HYBRID RECORDER

HR-700(PEN WRITING)
INSTRUCTION MANUALFor safe use
Thank you for purchasing our HR-701 (HR-702) Hybrid Recorder.
In order for this instrument to function effectively and correctly, please read and understand this
instruction manual thoroughly before using the instrument.
This instrument should be used in accordance with the instructions described in the manual. If it is not usedaccording to the instructions, it may malfunction or its protection function will be compromised.
The symbols below are used on this instrument for caution information.

Symbols used on the instrument
This indicates “Caution”. Please refer to this manual when handling parts with thissymbol for protection of both the human body and instrument.
This indicates “Protective grounding”.
Be sure to provide protective grounding prior to operation of this instrument.
This indicates “Risk of electric shock”. This symbol is used on the parts which have arisk of electric shock.

The symbols below are used on the warning and cautioning information in this manual.

Symbols used in this manual

WARNING

Failure to observe this information could result in death or injury.
Be sure to read this.

CAUTION

Failure to observe this information could damage the instrument.
Be sure to read this.

[Note]

This is cautionary information for correct use of the instrument.
Be sure to read this.

[Reference]

This is information about the functions of this instrument for effective use.

Be sure to observe the following warnings/cautions and those written in thismanual in order to ensure safety when using this instrument.

WARNING
GeneralProtective Grounding

Power Source

Working Environment

Input and Output Wiring(1) In order to prevent electric shock, be sure to disconnect this instrument fromthe main power source when wiring it.
(2) When this instrument is incorprated in a system, and if the system is expected
to be affected by malfunction from the recorder or external factor, takepreventive measures to secure the safety of the system.
(3) Take measures so that foreign objects cannot enter the opening section.
(1) In order to prevent electric shock, be sure to provide protective grounding prior
to turning on this instrument.
(2) Do not cut protective grounding conductors or disconnect protective grounding.
(1) Make sure that the supply voltage for this instrument conforms to the voltageof the supply source.
Rated power supply: 10 to 240 V AC
Allowable voltage fluctuation: 85 to 264 V AC
Power frequency: 50/60 Hz
Power consumption: 30 VA max.
(2) Attach protective covers prior to turning on this instrument.
(3) Power supply module used in this instrument complies with EN60950.
(1) Do not operate this instrument in an environment where it is exposed to
combustible/explosive/corrosive gas or water/steam.
(2) Use this instrument in the following environment.
Installation site: Indoor
Altitude: 2000 m max.
Ambient temperature: 0 to 50
Ambient humidity: 20 to 80 %RH (Non-condensing)
Overvoltage category
Pollution degree
Provide input and output wiring after turning off the power.

1

CAUTION
Input and Output Wiring

Transportation

Inside of Instrument

Do not use unused terminals for any other purposes.
When transporting this instrument or the equipment with this instrument
incorporated in it, take measures to prevent the door opening and the innermodule falling out.
Do not touch the switches, etc. inside this instrument. Also, do not replace themain unit or printed circuit boards, otherwise correct functioning of the instrument
cannot be guaranteed. Contact the dealer where you purchased the instrument or
our sales representative.

[Note]
Instruction Manual(1) Deliver this manual to an end user.
(2) Prior to handling this instrument, be sure to read this manual.
(3) If you have any questions about this manual or find any errors or omissions inthis manual, contact our sales representative.
(4) After reading this manual, keep it in close proximity to the instrument.
(5) If the manual becomes lost or stained, contact our sales representative.
(6) It is prohibited to copy or reproduce this manual without our permission.

Installation(1) When installing this instrument, put on protective gear such as safety shoes,
helmet, etc.
(2) Do not step on the installed instrument or get on top of it, as this is dangerous.

Maintenance

Only our serviceman or engineers authorized by Shinko Technos Co., Ltd. areallowed to remove and take apart the inner module, the main unit and printed
circuit boards.

Disposal(1) Dispose of the replaced batteries in a correct way.
(2) Do not incinerate plastics from the maintenance parts or replacement parts.
Harmful gases may be produced.

Cleaning(1) Use a dry cloth to clean the surface of this instrument.
(2) Do not use any organic solvents.
(3) Turn off the power to the instrument before cleaning.

RevisionsThis manual is subject to change without prior notice.

SAFETY PRECAUTIONS• To ensure safe and correct use, thoroughly read and understand this manual before using this instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring
equipment.
Verify correct usage after purpose-of-use consultation with our agency or main office.
(Never use this instrument for medical purposes with which human lives are involved.)• Proper periodic maintenance is required.
• This instrument must be used under the conditions and environment described in this manual.
Shinko Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring due
to the instrument being used under conditions not otherwise stated in this manual.
Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the
manufacture of weapons of mass destruction (i.e. military applications, military equipment, etc.),
please investigate the end users and the final use of this instrument. In the case of resale, ensurethat this instrument is not illegally exported.

2

Using this manual
Using this manualThis manual consists of “For safe use”, “Contents” and “Chapter 1 to Chapter 11” as below.
Please read the corresponding sections depending on your purpose of use.
Chapter and title

For safe use (page 1)
1. INTRODUCTION
2. CONSTRUCTION
3. INSTALLATION
4. WIRING
5. PREPARATIONS FOROPERATION
6. OPERATION
7. DEVICE SETTING
8. COMMUNICATION
9. MAINTENANCE
10. TROUBLESHOOTING
11. SPECIFICATIONS

For purchaseandinstallation

For initial
setting andsetting changeOO

For daily
operation

O
O
O

O

For maintenanceandtroubleshootingForcommunication

O
O

O

O

O

O
O

O
O
O
： Be sure to read this.
O： Be sure to read this if required.

O

Guide of Instruction manual
The instruction manuals of this instrument are as in the table below.
Name
This
manual1

2

HR-700 Hybrid Recorder(Pen type) instruction
manualEdition No.
HR72E

HR-700 Hybrid Recorder
Communication Command HR7CE
instruction manual

OutlineExplanation for installation,
wiring, standard operation,
setting and operation forusing this instrument.
Explanation for reading andwriting data of the recorderby communication function.

Initial setting checking list
After installing and wiring the HR-701, 702, the following items are the minimum required setting items
before running.
Checking item
1 Range setting
Check the range setting.

Default value
10 mV DC (All channels)

Scaling setting is required for the following cases.
• When converting the input into another physicalquantity• When changing the decimal point position for thethermocouple and RTD input• When changing unit
2 Unit setting
Check the unit.
3 Chart feed speed setting
Check the Chart feed speed.
4 Date and time setting
Check the date and time.
3

Scaling 0 to 100.0 (Unit(All channels)

Unit

)

(All channels)

1st chart speed: 20 mm/h
2nd chart speed: 20 mm/h
Jan. 1st, 2000,
00:00 (Hour:Minute)

Reference
pp.52 to 54
pp.55, 56

pp.65, 66
p.67
p.68

CONTENTS
1. INTRODUCTION ....................................................................6
1.1 Checking the Accessories ...............................................................6
1.2 Checking the Model Type and Specifications................................8
1.3 Temporary Storage ...........................................................................9
1.4 Location ID tag ..................................................................................9

6.5.2<Manual Display>..................................................... 48

6.5.3<Date Display> ..................................................... 48

6.5.4<Time Display> ..................................................... 48

6.5.5<Display OFF> .......................................................... 48

7. DEVICE SETTING............................................................... 49

2. CONSTRUCTION................................................................ 10

7.1 Setting the Setup Mode................................................................. 49

2.1 Appearance .....................................................................................10

7.1.1 Setting the Range.................................................................... 52

2.2 Display and Operation Keys.......................................................... 11(1) Setting Method ......................................................................... 52

2.2.1 Display ....................................................................................... 11(2)

2.2.2 Operation Keys .........................................................................12(Current/Voltage),(Thermocouple),(Resistance Temperature Detector)........................ 53

3. INSTALLATION ................................................................... 13(3)(Scaling) ............................................................ 55

3.1 External Dimensions and Panel Cutout .......................................13(4)(Square Root) ....................................................... 57

3.2 Panel Mounting ...............................................................................14(5)

3.2.1 Procedure for Panel Mounting ................................................14(6)

3.2.2 Panel Mounting in Compliance with the IP65........................15(Decade) ........................................................... 59(Difference),(Sum),(Average)............................................................... 61(7)

4. WIRING................................................................................. 16(Skip) ...................................................................... 62

7.1.2 Setting the Alarm ..................................................................... 63

4.1 Terminal Arrangement and Power Wiring ....................................16

7.1.3 Setting the Unit......................................................................... 65

4.1.1 Terminal Arrangement..............................................................16(1) Character Code Table ............................................................. 66

4.1.2 Power Wiring.............................................................................16

7.1.4 Setting the Chart Feed Speed ............................................... 67

4.1.3 Wiring Procedure......................................................................17

7.1.5 Setting the Date and Time ...................................................... 68

4.2 Input Wiring .....................................................................................18

7.1.6 Copying the Setting Data........................................................ 69

4.2.1 Wiring Procedure......................................................................18

7.1.7 Setting Other Functions .......................................................... 70

4.3 DI Function/Alarm Output Wiring (Optional) ................................20
4.3.1 DI Function/Alarm Output Wiring Example............................20
4.3.2 Alarm Output Wiring Procedure ..............................................21
4.3.3 DI Function Wiring Procedure .................................................21
4.4 Communication Wiring...................................................................224.4.1 RS-232C Wiring .......................................................................22(1)(Zone Recording).................................................. 71(2)(Partial Compression/Expansion)....................... 72(3)(Digital Printing) ................................................ 73(4)(Tag) ........................................................................... 74(5)(Comments) .......................................................... 75

7.2 Setting Engineering Mode ............................................................ 76

4.4.2 RS-485 Wiring ..........................................................................22

7.2.1 Alarm Hysteresis...................................................................... 79

5. PREPARATIONS FOR OPERATION ............................... 23

7.2.2 Burnout ON/OFF ..................................................................... 79

5.1 Setting the Chart Paper..................................................................23

7.2.3 Channel Offset ......................................................................... 80

5.2 Setting the Cartridge Pen...............................................................30

7.2.4 Reference Junction Compensation ....................................... 80

5.3 Setting the Ribbon Cassette..........................................................32

7.2.5 Digital Filter ............................................................................... 81
7.2.6 Settings Related to Recording ............................................... 82

6. OPERATION ....................................................................... 37(1) Recording RUN/STOP Trigger Setting.................................. 82

6.1 Operation .........................................................................................37(2) Channel/Tag Printing Selection .............................................. 82

6.1.1 Status after Initial Screen .........................................................37(3) Alarm Printing ON/OFF ........................................................... 83

6.2 Recording ........................................................................................38(4) Logging Printing ON/OFF ....................................................... 83

6.2.1 Recording Colors......................................................................38(5) Printing Gap Correction ON/OFF .......................................... 84

6.2.2 Printing Color ............................................................................38(6) Recording Start/Stop Printing ................................................ 85

6.2.3 Pen Gap ....................................................................................38

7.2.7 Setting the Communication Function .................................... 86

6.2.4 Gaps between Pens and Printer ............................................38

7.2.8 Initializing the Setup Data and Calibration Data ................... 86

6.2.5 Printing Gap Correction ...........................................................39

7.2.9 DI Functions ............................................................................. 87

6.3 How to Record ................................................................................39

7.2.10 Temperature Unit Selection .................................................. 89

6.3.1 Starting/Stopping Recording ...................................................39

7.2.11 Point Calibration..................................................................... 90

6.3.2 Feeding the Chart Paper .........................................................39

7.2.12 Data Calibration ..................................................................... 91

6.3.3 Printing Sample ........................................................................40

7.3 Terminating Engineering Mode .................................................... 93

6.4 Digital Printing .................................................................................41
6.4.1 Manual Printing .........................................................................41
6.4.2 List Printing ................................................................................42
6.4.3 Engineering List Printing ..........................................................44
6.4.4 Recording Start/Stop Printing ...............................................46
6.4.5 DI Manual Printing (DI option) ................................................46
6.4.6 DI Date/Time Printing (DI option) ...........................................46
6.4.7 DI Comment Printing (DI option) ............................................47
6.5 Changing the Display .....................................................................48
6.5.1<Auto Display> ......................................................48

4

11. SPECIFICATIONS ............................................................117

8. COMMUNICATION ............................................................. 94

11.1 Common Specifications .............................................................117

8.1 General Description........................................................................94

11.1.1 Input Signal............................................................................117

8.1.1 General Description of Functions ...........................................94

11.1.2 Performance and Characteristics .......................................117

8.1.2 Transmission Specifications ....................................................94

11.1.3 Structure ................................................................................118

8.1.3 Data Structure...........................................................................94

11.1.4 Power Source .......................................................................118

8.2 Original protocol ..............................................................................95

11.1.5 Normal Operating Conditions ..............................................118

8.2.1 Data Structure...........................................................................95

11.1.6 Alarm Output (Option: LH3) .................................................1198.2.2 Opening/Closing the Link ........................................................96

11.1.7 Low Voltage Directive, EMC Directive................................119

8.2.3 Outputting the Process Variable .............................................96

11.1.8 DI Function (Option: RE1) ...................................................119

8.2.4 Outputting the Unit and Decimal Point Position Data ...........99

11.1.9 Paper-empty Detection Function (Option: FL1) ................119

8.2.5 Outputting the Status ............................................................ 100

11.2 Standard Setting Specifications................................................ 120

8.2.6 Data Reception Example ..................................................... 101

11.2.1 Measurement Range .......................................................... 120

8.3 MODBUS RTU Protocol ............................................................. 102

11.2.2 Accuracy at the Computation ............................................. 123

8.3.1 Data Structure........................................................................ 102(1) Scaling .................................................................................... 123

8.3.2 Function Codes ..................................................................... 102(2) Square Root Computation.................................................... 123

8.3.3 Error Response ..................................................................... 102(3) Decade ................................................................................... 124

8.3.4 Reading from the Input Register Area ................................. 103(4) Difference, Sum and Average .............................................. 124

8.3.5 Reading from or Writing to the Holding Register Area....... 106

11.2.3 Individual Specifications ...................................................... 124

9. MAINTENANCE ................................................................ 107

11.2.4 Standard Functions ............................................................. 125

9.1 Inspection ..................................................................................... 107

11.3 Standard Setting Functions ...................................................... 126

9.2 Cleaning........................................................................................ 107

11.3.1 Standard Setting Functions ................................................ 126

9.3 Replacing Consumables............................................................. 108

11.4 Optional Functions ..................................................................... 126

9.4 Adjusting the Pen Recording Position (Point Calibration)........ 108

11.4.1 Remote Function of DI ........................................................ 126

9.5 Calibration (Data Calibration) ..................................................... 108

11.4.2 Alarm Output Function ........................................................ 126(1) Calibration of Voltage ............................................................ 109

11.4.3 Communication Function.................................................... 126(2) Calibration of RTD (Resistance Temperature Detector).... 109(3) Calibration of Reference Junction Compensation ..............110

10. TROUBLESHOOTING ....................................................111
10.1 Troubleshooting .......................................................................... 111
10.1.1 Problems ............................................................................... 111
10.1.2 When the Recorder Does not Work at All.......................... 111
10.1.3 When There is a Large Error ..............................................112
10.1.4 When Pen Alignment is Incorrect .......................................112
10.1.5 When Nothing is Printed......................................................113
10.1.6 When Printing is Faint..........................................................113
10.1.7 When the Chart Paper is not Fed at All..............................114
10.1.8 When the Chart Paper is not Fed Properly .......................114
10.2 Self-Diagnosis Function (ERROR) ...........................................115
10.2.1 Self Diagnosis Items ............................................................115
10.2.2 Error Display .........................................................................1165

1． INTRODUCTION
1.1 Checking the Accessories
Upon delivery of this instrument, unpack and check its accessories and appearance.
If there are any missing accessories or damage to the appearance of the instrument, please contact ourdealer where you purchased the instrument, or our sales representative.
The following accessories are included with the recorder.

1. Chart paper

2. Ribbon cassette

4. Mounting brackets

3. Cartridge Pen(s)5. Instruction manual

6. Packing

Fig. 1.1-1 Accessories

Table 1.1 List of Accessories
No.

Parts

Type

Quantity
Remarks
1 pen 2 pens
1
1
100 equal divisions

1

Chart paper

H-10100

2

Ribbon Cassette

HPSR001H0002C

1

1

3

Cartridge Pen

WPSR196A000001A

1

1

For 1 pen (Red)WPSR196A000002A―――

1

For 2 pens (Green)4

Mounting brackets

H4A14175

2

2

Panel mounting brackets

5

Instruction manual

HR72E

1

1

This manual

6

Packing

H4H14900

1

1For IP65[Note]The ribbon cassette has been set in the instrument before shipment.

6

[Note](1) This instrument is equipped with a ribbon cassette in the back of the chart holder. Open thedoor and pull out the chart holder to check it. (See Section 5.1 for taking out the chart holder.)(2) The chart paper and the cartridge pen are contained in the chart holder. Take the chart holder
out and open the chart cover to check it.
(3) Maximum angle of the fully opened door is 135 degrees. Do not open the door any further, as
the hinge will break.

This instrument is shippedwith the ribbon cassetteloaded in the back of thechart holder.

Take the chart holder out to check accessories.
See Section “5.1 Setting the Chart Paper” for taking out the chart holder.
Fig. 1.1-2 Position of the Ribbon CassetteChart CoverThe chart paper iscontained in the chart
holder.
Open the Chart Guide
to check it.

The cartridge pen iscontained in the storage
chamber.
Open the chart cover tocheck it.

Chart Guide

Fig. 1.1-3 Position of the Chart Paper and the Cartridge Pen7

1.2 Checking the Model Type and Specifications
A nameplate has been secured on the inside of the instrument. Remove the chart holder and makesure that the nameplate is put in the middle far side of the instrument.
Make sure that this instrument meets your requested specification, checking the following table.

Table 1.2 Model type
HR–70

144 (W) x 144 (H) x 150 (D) mm,
1

1 point

2

2 points

Measuring point
C5

Communication function (RS-485)

RE1

DI function

FL1

Paper-empty detection function

LH3

Alarm output function

Options8

1.3 Temporary Storage
Store the instrument in the following environment.
When incorporated into other industrial equipment, ensure that the operating environment conforms to thefollowing conditions.

CAUTION
Storage in a poor environment may damage the appearance, functions, and service life of the
instrument.
Storage Environment
・A place free from excess dust
・A place free from combustible, explosive, or corrosive gases (SO2, H2S, etc.)
・A place free from vibrations or shocks
・A place free from water, steam or high humidity (95 %RH max.)
・A place free from direct sunlight or high temperature (50・A place free from extremely low temperature (-20

max.)

min.)

1.4 Location ID Tag
A location ID tag is kept on the inside of the transparent door upon delivery. Enter location or other data as
required.

CAUTION
Note that if a non-approved name label or nameplate is attached, it may damage the door or
mounting part.

9

2． CONSTRUCTION
2.1 Appearance
Door

Case

Display
keyboard

Cartridge pen
Location ID tag

Chart Holder

Mounting bracket

Terminal block

Fig. 2.1 Appearance102.2 Display and Operation Keys
2.2.1 Display
The following describes the displays and indicators. The channel numbers and data are indicated by a
7-segment LED.
For the characters, see "Characters used in this manual" at [Reference] below.

Alarm Type Indicator
Indicates alarm type in red.
"H" is lit when the alarm is High. And“L” is lit when the alarm is Low. Neitherof them is lit when there is no alarm.

Channel No. Display
Indicates channel
number in orange.

Data Display
Indicates the process
variable, time, setting
screen, etc. in orange.

Status Indicator
The "REC" indicator (orange) is lit when recording.
Operation Keys
Use these keys for setting and

The "ALM" indicator (red) is lit while the alarm isbeing activated.
anddenotes that the indicator is off,

other operations.

denotes that it is lit.

Fig. 2.2.1 Display[Reference] Characters used in this manual
DisplayAlphabet

A

B

C

D

E

F

G

H

h

I

J

K

L

l

M

N

O

P

Q

R

S

T

U

V

W

X

Y

Z

DisplayAlphabet

11

2.2.2 Operation Keys
The following describes the functions of each operation key.
The actual operation keys are described as shown below.

ESC
REC
R
FEEDP
MENU
PRINT
M
E
R
E
C
I
N
Fig. 2.2.2 Operation Keys in ThisU Manual N
T
Table 2.2.2 Name of Operation Keys and Their Functions

ENT
F
E
E
D

Key

Name

FunctionREC“REC” key

Starts/stops recording. To stop, press the “REC” key for 3 seconds orlonger. (The “REC” indicator goes off.) To start, press the “REC” key.
(The “REC” indicator lights.)“MENU” key

MENUfunction

Selects engineering list printing and changes
the mode to the setting mode.

“ESC” key

ESC function

While selecting a function, exits that menu
without finishing.

ESC
MENU

“PRINT” key

PRINTING
function

Used to perform manual printing or list
printing.

PRINT
” key

function

Used to select a setting parameter (numeralor built-in command) (ascending direction).

“FEED” key

FEED function

The chart paper is fed while the key ispressed, and stopped when released.

“

FEED
“

ENT

“ key

“ENT” keyfunction

Used to shift a digit in setting a numeral.

Used to register a setting parameter (numeral or built-in command)
after selecting it, or executing a function. Pressing this key executes
the setting.

12

E
N
T

3． INSTALLATION
3.1 External Dimensions and Panel Cutout(Scale: mm)<Front>2.5<Rear>

144

CH,NO.

(100)

3.5

137

DATA

ESC

RUN

MENU PRINT FEED

ENT

1A

Dimension required
when the door is fully
opened (135 ).
(Wall on the left side).

1+
B

1B

1
A

2+
B

2B

1C

2
A

2A

L

N

2C 3A

3C

+

-

DI
C OM

DI
3

DI
2

DI
1

TD

RD SG

137

ALM

144

RUN

（56）

<Side><Panel Cutout>
200min.

（128）+1
138 -0200min.

7 MAX（Panel thickness）

（128）

150

25

138 +1-0

Fig. 3.1 External Dimensions and Panel Cutout

CAUTION
For the maintenance and safety of the instrument, the space/size indicated in the brackets ( ) is theminimum secure spacing.

13

3.2 Panel Mounting

WARNING
Do not install this instrument in a place exposed to combustible, explosive or corrosive gas (SO2,
H2S, etc.).

CAUTIONInstall the instrument only in the following conditions.
• A place free from frequent humidity fluctuations.
• A place at room temperature (around 25 ).
• A place exposed to as few mechanical vibrations as possible.
• A place with as little dust as possible.
• A place affected as little as possible by electromagnetic fields.
• A place not directly exposed to high radiant heat.
• A place of altitude below 2000 m.
• Humidity has an effect on the chart paper and ink. Use the instrument in a humidity range of 20 to
80 %RH (60 %RH is the optimum).
• This instrument needs to be installed inside.
Panel Mounting• A steel plate not thinner than 1.2 mm is recommended as a mounting panel.
• The maximum thickness of the mounting panel is 7 mm.
Inclination• Install the instrument horizontally.
• The instrument should be installed so that its inclination should be 0 at the front and within 30 at
the rear.
.
3.2.1 Procedure for Panel Mounting
1) Assemble the mounting bracket, referring to Fig. 3.2.1-1.
2) Insert this instrument through the front of the panel.
3) Hook the claws of the mounting bracket at the square holes on the top and bottom surfacesof the case.
4) Tighten the screw of the mounting bracket with a screwdriver to attach it to the panel. Whenthe mounting unit does not move back and forth any more, tighten the screw by 180 .

Assemble in such a mannerthat the screw nose will bealmost aligned with the platesurface.

Mounting Claw

Body
Mounting Claw

Fig. 3.2.1-1 Mounting bracket[Note]
If it is tightened with excessive force, the case may be distorted and the mounting brackets may bedamaged. An adequate tightening torque is about 0.2 to 0.3 N･m (2 to 3 kgf•cm).

14

PanelMounting bracket

Screw

Fig. 3.2.1-2 Panel Mounting

3.2.2 Panel Mounting in Compliance with the IP65
Prior to mounting the instrument to the panel, attach packing to the position shown in the figure.
The rest of the procedure is the same.
Panel
Packing
Mounting bracket

Mounting of Packing

Screw

Packing

Panel

Fig. 3.2.2 Panel Mounting (in compliance with the IP65)

15

4. WIRING
4.1 Terminal Arrangement and Power Wiring
4.1.1 Terminal Arrangement
Power source
terminal block
Alarm output terminal block(Option: LH3)

L

Communication
terminal block (RS-485)(Option: C5)

N

1A 1C 2A 2C 3A 3C +-DI terminal block(Option: RE1)

Input terminal block
DI
COM

1+
B

1B

1
A

2+
B

2B

2
A

DI
3

DI
2

DI
1

TD RD SGCommunication
terminal block (RS-232C)

Fig. 4.1.1 Terminal Arrangement (Rear view)

4.1.2 Power Wiring

WARNING(1) In order to prevent an electric shock, be sure to provide protective grounding prior to supplying
power to the instrument.
(2) Do not cut the protective grounding conductor or disconnect protective grounding.
(3) Make sure that the supply voltage for the instrument conforms to the voltage of the supply source.
(4) Attach a transparent protective cover prior to switching on power to the instrument.
(5) Any disruption to the protective conductor inside or outside the instrument, or disconnection of theprotective grounding terminal, is likely to make the instrument dangerous under conditions that areother than normal.
Intentional disruption is prohibited.

16

CAUTION(1) As an electric wire for the power source, use a 600 V vinyl insulated wire (IEC60227-3) or its
equivalent or above.
(2) Attach a ring-type solderless terminal with an insulation sleeve (for M3.5) to the end of the electric
wire.
(3) Connect a protective grounding (resistance: 100

or less, minimum diameter of a grounding

conductor: 1.6 mm) to the protective grounding terminal.
(4) If another instrument shares the protective grounding conductor, there may be interference caused
by noise coming from the grounding conductor. Do not share the grounding conductor with other
instruments.
(5) In order to comply with the requirements of safety standard EN61010-1, the recorder should haveone of the following as a disconnecting device, fitted within easy reach of the operator, and labeled as
the disconnecting device.
a) A switch or circuit breaker which complies with the requirements of IEC60947-1 and IEC60947-3.
b) A separable coupler which can be disconnected without the use of a tool.
c) A separable plug, without a locking device, to plug in a socket outlet in the building.
(6) This product has been designed to conform to EN61010-1, installation category

and pollution

degree 2.
(7) The main power supply should be used within a fluctuation range of

10 % of rated voltage.

(8) Transient current might flow to the main power supply when the power supply is turned on.
4.1.3 Wiring Procedure

CAUTIONTo remove the transparent protective covers, release the left and right (both sides) hooks simultaneously.
If they are released one after the other, the cover may be damaged.
1) Put your fingers on the left and right (both sides) hooks of the transparent protective cover on the
power source terminal block. Pushing them in, pull the cover toward you.
2) Connect the power source electric wire, referring to Fig. 4.1.3. Connect the protective grounding totheterminal. Connect the non-grounding side of the power source to "L" terminal.

Connect the grounding side to the "N" terminal.
3) Attach the transparent protective cover.
4) Make sure that protective grounding is properly done.

L

N

Power supply:
Rated power supply: 100 to 240 V AC
Allowable voltage fluctuation: 85 264 V AC
Power frequency: 50/60 Hz

Fig. 4.1.3 Wiring of Power Supply
17

4.2 Input Wiring

CAUTION(1) Precautions for the input electric wire
• Make sure that noise is eliminated when wiring. For input wiring, a shielded wire or twisted wire should
be used to eliminate noise.
• In the case of thermocouple input, connect a thermocouple wire directly or use a compensating leadwire. A shielded input line should be used.
• In the case of RTD input, dispersion of 3-wire line resistance should be lower than the value mentioned
below. A shielded input line should be used.
For Pt100, JPt100； 50 m max.
• When it is likely to be affected by induction noise, particularly when wiring near a high-frequency powersource, a shielded twisted wire should be used.
• Attach a ring-type solderless terminal with insulation sleeve (for M3.5) to the end of the electric wire.
(2) Precautions for wiring• The wiring between the instrument and measurement point should be kept away from the power circuit(25 V or higher of either power supply or DO circuit).
• Short-circuit unused input terminals. (Short-circuit between "+" and "-" for mV, V or thermocouple input,
and short-circuit among A, B, and B of RTD input.)• Be sure to ground the connecting wire shield.
4.2.1 Wiring Procedure

CAUTIONTo remove the transparent protective covers, release the left and right (both sides) hooks simultaneously.
If they are released one after the other, the covers may be damaged.
(1) Place fingers on both left and right hooks of the transparent protective cover on the input terminal block.
Pushing them in, pull the cover toward you.
(2) Wire the input lines referring to Fig. 4.2.1-1 (p.18), Fig. 4.2.1-2 and Fig. 4.2.1-3 (p.19).
(3) Attach transparent protective covers.

1+
B

1B

1
A

2+
B

CH1

2B

CH2

DC voltage

+

-

+

-

Thermocouple

+

-

+

-

RTD

B B

Direct current

+

A

-

2
A

B B

+

A

-

Fig. 4.2.1-1 Input Wiring
18

DI DI
COM 3

DI
2

DI
1

TD

RD

SG

(1) mV, V inputs

+
B

B

+

(2) Thermocouple input

+
B

A
-

B

+

A
-

Fig. 4.2.1-2 Input Wiring (mV, V, Thermocouple inputs)(3) RTD input

+
B
B

B

(4) mA input

+
B

A

B

B

A

A

Shunt resistor+-

Fig. 4.2.1-3 Input Wiring (RTD, mA inputs)

CAUTION(1) Attach a shunt resistor to the input terminal block of the instrument.
(2) A shunt resistor affects input accuracy. Use the following resistor.
Resistance: 250

Rated power: 1/4 W Tolerance:

Temperature coefficient:

50 ppm max.

19

0.1 % max.

4.3 DI Function/Alarm Output Wiring (Optional)

WARNING(1) Be sure to wire after turning POWER OFF.
(2) When the power source has been connected to the Alarm output, turn OFF that power source.
(3) When a hazardous voltage is supplied to the alarm terminal:a) Never touch terminals.
b) Attach covers to terminals.
c) Wires should be double shielded.
d) Use ring-type solderless terminal connectors with an insulation sleeve.

CAUTION
Precautions for DI Function Wiring(1) DI function input has a built-in drive power source. Do not apply an external voltage to DI function input
terminals.
(2) DI function input contact capacity should be a withstanding voltage of 50 V DC, 16 mA or more,
ON resistance of 20 max. (wiring resistance included).
(3) Do not use unused terminals as relay terminals.
Precautions for Alarm Output Wiring(1) An alarm output contact capacity is as follows.
250 V AC : 3 A max. (Resistive load)
30 V DC : 3 A max. (Resistive load)
125 V DC : 0.5 A max. (Resistive load)
0.1 A max. L/R= 7 ms max. (Inductive load)(2) Attach an anti-surge protective circuit (surge absorbers, etc.) to an output terminal, when required.
(3) Attach a ring-type solderless terminal with an insulation sleeve (for M3.5) to the end of an electric wire.
(4) Keep alarm output wiring away from the input wiring.
(5) Do not use unused terminals as relay terminals.

4.3.1 DI Function/Alarm Output Wiring Example+12V
Terminal Block
Terminal Block
DI1～3

1A～3A
1C～3C

DI.COM

Load

Recorder Side
Recorder SideCOM

Fig. 4.3.1-1 DI Function Wiring Example

Fig. 4.3.1-2 Alarm Output Wiring Example

CAUTIONThe DI function (Option: RE1) consists of 3 Digital inputs.
The alarm output consists of 3-Relay output (Normally open).

20

4.3.2 Alarm Output Wiring Procedure
Wire the Alarm output (option: LH3), referring to Fig. 4.3.2.

1A 1C

2A 2C

L

N

3A

3C

+

-

3-Relay output
1C

3C

…..

1A
Relay No.1..
(Normally open)

3A

Relay No.3
(Normally open)

Fig. 4.3.2 Alarm Output Wiring

CAUTION
Relay No.1 is a common output for Alarm output and Paper-empty detection sensor (optional)(OR output).
4.3.3 DI Function Wiring Procedure
Wire the DI function (option: RE1), referring to Fig. 4.3.3.

LN

1A 1C 2A 2C 3A 3C +

DI
COM

1+
B

1B

1
A

2+
B

2B

DI
3

DI
2

-

DI
1

TD RD SG

2
A

DI1
DI2
DI3
DI.COM(COMMON to DI1 to DI3)
Fig. 4.3.3 DI Function Wiring
21

4.4 Communication Wiring

CAUTION
Precautions for the communication wiring(1) Ensure that the communication wiring is unaffected by noise. For communication wiring, a shieldedwire effective against noise should be used.
(2) When it is likely to be affected by induction noise, particularly when wiring near a high frequency
power source, a shielded twisted wire should be used.
(3) Attach a ring-type solderless terminal with an insulation sleeve (for M3.5) to the end of the electric wire.
(4) Be sure to ground the shield of a connecting wire.
4.4.1 RS-232C Wiring
Wire RS-232C, referring to Fig. 4.4.1.
1A 1C 2A 2C 3A 3C +-［Note］
Cable length is a maximum of 15 m.

DI
COM

DI
3

DI
2

DI
1

Host computer
1+
B

1B

1
A

2+
B

2B

TD RD SG2
A

D sub 9-pin connector
3
2
5
1
4
6
7
8
9

Fig. 4.4.1 RS-232C Wiring

4.4.2 RS-485 Wiring
Wire RS-485, referring to Fig. 4.4.2.

TXD
RXD
GND
DCD
DTR
DSR
RTS
CTS
RI

D sub 25-pin connector

⑬ RD
⑫ TD
⑭ SG

2
3
7
1
4
5
6
20
8

TXD
RXD
SG
FG
RTS
CTS
DSR
DTR
CD

[Note]

1A 1C

2A

2C

L

N

3A

3C

+

(1) Be sure to attach terminating
resistance (Rt 200 ) to a hostcomputer side, and attach it to theremotest HR-701 or 702 from the hostcomputer.
(2) Maximum of 32 units, including ahost computer, connectable.
(3) Cable length is a maximum of 1.2 km.
(4) Use the UL20620-SB(M) cable madeby HITACHI CABLE LTD. or equivalent.

-

Host computer

Ｒｔ－

Fig. 4.4.2 RS-485 Wiring
Ｒｔ＋

HR-70122………

HR-701

5． PREPARATIONS FOR OPERATION
5.1 Setting the Chart Paper

CAUTION
Our original chart paper should be used to ensure proper recording.
If the chart holder is taken out and the recording operation is activated, the ink ribbon may bedamaged. To replace the chart paper, be sure to press the “REC” key to stop recording.
Be sure to move the cartridge pen up, otherwise, it will be damaged.
(1) Press the “REC” key for 3 seconds or longer.
To stop recording, press the “REC” key for 3 seconds or longer with the power turned on.

ＬＭ

Press the key for 3 seconds

The REC indicator goesor longer.

off if recording stops.

Fig. 5.1-1 Pressing the “REC” key
When recording stops, the pen(s) and printer will move automatically, discontinuing paper feed.
For the HR-701, the pen (red) records at the center (50%).
For the HR-702, No.1 pen (red) records on the left side (approx. 15%), and No.2 pen (green)
records on the right side (approx. 85%) respectively.

No.2 pen(green) to
the right

No.1 pen (red)The printer moves

to the left

to the right.

Fig. 5.1-2 Two Pens and Printer will automatically move.
23

(2) Open the Door.

CAUTION
Maximum angle of the fully opened door is 135 degrees.
Do not open the door any further, as the hinge will break.
Push the door hook to unlock the door. Pull the door to open it.

Push the door
hook in thearrow direction
to unlock.
Door Hook

Fig. 5.1-3 Opening the Door
(3) Push up the Pen.
Push the pen-up lever upward to move pen tips away from the chart paper.

Pen-up lever

The pen-up lever islocated at the inside rightend of the case.
Push it up using a finger.

Fig. 5.1-4 Moving up the Pen

24

(4) Pull out the Chart Holder.
Press the levers at both sides of the chart holder and pull it out.

Lever
Pull out the chart
holder.
Fig. 5.1-5 Pulling out the Chart Holder(5) Open the Chart Cover and Chart Guide.
Gently open the chart cover and chart guide doors, following the direction of the arrows.

Chart Cover (Black)

Open the chart cover in thearrow direction.

Chart Guide(Transparent)

Open the chart guide in
the arrow direction.

Fig. 5.1-6 Opening the Chart Cover and Chart Guide25

(6) Loosen the chart paper.
The chart paper may not feed properly, if stuck together. Be sure to loosen the paper.

Fig. 5.1-7 Loosening the Chart Paper(7) Set the Chart Paper into the Storage Chamber.
Unfold 2 sections of chart paper. Holding the printing surface upward, set it in the storage
chamber.

Rectangular Holes on the Right

Square Holes on the Left

Sprocket Drum
Storage Chamber

Fig. 5.1-8 Setting the Chart Paper26

(8) Align the Chart Paper with the Sprocket Drum.
Align the holes in the chart paper with the sprocket drum teeth. Set the chart paper along with the
sprocket drum. Put the first ply of the chart paper into the chart receiver.
Ensure that the left and rightholes are parallel with each other.

Chart Paper

Chart Receiver

Chart Paper

Storage Chamber

Fig. 5.1.9 Aligning the Chart Paper with the Sprocket Drum (Chart Holder Sectional View)(9) Close the Chart Cover and the Chart Guide.
Gently close the chart cover and the chart guide doors, following the direction of the arrows.
Chart CoverAlign the holes in thepaper properly with the
sprocket drum teeth.

Sprocket Drum Teeth

Chart Guide

Fig. 5.1-10 Closing the Chart Cover and Chart Guide27

(10) Turn the Sprocket Drum Gear.
Check paper feed by hand. Turn the sprocket drum gear to feed out the chart paper.
(4 sections of chart paper should be fed out.)

Turn the gear at the left endof the sprocket drum in thearrow direction.

Fig. 5.1-11 Checking Paper Feed(11) Put the Chart Holder back.
Insert the chart holder horizontally into the case until it is locked.

Chart Guide

Gently push the chart
receiver for insertion.
Chart ReceiverFig. 5.1-12 Putting Back the Chart Holder

28

(12) Move the Pen down to close the Door.

CautionBe careful when opening the door more than 90 degrees. Also make sure that the display cable
does not get caught.

Move the pen down with the pen-up lever. Close the door and check that it is locked.
The pen-up lever is located at the inside right end of the case. Push it down using a finger.

(13) Press the “FEED” key to check Paper Feed.
Press the “FEED” key on the display keyboard to feed the chart paper.

ＬＭ

The chart paper is fed
while pressing the key.
Releasing the key stops

Prior to the operation of

paper feed. The paper

the instrument, be sure to

begins a slow feed when

confirm that the chart

the key is pressed.

paper is being fed.

Fig. 5.1-13 Pressing the “FEED” key(14) Press the “REC” key to restart operation.

29

5.2 Setting the Cartridge Pen

CAUTIONTo replace the cartridge pen, be sure to press the “REC” key to stop recording.

(1) Press the “REC” key for 3 seconds or longer.
With the power turned on, press the “REC” key for 3 seconds or longer.
Once recording stops, the REC indicator goes off. (See Fig. 5.1-1 on page 23.)
The pen(s) and printer automatically move, discontinuing paper feed.
For the HR-701, the pen (red) records at the center (50%).
For the HR-702, No.1 pen (red) records on the left side (approx. 15%), and No.2 pen (green)
records on the right side (approx. 85%) respectively. (See Fig. 5.1-2 on page 23.)(2) Open the Door.
Push the door hook to unlock the door. Pull the door to open it. (See Fig. 5.1.3 on Page 24.)WARNINGTo avoid injury to fingers, be careful when opening/closing the door.

CAUTION
Maximum angle of the fully opened door is 135 degrees. Do not open the door any further, as the(3)
Detachthe Cartridge Pen.
hingewill break.
(3) Pull out the Cartridge Pen.
Pen Holder
Hold here using
your fingers.
Cartridge Pen

Pen Pipe

Fig. 5.2-1 Detaching the Cartridge Pen[Note]
When pulling out the cartridge pen, do not hold the pen pipe. If it becomes bent, improper
recording may result.

30

(4) Remove the Pen Cap.
Gently remove the pen cap from the new cartridge pen, taking care not to bend the pen pipe.

Pen Cap

Fig. 5.2-2 Removing the Pen Cap(5) Attach the new Cartridge Pen.
Attach the new cartridge pen to the pen holder. Push it in firmly until a clicking sound is heard.

CautionIf the cartridge pen is not attached properly, recording might skip when pens cross.

Hold here withPen Holder

your fingers.

Cartridge Pen

Pen Pipe

Attach

No.1

pen

(red)

above

and

No.2 pen (green)
Fig. 5.2-3 Reattaching the Cartridge Penbelow, respectively.

(6) Close the Door.

CautionBe careful when opening the door more than 90 degrees. Also make sure that the display cable
does not get caught.

(7) Press the “REC” Key to restart operation.

31

5.3 Setting the Ribbon Cassette

CAUTION
If recording operation is activated while the chart holder is removed, the ink ribbon may bedamaged. To replace the ribbon cassette, be sure to press the “REC” key to stop recording.
Be sure to move the cartridge pen up. Otherwise, it will be damaged.
(1) Press the “REC” key for 3 seconds or longer.
Recording stops. With the power turned on, press the “REC” key for 3 seconds or longer.
Once recording stops, the REC indicator goes off. (See Fig. 5.1-1 on page 23.)
The pen(s) and printer automatically move, discontinuing paper feed.
For the HR-701, the pen (red) records at the center (50%).
For the HR-702, No.1 pen (red) records on the left side (approx. 15%), and No.2 pen (green)
records on the right side (approx. 85%) respectively. (See Fig. 5.1-2 on page 23.)(2) Open the Door.
Push the door hook to unlock the door. Pull the door to open it. (See Fig.5.1-3 on Page 24.)WARNINGTo avoid injury to fingers, be careful when opening/closing the door.

CAUTION
Maximum angle of the fully opened door is 135 degrees. Do not open the door any further, as thehinge will break.
(3) Move the Pen up.
Push the pen-up lever upward to move pen tips away from the chart paper. (Move the pen up;
See Fig.5.1-4 on page 24)(4) Pull out the Chart Holder.
Put your fingers onto the levers at both sides of the chart holder and pull it out. (See Fig.5.1-5
on page 25.) The guide shaft, printer, and the main shaft are laid out as shown in the figure
below. (The main shaft is invisible if the ribbon cassette has been set.)
Guide Shaft

Main Shaft

PrinterFig. 5.3-1 Layout of Guide Shaft, Printer and Main Shaft

32

(5) Lower the Ribbon Cassette.
Detach both sides of the ribbon cassette from the guide shaft.

Guide Shaft

Lower both sides of
the ribbon cassette.
Detach it from theguide shaft, one side
at a time.
RibbonFig. 5.3-2 Detaching from the Guide Shaft(6) Pull the Ribbon Cassette down and out slantwise at an angle of approximately 30 degrees.
Detach the ribbon cassette from the main shaft.
Guide Shaft

Main Shaft

Ribbon CassetteFig. 5.3-3 Detaching from the Main Shaft
When viewed from the right side (section), the hook of the ribbon cassette has been engaged withthe main shaft.
Main Shaft

Detach the hook of theribbon cassette in such a

Ribbon Cassettemanner as to disengage itfrom the main shaft.
Hook

Fig. 5.3-4 Ribbon Cassette Sectional View

33

(7) Level the Ribbon Cassette.
Prior to pulling out the ribbon cassette, level it.

Lower the hook of theribbon cassette.

Main Shaft
Ribbon Cassette
HookFig. 5.3-5 Leveling the Ribbon Cassette (Ribbon Cassette Sectional View)(8) Pull out the Ribbon Cassette.
Gently pull the ribbon cassette out horizontally. If the gear of the ribbon cassette touches theprinter, move the printer further to the right.

(9) Tighten the Ink Ribbon.
Set the new ink ribbon. Turn the gear in the arrow direction to tighten the ink ribbon.

Turn the gear in thearrow direction.

Gear
Ink Ribbon

Fig. 5.3-6 Tightening the Ribbon

34

(10) Insert the Ribbon Cassette Horizontally.
Gently insert the ribbon cassette horizontally.

Guide Shaft

Main Shaft

Ribbon CassetteFig. 5.3-7 Inserting the Ribbon Cassette[Note]
If your hands become stained with ink when replacing the cartridge pen or ribbon cassette, washwith soap after finishing.

(11) Reattach the Ribbon Cassette to the Main Shaft.
Holding the ribbon cassette aslant, reattach it to the main shaft in such a manner that the hookof the ribbon cassette engages with the main shaft.

Hook
Insert it horizontally

Slant upward toengage with themain shaft.
Ribbon Cassette“Click” sounds whenengaging with themain shaft.
Main Shaft

Fig. 5.3-8 Reattaching to the Main Shaft (Ribbon Cassette Sectional View)

35(12) Push up the Ribbon Cassette.
Attach both sides of the ribbon cassette to the guide shaft. If the gear of the ribbon cassette is
not properly engaged with the printer, slide the printer slightly.

Attach both sides to the guideshaft, one side at one time.
A clicking sound is heard if
properly engaged.
Guide Shaft

RibbonFig. 5.3-9 Attaching to the Guide Shaft(13) Put the Chart Holder back.
Insert the chart holder horizontally into the case until it is locked.
(See Fig. 5.1-12 on page 28.)(14) Move the Pen down to Close the Door.

CautionBe careful when opening the door more than 90 degrees. Also make sure that the display cable
does not get caught.

Use the pen-up lever to lower the pen. Close the door and make sure that it is locked.
The pen-up lever is located at the inside right end of the case.
Push it down using a finger.
(15) Press the “REC” key to restart operation.

36

6． OPERATION
6.1 Operation

WARNING
Prior to switching the power on, make sure that the supply voltage meets the specifications for the
instrument and the instrument is properly grounded.

CAUTION
Prior to switching the power on, make sure that the chart paper is set in the chart holder.
If the printer is activated with no chart paper set, the sprocket drum (cylindrical part) of the chart
holder may be damaged.

WARNING
When calibrating data, connect the equipment written in Section 9.5 of the instruction manual.
Data calibration must be performed correctly.
Operators possessing knowledge of electricity and control related matters should perform thecalibration after careful reading of Sections 7.2 and 9.5 of the instruction manual.
(Shinko Technos Co., Ltd. is not liable for damages incurred if calibration is performed other thanprescribed in the instruction manual.)Once the power is turned on, the display on the front of the door lights. The instrument will be ready torun (user mode) in about 5 seconds, including the initial screen. If the REC indicator is not lit, press the“REC” key to start recording operation.
6.1.1 Status after Initial Screen
The “REC” indicator retains its status prior to power-off.
All printing data and analog recording data are cleared after the power has
been switched off.
(3) Alarm and Diagnosis： Alarm indications and output information are not recoverable after the
power has been switched off.
(4) Data and Channel number are indicated as the following Fig. 6.1.1.
(1) Display:(2) Printout data:

REC ALM

Returns to the mode before
power-OFF.
Fig. 6.1.1 Screen at Power-on[Note](1) If electric power failure occurs, initialization is performed after power recovery. The recorderreverts to the initial status above.
(2) The printout data is eliminated in initialization. Printout operation does not continue after power
recovery when the power is turned off during printout.
(3) When the power is turned on, the display indicates “ ” while reading data in Fig.6.1.1. The
number of the indicated digits differs depending on the setting.
37

6.2 Recording
6.2.1 Recording Colors
Table 6.2.1 shows the recording colors for each channel in analog recording.
Table 6.2.1 Recording Colors
Channel No.
1
2

Color
Red
Green

6.2.2 Printing Color
Various digital printings are recorded in purple.

6.2.3 Pen Gap
Fig.6.2.3 indicates the gap between No.1 and No.2 pens.

CH2（Green）

CH1
2 mm
（Red）

Fig. 6.2.3 Gap between Pens

6.2.4 Gaps between Pens and PrinterFig. 6.2.4 indicates the gap between No.1/No.2 pens and printer.
Printer

00 ABC

20.5 mm22.5 mm

CH2
CH1(Green)

2 mm

(Red)

Fig. 6.2.4 Gaps between Pens and Printer38

6.2.5 Printing Gap Correction
This is a function to correct the gap between the penpoint and the end of the tracing.
When setting this function ON, chart paper for the whole printout contents is automatically sent
to the position in front of Pen 1 after the asynchronous printout is completed. (Fig. 6.2.5)
The printout contents can be checked.
For this function, only DI asynchronous printing and asynchronous log printing are available.
• Printing gap correction OFF
12:00• Printing gap correction ON.

STOP MEAS.
Pen 1

Pen 1

12:00

STOP MEAS.
Gap Correction

Fig. 6.2.5 Printing gap correction

6.3 How to Record
6.3.1 Starting/Stopping Recording
To start/stop recording, press the “REC” key.
To stop, press the “REC” key for 3 seconds or longer.
The “REC” indicator goes off when recording stops.

[Note](1) When the power is turned on, the recorder continues from the state in which it was turned off.
(2) When controlling recording start/stop through Digital Input (DI function, option: RE1),
recording start/stop cannot be switched by the “REC” key.
(3) When recording stops, No.1 pen (red) holds a left-side position (approx. 15 %) and No.2 pen(green) a right-side position (approx. 85 %) respectively. Paper feed is suspended, but the
FEED key movement is enabled.

6.3.2 Feeding the Chart Paper
The chart paper is fed while the “FEED” key is pressed, and stops when released.

39

6.3.3 Printing Sample
Logging printing and Alarm printing
Recording Start/Stop printing examples
Recording End

Oct.20. 06 20 : 55 End

Alarm Printing
Printsalarm
occurrence/recovery,
channel number, alarm type and level.
denotes alarm occurrence, anddenotes alarm recovery, respectively.

Date, Time
Oct.20.06
20:00
1H1 20:29
2L2 20:28
34.0
20.1491H1 20 :19

1:
2:
Chart Speed

Process variable
Prints "" if skipped.
Prints no data on the channel
where digital print has been set
to OFF.

100mm/hOct. 20. 06 19 : 00 Start

See the [Reference] below.
Logging Printing
Prints date and time, channel
number,
Processvariable,
engineering unit and chart speed.

Recording Start

Fig. 6.3.3 Printing Sample[Reference]
When other printing is active with current job, Alarm printing, Comment printing and Date-and-Time
printing will queue.
HR-701 (HR-702) can store up to 6 items of Alarm Occurrence/Recovery printing and 5 items ofComment/Date-and-Time printing. If printing commands exceed those numbers, HR-701 (HR-702)
prints a " " at the end of the last printing. This mark means that any following items after the item witha “ ” mark could not be printed.
For priority of respective printings, see “Priority” in Reference on the next page.
Synchronous printing: Performs digital printing with the same speed as the analog recording.
Asynchronous printing: Stops analog recording and performs digital printing. After digital printing
iscompleted, restarts analog printing.

[Note]
Logging printing will not be in the printing queue. At the start time of Logging printing, if prior Logging
printing is printing, the next log will not print. In this case, prolonging the printing interval of the
Logging printing is recommended.

40

6.4 Digital Printing
Digital printings listed below are performed by key operation.
• Manual printing• List printing• Engineering list printing• Recording Start/Stop printing
Digital printings listed below are performed by DI input.
• DI Manual printing• DI Date/Time printing• DI Comment printing[Reference]
Printing is activated in the following priority. See the following Priority Order. When multiple types ofprinting are activated simultaneously, higher-priority printing takes effect first according to the following
order.
High

List Printing
Manual Printing
Engineering
List Printing

PriorityLow

DI Comment
printing (ASync.)

DI Manual
printing(ASync.)

Alarm Occurrence
printing

DI Comment
printing(Sync.)

Logging
Printing(Sync.)

DI Date and Time
printing (ASync.)

Logging printing(ASync.)Alarm Recovery
printing

DI Date and Time
printing(Sync.)

DI Manual
Printing(Sync.)

RecordingStart/Stop printing(ASync.)RecordingStart/Stop printing(Sync.)The Synchronous printout synchronizes with the analog recording, and executes digital printout.
In the case of the asynchronous printout, an analog recording stops and a digital printout is executed.
After the digital printout is completed, an analog recording resumes.

6.4.1 Manual Printing
Prints the following data on the chart paper:• Time/date (year, month, day)
• Channel number/Tag setting characters• Activated alarm type/Latest Process variable/Engineering unit
(1) Manual printing operating procedure
1) Press the “PRINT” key.
2) Use the “PRINT” key to display “”, and press the “ENT” key.

3) Use the “PRINT” key to select “”. Pressing the “ENT” key executes manual printing.

Once manual printing starts, the display automatically returns to the Data Display screen.
Upon termination of manual printing, the dispaly returns to the status prior to starting manual printing.

[Note](1) Analog recording continues while manual printing is running. Chart paper is fed.
(2) If an alarm is activated while manual printing is running, alarm printing will start from the time
when manual printing finished.
41

(2) Manual printing stopping procedure
1) Press the “PRINT” key.
2) Use the “PRINT” key to display “”, and press the “ENT” key.

3) Use the “PRINT” key to select “”. Pressing the “ENT” key stops the manual printing.

However, for the line during printing, printing will continue until the line is completely printed.
Then Manual printing will stop, and the display will return to the status prior to starting manual printing.
Mar.16.00 13:38
1:
23.3
2: 22.992mV

Fig. 6.4.1 Manual Printing[Note]For the channel where digital printing has been set to OFF, printing is not performed.

6.4.2 List Printing
Prints the following setup data of the instrument on the chart paper:• Date/time/Chart speed/2nd chart speed• Channel number/range/tag/scaling value/engineering unit.
• Alarm type
• Comment/Others(1) List printing operating procedure
1) Press the “PRINT” key.
2) Use the “PRINT” key to display “”, and press the “ENT” key.

3) Use the “PRINT” key to select “”. Pressing the “ENT” key executes the list printing.

Once list printing starts, the display automatically returns to the Data Display screen.
Upon termination of list printing, the display returns to the state prior to starting list printing.

[Note](1) Analog recording continues while list printing is running. Chart paper is feed.
(2) If an alarm is activated while list printing is running, alarm printing is activated from the time
when list printing finished.

42

(2) List printing stopping procedure
1) Press the “PRINT” key.
2) Use the “PRINT” key to display “”, and press the “ENT” key.

3) Use the “PRINT” key to select “”. Pressing the “ENT” key stops list printing.

However, printing continues until the line is completely printed. Then list printing stops. And thedisplay returns to the status prior to starting list printing.

Date and Time
for List Printing

Chart Speed Set
value.
The 2nd Chart
Speed
(2)
is
effective for DI
function (optional).

Prints alarm set
values and relays.

Mar.08.00 13:12
CHART (1)20 mm/hSPEED (2)12000 mm/h

CH1 Type K
TAG01 NORMAL
0.0 50.0
CH2 mA
SCALE
4.00 20.00mA
0.00 100.00%

Prints Input range, Tag,
Measurement range, andScaling value for each channel.
If skipped, the channel number
is prefixed by " ".

CH:LV ALM RLY
1:1 H 28.0 1
1:2 L 18.0
2:4 H 30.00 3

CH DIGITAL ZONE(mm)
1 ON
0- 50
2 OFF
50-100Digital printing ON/OFF, zone
recording, and partial
compression/expansion set
value for each channel.

CH PARTIAL
2 50 40.00%
COMMENT
2 = RM10 PEN
3 = 123456789012

Fig. 6.4.2 List Printing

43

Comment printing set
values.
Effective for the DI function(optional).

6.4.3 Engineering List Printing
Engineering list printing provides the following setting data of the instrument on the chart paper.
• Analog recording• Digital printing• Burnout/RJC, etc.
(1) Engineering list printing operating procedure
1) Press the “MENU” key.
2) Use the “PRINT” key to display “”. Then, press the “ENT” key.

3) Use the “PRINT” key to select “”. Pressing the “ENT” key executes engineering list

printing.
Once engineering list printing starts, the display automatically returns to the Data Display screen.
Upon termination of engineering list printing, the display returns to the status prior to starting
engineering list printing.

[Note](1) Analog recording continues while engineering list printing is running. Chart paper is fed.
(2) If an alarm is activated while engineering list printing is running, alarm printing is performed
from the time when engineering list printing finished.

44

(2) Engineering list printing stopping procedure
1) Press the “MENU” key.
2) Use the “PRINT” key to display “”. Then, press the “ENT” key.

3) Use the “PRINT” key to select “”. Pressing the “ENT” key stops engineering list printing.

However, for the line during printing, the printing continues until the line is completely printed.
Then Engineering list printing stops. And the display returns to the status prior to starting
engineering list printing.

RUN/STOPInternal/External
Switching setting:
INT (key)

Channel Number/Tag
Number setting: CH

Alarm Printing setting:ON1 (occurrence/recovery)

ENGINEERING LISTRUN CH/TAG ALM_PR
INT CHON1
D_PR START INT
ON 00:00 1h
SYSC/ASYNC
SYNC
PG_ADJ
ON
Alarm Hysteresis
setting: ON

Logging Printing setting ON/OFF,
Start Time, Interval:
Logging Printing: ON
Start time: 00:00
Interval: 1h
Logging printing Sync/ASync: Sync
PG_ADJ: ON

ALM_HYS TEMPON
Temperature Unit:

CH RJC V(REFCH)
B.OUT OFFSETFILTER
1 INT
ON 10.0
1.00002 EXT 805ON
0.00.8000
COM_ADR 01
SPEED 9600bps
DATA
8bit
PARITY NONESTOP 1bit

Setting of RJC, constant value inexternal RJC, Burnout Protection,
Offset, and Digital filter value foreach channel.
1CH
2CH
RJC
: INT
EXT
REF
: None805 V
B.OUT : ON
ON
Offset : 10.0
0.0Filter: 1.0
0.8

Communication settings

DI 1:COMMENT1
2:MAN_PRINT
3:TIMEPRINT

Fig. 6.4.3 Engineering List Printing

45

DI Function settings.
Effective for the DI
function (optional).

6.4.4 Recording Start/Stop Printing
Recording Start and Stop times are printed on the chart paper when recording starts and stops.
Recording Start/Stop printing can be set in Engineering Mode.
Refer to Section 7.2.6 (6) Recording Start/Stop printing.
[]: Recording Start synchronous printing
Recording Stop printing is asynchronous.
[]: Recording Start/Stop asynchronous printing[]:
Recording Start/Stop time are not printed.
(1) Recording start printing
Press the REC key in the Recording Stop status in order to enable recording status.
Printing will start.
If Recording Start/Stop printing is set to OFF, printing will be disabled.
(2) Recording stop printing
Press the REC key for 4 seconds or longer in the Recording Start status in order to enable
recording stop status.
Printing will start.
If Recording Start/Stop printing is set to OFF, printing will be disabled.

[Note]
If DI function is used for Recording Start/Stop, printing will start by DI function, not via the keypad.

6.4.5 DI Manual Printing (DI option)
DI Manual printing occurs by turning the DI from OFF to ON. Current measured value and time
will be printed on the chart paper.
Printing contents are the same as Section 6.4.1 Manual Printing.
(1) Select Manual Printing of DI function in Engineering Mode.
For detailed setting method, refer to Section 7.2.9 DI Functions.
[MAN-P]: Synchronous Manual printing[AMAN-P]: Asynchronous Manual printing(2) Turn the DI input ON.
DI Manual printing starts by turning the DI input from OFF to ON.

6.4.6 DI Date/Time Printing (DI option)
DI Date/Time printing starts by turning the DI input from OFF to ON. Current date and time will
be printed on the chart paper.
(1) Select Date/Time printing of DI function in Engineering mode.
For detailed setting method, refer to Section 7.2.9 DI Functions.
[TIM-P]: Synchronous Date/Time printing[ATIM-P]: Asynchronous Date/Time printing(2) Turn the DI input ON.
DI Date/Time printing starts by turning the DI input from OFF to ON.

46

6.4.7 DI Comment Printing (DI option)
DI Comment printing starts by turning the DI input from OFF to ON. Comment printing
can be performed on the chart paper.
(1) Select Comment Printing of DI function in Engineering mode.
For detailed setting method, refer to 7.2.9 DI Functions.
[CMNT1], [CMNT2], [CMNT3]: Synchronous comment printing[ACMT1], [ACMT2], [ACMT3]: Asynchronous comment printing(2) Turn the DI input ON.
DI Comment printing starts by turning the DI input from OFF to ON.

Date/Time Printing

May.10.06 20:30
20:10 No.01 OPEN

Comment Printing
Fig. 6.4.7 Date/Time Printing, Comment Printing

47

6.5 Changing the Display
Display selection procedure(1) Press the “MENU” key to display "", then press the “ENT” key.

(2) Use the “PRINT” key to select a required display from the menu below.
Then press the “ENT” key.
(3) "" is displayed for the manual display. Use the “PRINT” key to select a channel number,then press the “ENT” key.
““”(AUTO): Auto display”(MAN):

Manual display“”(DATE): Date display“”(TIME): Time display

“
6.5.1

”(OFF):

Display OFF<Auto Display>

Indicates the process variable of each channel sequentially at intervals of 2.4 seconds.
(Data renewal period: 120 ms.)

Channel Number (1-digit)

H

H : Upper limit Alarm
L : Lower limit Alarm

6.5.2

Process Variable (5-digit)<Manual Display>

Indicates the Process variable of the specific channel, and updates it every measurement
cycle (120 ms).
The data is the same as that of Auto display.
Pressing the “ENT” key changes to the next displayed channel number.
6.5.3<Date Display>

Indicates the month and day. Leap year is adjusted automatically.

(e.g.) October 18
Month
6.5.4

Day

<Time Display>

Indicates the hour and minute.
(e.g.) 12:15
Hour
6.5.5

Minute<Display OFF>Turns off the process variable display. Key operation is the same as usual. To switch to other displays,repeat steps (1) through (3) mentioned above.

48

7． DEVICE SETTING
7.1 Setting the Setup ModeWARNING
When calibrating data, connect the equipment according to Section 9.5 of the instruction manual.
Data calibration must be performed correctly.
Operators possessing knowledge of electricity and control related matters should perform thecalibration after careful reading of Sections 7.2 and 9.5 of the instruction manual.
(Shinko Technos Co., Ltd. is not liable for damages incurred if calibration is performed other thanprescribed in the instruction manual.)
Key Operation for Entering the Setup Mode
Press the “MENU” key for 3 seconds or longer to enter the setup mode.
The software version is indicated for approximate 1 second as below.
After that, the setting screen of the range is indicated.
(e.g.) Version number: 2.10
To return from the setup mode to the recording mode (user mode), press the “MENU” key for 3
seconds or longer again.
The following shows a display map of the setup mode. Use thekey to operate.

7.1.1 Setting the Range
Setting

ChannelMode

Current/Voltage Input
Thermocouple Input
RTD (Resistance temperature detector) Input
Scaling
Square Root
DecadeDifference
Sum
Average
Skip
Option7.1.2 Setting the Alarm
Setting

ChannelLevel

ON/OFF49

Type

Set value

Relay ON/OFF

Relay No.

7.1.3 Setting the Unit
Setting

ChannelChar. No.

Code No.

( )
( )
End

7.1.4 Setting the Chart Speed
Setting

1st/2nd Chart speed

Chart speed

7.1.5 Setting the Date and Time
Setting

Year

Month, DayTime

7.1.6 Copying the Setting Data
Setting

Channel to copy from

Channel to copy to

7.1.7 Setting Other Functions (Zone recording, Partial compression/expansion, Digital printing, Tag,
Comments)
Setting

Mode

Zone Recording
Partial Compression/Expansion
Digital Printing
Tag
Comments
Shift to the engineering mode
Setting

Password

Input a password to enter engeneering mode.
See the key operation on page 76.

50

Table 7.1 Default value of the setup mode
Setting items
Range (all channels)
Alarm (all channels)
Engineering unit (all channels)
Chart speed

Default value
10 mV DC
Scaling 0 to 100.0 ( )All levels alarm OFF, relay OFF(BF 43 00)
1st speed: 20 mm/h
2nd speed: 20 mm/h

Time

2000/01/01 00:00

Zone setting (all channels)

0 to 100 (%)

Partial compression/expansion(all channels)
Digital printing (all channels)
Tag printing character (allchannels)Comments printing (1 to 3)

RemarksOFF
ON“Blank up to the 5th character”“Blank up to the 12th character”

51

The present time has
been set.
(Japan standard time)
GMT + 09:00

7.1.1 Setting the Range(1) Setting method
With a multirange system, setting the range for each channel is possible.
Use thekey to shift the mode mode from 1 to 10 shown in the Table below.

Set the range from the following input signals. (Modes 1 to 3 )DC voltage:

10 mV DC, 0 to 20 mV DC, 0 to 50 mV DC,
1 V DC, 0 to 5 V DC,

Direct current:

200 mV DC,

10 V DC

4 to 20 mA DC (External shunt resistor: 250)

Thermocouple: B, R, S, K, E, J, T, C, Au-Fe, N, PR40-20, PL- , U, L
RTD:

Pt100, JPt100

For the thermocouple and RTD, an optimum range is automatically selected by the set
span point.
Sets scaling, square root, decade, interchannel sum/difference/average. (Modes 4 to 9 )Unnecessary channels can be skipped. (Mode 10 )Setting

ChannelMode

Key(Current/Voltage input)

1(Thermocouple input)

2(Resistance temperaturedetector input)

3
4(Scaling)

5(Square Root)

6(Decade)7(Difference)

8

(Sum)

9

(Average)10(Skip)

Key[Note]
A decimal point position can be set only in the "scaling" and "square root" modes. When altering thedecimal point position in Voltage/Current/Thermocouple/RTD input, set it in the "scaling" mode.
To fix the decimal point position, set it as follows.
Input

Digits after decimal point

Input

Digits after decimal pointmV

2 digits.00

Thermocouple

1 digit.0

1, 0 to 5 V

3 digits

.000

RTD

1 digit.0

10 V

2 digits.00

1 digit.0

2 digits

.00

mA

52

200 mV DC

(2)(Current/Voltage),(Thermocouple),(Resistance Temperature Detector)[Note]: When changing a decimal point position, change it during “(3) Scaling” (pages 55, 56).
Measures Current, Voltage, Thermocouple and RTD.
Example: When setting Thermocouple T for Channel 1 (T: -100 to 300
Display

Operation Keys)

Description

ESC

MENU

Press the MENU key for 3 seconds or longer to enter

3 sec

setup mode. After “ENT“ appears on the display,press the “ENT” key.
PRINT

Use theENT

key to select the desired channel. Press the“ENT” key.

(Zero setting)

PRINT

ENT

Use thekey to select the mode. Press the “ENT” key.

PRINT

ENT

Use thekey to select a type. Press the “ENT” key.

PRINT

FEED

Use thekey to select a sign and numeral.

Use thekey to shift a digit. Press the “ENT” key.

Use thekey to select a sign and numeral.

ENT

Use thekey to shift a digit. Press the “ENT” key.

ENT

Press the “ENT” key. Setting is completed.

ENT

PRINT

(Span setting)

FEED

To return to user mode, press the “MENU” key for 3
seconds or longer.

[Note](1) If an incorrect numeral is entered,oris displayed. Press the “ENT” key and re-enter a correct numeral.
(2) Perform range settings that do not need the scaling at this step of (2),,(Be especially sure to perform the thermocouple and RTD range setting for which decimal
point position change is not necessary) (One digit after the decimal point is standard.)
Do not perform scaling here.

53.

[Note]In,

Use themode, press the “ENT” key to select the range.

, or

key to select a range type.
Mode

Range

“ENT” key

「」

10 mV
0 to 20 mV
0 to 50 mV
200 mV
1V
0 to 5 V
10 V
4 to 20 mA
B
R
S
K
E
J
T
C
Au-Fe
N
PR40-20
PLU
L
Pt100
JPt100[Reference]
The following ranges (input types) have more than one measurement range, and their accuracy isdifferent depending on the measurement range. An optimum range (input type) is automaticallyselected by the set span point.
Thermocouple (
RTD (

):

):

R, K, E, J, T
Pt100, JPt100

54

(3)

(Scaling)

Converts the input of VOLT, TC and RTD to a corresponding value.
Setting the unit is possible. (See Section 7.1.3 Setting the Unit on page 65.)
Example: When setting the voltage of 0 to 40 mV and scaling of 000.00 to 100.00 for Channel 1
Display

Operation Keys
ESC

MENU

Description

Press the “MENU” key for 3 seconds or longer to

3 sec

enter setup mode. "ENT" will appear.

Then press the “ENT” key.
PRINT

Use theENT

key to select the desired channel. Press

the “ENT” key.
PRINT

Use theENT

key to select "".

Press the “ENT” key.
PRINT

Use theENT

key to select "Press the “ENT” key. If “".
” or “” is selected

here, temperature input scaling can be performed.
PRINT

Use theENT

key to select a range (50 mV).

Press the “ENT” key.
PRINT

(Zero input value)

FEED

Use thekey to select a numeral.

Use thekey to shift a digit. Press the “ENT” key.

ENT

[Note]For TC and RTD input, set the Zero input value equal to the Zero-side scale value.
(See 1 on the next page.)

PRINT(Span input value)

FEED

ENT

Use thekey to select a numeral.

Use thekey to shift a digit. Press the “ENT” key.

[Note]For TC and RTD input, set the Span input value equal to the Span-side scale value.
(See 2 on the next page.)55

PRINT

FEED(Zero-side scale value) ( 1)

FEED

(Decimal point)

key to select a numeral.

Use thekey to shift a digit.

Press thePRINT

key again in the lowest digit (at right) to

flash a decimal point. Use thekey to select a decimal

ENT

point position. Press the “ENT” key.

PRINT

(Span-side scale value) ( 2)

Use theFEEDENT

Use thekey to select a numeral.

Use thekey to shift a digit. Press the “ENT” key.

Press the “ENT” key. Setting is completed.

ENT

To return to user mode, press the “MENU” key for 3
seconds or longer.

[Note]
If an incorrect numeral is entered,or,is displayed.

Press the “ENT” key and re-enter a correct numeral.

56

(4)

(Square Root)Caluclates the square root of Volt input, and scales the value.
The unit can be set. (See Section 7.1.3 Setting the Unit on page 65.)
Example: Setting the voltage of 0 to 40 mV and scaling of 000.00 to 100.00 for Channel 1
Display

Operation Keys
ration Keys

Description

ESC

MENU

Press the “MENU” key for 3 seconds or longer to enter

3 sec

setup mode. "ENT" will appear. Then press the“ENT” key.
PRINT

Use theENT

key to select the desired channel. Press the“ENT” key.
". Press the “ENT” key.

PRINT

ENT

Use thekey to select "

PRINT

ENT

Use thekey to select a range (50 mV). Press the“ENT” key.
PRINT

(Zero input value)

Use theFEED

Use the

FEED

PRINT

Press theFEED

key to select a numeral. Use thekey tokey tokey again at the lowest digit (at right) to

flash a decimal point. Use theENT

key to select a decimal

point position. Press the “ENT” key.

PRINT

(Span-side scale value)

key to select a numeral. Use theshift a digit.

(Zero-side scale value)(Decimal point)

key toshift a digit. Press the “ENT” key.

ENT

PRINT

key to select a numeral. Use theshift a digit. Press the “ENT” key.

ENT

PRINT

(Span input value)

Use theFEED

Use theFEED

key to select a numeral. Use thekey toENT

shift a digit. Press the “ENT” key.

ENT

Press the “ENT” key. Setting is completed. To return touser mode, press the “MENU” key for 3 seconds orlonger.

[Note]
If an incorrect numeral is entered,or,is displayed.

Press the “ENT” key and re-enter a correct numeral.

57

About Square Root Computation
The square root computation is as follows:
Each item is defined as follows:
SPANL : Span lower limit value (Span L)
SPANR : Span upper limit value (Span R)
SCALL : Scaling lower limit value (Scale L)
SCALR : Scaling upper limit value (Scale R)
IN: Input voltage

OUT: Output (Scaling value)

When an input value is 1 % or more (1 to 100 %)IN SPANLSPANR SPANLOUT=(SCALR SCALL) x+ SCALL

When an input value is less than 1 %OUT=

10 x （SCALR SCALL）
SPANR SPANL

x (IN SPANL)+ SCALL

Example: When the setting on the previous page is performed, the display shows as follows.

Input voltage (mV)

0

10

20

30

40

Display (%)

0.00

50.00

70.71

86.63

100.00[Note]At near zero point, digital indicating value may not stabilize in relation to scaling factor and display
digits.

58

(5)(Decade)Performs scaling of VOLT input and displays the index number. The unit can be set as well.
(See Section 7.1.3 Setting the Unit on page 65.)
Example: Setting the voltage of 0 to 5 V and decade of 1.0 x 100 to 1.0 x 105 for Channel 1
Display

Operation Keys

Description

ESC

MENU

Press the “MENU” key for 3 seconds or longer to enter

3 sec

setup mode. "ENT" will appear. Then press the“ENT” key.
PRINT

Use theENT

key to select the desired channel. Press the“ENT” key.
PRINT

Use theENT

key to select "". Press the “ENT”

key.
PRINT

Use theENT

key to select a range (5 V). Press the “ENT”

key.
PRINT

Use the

FEED

PRINT

Use the

FEED

PRINT

Use the

FEED

PRINT

(Span-side scale value)

key tokey to select a numeral. Use thekey toshift a digit. Press the “ENT” key.

ENT

(Zero-side scale value)

key to select a numeral. Use theshift a digit. Press the “ENT” key.

ENT

(Span input value)

key toshift a digit. Press the “ENT” key.

ENT

(Zero input value)

key to select a numeral. Use theUse theFEED

key to select a numeral. Use the

ENTshift a digit. Press the “ENT” key.

ENT

Press the “ENT” key. Setting is completed.

key toTo return to user mode, press the “MENU” key for 3
seconds or longer.

[Note]
If an incorrect numeral is entered,
or

.

,

is displayed.

Press the “ENT” key and re-enter a correct numeral.

59

About Decade Display
Each item is defined as follows:
SPANL : Span lower limit value (Span L)
SPANR : Span upper limit value (Span R)
SCALL : Scaling lower limit value (Scale L)
SCALR : Scaling upper limit value (Scale R)
XXEYY
XX: Mantissa section (1.0 to 9.9)
YY: Exponent section (-19 to 19)
OUT : Output (Scaling value)
Up to 5 decades.
[(Scaling upper limit) – (Scaling lower limit)
is 1.0E5 or less.]
Decade display adheres to the following relational expression.
IN: Input voltage

OUT1 = INx

OUT = 10OUT1LGSCALR LGSCALLSPANR SPANL+SPANR x LGSCALL SPANL x LGSCALRSPANR SPANLLGSCALL : Log10(SCALL)LGSCALR : Log10(SCALR)[Note]
Decade output is used for display and printing. It is not reflected in pen recording.

Example: When the setting on the previous page is performed, the display is shown as follows.

Input voltage (V)

0.0

1.0

2.5

3.0

5.0

Display

1.0E0

1.0E1

3.2E2

1.0E3

1.0E5

60

(6)

(Difference),(Sum),(Average)

Caluclates the input of VOLT, TC, RTD or SCALE, and output it.
Example: Subtracting Ch1 input data from Ch2 input data to set the difference value to Ch2.
(0 to 40 mV). “Ch2 (input) - Ch1 (input)” is recorded and displayed on Ch2.

Display

Operation Keys

Description

ESC

MENU

Press the “MENU” key for 3 seconds or longer to enter

3 sec

setup mode. "ENT" will appear. Then press the“ENT” key.
PRINT

Use theENT

key to select the desired channel. Press the“ENT” key.
(Setting channel)". Press the “ENT” key.

PRINT

ENT

Use thekey to select "

PRINT

ENT

Use thekey to select the "channel you want tosubtract." Press the “ENT” key.
(Reference channel)
PRINT(Zero input value
after difference operation)

key to select a numeral. Use thekey toshift a digit. Press the “ENT” key.

ENT

PRINT

(Span input value
after difference operation)

Use theFEED

Use theFEED

key to select a numeral. Use thekey toshift a digit. Press the “ENT” key.

ENT

Press the “ENT” key. Setting is completed.

ENT

To return to user mode, press the “MENU” key for 3
seconds or longer.

[Note]
When an incorrect numeral is entered,,oris displayed.
Press the “ENT” key and re-enter a correct numeral.
About Difference, Sum and Average
The setting channel must be larger than the one which computes the "difference, sum or average"
value. Set up only Channel 2.
The range and scale of the channel to be set is the same as those of the reference channel which
computes the "difference, sum or average" value.
A range value after computation cannot exceed the maximum range of the instrument.
For the reference channel, only the voltage, current, thermocouple, RTD, and their scaling ranges
can be set.

61

(7)

(Skip)Set SKIP to the channel for which display and recording are not required.
Example: When skipping Channel 2

Display

Operation Keys

Description

ESC

MENU

Press the “MENU” key for 3 seconds or longer to enter

3 sec

setup mode. "
ENT

PRINT

" will appear. Then press the“ENT” key.
Use theENT

key to select the desired channel. Press the“ENT” key.
PRINT

Use theENT

key to select "". Press the “ENT” key.

Press the “ENT” key. Setting is completed.
To return to user mode, press the “MENU” key for 3

ENT

seconds or longer.

[Note]• If all channels are set to Skip,

is displayed.

At least one channel should be set to Measurement.
• Measurement can be conducted on a skipped channel, however, indication, printing or alarmassessment cannot be performed. If input is not connected, be sure to short-circuit the input
terminals.
• To return the skipped settings to the measurement settings, set the measurement range again.

62

7.1.2 Setting the Alarm
Setting items
The following 2 alarm types can be set to each channel.
Up to 4 alarm points (4 levels) can be set per channel.
Once alarm point is set, the “ALM” lights when PV (process variable) reaches alarm point, andsimultaneously the alarm is printed on the chart paper, indicating an alarm occurrence.
: Upper limit alarm When the process variable is higher than the alarm set point,an alarm is issued.
: Lower limit alarm When the process variable is lower than the alarm set point,an alarm is issued.
OptionOperation
Setting

ChannelLevel

ON/OFFType

Set Value

Relay ON/OFF

Relay No.

Example: When setting Channel 1 alarm point (Level 1) to the following:
Upper limit alarm, alarm set point of -2.000, Alarm output relay No. 1

Display

Operation Keys

Description

ESC

MENU

3 sec

PRINT

ENT

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "".

Press the “ENT” key.
PRINT

Use theENT

key to select the desired channel. Press the“ENT” key.
PRINT

Use theENT

key to select an alarm level. Press the “ENT”

key. Up to 4 levels can be set.
PRINT

Use theENT

key to select alarm setting ON. Press the“ENT” key. Setting to "" displays ""and completes setting.
PRINT

Use theENT

key to select an alarm type. Press the “ENT”

key.
PRINT

Set an alarm point. Use theFEED

use theENT

key to shift a digit, andkey to select a numeral. Once setting iscompleted, press the “ENT” key. A decimal point positionis fixed depending on the range.

[Note]
When the “ENT” key is pressed during the alarm point setting process, display will switch to the next
setting display.

63

Setting hereinafter is valid only for the model for which an alarm output (option: LH3) has beenordered. If the option has not been ordered, press the “ENT” key until "" is displayed.

Then, press the “ENT” key again. Setting is completed.
Display

Operation Keys
PRINT

Description

When an alarm occurs, the alarm output is issued fromENT

the alarm output relay. Use thekey to select alarm

output ON, and press the “ENT” key.
If the alarm output is not used, select OFF, and press the“ENT” key.

PRINT

If the alarm output is set to ON, set the relay number.

ENT

Use thekey to select the relay number out of 1 to 3,

and press the “ENT” key.

Press the “ENT” key. Setting is completed.

ENT

To continue to set other channels, press the

key toselect the channel, and start setting. To return to theuser mode, press the “MENU” key for 3 seconds orlonger.

[Note]For Alarm printing, the channel number is printed even if the tag printing has been set.
.

64

7.1.3 Setting the Unit
Setting Items
Sets the input unit for each channel.

[Note]To change the unit in the range ofOperation

Setting,or

Channel, set the range toChar. No.

.

Code No.

( )
( )
End

Example: Setting the unit ( ) for Channel 1
Display

Operation Keys

Description

ESC

MENU

3 sec

PRINT

ENT

PRINT

ENT

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "".

Then press the “ENT” key.
Use thekey to select the desired channel. Then,press the “ENT” key.

PRINT

Use theFEED

key to select the 1st character of thecode number, and press the “ENT” key.

ENT

(Code No.)or(See the Character Code Table on the next page.)
PRINT

Set the 2nd character in the same manner.

FEED

ENT

When the number of unit characters is less than 6, enter
PRINT

FEED

the code (ENT) to finish. Press the “ENT” key.

Press the “ENT” key. Setting is completed.

ENT

To return to user mode, press the “MENU” key for 3
seconds or longer.

[Note]The unit setting is effective only when the range setting isand(including

,

and

,
when

,

,

is selected as a range

setting of the reference channel).
For other range settings, the unit is automatically determined according to the range.

65,

(1) Character Code Table

2

3

4

5

6

0

SP

0

@

P

1

!

1

A

Q

2

"

2

B

3

#

3

4

$

5

7

A

B

p

0

a

q

1

R

b

r

2

C

S

c

s

3

4

D

T

d

t

4

%

5

E

U

e

u

5

6

&

6

F

V

f

v

6

7

'

7

G

W

g

w

7

8

(

8

H

X

h

x

8

9

)

9

I

Y

i

y

9

:

J

Z

j

z

[

k

{

l

A

C

D

E

F

Π

0

π

1

A

P

α

ρ

2

B

Σ

β

σ

3

Γ

T

γ

τ

Y

δ

υ

4
5

E

Φ

ε

φ

6

Z

X

ζ

χ

7

H

Ψ

η

ψ

8

Θ

θ

ω

9

I

ι

K

κ

Λ

λ

｜

M

μ

N

ν

-

Ξ

ξ

ｏ

Ο

o

B

+

;

K

C

,

<

L

D

–

=

M

]

m

}

E

.

>

N

＾

n

￣

F

/

?

O

＿

o

+

ｏ

Setting example: The code number for "C" is "43".

66+

7.1.4 Setting the Chart Feed Speed
Setting Items
Sets the chart feed speed. Select it from the table below.
Table 7.1.4 Chart feed speed (Unit: mm/h)
5
80
360
2400
12000

10
90
375
3000

15
100
450
3600

20
120
600
4500

25
150
720
4800

30
160
750
5400

40
180
900
6000

50
200
1200
7200

60
240
1500
9000

75
300
1800
10800

Operation
Setting

1st/2nd Chart speed

Chart speed

Example: When setting the 1st chart feed speed to 1500 mm/h

Display

Operation Keys

Description

Press the “MENU” key for 3 seconds or longer to enter

ESC

MENU

3 sec

PRINT

ENT

press the “ENT” key.

PRINT

ENT

Use thesetup mode. Use thekey to display "", andkey to select the 1st chart feed speed, andpress the “ENT” key.

[Note]To alter the chart feed speed with the DI function (option: RE1), set the 2nd chart feed speed.
If the DI function (option: RE1) is not ordered, the 2nd chart feed speed setting becomes invalid.

Display

Operation Keys

PRINT

Description

Use theENT

key to set the 1st chart feed speed, and press

the “ENT” key.

Press the “ENT” key. To return to user mode, press the

ENT“MENU” key for 3 seconds or longer.

67

［Note］

Chart speed(mm/h)

Printing type

Restriction of printing

Alarm occurrence printing, Alarm recovery
printing, Affix printing, DI Comment printing(Synchronous), DI Date/Time printing
5 to 100 (mm/h)(Synchronous), DI Manual printing (Synchronous),
Recording Start printing (Synchronous)

by chart feed speed:

Logging printing (Synchronous)

10 to 100 (mm/h)

When the chart feed speed is 0 mm/h, it is printed with forcible chart feed. (When the chart speed ishigher than 120 mm/h, the Table shown above is not printed.) List Printing, Engineering List Printing,
Manual Printing (key), DI Comment printing (Asynchronous), DI Date/Time printing (Asynchronous), DI
Manual printing (Asynchronous), Logging printing (Asynchronous) and Recording Start/Stop printing(Asynchronous) are printed regardless of the chart feed speed.
7.1.5 Setting the Date and Time
Setting Items
The following describes how to set the date/time of the internal clock.
Setting

Year

Month, DayTime

Example: When setting the date and time to January 1st, 2000, 6:00.
Display

Operation Keys
ESC

MENU

3 sec

PRINT

ENT

PRINT

FEED

Description

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "", andpress the “ENT” key.

Set the year. Use theuse theENT

key to select a numeral, andkey to shift a digit.

Press the “ENT” key.

PRINT

Set the month and day.

FEED

Follow the same process as above.

ENT

PRINT

Set the time (Hour:Minute).

FEED

ENT

Follow the same process as above.

ENT

Press the “ENT” key. To return to user mode, press the“MENU” key for 3 seconds or longer.

[Note]
If a non-existent date or time is entered,is displayed.

Press the “ENT” key and re-enter a correct numeral.

687.1.6 Copying the Setting Data
Setting Items
The following figure describes how to copy the setting data of Channel 1 to Channel 2.

Operation
Setting

Channel to copy from Channel to copy to

Example: When copying the setting data of Channel 1 to Channel 2.
Display

Operation Keys

Description

ESC

MENU

3 sec

PRINT

ENT

Press the MENU key for 3 seconds or longer to enter
setup mode. Use thekey to display "", andpress the “ENT” key.
PRINT

Use theENT

key to select the channel to copy from.

Press the “ENT” key.
PRINT

Use theENT

key to select the channel to copy to, andpress the “ENT” key.

Press the “ENT” key. To return to user mode, press the

ENT“MENU” key for 3 seconds or longer.

[Note]
Copy the setting data of Channel 1 to Channel 2.
Copying from Channel 2 to Channel 1 is impossible.

697.1.7 Setting Other Functions
Other functions can be set as the following.
Setting Items(1)(Zone recording)

The data for each channel can be recorded separately in another area so as not to overlap.
(2)(Partial compression/expansion)

Records the measured data of the channel, partially compressed or expanded.
(3)(Digital printing)Specify ON or OFF whether the measured data for each channel is printed or not.
(4)

(Tag)

Sets the tag, which is to be printed instead of a channel number, for logging printing or manual
printing.
For tag setting, up to 5 characters can be used (See page 66 for Character Code Table) for eachchannel.
(5)(Comments)

Sets the comments to be printed by the DI function (option: RE1).
There are 3 kinds of comments, and setting up to 12 characters (See Page 66 for Character Code
Table) for each channel is possible.

Operation
Setting

Mode

Zone Recording
Partial Compression/Expansion
Digital Printing
Tag
Comments70

(1)(Zone Recording)

Example: Setting the instrument so that Channel 1 zero/span will be recorded at the position of
20 to 50 %Display

Operation Keys
ESC

MENU

3 sec

PRINT

ENT

PRINT

ENT

Description

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "", andpress the “ENT” key.
Use thekey to display "", and press the “ENT”

key.
PRINT

Use theENT

key to select the desired channel, and press

the “ENT” key.
PRINT

(Left-end Position setting)Set a zero point printing position in percentage (%).

FEED

Use theENT

key to select a numeral, and use thekey to shift a digit. Press the “ENT” key.
PRINT

Set a span point printing position in percentage (%).

FEED

Use the same process as the zero point printing

ENT(Right-end Position setting)position setting.
Press the “ENT” key. Setting is completed.

ENT

To return to user mode, press the “MENU” key for 3
seconds or longer.

[Note]
If an incorrect numeral is entered,oris displayed. Press the “ENT” key and re-enter a correct numeral.

71

(2)

(Partial Compression/Expansion)

Example: Setting Channel 1 scale of “0 to 1000.0

Display” to 500.0

0

500.0

1000.0(0 %)(30 %)(100 %)

Operation Keys

at a boundary point of 30 %Description

ESC

MENU

3 sec

PRINT

ENT

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "", andpress the “ENT” key.
PRINT

Use theENT

key to select "", and press the “ENT”

key.
PRINT

Use theENT

key to select the desired channel, and press

the “ENT” key.
ENT

Use the

PRINT

FEED

Set a boundary point position in percentage (%).
Use theENT

key to select "", and press the “ENT” key.

PRINT

key to select a numeral, and use thekey to shift a digit. Press the “ENT” key.
PRINT

Set a boundary point position to "0500.0". Use the

FEED

key to select a numeral, and use theENT

key to shift adigit. Press the “ENT” key.
Press the “ENT” key. Setting is completed.

ENT

To return to user mode, press the “MENU” key for 3
seconds or longer.

[Note]
If an incorrect numeral is entered,is displayed.

Press the “ENT” key and re-enter a correct numeral.

72

(3)

(Digital Printing)

Example: Setting the channel data printing (logging printing, manual printing) to "ON" (printing
enabled) for all channels
Display

Operation Keys

Description

ESC

MENU

3 sec

PRINT

ENT

PRINT

ENT

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "", andpress the “ENT” key.
Use thekey to display "", and press the“ENT” key.
PRINT

Use theENT

key to select the desired channel, and press

the “ENT” key.
PRINT

Use theENT

key to switch from "" to "", and press

the “ENT” key.
Press the “ENT” key. Setting is completed.

ENT

To return to user mode, press the “MENU” key for 3
seconds or longer.
PRINT

ENT

Repeat the settings for channel 2.
ENT[Note]
When the channel, for which skip is selected during the range setting, is set ON, " " is printed.

73

(4)

(Tag)

Example: Setting "AB" for Channel 1

Display

Operation Keys

Description

ESC

MENU

3 sec

PRINT

ENT

PRINT

ENT

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "", andpress the “ENT” key.
Use thekey to display "", and press the “ENT”

key.
PRINT

Use theENT

key to select the desired channel, and press

the “ENT” key.
PRINT

(Setting of 1st Character)key to select a

key to shift a digit. Press the “ENT”

key.
Set the 2nd character in the same manner.

FEED

ENT

PRINT

(Setting of 3rd Character)" as the 1st character. Use thecode, and use the

ENT

PRINT

(Setting of 2nd Character)

Set "

FEED

Set "

FEED

" as the 3rd character and press the “ENT” key.

ENT

Press the “ENT” key. Setting is completed. To return to

ENTuser mode, press the “MENU” key for 3 seconds orlonger.

[Note]
When the number of tag characters is less than 5, add "See page 66 for Character Code Table.

74" code at the last character.

(5)(Comments)

Example: Setting "ON" for Comment 1 (Display)

Operation Keys

Description

ESC

MENU

3 sec

PRINT

ENT

PRINT

ENT

Press the “MENU” key for 3 seconds or longer to enter
setup mode. Use thekey to display "", andpress the “ENT” key.
Use thekey to display "", and press the “ENT”

key.
PRINT

PRINT

(Setting of the 1st Character)

Use theENT

FEED

key to select a comment number. Display"" and press the “ENT” key.

Set "" as the 1st character. Use thecode, and use theENT

key to select a

key to shift a digit. Press the “ENT”

key.
PRINT

(Setting of the 2nd Character)" as the 2nd character.

Use the same process as the 1st character setting.

ENT

PRINT

(Setting of the 3rd Character)

Set "

FEED

Set "

FEED

" as the 3rd character.

Use the same process as the 1st character setting.

ENT(Setting of the 9th Character)
Repeat the settings up to the 12th Character if required.

(Setting of the 10th Character)(Setting of the 11th Character)(Setting of the 12th Character)
Press the “ENT” key. Setting is completed. To return to

ENTuser mode, press the “MENU” key for 3 seconds orlonger.

[Note]• When the number of tag characters is less than 12, add "" code next to the last character.

See page 66 for Character Code Table.
• If DI (Option: RE1) is not ordered, comment printing does not function.
Do not set the comment printing.

75

7.2 Setting Engineering ModeWARNING
When calibrating data, connect the equipment as written in Section 9.5 of the instruction manual.
Data calibration must be performed correctly.
Operators possessing knowledge of electricity and control related matters should perform thecalibration after careful reading of Sections 7.2 and 9.5 of the instruction manual.
(Shinko Technos Co., Ltd. is not liable for any damages incurred if calibration is performed
other than described in the instruction manual.)

Key Operation to Enter Engineering Mode
Press the “MENU” key for 3 seconds or longer to enter setup mode.
", and press the “ENT” key then to display "0000".

Use thekey to select "Use thekey to alter a numeral, and thethen press the “ENT” key. "key to shift a digit. Enter the password "2222"," will be flashing.

Press the “ENT” key. After reset, the mode becomes engineering mode.

The following figure shows a display map of the Enginneering mode. Use thekey to operate.

7.2.1 Alarm Hysteresis
Setting

ON/OFF

Sets the hysteresis width
Cancels the hysteresis width

7.2.2 Burnout ON/OFF
Setting

ChannelON/OFF

Sets burnout.
Cancels burnout.

7.2.3 Channel Offset
Setting

ChannelOFFSETEnters an offset value

7.2.4 Reference Junction Compensation
Setting

ChannelFunctionInternal Compensation
External Compensation
External channel Input Compensation

767.2.5 Digital Filter
Setting

ChannelFunction

Digital Filter value

7.2.6 Settings Related to Recording
SettingFunction

Recording start/stop INT/EXT selection
Channel/Tag printing selection
Alarm printing ON/OFF
Logging printing ON/OFF
Printing gap correction ON/OFF
Recording Start/Stop printing (OFF/Synchronous/Asynchronous)

Recording start/stop INT/EXT selection
7.2.7 Setting the Communication Function
Tag/Channel printing selection
Setting
Setting the Communication
Alarm printing Function
ON/OFF
ProtocolLogging
printing ON/OFF
Local
Printing
gapAddress
correction ON/OFF
Communication
speed(OFF/Synchronous/Asynchronous)
RecordingStart/Stop printing
Data Length
Parity Bit
Stop Bit
Recordingstart/stop INT/EXT selection
Tag/Channel printing selection
printing
ON/OFF
7.2.8 Initializing the Setup Data and Alarm
Calibration
Data
Logging printing ON/OFF
Setting
YES/NO
Printing gap correction ON/OFF
Recording Start/Stop printing (OFF/Synchronous/Asynchronous)
Initializes Setup mode.
Returns
to thecalibration
valueselection
at the time of shipment.
Recordingstart/stop
INT/EXTTag/Channel printing selection
7.2.9 DI Functions
Alarm printing ON/OFF
Setting
DI No.
Function
Logging printing ON/OFF
Printing gap correction ON/OFF
Recording Start/Stop
printing (OFF/Synchronous/Asynchronous)
No function
RUN/STOP trigger
Chart feed speed selection
Recording start/stop
INT/EXT
selection (Synchronous)
Comment
1 printing
Tag/Channel printing
selection
Comment
2 printing (Synchronous)
Alarm printing ON/OFF
Comment 3 printing (Synchronous)
Logging printing
ON/OFF
Manualprinting(Synchronous)
Printing gap correction
ON/OFF
Date andtime printing (Synchronous)
Recording Start/Stop
printing(OFF/Synchronous/Asynchronous)
Comment
1 printing(Asynchronous)
Comment 2 printing (Asynchronous)
Comment 3 printing (Asynchronous)
Recording start/stop
INT/EXT
selection (Asynchronous)
Manualprinting
Tag/Channel printing
selection
Date andtime printing (Asynchronous)
Alarm printing ON/OFF
Logging printing ON/OFF
Printing gap correction ON/OFF
7.2.10 Temperature Unit selection Recording Start/Stop printing (OFF/Synchronous/Asynchronous)
Setting

Temp Unit

Recording start/stop INT/EXT selection
Centigrade
Unitprinting selection
Tag/Channel
Alarm printing
Fahrenheit
Unit ON/OFF
Logging printing ON/OFF
77
Printing gap correction ON/OFF
Recording Start/Stop printing (OFF/Synchronous/Asynchronous)

7.2.11 Point Calibration
SettingFunction

Zero point calibration
Span point calibration

7.2.12 Data Calibration
SettingZero point calibration
Span point calibration

Function

Voltage calibration
RTD (Resistance temperature detector) calibration
Internal reference junction compensation calibration

7.3 Terminating the Engineering Mode
SettingFunction

Saves the set value.
Cancels the set value.
スパ[Note]
ン点調整
When terminating Engineering mode, if "selecting "左" is selected and the power was turned off without", the setting will become invalid. For "" operation, see Section 7.3 on page93.
右移動時のヒステリシス（工Table
7.2 Default value of engineering mode
場設定項目）
Setting items
Alarm hysteresis
リ ボchannels)
Burnout (allChanneloffset (all channels)
ンセレクト調整RJC (all channels)（工場設定項
Digital filter value
目REC (recording):
RUN/STOP trigger
Channel/Tag printing
Alarm printing
Logging Printing
Logging Printing interval
Logging Printing reference time
Logging Printing Sync./ASync.
Printing gap correction
Recording Start/Stop printing
COM (communication):ProtocolAddress
Communication speed
Data length
Parity bit
Stop bit
DI function (Ch1 to Ch3)
Temperature unit

Default value

Remarks

ON (0.5 %)OFF
0.0Internal compensation INT
K=1.0000
INT
Ch
OFF
ON
6H
00:00
SYNC (Synchronous)OFFOFF
Original
01
9600
8 bits
None
1 bit
OFF
(centigrade unit)
78

Use the “REC” key.

7.2.1 Alarm Hysteresis
0.5 % hysteresis can be set as a value upon alarm activation and alarm recovery. (Common to all
alarms)Example: Turning alarm hysteresis OFF.
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT

PRINT

Description

Enter Engineering Mode” on p.76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to display "", and press the “ENT”

key.

7.2.2 Burnout ON/OFF
Burnout High for each channel (to deflect the recording over the span) can be set.
Example: Setting Burnout High for Channel 2
Display

Operation Keys

Description

Enter the engineering mode (See the “Key Operation to

ENTEnter Engineering Mode” on page 76). Use thePRINT

display "PRINT

key toENT

Use theENT", and press the “ENT” key.
key to display "", and press the “ENT”

key to select "", and press the “ENT” key.

key.
PRINT

Use theENT

79

7.2.3 Channel Offset
For channel offset setting, set the value to be added to an actual process variable.
The input value can be set from -19999 to 99999.
Example: Adding the value (3.0) to Channel 2
Display

Operation Keys

Description

Enter engineering mode (See the “Key Operation to

ENTEnter Engineering Mode” on page 76). Use thePRINT

display "PRINT

key toENT

Use theENT", and press the “ENT” key.
key to select "", and press the “ENT”

key.
PRINT

Use theFEED

key to alter a sign/numerical value or shift a

decimal point. Use theENT

key to shift a digit. Once thenumerical value is determined, press the “ENT” key.

7.2.4 Reference Junction Compensation
The following describes how to set the method for compensating an electromotive force generated
between a thermocouple wire or compensation lead wire and a terminal. There are threecompensating methods as follows:• A method of compensation using the built-in temperature sensing element (INT: Internal
compensation).
• A method of compensation by keeping an external compensator’s temperature constant (EXT:
External compensation). For external reference junction compensation voltage, -19999 to 19999 V
can be set as an input.
• By connecting the terminal of the external junction box to one input of the recorder, compensates the
input of the other channel. (CH: Channel input compensation)
For this recorder, internal compensation (INT) default value has been set.
Example: Setting the external compensator’s temperature constant voltage for Ch2 to 391 V(Thermocouple T input, external compensator’s compensating temperature 10 ,Electromotive force input 391 V)
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select "", and press the “ENT”

key to select "", and press the “ENT”

key.
PRINT

Use the

ENT

key.
PRINT

Use theFEED

key to alter a numerical value, and the

key to shift a digit. Once the numerical value is

ENTdetermined, press the “ENT” key.
Once setting is completed, the recorder will return tochannel setting mode.
80Example: When Channel 1 input compensates Channel 2 reference junction
Display

Operation Keys

Description

Enter engineering mode (See the “Key Operation to

ENTEnter Engineering Mode” on page 76). Use thePRINT", and press the “ENT” key.

display "PRINT

key toENT

Use theENT

key to select "", and press the “ENT”

key.
PRINT

ENT

Use thekey to select "", and press the “ENT” key.

PRINT

ENT

Use thekey to select "", and press the “ENT”

key.
Once setting is completed, the recorder will return toChannel setting mode.

7.2.5 Digital Filter
Apply a digital filter to a process variable to record the first lagged value.
From 99.5 % response time (TI), the filter constant (K) is obtained by the following formula.
K=

0.64
TI

Example: When 99.5 % response time is 50 seconds; K=
Display

Operation Keys

=0.0128

Description

Enter engineering mode (See the “Key Operation to

ENT
PRINT

0.64
50

Enter Engineering Mode” on page 76).

ENT

Use thekey to select “” and press the “ENT”

key.
PRINT

Use theFEED”.

Press the “ENT” key.

ENT

PRINT

key to select digital filter value “Setting is completed. Press the MENU key for 3 seconds

ENT

or longer to return to user mode.

81

7.2.6 Settings Related to Recording
Recording RUN/STOP trigger, channel/tag printing, alarm printing, logging printing, scale printing
ON/OFF and Recording start/stop printing can be set.

(1) Recording RUN/STOP Trigger Setting
Recording RUN/STOP trigger can be set by the “REC” key or by the DI.
Example: When setting Recording RUN/STOP trigger by the DI.
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select "", and press the

key to display "", and press the “ENT”“ENT” key.
PRINT

Use theENT

key.
Once setting is completed, the recorder will return to the
Recording RUN/STOP Setting screen.

[Note]
Default value:(REC key)
If Recording Start/Stop is set to “
(EXT)” as in the example above, select “RUN/STOP trigger(RCD)” as the DI function. See pages 87, 88 for the DI function.
If "" is selected, the “REC” key will not function.

(2) Channel/Tag Printing Selection
In the case of logging printing or manual printing, this sets tag printing or channel printing.
Example: Setting the tag printing
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select "", and press the “ENT”

key to display "", and press the “ENT”

key.
PRINT

Use theENT

key.
Once setting is completed, the recorder will return toChannel/Tag setting screen.

82

(3) Alarm Printing ON/OFF
Sets the alarm printing ON/OFF.
ON1: Prints the alarm occurring and alarm recovering.
ON2: Prints only the alarm occurring.
Example: When setting both alarm occurring and alarm recovering printings
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display “PRINT

Use theENT

key to”, and press the “ENT” key.
key to select “”, and press the

key to display “”, and press the “ENT”“ENT” key.
PRINT

Use theENT

key.
Once setting is completed, the recorder will return to
Alarm Printing Setting screen.

(4) Logging Printing ON/OFF
Sets the logging printing ON/OFF.
In the case of logging printing ON, this sets the printing start time and printing interval.
Logging printing Synchronous/Asynchronous can be set as well.

Setting

ON/OFF

Printing Start time

83

Interval

Sync/ASync

Example: Setting the logging printing Synchronous to start at 18:00 every day
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT", and press the “ENT” key.

display "PRINT

Use theENT

key tokey to select "", and press the

key to display "", and press the “ENT”“ENT” key.
PRINT

Use the

ENT

key.
PRINT

(Printing start time)

Use theFEED

key to shift adigit. Then, set the time and press the “ENT” key.

ENT

PRINT

key to alter a numeral, andUse theENT", and press the “ENT”

key to display "key. (Select from 10 min, 15 min, 20 min, 30 min, 1H, 2H,(Interval)

3H, 4H, 6H, 8H, 12H, 24H)
PRINT

SelectENT

with thekey, and press the “ENT” key.

(Synchronous/Asynchronous)
Once setting is completed, the recorder will return to the
Logging Printing setting screen.

(5) Printing Gap Correction ON/OFF
Sets printing gap correction. This function is available only for DI asynchronous printing andasynchronous logging printing.
Example: When setting the printing gap correction to ON.
Display

Operation keys

Enter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select "", and press the

key to display "", and press the “ENT”“ENT” key.
PRINT

Use theENT

key.
Once setting is completed, the recorder will return to thePrinting gap correction setting screen.

84

(6) Recording Start/Stop Printing (OFF/Synchronous/Asynchronous)
Prints recording start and stop time on the chart paper.
For Recording Start/Stop printing, OFF, Synchronous or Asynchronous can be selected.
Example: Recording Start/Stop Synchronous printing
Display

Operation keys

Enter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select "", and press the

key to display "", and press the “ENT”“ENT” key.
PRINT

Use theENT

key.
Once setting is completed, the recorder will return to the
Recording Start/Stop Printing Setting screen.

Selection item:(Recording Start/Stop OFF): Recording Start/Stop is not printed.
(Recording Start/Stop Synchronous printing):
Recording Start is printed at preset chart speed. However, Recording Stop can be printed
regardless of the preset Chart feed speed.
(Recording Start/Stop Asynchronous printing):
Recording Start/Stop is printed regardless of the preset Chart feed speed.
How to print Recording Start:
Press the REC key after recording stops.
Recording Start time will be printed automatically.
It will not be printed if “Recording Start/Stop “OFF” has been set.
How to print Recording Stop:
Stop recording by pressing the REC key for 3 sec or longer.
Recording Stop time will be printed automatically.
It will not be printed if Recording Start/Stop “OFF” has been set.
DI function:
If DI function is used for Recording RUN/STOP, Recording Start/Stop cannot be performed
using the REC key.
Perform Recording RUN/STOP using external DI terminals.
Printing sample:
Recording Stop printing(e.g.) 2006.5.10 20:00

May.10.06 20:00End

Recording Start printing(e.g.) 2006.5.10 19:00

May.10.06 19:00Start

85

7.2.7 Setting the Communication Function
Sets Protocol, Local Address, Communication speed, Data Length, Parity Bit and Stop Bit.
Example: Protocol: Original, Local Address: 02, Communication speed: 1200 bps,
Data Length: 7 bits, Parity Bit: Even, Stop Bit: 2 bits
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display “PRINT

Use theENT

key to”, and press the “ENT” key.
key to display "", and press the“ENT” key. (Select Original or MODBUS RTU.)(Protocol)
PRINT

Use the

ENT(Local address)
PRINT

ENT

key to display "", and press the“ENT” key. (Select fromto

Use the", and press the “ENT”

key to display ")key. (Select 1200, 2400, 4800, 9600, 19200 or 38400.)(Communication speed)
PRINT

Use theENT

key to display "", and press the “ENT”

key. (Select 7 bits or 8 bits.)(Data length)
PRINT

Use theENT

key to display "", and press the “ENT”

key. (Select from EVEN, ODD and NONE)(Parity bit)
PRINT

length)Use theENT

key to display "", and press the “ENT”

key. (Select 1 bit or 2 bits.)(Stop bit)

Once setting is completed, the recorder will return tolength)Communication Function Setting screen.

[Note]For MODBUS RTU (.

), be sure to set the Data length to 8 bits ().

7.2.8 Initializing the Setup Data and Calibration Data
If “
” is selected, the calibration data is not initialized, but the setup data reverts to the default
value. If “
” is selected, the calibration data reverts to the default value, but the setup data does
not initialize. Take extra caution when executing this function.
Example: Initializing the Setup Data.
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT", and press the “ENT” key.
key to select "

key.
PRINT

Initialization is completed.

ENT

86

key to", and press the “ENT”

7.2.9 DI Functions
Allocates functions to 3 DI terminals. There are 12 DI functions as follows.
No function:

Manual printing (Sync.):(Positive edge: Printing starts)

Manual printing (ASync.):(Positive edge: Printing starts)

RUN/STOP trigger:(ON: RUN, OFF: STOP)

Date/Time printing (Sync.):(Positive edge: Printing starts)

Date/Time printing (ASync.):(Positive edge: Printing starts)

Chart feed speed selection:(ON: Spd-1, OFF: Spd-2),,

Comment printing 1 to 3 (Sync.) :Comment printing 1 to 3 (ASync.) :,,(Positive edge: Printing starts)(Positive edge: Printing starts)

About DI function
By closing 3 DI terminals (D1, D2, D3) and the COM terminal, the following operations can be
carried out.
Setting method(1) Manual printing (Sync.)(MAN-P)

Temporary channel value, date/time are printed on the chart paper.
Printing channel will be determined depending on whether the Digital printing function isturned ON or OFF in Setup mode.
Refer to Section “7.1.7 Setting other functions: (4) Digital printing”.
Preset Chart feed speed is used for printing.
Printing starts upon closing the DI terminals and the COM terminal (ON).
(2) Manual printing (Async.)(AMAN.P)

Temporary channel value, date and time are printed on the chart paper.
Printing channel will be determined depending on whether the Digital printing function isturned ON or OFF in Setup mode.
Refer to Section “7.1.7 Setting other functions: (4) Digital printing”.
Temporary channel value, date/time are printed regardless of the preset Chart feed speed.
During printing, measured value recording will not be performed.
Printing starts upon closing the DI terminals and the COM terminal (ON).
(3) RUN/STOP trigger(RCD)

Recording RUN/STOP can be operated externally by DI terminals.
Recording starts upon closing the DI terminals and the COM terminal (ON).
Recording stops upon opening the DI terminals and the COM terminal (OFF).
(4) Date/Time printing (Sync.)(TIM-P)

Current date and time are printed on the chart paper.
Preset Chart feed speed is used for printing.
Printing starts upon closing the DI terminals and the COM terminal (ON).
(5) Date/Time printing (Async.)(ATIM.P)

Current date and time are printed on the chart paper.
Date and time will be printed regardless of the preset Chart feed speed.
During printing, measured value recording will not be performed.
Printing starts upon closing the DI terminals and the COM terminal (ON).

87(6) Chart feed speed selection(SPEED)

Chart feed speed can be operated in 2 stages by DI terminals. Refer to Section “7.1.4 Setting
the Chart feed speed”. The 1st chart feed speed can be selected by closing the DI terminals
and the COM terminal (ON). The 2nd chart feed speed can be selected by opening the DI
terminals and the COM terminal (OFF).
(7) Comment printing 1 to 3 (Sync.),,(CMNT 1 to 3)

The preset comments (comment 1, 2, 3) are printed on the chart paper.
Refer to Section “7.1.7 Setting other functions: (6) Comments”.
Comments are printed by the preset Chart feed speed.
Depending on the Chart feed speed, some comments may not be printed (e.g. where Chart
feed speed is too slow.). Refer to Section “7.1.4 Setting the Chart feed speed”.
Printing starts upon closing the DI terminals and the COM terminal (ON).
It is possible to print up to 16 characters for Dot printing type, and up to 12 characters for Pen
writing type.
(8) Comment printing 1 to 3 (Async.),,(ACMT 1 to 3)

The preset comments (comment 1, 2, 3) are printed on the chart paper.
Refer to Section “7.1.7 Setting other functions: (6) Comments”.
Comments are printed regardless of the preset Chart feed speed.
During printing, measured value recording will not be performed.
Printing starts upon closing the DI terminals and the COM terminal (ON).
It is possible to print up to 16 characters for Dot printing type, and up to 12 characters for Pen
writing type.
Notes when operating Recording RUN/STOP
When operating Recording RUN/STOP externally using the DI function, select “(3) RUN/STOP
trigger(RCD)” of the DI function (p.85), and DI setting in engineering mode has to be set
to(EXT). [Refer to Section “7.2.6 Settings related to Recording (1) Recording
RUN/STOP Trigger setting” on page 82.]
For priority order and buffer action of Comment, Manual and Date/Time Printing, refer to
Sections “6.3.3 Printing sample” and “6.4 Digital printing”.
Example: Setting comment printing 3 for DI3
Display

Operation Keys

Description

Enter engineering mode (See the “Key Operation to

ENTEnter Engineering Mode” on page 76). Use the
PRINT

PRINT

ENT

ENT

display "", and press the “ENT” key.

Use thekey to select "key to", and press the “ENT”

key.
PRINT

Use theENT

key to display "", and press the“ENT” key.
Once setting is completed, the recorder will return to the
DI3 Function Setting screen.

[Note](1) Although "DI synchronous printing" does not print in STOP status, "DI asynchronous printing"does print in STOP status.
(2) Be sure to set to OFF if the DI option is not ordered. Do not set to any other settings except OFF.

88

7.2.10 Temperature Unit Selection
Sets the temperature unit.
Example: Setting the temperature unit to Fahrenheit ( ).
Display

Operation keys

Enter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select " ", and press the “ENT” key.

Once setting is completed, the recorder will return to theTemperature Unit Setting screen.

[Note]
If the temperature unit is changed, all setup data will revert to default values.

897.2.11 Point Calibration
Calibrates the pen recording position.
: Calibration of the zero point position: Calibration of the span point position
Example: Calibrating the zero point of Channel 1 pen recording

Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select "", and press the “ENT”

key.
PRINT

Use theENT

key to select "", and press the “ENT”

key.
PRINT

The instrument starts to feed and trace the chart paper.

FEED

With theENT(move left) and(move right), adjust the 0mm position of the chart paper and the recording
position. Press the “ENT” key. Display indicates thepreset counter value.
Once setting is completed, the recorder will return to theZero Point Calibration screen.

Example: Calibrating the span point of Channel 1 pen recording
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the ENT key.
key to select "", and press the “ENT”

key.
PRINT

Use theENT

key to select "", and press the “ENT”

key.
PRINT

The instrument starts to feed and trace the chart paper.

FEED

With theENT(move left) and(move right), adjust the

100 mm position of the chart paper and the pen
recording position. Press the “ENT” key.
Display indicates the preset counter value.
Once setting is completed, the recorder will return to the
Span Point Calibration screen.

907.2.12 Data Calibration
Calibrate the voltage, RTD (resistance temperature detector), and reference junction compensation.
: Calibration of voltage: Calibration of RTD (resistance temperature detector): Calibration of RJC (reference junction compensation)[Note]• Every channel requires calibration.
• This instrument has been carefully calibrated. If an indicated value is not normal, check the settings
again.
Example: Calibrating the voltage at Channel 1
Display

Operation KeysEnter engineering mode (See the “Key Operation to

ENT
PRINT

Description

Enter Engineering Mode” on page 76). Use the

ENT

display "PRINT

Use theENT

key to", and press the “ENT” key.
key to select "", and press the “ENT”

key. Select a channel to which equipment for calibration
PRINT

such as an mV generator is connected, and press the

ENT“ENT” key.
Enter 0 mV. After 30 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
Enter 15 mV. After 10 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
Enter 25 mV. After 10 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
Enter 35 mV. After 10 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
Enter 55 mV. After 10 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
Enter 200 mV. After 10 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
Enter 1 V. After 10 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
Enter 5 V. After 10 seconds, make sure that the ALM

ENT

indicator is unlit and press the “ENT” key.
Enter 10 V. After 10 seconds, make sure that the ALM

ENT

indicator is unlit, and press the “ENT” key.
PRINT

If the calibration is correct, select "ENTkey, and if incorrect, select "Then, press the “ENT” key.
91

" with the", with thekey.

Example: Calibrating the RTD at Channel 2
Display

Operation Keys

Description

Enter engineering mode (See the “Key Operation to

ENT
PRINT

Enter Engineering Mode” on page 76). Use the

ENT", and press the “ENT” key.

display "PRINT

Use theENT

key to", and press the “ENT”

key to select "

key.
PRINT

Select a channel to which equipment for calibration suchENT

as a dial resistor is connected, and press the “ENT” key.
Enter 100

ENT

. After 10 seconds, make sure that the ALM

indicator is unlit, and press the “ENT” key.
Enter 150

ENT

. After 10 seconds, make sure that the ALM

indicator is unlit, and press the “ENT” key.
Enter 300

ENT

. After 10 seconds, make sure that the ALM

indicator is unlit, and press the “ENT” key.
PRINT

If the calibration is correct, select "ENT" with thekey, and if incorrect, select "" with theThen, press the “ENT” key.

[Note]
During calibration, the ALM indicator shows the following conditions.
(1) When the ALM indicator is unlit:(2) When the ALM indicators flashing:

The calibration is within the range.

The recorder is assessing the input value now.

CH.No

CH.No

CH.No

(3) When the ALM indicator is lit:
The calibration is out of the range.

: The indicator is unlit.

Make sure that the connection: The indicator is lit.

is proper and input is correct.

If calibration is performed outside of the input range,(voltage) or

displayed when the unit returns to Measurement mode.
In that case, calibrate again, or return the calibration value to the factory default value.
(Refer to “7.2.8 Initializing the Setup Data and Calibration Data”.)

92(RTD) iskey.

Example: Setting the terminal temperature for Channel 1
Display

Operation Keys

Description

Enter engineering mode (See the “Key Operation to

ENTEnter Engineering Mode” on page 76). Use thePRINT", and press the “ENT” key.

display "PRINT

key toENT

Use theENT

key to select "", and press the “ENT”

key.
PRINT

Select the Channel to be set, and press the “ENT” key.

ENT

Measures and displays the current terminal temperature.
Press the “ENT” key, and the display turns to

ENTtemperature setting.
PRINT

Use theFEED

key to alter a numeral, andkey to shift adigit. Then, press the “ENT” key.

ENT

For detailed calibration, see Section “9.5 Calibration(Data calibration): Calibration of Reference Junction
Compensation” on page 110.
PRINT

If the calibration is correct, select "ENTkey, and if incorrect, select "" with the" with the

key.

Then, press the ENT key.

7.3 Terminating Engineering Mode
Example: The following describes how to save the setting data.
Display

Operation Keys
PRINT

Description

Use theENT

key to display "", and press the “ENT”

key to select "", and press thekey.
Use the

ENT“ENT” key. The settings of the engineering mode havebeen saved. Recording starts under the new settings.

1)[Note]
When terminating Engineering mode, if "" is selected and the power was turned off without

selecting "", the settings will become invalid.

Selecting "" invalidates the settings, and starts recording under the previous settings.

93

8． COMMUNICATION
8.1 General Description
8.1.1 General Description of Functions
This Recorder has the following communication functions.
(1) Process variable output
Process variable, alarm status, etc.
(2) Setting value output
Reads the setup data such as a range and chart speed.
(3) Setting value input
Writes the setup data such as a range and chart speed.
(4) Controlling recorder via communication (Partial)
Printing start/stop, recording start/stop and switching display are possible via thecommunication function.
For functions (2) to (4), see the separate volume “Communication Command Manual”.
Settings in Engineering mode cannot be performed by the communication function.
8.1.2 Transmission Specifications(1) Communication method: Start-stop synchronization, Half-duplex communications(2) Connection:

One-to-one (RS-232C), One-to-N (RS-485); N = 1 to 32(3) Protocol:

Original, MODBUS RTU (Selectable)(4) Communication speed: 1200, 2400, 4800, 9600, 19200 or 38400 bps(5) Start bit:

1 bit(6) Stop bit:

1 or 2 bits

(7) Parity:

Even, Odd or None

(8) Data length:

7 or 8 bits (For MODBUS RTU: 8 bits fixed)

94

8.2 Original Protocol
8.2.1 Data Structure(1) Character structure
In order to send one data (byte) in start-stop synchronous communication, the following data
format is used.
Start Bit (1) + Data Bit (7 or 8) + Parity Bit (1) + Stop Bit (1 or 2)(2) Data format
As communication data, multiple characters are sent by adding terminators to them.
Upon receiving these terminators, the recorder starts analyzing the communication data.

A

B

C

CR

LF

Terminator
1 Character(3) Error detection
If the recorder receives data as a result of a signal error such as a parity error and framing error due
to noise or hardware problems, this data will be ignored. Consequently, a data format error will
result, as the data was not sent in the normal data format. That command will not be executed in thereceiving function, and no response will result in the transmitting function. The data format error
status can be confirmed by the ESC S command (described later).

(4) Control codes
ESC: 1B HEX (hexadecimal)This code is used for an escape sequence such as link opening/closing.
This code cannot be treated as a part of the text data.
CR: 0D HEX (hexadecimal)This data is added to the end of the text data together with the LF data.
This code cannot be treated as a part of the text data.
LF: 0A HEX (hexadecimal)This data is added to the end of the text data together with the CR data.
This code cannot be treated as a part of the text data.

958.2.2 Opening/Closing the Link
To communicate with the recorder, a connection between the host computer and the recorder mustbe established. In the case of one-to-N communications, it is necessary to inform each recorder towhich recorder the text is being sent. If a link open command is sent to multiple recorders throughtheir sending function, all recorders which have received the command will send data. As a result,normal data cannot be received.
(1) Open Command(ESC)O 01 (CR)(LF)
Address (01 to 32)
Space (20 Hex.)
To communicate with the recorder, the host computer must issue this command and must memorizethat it has issued this command.
(2) Close Command(ESC)C 01 (CR)(LF)
Address (01 to 32)
Space (20 Hex.)
When sending data to the recorder to which the open command has not been issued, the hostcomputer must issue this command to the reocrder to which the open command has been issued.

8.2.3 Outputting the Process Variable(1) Specifying the Process Variable Output
TS0(CR)(LF)
Once this command is received, the recorder transfers the process variable to the transmission
buffer when the (ESC) T command is received.
(2) Updating the Data
(ESC)T(CR)(LF)
Once this command is received, the recorder transfers the updated process variable to thetransmission buffer.
(3) Specifying the Process Variable Output Order (For BINARY Mode Output)
BO0(CR)(LF)

Outputs from the upper byte.

BO1(CR)(LF)

Outputs from the lower byte.

The recorder outputs from the lower byte, unless otherwise specified.
96

(4) Outputting the Data
FM0, S CH, E CH (CR) (LF)

Specifies the ASCII mode output.

FM1, S CH, E CH (CR) (LF)

Specifies the BINARY mode output.

S CH: Output start channel 01 to 02
E CH: Output end channel 01 to 02
When the number of channels is one, specify (S_CH) = (E_CH).
(e.g.) FM0, 01, 02(CR) (LF) ASCII, Ch1 to Ch2 output(5) Process Variable Transmission Format (ASCII)DATE (YY) (MM) (DD) (CR) (LF)

Date

Year Month Day
TIME (HH) (MM) (SS) (CR) (LF)

Time

Hour Minute Second(DS1) (DS2) (ALM1) (ALM2) (ALM3) (ALM4) (UNIT 1 to 6) (CHNo.), (DATA) (CR) (LF)1

DS1: Data information 1 (1 byte)
N: Normal
D: Difference computation data
S: Sum computation data
M: Average computation data
R: Square root computation data
C: Decade computation data
O: Overrange (Data = 99999)
S: Skip (All data: Space)

2 DS2: Data information 2 (1 byte)

E:
Final data_(Space): Other data
3 ALM 1 to 4: Alarm information (1 byte each, 4 bytes in total)

H:
Upper limit alarm
L:
Lower limit alarm_(Space): Alarm OFF or alarm setting OFF
4 UNIT 1 to 6: UNIT (6 bytes)

Returns the setting unit with codes. (When sending the 7-bit data, 1 bit of information
will be omitted.)
5 CH No.: Channel number (2 bytes)

01 to 02
6 DATA: Process variable

Sign (1 byte): “+” or “–“Data mantissa section (6 bytes): 00000 to 99999
Data exponent section (4 bytes): E (Sign: 1 byte) (Multiplier: 2 bytes)(Example) +99999E 02

97

(6) Process Variable Transmission Format (BINARY)
Output bytes

Date and time

Process variable 1

Process variable 6

Output bytes (2 bytes): 5 x n (specified number of channels)+61

Date and time (6 bytes):
1) Year

1)

2)

3)

4)

5)

6)

: 00H to 63H (00H for the year 2000)

2) Month : 01H to 0CH
3) Day: 01H to 1FH4) Hour: 00H to 17H (24-hour format)5) Minute : 00H to 3BH
6) Second : 00H to 3BH

2 Process variable (5 bytes):

CH No. A2 A1 A4 A3 DATA1 DATA2CH No. (1 byte): Channel number 01H to 02H
A1 to A4 (2 bytes): Alarm type activated at each alarm level
1: Upper limit alarm
2: Lower limit alarm
0: Alarm OFF or alarm setting OFF(Example) Bit Sequence.
0

0

1

0

0

0

0

1

0

0

0

1

0

0

1

0

LEVEL2HLEVEL1HLEVEL4H

LEVEL3HLEVEL2LLEVEL1L

LEVEL4L

LEVEL3L

LEVELxH: Upper limit alarm bit at the alarm level x
LEVELxL: Lower limit alarm bit at the alarm level x

DATA1, DATA2 (2 bytes): Process variable-32000 to +32000 (2-byte hexadecimal data)
The higher and lower bytes can be reversed by the BO command.
If Skip is specified, 8080H will be output.

[Note](1) Output data over -32000 to +32000 constitutes overflow. The overflow 7E7E is output on theplus side, and 8181 is output on the minus side (Excluding the Decade channel).
(2) Irrespective of the BO command, data output of the Decade channel is as follows. The upperbyte outputs a mantissa (10 to 99). The lower byte outputs an exponent (-19 to 19). A decimal
point position is fixed on the 1 digit after the decimal point.

988.2.4 Outputting the Unit and Decimal Point Position Data
(1) Specifying the Unit and Decimal Point Position Data Output
TS2(CR) (LF)
Once this command is received, the recorder transfers the unit and decimal point position data to thetransmission buffer, when the (ESC) T command is received.
(2) Updating the Data
(ESC)T(CR) (LF)
Once this command is received, the recorder transfers the updated unit and decimal point position
data to the transmission buffer.
(3) Outputting the Data
LF, S CH, E CH (CR) (LF)
S CH: Output starting channel 01 to 02
E CH: Output ending channel 01 to 02(e.g.) LF, 01, 02(CR) (LF) Unit and decimal point place data output for Ch1 and Ch2When the number of channels is 1, specify (S_CH) = (E_CH).
(4) Data Format(DS1)(DS2)(CHNo.) (UNIT1 to 6)(DP) (CR) (LF)1

DS1 : Data information 1 (1 byte)
N : Normal
D : Difference computation data
S : Sum computation data
M : Average computation data
R : Square root computation data
C : Decade computation data
O : Overrange (Data =99999)

S : Skip (All data: Space)
2 DS2 : Data information 2 (1 byte)

E : Final data
(Space): Other data
3 CH No.: Channel number (2 bytes); 01 to 02
4 UNIT 1 to 6: UNIT

The set unit is returned by codes.
(When sending the 7-bit data, 1 bit information is omitted.)
5 DP: Decimal point information (1 byte); 0 to 4

[Note]
Depending on the range setting, the decimal point position may differ from that of the display screen.
998.2.5 Outputting the Status
If the open-link recorder has a data error on its link, it will save the error as a communication error in
the internal status area. The (ESC S) command reads this status. Issuing this command clears an
on-going error.
(1) Status Output Command
ESC

S(CR) (LF)

Upon receiving this command, outputs will be the status of commands received so far.
(2) Status Output
ERX

X CR LF

XX: 00 to 19

Table 8.2.5 Status output list
Status factor
Status output
ER 00 CRLF
ER 01 CRLF
ER 02 CRLF
ER 03 CRLF
ER 04 CRLF
ER 05 CRLF
ER 16 CRLF
ER 17 CRLF
ER 18 CRLF
ER 19 CRLF

Status factor
Syntax error

A/D END

Chart paper-empty

O
O

O
O

O
O
O
O

O: Factor by which the status is ENABLED.
A/D END:

This occurs when AD conversion is completed. However, this does not occur
for the pen writing recorder.

Syntax error:

This occurs during a communication error or a command error.

Chart paper-empty: This occurs when chart paper-empty is detected.
(When chart paper-empty option has been ordered.)100

8.2.6 Data Reception Example

Host Computer Side

Recorder Side

Open commandOpens the link.
Process variable
output designationcommand (TS0)Send the command to therecorder from which datatransmission starts.

Prepares theprocess variable
output.
Send once to the recorder.
Then, it is not necessary tosend it any more.

Process variable
output order
designation (BO);BINARY mode only

Determines theprocess variable
output order.

Data update
command (ESC T)

Prepares the
updated data.
Send to the recorder eachtime.

Data output request
command (FM)Sends theupdated data.

Close command
Closes the link.

101

Send the command whensending to another recorder.

8.3 MODBUS RTU Protocol
The MODBUS protocol is a communication protocol developed for PLCs by Modicon Inc. (AEG Schneider
Automation International S.A.S). It is described in the protocol specification (PI-MBUS-300 Rev. J). Referto this document for information on the MODBUS protocol specification.

8.3.1 Data Structure
Slave address(1 byte)Function code
(1 byte)Data(variable)

CRC(2 bytes)

8.3.2 Function Codes
The following function codes are available on this product:

Setting data read

Maximum data
length123 words

Data read from holding register

04H

Input data read

123 words

Data read from input register

06H

One-time data write

1 wordData write to holding register

10H

Continuous data writes

123 words

Data write to holding register

CodeFunction

03H

MODBUS-specific function (reference)

8.3.3 Error Response
In accordance with the communication protocol for the pertinent function code, if an error
occurs during command transmission, an error response is returned in the fixed format
described in this section.

■ Example of slave response (function code = 06H, command error = 10H)
Data
length11125Component
Slave address (only for MODBUS RTU)
Function code + 80H
Error code
Error check (only for MODBUS RTU)
Total number of bytes

Data
86H
10H
CRC
-

■ Error codes and their occurrence conditions
Error
code

Description

Occurrence condition(s)01HInvalid function code

An unsupported function code is specified.

02H

Invalid register address

03H

Invalid number of registers

04H

Device error

10H

Command error

The relative address range exceeds "9999."- The length of the accessed data is "0" or the sum of therelative address and data length exceeds the limit.
- It is assumed that two or more function codes, each
executable for one area, are involved.
- Data longer than 2 words is specified for a one-time write
command.
- The data length exceeds 123 words.
The received data is shorter than the predefined data length.
An attempt at a write over an area exceeding the writable
range.

102

8.3.4 Reading from the Input Register Area
The input register area is a read-only area. The current measured value and thecurrent time are mapped. Specify the start address (relative) and data count(assuming that one word is two bytes) of the data to be read.

Function code: 04H■ Example of master transmission (with a start address of 0032H and a data count of 2 words)
Component
Slave address
Function code
Relative start address (high-order)
Relative start address (low-order)
Data
Read data count (high-order)
Read data count (low-order)
Error check
Total number of bytes

Data
length
1
1
1
1
1
1
2
8

Data
－
04H
00H
32H
00H
02H
CRC (16 bits)■ Example of slave response (with a start address of 0032H and a data count of 2)
Data
length
1
1
1
1
1
1
1
2
9

Component
Slave address
Function code
Number of data bytes
Data 1 (high-order)
Data 1 (low-order)
Data
Data 2 (high-order)
Data 2 (low-order)
Error check
Total number of bytes

103

Data
－
04H
04H
00H
09H
00H
0AH
CRC (16 bits)

[Input Register Area Map] Function code: 04H

Address

Relativeaddress
(HEX)

30001
30002
30003

0
1
2

30004
30005
30006
30007
30008

3
4
5
6
7

30009
30010
30011
30012
30013
30014
30015
30016
30017
30018
30019
30020
30021
30022
30023
30024

8
9
A
B
C
D
E
F
10
11
12
13
14
15
16
17

30025

18

30026
30027
30028
30029
30030
30031
30032
30033
30034
30035

19
1A
1B
1C
1D
1E
1F
20
21
22

30036
30037
30038

23
24
25

30039
30040
30041
30042
30043
30044
30045

26
27
28
29
2A
2B
2C

30046
30047
30048
30049

2D
2E
2F
30

Name
Model type(1/8)
Model type(2/8)
Model type(3/8)
Model type(4/8)
Model type(5/8)
Model type(6/8)
Model type(7/8)
Model type(8/8)
Software version(1/16)
Software version(2/16)
Software version(3/16)
Software version(4/16)
Software version(5/16)
Software version(6/16)
Software version(7/16)
Software version(8/16)
Software version(9/16)
Software version(10/16)
Software version(11/16)
Software version(12/16)
Software version(13/16)
Software version(14/16)
Software version(15/16)
Software version(16/16)
MODBUS map version
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve

Arrangement1
2
3

Description

Remarks

ASCII Multipoint type： "MULTI"
Pen type ： "PEN"
Subsequently blank

4
5
6
7
8
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16

ASCII Version information on a system.

1

BinaryReserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve
Reserve104

Address

Relativeaddress
(HEX)

30050
30051
30052

31
32
33

30053
30054
30055
30056
30057

Name

Arrangement

Description

34
35
36
37
38

Reserve
Year
Month
Day
Hour
MinuteSecondRecording status

0 to 99
1 to 12
1 to 31
0 to 24
0 to 59
0 to 59
0 to 130058

39

Chart sensor status

0 to 130059

3A

Manual print status

0 to 130060

3B

List print status

0 to 1

30061

3C

Engineering
list print status

0 to 1

30062

3D

30100
30101

Remarks

Every second update

0：Recording not in progress
1：Recording in progress
0：With chart
1：Without chart
0：Print stops
1：During printing
0：Print stops
1：During printing
0：Print stops
1：During printing

Reserve
63 Reserve
64

CH01
Channel status

30102

65

CH02

30103
30104

66
67

Reserve
Reserve

30105
30106

68
69

Reserve
Reserve

30107

6A

30108

6B

30109
30110
30111
30112
30113
30114
30115
30116
30117
30118
30119
30120
30121
30122
30123
30124
30125
30126
30127

6C
6D
6E
6F
70
71
72
73
74
75
76
77
78
79
7A
7B
7C
7D
7E

00bit：
Alarm 1, 1= ON 0= OFF
01bit：
Alarm 2, 1= ON 0= OFF
02bit：
Alarm 3, 1= ON 0= OFF
03bit：
Alarm 4, 1= ON 0= OFF

CH01
Measurement data(BIN)-32000 to 32000
CH02

Reserve
Reserve
Reserve
Reserve
Decimal point

CH01
CH02

0 to 4

Reserve
Reserve
Reserve
Reserve
Float (high-order 2
byte)
Float (low-order 2
CH02
byte)
CH01
Measurement data(Float)

Reserve
Reserve
Reserve
Reserve
Reserve105

For data larger or smaller than
32000, the minus side will be
8181H, and the plus side will be
7E7EH.

Address

Relativeaddress
(HEX)

30128
30129
30130
30131
30132
30133
30134
30135
30136
30137
30138
30139

7F
80
81
82
83
84
85
86
87
88
89
8A

・・・

・・・

Name
Reserve
Reserve
Reserve
Unit (1/4)
Unit (2/4)
Unit (3/4)
Unit (4/4)
Unit (1/4)
Unit (2/4)
Unit (3/4)
Unit (4/4)
Reserve

Arrangement

Description

Remarks

CH01

Current unit

CH02

Current unit

8.3.5 Reading from or Writing to the Holding Register Area
The holding register area is a read-write area. Parameter settings and the start and stop
command of the recording state are mapped. For read, specify the start address (relative) anddata count (assuming that one word is two bytes) of the data to be read. For write, specify thestart address and the data to be written.
For details, refer to the Communication Command Manual for the Hybrid Recorder HR-700.

106

9． MAINTENANCE
Check the following items in order to use this instrument effectively.
⚫ Inspection⚫ Cleaning⚫ Replacing Consumables⚫ Adjusting Pen Recording Position⚫ Calibration

9.1 Inspection
Inspect the condition of operation periodically for effective use.
When any defect is found, see Chapter “10. Troubleshooting”.
Check whether pen recording is out of range.
Check whether the recording indication is off course.
• Check whether there are any large errors in indicated values or the pen recording position.
• Check whether there is improper pen recording.
• Check whether there are any blurred pen recordings or printouts.
• Check whether printout is correct.
Check whether the chart paper has been fed properly.
• Check whether the chart paper is folded properly.
• Check whether the feed holes in the chart paper are torn off or broken.
• Check whether the chart speed is normal.
Check whether there are any abnormal sounds.

9.2 Cleaning
Many parts of this instrument are made of plastic. Use a dry cloth to wipe the parts clean.

[Note]Do not use any organic solvents.

1079.3 Replacing Consumables
Replace consumables according to the following list.
No.

Name

Type

Period1

Chart paper

H-10100

33 days2

Ribbon cassette

HPSR001H0002C

6 months

3

Cartridge pen

WPSR196A000001A 2 months

4

Cartridge pen

WPSR196A000002A 2 months

Remarks
Quantity
When chart speed is 20
1
mm/h
Logging printing every 3
1hours
1 pen, about 2km in a beeline
12-pen, about 2km in a beeline1[Note](1) Over time ribbon colors will begin to fade. To ensure clear recording, replace the ribbon
cassette as soon as possible.
(2) Use the ribbon cassette within one year after its purchase. Over time, ribbon colors will begin
to fade.
(3) If color is noticeably faint before the changing period, replace the cartridge pen.

9.4 Adjusting the Pen Recording Position (Point Calibration)
The following describes how to adjust pen position on the chart paper.
It is advised to adjust recording position annually for accurate recording.

How to Adjust
Leave the recorder as it is, and adjust it according to the instructions in Section 7.2.11 Point
calibration. (See p.90)

9.5 Calibration (Data Calibration)
Calibrate a process variable input every year in order to keep measurement accurate.

[Note]
Prior to calibration, allow warm-up for at least 30 minutes after switching the power ON.

Devices required for calibration⚫ Precision voltage (current) generator (Jointly use a digital voltmeter with accuracy ofor less as required)⚫ Precision dial resistor

108

0.02 %

(1) Calibration of Voltage
Conduct calibration for voltage, current or thermocouple input. For thermocouple input, calibrate
reference junction compensation together.
+
Digital Voltmeter
Wiring-Precision Voltage +(Current)
Generator-+-

HR-701 (702) Recorder
Input Terminals

Conduct calibration according to the instructions in Section 7.2.12 Data Calibration (Calibration of thevoltage). (See p.91)[Note]
1Calibrate at every channel.

2 When a shunt resistor is attached to the input terminal block in the case of current input,

detach it prior to calibration.

(2) Calibration of RTD (Resistance Temperature Detector)
WiringA
Precision Dial
Resistor

B

HR-701 (702) Recorder
Input Terminals

B
Conduct calibration according to the instructions in 7.2.12 Data Calibration (Calibration of the RTD).
(See p.91)[Note]The dispersion of resistance for each wire must be 6 m

109

or less.

(3) Calibration of Reference Junction Compensation
Calibrate the reference junction compensation in the case of thermocouple input.
Wiring
Thermocouple Wires or Compensation Lead Wires
Digital Voltmeter+Be sure to attach a terminal cover.

Precision
Voltage(Current)
Generator

+

+

+

+

+

-

-

-

-

+

-

-

HR-701 (702) Recorder
Input Terminals

Copper Wires
Freezing Point
Cold Junction
Compensator(Iced Water)

Calibration(Example) Connecting the input to Channel 1, and setting the calibration when the thermocouple
input value is -0.5

(input 0), and the measured RJC temperature is 24.5

1) Check the Process variable when impressing 0.000 mV 2
voltage generator beforehand. (In this example, it is -0.5.

V to Channel 1 by the.)2) Check the terminal temperature of Channel 1 at the data calibration in the engineeringmode (Reference junction compensation). (In this example, it is 24.5
3) Add the difference (0.5
[24.5-(-0.5)=25.0.)) to the true value, then set as a correct RJC temperature.

]

Calibrate RJC according to the instructions in Section 7.2.12 Data Calibration (Calibration of RJC).
(See p.91.)[Note](1) This calibration is conducted when compensation by the recorder's built-in temperaturesensing element (INT: internal compensation) is selected as a reference junction compensation
method.
(2) Wire Channels 1 and 2 to calibrate.
(3) After wiring, attach the terminal covers and wait for 5 minutes or longer. Then, perform
calibration.

110

10．TROUBLESHOOTING
10.1 Troubleshooting
10.1.1 Problems
Solution

Problems
Does not work at all

Flowchart 1 (p.111)

Large error

Flowchart 2 (p.112)

Recordingerror

Chart paper
feed error

Pen alignment incorrect

Flowchart 3 (p.112)

No printing at all

Flowchart 4 (p.113)

Faint printing

Flowchart 5 (p.113)

No chart paper is fed

Flowchart 6 (p.114)

Improper chart paper feed

Flowchart 7 (p.114)

Inaccurate feed rate

Contact our dealer where youpurchased the instrument or oursales representative.

10.1.2 When the Recorder Does not Work at All
Flowchart 1

Does not work at all

Power turned on?

NO
Turn on the power.

YES
Power terminal
connected
properly?

NO

Connect properly.
(See Section 4.1 on p.16)

YES
Power supplied
properly?

NO
Supply the power properly.
85 to 264 V AC

YES

45 to 65 Hz

Contact our dealer where youpurchased the instrument or oursales representative.

111

10.1.3 When There is a Large Error
Flowchart 2
Large error
YES(Recording error)NO

Display or printed
value matches pen
recording value?
YES

Calibration
correct?

NO

Calibrate again.
(See pages 108-110)

YES

YES

NO

Scaling setting
correct?

Pen recording
position of
Zero/Spancorrect?

NO

Calibrate again.
(See Section
7.2.11 on p.90.)

Set properly.
(See Section 7.1.1
on p.55.)

YES
Contact our dealer where youpurchased the instrument or oursales representative.

10.1.4 When Pen Alignment is Incorrect
Flowchart 3
Pen alignment incorrect (0 or 100 %)Input signal wiringcorrect?

NO(Sensor disconnection, etc)

Wire it correctly.
(See Section 4.2 on pp.18-19)

YES
Input value andsigns (+/-) arecorrect?

NO

Input them correctly.
(See Section 4.2 on pp.18-19)NO

Set them correctly.
(See Section 7.1.1 on pp.52-56)

YES
Range and Scalesetting correct?

YES
Contact our dealer where youpurchased the instrument or oursales representative.
112

10.1.5 When Nothing is Printed
Flowchart 4
No printing at all.

Error message
displayed?

YES

See Section 10.2 on p.115.

NO
REC indicator
is lit?

Press the REC key to light
the REC indicator.
(See Section 2.2.2 on p.12)

NO

YES
Ribbon cassette
set properly?

NO

Set properly.
(See Section 5.3 on p.32)

YES
Ribbon fed
properly?

NO

Replace the ribbon cassette.
(See Section 5.3 on p.32)

YES
Contact our dealer where youpurchased the instrument or oursales representative.

10.1.6 When Printing is Faint
Flowchart 5
Faint printing

YES

Ribbon ink shortage.
The ribbon is worn out.
of life?
NO
Ribbon fed
properly?

NO

Replace the ribbon cassette.
(See Section 5.3 on p.32)

YES
Contact our dealer where youpurchased the instrument or oursales representative.

113

10.1.7 When the Chart Paper is not Fed at All
Flowchart 6
No chart paper is fed.

Chart paper fed by
the FEED key?

YESNO
DI contact(RUN/STOP trigger)
OFF?

YES

Change Chart speed.
Close (ON) the DI
contact. (See Section
7.2.9 on p.87)

NO
Chart holder set
properly?

YES

Contact our dealer where youpurchased the instrument or oursales representative.

NO
Set it properly. (See Section 5.1 on p.23)

10.1.8 When the Chart Paper is not Fed Properly
Flowchart 7
Improper chart paper feed

Chart guideattached
properly?

NO

Attach properly.
(See Section 5.1 on p.23)

NO

Lock properly.
(See Section 5.1 on p.23)

NO

Set properly.
(See Section 5.1 on p.23)

NO

Secure an adequate
environment. (See
Section 11.1.5 on p.118)

YES
Chart coverlocked properly?

YES
Chart paper set
properly?
YES
Used in adequate
environment?
YES
Contact our dealer where youpurchased the instrument or oursales representative.

114

10.2 Self-Diagnosis Function (ERROR)This instrument always performs self-diagnosis on the items listed in Table 10.2.1 below. When an error
is found after self-diagnosis, a relevant error number is displayed.
10.2.1 Self-Diagnosis Items
Table 10.2.1 List of Self-Diagnosis Items

Hard ware error

Type

Error
message

Parts
Zero point
sensorZero point position of servomotor
cannot be detected.

ADC

Conversion end error, Initialization error

IC for clock

The clock battery has run down [Note1]

WDT

Watchdog timer error

EEP WRITE

Non-volatile memory writing error

EEP READ1EEP READ2EEP READ3
RJC error
Area error
Area error
Area error

Operation error

Time error
Setting range
error

Non-volatile memory reading error(Setup data)
Non-volatile memory reading error(Engineering data)
Non-volatile memory reading error(ADC Calibration data)
Measurement error of temperaturesensing element of RJC
Input range error when calibrating
voltage
Input range error when calibrating RTD

Span L

Span R

Area error

Scale L

Scale R

Setting channelerror
All channels
skip error

SolutionContact ourdealer where
you purchased
the instrument
or our sales
representative.

Review the input
connections, andperform
calibration within
the input range.

Input value is outside of the setting
range.
Clock time setting error
Setting the channel (CH1) to which thesum/difference/average operation
cannot be set.

Area error

Area error

Other

Diagnosis

Set properly.

Zone L Zone R or
Zone R–Zone L is less than 5 mm.
The reference channel range of thesum/difference/average operation is notfor Volt, TC, RTD and SCALE.
All channels have been set to Skip.

Calibration
Not calibrated with the specified input.
value input error

Calibrate it witha correct input.

[Note 1] The battery life lasts approx. 10 years, however, this is subject to the type of environment inwhich it is used.

115

10.2.2 Error Display⚫
⚫

When multiple errors occur, their serial error numbers are automatically displayed.
When an error is recovered, the error display can be cancelled by pressing the “REC” key.

CAUTION
While an error is indicated, Auto, Manual and Date/Time are not indicated.

116

11．SPECIFICATIONS
11.1 Common Specifications
11.1.1 Input signal
DC voltage:

10 mV DC, 0 to 20 mV DC, 0 to 50 mV DC,
1 V DC, 0 to 5 V DC,

200 mV DC,

10 V DC

Thermocouple: B, R, S, K, E, J, T, C, Au-Fe, N, PR40-20, PL- , U, L
RTD:

Pt100, JPt100

Direct current:

4 to 20 mA DC (External shunt resistor: 250(Sold separately)

11.1.2 Performance and Characteristics
Digital indication accuracy rating: See Section 11.2.1.
Dead band:

Within 0.2 %Input impedance: 10 M

min. in mV, TC input (without burnout)

200 kmin. in mV, TC input (with burnout)

1M

min. in voltage input

250

in mA input [Externally install a shunt resistor (sold separately).]

Allowable signal source resistance:
10 kmax. in mV, TC input (without burnout)

100

max. in mV, TC input (with burnout)

1kmax. in Voltage input

10max. (per line) in RTD input

Normal mode rejection ratio:

60 dB min. (50/60 0.1 Hz)

Common mode rejection ratio : 140 dB min. (50/60 0.1 Hz)
Insulation resistance: 0.5 kV DC 20 M
Withstand Voltage:min. between each terminal and grounding terminal

1.5 kV AC for 1 minute between power terminal and grounding terminal
0.5 kV AC for 1 minute between input terminal and grounding terminal
0.2 kV AC for 1 minute between input terminals
Interchannel maximum noise voltage: 200 V AC at 50/60 Hz

Vibration resistance: 10 to 60 Hz, 1 m/s2 max.
Shock resistance:
Clock precision:

2 m/s2 max.
50 ppm max. (Under the Reference operating conditions)
The error due to power ON/OFF is not included.

Chart feed accuracy:

0.1 % max.

117

11.1.3 Structure
Mounting: Panel mounting (vertical panel)
Allowable backward inclination: Within 30
Material (Color)
Case: Polycarbonate (Black), Glass 10 % UL94-V0
Door: Polycarbonate UL94-V2 (Transparent)
Dust-proof, drip-proof (IEC60529-IP65)
11.1.4 Power Source
Rated supply voltage range:

100 to 240 V AC

Working supply voltage range: 85 to 264 V AC
Rated power frequency:

50/60 Hz

Working frequency range:

45 to 65 Hz

Table 11.1.4 Power consumption
100 V AC

Max. power consumption

1 pen

Approx. 20 VA

Approx. 30 VA

2 pensApprox. 25 VA

Approx. 35 VA

11.1.5 Normal Operating Conditions
Ambient temperature: 0 to 50
Ambient humidity:

20 to 80 %RH

Supply voltage:

85 to 264 V AC

Supply frequency:

45 to 65 Hz

Vibration:

10 to 60 Hz 0.2 m/s2 max.

Shock:

Unacceptable

Magnetic field:

400 A/m max. (DC and AC: 50/60 Hz)

Noise:

Normal mode (50/60 Hz)
DC voltage, Thermocouple: Peak value lower than the span widthof the measurement range
RTD: 50 mV max.
Common mode (50/60 Hz): 250 V AC max.
Interchannel maximum noise voltage: 200 V AC at 50/60 Hz

Posture:Allowable backward inclination: Within 30 , Right-and-left levelWarm-up time:

30 minutes minimum from the time of power-on

11811.1.6 Alarm Output (Option: LH3)
Number of output:

3 points (Built-in option, normally open)

Alarm type:2 types (H, L), total 4 levels/channel

Contact point capacity: 250 V AC, 3 A max. (Resistive load)
30 V DC, 3 A max. (Resistive load)
125 V DC, 0.5 A max. (Resistive load)
Hysteresis width:

0.5 %Setting accuracy:

The same as Digital indication accuracy

11.1.7 Low Voltage Directive, EMC Directive
Low Voltage Directive: EN61010-1
EMC Directive:

EN61326-1 Class A
At the EN61000-4-3 test condition:
DC voltage, Direct current range：Accuracy Ratings + 3 digit
Thermocouple range：Accuracy Ratings + 5 digit
Resistance temperature detector range: Accuracy Ratings + 10 digit

11.1.8 DI Function (Option: RE1)
A maximum of 3 points can be set.
11.1.9 Paper-empty Detection Function (Option: FL1)
Detects the chart paper-empty to output an alarm.

11911.2 Standard Setting Specifications
11.2.1 Measurement Range
The arbitrary setup is possible by the front keypad.
Digital indication accuracy rating is described in Table11.2.1 below under the Reference operating
conditions.
Reference operating conditions:Ambient temperature:
23 2
Ambient humidity:
55 10 %RH
Power supply voltage:
85 to 264 V AC
Power-supply frequency: 50/60 Hz 1 %Warm-up time:
30 minutes minimum(Performance is based on a state of no vibration or shocks.)
Terminal block reference junction compensation accuracy is not included in the digital indication
accuracy rating.
Terminal block reference junction compensation accuracy:
B, R, S, PR40-20, Au-Fe:
1
K, E, J, T, C, N, PL- , U, L: 0.5
In addition, the fluctuations due to changes in ambient temperature are as follows.
- Fluctuations due to temperature changes: Indication accuracy rating x 0.8 / 10 °C or less

Table 11.2.1 Measurement Range
Measurement
Type

Range

DC voltage, current Input

AccuracyMax.
resolution-10 to 10 mV(0.2 % of rdg + 3 digit)

10V

0 to 20 mV(0.2 % of rdg + 3 digit)

10V

0 to 50 mV(0.2 % of rdg + 2 digit)

10V-200 to 200 mV(0.2 % of rdg + 3 digit)

100V-1 to 1 V

(0.1 % of rdg + 3 digit)

1 mV

0 to 5 V(0.2 % of rdg + 2 digit)

1 mV-10 to 10 V

(0.3 % of rdg + 3 digit)

10 mV(0.2 % of rdg + 2 digit)

0.01 mA

4 to 20 mA

0.0 to 1820.0
B

Thermocouple

Recording(Analog)

Indication (Digital)

Measurement range

32.0 to 3308.0

0.0 to 1760.0
R1
32.0 to 3200.0(0.15 % of rdg + 1 )400 to 600 , 2
Accuracy is not guaranteed
within a range of 0 to less than400
(0.15 % of rdg + 1.8 )752 to 1112 , 3.6
Accuracy is not guaranteed
within a range of 32 to less than 0.1
752(0.15 % of rdg + 1 )However, 0 to 100 , 3.7
100 to 300 , 1.5(0.15 % of rdg + 1.8 )However, 32 to 212 , 6.7
212 to 572 , 2.7

120

Accuracy

Digital indication
accuracy(0.3 % of
span)

/0.18

Measurement
Type

Range

Accuracy

0.0 to 1200.0
R2
32.0 to 2192.0

0.0 to 1760.0
S
32.0 to 3200.0-200.0 to 1370.0
K1-328.0 to 2498.0-200.0 to 600.0
K2

Thermocouple-328.0 to 1112.0-200.0 to 300.0
K3-328.0 to 572.0
E1
E2
E3

-200.0 to 800.0-328.0 to 1472.0-200.0 to 300.0-328.0 to 572.0-200.0 to 150.0-328.0 to 302.0-200.0 to 1100.0J1-328.0 to 2012.0-200.0 to 400.0
J2-328.0 to 752.0-200.0 to 200.0
J3-328.0 to 360.0

Recording(Analog)

Indication (Digital)

Measurement range(0.15 % of rdg + 0.8 )However, 0 to 100 , 3.7
100 to 300 , 1.5(0.15 % of rdg + 1.44 )However, 32 to 212 , 6.7
212 to 572 , 2.7(0.15 % of rdg + 1 )However, 0 to 100 , 3.7
100 to 300 , 1.5(0.15 % of rdg + 1.8 )However, 32 to 212 , 6.7
212 to 572 , 2.7(0.15 % of rdg + 0.7 )However, -200 to -100 ,(0.15 % of rdg + 1 )(0.15 % of rdg + 1.3 )However, -328 to -148 ,(0.15 % of rdg + 1.8 )(0.15 % of rdg + 0.4 )However, -200 to -100 ,(0.15 % of rdg + 1 )(0.15 % of rdg + 0.7 )However, -328 to -148 ,(0.15 % of rdg + 1.8 )(0.15 % of rdg + 0.3 )However, -200 to -100 ,(0.15 % of rdg + 1 )(0.15 % of rdg + 0.5 )However, -328 to -148 ,(0.15 % of rdg + 1.8 )(0.15 % of rdg + 0.5 )(0.15 % of rdg + 0.9 )(0.15 % of rdg + 0.4 )(0.15 % of rdg + 0.7 )(0.15 % of rdg + 0.3 )(0.15 % of rdg +0.5 )(0.15 % of rdg + 0.5 )However, -200 to -100 ,(0.15 % of rdg + 0.7 )(0.15 % of rdg + 0.9 )However, -328 to -1480 ,(0.15 % of rdg + 1.3 )(0.15 % of rdg + 0.4 )However, -200 to -100 ,(0.15 % of rdg + 0.7 )(0.15 % of rdg + 0.7 )However, -328 to -1480 ,(0.15 % of rdg + 1.3 )(0.15 % of rdg + 0.3 )However, -200 to -100 ,(0.15 % of rdg + 0.7 )(0.15 % of rdg + 0.5 )However, -328 to -1480 ,(0.15 % of rdg + 1.3 )

121

Max.
resolution

0.1

/0.18

Accuracy

Digital indication
accuracy(0.3 % ofspan)Measurement
Type

Range

Recording(Analog)

Indication (Digital)

Measurement range

AccuracyMax.
resolution

Accuracy(0.15 % of rdg + 0.5 )However, -200 to -100 ,(0.15 % of rdg + 0.7 )T1(0.15 % of rdg + 0.9 )-328.0 to 752.0
However, -328 to -1480 ,(0.15 % of rdg + 1.3 )(0.15 % of rdg + 0.4 )
0.1 /0.18-200.0 to 200.0
However, -200 to -100 ,(0.15 % of rdg + 0.7 )T2(0.15 % of rdg + 0.7 )-328.0 to 392.0
However, -328 to -148 ,(0.15 % of rdg + 1.3 )0.0 to 2320.0(0.15 % of rdg + 1 )C
32.0 to 4208.0(0.15 % of rdg + 1.8 )(0.15 % of rdg + 1 K)
Au-Fe
1.0 to 300.0 K
0.1 K
However, 1 to 20 K 2.4 K
0.0 to 1300.0(0.15 % of rdg + 0.7 )N
32.0 to 2372.0(0.15 % of rdg + 1.3 )(0.15 % of rdg + 1 )0.0 to 1880.0
However, 0 to 300 , 37.6
300 to 800 , 18.8
PR40-20(0.15 % of rdg + 1.8 )Digital indication
32.0 to 3416.0
However, 32 to 572 , 67.7
accuracy
572 to 1472 , 33.8
(0.3 % of
0.0 to 1390.0(0.15 % of rdg + 0.7 )span)
PL32.0 to 2534.0(0.15 % of rdg + 1.3 )(0.15 % of rdg + 0.5 )-200.0 to 400.0
However, -200 to -100 ,(0.15 % of rdg + 0.7 )U(0.15 % of rdg + 0.9 )-328.0 to 752.0
However, -328 to -148 ,
0.1 /0.18(0.15 % of rdg + 1.3 )(0.15 % of rdg + 0.5 )-200.0 to 900.0
However, -200 to -100 ,(0.15 % of rdg + 0.7 )L(0.15 % of rdg + 0.9 )-328.0 to 1652.0
However, -328 to -148 ,(0.15 % of rdg + 1.3 )-200.0 to 650.0(0.15 % of rdg + 0.3 )Pt100-1-328.0 to 1202.0(0.15 % of rdg + 0.6 )-200.0 to 200.0(0.15 % of rdg + 0.2 )Pt100-2-328.0 to 392.0(0.15 % of rdg + 0.4 )-200.0 to 630.0(0.15 % of rdg + 0.3 )JPt100-1-328.0 to 1166.0(0.15 % of rdg + 0.6 )-200.0 to 200.0(0.15 % of rdg + 0.2 )JPt100-2-328.0 to 392.0(0.15 % of rdg + 0.4 )For Input types (R, K, E, J, T, Pt100, JPt100), an input type is automatically selected depending on themeasurement range.
B,R,S,K,E,J,T,N: JIS C 1602-1995 (IEC584-1)
C(WRe5-26):
ASTM E988-1996
Au-Fe:
ASTM E1751-2009
PR40-20:
ASTM E1751-2009
PL- :ASTM E1751-2009
U:
DIN43710: 1985 (Cu-CuNi)
L:
DIN43710: 1985 (Fe-CuNi)
Pt100:JIS C 1604-1997 (IEC751)
JPt100:(JIS C 1604-1981)

Resistancetemperature detector

Thermocouple-200.0 to 400.0

12211.2.2 Accuracy at the Computation(1) Scaling
The accuracy rating at the scaling is shown as the following formula.
Scaling accuracy (digits) =[Range accuracy rating (digits) x Scaling factor +2 digits]

Scaling span (digits)
Measurement range span (digits)

But, Scaling factor =(Example 1) When VOLT range is 1.000 to 1.000V and Scaling range is 0.00 to 100.00.
Range accuracy rating =0.1 % x 1.000 V + 3 digits(Input +1.000 V) =4 digits

Scaling factor

10000 01000 ( 1000)==5Therefore, Scaling accuracy = (4 x 5 + 2)
= 22 digits

Analog recording accuracy =4+{1000 ( 1000)} x

0.3
100

= 10 digits= 0.01 V(Example 2) When Type K is 0.0 to 200.0
Range accuracy rating(Input 200)

and Scaling is 0.00 to 200.00.

=0.15 % x 200.0+0.3=0.6=6 digits

Scaling factor=

20000 02000 0=10

Therefore, Scaling accuracy = (6 x 10 + 2)
= 62 digits= 0.62(2) Square Root Computation
The square root computation is shown as the following formula.
Display value= 10

Process variable

Display value accuracy rating:
Process variable accuracy: e
Process variable: x (converting into %).
Display value

=10

x e

Therefore, Display value accuracy

10 x

=

5e
x

123

10e
2 x

For input 4 to 20 mA, the square root computation accuracy

is as the following formula.

Measuring 100 %

100

=

5e
100

=

5(0.2+0.125)
10= 0.16 % rdg

Measuring 50 %

50

=

5e
50

=

5(0.1+0.167)
7.07= 0.19 % rdg

Measuring 9 %

9

=

5e
9

=

5(0.018+1.39)
3= 2.3 % rdg

Measuring 1 %

1

=

5e
1

=

5(0.002+12.5)= 62.5 % rdg

If the process variable is low, as above, its accuracy rating is not valid.
(3) Decade
The accuracy rating at the decade computation is defined as the logarithm of display value.
Under normal operaitng conditions, the accuracy rating is less than 1 digit, and it cannot be read.
(4) Difference, Sum and Average
The accuracy rating of Difference, Sum and Average is shown as the following formula.
Difference and Sum: Reference channel accuracy rating x 2
Average: Reference channel accuracy rating(Example) When Reference channel VOLT is 00.00 to 10.00 V and Difference is -10.00 to 10.00.
Reference channel accuracy rating = (0.3 % x 10.00 V + 3 digits)= (1000 x 0.003 + 3)
= 6 digits
Difference accuracy rating = 6 x 2 = 12 digits

11.2.3 Individual Specifications
Table 11.2.3 Specification Items
Block
Input

Record&Printer

Item

Specification

Measuring Point

1, 2

Input Sampling

120 ms

Display Interval

2.4 sec (Data update period 120 ms)

Recording Form

Disposable felt pen

Printing Form

Wire dot (one color ink ribbon)

Recording Width

100 mmStep Response Time

1.0 sec max. (95 % response to IEC 1143)

Chart Paper

Chart Speed

Length: 16 m, Width: 114 mm, Folding width: 40 mm
The length of the Clean Chart is 12 m.
5, 10, 15, 20, 25, 30, 40, 50, 60, 75, 80, 90, 100, 120, 150, 160, 180,
200, 240, 300, 360, 375, 450, 600, 720, 750, 900, 1200, 1500, 1800,
2400, 3000, 3600, 4500, 4800, 5400, 6000, 7200, 9000, 10800, 12000
mm/h

Recording Color

1 pen (Red), 2 pen (Green)

Printing Color

PurpleWeight2.5 kg max.

Power Consumption

35 VA max.

124

11.2.4 Standard Functions
Table 11.2.4 Standard Functions
Item

Description

Analog indication

None

Analog recording

Analog recording by pen.

Engineering unit indication

None

Digital indication
Logging printing

List printing

Engineering list printing
Manual printing
Programming
Memory Backup

Indicates Channel No., Process variable, Chart speed, Alarm setting
value on the display.
Prints Date, Time, Chart speed, Process variable and Engineering unit ata programmed interval. Synchronous printing/Asynchronous printing canbe selected depending on the setting.
Prints Chart speed, Sensor type, Measurement range, Engineering unit,
Alarm value, Comment, Printing description, Logging printing ON/OFF,
Zone, Partial Compression/Expansion.
Prints Channel or Tag printing, Alarm function, Logging printing ON/OFF,
Reference time, Interval, Logging printing Sync./ASync., Printing gap
correction ON/OFF, Alarm hysteresis, Burnout, DI function (Option)
Prints Measurement result by key operation.
Programs Chart speed, Alarm value, Logging printing interval, Logging,
Date and Time.
A built-in lithium battery protects the clock function. The battery life lasts
10 years. (Total power off period of the instrument: 5 years) A non-volatilememory stores setting data and calibration data.

Alarm

Sets 2 types (H, L) of alarms per channel. Sets a total of 4 levels.

Chart Speed

Selects a Chart speed from 41 types.

Clock Indication
Self Diagnosis
Scaling

Indicates Year, Month, Day, Hour and Minute. Sets the year in A.D, andadjusts leap year automatically.
When malfunctions occur, “E Err” is indicated.
(: Type of error).
Indicates and records the input of a signal 4 to 20 mA, etc. by converting
it into engineering scales.

12511.3 Standard Setting Functions
11.3.1 Standard Setting Functions
Table 11.3.1 Standard Setting Functions
Function

Description
To deflect over toward 100 % with disconnecting input. Specifies every
channel’s burnout. (DC voltage of 50 mV max, TC input)
Sets a tag number by up to 5 figures per channel. (Printed at a logging
printing)

Burnout
Tag Number
Reference Junction
Compensation

Specifies an inside/outside of the reference junction compensation.

Copy Function

Copies a channel setting.

Input Offset

Sets the input offset per channel.

Zone Recording(Track Recording)PartialCompression/Expansion

Specifies a recording area per channel to separate the record into
Tracks.

Decade Recording Indication
Alarm Printing
Alarm Recovery Printing
Recording Start/Stop printing

Records Partial Compression/Expansion to 1-crease line per channel.
Records and indicates by up to 5 decades. 2 digits are effective forindication and printing.
Prints Occurrence time, Occurrence channel, alarm Setting No. andAlarm type in purple when an alarm occurs.
Prints Recovery time, Recovery channel, alarm Setting No. and Alarm
type in purple when the alarm recovers.
Prints Time and Start/End comment when recording starts and stops.
Synchronous, Asynchronous or OFF can be selected.

Computation

Computes Square root.

Interchannel Computation

Computes Sum, Difference and Average values between channels.

Alarm Hysteresis Width

Sets an alarm hysteresis width to 0 % FS or 0.5 % FS.

Digital Filter

Records the first order lag with the filtering Process variable.

11.4 Optional Functions
11.4.1 Remote Function of DI
Function

Description

Chart Feed Start/Stop

ON: Start, OFF: Stop

Changing Chart Speed

ON: 1st, OFF: 2nd

Comment Printing (Sync./Async.)

ON: Comment Printing

Manual Printing (Sync./Async.)

ON: Manual Printing

Date and Time Printing (Sync./Async.)

ON: Date and Time Printing

11.4.2 Alarm Output Function
3 relays
11.4.3 Communication Function
RS-232C (Standard)
RS-485 (Optional)
126

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office:2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, Japan[URL][E-mail]

https://shinko-technos.co.jp/e/overseas@shinko-technos.co.jp

Tel: +81-72-727-6100
Fax: +81-72-727-7006
No.HR72E14 2024.01