Hybrid Recorder

HR-700 series

More or Less?
How about both!

More: Bigger display

144

18mm LED allows a high level of visibility

144

175

Less: Overall size
Compact, lightweight:
Ideal for small scale control panels

Compact: 150mm in depth, 1.5kg in weight■ Display and operation keys①

Available for small panel mounting②③

Communication interface
Corresponds to FA (factory automation) system via communication interface,
RS-232C (standard) or RS-485 (option)

Larger LED display
LED size: 18mm in height

Dust-proof and Drip-proof (IEC60529 IP65)
Stands up to even harsh environments such as food related plants and kilns

Safety standard⑤④① : Channel number display

UL/C-UL and CE marking

Orange LED indicates Channel numbers 1 to 6.

② : Alarm indicator■ ModelHR-70 □ ，□□□ HR-700 (W144 H144 D150mm)
1
1-point (Pen)
Multi-range system
Input point 2
2-point (Pen)(Input types are selectable.)
6
6-point (Dot printing)
C5
Communication function (RS-485)
RE1
DI function (Pen)
RE6
DI function (Dot printing)
OptionFL
Paper-empty detection function
LH3
Alarm output function (Pen)
LH6
Alarm output function (Dot printing)• When ordering, select the alphanumeric characters from the table above for□.
• When adding options, enter the code using a “comma”.

Red LED indicates the type of the alarm.
[H] is lit when the alarm is High, and [L] is lit
when the alarm is Low. Neither of them is lit when
there is no alarm.

③ : Data display
Indicates the process variable, date and year, chart feed speed
or alarm value (orange).

④ : Status indicators
The [REC] (orange) is lit when recording. The [ALM] (red) is lit
when the alarm is being activated.

⑤ : Operation keys
Use these keys for setting and other operations.

■ Rated scale and accuracy
Range

Input

Thermocouple

RTD

DC voltage

DC current

K1
K2
K3

-200.0 to1370.0℃-200.0 to 600.0℃-200.0 to 300.0℃

E1
E2
E3
J1
J2
J3
T1
T2

-200.0 to 800.0℃-200.0 to 300.0℃-200.0 to 150.0℃-200.0 to 1100.0℃-200.0 to 400.0℃-200.0 to 200.0℃-200.0 to 400.0℃-200.0 to 200.0℃

R1
R2S
B

0.0 to 1760.0℃
0.0 to 1200.0℃
0.0 to 1760.0℃
0.0 to 1820.0℃

NC0.0 to 1300.0℃
0.0 to 2320.0℃

PR40-20

0 to 1880℃

Au-Fe
PL-ⅡUL
Pt100 1
Pt100 2
JPt100 1
JPt100 2

0 to 300 K-100 to 1390℃-200.0 to 400.0℃-200.0 to 900.0℃-200.0 to 650.0℃-200.0 to 200.0℃-200.0 to 630.0℃-200.0 to 200.0℃-10 to 10mV
0 to 20mV
0 to 50mV-200 to 200mV-1 to 1V
-10 to 10V
0 to 5V
4 to 20mA

Measurement (Digital display)
Measurement accuracy
K1: ± (0.15% of rdg＋0.7℃)
K2: ± (0.15% of rdg＋0.4℃)
K3: ± (0.15% of rdg＋0.3℃)
However, Range -200 to100℃, ± (0.15% of rdg＋1℃)± (0.15% of rdg＋0.5℃)± (0.15% of rdg＋0.4℃)± (0.15% of rdg＋0.3℃)
J1, T1 : ± (0.15% of rdg＋0.5℃)
J2, T2 : ± (0.15% of rdg＋0.4℃)
J3: ± (0.15% of rdg＋0.3℃)
However, Range -200 to100℃, ± (0.15% of rdg＋0.7℃)
R1, S, B : ± (0.15% of rdg＋1℃)
R2: ± (0.15% of rdg＋0.8℃)
However, R1, R2, S: Range 0 to 100℃, ± 3.7℃Range 100 to 300℃, ± 1.5℃
B: Range 400 to 600℃, ± 2℃
(Accuracy is not guaranteed below 400%)± (0.15% of rdg＋0.7℃)± (0.15% of rdg＋1℃)± (0.15% of rdg＋1℃)
However, Range 0 to 300℃, ± 4.0℃
Range 300 to 800℃, ± 3.0℃
± (0.15% of rdg＋1K)± (0.15% of rdg＋0.7℃)± (0.15% of rdg＋0.5℃)
However, Range -200 to100℃: ± (0.15% of rdg + 0.7℃)± (0.15% of rdg＋0.3℃)± (0.15% of rdg＋0.2℃)± (0.15% of rdg＋0.3℃)± (0.15% of rdg＋0.2℃)± (0.2% of rdg＋3 digits)± (0.2% of rdg＋3 digits)± (0.2% of rdg＋2 digits)± (0.2% of rdg＋3 digits)± (0.1% of rdg＋3 digits)± (0.3% of rdg＋3 digits)± (0.2% of rdg＋2 digits)± (0.2% of rdg＋2 digits)• In the case of DC current input, connect a shunt resistor (sold separately).

Resolution

Recording (analog)
Recording accuracy

0.1℃

Measurement accuracy:± (0.3% of recording span)

0.1K
0.1℃

0.1℃

10μ V
10 μ V1mV
10mV
1mV
0.01mA■ Standard specifications
Measuring point

Input

Display

Performance

Recording

Alarm

Communications
Supply voltage
Power consumption
Insulation resistanceDielectric strength
Operating environment
Safety standard
Material/Color
Door
Mounting
WeightAttached functions

Accessories

When ordering, one type can be selected from the following.
1-point (Pen): HR-701, 2-point (Pen): HR-702, 6-point (Dot printing): HR-706
Multi-range (Types are selectable.)• Thermocouple --------- K, J, R, S, B, E, T, C (W/Re5-26), N, PL-Ⅱ , PR40-20, Au-Fe, U, L• RTD ---------------------- Pt100, JPt100• DC voltage ------------- ±10mV DC, 0 to 20mV DC, 0 to 50mV DC, ±200mV DC, ±1V DC, ±10V DC, 0 to 5V DC• DC current -------------- 4 to 20mA DC (Connect external 250Ω shunt resistor which is sold separately.)
Scale ----------------------- Refer to the rated scale.
Input resistance --------- Thermocouple, DC voltage (mV) range without burnout alarm : 10MΩ or more
Thermocouple, DC voltage (mV) range with burnout alarm: 200kΩ or more
DC voltage (V) : 1MΩ or more
DC current (mA) : 250Ω (External shunt resistor required)
Allowable signal source resistance ---- Thermocouple, DC voltage (mV) range without burnout alarm : 10kΩ or less.
Thermocouple, DC voltage (mV) range with burnout alarm: 100Ω or less.
DC voltage (V) : 1kΩ or less
RTD: 10Ω or less per wire
Indication -------- Digital indication, 7-segment orange, LED 6 digits (Channel No. display: 1 digit, Data display: 5 digits)
Contents--------- Channel No., Process variable, Date and year, Chart feed speed, Alarm value
Others------------ REC: Lights while recording
ALM: Red LED lights when alarm occurs in any channel.
Digital accuracy------------------------- ±0.2% ±1digit or less (Within the measurement range of mV and V input. TC and RTD are excluded.)
Recording accuracy ------------------- Measurement accuracy ±(0.3% of recording span)
Dead band ------------------------------- Within 0.2%
Normal mode rejection rate --------- 60dB or more (50/60±0.1Hz)
Common mode rejection rate ------- 140dB or more (50/60±0.1Hz)
Interchannel maximum noise voltage------ 200V AC 50/60Hz
Vibration resistance ------------------- 10 to 60Hz 1m/s2
Shock resistance ----------------------- 2m/s2 or less
Clock accuracy ------------------------- ±50ppm or lessChart paper ------------------ Width: 100mm, Length: 16m, Weight: 83g per stack
Recording method---------- Pen: Disposable felt pen (1-pen: Red, 2-pen: Green)
Dot printing : Wire dot (6-color ink ribbon)(No.1: Purple, No.2: Red, No.3: Green, No.4: Blue, No.5: Brown, No.6: Black): Wire dot (1-color ink ribbon)
Printing method ------------- Pen
Dot printing : Wire dot (6-color ink ribbon)
Step response time -------- Pen: 1 second or less (IEC1143, 95% response): Continuous recording for each channel
Recording period ----------- Pen
Dot printing : 10 seconds (Selectable from 10sec, 20sec, 30sec, 60sec): Selectable from 46 types of speed by front key within the range of 5 to 12000mm/h
Chart speed ------------------ Pen
Dot printing : Selectable from 34 types of speed by front key within the range of 0 to 1500mm/h
Chart feed accuracy ------- Within ± 0.1% (Does not include expansion or shrink of paper, when it is fed 1000mm or more.)
Alarm output is not available. (ALM is lit.)
Specify the alarm option(LH3, LH6) when alarm output is required.
Setting points----------- Maximum 4 points (2 points for High limit and low limit each) can be set per channel.
Hysteresis width ------- 0.5%
Setting accuracy------- The same as Digital display accuracy
RS-232C: 1200/2400/4800/9600bps
100 to 240V AC, 50/60Hz, Allowable voltage fluctuation----- 85 to 264V AC
1-pen: Approx.25VA, 2-pen: Approx.28VA, 6-dot: Approx. 25VA
Between each terminal and ground: 20MΩ or more, at 500V DC
1.5kV AC for 1min between power terminal and ground terminal
500V AC for 1min between input terminal and ground terminal
200V AC for 1min between input terminal and input terminalTemperature: 0 to 50℃, Humidity: 20 to 80% RH
UL: Power input rating 100-240V AC File No. E195801
Case: Flame-resistant resin (Black) Door: Flame-resistant resin (Transparent)
Dust-proof and Drip-proof (IEC60529 IP65)
Panel mounting (Vertical panel), Allowable inclination angle: Backward 0 to 30° or less
HR-701: Approx. 2kg, HR-702: Approx. 2.5kg, HR-706: Approx. 1.5kg
Skipping, Servo-stop, Self-diagnosis, Zone recording, Partial compression/expansion recording, Decade recording and indication,Tag number setting, Copy function, Input offset setting, Computation, Interchannel computation, Scaling, Burnout,
Memory back up (Clock function is protected by the internal lithium battery. [Battery life: Approx. 10 years] Setting/Corrected data
is protected by non-volatile memory.), Asynchronous print mode, Printer gap correction function
Chart paper: 1 volume, Ribbon cassette (Dot printing: 1, Pen: 1),
Cartridge pen (1-pen: 1, 2-pen: 2), Packing: 1, Mounting brackets: 1 set, Instruction manual: 1 copy

■ Optional specifications
Communication
function [C5]Communication line: RS-485, Communication speed: 1200/2400/4800/9600bps

Settable at 3 points (Maximum)
Chart feed Start/Stop : Relay contact ON: Start Relay contact OFF: StopDI function
Changing chart speed : Changes 1st with tuning to ON and 2nd with OFF[RE1] [RE6]Comment printing: Prints comments with contact ON (Up to16 characters per line )Log printing: Prints with contact ON
Date and time printing : Prints date and time with contact ON
Paper-empty
Detects the paper tray is empty, stops recording, and activates the alarm.
detection function [FL] (When adding this option, [LH3] or [LH6] option needs to be added.)
Output number• Pen: 3 points (Built-in option, a contact)
Alarm output function • Dot printing: 6 points (Built-in option, a contact)[LH3] [LH6]
Contact capacity : 250V AC Maximum 3A (Resistive load)
30V DC Maximum 3A (Resistive load)
125V DC Maximum 0.5A (Resistive load)■ Terminal arrangement① : Alarm output terminal (Option)
Alarm1

Alarm 2

Alarm 3

Alarm 4

Alarm 5

Alarm 6③④①

② : Input terminal
Channel 4

Channel 5

Channel 6

Channel 1

Channel 2

Channel 3⑤②⑥④ : RS-485 Communication
terminal (Option)③ : Power supply terminal⑤ : DI Function terminal(Option)⑥ : RS-232C Communicationterminal

100 to 240V AC（50／60Hz）■ Panel cutout (Scale: mm)■ External dimensions (Scale: mm)
2.5

144

3.525150200

138＋10200137

144

Maximum 7(Panel thickness )

138

＋1
0

Allowable inclination angle: Backward 0 to 30° or less！

SAFETY
PRECAUTIONS● To ensure safe and correct use, thoroughly read and understand the manual before using this instrument.
● This instrument is intended to be used for industrial machinery, machine tools and measuring equipment. Verify
correct usage after purpose-of-use consultation with our agency or main office.
(Never use this instrument for medical purposes with which human lives are involved.)● External protection devices such as protection equipment against excessive temperature rise, etc. must be installed,
as malfunction of this product could result in serious damage to the system or injury to personnel. Also proper
periodic maintenance is required.
● This instrument must be used under the conditions and environment described in the manual. Shinko Technos Co.,
Ltd. does not accept liability for any injury, loss of life or damage occurring due to the instrument being used underconditions not otherwise stated in this manual.

Caution with respect toExport Trade Control Ordinance
To avoid this instrument from being used as acomponent in, or as being utilized in themanufacture of weapons of mass destruction(i.e. military applications, military equipment,etc.), please investigate the end users andthe final use of this instrument.
In the case of resale, ensure that thisinstrument is not illegally exported.

・This catalog is as of June 2021, and specifications are subject to change without notice.
・The photos in this catalog do not show actual usage.
・If you have any inquiries, please consult us or our agency.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISIONOﬃce : 2-5-1, Senbahigashi, Minoo, osaka, 562-0035, JapanTel: 81-72-727-6100

Fax: 81-72-727-7006

URL: https://shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

202106 E10