Output Sensor : RS02E-030E-PU

Redtype KRedWhiteWhiteTransmitter
RS02T-030-K300

Compensation lead wire(JIS) phi 0.9 mm x 2
All heat-resistant vinyl (90 deg. C)

13.5

2017. 03. 23

No.T315901B- The inzone signal is a preliminary signal for conﬁrming that the output
signal is established within the specification range. Please note that it
does not guarantee signals output outside the speciﬁcation range.

- Output is current source , therefore please connect the load
between output and GND.
- Please note that the signal may become unstable (false signal orchattering) when the transmission distance and the center oﬀset areoutside the speciﬁcation range.

- The measurement temperature should be lower than upper limitof the temperature shown at the code end.
example : RS02T-030-K300 ： 0...300 degree C

Hostdevice: (1) Detects the voltage of thermocouples which changes
depending on temperature.
(2)The internal CPU converts the temperature data of (1)into digital signals and transmits the signals to theOutput Sensor.

: Two thermocouples type K are used as a detector and itdetects temperature.

4...20mA

66

Output Sensor
RS02E-030E-PU-̲̲

83.595LRedBlackWhiteBlue

L=Cable length
The notation in meters
to the end of the model･･･PU-01 ⇒ 1mTransmitter and Output sensor
Operating Temperature
0...+80 degree C
Protection class
IP67
Housing
Nickel plated brass
Material
Active surface Nylon12

Applicable Transmitter For thermocouple K :RS02T-030-K300
24V DC +/- 5 % (include ripple)
Supply voltage
Current consumption <= 150 mA
LED
INZONE (data valid)
Output
4 ... 20 mA × 2 ch (current source)
Load resistance<= 400 Ohm
Resolution<= 0.04 % full schale range
Responce speed<= 0.5 sec.
Linearity<= +/- 0.8 % full schale range
Cable
PUR, Phi 5 mm/4x0.25 mm2Type code

13.55Output sensor：RS02E-030E-PU-̲ ̲

RS02T-030-K300：outer sheath color Blue (VX-G:7/0.3x2 K)- For a detector, please use a thermocouple K that meets JIS.

Cable

74

50.52 Compensation lead wire, outer diameter 3.2 x 5.1 mm
2 conductor, phi 0.9 mm

3005Signaltransmission

Output sensor
RS02EPower（24 VDC)

Output Sensor: Change the temperature data to analog signal (4...20mA)
and output to external unit and supplies power for
operation of Transmitter at the same time.

TransmitterDetector

Transmitter： RS02T-030-K300

Applicable Output sensor RS02E-030E-PU-̲̲
Rated transmitting distance 2 ... 6 mm+/- 4 mmCenter oﬀ-set
Input channel
2 (IN 1, IN 2)
Applicable thermal sensor Thermocouple per K
Measuring temperature range 0...300 deg.C.
Compensated cold junction +/- 0.5 deg. C

Type code

Speciﬁcation

M30×1.536Dimension◆ To avoid malfunction caused by induction noise, cableshould be kept apart from motor or other power cable.

◆ Ensure correct connections by reference to the wiring diagram.

◆ Use a regulated power supply, e.g. switch-model type.
Simpler power supplies, such as a full-wave rectiﬁcation type,will cause the permissible ripple rating to be exceed and may
cause malfunction.

◆ Ensure the power is switched oﬀ during installation or
maintenance operations.

Detectedsignal

Transmitter
RS02T

Power supply[Output sensor]

Power supply[Transmitter]【Function of each component】

Thernocouple

Transmitter: RS02T-030-K300 (type-K)

Attention for Installation(Read this section thoroughly before installation.)

Thernocouple

Remote Sensor system /Analog signal
Thermocouple / 2 signal transmission[Detector]

Construction of the system[Transmitter]㻼㻸㻯㻌㻌㼛㼞㻌㻼㼍㼚㼑㼘㻌㼙㼑㼠㼑㼞㻌㻻㼁㼀㻝㼍㼚㼐㻌㻞㻌㼍㼞㼑㻌㻠䈈㻞㻜㼙㻭㻌㼏㼡㼞㼞㼑㼚㼠㻌㼟㼛㼡㼞㼏㼑㻚㻻㼡㼠㼜㼡㼠㻌㼟㼑㼚㼟㼛㼞㻌㻮㼞㼛㼣㼚㻔㻗㻞㻠㼂㻕㻌㻮㼘㼍㼏㼗㻔㻻㼁㼀㻝㻕㻌㼃㼔㼕㼠㼑㻔㻻㼁㼀㻞㻕㻌㻌㻮㼘㼡㼑㻔㻳㻺㻰㻕㻌[Output Sensor]

Output current

234567mm

1X
10 8 6 4 2 0 2 4 6 8 10Y:;

Rated transmitting range

X:Center oﬀ-set(mm)Y:Transmitting distance(mm)【Example: Supply voltage at 24V DC】

Typical Transmitting Diagram

0 mA
４ ... 20 mA
Receiving the temperature（at 0 deg. C : 4mA, at the hightest temperature
data from Transmitterof the measurement range : 20 mA)
Thermocouple is notconnected, though the21 mA
transmitter is in thetransmitting range.

statusWithout Transmitter

Output sensor RS02E outputs the electric current as described below.

Current output- Measure to static electricity
When using RS02 in environment with the electrostatic discharge,
take the ground to prevent infuluence by the static electricity on
RS02.

【Causion】- Measure to surge
Please note that the cable length of an output sensor may not longer than 10m. The CE marking veriﬁes that our products comply
with the requirements of EMC directive. The surge test to an output sensor is not carried out.
When using RTT or RTE with cable length longer than10 m a measure to protect the sensor from surge current should be taken.

Cable length
Transmitter (RS02T-030-K300) ：max.3m
Output sensor (RS02E-030E-PU)：max.10m

㻵㻺㻞㻌㻔㻗㻕㻌 㼞㼑㼐㻌
Thermocoupl 㻔㻙㻕㻌 㼣㼔㼕㼠㼑㻌㻵㻺㻝㻌㼀㼞㼍㼚㼟㼙㼕㼠㼠㼑㼞㻌㼀㼔㼑㼞㼙㼛㼏㼛㼡㼜㼘㼑㻌 㻔㻗㻕㻌 㼞㼑㼐㻌㻔㻙㻕㻌 㼣㼔㼕㼠㼑㻌[Detector]

Wiring diagram

Internalcircuit

Remote SystemUser’s Guide

Before using the Remote Sensor, read this manual carefully.
During installation and operation, pay close attention to the safety aspect.

41.5

Internalcircuit

20200＊ Never pull the cablestrongly in installing.

50mm

The minimum bending radius
for Output sensor is 50mm.

30A(mm) B(mm) C(mm)C■ Parallel installation

E-mail b-plus@b-plus-kk.jp(Former NIHON BALLUFF co., Ltd.)

http://www.b-plus-kk.jp/

20Nm

Tightening troque forattached nut is20Nm(200kgf･cm).

Mutual interference

型式
RS02T-030-K300
RS02E-030E-PU

BA■ Surrounding metal

free zone as described below.

In order to avoid inﬂuence of surrounding metal, or to avoid mutualinﬂuence.between parallel-mounted sensors, provide the minimum

Inﬂuence of surrounding metal