SPEC SHEETSGR

RTD Transmitter (1-output)
Features:

SGR (1-input 1-output) converts RTD input signal (JP100, JPt100) into
isolated DC voltage/current. Indication displays are equipped.

Functions:Sensor correction (Input value correction)
Normal/Reverse modeFilter time constantManual mode
Input and output types can be changed.
Output high and low limits

Model: SGR - ① ② - ③ - ④

Model name
Input 1Output 1
Power supply
Option③ Power supply① INPUT 1
RTD

P0
P1P2P30Pt100

1JPt100

DC Current
DC Voltage24V AC/DC 50/60Hz
For SGRL (with serial communication function),
only [0: 100 to 240V AC] is selectable.

④ Option② OUTPUT 1
14 to 20mA (allowable load resistance 750Ω max.)

20 to 20mA (allowable load resistance 750Ω max.)※1

30 to 16mA (allowable load resistance 900Ω max.)※1

4

2 to 10mA (allowable load resistance 1500Ω max.)

50 to 10mA (allowable load resistance 1500Ω max.)※1A

0 to 10mV (allowable load resistance 10kΩ min.)※2

B0 to 100mV (allowable load resistance 100kΩ min.)※2

C0 to 1V (allowable load resistance 1000Ω min.)※2D

0 to 5V (allowable load resistance 5000Ω min.)※2E

1 to 5V (allowable load resistance 5000Ω min.)F

0 to 10V (allowable load resistance 10kΩ min.)※2G-5 to 5V (allowable load resistance 10kΩ min.)※1: 0mA or less: Out of base accuracy.
※2: 0V or less: Out of base accuracy.

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/

100 to 240V AC 50/60Hz

0No option needed

1

Multi-rotation trimmer

2

Moisture-proof treatment

3

Multi-rotation trimmer + Moisture-proof treatment

E-mail: overseas@shinko-technos.co.jp

SPEC SHEET
Performance

Base accuracy (at 25℃)
Display accuracy
Temperature coefficient
Effect of allowable lead wire resistance
Response time
Insulation resistanceDielectric strength

General specifications
Input
Zero adjustment range
Power supply
Power consumption
Operating temperature, humidity
Storage temperature
Material
Mounting method
External dimensions
WeightMeasurement range
Input code

±0.1% of each input span
Base accuracy ±1 digit± 0.015%/℃ (0 to 10mV output:±0.02%/℃)
Less than 20Ω per wire: Base accuracy20Ω or more per wire: Base accuracy + 0.005 %/Ω
0.5 sec. max. (0→90%)
100MΩ minimum, at 500V DC
2.0kV AC for 1 minutePt100, JPt100
Input detection current: Approx. 200 μA, Allowable lead wire resistance: 200Ω or less per wire
-5 to 5%(Adjustable from front panel.)
Span adjustment range
95 to 105% (Adjustable from front panel.)
100 to 240V AC 50/60Hz
85 to 264V AC
Allowable voltage range
24V AC/DC
20 to 28 V AC/DC
100 to 240 V AC: Approx. 9 VA max. (SGRL: Approx. 10 VA max.)
24 V AC: Approx. 6 VA max.
24 V DC: Approx. 3 W max.
-10 to 55℃ (Non-condensing, No icing), 35 to 85%RH (Non-condensing)-10 to 60℃
Case: Flame-resistant resin, Black color, Panel: Polycarbonate
DIN rail22.5(W) x 89(H) x 70(D)mm (without socket)
Approx. 77g (without socket)

Terminal arrangement

Measurement range ※2Resolution

P0-200 to 650 ℃-328 to 1202 ˚F

1℃(˚F)

P1-100 to 100 ℃-148 to 212 ˚F

1℃(˚F)※1P2-200 to 500 ℃-328 to 932 ˚F1℃(˚F)

P3-100 to 100 ℃-148 to 212 ˚F1℃(˚F)※1Power supply voltage

PWR ⑬ - ⑭

100 to 240V AC, 24V AC/DC

OUT ⑨ - ⑫Output

RTD ① - ④ - ⑤

RTD input
Serial communication RS-485(For SGRL)

RS-485※1: When ‘1 digit after decimal place’ is selected in [decimal point place], it becomes 0.1.
※2: ℃ or ˚F can be selected in [Input unit].
Please specify the input span. Minimum span is 50℃ or 100˚F.

Circuit configuration(8P Socket)

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20190507