DIGITAL INDICATING CONTROLLER

FCR-13A
INSTRUCTION MANUALPreface

Thank you for purchasing our Digital indicating controller FCR-13A.
This manual contains instructions for the mounting, functions, operations and notes when
operating the FCR-13A.
To prevent accidents arising from the misuse of this controller, please ensure the operator
receives this manual.

Abbreviations Used in This Manual
Symbol
PV
SV
MV
OUT1
OUT2
AT
DC input

Term
Process variable
Desired value
Output manipulated variable
Control output 1
Control output 2 (option)
Auto-tuning
Direct current input
DC voltage input

Characters Used in This Manual
Indication
Number,

/

-1

0

1

2

3

4

5

6

7

8

9

A

B

C

D

E

F

G

H

I

J

K

L

M

N

O

P

Q

R

S

T

U

V

W

X

Y

Z

Indication
Alphabet
Indication
Alphabet

2

Notes• This instrument should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed, serious injury or
malfunction may occur.
• Specifications of the FCR-13A and the contents of this instruction manual are subject to change
without notice.
• Care has been taken to assure that the contents of this instruction manual are correct, but if
there are any doubts, mistakes or questions, please inform our sales department.
• This instrument is designed to be installed through a control panel. If it is not, measures must be
taken to ensure that the operator cannot touch power terminals or other high voltage sections.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos CO., LTD. is not liable for any damage or secondary damage(s) incurred as aresult of using this product, including any indirect damage.

SAFETY PRECAUTIONS (Be sure to read these precautions before using our products.)
The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on the circumstances, procedures indicated by
Caution may cause serious
results, so be sure to follow the directions for usage.

Warning

Procedures which may lead to dangerous conditions and cause
death or serious injury, if not carried out properly.

CautionProcedures which may lead to dangerous conditions and cause
superficial to medium injury or physical damage or may degrade
or damage the product, if not carried out properly.

Warning• To prevent an electric shock or fire, only Shinko or other qualified service personnel may
handle the inner assembly.
• To prevent an electric shock, fire or damage to the instrument, parts replacement may only
be undertaken by Shinko or other qualified service personnel.

SAFETY PRECAUTIONS• To ensure safe and correct use, thoroughly read and understand this manual before using
this instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring
equipment. Verify correct usage after purpose-of-use consultation with our agency or main office.
(Never use this instrument for medical purposes with which human lives are involved.)• External protection devices such as protection equipment against excessive temperature rise,etc. must be installed, as malfunction of this product could result in serious damage to thesystem or injury to personnel. Also proper periodic maintenance is required.
• This instrument must be used under the conditions and environment described in this manual.
Shinko Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring
due to the instrument being used under conditions not otherwise stated in this manual.
Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the
manufacture of weapons of mass destruction (i.e. military applications, military equipment,etc.), please investigate the end users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.
3

1. Installation Precautions

CautionThis instrument is intended to be used under the following environmental
conditions (IEC61010-1): Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of 0 to 50 (32 to 122 )that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85%RH• No large capacity electromagnetic switches or cables through which large current isflowing.
• No water, oil or chemicals or where the vapors of these substances can come into
direct contact with the unit
• Take note that ambient temperature of this unit - not the ambient temperature of the control
panel - must not exceed 50 (122 ) if mounted through the control panel. Otherwise the lifeof electronic components (especially electrolytic capacitor) may be shortened.
Note: Avoid setting this instrument directly on or near flammable material eventhough the case of this instrument is made of flame-resistant resin.

2. Wiring Precautions

Caution• Do not leave bits of wire in the instrument, because they could cause a fire or malfunction.
• Use the solderless terminal with an insulation sleeve in which an M3 screw fits whenwiring the FCR-13A.
• The terminal block of this instrument is designed to be wired from the left side.
The lead wire must be inserted from the left side of the terminal, and fastened with theterminal screw.
• Tighten the terminal screw within the specified torque. If excessive force is applied tothe screw when tightening, the screw or case may be damaged.
• Do not apply a commercial power source to the sensor connected to the input terminal
nor allow the power source to come into contact with the sensor, as the input circuit
may be burnt out.
• This controller has no built-in power switch, circuit breaker or fuse. It is necessary to
install them near the controller.
(Recommended fuse: Time-lag fuse, rated voltage 250V AC, rated current 2A)• For a 24V AC/DC power source, do not confuse polarity when using direct current (DC).

3. Operation and Maintenance Precautions

Caution• It is recommended that AT (auto-tuning) be performed on the trial run.
• Do not touch live terminals. This may cause electric shock or problems in operation.
• Turn the power supply to the instrument OFF when retightening the terminal andcleaning. Working on or touching the terminal with the power switched ON may result
in severe injury or death due to electric shock.
• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)• As the display section is vulnerable, do not strike or scratch it with a hard object or put
pressure on it.
4

--- CONTENTS --1. Model
1.1 Model ---------------------------------------------------------------------------------------- 8
1.2 Rated Input --------------------------------------------------------------------------------- 9
1.3 How to Read the Model Label --------------------------------------------------------- 9

2. Name and Functions of Sections ----------------------------------------------- 10
3. Setup
3.1 Drawing the Internal Assembly Out ------------------------------------------------ 12
3.2 Switch Setting (Multi-function) ------------------------------------------------------- 12
3.3 Inserting the Internal Assembly ----------------------------------------------------- 15

4. Mounting to the Control Panel
4.1 Site Selection ---------------------------------------------------------------------------- 15
4.2 External Dimensions (Scale: mm) -------------------------------------------------- 15
4.3 Panel Cutout (Scale: mm) ------------------------------------------------------------ 16
4.4 Current Transformer (CT) Dimensions (Scale: mm) --------------------------- 16
4.5 Mounting ---------------------------------------------------------------------------------- 17

5. Wiring
5.1 Terminal Arrangement ----------------------------------------------------------------- 19
5.2 Wiring Examples ------------------------------------------------------------------------ 20
5.2.1 Current Transformer 1(CT1), 2(CT2) Input Wiring (W/W3 Option) --- 21

6. Settings
6.1 Operation Flowchart ------------------------------------------------------------------- 22
6.2 Settings ----------------------------------------------------------------------------------- 24
6.2.1 Main Setting Mode ----------------------------------------------------------------- 25
SV ----------------------------------------------------------------------------------------- 25
6.2.2 Sub Setting ModeSet value memory number ---------------------------------------------------------- 25
AT Perform/Cancel -------------------------------------------------------------------- 25
OUT1 proportional band ------------------------------------------------------------- 25
OUT2 proportional band ------------------------------------------------------------- 25
Integral time ----------------------------------------------------------------------------- 26
Derivative time ------------------------------------------------------------------------- 26
OUT1 proportional cycle ------------------------------------------------------------- 26
OUT2 proportional cycle ------------------------------------------------------------- 26
Manual reset ---------------------------------------------------------------------------- 26
A1 value ---------------------------------------------------------------------------------- 26
A2 value ---------------------------------------------------------------------------------- 26
Heater burnout alarm value --------------------------------------------------------- 27
Loop break alarm time ---------------------------------------------------------------- 27
Loop break alarm span --------------------------------------------------------------- 27

56.2.3 Auxiliary Function Setting Mode 1
Set value lock --------------------------------------------------------------------------- 28
SV high limit ----------------------------------------------------------------------------- 29
SV low limit ------------------------------------------------------------------------------ 29
Sensor correction ---------------------------------------------------------------------- 29
Overlap/Dead band ------------------------------------------------------------------- 29
Remote/Local setting ----------------------------------------------------------------- 29
Instrument number -------------------------------------------------------------------- 30
Communication speed --------------------------------------------------------------- 30
Communication protocol ------------------------------------------------------------- 30
6.2.4 Auxiliary Function Setting Mode 2
Scaling high limit value --------------------------------------------------------------- 30
Scaling low limit value ---------------------------------------------------------------- 30
Decimal point place ------------------------------------------------------------------- 31
PV filter time constant ---------------------------------------------------------------- 31
OUT1 high limit value ----------------------------------------------------------------- 31
OUT1 low limit value ------------------------------------------------------------------ 31
OUT1 ON/OFF hysteresis ----------------------------------------------------------- 31
OUT2 action mode -------------------------------------------------------------------- 31
OUT2 high limit value ----------------------------------------------------------------- 31
OUT2 low limit value ------------------------------------------------------------------ 31
OUT2 ON/OFF hysteresis ----------------------------------------------------------- 32
A1 action Energized/De-energized ------------------------------------------------ 32
A2 action Energized/De-energized ------------------------------------------------ 32
A1 hysteresis --------------------------------------------------------------------------- 32
A2 hysteresis --------------------------------------------------------------------------- 32
A1 action delay time ------------------------------------------------------------------ 32
A2 action delay time ------------------------------------------------------------------ 32
External setting input high limit ----------------------------------------------------- 33
External setting input low limit ------------------------------------------------------ 33
Transmission output ------------------------------------------------------------------- 33
Transmission output high limit ------------------------------------------------------ 33
Transmission output low limit ------------------------------------------------------- 33
Indication when control output OFF ----------------------------------------------- 33
SV rise rate ------------------------------------------------------------------------------ 33
SV fall rate ------------------------------------------------------------------------------- 34
Output status when input error ----------------------------------------------------- 34
6.2.5 Program Mode ----------------------------------------------------------------------- 34
Fixed value control/Program control switching --------------------------------- 35
Step 1 time ------------------------------------------------------------------------------- 35
Step 2 time ------------------------------------------------------------------------------- 35
Step 3 time ------------------------------------------------------------------------------- 35
Step 4 time ------------------------------------------------------------------------------- 35
Step 5 time ------------------------------------------------------------------------------- 35
Step 6 time ------------------------------------------------------------------------------- 36
Step 7 time ------------------------------------------------------------------------------- 366

6.2.6 Auto/Manual Control Switching -------------------------------------------------- 36
6.2.7 Control Output OFF Function ---------------------------------------------------- 36
6.2.8 MV, Remaining Step Time Indication ------------------------------------------- 36

7. Set Value Memory Function (SM Option) -------------------------------------- 37
8. Operation
8.1 When Using the FCR-13A as a Temperature Controller ---------------------- 38
8.2 When Using the FCR-13A as a Simplified Programmable Controller ----- 39

9. Action Explanations
9.1 OUT1 Standard Control Action ------------------------------------------------------ 40
9.2 Heater Burnout Alarm Action (Optional) ------------------------------------------ 40
9.3 OUT1 ON/OFF Control Action ------------------------------------------------------ 41
9.4 Pattern End Action --------------------------------------------------------------------- 41
9.5 Heating/Cooling Control Action (DR, DS, DA Option)
9.5.1 Heating/Cooling Control Action ------------------------------------------------- 42
9.5.2 When Setting Dead Band -------------------------------------------------------- 43
9.5.3 When Setting Overlap Band with Relay Contact Output ----------------- 44
9.6 Alarm 1 (A1), Alarm 2 (A2) Action -------------------------------------------------- 45

10. Control Actions
10.1 Fuzzy Self-tuning --------------------------------------------------------------------- 47
10.2 PID --------------------------------------------------------------------------------------- 47
10.3 AT of This Controller ----------------------------------------------------------------- 48

11. Specifications
11.1 Standard Specifications ------------------------------------------------------------- 49
11.2 Optional Specifications -------------------------------------------------------------- 54

12. Troubleshooting ---------------------------------------------------------------------- 57
13. Character Table ------------------------------------------------------------------------ 59

7

1. Model
1.1 Model
Alphanumeric characters to represent the control output (OUT1), input or options areentered where underlined.
[Example]

FCR - 13A - R / M,

A2
Alarm 2 (A2) output
Multi-range input
Relay contact output

Standard specifications
FCR–1 3 AControl action 3
Alarm 1 (A1)A
Control output(OUT1)
Input

/

R
S
A
M

PID control *1
Alarm types can be selected. *2
Relay contact
Non-contact voltage (for SSR drive)
Direct current
Multi-range *3*1: Fuzzy self-tuning PID, PID, PD, ON/OFF control can be selected by internal DIP switch.
*2: 12 types of alarm plus No alarm action can be selected by internal DIP and Rotaryswitches.
*3: An input type can be selected by DIP and Rotary switches from a choice of:Thermocouple (10 types), RTD (3 types), Direct current (2 types) and DC voltage(1 type).
Optional specifications
Code
Option name
A2
Alarm 2 (A2) output (including Pattern end 2 output)
DR
Relay contact
Heating/CoolingDS
Non-contact voltage
control output (OUT2)
DA
Direct current
TA
Direct current (4 to 20mA DC)
Transmission output
TV
DC voltage (0 to 1V DC)
C5
Serial
RS-485
communicationC
RS-232CSM
Set value memory external selection
EA
Direct current (0 to 20mA DC, 4 to 20mA DC)
External setting input
EV
DC voltage (0 to 1V DC, 1 to 5V DC)W
HB (Heater burnout
Single-phasealarm)
W3
Three-phases
LA
Loop break alarm
P24
Insulated power output
BL
Screw type mounting bracket
BK
Color: BlackIP
Drip-proof/Dust-proofTC
Terminal cover(For options in detail, see Section “11.2 Optional Specifications”.)
8Warning
Do not take the inner assembly out nor touch the terminal with the power
supply ON.
Touching the terminal with the power switched ON may result in severe
injury or death due to electric shock.

1.2 Rated Input
Input type
K
J
R
B
PLN
S
E
T
C(W/Re5-26)
Pt100
JPt1004 to 20mA DC
0 to 20mA DC
0 to 1V DC–200 to 1370
–200 to 1000
0 to 1760
0 to 1820
0 to 1390
0 to 1300
0 to 1760
0 to 1000–199.9 to 400.0
0 to 2315–199.9 to 850.0–200 to 850–199.9 to 500.0

Input range–320 to 2500
–320 to 1800
0 to 3200
0 to 3300
0 to 2500
0 to 2300
0 to 3200
0 to 1800–199.9 to 750.0
0 to 4200–199.9 to 999.9–320 to 1560
–199.9 to 900.0–1999 to 9999 *–1999 to 9999 *–1999 to 9999 *Resolution
1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
0.1 ( )1 ( )
0.1 ( )1 ( )
0.1 ( )
1
1
1*: For DC input, input range and decimal point place can be changed.
1.3 How to Read the Model Label
Model labels are attached to the case and the inner assembly.

Model label
Model
OptionInput
OutputPower supply, power consumption
Ambient temperature
Serial number

92 Name and Functions of Sections
(1)
(2)

(4)
(5)
(6)
(11)
(12)
(13)
(14)
(15)
(16)
(17)

(3)
(7)
(8)
(9)
(10)

(Fig. 2-1)
(1) PV display (Red)
Indicates the PV or setting characters in the setting mode.
(2) SV/MV/TIME display (Green)
Indicates the SV, MV, Time or set values in each setting mode.
(3) MEMO display (Yellow)
Indicates the Set value memory number.
(4) SV indicator (Green)
Lights while the SV is indicated on the SV/MV/TIME display.
(5) MV indicator (Red)Lights while the MV is indicated on the SV/MV/TIME display.
(6) TIME indicator (Yellow)
Lights while the Time is indicated on the SV/MV/TIME display.
(7) MAN indicator (Red)Lights during Manual control.
(8) OUT1 indicator (Green)
Lights when OUT1 (control output 1) is on.
(For Direct current output type, flashes corresponding to the MV in 125ms cycles.)10(9) OUT2 indicator (Yellow)
Lights when OUT2 (control output 2) is on.
(For Direct current output type, flashes corresponding to the MV in 125ms cycles.)(10) TX/RX indicator (Green)
Lights during Serial communication [TX (transmitting) output].
(11) A1 indicator (including Pattern end 1 output) (Red)Lights when the Alarm 1 (A1) output or Pattern end 1 output is on.
(12) A2 indicator (including Pattern end 2 output) (Red) (Option)
Lights when the Alarm 2 (A2) output or Pattern end 2 output is on.
(13) HB indicator (Red)Lights when the Heater burnout alarm output is on.
(14) SB indicator (Red)Lights when Sensor is burnt out.
(15) REM indicator (Red)Lights during Remote operation.
(16) LA indicator (Red)Lights when the Loop break alarm output is on.
(17) AT indicator (Yellow)
Flashes during AT (auto-tuning).
Keys
Increase key:

Increases the numeric value on the SV/MV/TIME display
in the setting mode.
Decrease key:Decreases the numeric value on the SV/MV/TIME display
in the setting mode.
Mode key:
Switches a setting mode and registers set values.
[To register each set (or selected) value, press this key.]
OUT/OFF key:
Performs the control output ON or OFF (Fixed value control).
Starts/Stops the program control (Program control).
Auto/Manual key: Switches either Automatic control or Manual control.
Fast key:Makes the numerical value change faster by pressing the
key with theor
key simultaneously.

11

3. Setup
3.1 Drawing the Internal Assembly Out
Before the power supply to this instrument is turned on, take the internal assembly out
from the case in the direction indicated by the arrow by pushing the latch (bottom of the
instrument) while holding the instrument by the top and bottom.

Latch

Draw out the FCR-13A toward you
by pushing the latch.

(Fig. 3.1-1)

3.2 Switch Setting (Multi-function)
Using a small flat blade screwdriver and tweezers, set the following with the DIP and
Rotary switches.
Sensor input, Alarm 1 (A1) type, Alarm 2 (A2) type, Control action,
Heating (reverse)/Cooling (direct) action, Alarm 1 (A1) and Alarm 2 (A2)standby actions, / , Program start Auto/Manual
Rotary switch (SW301) will be equipped only when A2 option is ordered.
Control action

Alarm 2 typeHeating/Cooling action

Alarm 1 type

Alarm 1 standby action

Sensor input

Alarm 2 standby action/
Sensor input
Program start Auto/Manual

Rotary switch

DIP switch(Fig. 3.2-1)
12

The following items can be selected by the DIP switch (SW304).
Default value: All switches OFF.
(Table 3.2-1)
DIP
SW304
Item
Selection
Switch status
No.
Fuzzy self-tuning PID control
No.1: OFF No.2: OFF1
PID control
No.1: ON
No.2: OFF
Control action
andPD control
No.1: OFF No.2: ON
2
ON/OFF control
No.1: ON
No.2: ON
Heating/Cooling
Heating (reverse) action
No.3: OFF3
actionCooling (direct) action
No.3: ON
Alarm 1 (A1)
Without standby action
No.4: OFF4
standby action
Standby action
No.4: ON
Alarm 2 (A2)
Without standby action
No.5: OFF5
standby action
No.5: ON
Standby action(*1)
No.6: OFF6/
No.6: ONK, J, R, B, N, PL- , Pt100, JPt100 No.7: OFF
Sensor input
7
S, E, T, C, 4 to 20mA, 0 to 20mA, No.7: ON(*2)
0 to 1V, Pt100
Program start
Manual start
No.8: OFF
8
Auto/Manual
No.8: ON
Automatic start(*3)(*1) The standby function will work only when the A2 option is ordered.
(*2) Use the Rotary switch (SW303) and DIP switch (SW304) together for making a
selection.
(*3) Program start Auto/Manual: For program control
Manual start:
The preset program starts by pressing the
key.
Automatic start: The preset program automatically starts from Step 1 after 2 seconds
of warm-up status after power-on.

13

Select a sensor type using the Rotary switch (SW303) and DIP switch (SW304).
Default value: K -200 to 1370
Note: If the input type is changed, Scaling high/low limit, External setting
input high/low limit and transmission output high/low limit values(optional) will become the altered input range high/low limit value.
(Table 3.2-2)
RotaryDIP
Range (DIP SW304 No.6)
Sensor type
SW303
SW304
OFF
ON
No.
No. 7
0
OFF
K
-200 to 1370
-320 to 2500
1
2
3
4
5
6
7

OFF
OFF
OFF
OFF
OFF
OFF
OFF

J
R
B
PLN
Pt100
JPt100-200 to 1000
0 to 1760
0 to 1820
0 to 1390
0 to 1300-199.9 to 850.0-199.9 to 500.0-320 to 1800
0 to 3200
0 to 3300
0 to 2500
0 to 2300-199.9 to 999.9-199.9 to 900.0

0
1
2
3
4
5
6
7

ON
ON
ON
ON
ON
ON
ON
ON

S
E
T
C (W/Re5-26)
4 to 20mA DC
0 to 20mA DC
0 to 1V DC
Pt100

0 to 1760
0 to 3200
0 to 1000
0 to 1800-199.9 to 400.0-199.9 to 750.0
0 to 2315
0 to 4200-1999 to 9999
-1999 to 9999
-1999 to 9999
-200 to 850-320 to 1560

The alarm type and the pattern end output for program control can be selectedby the Rotary switches (SW302) and (SW301).
Rotary switch (SW301) will be equipped only when A2 option is ordered.
Rotary SW302: Alarm 1 (A1) type, Pattern end 1 output
Rotary SW301: Alarm 2 (A2) type, Pattern end 2 output
Note: If an alarm type is changed, the alarm set value becomes 0 (0.0).
Default value: No alarm action(Table 3.2-3)
Alarm 1 (A1) typeNo alarm action
High limit alarm
Low limit alarm
High/Low limits alarm
High/Low limit range alarm
Process high alarm
Process low alarm
Pattern end 1 output

Rotary
SW302
No.
0
1
2
3
4
5
6
7
14

Alarm 2 (A2) typeNo alarm action
High limit alarm
Low limit alarm
High/Low limits alarm
High/Low limit range alarm
Process high alarm
Process low alarm
Pattern end 2 output

Rotary
SW301
No.
0
1
2
3
4
5
6
7

3.3 Inserting the Internal Assembly
After the setup is completed, insert the internal assembly into the case.
Firmly insert the assembly until it is locked by the latch at the bottom of the instrument.
(There will be a clicking sound.)Caution
Do not confuse the top and bottom of the internal assembly.
If inserting the assembly into the case by force in the wrong direction, the PCB may
be damaged.

4. Mounting to the Control Panel
4.1 Site Selection
Ensure the mounting location corresponds to the following conditions(IEC61010-1): Overvoltage category , Pollution degree 2(1) A minimum of dust, and an absence of corrosive gases(2) No flammable, explosive gases(3) No mechanical vibrations or shocks(4) No exposure to direct sunlight, an ambient temperature of 0 to 50 (32 to 122 )that does not change rapidly, and no icing(5) An ambient non-condensing humidity of 35 to 85%RH(6) No large capacity electromagnetic switches or cables through which large current
flows(7) No water, oil or chemicals or where the vapors of these substances can come into
direct contact with the unit
(8) Take note that ambient temperature of this unit - not the ambient temperature of thecontrol panel - must not exceed 50 (122 ) if mounted through the control panel.
Otherwise the life of electronic components (especially electrolytic capacitor) may be
shortened.

48

10

100
106 (for TC option)(Fig. 4.2-1)
15

109

106.2 (for BL option)

96

91

4.2 External Dimensions (Scale: mm)

13092+0.80

4.3 Panel Cutout (Scale: mm)

92

+0.8
0

n x 48-3 +0.5
0

+0.5

45 0

Lateral close mounting
n: Number of units mounted(Fig. 4.3-1)

4.4 Current Transformer (CT) Dimensions (Scale: mm)

CTL-6S (for 20A)

CTL-12-S36-10L1U (for 50A)
(Fig. 4.4-1)

16

4.5 Mounting

Warning
As the case is made of resin, do not use excessive force while screwing in themounting bracket, or the case could be damaged.
The torque should be 0.12N•m.
• When using the One-touch type mounting bracket
Mountable panel thickness: 1 to 3mm.
If Soft front cover (FC-R-S) is used, mounting
panel thickness will be 1 to 2.5mm.
Mount one-touch mounting bracket 1 to the
instrument first, and then insert the FCR-13A
2 from the front of the panel.

12(Fig. 4.5-1)

1• When using the Screw type mounting bracket (BL option)
Mountable panel thickness: 1 to 8mm.
If Soft front cover (FC-R-S) is used, mounting panel thickness will be 1 to 7.5mm.
Insert the FCR-13A from the front of the panel.
Attach the mounting brackets to the slots at the top and bottom of the case,and secure the controller in place with the screws provided.

(Fig. 4.5-2)
17

5. Wiring
Warning
Turn the power supply to the instrument off before wiring or checking.
Working on or touching the terminal with the power switched on may result insevere injury or death due to electric shock.
Moreover, the instrument must be grounded before the power supply to the
instrument is turned on.

Caution• Do not leave wire remnants in the instrument, as they could cause a fire or a malfunction.
• Use a solderless terminal with an insulation sleeve in which the M3 screw fits whenwiring the controller.
• The terminal block of this instrument is designed to be wired from the left side.
The lead wire must be inserted from the left side of the terminal, and fastened by theterminal screw.
• Tighten the terminal screw using the specified torque. If excessive force is applied tothe screw when tightening, the terminal screw may be damaged.
• Do not apply a commercial power source to the sensor which is connected to the input
terminal nor allow the power source to come into contact with the sensor.
• This controller does not have a built-in power switch, circuit breaker or fuse.
Therefore, it is necessary to install them in a circuit externally, near the controller.
(Recommended fuse: Time-lag fuse, rated voltage 250V AC, rated current 2A)• For a 24V AC/DC power source, do not confuse polarity when using direct current (DC).
• Use a thermocouple and compensating lead wire according to the sensor input
specifications of this controller.
• Use a 3-wire RTD according to the sensor input specifications of this controller.
• When using a relay contact output type, use a relay according to the capacity of theload to protect the built-in relay contact.
• When wiring, keep input wires (thermocouple, RTD, etc.) away from AC sources orload wires.
• Use a thick wire (1.25 to 2.0mm2) for grounding.
Lead Wire Solderless Terminal
Use a solderless terminal with an insulation sleeve in which an M3 screw fits as shown
below. The torque should be 0.63N•m.
Solderless
Manufacturer
Model
Torqueterminal
Nichifu Terminal Industries CO.,LTD.
TMEV1.25Y-3
Y typeJapan Solderless Terminal MFG CO.,LTD. VD1.25-B3A
0.63N•m
Nichifu Terminal Industries CO.,LTD.
TMEV1.25-3
Ring type
Japan Solderless Terminal MFG CO.,LTD. V1.25-3
3.2mm
5.8mm or less

5.8mm or less

3.2mm

(Fig. 5-1)
18

5.1 Terminal Arrangement

OUT1(S/M, A/M)

3 -

-

6

b0 13

CT1

RS-232C (RS-485)

CT2

SM

COM 1617
A+
TA or TV

9B-

A1/P.END1

B

10

OUT2(DS,DA)/A2/P.END2/LA/P24

8

b1 14
b2 15

L W or W3

OUT2(DR)/A2/ 7
P.END2/LA/P24
87RX
12 [YB( + )]-H

OUT1 C
5 + (R/M) 5
6

EA or EV

2 +

4

TX
11 [YA( - )]

+

RTD

18

17

+

+

18 mA DC 18 +
TC
V DC
19
19

・

100 to 240V AC
or 24V AC/DC

1・

Ground

19

-

-

20

+

(Fig. 5.1-1)

Dotted lines show options, and no terminal is equipped unless specified.
If Alarm 2 (A2 option/Pattern end 2) and Loop break alarm (LA option)
are ordered together, they will utilize common output terminals.
OUT1:
Control output 1
OUT2:
Control output 2
A1:
Alarm 1 output
A2:
Alarm 2 output
P.END1: Pattern end 1 output
P.END2: Pattern end 2 output
LA:
Loop break alarm output
P24:
Insulated power output
RS-232C (RS-485): Serial communicationSM:
Set value memory number external selection
TC:
Thermocouple input
RTD:
RTD inputDC:
DC voltage, current input
EA or EV: External setting input
CT1, CT2: CT1 input (W, W3 option), CT2 input (W3 option)
W or W3: Heater burnout alarm output
TA or TV: Transmission output
19-

5.2 Wiring Examples[FCR-13A-R/M]
3-phase100 to 240V AC

Ground*Electromagnetic
switchPower supply
for Alarm unit

Alarm unit

Heater

Thermocouple

Electric furnace* To prevent the unit from harmful effects of unexpected level noise, it is recommended
that a surge absorber be installed between the electromagnetic switch coils.
(Fig. 5.2-1)[FCR-13A-S/M, W

Power supply: 24V DC]

Single-phase

Alarm unit
24V
CT1
SSR
W

CT

Thermocouple
HeaterElectric furnace• The connectable SSRs in parallel are 2 units if the Shinko SSRs (SA-500 series) areused.
• For a 24V AC/DC power source, do not confuse polarity when using direct current (DC).
(Fig. 5.2-2)
20

5.2.1 Current Transformer 1 (CT1), 2 (CT2) Input Wiring (W/W3 Option)[Heater burnout alarm output](1) This alarm is not usable for detecting current under phase control.
(2) Use the current transformer (CT) provided, and pass one lead wire of heater
circuit into the hole of the CT.
(3) When wiring, keep CT wire away from AC sources and load wires to avoid theexternal interference.
[Single-phase heater]
CT1 input
connector
CT1
Power supply
Heater(Fig. 5.2-3)[Three-phase heater]
CT1 input connector

3-phase heater

CT2 input connector
CT1
R
S Power supply

CT2

T(Fig. 5.2-4)[Mounting the wire harness]

CT1

CT2
W or W3(Fig. 5.2-5)
21

6. Settings
6.1 Operation Flowchart
Power ON(*1)(*1)(approx. 1sec)

Control output OFF function (Fixed value control)
Manual control (Fixed value control)

PV/SV display mode(*1)
Standby mode (Program control)(approx. 1sec)(approx. 3sec)

MV indication

Remaining step time indication (*2)+[Main setting mode]
[
] SV

+

[Sub setting mode](appprox. 3sec)[] Set value memory number[Auxiliary function setting mode 1][] Set value lock[] AT Perform/Cancel (*3)[

] SV high limit[] OUT1 proportional band[

] SV low limit[] OUT2 proportional band[

] Sensor correction[] Integral time[

] Overlap/Dead band[

] Derivative time[

] Remote/Local[] OUT1 proportional cycle[] Instrument number[] OUT2 proportional cycle[] Communication speed[] Manual reset[] Communication protocol[] Alarm 1 (A1) value[] Alarm 2 (A2) value[ xx.x]Heater burnout alarm value[] Loop break alarm time[

] Loop break alarm span22

: Setting items with dotted lines are optional, and they appear only when the options are ordered.

: This means that if theis pressed, the unit proceeds to the next setting item.
+: Press the
key while holding down the
key.
+(approx.3sec): Press the
key for approx. 3 sec while holding down the
key.
++(approx.3sec): Press thefor approx.3 sec while holding down the,.
+(approx.3sec): Press thefor 3 sec while holding down thekey.
If theis pressed for 3 sec, the unit reverts to the PV/SV display mode from any mode.
(*1): Warm-up status lasts 2 sec after power-on, then starts from previous controller status (last shutdown).
(*2): Remaining step time is indicated only during program control.
(*3): If AT is performed, the unit reverts to the PV/SV display mode.
If AT is cancelled, the unit proceeds to the [OUT1 proportional band].
(*4): Selectin this mode and press the. The unit reverts to the PV/SV display mode.

++(Appprox. 3sec)+[Auxiliary function setting mode2][] Scaling high limit[] A1 action delay time[

] Scaling low limit[] A2 action delay time[

] Decimal point place[] External setting
input high limit[] PV filter time constant[] External setting
input low limit(Appprox. 3sec)[Program mode][] Program control(*4)[] Step 1 time[

] Step 2 time[

] Step 3 time[

] Step 4 time[

] OUT1 high limit[] OUT1 low limit[] Transmission output[] Step 5 time[

] OUT1 ON/OFF hysteresis[] Transmission
output high limit[] Step 6 time[

] OUT2 action mode[

] Step 7 time[

] Transmission
output low limit[] Indication whencontrol output OFF[] SV rise rate[

] SV fall rate[

] Output status when
input error[] OUT2 high limit[] OUT2 low limit[] OUT2 ON/OFF hysteresis[] A1 Energized/De-energized[] A2 Energized/De-energized[] A1 hysteresis[] A2 hysteresis236.2 Settings
The PV display indicates the sensor characters selected in [Sensor input], and the
SV/MV/TIME display indicates input range high limit (or Scaling high limit value for DC
voltage, current inputs) for approx. 2 seconds (warm-up status) after power-on.
See (Table 6.2-1).
During this time, all outputs and LED indicators are in OFF status.
Control will then start, indicating PV on the PV display and SV on the SV/MV/TIME display.
While control output OFF function is working, the PV display indicates the item selected in[Indication when control output OFF].
(Table 6.2-1)
Input

PV display

K
J
R
B
PLN
S
E
T
C(W/Re5-26)
Pt100
JPt100Pt100
4 to 20mA DC
0 to 20mA DC
0 to 1V DC

SV/MV/TIME
display

Scaling highlimit value

PV display

SV/MV/TIME
display

Scaling highlimit value

Notes before key operation• Pressing the
key for approx. 1 second from any mode enables the control output
OFF function.
Once the control output OFF function is enabled, the function cannot be released even
if the power to the instrument is turned OFF and ON again. To cancel the function, press
thekey again for approx. 1 second.
In the program control, the
key works as program Start/Stop key.
The control output OFF function does not work.
• If themode.

is pressed for 3 sec, the unit will revert to the PV/SV display mode from any

24

6.2.1 Main Setting ModeIn the PV/SV display mode, if the
key is pressed, the unit proceeds to the Main
setting mode.
The SV can be increased or decreased using the
,
and
keys.
If thekey is pressed, the SV will be registered, and the controller will revert to the
PV/SV display mode.
Character
Name, Function, Setting range
Default valueSV0• Sets SV (desired value).
• Setting range: SV low limit to SV high limit
6.2.2 Sub Setting ModeIn the PV/SV display mode, if the
key is pressed while holding down thekey, the unit moves to the Sub setting mode.
The set value can be increased or decreased using the
,
and
keys.
Pressing the
key registers the set value, and proceeds to the next setting item.
Pressing the
key at the last setting item registers the set value, and the unit
reverts to the PV/SV display mode.
Character
Name, Function, Setting range
Default value
Set value memory number
1• Selects the number (file) to be set or to be retrieved.
• Selection range: 1 to 7
AT Perform/Cancel
AT Cancel• Sets AT (Auto-tuning) Perform or Cancel.
If thekey is pressed after selecting AT Perform, the unit reverts to
the PV/SV display mode.
• Not available if ON/OFF or PD control is selected in [Control action].
• During AT, none of the settings can be performed.
• If the AT is cancelled during the process, P, I, D values revert to thevalues before AT was performed.
• Selection item:: AT Cancel: AT Perform
OUT1 proportional band
2.5%
• Sets OUT1 proportional band.
OUT1 proportional band depends on the scaling span.
(e.g.)
When OUT1 proportional band: 2.5%, -200 to 1370 :OUT1 proportional band converted value: (1370-(-200))x2.5/100=39.25
When scaling high limit value: 100 , scaling low limit value: 0
OUT1 proportional band converted value: (100-0)x2.5/100=2.5• Not available if ON/OFF control is selected in [Control action].
• Setting range: 0.1 to 999.9%
1.0 times
OUT2 proportional band
(2.5%)• Sets OUT2 proportional band.
• OUT2 becomes ON/OFF control when set to 0.0.
• Not available if Heating/Cooling control (DR, DS, DA option) is notordered, or if ON/OFF control is selected in [Control action].
• Setting range: 0.0 to 10.0 times (multiplied value of OUT1 proportional band)

25

Character

Name, Function, Setting range
Default value
Integral time
200 sec• Sets the integral time.
Setting the value to 0 disables the function (PD control).
• Not available if ON/OFF control or PD control is selected in [Control
action].
• Setting range: 0 to 3600 seconds
Derivative time
50 sec• Sets the derivative time.
Setting the value to 0 disables the function (PI control).
• Not available if ON/OFF control is selected in [Control action].
• Setting range: 0 to 3600 seconds
OUT1 proportional cycle
Relay contact output: 30 sec• Sets OUT1 proportional cycle.
Non-contact voltage output: 3 sec• Not available for Direct current output type or if ON/OFF control isselected in [Control action].
• For the relay contact output type, if the proportional cycle time isdecreased, the frequency of the relay action increases, and the lifeof the relay contact is shortened.
• Setting range: 1 to 120 seconds
OUT2 proportional cycle
Relay contact output: 30 sec• Sets OUT2 proportional cycle.
Non-contact voltage output: 3 sec• Not available for Direct current output type.
Not available if Heating/Cooling control (DR, DS, DA option) is not ordered,or if ON/OFF control is selected in [Control action].
• For the relay contact output type, if the proportional cycle time isdecreased, the frequency of the relay action increases, and the lifeof the relay contact is shortened.
• Setting range: 1 to 120 seconds
Manual reset
0.0• Sets the reset value to correct the offset (deviation between SV and PV
in equilibrium status).
• Available only when PD control is selected in [Control action].
• Setting range: Proportional band converted value
Thermocouple, RTD inputs: -199.9 to 999.9
DC voltage input: -1999 to 9999 (The placement of the decimal point
follows the selection.)
How to set manual reset:
When SV>PV, set the positive (+) value (SV–PV).
When SV<PV, set the negative (–) value (SV–PV).
A1 value0• Sets the action point of Alarm 1 (A1) output.
• Setting the value to 0 or 0.0 disables the function (except Process
high and Process low alarm).
• Not available if the Rotary switch is set to No.0 or No.7
• Setting range: Refer to (Table 6.2.2-1).
0
A2 value• Sets the action point of Alarm 2 (A2) output.
• Setting the value to 0 or 0.0 disables the function (except Process
high and Process low alarm).
• Not available if A2 output (option) is not ordered, or if Rotary switch is
set to No.0 or No.7 in [A2 type] even if it is ordered.
• Setting range: Refer to (Table 6.2.2-1).
26

Characterxx.x(xx.x:
Heatercurrent
value)

Name, Function, Setting range
Default value
Heater burnout alarm value
0.0A• Sets the heater current value for Heater burnout alarm.
• Setting the value to 0.0 disables the function.
• Available only when Heater burnout alarm (W, W3 option) is ordered.
• It is recommended to set approx. 80% of the heater current value (setvalue) considering the voltage fluctuation.
• Upon returning to set limits, the alarm will stop.
• Setting range:
Rated current 20A: 0.0 to 20.0A (Indication: 0.0 to 50.0)
Rated current 50A: 0.0 to 50.0A
0 minutes
Loop break alarm time
• Sets the time to assess the Loop break alarm.
• Available only when Loop break alarm (LA option) is ordered• Setting range: 0 to 200 minutes
Loop break alarm span0
• Sets the span to assess the Loop break alarm.
• Available only when Loop break alarm (LA option) is ordered• Setting range:
Thermocouple, RTD without decimal point: 0 to 150
Thermocouple, RTD with decimal point: 0.0 to 150.0
DC input: 0 to 1500 (The placement of the decimal point follows theselection.)[A1, A2 setting range]
Alarms with the standby function have the same setting range.
(Table 6.2.2-1)
Alarm type
Setting range
High limit alarm–Input span to Input span ( )Low limit alarm–Input span to Input span ( )High/Low limits alarm
0 to Input span ( )High/Low limit range alarm
0 to Input span ( )Process high alarm
Input range low limit to Input range high limit
Process low alarm
Input range low limit to Input range high limit

*1
*1
*1
*1
*2
*2

• For RTD input, the negative low limit value is –199.9.
• For DC input, the negative low limit value is –1999. (The placement of the decimal
point follows the selection.)*1: For DC input, the Input span is the same as the Scaling span.
*2: For DC input, Input range low (or high) limit value is the same as the Scaling low(or high) limit value.
[Loop break alarm]
Detects the breaking status on the loop such as heater burnout, sensor burnout or
actuator trouble.
When the control action is Heating (Reverse) control:• The alarm will be activated when the PV does not rise as much as the alarm span ormore within the loop break alarm time after the MV has reached 100% or the output
high limit value.
• The alarm will also be activated when the PV does not fall as much value as the spanor more within the loop break alarm time after the MV has reached 0% or the output
low limit value.
• When the control action is Direct (Cooling), read “fall” for “rise” and vice versa.
27

6.2.3 Auxiliary Function Setting Mode 1
In the PV/SV display mode, if the
key is pressed for approx. 3 seconds while
holding down thekey, the unit proceeds to Auxiliary function setting mode 1.
The set value can be increased or decreased by pressing the
,
and
keys.
Pressing the
key registers the set value and proceeds to the next setting item.
If thekey is pressed at the last setting item, the set value will be registered andthe unit will revert to the PV/SV display mode.
Character
Name, Function, Setting range
Default value
Set value lock
Unlock• Locks the set value to prevent setting errors.
The setting item to be locked differs depending on the selection.
• When selecting Lock, set the necessary items in the status Unlock,then select Lock 1, Lock 2 or Lock 3.
• Fuzzy self-tuning or AT will not function if [] or [] isselected.
• Be sure to select Lock 3 when changing the set value frequently via
communication function considering the life of non-volatile memory.
• Selection item:(Unlock): All set values can be changed.
(Lock 1): None of set values can be changed.
(Lock 2): Only SV can be changed.
(Lock 3): All set values can be changed temporarily. However,changed values revert to their previous value after
power-off because they are not saved in the
non-volatile memory.
Since this function has no relation to the memory life,
it is well suited when using with Shinko programmablecontrollers (with SVTC option).
[About Lock 3]
When using the FCR -13A as a Fixed value controller
The data in the selected Set value memory number can be temporarilychanged.
However, if the memory number is changed, the data in the previous
memory number is cancelled, and returns to the previous values.
When using the FCR -13A as a Programmable controller
The data in the currently performing step number can be temporarilychanged.
However, if the step number is changed, the data is cancelled, andreturns to the previous values.
During program standby status, the data changes will be invalidated.

28

Character

Name, Function, Setting range
SV high limit

Default value
400• Sets SV high limit value within the Scaling low limit value/Scaling highlimit value range.
• Setting range: SV low limit to Scaling high limit value(For DC input, the placement of the decimal point follows
the selection.)
SV low limit0• Sets SV low limit value within the Scaling low limit value/Scaling highlimit value range.
• Setting range: Scaling low limit value to SV high limit(For DC input, the placement of the decimal point follows
the selection.)
Sensor correction
0.0• Sets the sensor correction value.
• Setting range: –100.0 to 100.0 ( )[Sensor correction function]
This corrects the input value from the sensor. When a sensor cannot be
set at the exact location where control is desired, temperatures
measured by the sensor may deviate from the temperature in thecontrolled location. When controlling with plural controllers, sometimes
the measured temperatures (input value) do not concur due todifferences in sensor accuracy or dispersion of load capacities.
In such a case, the control can be set at the desired temperature byadjusting the input value of sensors. However, it is effective within the
input rated range regardless of the sensor correction value.
PV after sensor correction= Current PV+ (Sensor correction value)
Overlap/Dead band
0.0
• Sets the Overlap or Dead band for OUT1 and OUT2.
• + Set value: Dead band
– Set value: Overlap band
• Not available if Heating/Cooling control (DR, DS, DA option) is notordered, or if ON/OFF control is selected in [Control action].
• Setting range: OUT1 proportional band converted value
Thermocouple, RTD : -199.9 to 999.9 ( )DC voltage, current : -1999 to 9999 (The placement of
the decimal point follows the selection.)
Local setting
Remote/Local setting• SV can be set with either Remote or Local method.
• Available only when External setting input (EA, EV option) is ordered.
• Selection item:: Local setting. The SV can be set by the front keypad.
: Remote setting. The SV can be set by external remote operation.

29

Character

Name, Function, Setting range
Default value
Instrument number0• Sets the instrument number of this unit. (The instrument numbers should
be set one by one when multiple instruments are connected in series
in Serial communication, otherwise communication is impossible.)• Available only when the Serial communication (C, C5 option) is ordered• Setting range: 0 to 95
Communication speed
9600bps• Selects the communication speed of this instrument.
(The communication speed of this instrument must be equal to that ofhost computer, otherwise communication is impossible.)• Available only when the Serial communication (C, C5 option) is ordered• Selection item:: 2400bps: 4800bps: 9600bps: 19200bps
ShinkoCommunication protocol
protocol• Selects the communication protocol of this instrument.
• Available only when the Serial communication (C, C5 option) is ordered• Selection item:: Shinko protocol: Modbus ASCII mode

6.2.4 Auxiliary Function Setting Mode 2
In the PV/SV display mode, if the
key is pressed for approx. 3 seconds while
holding down theandkeys, the unit proceeds to Auxiliary function setting
mode 2.
The set value can be increased or decreased using the
,
and
keys.
Pressing the
key registers the set value, and proceeds to the next setting item.
If thekey is pressed at the last setting item, the set value is registered and theunit reverts to the PV/SV display mode.
Character

Name, Function, Setting range
Default value
1370
Scaling high limit value• Sets scaling high limit value.
If scaling high limit value is changed, SV high limit automaticallychanges to the scaling high limit value.
• Setting range: Scaling low limit to Input range high limit value
Scaling low limit value–200• Sets scaling low limit value.
If scaling low limit value is changed, SV low limit automatically changes
to the scaling low limit value.
• Setting range: Input range low limit to scaling high limit value

30

Character

Name, Function, Setting range
Default value
No decimal
Decimal point placepoint• Selects the decimal point place.
• Not available if RTD or thermocouple is selected in [Sensor input].
• Selection item:: No decimal point: 1 digit after the decimal point: 2 digits after the decimal point: 3 digits after the decimal point
PV filter time constant
0.0 sec• Sets PV filter time constant.
However, if the set value is set too large, it affects control result due tothe delay of response.
• Setting range: 0.0 to 10.0 sec
OUT1 high limit value
100%• Sets the high limit value for OUT1.
• Not available if ON/OFF control is selected in [Control action]• Setting range: OUT1 low limit value to 100%
(Direct current output: OUT1 low limit value to 105%)
OUT1 low limit value0%• Sets low limit value for OUT1.
• Not available if ON/OFF control is selected in [Control action]• Setting range: 0% to OUT1 high limit value(Direct current output: –5% to OUT1 high limit value)
OUT1 ON/OFF hysteresis
1.0• Sets ON/OFF hysteresis for OUT1.
• Available only when ON/OFF control is selected in [Control action]• Setting range: 0.1 to 100.0 ( )OUT2 action mode
Air cooling• Selects OUT2 cooling action from a choice of: Air cooling, oil coolingand water cooling.
• Available only when Heating/Cooling control (DR, DS, DA option) is ordered• Selection item:: Air cooling (linear characteristic): Oil cooling (1.5th power of the linear characteristic): Water cooling (2nd power of the linear characteristic)
OUT2 high limit value
100%• Sets the high limit value for OUT2.
• Not available if Heating/Cooling control (DR, DS, DA option) is notordered, or if OUT2 is ON/OFF control action• Setting range: OUT2 low limit value to 100%
(Direct current output: OUT2 low limit value to 105%)
OUT2 low limit value0%• Sets the low limit value for OUT2.
• Not available if Heating/Cooling control (DR, DS, DA option) is notordered, or if OUT2 is ON/OFF control action• Setting range: 0% to OUT2 high limit value(Direct current output: –5% to OUT2 high limit value)

31

Character

Name, Function, Setting range
OUT2 ON/OFF hysteresis

Default value
1.0• Sets ON/OFF hysteresis for OUT2.
• Not available if Heating/Cooling control (DR, DS, DA option) is notordered, or if OUT2 is PID or PD control action.
• Setting range: 0.1 to 100.0 ( )A1 action Energized/De-energized
Energized• Selects A1 action Energized/De-energized.
• Not available if Rotary switch is set to No.0 or No.7 in [A1 type].
• Selection item:: Energized,: De-energizedA2 action Energized/De-energized
Energized• Selects A2 action Energized/De-energized.
• Not available if A2 option is not ordered, or if Rotary switch is set to No.0
or No.7 in [A2 type].
• Selection item:: Energized,: De-energizedA1 hysteresis
1.0• Sets A1 hysteresis.
• Not available if Rotary switch is set to No.0 or No.7 in [A1 type].
• Setting range: Thermocouple, RTD input: 0.1 to 100.0 ( )DC input: 1 to 1000 (The placement of the decimal point
follows the selection.)A2 hysteresis
1.0• Sets A2 hysteresis.
• Not available if A2 option is not ordered.
Not available if Rotary switch is set to No.0 or No.7 in [A2 type].
• Setting range: Thermocouple, RTD input: 0.1 to 100.0 ( )DC input: 1 to 1000 (The placement of the decimal point
follows the selection.)A1 action delay time
0 sec
• Sets A1 action delay time.
When setting time has elapsed after the input enters the alarm output
range, the alarm is activated.
• Not available if the Rotary switch is set to No.0 or No.7 in [A1 type].
• Setting range: 0 to 9999 seconds
A2 action delay time
0 sec
• Sets A2 action delay time.
When setting time has elapsed after the input enters the alarm output
range, the alarm is activated.
• Not available if A2 option is not ordered, or if the Rotary switch is set toNo.0 or No.7 in [A2 type] even if A2 option is ordered.
• Setting range: 0 to 9999 seconds

32

Character

Name, Function, Setting range
External setting input high limit

Default value
400• Sets External setting input high limit value.
For EA option (4 to 20mA), the value corresponds to 20mA input.
• Available only when External setting (EA, EV option) is ordered.
• Setting range: External setting input low limit value to
Input range high limit value
External setting input low limit0• Sets External setting input low limit value.
For EA option (4 to 20mA), the value corresponds to 4mA input.
• Available only when External setting (EA, EV option) is ordered.
• Setting range: Input range low limit value toExternal setting input high limit value
Transmission output
PV• Selects a Transmission output type.
transmission• Available only when Transmission output (TA, TV option) is ordered• Selection item:: PV transmission: SV transmission: MV transmission
Transmission output high limit
400• Sets the Transmission output high limit value.
For TA option, the value correponds to 20mA output.
• Available only when Transmission output (TA, TV option) is ordered• Setting range: Transmission output low limit value to Input range highlimit value
Transmission output low limit0• Sets the Transmission output low limit value.
For TA option, the value correponds to 4mA output.
• Available only when Transmission output (TA, TV option) is ordered• Setting range: Input range low limit value to Transmission output highlimit value
OFF isIndication when control output OFF
indicated• Selects the indication when the control output is off.
on the PV• Selection item:
display.
: OFF is indicated on the PV display.
: No indication: Only PV is indicated.
0 /minuteSV rise rate
• Sets the SV rise rate (Rising value per minute).
Setting the value to 0 or 0.0 disables the function.
• Setting range: 0 to 9999 /min.
With a decimal point: 0.0 to 999.9 /min.
DC input: 0 to 9999 (The placement of the decimal
point follows the selection.)
33

Character

Name, Function, Setting range

Default value
0 /minuteSV fall rate

• Sets the SV fall rate (Falling value per minute).
Setting the value to 0 or 0.0 disables the function.
• Setting range: 0 to 9999 /min.
With a decimal point: 0.0 to 999.9 /min.
DC input: 0 to 9999 (The placement of the decimal point
follows the selection.)
Output status when input error
Outputs• Selects control output (OUT1/OUT2) status when DC
OFF(4mA) or
input is in overscale or underscale.
• Available only for Direct current output type with DC input OUT1(OUT2)
low limit value.
• Selection item:: Outputs OFF (4mA) or OUT1 (OUT2) low limit value.
: Outputs a value between OFF (4mA) and ON (20mA) or
between OUT1 (OUT2) low limit value and OUT1 (OUT2) highlimit value, depending on a deviation.
Alarm 1 (A1), 2 (A2) Energized/De-energized function[If the alarm action Energized is selected]
When the alarm output indicator is lit, the alarm output (terminals 7-8, 9-10) is
conducted (ON). When the alarm output indicator is unlit, the alarm output is notconducted (OFF).
[If the alarm action De-energized is selected]
When the alarm output indicator is lit, the alarm output (terminals 7-8, 9-10) is notconducted (OFF). When the alarm output indicator is unlit, the alarm output isconducted (ON).
High limit alarm (when Energized is set) High limit alarm (when De-energized is set)
A1 hysteresis

A1 hysteresis
ON

ON
OFF

OFF
SV

+ A1 value(Fig. 6.2.4-1)

SV+ A1 value

A1: Alarm 1.
For Alarm 2(A2),read “A2” for “A1”.

6.2.5 Program Mode
In PV/SV display mode, if the
key is pressed for approximately 3 seconds while
holding down thekey, the units moves to the Program mode.
The
,
and
keys increase or decrease the set values (numeric value).
Pressing the
key registers the set value, and proceeds to the next setting item.
If thekey is pressed at the last setting item, the set value is registered and the unit
reverts to the PV/SV display mode.
• Data in Set value memory numbers (1 to 7, set with Fixed value control method) is
assigned to Steps 1 to 7 respectively.
For example, data in Set value memory number 1 becomes Step 1 data, and data in Set
value memory number 2 becomes Step 2 data, and so on.
• If the Pattern end output is selected and the program control is performed, the Patternend output is turned on when the program is completed.
• If thekey is pressed while the Pattern end output is on, the Pattern end output isturned off. If the
key is pressed again, the program will be performed.
34• The following shows the program pattern example.
Set the step time to 00.00 for the unnecessary steps.
[Program Example]
Step number
1
2
3
4
5
6
7
(Set value memory number)
Step SV500
500 1000 1000 200
200
100
OUT1 proportional band
2.0
1.8
2.0
1.8
2.5
1.8
2.0
Integral time
180
80
180
80
200
80
200
Derivative time
30
20
30
20
50
20
50
OUT1 proportional cycle
30
30
30
30
30
30
30
A1 value
2
5
2
5
10
0
5
A2 value
2
5
2
5
10
0
5
Step time
00:30 01:00 00:40 01:00 02:00 00:30 01:00
1000

500

0
Character

Name, Function, Setting range
Default value
Fixed value control/Program control switching
Fixed value• Fixed value control or program control can be selected.
control• If thekey is pressed after selecting the fixed value control, thecontroller will revert to the PV/SV display mode.
If thekey is pressed after selecting the program control, steptime from Step 1 to Step 7 can be set.
• Selection item:: Fixed value control: Program control
Step 1 time
00.00
• Sets Step 1 time. (Available only for program control)(e.g.) When setting 1 hour 58 minutes, set as [].
• Setting range: 00.00 to 99.59
Step 2 time
00.00
• Sets Step 2 time. (Available only for program control)• Setting range: 00.00 to 99.59
Step 3 time
00.00
• Sets Step 3 time. (Available only for program control)• Setting range: 00.00 to 99.59
Step 4 time
00.00
• Sets Step 4 time. (Available only for program control)• Setting range: 00.00 to 99.59
Step 5 time
00.00
• Sets Step 5 time. (Available only for program control)• Setting range: 00.00 to 99.59
35

Step 6 time

00.00

• Sets Step 6 time. (Available only for program control)• Setting range: 00.00 to 99.59
Step 7 time
• Sets Step 7 time. (Available only for program control)• Setting range: 00.00 to 99.59

00.00

6.2.6 Auto/Manual Control Switching
MV can be changed manually.
As a proportional action, the MV is automatically outputted in proportion to thedeviation. This is called an automoatic control.
In comparison with the above, in the manual control, the value set by front keypadcan be outputted regardless of the deviation.
If control action is switched from automatic to manual and vice versa, balancelessbumpless function works to prevent sudden change of MV.
Each time thekey is pressed, the control switches from Automatic to Manualand vice versa.
• Setting range: OUT1 low limit value to OUT1 high limit value
When Heating/Cooling control (DR, DS, DA option) is ordered: OUT2
low limit value to OUT1 high limit value
6.2.7 Control Output OFF Function
Control output OFF function []• The control action and output of an instrument (or any instruments in series) can beturned off without turning off the power supplies using this function.
• Pressing the
key for approx. 1 second from any mode enables the control
output OFF function. The PV display indicates the setting item selected in [Indication
when control output OFF].
To cancel the function, press the
key again for approx. 1 second.
• Once the control output OFF function is enabled, the function cannot be released
even if the power to the instrument is turned OFF and ON again.
To cancel the function, press the
key again for approx. 1 second.
• During program control, thekey becomes the Program Start/Stop key, andthe control output OFF function is disabled.
6.2.8 MV, Remaining Step Time Indication
MV indication
In the PV/SV display mode, press the
key for approx. 3 seconds.
Keep pressing the
key until the MV appears, though the SV setting mode
appears during the process.
The MV indicator lights. The SV/MV/TIME display indicates the output MV. The 2nd
decimal point from the right on the SV/MV/TIME display flashes.
For the fixed value control, when the
key is pressed again, the unit reverts
to the PV/SV display mode.
Remaining step time indication
For the program control, if the
key is pressed while in MV indication,
remaining step time will be indicated on the SV/MV/TIME display. The TIME indicatorlights. By pressing the
key again, the unit reverts to the PV/SV display mode.
36

7. Set Value Memory Function (SM Option)
If the SM option (Set value memory number external selection) is ordered, a maximum of
7 files (12 pieces of data per file) of data can be memorized.
Control can be performed by selecting the desired file.
One file comprises 12 kinds of data:
SV, OUT1 proportional band, Integral time, Derivative time, OUT2 proportional band,A1 value, A2 value, Overlap/Dead band, OUT1 high limit value, OUT1 low limit value,
OUT2 high limit value, OUT2 low limit value
To select the Set value memory number (file number), connect terminals 13 to 16 asshown below (Table 7-1).
Up to 50 units of the FCR -13A can be connected in parallel.
Terminal connection for Set value memory number selection(Table 7-1)
Set value
memory No.
1
2
3
4
5
6
7
Connecting
terminals
13 - 16 (b0-COM)
Closed Open Closed Open Closed Open Closed
14 - 16 (b1-COM)
Open Closed Closed Open
Open Closed Closed
15 - 16 (b2-COM)
Open
Open
Open Closed Closed Closed Closed• If the Set value memory number is selected externally, the number cannot be selectedby front keypad.
• To select the memory number by front keypad, all terminals (b0, b1, b2 and COM)
should be open.
• The Set value memory number cannot be changed during setting mode or AT.
Operation procedures for Set value memory function(1) In the PV/SV display mode, select a Set value memory number by connecting
terminals.
(2) In the setting mode, set each value:
SV, OUT1 proportional band, Integral time, Derivative time, OUT2 proportionalband, A1 value, A2 value, Overlap/Dead band, OUT1 high limit value,
OUT1 low limit value, OUT2 high limit value, OUT2 low limit value.
(3) After settings are complete, press the
key to revert to the PV/SV display mode.
Registering Data
• Data is registered in the file number indicated by the MEMO display.
• When any number is retrieved by connecting terminals, the selected number is indicated,
and the control is performed using data (set values) of the indicated file number.
• To change set values, repeat the Operation procedures above.

37

8 Operation
After the controller is mounted to the control panel and wiring is completed, operate theunit following the procedures below.
8.1 When Using the FCR-13A as a Temperature Controller(1) Turn the power supply to the FCR-13A ON.
For approx. 2 seconds after power-on, the sensor characters and the temperature unit
selected in [Sensor input] will be indicated on the PV display, and input range high limit
value (or Scaling high limit value for DC input) will be indicated on the SV/MV/TIME
display. See (Table 8.1-1).
During this time, all outputs and LED indicators are in OFF status.
After that, control starts, indicating PV on the PV display and SV on the SV/MV/TIME
display.
When the Control output OFF function is working, the item selected in [Indication whencontrol output OFF] mode is indicated on the PV display.
(Table 8.1-1)
Input
K
J
R
B
PLN
S
E
T
C(W/Re5-26)
Pt100
JPt100Pt100
4 to 20mA DC
0 to 20mA DC
0 to 1V DC

PV display

SV/MV/TIME
display

PV display

Scaling highlimit value

SV/MV/TIME
display

Scaling highlimit value(2) Input each set value.
Refer to Section “6. Settings”.
When controlling by Fuzzy self-tuning PID, select ”Perform” in [AT Perform/Cancel(p.25)] mode to start control in optimal conditions.
(3) Turn the load circuit power ON.
Control action starts so as to keep the control target at the SV.

38

8.2 When Using the FCR-13A as a Simplified Programmable Controller(1) Turn the power supply to the FCR-13A ON.
For approx. 2 seconds after power-on, the sensor type and the temperature unit
selected in [Sensor input] are indicated on the PV display, and input range high limit
value (or scaling high limit value for DC input) is indicated on the SV/MV/TIME display.
See (Table 8.1-1).
During this time, all outputs and LED indicators are in OFF status.
After that, the PV display indicates the PV, and the unit enters standby mode.
(2) Input each set value and step time
Refer to Section “6. Settings”.
(3) Turn the load circuit power ON.
(4) Program control start
If “Automatic start” is selected in [Program start Auto/Manual], the unit will switch towarm-up status for approx. 2 seconds after power-on. Program control automaticallystarts from Step 1.
If “Manual start” is selected in [Program start Auto/Manual], the unit will switch towarm-up status for approx. 2 seconds after power-on. The unit proceeds to thestandby status. In this status, if the
key is pressed, the program control starts
from Step 1.
For the program control start, SV start (0 ) is used.
During program control (RUN), the Step number (Set value memory number) change
is not effective.
To make the step time progress faster
The step time progress becomes 60 times faster than usual when thepressed.

key isTo terminate the program control during the process
The program control will be terminated if the
key is pressed for approx. 1 sec or
longer.
To switch the indication of MV and Remaining step time
In the PV/SV display, if the
key is pressed for approximately 3 seconds, the MV
is indicated. If the
key is pressed again, the remaining step time will be
indicated.
Instrument status when power is restored
After restoration following a power failure during program control, the FCR-13A
resumes program performance from where it stopped.
(The PV display flashes until the step at which the power failure occurred is finished.)39

9 Action Explanations
9.1 OUT1 Standard Control Action
Heating (reverse) action

Cooling (direct) action

Proportional band

Proportional band

ON

ON

Control action

OFF

OFF
SV

R/

SV

H 4

H 4

H 4

H 4

H 4

H 4

C 5

C 5

C 5

C 5

C 5

C 5

L 6

L 6

L 6

L 6

L 6

L 6

Cycle action is performed
according to deviation.

+ 5

+ 5

12V DCS/+ 5

12/0V DC

- 6

Cycle action is performed
according to deviation.

6

+ 5

+ 5

+ 5

20mA DC

20 to 4mA DC

6

0/12V DC

12V DC

66

6

6

Cycle action is performed
according to deviation.

+ 5

6

+ 5

0V DC

Cycle action is performed
according to deviation.

A/

+ 5

0V DC

+ 5

4mA DC

4mA DC

6

6

Changes continuously
according to deviation.

+ 5

+ 5

4 to 20mA DC

6

20mA DC

6

Changes continuously
according to deviation.

Indicator(OUT1) Green

Lit

Unlit

: Turns ON (lit) or OFF (unlit).

9.2 Heater Burnout Alarm Action (Optional)
ON
Heater burnout OFF
alarm action
Heater burnout alarm value
Small

Load current

Large

Heater burnout
alarm output
Indicator (HB)
Red

Lit

Unlit

40

Unlit

Lit

9.3 OUT1 ON/OFF Control Action
Heating (reverse) action

Cooling (direct) action

Hysteresis

Hysteresis

ON

ON

Control action

OFF

OFF
SV

R/

SV

H 4

H 4

H 4

H 4

C 5

C 5

C 5

C 5

L 6

L 6

L 6

L 6

+ 5

+ 5

+ 5

+ 5

12V DC

0V DC

6

6

6

6

+ 5

+ 5

+ 5

+ 5

S/

A/

0V DC

12V DC

20mA DC

4mA DC

4mA DC

20mA DC

6

6

6

6

Indicator(OUT1) Green

Lit

Unlit

Unlit

Lit

: Turns ON (lit) or OFF (unlit).

9.4 Pattern End Action
ON
Pattern end
actionOFF
Pattern end output isreleased by pressing

Time
Program control end
9

9

the
9

Pattern end
output10

10

10

Indicator (A1)
Red

Unlit

Lit

key.

OUTOFF

Unlit

Pattern end output is turned ON when the program control is completed,
and it can be released when the

key is pressed.

Use terminals 7 and 8 for Pattern end 2 output.
The A2 indicator lights when Pattern end 2 output is ON.

41

9.5 Heating/Cooling Control Action (DR, DS, DA Option)
9.5.1 Heating/Cooling Control Action
OUT1 P-band(OUT2 P-band)

ON

ON
HeatingactionControlaction(Cooling
action)

OFF

OFF
SV

R/

H 4

H 4

H 4

C 5

C 5

C 5

L 6

L 6

L 6

Cycle action is performed
according to deviation.

7
DR

8

7

7

8

8

Cycle action is performed
according to deviation.

+ 5
S/

+ 5

+ 5

12V DC

12/0V DC

6

6

0V DC

6

Cycle action is performed
according to deviation.

+ 7

+ 7
0V DC

DS

+ 7

0/12V DC

8

12V DC

8

8

Cycle action is performed
according to deviation.

+ 5
20mA DC

A/

6

+ 5

+ 5

20 to 4mA DC

4mA DC

6

6

Changes continuously
according to deviation.

+ 7

+ 7

4mA DCDA+ 7

4 to 20mA DC

8820mA DC

8

Changes continuously
according to deviation.

Indicator(OUT1) Green
Indicator(OUT2) Yellow

Lit

Unlit

Unlit

Lit

: Turns ON (lit) or OFF (unlit).
: Represents Heating control action.
: Represents Cooling control action.

42

9.5.2 When Setting Dead BandOUT1 P-band

Dead band(OUT2 P-band)

ON

ON

ControlactionHeatingaction(Cooling
action)

OFF

OFF
SV

R/

H 4

H 4

H 4

C 5

C 5

C 5

L 6

L 6

L 6

Cycle action is performed
according to deviation.

7
DR

8

7

7

8

8

Cycle action is performed
according to deviation.

+ 5
S/

+ 5

+ 5

12V DC

12/0V DC

6

6

0V DC

6

Cycle action is performed
according to deviation.

+ 7

+ 7
0V DC

DS

+ 7

0/12V DC

8

12V DC

8

8

Cycle action is performed
according to deviation.

+ 5

+ 5

20mA DC

A/

6

+ 5

20 to 4mA DC

64mA DC

6

Changes continuously
according to deviation.

+ 7

+ 7
4mA DCDA

8

4 to 20mA DC

8+ 7
20mA DC

8

Changes continuously
according to deviation.

Indicator(OUT1) Green

Lit

Unlit

Unlit

Lit

Indicator(OUT2) Yellow: Turns ON (lit) or OFF (unlit).
: Represents Heating control action.
: Represents Cooling control action.

43

9.5.3 When Setting Overlap Band with Relay Contact Output.
OUT1 P-band
OUT2 P-band
Overlap
band

ON
ControlactionON

Heatingaction(Cooling
action)

OFF

OFF
SV

R/

H 4

H 4

H 4

C 5

C 5

C 5

L 6

L 6

L 6

Cycle action is performed
according to deviation.

DR

7

7

7

8

8

8

Cycle action is performed
according to deviation.
Indicator(OUT1) Green
Indicator(OUT2) Yellow

Lit

Unlit

Unlit

Lit

: Turns ON (lit) or OFF (unlit).
: Represents Heating control action.
: Represents Cooling control action.

44

9.6 Alarm 1 (A1), Alarm 2 (A2) Action
High limit alarm

Low limit alarm
A1
hysteresis

ONA1hysteresis
ON

Alarm action
OFF

OFF

-A1 value

Alarm output

SV+A1 value-A1 value

+side

+side

-side

-side

SV

+A1 value

High/Low limit range alarm

High/Low limits alarm
A1
hysteresis

A1
hysteresis
ON

ON

Alarm action
OFF

OFF

SV

A1 valueA1 valueA1 valueSV

A1 value

Alarm output
Process high alarm

Process low alarm
A1
hysteresis

A1
hysteresis
ON

ON

Alarm action
OFF

OFF

A1 valueA1 value
Alarm output
High limit alarm with standby

Low limit alarm with standby
A1hysteresis

A1
hysteresis
ON

ON

Alarm action
OFF

OFF

-A1 value
Alarm output

SV+A1 value-A1 value

+side

+side

-side

-side

45

SV

+A1 value

High/Low limits alarm with standby

High/Low limit range alarm with standby

A1hysteresis

A1
hysteresis
ON

ON

Alarm action
OFF

OFF

A1 valueSV

A1 valueA1 valueSV

A1 value

Alarm output
Process high alarm with standby

Process low alarm with standby
A1hysteresis

A1
hysteresis
ON

ON

Alarm action
OFF

OFF

A1 valueA1 value

Alarm output: A1output terminals 9 and 10 are closed.
: A1output terminals 9 and 10 are closed or opened.
: A1 output terminals 9 and 10 are opened.
: Standby functions.

A2 output terminals: 7 and 8
A1 and A2 indicators light when their output terminals are closed,
and go off when their output terminals are opened.
For A2, read “A2” for “A1”.

46

10 Control Actions

10.1 Fuzzy Self-tuning
Fuzzy self-tuning is a function that performs fine adjustment of PID values automatically.
Stable control can be carried out even if the conditions of the production process arechanged due to various external factors (types and rates of production).
(1) When using the controller for the first time, perform the AT (auto-tuning) or set theproper PID values by keypad operation.
(2) When control initiates, the controller performs this function by the previously adjusted
PID values.
(3) When control result is disordered by disturbance or a change in the process, thecontroller checks the convergence status, and performs a fine adjustment of PID
values if necessary.
The instrument is constantly in self-tuning status, and when deviation occurs, the tuning
starts.
• If the convergence is performed smoothly, the PID values are not changed.
• If the convergent speed is slow, the controller corrects the PID values to accelerate theconvergence.
• When overshoot is generated during the convergence, the controller changes the PID
values to correct this.
• When hunting occurs, the controller checks its waveform and performs a fine adjustment
of PID values.
Even in Fuzzy self-tuning status, when very large hunting occurs and the control is notstabilized, AT automatically starts.
When the AT “Perform” is selected by the keypad, AT initiates, and when the control isstabilized, the AT is released and the controller returns to self-tuning status.
When lock mode [] or [] is selected, Fuzzy self-tuning or AT does not work.

For a control system in which load fluctuation periodically occurs, the Fuzzy
self-tuning PID control may malfunction. In such a case, use the controller with the
PID control.
10.2 PID(1) Proportional band (P)Proportional action is the action which the control output varies in proportion to thedeviation between the SV (desired value) and the processing temperature (PV).
If the proportional band is narrowed, even if the output changes by a slight variation of
the processing temperature, and better control results can be obtained as the offset
decreases.
However, if the proportional band is narrowed too much, even slight disturbances may
cause variation in the processing temperature, and control action changes to ON/OFF
control action and the so-called hunting phenomenon occurs. Therefore, when theprocessing temperature comes to the balanced position near the SV and a constanttemperature is maintained, the most suitable value is selected by gradually narrowing the
proportional band while observing the control results.
(2) Integral time (I)Integral action is used to eliminate offset. When the integral time is shortened, thereturning speed to the setting point is accelerated. However, the cycle of oscillation is
also accelerated and control becomes unstable.
(3) Derivative time (D)Derivative action is used to restore the change in the processing temperature according
to the rate-of-change. It reduces the amplitude of overshoot and undershoot width.
If the derivative time is shortened, the restoring value becomes small, and if the derivative
time is extended, an excessive returning phenomenon may occur and the control system
may oscillate.
47

10.3 AT of This Controller
In order to set each value of P, I and D automatically, the AT process has been madeto fluctuate in order to achieve an optimal value.
Sometimes the AT process will not fluctuate if AT is performed at or near room
temperature. Therefore AT may not finish normally.
(A) If there is a large difference between the SV and PV as the temperature is rising
The AT process will fluctuate at the temperature 1.5% of input span lower than the SV.
(DC input: 1.5% of scaling span)
1.5% of input span lower than the SV
SV(4)
TemperatureTime

AT starting point
(1)

(2)

(1) Calculating PID
constant(2) PID constantcalculated(3) Controlled by the
PID constant set
by AT.
(4) 1.5% of input span

(3)

(Fig. 10.3-1)(B) When the control is stable or when PV is within SV (1.5% of input span).
The AT process will fluctuate around the SV. [DC input: (1.5% of scaling span)]SV
TemperatureTime

AT starting point
(1)

(2)

(1) Calculating PID
constant(2) PID constantcalculated(3) Controlled by the
PID constant set
by AT.

(3)(Fig. 10.3-2)(C) When the PV is 1.5% of input span higher than the SV
The AT process will fluctuate at the temperature 1.5% of input span higher than the SV.
(DC input: 1.5% of scaling span)(1) Calculating PID
1.5% of input span higher than the SVconstant
Temperature(2) PID constantcalculated(4)(3)
Controlled
by theSV
PID constant set
by AT.
(4) 1.5% of input spanTime
AT starting point
(1)

(2)

(Fig. 10.3-3)
48(3)

11 Specifications

11.1 Standard Specifications
Mounting:
Flush
Mountable panel thickness: 1 to 3mm [If Soft front cover(FC-R-S) is used, 1 to 2.5mm]
Setting:
Input system using membrane sheet key
Display:
PV display:
Red LED, 4 digits, character size, 8(H) x 4(W)mm
SV/MV/TIME display: Green LED, 4 digits, character size, 8(H) x 4(W)mm
MEMO display (Set value memory number):Yellow LED, 1 digit, character size, 8(H) x 4(W)mm
Accuracy (Indication, setting)
Thermocouple:
Within 0.2% of each input span 1 digit
K, J, T inputs, range less than 0 (32 ):Within 0.4% of each input span 1 digit
R, S inputs, range 0 to 200 (0 to 400 ): Within 4 (8 )B input, range 0 to 300 (0 to 600 ): Accuracy is notguaranteed.
Cold junction temperature compensation accuracy:
1 (at 0 to 50 )RTD:
Within 0.2% of each input span 1 digit
Direct current:
Within 0.2% of each input span 1 digit
DC voltage:
Within 0.2% of each input span 1 digit
Input sampling period:
125ms(When EA, EV or W, W3 option is ordered: 500ms)
Input
Thermocouple:
K, J, R, S, B, E, T, N, PL- , C (W/Re5-26)
External resistance, 100 or less
When input is burnt out, Overscale
RTD:
Pt100, JPt100, 3-wire system
Allowable input lead wire resistance: 10 or less per wire
When input is burnt out, Overscale
Direct current:
0 to 20mA DC, 4 to 20mA DC
Input impedance, 50
Allowable input current, 100mA or less
When input is disconnected:
0 to 20mA: The same as 0mA
4 to 20mA: Underscale
DC voltage:
0 to 1V DC
Input impedance, 1M or more
Allowable input voltage, 5V or less
Allowable signal source resistance, 2k or less
When input is disconnected, Overscale
OUT1 (Control output 1)
Relay contact:
1a1b
Control capacity:
3A 250V AC (resistive load)
1A 250V AC (inductive load cos =0.4)
Electrical life: 100,000 cycles
Non-contact voltage: For SSR drive+2
12 0V DC maximum 40mA DC (short circuit protected)
Direct current:
4 to 20mA DC (Isolated type)
Load resistance, maximum 550
49

Alarm 1 output
The alarm action point is set by the deviation from the SV (except Process alarm).
[When A1 action Energized is selected]
When the input goes outside the range, the output turns ON or OFF (in the case of
High/Low limit range alarm).
[When A1 action De-energized is selected]
The output acts conversely.
An alarm can be selected by Rotary switch and DIP switch from 13 types of alarm(including No alarm) plus Pattern end output: High limit alarm, Low limit alarm,
High/Low limits alarm, High/Low limit range alarm, Process high alarm and Process
low alarm, the standby function which is applied to them respectively, as well as
Pattern end output. [See Section 9.6 Alarm 1 (A1), Alarm 2 (A2) Action.]
Setting accuracy: Within 0.2% of each input span 1 digit
Action:ON/OFF action
Hysteresis:
Thermocouple, RTD inputs: 0.1 to 100.0 ( )DC input: 1 to 1000 (The placement of the decimal point follows
the selection.)
Output:Relay contact 1aControl capacity: 3A 250V AC (resistive load)
1A 250V AC (inductive load cos =0.4)
Electrical life: 100,000 cycles
Pattern end output (A1 output)
If the Pattern end output [Rotary switch A1 (SW302) No.7] is selected, and theprogram control is performed, the Pattern end output is turned ON when the program
is completed. The Pattern end output can be turned OFF by pressing the
key.
(See Section 9.4 Pattern End Action.)Controlling action
The fuzzy self-tuning PID, PID, PD or ON/OFF control is selectable by DIP switch.
Fuzzy self-tuning PID control
OUT1 proportional band (P): Automatic
Integral time (I):
Automatic
Derivative time (D):
Automatic
Anti-reset windup (ARW):
Automatic
OUT1 proportional cycle:
1 to 120 sec
OUT1 high/low limit:
0 to 100% (Direct current output: –5 to 105%)
PID control (with AT function)
OUT1 proportional band (P): 0.1 to 999.9%
Integral time (I):
0 to 3600 sec (off when set to 0)
Derivative time (D):
0 to 3600 sec (off when set to 0)
Anti-reset windup (ARW):
Automatic
OUT1 proportional cycle:
1 to 120 sec
OUT1 high/low limit:
0 to 100% (Direct current output, –5 to 105%)
PD control
OUT1 proportional band (P): 0.1 to 999.9%
Derivative time (D):
0 to 3600 sec (off when set to 0)
OUT1 proportional cycle:
1 to 120 sec
Reset:
Proportional band converted value
Thermocouple, RTD input: –199.9 to 999.9 ( )DC input: –1999 to 9999 (The placement of thedecimal point follows the selection.)
OUT1 high/low limit:
0 to 100% (Direct current output: –5 to 105%)50

ON/OFF control
OUT1 ON/OFF hysteresis:

Thermocouple, RTD input: 0.1 to 100.0 ( )DC input: 1 to 1000 (The placement of the decimal
point follows the selection.)
Supply voltage:
100 to 240V AC, 50/60Hz, 24V AC/DC, 50/60Hz
Allowable voltage fluctuation: 100 to 240V AC: 85 to 264V AC
24V AC/DC: 20 to 28V AC/DC
Ambient temperature:
0 to 50 (32 to 122 )Ambient humidity:
35 to 85%RH (non-condensing)Power consumption:
Approx. 15VA
Circuit insulation configuration:
Ground terminal

Power source

Insulated
Insulated

Insulated

OUT1

A

Insulated(*1)

OUT2, A2,
LA or P24
outputInsulated

B

Insulated

CPU

Communication (*2)

Non-insulated

Non-insulated

F

Insulated
Transmission
outputInput
C
CT inputD
External setting input

Set value memory
number external
selection (*2)

EG

A1 output
Insulated

Insulated

Insulated

HB output(*1) When both OUT1 and OUT2 are Direct current output type or Non-contact voltage
output type, A is not electrically insulated from B.
(*2) When OUT1 is Direct current output type or Non-contact voltage output type,
A is not electrically insulated from F, and A is not electrically insulated from G.
When OUT2 is Direct current output type or Non-contact voltage output type,
B is not electrically insulated from F, and B is not electrically insulated from G.
Insulation resistance
10M or more, at 500V DC
Insulation test must not be carried out between A-B in the case of (*1) above,and between A-F, A-G, B-F, B-G, C-D-E and F-G in the case of (*2) above
because they are not insulated from each other.
Dielectric strength
Between input terminal and ground terminal: 1.5kV AC for 1 minuteBetween input terminal and power terminal: 1.5kV AC for 1 minuteBetween output terminal and ground terminal: 1.5kV AC for 1 minuteBetween output terminal and power terminal: 1.5kV AC for 1 minuteBetween power terminal and ground terminal: 1.5kV AC for 1 minute
Weight:Approx. 320g
External dimensions: 48 x 96 x 100mm (W x H x D)
Material:
Case, Front panel: Flame-resistant resin
Color:
Case, Front panel: Light gray51

Attached functions: [Control output OFF function], [Set value lock], [SV high/low limit][Sensor correction], [Multi-range], [Multi-function], [Simplified programmable controller][Burnout]
When the thermocouple or RTD input is burnt out, OUT1 and OUT2 are turned off (forDirect current output type, OUT1, OUT2 low limit value), and the PV display flashes“”.
For manual control, the preset MV is outputted.
When DC input is disconnected, the PV display flashes “” for 4 to 20mA DC
input, and “” for 0 to 1V DC input.
For 0 to 20mA DC, the PV display indicates the value corresponding with 0mA input.
[Input error indication]
Output statuswhen input
error (*1)

Contents andindication
Overscale
Measured value
has exceeded
indication range
high limit value.
"" flashes.
Underscale
Measured value
has dropped
below indication
range low limit
value.
"" flashes.

Output status

OUT1
OUT2
Direct action Reverse action Direct action Reverse action
ON (20mA)
ON (20mA)

or OUT1high limit

value (*2)
OFF (4mA)
or OUT1 low
limit value
OFF (4mA)
or
OUT1 low
limit value

OFF (4mA)
or
OUT1 low
limit value

OFF (4mA)
or
OUT2 low
limit value

ON (20mA)
or OUT1high limit
value(*2)
OFF (4mA)
or OUT1 low
limit value

ON (20mA)
or OUT2high limit
value(*2)
OFF (4mA)
or OUT2 low
limit value

or OUT2high limit

value (*2)
OFF (4mA)
or OUT2 low
limit value
OFF (4mA)
or
OUT2 low
limit value(*1) This is available only for DC input.

For manual control, the preset MV is outputted.
(*2) Outputs a value between OFF (4mA) and ON (20mA) or between OUT1 (or OUT2)
low limit value and OUT1 (or OUT2) high limit value, depending on deviation.
TC, RTD inputs
Input
Input range
Indication range
Control range–199.9 to 400.0–199.9 to 405.0–205.0 to 405.0
T–199.9 to 750.0–199.9 to 759.0–209.0 to 759.0–199.9 to 850.0–199.9 to 860.0–210.0 to 860.0–200 to 850–210 to 860–210 to 860
Pt100–199.9 to 999.9–199.9 to 999.9–211.0 to 1010.9–320 to 1560
–338 to 1578
–338 to 1578
–199.9 to 500.0–199.9 to 506.0–206.0 to 506.0
JPt100–199.9 to 900.0–199.9 to 910.9–211.0 to 910.9
Indication range and Control range for thermocouple inputs other than the above:[Input range low limit value – Input span x 1%] to [Input range high limit value + Input
span x 1%]• DC input
Indication range: [Scaling low limit value – Scaling span x 1%] to [Scaling highlimit value + Scaling span x 10%]
However, if the input value is out of the range –1999 to 9999,
the PV display flashes “” or “”.
Control range:[Scaling low limit value – Scaling span x 1%] to [Scaling highlimit value + Scaling span x 10%]52

[Self-diagnosis]
The CPU is monitored by a watchdog timer, and when an abnormal status is foundon the CPU, the controller is switched to warm-up status.
[Automatic cold junction temperature compensation] (Thermocouple input type)This detects the temperature at the connecting terminal between thermocouple andthe instrument, and always maintains it at the same status as if the reference
junction temperature was at 0 (32 ).
[Warm-up indication]
For approximately 2 seconds after power-on, the input type and the temperature unit
are indicated on the PV display, and the input range high limit value (for DC input,
scaling high limit value) is indicated on the SV/MV/TIME display.
[SV ramp function]
When the SV is adjusted, it approaches the new SV by the preset rate-of-change(Rising/Falling value per minute).
When the power is turned on, the control starts from the PV and approaches the SV
by the rate-of-change.
For the program control, this function will not work.
Accessories:
One-touch type mounting brackets: 1 set
Instruction manual:
1 copyUnit label:
1 labelCurrent transformer (CT):
1 piece(CTL-6S) (When W (20A) option is ordered.)(CTL-12-S36-10L1U) (When W (50A) option is ordered.)
Current transformer (CT):
2 pieces(CTL-6S) (When W3 (20A) option is ordered.)(CTL-12-S36-10L1U) (When W3 (50A) option is ordered.)
Wire harness:
3m (When the W/W3, TA/TV or EA/EV option isordered.)Screw type mounting brackets:
1 set (When the BL option is ordered.)
Gasket:1 piece (When the IP option is ordered.)Terminal cover:
1 piece (When the TC option is ordered.)53

11.2 Optional Specifications
Alarm 2 (Option code: A2)
The alarm action point is set by the deviation from the SV (except Process alarm).
[When A2 action Energized is selected]
When the input goes outside the range, the output turns ON or OFF (in thecase of High/Low limit range alarm).
[When A2 action De-energized is selected]
The output acts conversely.
One alarm can be selected by Rotary switch and DIP switch from 13 types of alarm(including No alarm) and Pattern end output: High limit alarm, Low limit alarm,
High/Low limits alarm, High/Low limit range alarm, Process high alarm andProcess low alarm, the standby function which is applied to them respectively, as
well as Pattern end output.
• When Alarm 2 (A2 option) and Loop break alarm (LA option) are ordered together,
they utilize common output terminals.
• If Alarm 2 (A2 option) is ordered, Heating/Cooling control (DR, DS, DA option) or
Insulated power output (P24 option) cannot be ordered.
Setting accuracy: Within 0.2% of each input span 1 digit
Action:ON/OFF action
Hysteresis:
Thermocouple, RTD inputs: 0.1 to 100.0 ( )DC input: 1 to 1000 (The placement of the decimal point follows
the selection.)
Output:Relay contact, 1a
Control capacity: 3A 250V AC (resistive load)
1A 250V AC (inductive load cos =0.4)
Electrical life: 100,000 cycles
Heating/Cooling control (Option code: DR, DS, DA)
If this option is ordered, Alarm 2 (A2 option), Loop break alarm (LA option) or Insulated
power output (P24 option) cannot be ordered.
OUT2 proportional band: 0.0 to 10.0 times OUT1 proportional band
(ON/OFF control action when setting the value to 0.0.)
OUT2 integral time:
The same as OUT1 integral time
OUT2 derivative time:
The same as OUT1 derivative time
OUT2 proportional cycle: 1 to 120 sec
Overlap/Dead band:
OUT1 proportional band converted value
Thermocouple, RTD inputs: –199.9 to 999.9 ( )DC input: –1999 to 9999 (The placement of the decimal point follows theselection.)
OUT2 ON/OFF hysteresis:
Thermocouple, RTD inputs: 0.1 to 100.0 ( )DC input: 1 to 1000 (The placement of the decimal point follows the selection.)
Output[DR] Relay contact, 1a
Control capacity: 3A 250V AC (resistive load)
1A 250V AC (inductive load cos =0.4)
Electrical life: 100,000 cycles[DS] Non-contact voltage (for SSR drive)+2
12 0 V DC Maximum 40mA DC (short circuit protected)[DA] Direct current: 4 to 20mA DC (Isolated type)
Load resistance: Maximum 550
OUT2 high/low limit: 0 to 100% (Direct current output: -5 to 105%)
OUT2 action mode: Selectable by keypad
Air cooling (Linear characteristic), Oil cooling (1.5th power of the linear
characteristic), Water cooling (2nd power of the linear characteristic).
54

Transmission output (Option code: TA, TV)
Converting the value (PV transmission, SV transmission, MV transmission) toanalog signal every 125ms, outputs the value in current or voltage.
One transmission output (PV, SV, MV) can be selected by keypad.
Resolution:
1/10000
Current (TA):
4 to 20mA DC (load resistance maximum 500 )Voltage (TV):
0 to 1V DC (load resistance minimum 100k )Output accuracy: Within 0.3% of Transmission output span
Serial communication (Option code: C5, C)
The following operations can be executed from an external computer.
(1) Reading and setting of the SV, PID values and various set values.
(2) Reading of the PV and action status.
(3) Function change
Communication line:
EIA RS-485 (C5 option)
EIA RS-232C (C option)
Communication method: Half-duplex communication
Synchronization method: Start-stop synchronization
Communication speed:
2400, 4800, 9600, 19200bps (Selectable by keypad)
Data format
Start bit: 1
Data bit: 7
Parity: Even parity
Stop bit: 1
Communication protocol: Shinko protocol, Modbus ASCII (Selectable by keypad)(When Modbus protocol is selected, the Digital externalsetting can not be used.)
Digital external setting:Receives digital set value from Shinko Programmablecontroller PC-900, PCD-33A (with SVTC option).
Set value memory number external selection (Option code: SM)
Selects the Set value memory number from 7 files (the below-mentioned data as onefile) by external terminals:
SV, OUT1 proportional band, Integral time, Derivative time, OUT2 proportionalband, A1 value, A2 value, Overlap/Dead band, OUT1 high limit value, OUT1 low
limit value, OUT2 high limit value, OUT2 low limit value
Memory number: 1 to 7 (7 files)
Data:
12
External setting input (Option code: EA, EV)
SV can be set by External analog signal.
Setting signal : Direct current [EA]
0 to 20mA DC, 4 to 20mA DC
Allowable input current: 100mA DC or less
Input impedance, 50 (non-isolated from input): DC voltage [EV]0 to 1V DC, 1 to 5V DC
Allowable input voltage: 0 to 1V DC: 5V DC or less
Input impedance, 100k (non-isolated from input)
Setting signal sampling period: 500ms(If this External setting input is ordered, the input sampling period also changes
to 500ms.)
Heater burnout alarm (Option code: W, W3)
Watches the heater current with CT (current transformer), and detects the burnout.
• For the direct current output type, this option cannot be ordered.
• If this option is ordered, the input sampling period also changes to 500ms.
Rated current: 20A [W (20A), W3 (20A) option] or
50A [W (50A), W3 (50A) option]
Must be specified
55

Setting range:

20A: 0.0 to 20.0A (Indication: 0.0 to 50.0), 50A: 0.0 to 50.0A
The alarm will be disabled when set to 0.0.
Setting accuracy: Within 5% of heater rated current
Action:ON/OFF action
Output:Relay contact, 1a
Control capacity: 3A 250V AC (resistive load)
1A 250V AC (inductive load, cos =0.4)
Electrical life: 100,000 cycles
Loop break alarm (Option code: LA)
Detects the breaking status on the loop such as heater burnout, sensor burnout or
actuator trouble.
• When this option and Alarm 2 (A2) option are ordered together, they utilize common
output terminals.
• If this option is ordered, Heating/Cooling control (DR, DS, DA option) or Insulated
power output (P24 option) cannot be ordered.
Output: Relay contact, 1a
Control capacity: 3A 250V AC (resistive load)
1A 250V AC (inductive load, cos =0.4)
Electrical life: 100,000 cycles
Insulated power output (Option code: P24)
Can be used as a small capacity power source for each sensor and converter.
If this option is ordered, Alarm 2 (A2 option), Heating/Cooling control (DR, DS, DA
option) or Loop break alarm (LA option) cannot be ordered.
Output voltage:
24 3V DC (when the load current is 30mA.)
Ripple voltage:
Within 200mV DC (when the load current is 30mA DC.)Max load current: 30mA DC
Screw-type mounting brackets (Option code: BL)
Mountable panel thickness: 1 to 8mm (When soft front cover FC-R-S is used: 1 to 7.5mm)
Color black (Option code: BK)
Front panel: Dark gray, Case: Black
Drip-proof/Dust-proof (IP54) (Option code: IP)
Effective only for front panel surface, case section is excluded.
Mount the controller vertically to the flat, rigid panel to ensure it adheres to theDrip-proof/Dust-proof specification.
The front cover (soft type, FC-R-S, sold separately) is recommended forcomprehensive Drip-proof/Dust-proof protection.
Terminal cover (Option code: TC)
Electrical shock protection terminal coverUser specified
Input, Scale range: Shipped as specified range.
Alarm type:
Shipped as specified alarm type [A1, A2 (A2 is optional)].
Control action:
Shipped as specified control action.
OUT2 action mode: Shipped as specified cooling action mode (DR, DS, DA option).
Transmission output: Shipped as specified output (TA, TV option).
External setting input: Shipped as specified input (EA, EV option).
Transmission output scaling: Shipped as specified scaling range.

56

12 Troubleshooting
If any malfunctions occur, refer to the following items after checking the power andthe wiring.

Warning
Turn the power supply to the instrument off before wiring or checking.
Working on or touching the terminal with the power switched on may result insevere injury or death due to electric shock.
<Indication>
Problem
The PV display is
indicating [], noindication or only PV.
The PV display is
flashing [].

The PV display is
flashing [].

The indication of the
PV display is irregularor unstable.

Possible Cause and Solution• Control output OFF function is working.
To release the function, press the
key for approx.
1 sec.
• Thermocouple or RTD is burnt out.
Change each sensor.
[Thermocouple]
If the input terminals of the instrument are shorted, and if avalue around room temperature is indicated, theinstrument is likely to be operating normally, however, thesensor may be burnt out.
[RTD]
If approximate 100 of resistance is connected to the
input terminals between A-B of the instrument andbetween B-B is shorted, and if a value around 0 (32 )is indicated, the instrument is likely to be operating
normally, however, the sensor may be burnt out.
• Check whether the lead wire of thermocouple or RTD issecurely mounted to the instrument input terminals.
• Check if polarity of thermocouple or compensating lead wireis correct.
• Check whether codes (A, B, B) of RTD agree with the
instrument terminals.
• Suitable sensor input has not been selected.
Set the sensor input properly using the Rotary switch(SW303) and DIP switch (SW304).
• Temperature unit ( or ) is not correct.
Set it to a suitable unit with the DIP switch (SW304).
• Sensor correction value is not suitable.
Set it to a suitable value.
• Specification of thermocouple or RTD is improper.
• AC leaks into the thermocouple or RTD circuit.
Use an ungrounded type sensor.
• There may be equipment that interferes with, or makes
noise near the controller.
Keep equipment that interferes with or makes noise awayfrom the controller.
57

<Key operation>
Problem
The setting modecannot be selected.
• Unable to set the SV,
P, I, D, OUT1
proportional cycle,
alarm value, etc.
• The values do notchange by the,keys.
The setting indication
does not change within
the input range even if
the
,
keys
are pressed, and newvalues are unable to be
set.
Program control does
not start even if the
key is pressedin the Program mode.

Possible Cause and Solution• Manual control is selected.
Change the mode to Automatic control.
• Set value lock (Lock 1 or Lock 2) is selected.
Release the lock selection.

• AT (Auto-tuning) is performing.
Cancel AT.
• SV high limit or SV low limit in Auxiliary function setting
mode 1 may be set at the point where the value does notchange.
Set it to a suitable value in Auxiliary function setting mode 1.

• Step time has not been set.
Set the step time.

<Control>
ProblemPV does not rise.

Possible Cause and Solution• Thermocouple or RTD is burnt out.
Replace the sensor.
• Check whether the sensor or control output terminals
are securely mounted to the instrument terminals.
• Check whether the wiring of sensor or control output
terminals is correct.
Control is not• Control output OFF function is working.
performing.
Press thekey to cancel Control output OFF function.
(Only PV display is• Program mode is selected.
indicated)
To start Program control, press thekey.
To perform Fixed value control, press the
key forapprox. 3 sec while holding down the
key.
OUT1 or OUT2 remains • OUT1 low limit value or OUT2 low limit value is set to 100%in an ON status.
or higher in Auxiliary function setting mode 2.
Set it to a suitable value.
OUT1 or OUT2 remains • OUT1 high limit value or OUT2 high limit value is set to 0%
in an OFF status.
or less in Auxiliary function setting mode 2.
Set it to a suitable value.

For all other malfunctions, please contact our main office or dealers.

58

13 Character Table<Main setting mode>
Character
Item
SV<Sub setting mode>
Character
Item
Set value memory number
AT Perform/Cancel
OUT1 proportional band
OUT2 proportional band
Integral time
Derivative time
OUT1 proportional cycle
OUT2 proportional cycle

xx.x
xx.x: Heater

Manual reset
Alarm 1 value
Alarm 2 value
Heater burnout alarm value

Default

Data

Default

Data

01
Cancel2.5%
1.0 times
200 sec
50 secR/M: 30 sec
S/M: 3 sec
DR: 30 sec
DS: 3 sec
0.0
0
0
0.0A

current value

Loop break alarm time
Loop break alarm span
<Auxiliary function setting mode 1>
Character
Item
Set value lockSV high limit
SV low limit
Sensor correction
Overlap/Dead band
Remote/Local setting
Instrument number
Communication speed
Communication protocol<Auxiliary function setting mode 2>
Character
Item
Scaling high limit value
Scaling low limit value
Decimal point place
PV filter time constant
OUT1 high limit value
OUT1 low limit value
OUT1 ON/OFF hysteresis
OUT2 action mode
OUT2 high limit value
OUT2 low limit value
OUT2 ON/OFF hysteresis

0 minutes
0
Default
Unlock
400
0
0.0
0.0
Local
0
9600bps
Shinko protocol

Default
1370-200
No decimal point
0.0 sec
100%0%1.0
Air cooling
100%
0%
1.0
Alarm 1 action Energized/De-energized Energized
Alarm 2 action Energized/De-energized Energized
59

Data

Data

Alarm 1 hysteresis
1.0
Alarm 2 hysteresis
1.0
Alarm 1 delay time
0 sec
Alarm 2 delay time
0 sec
External setting input high limit
400
External setting input low limit
0Transmission output
PV transmission
Transmission output high limit
400
Transmission output low limit0
Indication when control output OFF OFF on the PV display
SV rise rate
0 /minuteSV fall rate
0 /minuteOutput status when input error
Outputs OFF(4mA)or OUT1(OUT2)
low limit value.
<Program mode>
Character
Item
Default
Fixed value control/Program control Fixed value
control
Step 1 time
00.00
Step 2 time
00.00
Step 3 time
00.00
Step 4 time
00.00
Step 5 time
00.00
Step 6 time
00.00
Step 7 time
00.00

Data

***** Inquiries *****For any inquiries about this unit, please contact the vendor
where you purchased the unit after checking the following.
[Example]• Model --------------------------- FCR -13A-R/M• Input type ---------------------- K• Option --------------------------- A2, TV, C5, W (20A)• Serial number ----------------- No. 123F05000
In addition to the above, please let us know the details of themalfunction, if any, and the operating conditions.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION

Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, Japan
http://www.shinko-technos.co.jp
URL:
Tel : 81-72-727-6100
E-mail:overseas@shinko-technos.co.jp
Fax: 81-72-727-7006
No. FCR11E15 2012.10