HR-706 Operation Flow Chart
CONTENTS
Normal Operation Mode : Recording, Changing the Display, Engineering ListPrinting, Manual Printing, List Printing....................................... 3
Setup Mode : Setting the Range (Current/Voltage, Thermocouple, Resistance

Setup Mode : Setting Other Functions(Comments) ............................................ 18
Engineering Mode : Alarm Hysteresis, Burnout ON/OFF .................................... 19
Engineering Mode : Channel Offset .................................................................... 20

temperature detector) ................................................................ 4

Engineering Mode : Reference Junction Compensation ..................................... 21

Setup Mode : Setting the Range (scaling) ............................................................. 5

Engineering Mode : Changing the Printing Color ................................................ 22

Setup Mode : Setting the Range (Square Root) .................................................... 6

Engineering Mode : Recording RUN/STOP Trigger Setting, Channel/Tag Printing

Setup Mode : Setting the Range (Decade) ............................................................ 7

Selection, Alarm printing ON/OFF........................................... 23

Setup Mode : Setting the Range (Difference, Sum, Average) ............................... 8Engineering Mode : Logging printing ON/OFF, Scale printing ON/OFF,Setup Mode : Setting the Range (Skip) ................................................................. 9Setup Mode : Setting the Alarm ........................................................................... 10

Synchronous/Asynchronous ................................................... 24
Engineering Mode : Setting the Communication Function, Initializing the Setup

Setup Mode : Setting the Unit.............................................................................. 11

Data ........................................................................................ 25

Setup Mode : Setting the Chart Feed Speed, Setting the Date and Time ........... 12Engineering Mode : DI Functions ........................................................................ 26

Setup Mode : Copying the Setting Data .............................................................. 13

Engineering Mode : Temperature Unit Selection, Point Calibration..................... 27

Setup Mode : Setting Other Functions(Printing Cycle, Zone Recording) ............. 14

Engineering Mode : Data Calibration(Voltage) .................................................... 28

Setup Mode : Setting Other Functions(Partial Compression/Expansion) ............ 15

Engineering Mode : Data Calibration (Resistance temperature detector) ........... 29

Setup Mode : Setting Other Functions(Digital Printing) ....................................... 16

Engineering Mode : Data Calibration (Terminal Temperature) ............................ 30

Setup Mode : Setting Other Functions(Tag) ........................................................ 17

Default value................................................................................................ 31

WARNING
When calibrating data, connect the equipment according to Section 9.5 of the instruction manual. Data calibration must be performed correctly.
Operators possessing knowledge of electricity and control related matters should perform the calibration after careful reading of Sections 7.2 and 9.5 of theinstruction manual.
(Shinko Technos Co., Ltd. is not liable for damages incurred if calibration is performed other than prescribed in the instruction manual.)

Normal Operation Mode : Recording, Changing the Display, Engineering List Printing, Manual Printing, List Printing－3－Setup Mode : Setting the Range (Current/Voltage, Thermocouple, Resistance temperature detector)－4－Setup Mode : Setting the Range (scaling)－5－

Setup Mode : Setting the Range (Square Root)－6－

Setup Mode : Setting the Range (Decade)－7－

Setup Mode : Setting the Range (Difference, Sum, Average)－8－

Setup Mode : Setting the Range (Skip)－9－

Setup Mode : Setting the Alarm－10－

Setup Mode : Setting the Unit

－11－

Setup Mode : Setting the Chart Feed Speed, Setting the Date and Time

－12－

Setup Mode : Copying the Setting Data

－13－

Setup Mode : Setting Other Functions(Printing Cycle, Zone Recording)－14－

Setup Mode : Setting Other Functions(Partial Compression/Expansion)－15－

Setup Mode : Setting Other Functions(Digital Printing)－16－

Setup Mode : Setting Other Functions(Tag)－17－Setup Mode : Setting Other Functions(Comments)－18－

Engineering Mode : Alarm Hysteresis, Burnout ON/OFF－19－

Engineering Mode : Channel Offset－20－

Engineering Mode : Reference Junction Compensation－21－

Engineering Mode : Changing the Printing Color－22－

Engineering Mode : Recording RUN/STOP Trigger Setting, Channel/Tag Printing Selection, Alarm printing ON/OFF－23－

Engineering Mode : Logging printing ON/OFF, Scale printing ON/OFF, Synchronous/Asynchronous－24－

Engineering Mode : Setting the Communication Function, Initializing the Setup Data

－25－

Engineering Mode : DI Functions－26－

Engineering Mode : Temperature Unit Selection, Point Calibration－27－

Engineering Mode : Data Calibration(Voltage)

WARNING
When calibrating data, connect the equipment according to Section 9.5 of the instruction manual.
Data calibration must be performed correctly.
Operators possessing knowledge of electricity and control related matters should perform thecalibration after careful reading of Sections 7.2 and 9.5 of the instruction manual.
(Shinko Technos Co., Ltd. is not liable for damages incurred if calibration is performed other than prescribed in theinstruction manual.)－28－

Engineering Mode : Data Calibration (Resistance temperature detector)

WARNING
When calibrating data, connect the equipment according to Section 9.5 of the instruction manual.
Data calibration must be performed correctly.
Operators possessing knowledge of electricity and control related matters should perform thecalibration after careful reading of Sections 7.2 and 9.5 of the instruction manual.
(Shinko Technos Co., Ltd. is not liable for damages incurred if calibration is performed other than prescribed in theinstruction manual.)－29－

Engineering Mode : Data Calibration (Terminal Temperature)

WARNING
When calibrating data, connect the equipment according to Section 9.5 of the instruction manual.
Data calibration must be performed correctly.
Operators possessing knowledge of electricity and control related matters should perform thecalibration after careful reading of Sections 7.2 and 9.5 of the instruction manual.
(Shinko Technos Co., Ltd. is not liable for damages incurred if calibration is performed other than prescribed in theinstruction manual.)－30－

Default value
Setup mode

Engineering mode

Display

Setting items
Range(all channels)
Alarm(all channels)
Engineering unit(all channels)
Chart feed speed

Time

Default value

Remarks

Display－10～10mV
0～100.0

Setting items

Default value

Alarm hysteresis

ON(0.5%)

Burnout(all channels)

OFF

Channel offset(all channels)

0.0

RJC (all channels)

Internal
compensation INT

Dot printing color(Ch1 to Ch6)

Purple(1), Red(2),Green(3), Blue(4),Brown(5), Black(6)Remarks

All levels alarm OFF,
Relay OFF℃
(1)20mm/h(2)20mm/h
2000/01/01 00:00

The present time has been set.
(Japan standard time)
GMT + 09:00

Setting other recordings
Dot point interval10(sec)Zone setting(all channels)

0～100(%)

Partialcompression/expansion OFF(all channels)
Tag printing character(all channels)

Blank up to the 7thcharacter

Comment printing word
(1 to 3)

Blank up to the 16thcharacter

REC (recording)
RUN/STOP trigger

INT

Channel/Tag printing

Ch

Alarm printing

OFF

Logging printing

ONLogging printing
reference time

00:00

Logging printing interval 6H
Logging scale printing

OFFDI function (Ch1 to Ch3) OFF－31－

Use the“REC” key

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office : 2-5-1, Senbahigashi, Minoo, Osaka, Japan
http://www.shinko-technos.co.jp/e/
URL:Tel : +81-72-727-6100
E-mail:overseas@shinko-technos.co.jp
Fax: +81-72-727-7006

No. HR7ZE1FLOW

2021.02