Discontinued Model Replacement 2018.05.08
This data mainly describes the differences at the time of product replacement.
For correct and safe use, be sure to read the instruction manual for precautions on use.
●Notification of Program Controller PCD-33A Discontinuation
Discontinued model
Programmable Controller
PCD-33A

Replacement Model
Programmable Controller
PCB1●Notes When Using Replacement Model
・The number of patterns is different.
9 patterns (linkable) → 10 patterns (linkable)
・The number of steps is different.
81 steps (9 steps/pattern) → 100 steps (10 steps/pattern)
・SV high limit and SV low limit functions are not available. → Scaling high limit, scaling low limit
・Case color black (OP: BK) is standard.
Case: Black color only・Console communication (CMD-001) is available.
・Control output: Non-contact voltage is different.
+2

12 0 V DC → 12 V DC

15 %・Serial communication: The selection of communication speed is different.
MAX 19200 bps (2400, 4800, 9600, 19200 bps) → MAX 38400 bps (9600, 19200, 38400 bps)
・The rated range of the input range is different. (See Input Range on p. 6)
・Depth of control panel interior is different.
Depth of control panel interior: 98.5 mm → 58.8 mm
・Alarm 1, 2 (standard) → Event output EV1 (standard), Event output EV2 (option: EV2 (DR)),
Event output EV3 (option: EV3 (D□))
・The power supply voltage terminals are different.
②(+)－③(-) → ⑬(+)－⑭(-)
・Control output (OUT1) terminal and contact structure (relay contact output) are different.
⑤(+)－⑥(-) → ⑮(+)－⑯(-)
1a1b → 1a・Thermocouple input terminals are different.
⑱(+)－⑲(-) → ㉒(+)－㉔(-)
・RTD input terminals are different.
⑱(A)－⑲(B)－⑳(B) → ㉒(A)－㉓(B)－㉔(B)・DC voltage input terminals are different.
⑱(+)－⑲(-) (0 to 1 V DC, 0 to 5 V, 1 to 5 V, 0 to 10 V DC) → ㉒(+)－㉔(-) ( 0 to 1 V DC)㉑(+)－㉔(-) (0 to 5 V, 1 to 5 V, 0 to 10 V DC)
・Direct current input terminals are different. Also, a 50 Ω receiving resistor is not required.
⑱(+)－⑲(-) → ㉒(+)－㉔(-)
・Serial communication terminals are different.
⑪YA(-)－⑭YB(+)－⑰COM → ⑩YA(-)－⑪YB(+)－⑫SG
・External operation terminals are different.

Select input terminals ⑧-⑫ or ⑨-⑫.

1

・Weight is different.
Approx. 370 g → Approx. 220 g
・OUT1 proportional band setting range is different.
0 to 1000 ℃, 0.0 to 999.9 ℃, DC input 0.0 to 100.0 % → 0 to input span or 0.0 to input span，

DC input 0.0 to 1000.0 %・The setting range of sensor correction is different. The slope of the sensor input value can be set using thesensor correction coefficient.
( )，DC input: -1000 to 1000 → -1000.0 to 1000.0-100.0 to 100.0( )，DC input: -10000 to 10000
・Terminal covers
Option: TC → Sold separately (TC-BCD2)●Differences Between Discontinued Model and Replacement Model
Case Color

External
dimensions

Panel cutdimensions

Wiringconnection

Ratingperformance

Actioncharacteristics

Operation
method

○

○

◎

×

○

○

○

(◎: Compatible (no change)○: Slight change

×: Large change)●Discontinued Model and Replacement Model
Discontinued model
PCD-33A-R/M
PCD-33A-S/M
PCD-33A-A/M

Replacement Model
PCB1R00-00
PCB1S00-00
PCB1A00-00

PCD-33A-□/M, 1

PCB1□10-00

Power supply voltage 24 V AC/DC

PCD-33A-□/M [Alarm 1, 2(A1, A2)]

PCB1□□0-10

PCD-33A-□/M (External operation)

PCB1□□0-19

PCD-33A-□/M, C5

PCB1□□0-□6

Alarm 1 and 2 (A1, A2) are selectable byevent output EV1 and EV2 allocation
External operation are selectable by event
input DI1 and DI2 allocation
Serial communication

PCD-33A-□/M, C5, P24

PCB1□□0-46

PCD-33A-□/M, SVTC

PCB1□□0-06

PCD-33A-□/M, SVTC, P24

PCB1□□0-46

PCD-33A-□/M, P24

PCB1□□0-40

SVTC is selectable by communication
protocol selection
SVTC is selectable by communication
protocol selection
Isolated power supply output

PCD-33A-□/M, TC

PCB1□□0-□□

Terminal cover sold separately (TC-BCD2)

PCD-33A-□/M, BK

PCB1□□0-□□Standard case color is black2

Remarks

●Mounting Dimensions, Panel Cut Dimensions (unit: mm)

91
106.2

130

PCD-33A

92

＋0.8
0

□92＋0.8
0

98.5
104.5( オプション:
104.5 TC付加時)(When optional TC is added)

n×96－3＋0.5
0Horizontal close mounting
n: Number of units installed

130

PCB1

＋0.8
0

□92＋0.8
0

92

11.5
96

n×96－3＋0.5
0Horizontal close mounting
n: Number of units installed3●Terminal Arrangement
PCD-33A

通信(RS-485)"C5"

Communication (RS-485) "C5"
または設定値デジ
Setting value digitaltransmission "SVTC"

接地
Power
supply
計器電源

GND

+

2

24V DC

Control output
制御出力+
Non-contact
無接点電圧
voltage

直流電流Directcurrent3

5100～240VAC
100
to 240V AC
or
24V
AC/DC
または 24V AC

リレー接点

11

2

12

イベント出力Event output (EVT)
(EVT)
13

3

H 4

14

C 5

15

Relay contact

-

6

温度警報1
Temperaturealarm(A1) 1 (A1)
Isolated
絶縁電源
power output :出力"P24"
"P24"

1

+

-

9
10

Temperature
温度警報2
alarm
(A2) 2 (A2)タル伝送"SVTC"
YA(-)
11

L 6

16

7

17

8

18

9

19

10

YB(+)

14

External operation

外部操作入力
input

COM

17

+

-

18
19
20

20

TC

Power supply: 100 to 240V AC or 24V AC/DC

Control output: Relay contact, non-contact voltage, DC current

Temperature Alarm 1: (A1) output

Temperature Alarm 2: (A2) output

Isolated power output: (OP: P24)

Serial communication: RS-485(OP: C5)

A
B

18
19

+

-

B
RTD

DC

Setting value digital transmission : (OP: SVTC)TC: Thermocouple input

RTD: Resistance thermometer input

DC: DC voltage or direct current input

Event output: (EVT) output Time signal, pattern end, RUN output

External operation input: Program control execution (RUN), Program control termination (STOP)

・The dotted line indicates the option.
・When option "C5" or "SVTC" is added, the external operation does not operate.
C5" and "SVTC"
cannot be added at the same time.
・When using DC current, connect a receiving resistor (50 ) between the input terminals.

4

PCB1

Terminal symbol
PWR
O1

Description
Power supply voltage 100 to 240 V AC or 24 V AC/DC (For a 24 V AC/DC
power source, ensure polarity is correct when using DC).
Control output OUT1

EV1
EV2
O2
P24

Event output EV1
Event output EV2 [Option: EV2, EV3(DR)]Control output OUT2 (Option: EV2, DS, DA, EV3D )Isolated power supply output 24 V DC (Option: P24)

TC
RTD
DC
CT1
CT2
RS-485EVENT INPUT
EV3

Thermocouple input
RTD input
Direct current・DC voltage input
Current transformer (CT) input 1 (Option: C5W, EIW, W)
Current transformer (CT) input 2 (Option: C5W, EIW, W)
Serial communication RS-485 (Option: C5W, C5)
Event input DI1 (Option: C5W, EIW, EIT, C5, EI)
Event input DI2 (Option: C5W, EIW, EIT, C5, EI)
Event output EV3 (Option: EV3D , EI)

TRANSMIT OUTPUT

Transmission output (Option: EIT)

Main Terminal Number Replacement
PCD-33A

PCB1Power supply voltage②(+)－③(-)⑬(+)－⑭(-)

Control output OUT1⑤(+)－⑥(-)⑮(+)－⑯(-)

Thermocouple input⑱(+)－⑲(-)

㉒(+)－㉔(-)

RTD input⑱(A)－⑲(B)－⑳(B)㉒(A)－㉓(B)－㉔(B)DC voltage input⑱(+)－⑲(-) (0 to 1 V，0 to 5 V,㉒(+)－㉔(-)(0 to 1 V DC)

1 to 5 V, 0 to 10 V DC)㉑(+)－㉔(-)(0 to 5 V, 1 to 5 V, 0 to 10 V DC)⑱(+)－⑲(-)

㉒(+)－㉔(-)

DC current input(Receiving resistor (50Serial communication⑪YA(-)－⑭YB(+)－⑰COM

5) is not required)⑩YA(-)－⑪YB(+)－⑫SG

●Input Range
Rated Scale
PCD-33A

Input
K
J
R
S
Thermocouple B
E
T
N
PL-Ⅱ
C(W/Re5-26)
Pt100
RTD

JPt100

Direct current
DC voltage

Input Range-200 to 1370
-320 to 2500
-199.9 to 400.0-199.9 to 750.0-200 to 1000
-320 to 1800
0 to 1760
0 to 3200
0 to 1760
0 to 3200
0 to 1820
0 to 3300-200 to 800-320 to 1500
-199.9 to 400.0-199.9 to 750.0-200 to 1300
-320 to 2300
0 to 1390
0 to 2500
0 to 2315
0 to 4200-199.9 to 850.0-199.9 to 999.9-200 to 850-300 to 1500
-199.9 to 500.0-199.9 to 900.0

Resolution
1 ( )0.1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
0.1 ( )
1 ( )
1 ( )
1 ( )
0.1 ( )1 ( )
0.1 ( )

-200 to 500

1
1
1
1
1
1
1

4 to 20 mA DC
0 to 20 mA DC
0 to 1 V DC
0 to 5 V DC
1 to 5 V DC
0 to 10 V DC

-300 to 900-1999 to 9999(*1)(*2)-1999 to 9999(*1)(*2)-1999 to 9999(*1)-1999 to 9999(*1)-1999 to 9999(*1)-1999 to 9999(*1)( )(*1): For direct current/voltage input, scaling and decimal point position can be selected.
(*2): Direct current input requires an optional receiving resistor (50 ).

PCB1

Input
K

Thermocouple

J
R
S
B
E
T
N
PLC(W/Re5-26)
Pt100

RTD

JPt100

Direct current
DC voltage

4 to 20 mA
0 to 20 mA
0 to 1 V
0 to 5 V
1 to 5 V
0 to 10 V

Input Range-200 to 1370
-328 to 2498
-200.0 to 400.0-328.0 to 752.0-200 to 1000
-328 to 1832
0 to 1760
32 to 3200
0 to 1760
32 to 3200
0 to 1820
32 to 3308
-200 to 800-328 to 1472
-200.0 to 400.0-328.0 to 752.0-200 to 1300
-328 to 2372
0 to 1390
32 to 2534
0 to 2315
32 to 4199
-200.0 to 850.0-328.0 to 1562.0-200 to 850-328 to 1562
-200.0 to 500.0-328.0 to 932.0-200 to 500-328 to 932-2000 to 10000(*)-2000 to 10000(*)-2000 to 10000(*)-2000 to 10000(*)-2000 to 10000(*)-2000 to 10000(*)(*): Scaling and decimal point position can be selected.

6

Resolution
1 ( )0.1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
1 ( )
0.1 ( )
1 ( )
1 ( )
1 ( )
0.1 ( )1 ( )
0.1 ( )
1 ( )
1
1
1
1
1
1

●Rating Performance
Item
Depth of control panel interior
WeightPower
Commercial
supplyvoltagevoltageLow voltage
Power consumption

Input

Thermocouple
(TC)

RTD

Direct current(DC)

DC voltage
(DC)

Range
PV filter
Display

PV display
SV display
PTN display
STEP display

PCD-33A

PCB1

98.5 mmApprox. 370 g
100 to 240 V AC (85 to 264 V AC)

58.8 mmApprox. 220 g
100 to 240 V AC (85 to 264 V AC)

24 V AC/DC (20 to 28 V AC/DC)Approx. 8 VA

K, J, R, S, B, E, T, N, PL-Ⅱ,
C(W/Re5-26)
External resistance: 100
or less(B: 40or less)
Pt100, JPt100 3-conductor type
Allowable input conductor
resistance: 10or less per line
4 to 20 mA, 0 to 20 mA DC(sold separately shunt resistor)
Input impedance: 50

24 V AC/DC (20 to 28 V AC/DC)
100 to 240 V AC: Approx. 8 VA max.
(When the maximum number of
options are added: Approx. 11 VA
max.)
24 V AC: Approx. 5 VA max.
(When the maximum number of
options are added: Approx. 8 VA max.)
24 V DC: Approx. 5 W max.
(When the maximum number of
options are added: Approx. 8 W max.)K, J, R, S, B, E, T, N, PL-Ⅱ,
C(W/Re5-26)
External resistance: 100
or less(B: 40or less)
Pt100, JPt100 3-conductor type
Allowable input conductor
resistance: 10or less per line
4 to 20 mA, 0 to 20 mA DC(Built-in shunt resistor)
Input impedance: 50

Allowable input current: 50 mA DC

Allowable input current: 50 mA DC

or less
0 to 1 V DC
Input impedance: 1 M or more
Allowable input voltage: 5 V DC or
less
Allowable signal source resistance:
2 k or lessor less
0 to 1 V DC
Input impedance: 1 M or more
Allowable input voltage: 5 V DC or
less
Allowable signal source resistance:
2 k or less0 to 5 V, 1 to 5 V, 0 to 10 V DC

0 to 5 V, 1 to 5 V, 0 to 10 V DC

Input impedance: 100 k 以上
Allowable input voltage: 15 V DC or
less
Allowable signal source resistance:
100or less
Multi-range
0 to 10.0 sec
7-segment LED Red 4-digits
18×8 mm (H x W)
7-segment LED Green 4-digits
12.6×6 mm (H x W)
7-segment LED Green 1-digit
12.6×6 mm (H x W)
7-segment LED Green 1-digit
12.6×6 mm (H x W)

Input impedance: 100 k 以上
Allowable input voltage: 15 V DC or
less
Allowable signal source resistance:
100or less
Multi-range
0.0 to 10.0 sec
7-segment LED Red 4.5-digits
24×11 mm (H x W)
7-segment LED Green 4.5-digits
14×7 mm (H x W)
7-segment LED Orange 1.5-digits
14×7 mm (H x W)
7-segment LED Orange 1.5-digits
14×7 mm (H x W)

7

Indication
accuracy

Item
TC

RTD
DC
Sampling period
Time indication accuracy
ProgramNumber of
performance patterns
Number ofsteps
Number of
repetitions
Step time range
Time setting
accuracy
Setting
resolution
Wait value

External
operation

PCD-33A
Within 0.2% of each input span 1 digit
R, S inputs 0 to 200(0 to 400 )within6(12 )B input 0 to 300(0 to 600 ) is outof accuracy guarantee range
K, J, E, N, T inputs less than 0(32) are within 0.4% of the input span 1
digit
Within0.1% of each input span1
digitWithin0.2% of each input span
1
digit
250 msWithin0.5 % of setting time
9 patterns (Linkable)

PCB1
Within 0.2% of each input span 1 digit
R, S inputs 0 to 200(32 to 392 )within6(12 )B input 0 to 300(32 to 572 ) isout of accuracy guarantee range
K, J, E, N, T inputs less than 0(32) are within 0.4% of the input span 1
digit
Within0.1% of each input span1
digitWithin0.2% of each input span
1
digit
125 msWithin0.5 % of setting time
10 patterns (Linkable)

81 (9 steps/pattern)

100 (10 steps/pattern)

0 to 99 hours 59 minutes/step, or 0 to
99 minutes 59 seconds/stepWithin0.5 % of setting time

0 to 10000 times (Repetitions disabled
when set to 0)
0 to 99 hours 59 minutes/step, or 0 to
99 minutes 59 seconds/stepWithin0.5 % of setting time

Temperature: See Rated Scale
Time: 1 minute or 1 second(0 to 100)( ) or(0.0 to100.0) ( )DC input: (0 to 1000)

Temperature: See Rated Scale
Time: 1 minute or 1 second

Input terminals 14 - 17
open to closed: RUN
open from closed: STOP

(OP:EV2, EI) are added, and program
control run (RUN)/stop (STOP)
switching is selected by event input
DI1 or DI2 allocation selection.
Select input terminals ⑧-⑫ or ⑨-⑫.
open to closed: RUN
open from closed: STOP

8

0 to 20 % of input span
DC input: 0 to 20 % of scaling span

Item
Control method
Control
OUT1
parameters Proportionalband (P)Integral time (I)Derivative time(D)
ARW
ON/OFFhysteresis
Output high limit,
Output low limit
Proportional cycle
Control
Relay contact
output(OUT1)
Non-contact
voltage
Direct current
Alarm output

PCD-33A

PCB1

PID control
ON/OFF control
0 to 1000(2000 ) or
0.0 to 999.9(999.9 )DC input: 0.0 to 100.0 %0 to 1000 sec
0 to 300 sce

PID control
ON/OFF control
0 to Input span( ) or
0.0 to Input span( )DC input: 0.0 to 1000.0 %0 to 3600 sec
0 to 1800 sec

0 to 100 %0.1 to 100.0

0 to 100 %0.1 to 1000.0

( )

( )

0 to 100 %0 to 100 %Direct current output: -5 to 105 %Direct current output: -5 to 105 %1 to 120 sce
0.5, 1 to 120 sec
1a1b
1a
3 A 250 V AC (resistive load)
3 A 250 V AC (resistive load)
1 A 250 V AC (inductive load cosφ=0.4) 1 A 250 V AC (inductive load cosφ=0.4)+2
12 0 V DC
12 V DC 15 %Max. 40 mA DC (short circuit protected) Max. 40 mA DC (short circuit protected)
4 to 20 mA DC
4 to 20 mA DC
Load resistance: Max. 550
Load resistance: Max. 550
A1, A2, 2 points standard
Event output EV1 1 point standard
Factory default
Event output EV2 (option)
A1: No action
Event output EV3 (option)
A2: No action
Select alarm output, heater burnout
alarm output, loop break alarm output,
Relay contact 1atime signal output, output during AT,
3 A 250 V AC(resistive load)
pattern end output, output by
1 A 250 V AC(inductive load cosφ=0.4)
communication command, RUN
output, or heating/cooling control
relay contact output from event output
allocation
Factory default
EV1: No action

Relay contact 1a
3 A 250 V AC(resistive load)
1 A 250 V AC(inductive load cosφ=0.4)
Attached
Sensor-100.0 to 100.0( )-1000.0 to 1000.0( )Functionscorrection
DC input: -1000 to 1000
DC input: -10000 to 10000
SV high limit, SV SV high limit value: SV low limit value to Not available
low limit function input range high limit value
Can be replaced by scaling upper limit
SV low limit value: input range low limit and scaling lower limit settingsvalue to SV high limit value
Fixed value
Whenis set for the step time, Whenis set for the step
control
fixed value control is performed in step time, fixed value control is performed inSV.
step SV.
ConsoleNot available
Available (CMD-001)
communication
Environmental Ambient
0 to 50-10 to 55
specification temperature
Ambient
35 to 85 %RH (Non-condensing)
35 to 85 %RH (Non-condensing)humidity

9

●Optional Specifications
Item

PCD-33A

PCB1

Serial communication

EIA RS-485

EIA RS-485(OP: C5, C5W)

Communication protocol

Communication protocol

Shinko protocol

Shinko protocol

SV digital transmission

SV digital transmission

SV digital reception

SV digital reception

Modbus ASCII

Modbus ASCII

Modbus RTU

Modbus RTU

Communication speed

Communication speed

MAX 19200 bps

MAX 38400 bps(2400, 4800, 9600, 19200 bps)

(9600, 19200, 38400 bps)

SV digital transmission

Selectable by communication protocol

Selectable by communication protocol(OP: SVTC)

selection

selection

When SV digital transmission is

When SV digital transmission is

selected by communication protocol

selected by communication protocol

selection, step SV can be transmitted

selection, step SV can be transmitted

digitally in combination with serial

digitally in combination with serial

communication (OP: C5) of ourcommunication (OP: C5, C5W) of ourprogrammable controller.

programmable controller.

When SV digital reception is selected,

When SV digital reception is selected,

step SV can be received by SVTC

step SV can be received by SVTC

command in combination with ourcommand in combination with ourprogrammable controllers [PCA1 orprogrammable controllers [PCA1 or

PCB1 (SV digital transmission is

PCB1 (SV digital transmission is

selected in communication protocol

selected in communication protocol

selection)].

selection)].

Case base color: blackStandard case color: black

Output voltage: 24 V 3 V DC (whenload current is 30 mA DC)
Ripple voltage: Within 200 mV DC(when load current is 30 mA DC)
Max. load current: 30 mA DC

Output voltage: 24 V 3 V DC (whenload current is 30 mA DC)
Ripple voltage: Within 200 mV DC(when load current is 30 mA DC)
Max. load current: 30 mA DC

Exterior color: black(OP: BK)
Insulated power output(OP: P24)

Terminal cover

Electric shock protection terminal cover Sold separately (TC-BCD2)(OP: TC)

10