3-90°11610289(取L径)

ﾌﾞ内付ｽﾘｰ

Detectedsignal

Output sensor
RS02E

Signaltransmission

Transmitter
RS02T: (1) Detects the voltage of thermocouples which changes
depending on temperature.
(2)The internal CPU converts the temperature data of (1)into digital signals and transmits the signals to theOutput Sensor.

Transmitter

282m0...300℃

No.T315904B

2017.03.23

PUR、 φ 5mm/4x0.25mm2◆ The inzone signal is a preliminary signal for conﬁrming that the output signal is established within the speciﬁcation range. Please note that it does notguarantee signals output outside the speciﬁcation range.

Cable◆ Please note that the signal may become unstable (false signal or chattering) when the transmission distance and the center oﬀset are outside thespeciﬁcation range.

<= 0.5 sec.

Resolution<= +/- 0.8 % full schale range

Load resistanceLinearity≦ 400 Ω<= 0.04 % full schale range

OutputResponce speed

INZONE (data valid)
4 ... 20 mA × 2 点

LED

Current consumption ≦ 150mA

Transmitter and Output sensor
Operating Temperature 0...+50 degree C
Protection class
IP67
Material
PURCable

Compensation lead wire(JIS) phi 0.9 mm x 2
All heat-resistant vinyl (90 deg. C)

Compensated cold junction ≦± 0.5℃

Thermocouple per J

Measuring temperature range 0...1000℃0...300℃

Applicable thermal sensor Thermocouple per K(1CH, 2CH)

24V DC +/- 5 % (include ripple)

Supply voltage± 8mm (Setting metal shaft)Applicable Output sensor RS02E-R01E-PU-̲̲

2signals

L=Cable length
The notation in meters to
the end of the model･･･PU-01 ⇒ 1m

Input channel

RS02E-R01E-PU-̲̲

Output Sensor9(取径er)ametneｰrﾌdiﾞ内付e ｽinﾘSleev

LED

14

Center oﬀ-set

Type code

113

55

5044-

Output sensor：RS02E-R01E-PU-̲ ̲

Output Sensor: Change the temperature data to analog signal (4...20mA)
and output to external unit and supplies power for
operation of Transmitter at the same time.

: Two thermocouples type K or J are used as a detector
and it detects temperature.

4...20mA

Hostdevice（24 VDC)PowerDetector【Function of each component】

Thernocouple

Power supply[Output sensor]

Power supply[Transmitter]

Rated transmitting distance 0...8 mm（Setting metal shaft)RS02T-R01-K1000 RS02T-R01-K300 RS02T-R01-J300

Transmitter

11355504-4-Thernocouple

6A p p l i c a b l e RS02T-R01-K1000、RS02T-R01-K300
Transmitter
RS02T-R01-J300

Type code

Speciﬁcation of the System

356914

Transmitter： RS02T-R01-K1000、RS02T-R01-K300
RS02T-R01-J300

Dimension◆ To avoid malfunction caused by induction noise, cableshould be kept apart from motor or other power cable.

◆ Ensure correct connections by reference to the wiring diagram.

◆ Use a regulated power supply, e.g. switch-model type.
Simpler power supplies, such as a full-wave rectiﬁcation type,will cause the permissible ripple rating to be exceed and may
cause malfunction.

◆ Ensure the power is switched oﬀ during installation or
maintenance operations.

Before using the Remote Sensor, read this manual carefully.
During installation and operation, pay close attention to the safety aspect.

Attention for Installation(Read this section thoroughly before installation.)

RS02T-R01-K300
RS02T-R01-J300

Output Sensor : RS02E-R01E-PU
Transmitter:
RS02T-R01-K1000

Remote Sensor system /Analog signalThermocouple / 2 signal transmission

13.5[Detector]

10[Transmitter]㻼㻸㻯㻌㻌㼛㼞㻌
㻼㼍㼚㼑㼘㻌㼙㼑㼠㼑㼞㻌㻻㼁㼀㻝㼍㼚㼐㻌㻞㻌㼍㼞㼑㻌㻠䈈㻞㻜㼙㻭㻌㼏㼡㼞㼞㼑㼚㼠㻌㼟㼛㼡㼞㼏㼑㻚㻻㼡㼠㼜㼡㼠㻌㼟㼑㼚㼟㼛㼞㻌㻮㼞㼛㼣㼚㻔㻗㻞㻠㼂㻕㻌㻮㼘㼍㼏㼗㻔㻻㼁㼀㻝㻕㻌㼃㼔㼕㼠㼑㻔㻻㼁㼀㻞㻕㻌㻌㻮㼘㼡㼑㻔㻳㻺㻰㻕㻌[Output Sensor]-20 -15 -10 -5

Y0181614121086425 10 15 20mm

X

X

YSetting metal shaft

X:Center offset(mm)Y:Operating distance(mm)RS02T-R01-K1000 / RS02E-R01E-PU-̲ ̲ , Xxxx

Typical Transmitting Diagram (Supply voltage at 24V /non-ﬂush mount）- Output is current source , therefore please connect the load
between output and GND.

- For a detector, please use a thermocouple K that meets JIS.
- The measurement temperature should be lower than upper
limit of the temperature shown at the code end.
example : RS02T-R01-K300 ： 0...300 degree C【Causion】- Measure to static electricity
When using RS02 in environment with the electrostatic
discharge, take the ground to prevent infuluence by the staticelectricity on RS02.

Cable length
Transmitter (RS02T-R01-K̲̲̲̲、RS02T-R01-J̲̲̲̲) ：max.3m
Output sensor (RS02E-R01E-PU)：max.10m

㻵㻺㻞㻌㻔㻗㻕㻌 㼞㼑㼐㻌Thermocoupl 㻔㻙㻕㻌 㼣㼔㼕㼠㼑㻌㻵㻺㻝㻌㼀㼞㼍㼚㼟㼙㼕㼠㼠㼑㼞㻌㼀㼔㼑㼞㼙㼛㼏㼛㼡㼜㼘㼑㻌 㻔㻗㻕㻌 㼞㼑㼐㻌㻔㻙㻕㻌 㼣㼔㼕㼠㼑㻌[Detector]

Wiring diagram

Internal
circuit

Construction of the system

3-90°356913.5

Internalcircuit

Remote System
User’s Guide

1150mm35100(mm)300

C

Web：http://www.b-plus-kk.com(Former NIHON BALLUFF co., Ltd.)BAC■ Parallel installation

Mail ：b-plus-usa@b-plus-kk.com

Wireless Power Supply byType code
RS02T-R01-K1000
RS02T-R01-K300
RS02T-R01-J300
RS02E-R01E-PU-̲ ̲

BA■ Surrounding metal

In order to avoid inﬂuence of surrounding metal, or to avoid mutualinﬂuence between parallel-mounted sensors, keep the minimum freezone as described below.

Installation notes＊ Never pull the cable strongin installing

The minimum
bendingradius forthesensors
are 50mm.

Bending radius of Cable

Output current
0 mA
４ ... 20 mA
Receiving the temperature（at 0 deg. C : 4mA, at the hightest temperature
data from Transmitterof the measurement range : 20 mA)
Thermocouple is notconnected, though the21 mA
transmitter is in thetransmitting range.

statusWithout Transmitter

Output sensor RS02E outputs the electric current as described below.

Current output