Installation Instruction Manual

Indoor CO Sensor

DSW-100-CO
No.DSW14JE1 2012.08

PrefaceThank you for purchasing our DSW-100-CO, Indoor CO Sensor.
This manual contains instructions for installation and handling of the DSW-100-CO. To ensure safe and correctuse, thoroughly read and understand this manual before using this sensor. To prevent accidents arising fromthe misuse of this sensor, please ensure the operator receives this manual.

Notes• This sensor should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow all of the warnings, cautions and notices. If they are not observed, serious injury ormalfunction may occur.
• The contents of this instruction manual are subject to change without notice.
• Care has been taken to assure that the contents of this instruction manual are correct, but if there are any
doubts, mistakes or questions, please inform our sales department.
• This sensor is designed to be used in close proximity to the measurement target. Measures must be taken toensure that the operator cannot touch power terminals or other high voltage sections.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos Co., Ltd. is not liable for any damage or secondary damage(s) incurred as a result of using
this product, including any indirect damage.

Safety Precautions (Be sure to read these precautions before using our products.)

The safety precautions are classified into 2 categories: “Warning” and “Caution”.
Depending on the circumstances, procedures indicated byCaution may cause serious results, so be sureto follow the directions for usage.

Warning
Procedures which may lead to dangerous conditions and cause death or serious injury, if not carriedout properly.

CautionProcedures which may lead to dangerous conditions and cause superficial to medium injury or physicaldamage or may degrade or damage the product, if not carried out properly.

Warning• To prevent an electric shock or fire, only Shinko or other qualified service personnel may handle theinner assembly.
• To prevent an electric shock, fire or damage to the instrument, parts replacement may only beundertaken by Shinko or other qualified service personnel.

SAFETY PRECAUTIONS• To ensure safe and correct use, thoroughly read and understand this manual before using thisinstrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring
equipment. Verify correct usage after purpose-of-use consultation with our agency or main office.
(Never use this instrument for medical purposes with which human lives are involved.)• External protection devices such as protective equipment against excessive temperature rise, etc.
must be installed, as malfunction of this product could result in serious damage to the system or injury
to personnel. Also proper periodic maintenance is required.
• This instrument must be used under the conditions and environment described in this manual.
Shinko Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring due tothe instrument being used under conditions not otherwise stated in this manual.
Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the manufacture ofweapons of mass destruction (i.e. military applications, military equipment, etc.), please investigate the end
users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.
11. Model
1.1 Model
Model
DSW-100-CO-A
DSW-100-CO-V

Measurement Range
0 to 250 ppm

Output4 to 20 mA DC
1 to 5 V DC

1.2 How to Read the Model Label
Model label is attached to the inside of the case.
1 : Model

123

2 : Measurement range, Output
3 : Supply voltage, Power consumption, Serial number(Fig. 1.2-1)

2. Mounting
CautionInstallation site should be examined, giving careful consideration to the following conditions.
This sensor is for interior use only. Do not install outside.
[This instrument is intended to be used under the following environmental conditions.]• Free air flow
• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of 0 to 50(32 to 122 ) that does notchange rapidly and no icing• An ambient non-condensing humidity of 5 to 95 %RH• An absence of chlorinated and sulfidizing gases• No large capacity electromagnetic switches or cables through which large current is flowing• No water, oil or chemicals or where the vapors of these substances can come into direct contact withthe unit

Maintenance• Be sure to conduct periodic inspection once a year (once every six months in a highly CO concentratedplace).
• Change the sensor if the measured value does not drop below 30 ppm in air (fresh air).
Also change the sensor when no output occurs.

Inquiries
For any inquiries about this unit, please contact our agency or the vendor where you purchased theunit after checking the model and serial number. Please let us know the details of the malfunction, or
discrepancy, and the operating conditions.
SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, JapanTel:
81-72-727-6100
Fax:
81-72-727-7006
URL:
http://www.shinko-technos.co.jp
2(1) Fixing the Mounting Plate
Screw either (A) or (B), and fix the Mounting Plateto the outlet box, via outlet box cover located insidethe wall.
For screws, use M4 screw (2 pieces) and spring
washer provided.

(3) Mounting Sensor
Screw 3 places as indicated by thedotted lines, and fix the sensor to theMounting Plate.
For screws, use M3 screw (3 pieces)provided.

(A)
(B)

(B)
(A)

(2) Wiring
Connect the sensor lead wire
and the wire from the wall side.

(4) Fitting Sensor CoverInsert the sensor cover protrusion (2 places) into the cover
mounting holes (2 places) located on the upper part of thesensor main unit, then fit the cover to the sensor main unit.

3. Wiring(Fig. 3-1)

Lead Wire Color
BRW: Brown
BLK: Black
RED: RedWHT: White

Lead Wire Type++-Power supply
Output3

4. Specifications
Measurement Range
OutputSupply Voltage
Measurement Method
Mounting
MaterialWiring
External Dimensions

WeightPerformance

0 to 250 ppm
4 to 20 mA DC: Maximum allowable load: 500or less
1 to 5 V DC: Maximum allowable load: 100 k or more (must be specified.)
24 V DC 10 %Semiconductor
To the outlet box (inside wall), via outlet box cover: Mounting dimension: 66.7 mm(JIS-C8340: 1999) (JIS: Japan Industrial Standards.)
Flame-resistant PC resin, Color: White
Lead wire: 300 mm Cross-section: 0.5 mm 2 See “3. Wiring” for details.
W90 x H90 x D32 mm (excluding lead wire)Approx. 100 g
AccuracyStartcharacteristics
Responsecharacteristics

Power Consumption
Operating
Environment

Storage Environment

Insulation ResistanceDielectric Strength
Accessories
EnvironmentalSpecification

0 to 100 ppm:
5 %FS at 23
100 to 250 ppm: 10 %FS at 23
2 hours of warm-up

50 %RH
50 %RHWithin 120 seconds (Varies depending on the air flow ofmeasurement environment.)

6W
Temperature: -10 to 50
Humidity: 10 to 95 %RH (non-condensing)[Caution]
Do not use this sensor in an environment where dew condensation occurs.
Do not use this sensor in an environment where chlorinated and sulfidizing gases
are being generated.
Temperature: -30 to 60
Humidity: 10 to 95 %RH (non-condensing)[Caution]If the sensor is exposed to a temperature lower than 0or higher than (andincluding) 40
for several hours, sensor measurement value will be affected.
If the sensor is stored at high humidity (90 % or more) over a long period without
power supplied, the sensor will deteriorate. If the sensor is left in an environment
where there is no oxygen (only nitrogen), errors will occur in sensor measurement
values.
Between Case - Output: 500 M or more, at 500 V DC
Between Case - Output: 1.5 kV for 1 minute, 3 mA or lessSensor main unit mounting screw: M3 x 4 (3 pieces)
Mounting Plate, Mounting Plate screw: M4 x 8 (2 pieces), Spring washer (2 pieces)
Instruction manual: 1 copy
RoHS directive conformance4