LASER DIODE ANALYZERFeatures:• Integrated Temperature Controller• Compact Plug-and-Play Laser DiodeAnalyzer• Fast and accurate LIV curve generation
with 0.1 mA resolution• 15-bit power resolution• Laser diode drive current up to 500 mA• 3 W output temperature controller• Password protection, hardware interlock
and overcurrent shut-off circuitry for safety• Built-in smart driver for diode pinout auto
detection and temperature control• Touchscreen control and accompanying
Windows software

Applications:• Laser Diode Characterization• Laser Diode Quality Control• Laser Source Calibration

Product Description:
This low cost, compact and standalone laser diode analyzer provides quick and versatile laser diode characterization and analysis. Thelaser diode analyzer is well suited for complete laser diode parametric analysis. The unit features a high resolution LIV (light, current,voltage) measurement system, providing all main laser diode parameters including threshold current and slope efficiency. The setup andmeasurement takes only a few seconds.
The measured parameters are displayed on a color touch screen, and the data can be downloaded to any host computer for further
processing and analysis through a built-in USB interface. The unit features a built-in smart driver that will automatically detect the laser
diode pinout. The user can select to operate the laser diode under test in either auto-current or auto-power control modes. The integrated
TEC controller maintains your diode at its operating temperature, ensuring accurate and repeatable measurements.
The smart laser diode analyzer comes with multiple features for safe and accurate measurements. These features include current limit
protection, automatic pinout detection, sensor range checks and manager passcode level to prevent any accidental changes of thepreset settings that could otherwise harm the laser diode under test, or invalidate the results. External interlocks can be connected todisable the laser diode in response to safety concern situations.
Each unit is provided as a set that includes all required accessories for direct plug-and-play operation. The kit includes a calibrated
optical power meter head and a thermally controlled laser diode holder for 5.6 mm package laser diodes (holders for 3.8 mm and 9 mmpackages are offered separately). A simple sliding mechanism allows easy laser diode replacement. Special laser diode adapters canbe customized upon request for any laser diode package.

DTS0186

OZ Optics reserves the right to change any specifications without prior notice.

09 December 2021

1

LASER DIODE ANALYZER
Part Number: LDA-100
Technical Specifications:Parameters

Units

Values

Setpoint Resolution :

mA

0.1

Setpoint Accuracy:%±0.05%Laser Diode Compliance Voltage:V

0.0–8.0

Photodiode Current Range:

μA0–3600

Photodiode Current Resolution:

μA

0.1

TEC Current Range:

A1

TEC Voltage Range:

V±8.00TEC Output Power:W

3Temperature Control Range:

°C

10-50

Operating Rel.Humidity:%<85%, non-condensing

Analyzer Onlymm(inches)

110x63x30 (4.3x2.5x1.2)

Analyzer +Accessories

mm(inches)

110x115x45 (4.3x4.5x1.8)

Analyzer Onlyg (lb)

300 (0.66)

Analyzer +Accessories

g (lb)

1000 (2.20)

Size

WeightAccessories Included:• Optical Powermeter• Thermally Controlled Laser Head for 5.6 mm laser diode packages• Quick-Change Sliding Baseplate for Powermeter and Laser Head• 12 V DC, 1 A Power Supply• USB Cable• Operating software compatible with Microsoft Windows2