6).
(3) E-mail send function
Permits E-mail transmission in a fixed period and also on occurrence of an alarm.
(4) MODBUS TCP/IP function
Permits exchange of data with host computer, programmable controller, graphic display panel, etc. by
MODBUS TCP/IP communication.
Note: The other browsers, for example Netscape, Mozilla Fire-fox, are not available.

－2－2. SPECIFICATIONS
2.1 Communication Specifications
Item
Electrical specification
Transmission system
Synchronizing system
Connection format
Number connectable units
Transmission distanceTransmission speed
Data format
Transmission code
Error detection
Isolation

Specification
Based on EIA RS-485
2-wire, semi-duplicate
Start-stop synchronous system
1:N
Up to 31 units
500m max. (total extension distance)
9600，19200 bps
Data length
8 bitsStop bit
1 bit
Paritynone, even, odd (selectable)
HEX value (MODBUS RTU mode)
CRC-16Functional isolation between transmission circuit
and ground (withstand voltage : 500V AC)－3－3. CONNECTIONWARNINGFor avoiding electric shock and malfunctions, do not turn on the power
supply until all wiring have been completed.

3.1 Communication Terminal Allocation
Signal name
TRX2 –TRX1 +

Communication
Terminal

Communication Terminal

ShieldTRX2 TRX1 +－4－

3.2 Wiring
・ Use twisted pair cables with shield.
Recommended cable: UL2464, UL2448, etc.
・ The total extension length of the cable is up to 500 m.

A master station and up to 31 units of the GR200 can beconnected per line.
・ Both ends of the cable should be terminate with terminating resistors 100Ω (1/2W).
・ The shield wire of the cable should be grounded at one place on the master station unit side.
・ If the GR200 is to be installed where the level of noise applied to the GR200 may exceed 1000 V, it isrecommended to install a noise filter in the master station side as below.
Recommended noise filter: ZRAC2203-11/TDK

Master station(PC, etc.)

RS-232C⇔RS-485

Noise filter－5－

Transmission
cable

GR200

4. SETTING OF COMMUNICATION CONDITION
In order that the master station and instrument (GR200) can correctly communicate, following
settings are required.
・All communication condition settings of the master station are the same as those of
instruments (GR200).
・All instruments (GR200) connected on a line are set to " MODBUS Station No." which are different fromeach other.

(Any " MODBUS Station No." is not shared by more than one instrument.)

4.1 Set Items
The parameters to be set are shown in the following table.
Item
Station No.
Transmission
speed

Set them by operating the front panel keys.

Value atdelivery

Setting range

Remarks

1

0 to 255(0:communication function stop)

Set a different value toeach station.

19200bps 9600bps, 19200bps

Parity setting

Odd

Data length
Stop bit

8bit
1bit

None:None parity
Odd:Odd parity
Even:Even parity
Fixed (can not be changed)
Fixed (can not be changed)－6－

Set the same communication condition to themaster station and all
stave stations.

4.2 Setting Operation Method
The following example shows how to set the communication conditions.
Example: Selecting an even parity and “STno=10 and 9600bps” on a station.
Keys used

Display
Trend display

Meanings
Operation state (Trend display)

SEL

Parameter Setting

Press the [SEL] key to display the Parameter Setting screen.

∨

Main unit

ENT

Register data

Press the [∨] key two times to select Main unit. (When the key pressedmore than two times, use the [∧] key to back.)
Press the [ENT] key to select Register data∨

Communication

Press the [∨] key seven times to select Communication.

ENT

Communication setting

Press the [ENT] key to display the Communication setting screen.

ENT

Numeric value entering
Press the [ENT] key to display the Numeric value entering screen.
screen∨10

Use [<], [>] ,[∧], or [∨] key to change the numeric value to 10.

ENT

MODBUS Station No.

Press the [ENT] key to confirm the MODBUS Station No.

∨

MODBUS baud rate

Press the [∨] key to select the MODBUS baud rate.

<

9600

Press the [<] key to select "9600".

∨

MODBUS parity

Press the [∨] key to select the MODBUS parity.

<

Even

Press the [<] key to select "Even".

∨

Front communication

Press the [∨] key to select the Front communication.

DISP

Select screen for saving
the settings

Press the [DISP] key to display a screen asking you want to save thesetting.

ENT

Confirmation screen for Press the [ENT] key to save the setting. (The confirmation screen
appears.)saving the settings

ENT

Trend display

Power OFFPower ON───
Trend display

The Trend screen appears.
Turn off the power.
Turn on the power once again to complete the setting.

－7－

5. MODBUS COMMUNICATION PROTOCOL
5.1 General
The communication system by the MODBUS protocol is that the communication is always started from the masterstation and a slave station responds to the received message.
Transmission procedures is as shown below.
1） The master station sends a command message to a slave station.
2） The slave station checks that the station No. in the received message matches with the own station No. ornot.
3） If matched, the slave station executes the command and sends back the response
message.
4)

If mismatched, the slave station leaves the command message and wait for the next
command message.

a)

In case when the station No. in the received command message matches with the own slave station No.

Master to slave

Command message

Slave to masterb)

Response message

In case when the station No. in the received command message mismatches with the own slave station
No.

Master to slave

Command message(Not respond)

Slave to master

5)

Data onthe line

Data onthe line

To assure safety, provide a structure where the response message is checked andretry is made three (3) times or more if no response is made or an error occurs.

The master station can individually communicate with any one of slave stations connected on the same line uponsetting the station No. in the command message.

－8－5.2 Composition of Message
Command message and response message consist of 4 fields ; Station No., Function code, Data and Error checkcode. And these are send in this order.
Station No. (1 byte)
Function code (1 byte)
Data (2 to 133 bytes)
Error check code (CRC-16) (2 bytes)

Fig. 5-1 Composition of message
In the following, each field is explained.

(1)

Station No
Station No. is the number specifiing a slave station. The command message is received and operated only by
the slave station whose station No. matches with the No. set in the parameter " MODBUS Station No. "For details of setting the parameter " MODBUS Station No.", refer to chapter 4.

(2)Function code
This is a code to designate the function executed at a slave station.
For details, refer to section 5.4.

(3)

Data
Data are the data required for executing function codes. The composition of data varies with function codes.
For details, refer to chapter 6.
A register number is assigned to each data in the recorder. For reading/writing
the data by communication, designate the register number.
Note that the register number transmitted on message is expressed as its relative address.
The relative address is calculated by the following expression.

Relative address

⎛

⎞

⎝

⎠

＝ ⎜The lower 4 digits of the register number ⎟ －1

For example, when the resister number designated by a function code is 40003,
Relative address = (lower 4 digits of 40003) – 1= 0002is used on the message.

(4)

Error check code
This is the code to detect message errors (change in bit) in the signal transmission.
On the MODUBUS protocol (RTU mode), CRC-16 (Cycric Redundancy Check) is applied.
For CRC calculation method, refer to section 5.5.

－9－

5.3 Response of Slave Station(1)

Response for normal command
To a relevant message, the slave station creates and sends back a response message which corresponds to thecommand message. The composition of message in this case is the same as in section 5.2.
Contents of the data field depend on the function code. For details, refer to Chapter 6.

(2)

Response for abnormal command
If contents of a command message have an abnormality (for example, non-actual function code is designated)other than transmission error, the slave station does not execute that command but creates and sends back a
response message at error detection.
The composition of response message at error detection is as shown in Fig. 5-2 The value used for function
code field is function code of command message plus 80H.
Table 5-1 gives error codes.
Station No.
Function code + 80H
Error code
Error check（CRC-16）

Fig. 5-2

Response message at error detection

Table 5-1 Error Code(3)

Error code
01H

Contents
Illegal function code

02H

Illegal data address

03H

Illegal data number

Description
Non-actual function code is designated.
Check for the function code.
A relative address of a resister number to which thedesignated function code can not be used.
Because the designation of number is too much,the area where resister numbers do not exist isdesignated.

No response
Under any of the following items, the slave station takes no action of the command message and sends back no
response.
・ A station number transmitted in the command message differs from the station number specified to the slave
station.
・ A error check code is not matched, or a transmission error (parity error, etc.) is detected.
・ The time interval between the composition data of the message becomes longer than the timecorresponding to 24 bits. (Refer to section 5.6 Transmission Control Procedure)・ Station No. of a slave station is set to 0.

－10－

5.4 Function CodeAccording to MODBUS protocol, register numbers are assigned by function codes.
Each function code acts on specific register number.
This correspondence is shown in Table5-2, and the message length by function is shown in Table5-3.

Table5-2
No.
03H
04H
10H

Correspondence between function codes and objective address

Function code
FunctionObjectRead-out
Holding register(continuously)
Read-out
Input register(continuously)
Write-inHolding register(continuously)

Resister No.
Contents

No.
4xxxx

Read-out/write-inword data

3xxxx

Read-out

word data

4xxxx

Read-out/write-inword data

Table5-3 Function code and message length
Function
code
03H
04H
10H

Contents
Read-out of word data
Read-out of word data(read-out only)

Write-in of continuous word data

[Unit:byte]
Number ofCommand message
Response message
designatable
Minimum Maximum Minimum Maximum
data
64 words
887
133
64 words

8

8

7

133

64 words

11

137

8

8

－11－

5.5 Calculation of Error Check Code (CRC-16)
CRC-16 is the 2-byte (16-bits) error check code. From the top of the message (station No.) to the end of the data
field are calculated.
The slave station calculates the CRC of the received message, and does not respond if the calculated CRC isdifferent from the contents of the received CRC code.
Fig. 5-3 shows the flow of the CRC-16 calculation system.
Start
Set FFFFH (hexadecimal number) in CR.

Explanation of variables
CR : CRC error check data (2 bytes)
I : Digits of calculation charactersin command message
J : Check on the number of timesof CR calculation

Set 1 in I.

Exclusive logical sum (XOR) is executedwith CR and one character (1 byte) of the I
characters, and its results is set in CR.
Set 1 in J.

Bit at right endof CR is 1?

NO

YES
Shift CR to right by 1 bit, and A001H and
exclusive logical sum (XOR) are executedand its result is set in CR.

Shift CR to right by 1 bit.

Add 1 to J.

Calculation (8 times)
is finished?
J>8

NO

YES
Add 1 to I.

NO

Calculation of all characters iscompleted?
I>All characters
YES
End

Fig. 5-3

Flow of CRC-16 calculation－12－(Calculation is executed in the orderof command message station No.,function code and data.)
CR calculation result shall be added
to the last command message in theorder of LOW byte and HIGH byte.

5.6 Transmission Control Procedure(1)

Transmission procedure of master station
The master station must proceed to a communication upon conforming to the following items.
(1-1) Before sending a command message, provide 48 bits time or more vacant status.
(1-2) For sending, the interval between bytes of a command message is below 24 bits time.
(1-3) Within 24 bits time after sending a command message, the receiving status is posted.
(1-4) Provide 48 bits time or more vacant status between the end of response message reception andbeginning of next command message sending [same as in (1-1)].
(1-5) For ensuring the safety, make a confirmation of the response message and make an arrangement so
as to provide 3 or more retries in case of no response, error occurrence, etc.
Note) The above definition is for most unfavorable value. For ensuring the safety, it’s recommended theprogram of the master to work with safety factors of 2 to 3. Concretely, it is advised to arrange theprogram for 9600 bps with 10 ms or more for vacant status (1-1), and within 1 ms for byte interval(1-2) and changeover from sending to receiving (1-3).

(2)

Description
1) Detection of the message frame
Since the communication system uses the 2-wire RS-485 interface, there may be 2 statuses on a line below.
(a) Vacant status (no data on line)(b) Communication status (data is existing)Instruments connected on the line are initially at a receiving status and monitoring the line. When 24 bits
time or more vacant status has appeared on the line, the end of preceding frame is assumed and, within
following 24 bits time, a receiving status is posted. When data appears on the line, instruments receive it
while 24 bits time or more vacant status is detected again, and the end of that frame is assumed. I.e., data
which appeared on the line from the first 24 bits time or more vacant status to the next 24 bits time or morevacant status is fetched as one frame.
Therefore, one frame (command message) must be sent upon confirming the following.
(1-1) 48 bits time or more vacant status precedes the command message sending.
(1-2) Interval between bytes of 1 command message is smaller than 24 bits time.
2) Response of this instrument (GR200)
After a frame detection (24 bits time or more vacant status), this instrument carries out processing with thatframe as a command message. If the command message is destined to the own station, a response
message is returned. Its processing time is 300 to 500 ms (depends on contents of command message).
After sending a command message, therefore, the master station must observe the following(1-3) Receiving status is posted within 24 bits time after sending a command message.

－13－

Space time of longer than 5ms isneeded.
(longer than 10ms is recommended.)
Masterstation → GR200

POL1

POL2
500ms max

Masterstation ←GR200

POL1 response data

Data on line
POL1

POL1 response data

－14－

POL2

5.7 FIX Processing (Cautions in data write)
The instrument is provided inside with a non-volatile memory (F-ROM) for holding the setting parameters.
Data written in the non-volatile memory is not lost even if turning off the power.
To hold parameters that were written in the internal memory via communication after turning off the power, the FIX
process is effective. It allows parameters to be written in nonvolatile memory.
Fig.5-4 shows the FIX procedure.

Cautions:
・ Write in the non-volatile memory takes approximately 2 seconds.
・ While writing, do not turn off the power of the GR200. Otherwise, the data in the non-volatile memory will
be destroyed, whereby the GR200 could not be used any longer.
・ Don’t change parameters on the front panel when performing the FIX procedure, or memory error may result.
・ The non-volatile memory (F-ROM) is a device where the number of write-in times is limited. The
guaranteed number of write-in times of the non-volatile memory used on the instrument is 100,000 minimum.
Therefore, limit the times of change of parameter setting to absolute minimum. Refrain from carrying out
the FIX processing periodically for example or while such is not absolutely required.

Start FIX

Read the FIX datawith function code
relative address

No

: 03H
: 1387H

FIX=0?
Yes

Write ‘1’ into FIX datawith function code
relative address: 10H: 1387H

Read the FIX datawith function code
relative address: 03H: 1387H

No

FIX=0?
Yes
End FIX

Fig.5-4 FIX procedure－15－

6. DETAILS OF MESSAGE
6.1 Read-out of Word Data [Function code:03H]
Function codeMax. word number read-out
in one message

Relative data address

Register No.

Kind of data

03H

64 words

0000H－1386H
1387H－176FH

40001－44999
45000－46000

Storage enable data
Storage disable data

(1)

Message composition
Command message composition (byte)
Station No.
Function code
Read-out start
Upper
No.
(relative address) Lower
Read-out word
Upper ⎫
⎬ 1 to 64
numberLower ⎭Lower
CRC data
Upper

Response message composition (byte)
Station No.
Function code
Read-out byte number
Contents of thefirst word data
Contents of thenext word data～
Contents of
the last worddata
CRC data

Read-out word number×2

Upper
Lower
Upper
Lower
～

Upper
Lower
Lower
Upper

* Arrangement of read-out word data
MSB
LSB
Upper byte of contents of the first word data
Lower byte of contents of the first word data
Upper byte of contents of the next word data
Lower byte of contents of the next word data～～
Upper byte of contents of the last word data
Lower byte of contents of the last word data

(2)

Function explanations
Word data of continuous word numbers from the read-out start No. can be read.
transmitted from the slave station in the order of upper and lower bytes.

－16－

Read-out word data are(3)Message transmission (example)
Reading range start and range end in Channel 1 from No. 2 station is shown below.
Relative address of range start in Channel 1: 001BH (Register No.40028),
Data number: 02H

Command message composition (byte)
Station No.
02H
Function code
03H
Read-out start No. Upper
00H(relative address)
Lower
1BH
Read-out word
Upper
00H
number
Lower
02H
Lower
B4H
CRC data
Upper
3FH

Response message composition (byte)
Station No.
02H
Function code
03H
Read-out byte number
04H
Contents of the Upper
00H
first word data
Lower
00H
Contents of the Upper
0FH
next word data
Lower
A0H
Lower
CCH
CRC data
Upper
BBH
B

* Meaning of data to be read
Channel 1 Range start(contents of the first word data)
Channel 1 Range end(contents of the next word data)

00 00H = 00F

A0H = 4000

Where the unit is ℃ with decimal point position set at 1,
Channel 1 Range start = 0.0℃
Channel 1 Range end = 400.0℃

Point

For “Point” decimal point, refer to Section 7.1－17－

6.2 Read-out of Read-out Only Word Data [Function code:04H]
Function code
04H

(1)

Max. word number readout
in one message
64 words

Relative data address

Register No.

0000H－07CFH

30001－32000

Message composition
Command message composition (byte)
Station No.
Function code
Read-out start No. Upper(relative address) Lower
Read-out wordUpper⎫
⎬ 1 to 64
number
⎭
Lower
Lower
CRC data
Upper

Response message composition (byte)
Station No.
Function code
Read-out byte number
Read-out word number×2
Contents of the Upper
first word data
Lower
Contents of the Upper
next word data
Lower
～
～
Contents ofUpper
the last word
Lower
data
Lower
CRC data
Upper* Arrangement of read-out word data
MSB
LSB
Upper byte of contents of the first word data
Lower byte of contents of the first word data
Upper byte of contents of the next word data
Lower byte of contents of the next word data～～
Upper byte of contents of the last word data
Lower byte of contents of the last word data

(2)

Function explanations
Word data of continuous word numbers from the read-out start No. can be read. Read-out word data aretransmitted from the slave station in the order of upper and lower bytes.

－18－

(3)

Message transmission (example)
Reading measured values in Channel 2 from No. 1 station is shown below.
Relative address of measured value in Channel 2: 0065H (Register No.30102),

Command message composition (byte)
Station No.
01H
Function code
04H
Read-out start No.
Upper
00H
(relative address)
Lower
65H
Read-out word
Upper
00H
number
Lower
01H
Lower
21H
CRC data
Upper
D5H* Meaning of data to be read
Channel 2 Measured value(contents of the first word data)

01 4FH

Response message composition (byte)
Station No.
01H
Function code
04H
Read-out byte number
02H
Contents of the
Upper
01H
first word data
Lower
4FH
Lower
F9H
CRC data
Upper
54H

= 335

Where the unit is °C with decimal point position set at 1
Channel 2 Measured value = 33.5℃

Point

For “Point” decimal point, refer to Section 7.1.

－19－

Data number: 01H

6.3

Write-in of Continuous Word Data [Function code:10H]

Function codeMax. word number write-in
in one message

10H

64 words(1)

Relative data
address
0000H－1386H
1387H－176FH

Register No.

Kind of data

40001－44999
45000－46000

Storage enable data
Storage disable data

Message composition
Command message composition (byte)
Station No.
Function code
Write-in start No. Upper(relative address) Lower
Write-in wordUpper⎫
⎬ 1 to 64
number⎭Lower

Response message composition (byte)
Station No.
Function code
Write-in start No.
Upper(relative address)
Lower
Write-in word
Upper
number
Lower
Lower
⎫
⎬ Write-in word number×2 CRC data⎭
Upper

Write-in byte numberFirst write-inword data
Next write-inword data～
Last write-inword data

CRC data

Upper
Lower
Upper
Lower
Upper
Lower
Lower
Upper

～

* Arrangement of write-in word data
MSB
LSB
Upper byte of contents of the first word data
Lower byte of contents of the first word data
Upper byte of contents of the next word data
Lower byte of contents of the next word data～～
Upper byte of contents of the last word data
Lower byte of contents of the last word data

(2)

Function explanation
Word data of continuous word number is written from write-in start address.
transmitted from master station in the order of upper and lower bytes.

－20－

Write-in word data are(3)Message transmission (example)
Writing Subtract channel = channel 2, PV shift = 20.0°C, and PV gain = 110.0% in
Channel 1 of No. 1 station is shown below.
Subtract channel = 0002H (= 2D : channel 2)
PV shift = 00C8H (= 200D)
Input filter = 044CH (= 1100D)
Relative address of Subtract channel in Channel 1: 0014H (Register No.40021),

Command message composition (byte)
Station No.
01H
Function code
10H
Write-in start No. Upper
00H(relative address) Lower
14H
Write-in word
Upper
00H
numberLower
03H
Write-in byte number
06H
First write-in
Upper
00H
word data
Lower
02H
Next write-in
Upper
00H
word data
Lower
C8H
Last write-in
Upper
04H
word data
Lower
4CH
Lower
5DH
CRC data
Upper
CBH

Data number: 03H

Response message composition (byte)
Station No.
01H
Function code
10H
00H
Write-in start No. Upper(relative address) Lower
14H
Upper
00H
Write-in word
number
Lower
03H
Lower
41H
CRC data
Upper
CDH

Point

Since the transmission data can not include a decimal point, data of 110.0 istransmitted as "1100".
For transmission format of each data, refer to the address map (Chapter7)

CautionIf the write-in command message is sent to any slave station during the
FIX process, response is not returned from it.

－21－

7. ADDRESS MAP AND DATA FORMAT
7.1 Data Format
7.1.1 Transmission data format
The MODBUS protocol used in this instrument (GR200) is RTU (Remote Terminal Unit) mode.
Transmitted data is "numeric value" and not "ASCII code".

－22－7.1.2 Control of decimal point
A decimal point is not included on the transmission data.
Align decimal point for data that have decimal point (decimal point is eliminated in transmission, and added inreceiving).

7.1.3 Data with input error
When input error (Over, Under, Burnout or Error) occurs in display data, read data from measured values are as
follows.

Display data
Over
Under
BunoutError

Read data
32767–32767–32768–32768Detection of input error during communication can be performed at address 30131 = Channel status.

7.1.4

Range of write-in data

When data is written in each parameter, the write-in data should be kept within the setting range. GR200 accepts
the write-in data beyond the range. However, be careful since the GR200 performance will not be guaranteed.

－23－

7.2 Address Map
For detailed contents about individual parameter function or setting range, refer to the operation manual.
Data type Long: long data The data of this address is manipulated in unit of word. 1 data/2 address
Word: word data The data of this address is manipulated in unit of word. 1 data/1 address
Byte: byte data The data of this address is manipulated in unit of byte. A maximum of 2 data/1 address
Bit: Bit data The data of this address is manipulated in unit of bit. A maximum of 16 data/1 address

7.2.1 Word data [ read-out / write-in ] : Function code [ 03H, 10H ]Register
No.

Data
type

4XXXX
40001 Byte
40002 Byte
40003 Byte
40004 Byte
40005 Byte
40006 Byte
40007 Byte
40008 Byte
40009 Word
40010 Word
40011 Word
40012 Word
40013 Word
40014 Word
40015 Word
40016 Word
40017 Word
40018 Word
40019 Word
40020 Word
40021 Word
40022 Word
40023 Word
40024 Word
40025 Word
40026 Word
40027 Word
40028 Word
40029 Word
40030 Word
40031 Word
40032 Word
40033 Word
40034 Word
40035 Word
40036 Word
40037 Word

Memory contents

Tag 1

Tag 2

1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters

Color
Input type
Input filter
Unit
Scaling
Masuring start
Masuring end
Engineering startEngineering end
Decimal point
Square rooter
Logarithmic display
Subtract channel
PV shiftPV gain

Channel 1
setting(64
Recording modewords)
Recording type
Range startRange end

Remarks

Set Tag 1 (8 characters) by the ASCII code.

Set Tag 2 (8 characters) by the ASCII code.

1 to 14 ( Please refer to Table 1)
0 to 40 ( Please refer to Table 2)
0 to 900 (0 to 900 sec)
0 to 167 ( Please refer to Table 3)
0:OFF, 1:ON
–1000 to 5500 ( Please refer to Table 4)–1000 to 5500 ( Please refer to Table 4)–32767 to 32767–32767 to 32767
0 to 4 ( Please refer to Table 5)
0:OFF, 1:ON
0:OFF, 1:ON
0 to 30(0:Subtract OFF, 1 to 30:channel 1 to 30)–32767 to 32767
0 to 32767 (0.00 to 327.67%)
Reserve
Reserve
0:With record, 1:Display only
0:Min-Max rec., 1:Point record, 2:Average rec.
–32767 to 32767 ( Please refer to Table 6)–32767 to 32767 ( Please refer to Table 6)
Reserve

Input channel

0 to 29 (channel 1 to 30)
Reserve
Reserve
Reserve
Reserve

Fvalue caluculation
Totalize caluculation

40038 Byte
40039 Byte
40040 Byte
40041 Byte
40042 Word
40043 Word
40044 Word
40045 Word
40046 Word
40047 Word
40048 Word

Read-out data / Write-in data setting range

1st, 2nd characters
Totalizetag

3rd, 4th characters
5th, 6th characters
7th, 8th characters
Totalize unit
Totalize cut value
Totalize scale value
Totalize type
External input
Totalize base time
Reset operation

0:OFF, 1:ON
1:Totalizer, 2:Counter, 3:Timer
Set Totalize Tag (8 characters) by the ASCII
code.

0 to 167 ( Please refer to Table 3)–32767 to 32767 ( Please refer to Table 6)
1 to 32767
0 to 15 ( Please refer to Table 7)
0 to 81 ( Please refer to Table 8)
0:/s, 1:/min, 2:/h, 3:/day0:OFF, 1:ON

－24－

Register
No.

Data
type

Read-out data / Write-in data setting range

Totalize reset input

0 to 130 ( Please refer to Table 9)

Channel 2 setting

Same allocation as in Channel 1

Channel 3 setting

Same allocation as in Channel 1

Channel 4 setting

Same allocation as in Channel 1

Channel 5 setting

Same allocation as in Channel 1

Channel 6 setting

Same allocation as in Channel 1

Channel 7 setting

Same allocation as in Channel 1

Channel 8 setting

Same allocation as in Channel 1

Channel 9 setting

Same allocation as in Channel 1

Channel 10 setting

Same allocation as in Channel 1

Channel 11 setting

Same allocation as in Channel 1

Channel 12 setting

Same allocation as in Channel 1

Channel 13 setting

Same allocation as in Channel 1

Channel 14 setting

Same allocation as in Channel 1

Channel 15 setting

Same allocation as in Channel 1

Channel 16 setting

Same allocation as in Channel 1

Channel 17 setting

Same allocation as in Channel 1

Channel 18 setting

Same allocation as in Channel 1～

40065 Word

Remarks
Reserve
Reserve～

40049 Word
40050 Word

Memory contents～

40129 Word～40193 Word～

40257 Word～

40321 Word～

40385 Word～

40449 Word～

40513 Word～

40577 Word～

40641 Word～

40705 Word～

40769 Word～

40833 Word～

40897 Word～

40961 Word～

41025 Word～

41089 Word

Reserve

～

41153

Alarm
No.1

Alarm
Channel 1 No.2
alarm
setting(16
Alarm
words)
No.3

Alarm
No.4

Alarm typeSet point

0:OFF, 1:H alarm, 2:L alarm–32767 to 32767 ( Please refer to Table 6)

DO relay No.
Alarm typeSet point

0 to 28 (0:None, 1 to 28:DO1 to 28)
0:OFF, 1:H alarm, 2:L alarm–32767 to 32767 ( Please refer to Table 6)

DO relay No.
Alarm typeSet point

0 to 28 (0:None, 1 to 28:DO1 to 28)
0:OFF, 1:H alarm, 2:L alarm–32767 to 32767 ( Please refer to Table 6)

DO relay No.
Alarm typeSet point

0 to 28 (0:None, 1 to 28:DO1 to 28)
0:OFF, 1:H alarm, 2:L alarm–32767 to 32767 ( Please refer to Table 6)

Reserve

Reserve

Reserve

Reserve

DO relay No.
Channel 2 alarm setting

0 to 28 (0:None, 1 to 28:DO1 to 28)
Same allocation as in Channel 1

Channel 3 alarm setting

Same allocation as in Channel 1～

41921 Word
41922 Word
41923 Word
41924 Word
41925 Word
41926 Word
41927 Word
41928 Word
41929 Word
41930 Word
41931 Word
41932 Word
41933 Word
41934 Word
41935 Word
41936 Word
41937 Word～

41953 Word－25－

Register
No.

Data
type

Read-out data / Write-in data setting range

Channel 4 alarm setting

Same allocation as in Channel 1

Channel 5 alarm setting

Same allocation as in Channel 1

Channel 6 alarm setting

Same allocation as in Channel 1

Channel 7 alarm setting

Same allocation as in Channel 1

Channel 8 alarm setting

Same allocation as in Channel 1

Channel 9 alarm setting

Same allocation as in Channel 1

Channel 9 alarm setting

Same allocation as in Channel 1

Channel 10 alarm setting

Same allocation as in Channel 1

Channel 11 alarm setting

Same allocation as in Channel 1

Channel 12 alarm setting

Same allocation as in Channel 1

Channel 13 alarm setting

Same allocation as in Channel 1

Channel 14 alarm setting

Same allocation as in Channel 1

Channel 15 alarm setting

Same allocation as in Channel 1

Channel 16 alarm setting

Same allocation as in Channel 1

Channel 17 alarm setting

Same allocation as in Channel 1

Channel 18 alarm setting

Same allocation as in Channel 1

Remarks～

41969 Word

Memory contents～

41985 Word～

42001 Word～

42017 Word～42033 Word～

42049 Word～

42049 Word～

42065 Word～

42081 Word～

42097 Word～

42113 Word～42129 Word～

42145 Word～

42161 Word～

42177 Word～

42193 Word

Reserve～

42209
42425 Byte

3rd, 4th characters
5th, 6th characters
Display
7th, 8th characters
name9th, 10th characters
11th, 12th characters
13th, 14th characters
Display
15th, 16th charactersgroup 1setting
Display No.1
(18
Display No.2
words)
Display No.3
Display No.4
Display No.5
Display No.6
Display No.7
Display No.8
Display No.9
Display No.10
Display group 2 setting

0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
0:None, 1 to 18:ch1 to 18
Same allocation as Display group 1

Display group 3 setting

Same allocation as Display group 1

Display group 4 setting

Same allocation as Display group 1～42426 Byte
42427 Byte
42428 Byte
42429 Byte
42430 Byte
42431 Byte
42432 Byte
42433 Word
42434 Word
42435 Word
42436 Word
42437 Word
42438 Word
42439 Word
42440 Word
42441 Word
42442 Word
42443 Word

Set Display name (16 characters) by the ASCII
code.

1st, 2nd characters～

42461 Word～

42479 Word
42497 Word

Reserve－26－

Register
No.

Data
type

Displaygroup 1setting2(5 words)

Read-out data / Write-in data setting range

Trend direction
Channel index
Scale display

Remarks

0:Vertical, 1:Horizontal
0:CH No.disp., 1:Tag No.disp., 2:Unit Disp.
0:OFF, 1:ON
Reserve

Display divided
Display group 2 setting2

1 to 20Same allocation as Display group 1

Display group 3 setting2

Same allocation as Display group 1

Display group 4 setting2

Same allocation as Display group 1～42498 Word
42499 Word
42500 Word
42501 Word
42502 Word
42503 Word

Memory contents～

42508 Word～

42513 Word

Reserve
Reserve
Reserve

Displaygroup 1setting3(4 words) Analog meter
Display group 2 setting2

0:Bar graph, 1:Analog meter
Same allocation as Display group 1

Display group 3 setting2

Same allocation as Display group 1

Display group 4 setting2

Same allocation as Display group 1～

42518 Word
42519 Word
42520 Word
42521 Word
42522 Word～

42526 Word～

42530 Word

Reserve
Reserve～

42534 Word
Totalize cycle

0 to 9 ( Please refer to Table 10)

Monthly base day

1 to 31 ( 1 to 31day)

Reserve
Reserve
Reserve
External input
H-P, L-P timer cycle
AVG timer cycle
SUM timer cycle
Formula1

Formula1

Math
CH 19
setting(36 words)

Formula1

Formula2

1st calculationargument1
argument2
2nd calculationargument1
argument2
3rd calculationargument1
argument2
1st calculationargument1
argument2

0 to 81 ( Please refer to Table 8)
1 to 32767 (1 to 32767min)
1 to 32767 (1 to 32767min)
1 to 32767 (1 to 32767min)( Please refer to Table 11)～42542 Word
42543
42544 Word
42545
42546
42547 Word
42548 Word42549 Word
42550 Word
42551 Word
42552 Word
42553 Word
42554 Word42555 Word
42556 Word
42557 Word42558 Word42559 Word
42560 Word42561 Word
42562 Word

Formula4

3rd calculationargument1
argument2

Math channel 20 setting

Same allocation as Channel 19

Math channel 21 setting

Same allocation as Channel 19

Math channel 22 setting

Same allocation as Channel 19

Math channel 23 setting

Same allocation as Channel 19

Math channel 24 setting

Same allocation as Channel 19

Math channel 25 setting

Same allocation as Channel 19

Math channel 26 setting

Same allocation as Channel 19

Math channel 27 setting

Same allocation as Channel 19～42584 Word42585 Word
42586 Word
42587 Word

～

42623

～

42659

～

42695

～

42731

～

42767

～

42803

～

42839

－27－

ReserveMath channel 28 setting

Same allocation as Channel 19

Math channel 29 setting

Same allocation as Channel 19

Math channel 30 setting

Same allocation as Channel 19

42983
42984

Constant1

～

～

42875
～
42911
～
42947

Constant20'-32767 to 32767 ( Please refer to Table 12)
0 to 4

Value
Decimal point'-32767 to 32767 ( Please refer to Table 12)
0 to 4
Reserve
Reserve

～

43021
43022
43023

Value
Decimal point

43032 Word
43033 Word
43034 Word
43035 Word
43036 Word
43037 Word
43038 Word
43039 Word
43040 Word
43041 Word
43042 Word
43043 Word
43044 Word
43045 Word
43046 Word
43047 Word
43048 Word
43049 Word
43050 Word
43051 Word
43052 Word
43053 Word
43054 Word
43055 Word
43056 Word
43057 WordTime
settingTimesettingTime set request

1 to 12 ( 1 to 12month)Year

1 to 31 ( 1 to 31day)

Attention:
Don’t change thetime absolutely
during recording
and totalizing.

Month
Day
Hour
Minute0 to 19 ( Please refer to Table 13 )Refreshment cycle

Reserve
LCD lights-out time
File division cycle
Memory full alarm
Record data format

FValue calculation

Target temperture
Z value
Decimal point

Refreshment cycle
0 to 4 (Please refer to Table 14)
0 to 28 (0:None, 1 to 28:DO1 to 28)
0:Ascii, 1:Binary–2300 to 32767 (–230.0 to 3276.7℃)–2300 to 32767 (–230.0 to 3276.7℃)
0 to 4 ( Please refer to Table 15)
Reserve

Reset temperature
Battery alarm
Date format
File overwrite
Display compression–2300 to 32767 (–230.0 to 3276.7℃)
0 to 28 (0:None, 1 to 28:DO1 to 28)
0 to 4 ( Please refer to Table 12)
0:OFF, 1:ON
0:1/1, 1: 1/10, 2:1/30, 3:1/60

Alarm hysteresis
Alarm latch

0 to 10000 ( 0.00 to 100.00%)
0:OFF, 1:ON

Reserve－28－

Register
No.

Data
type

Read-out data / Write-in data setting range

MODBUS Station No.
MODBUS baud rate
MODBUS parity

0 to 255 ( 0: Communication OFF)
0:9600bps, 1:19200bps
0:None, 1:Odd, 2:Even

Configuration password
CF manager password
Record password

0 to 9999
0 to 9999
0 to 9999
Reserve
Reserve
Reserve

Trend back color
Historical back color
Message Message
No.1
setting(22
Words)

1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters

0 : White , 1 : Black
0 : White , 1 : BlackSet Message (32 characters) by the ASCII code.

Reserve
Reserve
Message timing
Message timing argument 1
Message timing argument 2

0 to 2 ( Please refer to Table 16)
0 to 9/0 to 29 ( Please refer to Table 16)
0 to 3 ( Please refer to Table 16)
Reserve

Message No.2 setting

Same allocation as Message No. 1

Message No.3 setting

Same allocation as Message No. 1

Message No.4 setting

Same allocation as Message No. 1

Message No.5 setting

Same allocation as Message No. 1

Message No.6 setting

Same allocation as Message No. 1

Message No.7 setting

Same allocation as Message No. 1

Message No.8 setting

Same allocation as Message No. 1

Message No.9 setting

Same allocation as Message No. 1

Message No.10 setting

Same allocation as Message No. 1～

43061 Word
43062 Word
43063 Word
43064 Word
43065 Word
43066 Word
43067 Word
43068 Word
43081 Byte
43082 Byte
43083 Byte
43084 Byte
43085 Byte
43086 Byte
43087 Byte
43088 Byte
43089 Byte
43090 Byte
43091 Byte
43092 Byte
43093 Byte
43094 Byte
43095 Byte
43096 Byte
43097 Word
43098 Word
43099 Word
43100 Word
43101 Word
43102 Word
43103 Word

Remarks

Reserve～

43058 Word
43059 Word
43060 Word

Memory contents～

43125 Word～

43147 Word～

43169 Word～

43191 Word～

43213 Word～

43235 Word～

43257 Word～

43279 Word
43301 Byte

1st, 2nd characters

43302 Byte
43303 Byte
43304 Byte
43305 Byte
43306 Byte
43307 Byte
43308 Byte

3rd, 4th characters
5th, 6th characters
Original unit 1 setting
7th, characters

Set original unit (7 characters) by the ASCII
code.

Reserve
Reserve
Reserve
Same allocation as Original unit 1

Original unit 3 setting

Same allocation as Original unit 1～

Original unit 2 setting

43315 Byte

－29－

～

Original unit 4 setting

Same allocation as Original unit 1

Original unit 5 setting

Same allocation as Original unit 1

Memory contents

Read-out data / Write-in data setting range

Original unit 6 setting

Same allocation as Original unit 1

Original unit 7 setting

Same allocation as Original unit 1

Original unit 8 setting

Same allocation as Original unit 1

Original unit 9 setting

Same allocation as Original unit 1

Original unit 10 setting

Same allocation as Original unit 1

Original unit 11 setting

Same allocation as Original unit 1

Original unit 12 setting

Same allocation as Original unit 1～

43322 Byte～

43329 Byte

Register
No.

Data
type

～

43336 Byte

Remarks～

43343 Byte～

43350 Byte～

43357 Byte～

43364 Byte～

43371 Byte～

43378 Byte

Reserve
Reserve～

43385 Word
DI 1 function
DI 2 function
DI 3 function
DI 4 function
DI 5 function
DI 6 function
DI 7 function
DI 8 function
DI 9 function
DI 10 function

0 to 5 ( Please refer to Table 17)
0 to 50 to 50 to 50 to 50 to 50 to 50 to 50 to 50 to 5～43496 Word
43497 Word
43498 Word
43499 Word
43500 Word
43501 Word
43502 Word
43503 Word
43504 Word
43505 Word
43506 Word

PILC data

Serial number

1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters
1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters～44001 Byte
44002 Byte
44003 Byte
44004 Byte
44005 Byte
44006 Byte
44007 Byte
44008 Byte
44009 Byte
44010 Byte
44011 Byte
44012 Byte
44013 Byte
44014 Byte
44015 Byte
44016 Byte
44017 Byte
44018 Byte
44019 Byte
44020 Byte
44021 Byte
44022 Byte
44023 Byte
44024 Byte
44025 Word－30－

Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write
Do not write

The following register No.45000 to 45500 will not be recorded in the main unit.
Register
No.

Data
type

Read-out data / Write-in data setting range

Register data request

1:Register data (Automatically clear）

Fvalue calculation reset request
Prohibiting the writing to the memory card
Recorder control
Message request
Totalize reset request
Alarm latch clear request

1:Fvalue reset (Automatically clear）
0:Writing permission, 1:Writing prohibition( Please refer to Table 18)( Please refer to Table 19)
1:Totalize reset (Automatically clear）
1:Alarm latch clear (Automatically clear）

Remarks
Reserve～

Do not write～

45000 Word
45001 Word
45002 Word
45003 Word
45004 Bit
45005 Bit
45006 Word
45007 Word
45008 Word

Memory contents

Totalize reset request Channel 1 to 16
Channel 17 to 30( Please refer to Table 20)
Reserve
Reserve～

45051 Bit
45052 Bit
45053 Word

Communication input 1 : M01
Communication input 2 : M02
Communication input 3 : M03
Communication input 4 : M04
Communication input 5 : M05
Communication input 6 : M06
Communication input 7 : M07
Communication input 8 : M08
Communication input 9 : M09
Communication input10 : M10
Communication input11 : M11
Communication input12 : M12-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767-32767 to 32767～

45061 Word
45062 Word
45063 Word
45064 Word
45065 Word
45066 Word
45067 Word
45068 Word
45069 Word
45070 Word
45071 Word
45072 Word

Do not write

The following addreses are recorded in the main unit.
Register
No.

Data
type

45501 Word
45502 Word
45503 Word
45504 Word
45505 Word
45506 Word
45507 Word
45508 Word
45509 Word
45510 Word
45511 Word
45512 Word
45513 Word
45514 Word
45515 Word
45516 Word
45517 Word
45518 Word
45519 Word
45520 Word
45521 Word
45522 Word45523 Word
45524 Word45525 Byte
45526 Byte
45527 Byte
45528 Byte

Memory contents

Read-out data / Write-in data setting range

E-mail function
FTP server function
FTP access control
Web server function

0: OFF, 1. ON
0: OFF, 1: ON
0: OFF, 1: ON
0: OFF, 1: ON

Remarks
Reserve

Reserve
Reserve
MODBUS TCP/IP function
IP address
1st number
2nd number
3rd number
4th number
Subnet mask
1st number
2nd number
3rd number
4th number
Default gateway
1st number
2nd number
3rd number
4th number
SMTP IP address
1st number
2nd number
3rd number
4th number
Sender’s mall address 1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters

0: OFF, 1. ON
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
0 to 255
Set address (64 characters) by the ASCII code.

－31－Register
No.

Data
type

45529 Byte
45530 Byte
45531 Byte
45532 Byte
45533 Byte
45534 Byte
45535 Byte
45536 Byte
45537 Byte
45538 Byte
45539 Byte
45540 Byte
45541 Byte
45542 Byte
45543 Byte
45544 Byte
45545 Byte
45546 Byte
45547 Byte
45548 Byte
45549 Byte
45550 Byte
45551 Byte
45552 Byte
45553 Byte
45554 Byte
45555 Byte
45556 Byte
45557 Byte
45558 Byte
45559 Byte
45560 Byte
45561 Byte
45562 Byte
45563 Byte
45564 Byte
45565 Byte
45566 Byte
45567 Byte
45568 Byte
45569 Byte
45570 Byte
45571 Byte
45572 Byte
45573 Byte
45574 Byte
45575 Byte
45576 Byte
45577 Byte
45578 Byte
45579 Byte
45580 Byte
45581 Byte
45582 Byte
45583 Byte
45584 Byte
45585 Byte
45586 Byte
45587 Byte
45588 Byte
45589 Byte
45590 Byte

Memory contents

Read-out data / Write-in data setting range

9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters
33th, 34th characters
35th, 36th characters
37th, 38th characters
39th, 40th characters
41th, 42th characters
43th, 44th characters
45th, 46th characters
47th, 48th characters
49th, 50th characters
51th, 52th characters
53th, 54th characters
55th, 56th characters
57th, 58th characters
59th, 60th characters
61th, 62th characters
63th, 64th characters
Sender’s mail name
1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters
Receiver’s mail1st, 2nd characters
address 1 3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters
33th, 34th characters
35th, 36th characters

Set name (32 characters) by the ASCII code.

Set address (64 characters) by the ASCII code.

－32－

Remarks

Read-out data / Write-in data setting range

45591 Byte
45592 Byte
45593 Byte
45594 Byte
45595 Byte
45596 Byte
45597 Byte
45598 Byte
45599 Byte
45600 Byte
45601 Byte
45602 Byte
45603 Byte
45604 Byte
45605 Byte

37th, 38th characters
39th, 40th characters
41th, 42th characters
43th, 44th characters
45th, 46th characters
47th, 48th characters
49th, 50th characters
51th, 52th characters
53th, 54th characters
55th, 56th characters
57th, 58th characters
59th, 60th characters
61th, 62th characters
63th, 64th characters
Receiver's mail address 2

Same allocation as Receiver's mail address 145637 Byte

Receiver's mail address 3

Same allocation as Receiver's mail address 1

45669 Byte

Receiver's mail address 4

Same allocation as Receiver's mail address 1

45701 Byte～

Receiver's mail address 5

Same allocation as Receiver's mail address 145733 Byte～

Receiver's mail address 6

Same allocation as Receiver's mail address 145765 Byte～

Receiver's mail address 7

Same allocation as Receiver's mail address 1

45797 Byte

Receiver's mail address 8

Same allocation as Receiver's mail address 1～

Memory contents

～

Data
type

～

Register
No.

～

45829 Word～

Reserve
Reserve
User1
setting～

User 2 setting

Same allocation as User 1

45929 Byte～

User 3 setting

Same allocation as User 1

45943 Byte～

User 4 setting

Same allocation as User 1

45957 Byte～

User 5 setting

Same allocation as User 1

45971 Byte～

User 6 setting

Same allocation as User 1

45985 Byte～

User 7 setting

Same allocation as User 1

45999 Byte

User 8 setting

Same allocation as User 1～

45901 Byte
45902 Byte
45903 Byte
45904 Byte
45905 Byte
45906 Byte
45907 Byte
45908 Byte
45909 Byte
45910 Byte
45911 Byte
45912 Byte
45913 Word
45914 Word
45915 Byte

User name

Password

1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters

User levelSet name (16 characters) by the ASCII code.

Set name (8 characters) by the ASCII code.

0: administrator, 1: guest
Reserve

46013 Word

Reserve
Reserve～

46101 Byte
46102 Byte

Remarks

E-mailtrigger 1

Title

1st, 2nd characters
3rd, 4th characters

Set title (32 characters) by the ASCII code.

－33－Register
No.

Data
type

Memory contents

Read-out data / Write-in data setting range

46103 Byte
46104 Byte
46105 Byte
46106 Byte
46107 Byte
46108 Byte
46109 Byte
46110 Byte
46111 Byte
46112 Byte
46113 Byte
46114 Byte
46115 Byte
46116 Byte
46117 Byte
46118 Byte
46119 Byte
46120 Byte
46121 Byte
46122 Byte
46123 Byte
46124 Byte
46125 Byte
46126 Byte
46127 Byte
46128 Byte
46129 Byte
46130 Byte
46131 Byte
46132 Byte
46133 Byte
46134 Byte
46135 Byte
46136 Byte
46137 Byte
46138 Byte
46139 Byte
46140 Byte
46141 Byte
46142 Byte
46143 Byte
46144 Byte
46145 Byte
46146 Byte
46147 Byte
46148 Byte
46149 Word
46150 Word
46151 Word
46152 Word
46153 Word
46154 Word
46155 Byte

setting～E-mail trigger 2 setting

Same allocation as E-mail trigger 1～

46209 Byte

E-mail trigger 3 setting

Same allocation as E-mail trigger 1～46263 Byte

E-mail trigger 4 setting

Same allocation as E-mail trigger 1～

46317 Byte

E-mail trigger 5 setting

Same allocation as E-mail trigger 1～

46371 Byte

E-mail trigger 6 setting

Same allocation as E-mail trigger 1～

5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters
Text 1
1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters
Text 2
1st, 2nd characters
3rd, 4th characters
5th, 6th characters
7th, 8th characters
9th, 10th characters
11th, 12th characters
13th, 14th characters
15th, 16th characters
17th, 18th characters
19th, 20th characters
21th, 22th characters
23th, 24th characters
25th, 26th characters
27th, 28th characters
29th, 30th characters
31th, 32th characters
Trigger timing
Trigger timing argument 1
Trigger timing argument 2
PV value affixation
Receiver's mail address No.

46425 Byte

E-mail trigger 7 setting

Same allocation as E-mail trigger 146479 Byte

E-mail trigger 8 setting

Same allocation as E-mail trigger 1

Remarks

Set text 1 (32 characters) by the ASCII code.

Set text 2 (32 characters) by the ASCII code.

(Please refer to Table 21)

0: OFF, 1: ON(Please refer to Table 22)
Reserve

－34－

Data
type

Read-out data / Write-in data setting range

46533 Byte

E-mail trigger 9 setting

Same allocation as E-mail trigger 146587 Byte

E-mail trigger 10 setting

Same allocation as E-mail trigger 1～

Memory contents～

RegisterNo.

～

Reserve
Reserve
Reserve～

46641 Word
47000 Word

Remarks

Final address－35－

7.2.2 Word data [ read-out only ] : Function code [ 04H ]Register
No.

Data
type

～

3xxxx
30001 Bit
30002 Bit
30003 Bit
30004 Bit
30005 Bit
30006 Bit～

30086 Word

Read-out data

System information( Please refer to Table 23)

DO information( Please refer to Table 24)
Reserve

DI information( Please refer to Table 25)

Memory cord utilization

0 to 1000 (0.00 to 100.0%, 100.0% = Memory Full)Reserve
Reserve
Channel Alarm
information

Channel 1 to 4
Channel 5 to 8
Channel 9 to 12
Channel 13 to 16
Channel 17 to 18( Please refer to Table 26)

Reserve
Reserve
Reserve
Channel 1
Channel 2
Channel 3–32767 to 32767 （ No decimal point )–32767 to 32767 （ No decimal point )–32767 to 32767 （ No decimal point )30117 Word
30118 Word

Channel 17
Channel 18–32767 to 32767 （ No decimal point )–32767 to 32767 （ No decimal point )30131 Word
30132 Word
30133 Word

Channel 1
Channel 2
Channel 3

0:Normal, 1:Burnout, 2:Over, 3:Under, 4:Error
0:Normal, 1:Burnout, 2:Over, 3:Under, 4:Error
0:Normal, 1:Burnout, 2:Over, 3:Under, 4:Error

Channel 18

0:Normal, 1:Burnout, 2:Over, 3:Under, 4:Error-999999999 to 999999999 (No decimal point)～

Channel 1
Channel 1
Channel 2
Channel 230195 Long
30196 Long

Channel 18
Channel 18-999999999 to 999999999 (No decimal point)

Greenwich Time

Measured value

Reserve
Channel status～～

Remarks

Reserve
DO1 to 16
DO17 to 28～

30093 Bit
30094 Bit
30095 Bit
30096 Bit
30097 Bit
30098 Bit
30099 Bit
30100 Bit
30101 Word
30102 Word
30103 Word

Memory contents

30148 Word～

Reserve
Totalizing value-999999999 to 999999999 (No decimal point)

Reserve～

30161 Long
30162 Long
30163 Long
30164 Long～

Totalizing start time

Channel 1
Channel 1
Channel 2
Channel 230255 Long
30256 Long

Channel 18
Channel 18

Greenwich Time

Greenwich Time

Greenwich Time

Reserve～

30221 Long
30222 Long
30223 Long
30224 Long～

Channel 1
Channel 1
Channel 2
Channel 230315 Long
30316 Long

Channel 18
Channel 18

Greenwich Time

Channel 1
Channel 1
Channel 2
Channel 2-999999999 to 999999999 (No decimal point)～

Totalizing end time

30375 Long
30376 Long

Channel 18
Channel 18-999999999 to 999999999 (No decimal point)

Greenwich Time

Reserve～

30281 Long
30282 Long
30283 Long
30284 Long

30341 Long
30342 Long
30343 Long
30344 Long

Previous totalizedvalue-999999999 to 999999999 (No decimal point)

－36－

～
Previous totalized
start time

Channel 1
Channel 1
Channel 2
Channel 230435 Long
30436 Long

Channel 18
Channel 18

Greenwich Time

Channel 1
Channel 1
Channel 2
Channel 2

Greenwich Time～～

30401 Long
30402 Long
30403 Long
30404 Long

Previous totalized
end time～

30461 Long
30462 Long
30463 Long
30464 Long

Channel 18
Channel 18

Greenwich Time

Greenwich Time

Greenwich Time
Reserve
Reserve
Reserve～

30495 Long
30496 Long
30497 Long

Greenwich Time

32000 Word

Final address

Notes) ・The area marked (Do not write) is a system area.

－37－

Do not write in there.

7.3 Additional Explanation of Address Map
Table 1 Channel color code
Data
1
2
3
4
5
6
7
8
9
10
11
12
13
14

color
Red
Blue
VioletGreenSky blue
Yellow
Gray
IndigoDark redPurpleDeep green
Pale blue
Yellowish green
SilverTable 2 Input type code
Data
0
1
2
3
4
5
6
7
8
9
10
11
12
20
21
30
31
32
33
40

Input type
Skip
K-Type TC
E-Type TC
J-Type TC
T-Type TC
R-Type TC
S-Type TC
B-Type TC
N-Type TC
W-Type TC
L-Type TC
U-Type TC
PN-Type TC
Pt100
JPt100
50mV
500mV
1-5V
0-5V
Other channel

Skip

Initial decimal point
0

Thermocouple

1

Resistance bulb

1
2
1

DC voltage

3
Other channnel

0Note) When position of decimal point varies with input type, initialize it.

－38－

Table 3 Unit code
Data
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
Data
90
91
92
93
94
95
96
97
98
99
100
101
102
103
104
105
106
107

Unit
ﾟC
ﾟF
%RH
vol%

Data

t/d
kg/d
g/d
m3/d
l/d
t/h
kg/h
g/h
m3/h
l/h

Unit
mN
N
N･m
J
kJ
mm/s
mm/min
mm/h
m/s
m/min
m/h
rps
rpm
rph
m/s2
rad/s
km/h

Unit
t/min
kg/ming/min
m3/min
l/min

18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35

t/s
kg/s
g/s
m3/s
l/s

Data
108
109
110
111
112
113
114
115
116
117
118
119
120
121
122
123
124
125

Unit
us
ms
s
min
h
day
mV
V
kV
uA
mA
A
Hz
dB
W
kW
VA
kVA

Data
36
37
38
39
40
41
42
43
44
45
46
47
48
49
50
51
52
53

mbar
bar
N/mm2
N/m2

Data
126
127
128
129
130
131
132
133
134
135
136
137
138
139
140
141
142
143

Unit
mPa
Pa
kPa
MPa

mm
cm
m

ml
L
kl
mm3
cm3
m3
Unit
Var
kVar
uS/cm
uF
F
C
mH
H
m ohm
ohm
k ohm
M ohm
lx
cd
lm
cd/m2

Note) Unit 1 to 12: Original unit.

Table 4 Masuring start, Masuring end setting limit
Input type
50mV
500mV
1-5V
0-5V

Masuring start, Masuring end limit–1000 to 5500 (–10.00 to 55.00mV)–100 to 5500 (–10.0 to 550.0mV)
500 to 5500 (0.500 to 5.500V)–100 to 5500 (–0.100 to 5.500V)

Table 5 Decimal point code
Decimal point data
0
1
2
3
4

Setting data
–32767 to 32767–3276.7 to 3276.7–327.67 to 327.67–32.767 to 32.767–3.2767 to 3.2767

－39－

Data
54
55
56
57
58
59
60
61
62
63
64
65
66
67
68
69
70
71
Data
144
145
146
147
148
149
150
151
152
153
154
155
156
157
158
159
160
161

Unit
mm2
cm2
m2
g
kg
t
g/cm3
kg/cm3
g/m3
kg/m3

g/l
kg/l
g/ml

Unit
uSv/h
mSv/h
nGy/h
uGy/h
um
Pa･s
mPa･s

*Unit 1
*Unit 2
*Unit 3
*Unit 4
*Unit 5
*Unit 6

Data
72
73
74
75
76
77
78
79
80
81
82
83
84
85
86
87
88
89

Unit
ppm
ppmNH3
ppmSO2
ppmH2S
ppmCO
ppmO2
ppmNOx
ppb
pH
mol
%
%H2
%CO2
%He
%Ar
%O2
%NaCl
%CO

Data
162
163
164
165
166
167

Unit
*Unit 7
*Unit 8
*Unit 9*Unit 10
*Unit 11
*Unit 12

Table 6 Data setting limit
・With Fvalue calculation OFF
Input type TC,Pt

K-Type TC
E-Type TC
J-Type TC
T-Type TC
R-Type TC
S-Type TC
B-Type TC
N-Type TC
W-Type TC
L-Type TC
U-Type TC
PN-Type TC
Pt100
JPt100℃ (Centigrade)
Range start, Range end
Alarm No.1 to 4 set point
Totalize cut value–2300 to 14000 (–230.0 to 1400.0℃)–2300 to 8300 (–230.0 to 830.0℃)–2300 to 11300 (–230.0 ot 1130.0℃)–2300 to 4300 (–230.0 to 430.0℃)–300 to 17900 (–30.0 to 1790.0℃)–300 to 17900 (–30.0 to 1790.0℃)
3700 to 17900 (370.0 to 1790.0℃)–300 to 13300 (–30.0 to 1330.0℃)–300 ot 17900 (–30.0 ot 1790.0℃)–2300 to 9300 (–230.0 to 930.0℃)–2300 to 4300 (–230.0 to 430.0℃)–300 to 13300 (–30.0 to 1330.0℃)–2300 to 6300 (–230.0 to 630.0℃)–2300 to 6300 (–230.0 to 630.0℃)

ﾟ F (Fahrenheit)
Range start, Range end
Alarm No.1 to 4 set point
Totalize cut value–3820 to 25520 (–382.0 to 2552.0 ﾟ F)
–3820 to 15260 (–382.0 to 1526.0 ﾟ F)
–3820 to 20660 (–382.0 to 2066.0 ﾟ F)
–3820 to 8060 (–382.0 to 806.0 ﾟ F)
–220 to 32540 (–22.0 to 3254.0 ﾟ F)
–220 to 32540 (–22.0 to 3254.0 ﾟ F)
6980 to 32540 (698.0 to 3254.0 ﾟ F)
–220 to 24260 (–22.0 to 2426.0 ﾟ F)
–220 to 32540 (–22.0 to 3254.0 ﾟ F)
–3820 to 17060 (–382.0 to 1706.0 ﾟ F)
–3820 to 8060 (–382.0 to 806.0 ﾟ F)
–220 to 24260 (–22.0 to 2426.0 ﾟ F)
–3820 to 11660 (–382.0 to 1166.0 ﾟ F)
–3820 to 11660 (–382.0 to 1166.0 ﾟ F)

Scaling OFF
Range start, Range end
Alarm No.1 to 4 set point
Totalize cut value–1000 to 5500 (–10.00 to 55.00mV)–100 to 5500 (–10.0 to 550.0mV)
500 to 5500 (0.500 to 5.500V)–100 to 5500 (–0.100 to 5.500V)

Scaling ON
Range start, Range end
Alarm No.1 to 4 set point
Totalize cut value

Input type Volt

50mV
500mV
1-5V
0-5V

・With Fvalue calculation ON or Input type other channel
Range start, Range end
Alarm No.1 to 4 set point
Totalize cut value–32767 to 32767 (Please refer to Table 15)

All type

Table 7 Totalize type code
Data
0
1
3
6
15

Totalize type
Daily
Monthly
AnnualExternal
OFF－40－–32767 to 32767 (Please refer to Table 5)Table 8 External input code
Data
0
1
2
3
4
5
6
7
8
9

Input
DI 1
DI 2
DI 3
DI 4
DI 5
DI 6
DI 7
DI 8
DI 9
DI 10

Data
10
11
12
13
14

Input
CH 1 Alarm No.1
CH 1 Alarm No.2
CH 1 Alarm No.3
CH 1 Alarm No.4
CH 2 Alarm No.1

126
127
128
129

CH30 Alarm No.1
CH30 Alarm No.2
CH30 Alarm No.3
CH30 Alarm No.4

Table9 Totalize reset input code
0

Input
None

1
2
3
4
5
6
7
8
9
10

DI 1
DI 2
DI 3
DI 4
DI 5
DI 6
DI 7
DI 8
DI 9
DI 10

Data

12
13
14
15

Input
CH 1 Alarm No.1
CH 1 Alarm
No.2
CH 1 Alarm No.3
CH 1 Alarm No.4
CH 2 Alarm No.1

127
128
129
130

CH30 Alarm №1
CH30 Alarm №2
CH30 Alarm №3
CH30 Alarm №4

11

……

Data

Table 10 Totalize cycle code
Data
0
1
2
3
4
5
6
7
8
9

Totalize type
10min
20min
30min
1hour
2hour
3hour
4hour
6hour
12our
24hourTable11 Formula code
Calculation data = four rules calculation data + function data * 256
Data
0
1
2
3
4
5
6
7
8
9
10
11
12
13

Function data
None
ABS
POW
SQR
LOG
LN
EXP
RH
MAX
MIN
H-P
L-P
AVG
SUM

Data
0
1
2
3
4

Four rules calculation data
Formula end+ ( Please set it to the 1st calculation )
*
/

－41－

argument1,2 data = argument type * 256 + data number
Data

Argument type

Data number limit

0

Input channel

0 to 29 (Channel 1 to 30 : C01 to C30 )1

Totalizer input

0 to 29 (Channel totalizer 1 to 30 : T01 to T30 )2

Digital input

0 to 9 (DI1 to 10 : D01 to D10 )3Communication input

0 to 11(Communication input 1 to 12 : M01 to M12)4

Constant

0 to 19 (Constant 1 to 20 : K01 to K20 )5

Temporary data

0 to 2 (Temporary data 1 to 3 : B01 to B03 )When setting “POW(C01, T02)*K03” in Formula 1 of Channel 19, set the following data.
Address

Data

Display

Breakdown(+)POW

Four rules calc. data : "+" (1)C01

Argument type : Input channel(0 * 256) + Data number : 1 (0)

42551

513

42552

0

42553

257

T02

Argument type : Totalizer input(1 * 256) + Data number : 2 (1)

42554

3

*

Four rules calc. data : "*" (3)

42555

1026

K03

Argument type : Constant(4 * 256) + Data number : 3 (2)

42556

0

(none)

42557

0

(End)

42558

0

(none)

42559

0(none)

Table12 Constant data
Decimal point data

Constant data
0-32767 to 32767

1-3276.7 to 3276.7

2-327.67 to 327.67

3-32.767 to 32.767

4-3.2767 to 3.2767

Table 13 Refreshment cycle code
Data
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17
18
19

Refreshment cycle
1sec
2sec
3sec
5sec
10sec
20sec
30sec
1min
2min
3min
5min
10min
20min
30min
1hour
2hour
3hour
4hour
6hour
12our

－42－

+

+

Function data : "POW"(2 * 256)

Function data : none (0 * 256)

Table14 File division cycle code
Data
0
1
2
3
4

File division cycle
No division
1 hour
1 day
1 week1 monthTable 15 FValue calculation decimal point code
Decimal point data
0
1
2
3
4

FValue calculation ON channel data
–32767 to 32767–3276.7 to 3276.7–327.67 to 327.67–32.767 to 32.767–3.2767 to 3.2767

Table 16 Message timing data
Message timing argument 1 and 2 have a significant difference according to the contents of Message timing.
Data
0
1
2
3
4

Message timing
None
DI ON
DI OFF
Alarm start
Alarm cancel

argument 1
None
0 to 9 ( DI1 to 10)
0 to 9 ( DI1 to 10)
0 to 17 ( channel 1 to 18 )0 to 17 ( channel 1 to 18 )argument 2
None
None
None
0 to 3 ( Alarm No.1 to 4 )0 to 3 ( Alarm No.1 to 4 )Table 17 DI function code
Data
0
1
2
3
4
5

DI function
Function invalid
Rec start/Rec stop
Fvalue calc. reset
Totalize start/stop
Totalize resetLCD ON

Table 18 Recorder control
Bit
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Contents
Record start/stop
Reserve
Totalize start/stop
LCD Lighting
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve

Write data
0:Record stop, 1:Record start
0:Totalize stop, 1:Totalize start
0: No change, 1:LCD Lighting－43－

Table 19 Message request
Bit
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Contents
Message No.1 request
Message No.2 request
Message No.3 request
Message No.4 request
Message No.5 request
Message No.6request
Message No.7request
Message No.8request
Message No.9request
Message No.10request
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve

Write data
0: No change,
0: No change,
0: No change,
0: No change,
0: No change,
0: No change,
0: No change,
0: No change,
0: No change,
0: No change,

Totalize reset of each Table 20 channel
Write data 0:No change, 1:Totalize reset request
Bit

Address 45051

Address 45052

0 Channel 1

Channel17

1 Channel 2

Channel18

2 Channel 3

Channel19

3 Channel 4

Channel20

4 Channel 5

Channel21

5 Channel 6

Channel22

6 Channel 7

Channel23

7 Channel 8

Channel24

8 Channel 9

Channel25

9 Channel10

Channel26

10 Channel11

Channel27

11 Channel12

Channel28

12Channel13

Channel29

13Channel14

Channel30

14Channel15

Reserve

15Channel16

Reserve－44－

1:Message request
1:Message request
1:Message request
1:Message request
1:Message request
1:Message request
1:Message request
1:Message request
1:Message request
1:Message request

Table 21 E-mail trigger timing data
Triggaer timing argument 1 and 2 have a significant difference according to the contents of Trigger timing.
Data
0
1
2
3
4
5

Trigger timing
None
DI ON
DI OFF
Alarm ON
Alarm OFF
Warning

6Timer cycle

Argument 1
None
0 to 9 (DI 1 to 10)
0 to 9 (DI 1 to 10)
0 to 17 (Channel 1 to 30)
0 to 17 (Channel 1 to 30)
0 to 30: Alarm ON (All ch)
1: All warning
2: No battery
3: CF full
0 to 6
0: 1 hour
1: 2 hour
2: 3 hour
3: 4 hour
4: 6 hour
5: 12 hour
6: 1 day

Table 22 Receiver’s mail address No.
Bit
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Contnets
E-mail address No. 1
E-mail address No. 2
E-mail address No. 3
E-mail address No. 4
E-mail address No. 5
E-mail address No. 6
E-mail address No. 7
E-mail address No. 8
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve

Data
0: No receive; 1: Address to receive
0: No receive; 1: Address to receive
0: No receive; 1: Address to receive
0: No receive; 1: Address to receive
0: No receive; 1: Address to receive
0: No receive; 1: Address to receive
0: No receive; 1: Address to receive
0: No receive; 1: Address to receive－45－

Argument 2
None
None
None
0 to 3 (Alarm No. 1 to 4)
0 to 3 (Alarm No. 1 to 4)
None

0 to 23(Base time 0:00 to 23:00)

Table 23 System information
Bit
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Contents
Recording status
CF card capacity
Channel alarming status
Reserve
Reserve
LCD state
Reserve
Totalizing condition
Battery condition
Reserve
CF card status
Reserve
Reserve
Reserve
Reserve
Reserve

Read data
0: Stop, 1: Recording
0: capacity available, 1: No capacity
0: OFF, 1: ON

0: ON, 1: OFF
0: Stop, 1: Totalizing
0: Provided, 1: Not provided
0: None, 1: Exist

Table 24 DO information
Bit
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Address 30003
Contents
DO 1 information
DO 2 information
DO 3 information
DO 4 information
DO 5 information
DO 6 information
DO 7 information
DO 8 information
DO 9 information
DO 10 information
DO 11 information
DO 12 information
DO 13 information
DO 14 information
DO 15 information
DO 16 information

Address 30004
Contents
DO 17 information
DO 18 information
DO 19 information
DO 20 information
DO 21 information
DO 22 information
DO 23 information
DO 24 information
DO 25 information
DO 26 information
DO 27 information
DO 28 information
Reserve
Reserve
Reserve
Reserve

Read data
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON

Table 25 DI information
Bit
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Contents
DI 1 information
DI 2 information
DI 3 information
DI 4 information
DI 5 information
DI 6 information
DI 7 information
DI 8 information
DI 9 information
DI 10 information
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve

Read data
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON

－46－

Read data
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON
0:OFF, 1:ON

Table 26 Channel Alarm information
All data
Bit

0:Alarm OFF, 1:Alarm ON

0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Address 30093
Channel 1
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 2
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 3
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 4
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4

Address 30094
Channel 5
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 6
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 7
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 8
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4

Address 30095
Channel 9
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 10 Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 11 Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 12 Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4

Address 30096
Channel 13 Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 14 Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 15 Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel 16 Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4

0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15

Address 30097
Channel17
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel18
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel19
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel20
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4

Address 30098
Channel21
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel22
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel23
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel24
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4

Address 30099
Channel25
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel26
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel27
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel28
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4

Address 30100
Channel29
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Channel30
Alarm No.1
Alarm No.2
Alarm No.3
Alarm No.4
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
ReserveBit－47－

8. SAMPLE PROGRAM
This section concerns data read-out/write-in sample program by GW-BASIC*1 which operated on Windows 98*1
MS-DOS*1 PROMPT.
Note that the program shown here is for reference for you to create a program and not for guaranteeing all actions.
Before executing the program, make sure of the communication conditions in the following procedure.
・ Communication speed (baud rate), data length, stop bits and parity bit:
Set in this program.
Match the conditions with this instrument.
*1:

GW-BASIC, Windows 98 and MS-DOS are the registered trademarks of Microsoft Corporation.

－48－

(a) Example of data read-out

Operation : allows measured values in Channel 1 to 9 to be read out at a time.
（Continuous word read-out from read-out only area)
Used function code
Read-out start register No.
Read-out word number: 04H: 30101:91000 '------------------------------------------------------1010 ' READ CONTINUOUS WORDSSAMPLE PROGRAM
1020 '------------------------------------------------------1030 '
1040 '
1050 '
1060 CLS
1070 DIM CC(255)1080 '
1100 '-------------- Send data setting ----------------------1110 CC(1)=&H01'Station No. = 11120 CC(2)=&H04'Function code = 04H
1130 CC(3)=&H00'Upper byte of relative address(0064H) of resister No.30101
1140 CC(4)=&H64'Lower byte of relative address(0064H) of resister No.30101
1150 CC(5)=&H00'Upper byte of read-out word number(0009H)
1160 CC(6)=&H09'Lower byte of read-out word number(0009H)
1170 COUNT=61200 '
1210 '------------- CRC code calculation of send data -------------1220 GOSUB 3020
1230 CC(7)=CRC.L'Lower byte of CRC calculation result -> Upper byte in message
1240 CC(8)=CRC.H'Upper byte of CRC calculation result -> Lower byte in message
1250 COUNT=COUNT+2
1300 '
1310 '------------- Send data ----------------------------1320 PRINT "Sending data > ";1330 OPEN "COM1:9600,o,8,1" AS #1 '9600bps, Odd parity, Data Length=8, Stop bit=11340 FOR I=1 TO COUNT
1350 PRINT #1,CHR$(CC(I));'Writing in transmission port
1360 PRINT RIGHT$("0"+HEX$(CC(I)),2);" "; 'Displaying on screen
1370 NEXT I
1380 '
1390 FOR I=O TO 300000 :NEXT I'Interval time1500 '
1510 '------------- Data receive ---------------------------1520 PRINT
1530 LENGTH=LOC(1)'Number of data in receiving buffer
1540 IF LENGTH=0 THEN PRINT "No answer" :END
1550 PRINT "Receiving data < ";1560 FOR I=1 TO LENGTH
1570 X$=INPUT$(1,#1)'Taking data from receiving buffer
1580 CC(I)=ASC(X$)'Digitizing and storing
1590 PRINT RIGHT$("0"+HEX$(CC(I)),2);" "; 'Displaying on screen
1600 NEXT I
1610 CLOSE #1
1620 COUNT=LENGTH-2
1630 GOSUB 3020
'GOSUB *CRC.CALC
1700 '
1710 '------------- Transmission error check ----------------------1720 PRINT
1730 CRC.L$=RIGHT$("0"+HEX$(CRC.L),2)
1740 CRC.H$=RIGHT$("0"+HEX$(CRC.H),2)
1750 PRINT "CRC calculation = ";CRC.L$;" ";CRC.H$
1760 IF CC(LENGTH-1)<>CRC.L THEN GOTO 1790
'GOTO *ER.MESSAGE
1770 IF CC(LENGTH)<>CRC.H THEN GOTO 1790
'GOTO *ER.MESSAGE
1780 GOTO 1920
'GOTO *PRT.RESULT
1790 '*ER.MESSAGE
1800 PRINT "Communication error"
1810 END
1900 '
1910 '------------- Display of result --------------------------1920 '*PRT.RESULT
1930 ' In case of decimal point position = 1 and unit = mV1940 PRINT－49－

1950 CH1$=HEX$(CC(4))+RIGHT$("0"+HEX$(CC(5)),2)'2byte -> 1word
1960 CH2$=HEX$(CC(6))+RIGHT$("0"+HEX$(CC(7)),2)'2byte -> 1word
1970 CH3$=HEX$(CC(8))+RIGHT$("0"+HEX$(CC(9)),2)'2byte -> 1word
1980 CH4$=HEX$(CC(10))+RIGHT$("0"+HEX$(CC(11)),2) '2byte -> 1word
1990 CH5$=HEX$(CC(12))+RIGHT$("0"+HEX$(CC(13)),2) '2byte -> 1word
2000 CH6$=HEX$(CC(14))+RIGHT$("0"+HEX$(CC(15)),2) '2byte -> 1word
2010 CH7$=HEX$(CC(16))+RIGHT$("0"+HEX$(CC(17)),2) '2byte -> 1word
2020 CH8$=HEX$(CC(18))+RIGHT$("0"+HEX$(CC(19)),2) '2byte -> 1word
2030 CH9$=HEX$(CC(20))+RIGHT$("0"+HEX$(CC(21)),2) '2byte -> 1word
2040 PRINT "CH1 =";VAL("&H"+CH1$)/10;"mV"'1 place of decimal
2050 PRINT "CH2 =";VAL("&H"+CH2$)/10;"mV"'1 place of decimal
2060 PRINT "CH3 =";VAL("&H"+CH3$)/10;"mV"'1 place of decimal
2070 PRINT "CH4 =";VAL("&H"+CH4$)/10;"mV"'1 place of decimal
2080 PRINT "CH5 =";VAL("&H"+CH5$)/10;"mV"'1 place of decimal
2090 PRINT "CH6 =";VAL("&H"+CH6$)/10;"mV"'1 place of decimal
2100 PRINT "CH7 =";VAL("&H"+CH7$)/10;"mV"'1 place of decimal
2110 PRINT "CH8 =";VAL("&H"+CH8$)/10;"mV"'1 place of decimal
2120 PRINT "CH9 =";VAL("&H"+CH9$)/10;"mV"'1 place of decimal
2130 END
3000 '
3010 '------------ CRC calculation -----------------------------3020 '*CRC.CALC'For contents, refer to CRC calculation flow chart
3030 CR=&HFFFF
3040 FOR I=1 TO COUNT
3050 CR=CR XOR CC(I)3060 FOR J=1 TO 8
3070
CT=CR AND &H1
3080
IF CR<0 THEN CH=1 ELSE CH=0:GOTO 3100
3090
CR=CR AND &H7FFF
3100'*CRC.CALC.10
3110
CR=INT(CR/2)
3120
IF CH=1 THEN CR=CR OR &H4000
3130
IF CT=1 THEN CR=CR XOR &HA001
3140 NEXT J
3150 NEXT I
3160 CRC.L=CR AND &HFF
'Lower byte of CRC calculation
3170 CRC.H=((CR AND &HFF00)/256 AND &HFF)'Upper byte of CRC calculation
3180 RETURN－50－

(b) Data write-in example

Operation : allows alarm latch in Station No. 1 to be released via communication.
（Word write-in）
Used function code
Write-in start register No.
Write-in word number
Write-in data

: 10H
: 45007:1: 1 (Alarm latch clear)

1000 '------------------------------------------------------1010 ' WRITE CONTINUOUS WORDSSAMPLE PROGRAM
1020 '------------------------------------------------------1030 '
1040 '
1050 '
1060 CLS
1070 DIM CC(255)1080 '
1100 '-------------- Send data setting ----------------------1110 CC(1)=&H01'Station No. = 11120 CC(2)=&H10'Function code = 10H
1130 CC(3)=&H13'Upper byte of relative address(138EH) of resister No.45007
1140 CC(4)=&H8E'Lower byte of relative address(138EH) of resister No.45007
1150 CC(5)=&H00'Upper byte of write-in word number(0001H)
1160 CC(6)=&H01'Lower byte of write-in word number(0001H)
1170 CC(7)=&H02'Write-in word number * 21180 CC(8)=&H00'Upper byte of write-in data(0001H)
1190 CC(9)=&H01'Lower byte of write-in data(0001H)
1200 COUNT=91210 '------------- CRC code calculation of send data -------------1220 GOSUB 3020
1230 CC(10)=CRC.L'Lower byte of CRC calculation result -> Upper byte inmessage
1240 CC(11)=CRC.H'Upper byte of CRC calculation result -> Lower byte inmessage
1250 COUNT=COUNT+2
1300 '
1310 '------------- Send data ----------------------------1320 PRINT "Sending data > ";1330 OPEN "COM1:9600,o,8,1" AS #1 '9600bps, Odd parity, Data Length=8, Stop bit=11340 FOR I=1 TO COUNT
1350 PRINT #1,CHR$(CC(I));'Writing in transmission port
1360 PRINT RIGHT$("0"+HEX$(CC(I)),2);" "; 'Displaying on screen
1370 NEXT I
1380 '
1390 FOR I=O TO 300000 :NEXT I'Interval time1500 '
1510 '------------- Data receive ---------------------------1520 PRINT
1530 LENGTH=LOC(1)'Number of data in receiving buffer
1540 IF LENGTH=0 THEN PRINT "No answer" :END
1550 PRINT "Receiving data < ";1560 FOR I=1 TO LENGTH
1570 X$=INPUT$(1,#1)'Taking data from receiving buffer
1580 CC(I)=ASC(X$)'Digitizing and storing
1590 PRINT RIGHT$("0"+HEX$(CC(I)),2);" "; 'Displaying on screen
1600 NEXT I
1610 CLOSE #1
1620 COUNT=LENGTH-2
1630 GOSUB 3020
'GOSUB *CRC.CALC
1700 '
1710 '------------- Transmission error check ----------------------1720 PRINT
1730 CRC.L$=RIGHT$("0"+HEX$(CRC.L),2)
1740 CRC.H$=RIGHT$("0"+HEX$(CRC.H),2)
1750 PRINT "CRC calculation = ";CRC.L$;" ";CRC.H$
1760 IF CC(LENGTH-1)<>CRC.L THEN GOTO 1790
'GOTO *ER.MESSAGE
1770 IF CC(LENGTH)<>CRC.H THEN GOTO 1790
'GOTO *ER.MESSAGE
1780 GOTO 1920
'GOTO *PRT.RESULT
1790 '*ER.MESSAGE
1800 PRINT "Communication error"－51－

1810 END
1900 '
1910 '------------- Display of result --------------------------1920 '*PRT.RESULT
1930 PRINT
1940 PRINT "Completion of alarm latch clear"1950 END
3000 '
3010 '------------ CRC calculation -----------------------------3020 '*CRC.CALC'For contents, refer to CRC calculation flow chart
3030 CR=&HFFFF
3040 FOR I=1 TO COUNT
3050 CR=CR XOR CC(I)3060 FOR J=1 TO 8
3070
CT=CR AND &H1
3080
IF CR<0 THEN CH=1 ELSE CH=0:GOTO 3100
3090
CR=CR AND &H7FFF
3100'*CRC.CALC.10
3110
CR=INT(CR/2)
3120
IF CH=1 THEN CR=CR OR &H4000
3130
IF CT=1 THEN CR=CR XOR &HA001
3140 NEXT J
3150 NEXT I
3160 CRC.L=CR AND &HFF
'Lower byte of CRC calculation
3170 CRC.H=((CR AND &HFF00)/256 AND &HFF)'Upper byte of CRC calculation
3180 RETURN－52－

9. ETHERNET COMMUNICATIONFUNCTIONSFTP server function, web server function, E-mail send function and MODBUS TCP/IP function can be used for
Ethernet communication.
Setting of IP address, etc. is essential for connection of a paperless recorder to Ethernet. Be sure to consult with thesystem manager of your company.

9.1 LAN port specification
Item
Transmission rate
Transmission method
Maximum network length or maximum node interval
Maximum segment length
Cable for connection
ProtocolSpecification 10BASE-T
10 MbpsBase band
500 m (cascade in 4 stages)
100 m (between node and hub)UTP (twisted-pair cable without shield) 22-26 AWG
TCP/IP

9.2 Connection to the terminal
Ethernet terminal

9.3 Connection

Ethernet

HUB
10 BASE-T
straight cable

Node to hub distance:
Maximum number of nodes per network:
Recommended cable:

Up to 100 m
100 nodes
10 BASE-T twisted-pair cable, Category 5－53－

9.4 Setting Ethernet communicating conditions•

Set IP address, subnet mask and default gateway for connection of the paperless recorder to Ethernet.
with the system manager of your company for the values to be set.)(Consult•

Communicating conditions setting items
Item
Value at delivery
IP address
192.168. 1. 1
Sugnet mask
255.255.255. 0
Default gateway
0. 0. 0. 0

Setting range
0 to 255 for each digit
0 to 255 for each digit
0 to 255 for each digit

Remarks
Turn ON the power again after
setting change.

9.5 Ethernet communicating conditions setting operation(1) Select Main unit” → “Ethernet communication setting” on the parameter setting screen, and the “Ethernet
communication setting” screen appears.

(2) Select “Ethernet setting 1”, and the following screen appears.

(3) Move the cursor to “IP address” and set an IP address.
(4) Move the cursor to “Subnet mask”, and set a subnet mask.
(5) Move the cursor to “Default gateway”, and set a default gateway.

－54－

10. FTP SERVER FUNCTION
10.1 Description of FTP server function•

This function permits take-out of record files from the compact flash of the recorder, using browser or DOS
prompt.

•This function permits take-out the record data from the internal memory of the recorder, using browser or DOS
prompt.

•This function permits deletion of record files from the compact flash of the recorder, using browser or DOS
prompt. (This function is available only to a user of administration level.)•This function permits changing names of record files recorded on the compact flash of the recorder, usingbrowser or DOS prompt. (This function is available only to a user of administration level.)•

Use Internet Explorer made by Microsoft as the personal computer's browser.

•

Up to eight (8) user names and passwords may be set for those who are permitted to log in the FTP server.

•If the FTP access control function is OFF, log-in to the FTP server is permitted with common user name “ftp”(without password).

•

When log-in or log-off to/from the FTP server is executed, the subject information is displayed on the Ethernet
log screen.

•

The FTP server permits log-in by one user only at a time.

•

The recorded data of an internal memory record meter becomes a binary form regardless of "Recorded dataform" setting of the main body.

•The folder structure of the FTP server is shown in the figure below[Caution]• The communication automatically disconnects, if no FTP communication request is made for 10 minutes.
•

The display motion of the paperless recorder may slow down when taking out a file of large size.

•

While the compact flash of the paperless recorder is accessed by FTP communication, do not take out thecompact flash.
Furthermore, when the FTP server function is used, inhibit access to the compact flash in the "Memory card
abstract" screen, before taking out the compact flash.

•

Do not delete or change the name of a file while the file is being recorded or integrated.

•

Attributes of all files in the FTP server are displayed as read-only as hidden files.

•

If the Ethernet communication is shut down while the FTP server is in log-in status, log-in is not permitted
until the communication is automatically disconnected ten (10) minutes later.

•

Data files cannot be written into the compact flash installed in the paperless recorder via FTP
communication.

•

Update time that appears on the list of files contained in the compact flash may differ from the actual update
time if the file list is displayed via FTP communication.

－55－

FTP server folder configuration
Compact flash card
Root
Folder

S000000

S******.FDT

Record file

A******.FDT

Event file

Folder name
T000000

File nameT******.FDT

Daily totalizing data file

D******.FDT

Monthly totalizing data file

M******.FDT

Annual totalizing data file

E******.FDT

External signal totalizing data file

PA00000.PHR

Setting value file

Folder name
PARAM

Internal memory (The name of record data in the internal memory is fixed.)

MEMORY

SI00000.FDT, SI00001.FDT, SI00002.FDT, SI00003.FDT
Record data of internal memory (4 pieces max.)
AI00000.FDT, AI00001.FDT, AI00002.FDT, AI00003.FDT
Event data of internal memory (4 pieces max.)

Folder name

File name

T000000

TI00000.FDT, TI00001.FDT
DI00000.FDT, DI00001.FDT
MI00000.FDT, MI00001.FDT
EI00000.FDT, EI00001.FDT－56－

Daily totalizing data file of internal memory(2 pieces max.)
Monthly totalizing data file of internal memory(2 pieces max.)
Annual totalizing data file of internal memory(2 pieces max.)External input totalizing data of internal
memory(2 pieces max.)

10.2 Setting FTP server function•

Execute setting of FTP server function and of access verification, for using the FTP server function.
Furthermore, set names and passwords of those who use the FTP server function.

•

FTP server function setting items
Item
FTP server function
FTP access control

•

OFF
OFF

Value at delivery

Setting range
ON, OFFON, OFF

Value at delivery(Blank)(Blank)Administrator

Setting range
Up to 16 letters may be set.
Up to 8 letters may be set.
Administrator, guest

Remarks
Turn ON the power again
after setting is changed.

User name setting items
Item
User 1 to 8 name
User 1 to 8 password
User 1 to 8 level

Remarks

10.3 FTP server function setting operation• FTP server function setting(1) Select “Main unit” → “Ethernet communication setting” → “Ethernet setting 2” on the parameter setting screen,
and the following screen appears.

(2) Move the cursor to “FTP server function”, and select FTP server function ON/OFF. The FTP server function
can be used, if ON is set.
(3) Move the cursor to “FTP access control”, and select FTP access control function ON/OFF. No password isrequired at the time of log-in to the FTP server, if OFF is set.

－57－

• User name setting(1) Select “Main unit” → “Ethernet communication setting” → “User account setting” on the parameter setting
screen, and the following screen appears.

(2) Select a user No. to be set.
(3) Set a user name using up to 16 letters.
(4) Set a password using up to 8 letters.
(5) Select a user access level out of “administrator” and “guest”.
If “guest” was selected, file deletion is not permitted, although log-in to the FTP server is permitted.

10.4 FTP server operation•

Connect the FTP server to the paperless recorder from the browser, by performing operation in the sequence
indicated below.
(1) Start Internet Explorer from a personal computer on Ethernet.
(2) Enter the IP address of the paperless recorder in the address column in the following manner.
FTP: //(paperless recorder's IP address)
Enter FTP: //192.168.1.2 in case the IP address of the paperless recorder is 192.168.1.2.
(3) The screen that requests entry of user name and password appears. Enter a user name and password.
(4) "S000000" and "MEMORY" directory are displayed on a browser.
Select "S000000" folder when you want to display the file of the record in CompactFlash.
Select "MEMORY" folder when you want to display recorded data in an internal memory.
(5) The record file recorded on a browser is displayed.
(6) Select the file to be taken out, and copy it onto an arbitrary folder in the personal computer.
(7) When CompactFlash has been selected, the deletion of the record file is also possible.
Select a desired file and delete it.

－58－

11. Web server function
11.1 Description of web server function•

The web server function permits monitoring of measured values and event log in the paperless recorder using
personal computer's browser.

•

Use Internet Explorer made by Microsoft as the personal computer's browser.

[Caution] • Monitoring from cell phone's browser is not permitted. If connection to recorder's web server ismade from a cell phone, the recorder may halt in the worst case. Do not attempt to monitor data inthe paperless recorder from a cell phone.
• An error may arise depending on the circumstances of the communication, as the period of update of
the browser is 10 seconds.
The screen of the PC is displayed again, if the update button of the browser is pressed in such a case.
• The characters may not be displayed normally depending on the setting of the browser.
• With the type of 9-channel specifications, 0.0°C is displayed as the reading of channels 10 to 18 on
the measured value display screen.

11.2 Setting web server function•

Set the web server function for permitting its use.

•

Set items
Item
Web server function

Value at delivery
OFF

Setting range
ON, OFF

Remarks
Turn ON the power again after setting is changed.

11.3 Web server function setting operation(1) Select “Main unit” → “Ethernet communication setting” → “Ethernet setting 2” on the parameter setting screen,
and the follow screen appears.

(2) Move the cursor to “Web server function”, and select web server function ON/OFF.
function is permitted if ON is set.

－59－

Use of the web server11.4 Web server operation•

Connect the web server to the paperless recorder from the personal computer's browser, by performing
operation in the sequence indicated below.
(1) Start Internet Explorer from a personal computer on Ethernet.
(2) Enter the IP address of the paperless recorder in the address column in the following manner.
http: //(paperless recorder's IP address)
Enter http: //192.168.1.2 in case the IP address of the paperless recorder is 192.168.1.2.
(3) The paperless recorder monitor screen is displayed on the browser.

11.5 Web monitor screen•

Connect to the web server of the paperless recorder, and the following screen appears.
(1) Web monitor menu screen－60－(2) PV display screen(3) Totalize display screen－61－(4) Event Summary display screen－62－12. E-MAIL SEND FUNCTION
12.1 Description of E-mail send function•

E-mails can be transmitted from the paperless recorder.

•

E-mails can be transmitted in any of the states indicated below.
(1) An alarm arose or was cancelled.
(2) An external input (DI) was ON or OFF.
(3) Any error occurred to the main unit. (When battery end, memory card full, or one of the alarms of all thechannels should occur)(4) Once every fixed period (The period may be selected out of 1 hour, 2 hours, 3 hours, 4 hours, 6 hours, 12hours and 24 hours.)•

Up to eight (8) addresses can be registered for transmission of E-mails.

•

Measured values of each channel can be attached to each E-mail.

(Receipt of E-mails is not permitted.)• E-mail transmission test can be conducted in the E-mail trigger setting screen.
[Caution]•

Up to sixteen (16) E-mail send requests can be transmitted continuously, but not more than 16.
No transmission will be implemented if the number of E-mail send requests exceeds 16. Therefore, makesetting so that E-mail send requests will not occur continuously.

•

For sending E-mails, it is necessary to register the paperless recorder in the mail server.
Consult with the system manager of your company, for registration to the mail server.

•

If E-mail send fails, the E-mail send requests are cancelled.

•

Even if E-mail is sent, there is a possibility where the E-mail does not reach the destination because ofincorrect address, etc.

•

If two or more E-mail addresses are set as send destinations in the E-mail trigger setting, the error message
is not recorded on the Ethernet log screen unless all the attempts to send E-mails fail.

12.2 Setting E-mail function•

Set E-mail send/receive addresses and E-mail send trigger, for permitting use of the E-mail function.
(Consult with the system manager of your company, for the values to be set.)•E-mail function set items
Item
E-mail function•

Value at delivery
OFF

Setting range
ON, OFF

Remarks
Turn ON the power again after settingchange.

E-mail send/receive address set items
Item
SMPT IP address
Sender’s mail address
Sender’s mail name
Receiver’s mail address
1 to 8

Value at delivery
0. 0. 0. 0
(Blank)(Blank)(Blank)Setting range
0 to 255 for each digit
Up to 64 letters may be set.
Up to 32 letters may be set.
Up to 64 letters may be set.

－63－

Remarks•E-mail send trigger set items
Item
Title
Trigger timing

Value at delivery(Blank)None

DI No.
Alarm Channel
Alarm No.
Warning type

DI 1
Channel 1
1
Alarm ON (All ch)

Time cycle
Time base (hour)
Text 1
Text 2
PV value affixation
Receiver’s add No.

1 hour
0:00(Blank)(Blank)OFF
None

Setting range
Up to 32 letters may be set.
None, DI ON, DI OFF,
Alarm ON, Alarm OFF,
Warning, Timer cycle
DI 1 to 10
Channel 1 to 18
1 to 4
Alarm ON (All ch),All warning, No battery,CF full
1, 2, 3, 4, 6, 12 hour, 1 day
0:00 to 23:00
Up to 32 letters may be set.
Up to 32 letters may be set.
ON, OFF
Receiver’s address
No. 1, 2, 3, 4, 5, 6, 7, 8

Remarks

Trigger timing = DI ON, DI OFF
Trigger timing = Alarm ON, OFF
Trigger timing = Warning
Trigger timing = Timer cycle

12.3 E-mail function setting operation• Setting E-mail function(1) Select “Main unit” → “Ethernet communication setting” → “Ethernet setting 2” on the parameter setting screen,
and the following screen appears.

(2) Move the cursor to “E-mail function”, and select E-mail function ON/OFF.
used, if ON is set.

－64－

The E-mail send function can be• Setting E-mail send/receive addresses(1) Select “Main unit” → “Ethernet communication setting” → “E-mail setting 1” on the parameter setting screen,
and the following screen appears.

(2) Move the cursor to “SMTP IP address”, and set the IP address of the mail server.
(3) Move the cursor to “Sender’s mail Add”, and set the sender's mail address.
(4) Move the cursor to “Sender’s mail Name”, and set the sender's name.
(5) Select “Main unit” → “Ethernet communication setting” → “E-mail setting 2” on the parameter setting screen,
and the following screen appears.

(6) Move the cursor to “Receiver’s mail Add”, and set up to eight (8) receivers' mail addresses.

－65－• Setting E-mail send trigger(1) Select “Main unit” → “Ethernet communication setting” → “E-mail trigger setting” on the parameter setting
screen, and the following screen appears.

(2) Up to ten (10) patterns of E-mail send timing may be set.
Select a send timing to be selected, and press the [ENT] key.
(3) Move the cursor to “Title”, and set the E-mail title.
(4) Move the cursor to “Trigger timing”, and select an E-mail send timing.

Select one of the followings as an E-mail send timing.
When a timing is selected, particulars set items are displayed for each timing type. Set these items also.
1) DI ON, DI OFF
E-mails can be sent by DI ON/OFF.
When DI ON/OFF timing is selected, set items for the DI No. to be used are displayed. Set the DI No. to be
used for judgment.
2) Alarm ON, Alarm OFF
E-mails can be sent by the alarm occur/cancel information.
When alarm ON/OFF timing is selected, set items for the channel No. and alarm No. to be used are displayed.
Set the channel No. and alarm No. to be used for judgment.
3) Warning
E-mails can be sent by warning occur information.
When warning is selected, set items for the warning information to be used are displayed. Set the warning
information to be used for judgment.

－66－4) Timer cycle
E-mails can be sent in a fixed period.
When timer cycle is selected, set items for the send period and reference time are displayed. Set the Email send period and reference time.
(5) Move the cursor to “Text 1”, “Text 2”, and set a comment of two (2) lines to be described in the E-mail.
(6) Move the cursor to “PV value affixation”, and set whether to indicate measured values of all the channels in theE-mail. All the channels can be indicated , ON is set.
(7) Move the cursor to “Receiver’s add No.”, and select an address No. to receive the E-mail.
The E-mail is sent to each address No. for which ON was set.

12.4 E-mail send test operation•

Conduct an E-mail send test with the paperless recorder, by performing operation in the sequence indicatedbelow.
(1) Select “Main unit” → “Ethernet communication setting” → “E-mail trigger setting”.
(2) Select an E-mail trigger setting No. to conduct a send test.
(3) Move the cursor to “Mail send test”, and press the [ENT] key. Then, an E-mail send test can be conducted.

12.5 E-mail send contents•

The paperless recorder sends an E-mail with following contents.
Sender's mail nameMail title

Mail trigger timing & time

2002/ 4/ 3 20:00:00 Operational report
1hour
PVON

Mail text 1
Mail text 2
PV value

CH1 = 862.6mH
CH2 = 862.5mm/s
CH3 = 862.5mV
CH4 = 862.5mV
CH5 = 6.270V
CH6 = 6.270V
CH7 = 6.270V
CH8 = 6.270V
CH9 = 6.270V－67－

13. MODBUS TCP/IP FUNCTION
13.1 Description of MODBUS TCP/IP function•

The MODBUS TCP/IP protocol permits use of MODBUS protocol (MODBUS RTU), which is used with RS485 interface, on an Ethernet interface.
* See Chapter 5 through Chapter 7 for MODBUS protocol (MODBUS RTU) of RS-485 interface.

•

MODBUS TCP/IP communication is executed through port 502.

•

The MODBUS TCP/IP function permits read/write of set values from/to the paperless recorder.

13.2 Setting MODBUS TCP/IP function•

Make MODBUS TCP/IP function setting to permit the use of MODBUS TCP/IP function.

•

Specify station No. to evaluate the device with which communication is to be carried out.

•

Set items
Item
MODBUS TCP/IP
MODBUS Station NO.

Factory default
OFF
1

Setting range
ON, OFF
0 to 255

Remarks
Turn on the power after the setting is changed.
Communication is not carried out if 0 is selected.

13.3 MODBUS TCP/IP function setting operation(1) Select Select “Main unit” → “Ethernet communication setting” → “Ethernet setting 2” on the parameter setting
screen, and the following screen appears.

(2) Move the cursor to “MODBUS TCP/IP”, and select MODBUS TCP/IP function ON/OFF.
TCP/IP function can be used, if ON is set.
(3) Select “Main unit” → “RS485 communication setting” on the parameter setting screen.
(4) Move the cursor to “MODBUS Station No.” and select a desired station No.

－68－

The MODBUS

13.4 MODBUS TCP/IP communication protocol•

The MODBUS TCP/IP communication protocol permits use of MODBUS protocol, which is used with RS-485
interface, on the same interface.
* See Chapter 5 through Chapter 7 for MODBUS protocol (MODBUS RTU) of RS-485 interface.

•

MODBUS TCP/IP communication is executed through port 502.

•

A message used by the MODBUS TCP/IP communication protocol is what is produced by deleting two bytes oferror check code MODBUS TCP/IP from a message used by MODBUS RTU and by adding a 6-byte header.
A MODBUS RTU message can be converted to a message for MODBUS TCP/IP as indicated below.
Data section

CRC

A

MODBUS RTU message
MODBUS TCP/IP message•

Data section

The header of a MODBUS TCP/IP message is composed of transfer ID, protocol ID and total number of bytesof the data section.
Byte No.
Data content

01
Transaction Identifier

Data meaning:
Transaction Identifier:Protocol Identifier:Length Unit Identifier:•

Header2
3
Protocol Identifier

4
5
Length Unit Identifier

Identification of a MODBUS Request/Response transaction.
Set 0 for MODBUS TCP/IP.
Number of bytes of data part

Conversion of what is shown in Chapter 6.2 (3) Message transmission (example) into a MODBUS TCP/IP
message, for instance, can be accomplished as described below.
For reading the measured value of channel 2 from station No. 1:
Relative address of channel 2 measured value: 0065H (register No. 30102), number of data: 01H

Composition of command message (bytes)
MODBUS RTU message (bytes)

Datasection

CRC

Station No.
Function code
Read start No.
(relative address)
Number of readdata
CRC data

More significant
Less significant
More significant
Less significant
More significant
Less significant

MODBUS TCP/IP message (bytes)
More significant
Transfer ID
HeaderLess significant
More significant
Protocol ID
Less significant
More significant
Number of bytes
Less significant
Station No.
Data
Function code
sectionRead start No. More significant(relative address) Less significant
Number of read More significant
dataLess significant

01H
04H
00H
65H
00H
01H
21H
D5H

－69－

00H
00H
00H
00H
00H
06H
01H
04H
00H
65H
00H
01H

Composition of response message (bytes)
MODBUS RTU message (bytes)

Datasection

CRC•

Station No.
Function code
Number of bytes of read data
Contents of first More significant
word data
Less significant
More significant
CRC data
Less significant

MODBUS TCP/IP message (bytes)
More significant
Transfer ID
HeaderLess significant
More significant
Protocol ID
Less significant
More significant
Number of bytes
Less significant
Station No.
Data
Function code
sectionNumber of bytes of read data
Contents of first More significant
word data
Less significant

01H
04H
02H
01H
4FH
F9H
54H

00H
00H
00H
00H
00H
05H
01H
04H
02H
01H
4FH

The following error code is returned on occurrence of a communication error.

01H

Error code

Content
Illegal function code

02H

Illegal data address

03H

Illegal data number

04H

Device error

Explanation
Non-actual function code is designated.
Check for the function code.
A relative address of a resister number to which the designated function code
can not be used.
Because the designation of number is too much, the area where resister
numbers do not exist is designated.
Communication with slave equipment failed. Check the communication
specification for the slave equipment.

－70－

14. TROUBLESHOOTING
If the communication is unavailable, check the following items.
Case of RS-485 MODBUS communication
Whether the power is turned ON again after communication setup change.
Whether all devices related to communication are turned on.
Whether connections are correct.
Whether the number of connected instruments and connection distance are as specified.
Whether communication conditions coincide between the master station (host computer) and slave stations(GR200).
Transmission speed :9600bps
19200bps
Data length: 8 bits
Stop bit: 1 bit
Parity
:
odd
even
none
Whether send/receive signal timing conforms to Section 5.4 in this manual.
Whether the station No. designated as send destination by the master station coincides with the station No. of
the connected GR200.
Whether more than one instrument connected on the same transmission line shares the same station No.
Whether the station No. of instruments is set at other than 0.
If it is 0, the communication function does not work.
Whether the 12th digit of type cord of this Recorder is R or W ?
R(GR200W - )Case of Ethernet communication (common to FTP, web, E-mail and MODBUS TCP/IP)
Whether the power is turned ON again after communication setup change.
Whether all devices related to communication are turned ON.
Whether connections are correct.
Whether the number of connected instruments and connection distance are as specified.
Whether conditions for communication are correct.
IP address
Subnet mask
Default gateway
Whether the 12th digit of type code of this Recorder is E or W?
E(GR200W - )Case of FTP server function
Whether the user name, the password, and the user level are correct?
Whether a compact flash has been inserted to the main unit.
Case of E-mail send function
Whether conditions for communication are correct.
SMTP address
Sender’s mail address
Receiver’s mail address
Whether E-mail send conditions are correct.

－71－

Case of MODBUS TCP/IP communication function
Whether the station No. designated as send destination by the master station coincides with the station No. of
this Recorder been connected.
Whether the station No. of this Recorder is set other than 0.
If it is 0, the communication function does not work.

－72－

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Reg. Office : 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel: 81-72-727-6100
Fax: 81-72-727-7006
URL: http://www.shinko-technos.co.jp
E-mail: overseas@shinko-technos.co.jp