和文は裏面をご覧下さい。

QUICK REFERENCE [Applicable model: SGAU]
Model NameS GAU - □ □□ □ - 0 -□Option0: No option needed, 2: Moisture-proof treatment
Power supply
100 to 240 V AC
Output2A: 2-outputs (a contact) 2B: 2-outputs (b contact)
4A: 4-outputs (a contact) 4B: 4-outputs (b contact)
Input
Series name（For detailed input/output spec, refer to the full Instruction Manual.)

Terminal Arrangement
2-outputs (a contact) 2-outputs (b contact)

4-outputs (a contact) 4-outputs (b contact)

No. SGAU12JE1 2018.10

Characters
Indication
Alphanumeric, °C/°F

A

B

C

DE

FGH

IJKL

M

NO

P

QRS

TU

VW

XY

Z

-10123456789°C°F

Selection Item List
Input group
DC input
Thermocouple input
RTD input
Input typeK-200 to 1370 °C
K -200 to 200 °C
K
0 to 400 °C
J -200 to 1000 °C
J -200 to 200 °C
J
0 to 400 °C
R -50 to 1760 °C
S -50 to 1760 °C
B 0 to 1820 °C
E -200 to 800 °C

T -200 to 400 °C
T -100 to 100 °C
N -200 to 1300 °C
PL- 0 to 1390 °CJPt100-100 to 100 °C

0 to 10 mA(100Ω shunt resistor)

0 to 100 mV(Input resi. 1 MΩ)

4 to 20 mA (Built-in50 Ω shunt resistor)

1 to 5 mA(100Ω shunt resistor)

0 to 1 V(Input resi. 1 MΩ)W5Re/W26Re
0 to 2315 °C

4 to 20 mA(250Ω shunt resistor)

0 to 1 mA(1000Ω shunt resistor)

0 to 5 V(Input resi. 1 MΩ)W3Re/W25Re
0 to 2315 °C

4 to 20 mA(50 Ω shunt resistor)

10 to 50 mA(10Ω shunt resistor)

1 to 5 V(Input resi. 1 MΩ)

Pt100-200 to 650 °C

0 to 20 mA(250Ω shunt resistor)

0 to 10 mV(Input resi. 1 MΩ)-5 to 5 V
(Input resi. 1 MΩ)

Pt100-100 to 100 °C

0 to 16 mA(62.5Ω shunt resistor)

0 to 50 mV(Input resi. 1 MΩ)

0 to 10 V(Input resi. 1 MΩ)JPt100-200 to 500 °C

2 to 10 mA(250Ω shunt resistor)

0 to 60 mV(Input resi. 1 MΩ)-10 to 10 V
(Input resi. 1 MΩ)(Abbreviation: Input resi.: Input resistance)

Energized
Input unit
Save settings
Save
De-energized°C
Not save°F
Set value lock
Decimal point placeAlarm 1 to Alarm 4 type
UnlockNo decimal point
No alarm action
Lock 1
1 digit after point
High limit alarm
Lock 2
2 digits after point
Low limit alarm
Input burnout status
3 digits after point
High limit w/ standby
Overscale
Indication unit
Low limit w/ standby
Underscale
Alarm 1 to Alarm 4
No unitAuto/Manual
Holding/Not Holding%
Auto
Not HoldingmA
ManualHolding
V
Alarm indication Enabled/Disabled
Alarm 1 to Alarm 4°C
Disabled
Energized/De-energized
EnabledKey Operation Flowchart
Default Display
RUN display
mode 1Circuit Configuration
2-outputs (a contact)
132

6

RUN display
mode 2
DISP
key

45
107

112-outputs (b contact)

DISP
key

RUN display
mode 3
DISP
key

Custom display
mode 2DISP
key

RUN display
mode 4Unlit
display mode
DISP
key

3

2

6

DISP
key

5

RUN display
mode 5All unlit
display mode

DISP
key

DISP
key

RUN display
mode 6Model
display mode

DISP
key

DISP
key

10

7

114-outputs (a contact)
91264381011754-outputs (b contact)
91264381075

Manual Mode

Custom display
mode 11411

SET key for 5 sec

DISP
key

Input
settingmode
DISP
keyfor 3 sec

SETkeyAlarm 1 to Alarm 4
setting mode
DISP key for 3 sec

SETkeyInstrument
settingmode
DISP
keyfor 3 sec

Multi-Display
A

DISP
key

DISP
key

Filter time
constant

Multi-Display
B

DISP
key

DISP
key

Sensorcorrection

Savesettings

DISP
key

DISP
key

Input group

DISP
key

DISP
key

Alarm 2
outputInput type

DISP
key

DISP
key

Alarm 3
outputInput unit

DISP
key

DISP
key

Alarm 4
outputDecimal point
placeDISPkey

DISPkey

DISP key

DISPkey

Reverts to the Default Display
by pressing the SET key, orafter Manual mode auto return
time has elapsed.

Indication
value atinput 0%Alarm 1 to Alarm 4
ON delay time

Indication
time

DISP
key

DISP key

DISPkey

RUN display
mode 7
DISP
key

Display Mode•
: Available only for 4-outputs spec.
• For the details of the display mode, refer to the full Instruction Manual.

Default Display• If the SHIFT and DISP keys (in that order) are pressed together for approx. 3
seconds in any display mode, the display mode will become the Default Display.
• Once the Default Display is set, the DISP key will be in lock status.
• If the DISP key is pressed for approx. 3 seconds, the key lock status will becancelled.
• If the DISP key is pressed while the DISP key is in lock status, Multi-Display Awill indicate.

Setting Mode
• Alarm 1 setting mode to Alarm 4 setting mode: Shifts to the next mode bypressing the SET key.
• Alarm 3 setting mode and Alarm 4 setting mode are available for 4-outputs spec.
• If the SET key is pressed for approx. 5 seconds in each setting mode, the unit
will move to the Default Display.

Indication
value atinput 100%

Alarm 1 to Alarm 4 type
DISP key
Alarm 1 to Alarm 4 value
DISP key
Alarm 1 to Alarm 4
upper side hysteresis
DISP key
Alarm 1 to Alarm 4
lower side hysteresis

Alarm 1 to Alarm 4
OFF delay time

DISP
key

DISP key

Indication
unit

Alarm 1 to Alarm 4
Holding/Not Holding

DISP
key

DISP key

Savesettings
DISPkey

Default Display

Alarm 1 to Alarm 4
Energized/De-energizedDISP key
Savesettings
DISP key

Default Display

DISP
keyfor 3 sec

Set valuelock

Alarm 1
outputDOWN
key for3 sec

SETkeyCustom
displaysetting mode

Input
burnout
statusAuto/Manual
DISPkey
Manual mode
auto return
time
DISP
key
Alarm indication
Enabled/Disabled
DISP
key
Savesettings
DISP
key

Default Display

Default Display

SETkey