SPEC SHEET

SHT

Thermocouple Transmitter (1-output)
Features:

SHT (1-input 1-output) converts thermocouple signal into isolated
DC voltage/current.

Functions: (Can be changed via the Setting Software) Temperature range
 Sensor correction (Input value correction)
 Normal/Reverse mode
 Filter time constant Input and output types
 Output high and low limits

Model: SHT - ① ② - 0 - ④

Model name
Input 1Output 1Power supply
OptionThermocouple① INPUT 1 ※
K0
K1
K2
J0
J1J2RSBE
T0T1NPLW5W3Power Supply
K0

100 to 240V AC 50/60Hz

J④ Option

RSBET

0No option needed

1Multi-rotation trimmer

2Moisture-proof treatment

3Multi-rotation trimmer + Moisture-proof treatment

NPL-Ⅱ
W5Re/W26Re
W3Re/W25Re※ For the measurement range, refer to the next page.

DC Voltage

DC Current② OUTPUT 1
1
2
345A
BC
DEFG

4 to 20mA (allowable load resistance 750Ω max.)
0 to 20mA (allowable load resistance 750Ω max.)※1
0 to 16mA (allowable load resistance 900Ω max.)※1
2 to 10mA (allowable load resistance 1500Ω max.)
0 to 10mA (allowable load resistance 1500Ω max.)※10 to 10mV (allowable load resistance 10kΩ min.)※20 to 100mV (allowable load resistance 100kΩ min.)※2
0 to 1V (allowable load resistance 1000Ω min.)※2
0 to 5V (allowable load resistance 5000Ω min.)※2
1 to 5V (allowable load resistance 5000Ω min.)0 to 10V (allowable load resistance 10kΩ min.)※2-5 to 5V (allowable load resistance 10kΩ min.)※1: 0mA or less: Out of base accuracy.
※2: 0V or less: Out of base accuracy.

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

SPEC SHEETPerformance
Base accuracy (at 25℃)

±0.1% of each input span
When input is 0℃ or less: Base accuracy ±0.1% of each input span.
When input has a decimal point: Base accuracy ±0.05% of each input span
R, S input, -50 to 200℃ (-58 to 392˚F): ±0.3% of each input span,
B input, 0 to 300℃ (32 to 572˚F): Accuracy is not guaranteed.

Temperature coefficient
Cold junction compensation accuracy
Response time
Insulation resistanceDielectric strength±0.015%/℃ (0 to 10mV output: ±0.02%/℃)±0.5℃ (1.0˚F) at 20±10℃
0.5 sec. max. (0→90%)
100MΩ minimum, at 500V DC
2.0kV AC for 1 minute

General Specifications
Input
Zero adjustment range
Power supply
Power consumption
Operating temperature and humidity
Storage temperature
Material
Mounting method
External dimensions
WeightK, J, R, S, B, E, T, N, PL-Ⅱ, W5Re/W26Re, W3Re/W25Re
External resistance: 100Ω max. (However, B: 40Ω max.)
Span adjustment range95 to 105% (Adjustable from front panel)-5 to 5% (Adjustable from front panel)
Allowable voltage
100 to 240V AC 50/60Hz
85 to 264V ACrange
Approx. 9 VA max.
-10 to 55℃ (Non-condensing, No icing), 35 to 85%RH (Non-condensing)-10 to 60℃
Case: Flame-resistant resin, Black color, Panel: Polycarbonate
DIN rail22.5(W) x 89(H) x 70(D)mm (without socket)
Approx. 76g (without socket)

Measurement Range
Input CodeK0
K1
K2
J0
J1
J2RSBE
T0
T1N
PLW5W3

Measurement Range ※-200 to 1370 ℃-328 to 2498 ˚F-200 to 200 ℃-328 to 392 ˚F
0 to 400 ℃
32 to 752 ˚F-200 to 1000℃-328 to 1832 ˚F-328 to 392 ˚F-200 to 200 ℃
0 to 400 ℃
32 to 752 ˚F-50 to 1760 ℃-58 to 3200 ˚F-50 to 1760 ℃-58 to 3200 ˚F
0 to 1820 ℃
32 to 3308 ˚F-200 to 800 ℃-328 to 1472 ˚F-200 to 400 ℃-328 to 752 ˚F-100 to 100 ℃-148 to 212 ˚F-200 to 1300℃-328 to 2372 ˚F
0 to 1390 ℃
32 to 2534 ˚F
0 to 2315 ℃
32 to 4199 ˚F0 to 2315 ℃
32 to 4199 ˚FTerminal Arrangement
PWR ⑬ - ⑭
OUT ⑨ - ⑫
TC ① - ④
CJA ④ - ⑤

Power supply voltage
100 to 240V ACOutput
Thermocouple input
Cold junction compensation input※ Please specify the input span. Minimum span is 50℃ or 100˚F.

Circuit configuration(8P Socket)

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp