Instruction Manual

SIGNAL CONDITIONERSB SERIES
No. SB1E5 2022.11

PrefaceThank you for purchasing the Signal Conditioner SB series.
This manual contains instructions for the mounting, functions, operations and notes
for operating the SB series.
For model confirmation and unit specifications, please read this manual carefully
before starting operation.
To prevent accidents arising from the misuse of this instrument, please ensure
the operator receives this manual.

Notes• This instrument should be used in accordance with the specifications described inthe manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed,
serious injury or accidents may occur.
• The contents of this instruction manual are subject to change without notice.
• Care has been taken to assure that the contents of this instruction manual arecorrect, but if there are any doubts, mistakes or questions, please inform oursales department.
• This instrument is designed to be installed on a DIN rail within a control panel.
Measures must be taken to ensure that power terminals or other high voltage
sections cannot be touched.
• Any unauthorized transfer or copying of this document, in part or in whole, isprohibited.
• Shinko Technos Co., Ltd. is not liable for any damages or secondary damages
incurred as a result of using this product, including any indirect damages.

Safety Precautions (Be sure to read this before using units)

The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on circumstances, procedures indicated byCaution may cause
serious results, so be sure to follow the directions for usage.

which may lead to dangerous conditions andWarning Procedurescause death or serious injury, if not carried out properly.

Wiring precautions

Caution• Do not leave bits of wire in the instrument, because they could cause a fire ormalfunction.
• When wiring terminals, use ferrules with an insulation sleeve and crimping pliers
made by Phoenix Contact GMBH & CO. applicable to terminals.
• This instrument has no built-in power switch, circuit breaker or fuse. It isnecessary to install them near the instrument.
(Recommended fuse: Time-lag fuse, rated voltage 250V AC, rated current 2A)• For wiring of AC power source, be sure to use exclusive terminals as describedin this manual. If AC power source is connected to incorrect terminals, the unit
will burn out.
• For a DC power supply, do not confuse polarity.
• Do not apply a commercial power source to the sensor connected to the input
terminal nor allow the power source to come into contact with the sensor, as theinput circuit may burn out.
• Use a thermocouple, compensating lead wire and 3-wire system RTD
according to the sensor input specifications of this unit.
• When using DC voltage and current input, do not confuse polarity when wiring.
• Keep the input wire (TC, RTD, etc.) and power line away from each other.

Operation and maintenance precautions

Caution• Do not touch live terminals. This may cause electric shock or problems in operation.
• Turn the power supply to the instrument OFF before cleaning.
Working or touching the terminal with the power switched ON may result insevere injury or death due to Electric Shock.
• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)

1. Name and functions of sections
SBA, SBV, SBE, SBR, SBD
POWER indicator (Green): Lights when power-on.
Flashes in 0.5 second cycles if an error has occurredin non-volatile IC memory.
ERROR indicator (Red): When the input is equal to or morethan the upper rated limit of the input range + rated span ×10%, it flashes in cycles of approx. 0.25 sec, when the input
is equal to or less than the lower rated limit of the input
range + rated span × -10%, it flashes in cycles of approx. 0.5
sec.
Zero adjustment potentiometer

which may lead to dangerous conditions andCaution Procedurescause superficial to medium injury or physical damage or may
degrade or damage the product, if not carried out properly.

Warning• To prevent an electric shock or fire, only Shinko or qualified service personnel
may handle the inner assembly.
• To prevent an electric shock, fire or damage to instrument, parts replacement
may only be undertaken by Shinko or qualified service personnel.

Span adjustment potentiometer
Spring type plug
DIP switch No.1: Selects the output status.
OFF: Normal, ON: Reverse
No.2: Selects output status when input burnout(for SBE, SBR)
OFF: Upscale, ON: Downscale
No.3, 4: Not used

Safety precautions• To ensure safe and correct use, thoroughly read and understand this manual
before using this instrument.
• This instrument is intended to be used for industrial machinery, machine tools
and measuring equipment. Verify correct usage after consulting purpose of use
with our agency or main office. (Never use this instrument for medical purposeswith which human lives are involved.)• External protection devices must be installed, as malfunction of this product could
result in serious damage to the system or injury to personnel.
Also proper periodic maintenance is required.
• This instrument must be used under the conditions and environment described inthis manual. Shinko Technos Co., Ltd. does not accept liability for any injury, lossof life or damage occurring due to the instrument being used under conditions
not otherwise stated in this manual.
Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in themanufacture of weapons of mass destruction (i.e. military applications, militaryequipment, etc.), please investigate the end users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.

Installation precautions

CautionThis instrument is intended to be used under the following environmental
conditions (IEC61010-1): Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable or explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of -5 to 55 (23 to131 ) that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85%RH• No large capacity electromagnetic switches or cables through which large
current is flowing• No water, oil or chemicals or where the vapors of these substances cancome into direct contact with the unit
Note: Avoid setting this instrument directly on or near flammable material
even though the case of this instrument is made of flame-resistant resin.

(Fig.1-1)

SBP

POWER indicator (Green): Lights when power-on.
Flashes in 0.5 second cycles if an error has occurredin non-volatile IC memory.
ERROR indicator (Red): With the zero adjustment point
at 0% and the span adjustment point at 100%, when theinput is 110% or more, it flashes in approx. 0.25 sec.
cycles. When the input is -10% or less, it flashes inapprox. 0.5 sec. cycles.
Zero adjustment button
Span adjustment button
Spring type plug
DIP switch No.1: Selects the output status.
OFF: Normal, ON: Reverse
No.2 to No.4: Not used

(Fig.1-2)

Input Range Rating
ModelInput signal
nameSBE
Thermocouple
KJ
R
S
B
E
T
N
PL-Ⅱ
W5Re/W26Re
SBR
RTD
Pt100
JPt100Pt1000
Cu50
Cu100
SBA
Direct current

SBV

SBD

DC voltage

Direct current(Table 3-1)
Ferrules with insulation sleeve Conductor cross sections
Crimping pliers2
AI 0.25-6 BU
0.2 to 0.25mm
AI 0.34-8 TQ
0.25 to 0.34mm2
AI 0.5-8 WH
0.34 to 0.5mm2
CRIMPFOX ZA 3
AI 0.75-8 GY0.5 to 0.75mm2
CRIMPFOX UD 6
AI 1.0-8 RD
0.75 to 1.0mm2
AI 1.5-8 BK
1.0 to 1.5mm2
AI 2.5-8 BU
1.5 to 2.5mm2
For terminal wiring, refer to the following. (Fig. 3-1, Fig. 3-2)
SBA, SBV, SBE, SBR, SBP
TC : Thermocouple input (SBE)
RTD: RTD input (SBR)
V : DC voltage input (SBV)mA : DC current input (SBA)POTENTIOMETER: Potentiometer input(SBP)

Input range rating-200 to 1370
-328 to 2498
-200 to 1000
-328 to 1832
-50 to 1760
-58 to 3200
-50 to 1760
-58 to 3200
0 to 1820
32 to 3308
-200 to 800-328 to 1472
-200 to 400-328 to 752-200 to 1300
-328 to 2372
0 to 1390
32 to 2534
0 to 2315
32 to 4199
-200 to 850-328 to 1562
-200 to 500-328 to 932-200 to 850-328 to 1562
-50 to 150-58 to 302-50 to 150-58 to 302
4 to 20mA DC
0 to 20mA DC
2 to 10mA DC
0 to 10mA DC
0 to 100mV DC
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC
4 to 20mA DC(Fig. 3-1)
SBD

2. Mounting

mA: DC current input

2.1 External dimensions (Scale: mm)
DIN rail75(Fig. 3-2)
For spring type plug, insert the wire while pressing the lever (orange color) with aflat blade screwdriver. (Fig. 3-3)

4

The wire to which theferrule with an insulation
sleeve is clamped

85

18.33

4

(Fig. 2.1-1)

Notice• Mount the DIN rail laterally.
• The DIN rails compatible with the SB series
are shown on the right. (Fig. 2.2-1)• To fix this unit, use commercially available
fastening plates at both ends of the unit.

(Fig. 3-3)
35 0.1mm
23mm or more

7.5mm or more2.2 Mounting to DIN rail

Mounting screw height
3.4mm or less

17.5

Recommended fastening plate:Fig. 2.2-1
Manufacturer's nameProduct name and model name
OMRON Corp.
End plate:PFP-M
IDEC Corp.
Fastening plate: BNL6
Panasonic Electric Works Co., Ltd.
Fastening plate: ATA4806• Do not confuse the top and bottom of this unit.
• When mounting to or removing the unit from the DIN rail, the units must be
slanted a little.
Therefore do not mount any other instruments within 50mm of space at theupside and downside from the unit, considering wiring space and heat radiation.
(1) Pull the lock lever of the unit down ( 1 ), hook the unit to the DIN rail at anangle of approx. 20 degrees upward ( 2 ) and mount it ( 3 ).
(2) Push the lock lever of the unit up until a clicking sound is heard ( 4 ).

Insert the wire aspressing the lever(orange color) with aflat blade screwdriver.

4. Adjustment

The output of this unit has been already adjusted before shipping. Therefore, it is
not required to adjust the unit in the case users use it with the ordered Input/Output
specification.
However, in the case of fine adjustment or calibration between connected units,adjust the value following the procedures described below.
Connect an mV generator or Dial resistor to the input terminals of this unit.
Connect a digital multimeter to the output terminals.
For SBP, register Zero and Span position after connecting a potentiometer.

SBA, SBV, SBE, SBR, SBD (zero, span adjustment)(1) Input the Output 0% value, then adjust the value by turning the potentiometer forzero adjustment, while viewing the output value (digital multimeter indication).
(2) Input the Output 100% value, then adjust the value by turning the potentiometerfor span adjustment, while viewing the output value (digital multimeterindication).
(3) Input the Output 0%, 100% value again, and check the output value (digitalmultimeter indication).
(4) If Output 0% (or 100%) value is not at 0% (or 100%), repeat steps from (1) to (3)until output corresponds to the input value.
SBP (zero, span registration)(1) Set the potentiometer to any position of the MIN side, then hold down the Zeroadjustment button for approximately 2 seconds. Automatic adjustment will
begin, and Zero position will be registered.
(2) Set the potentiometer to any position of the MAX side, then hold down the Spanadjustment button for approximately 2 seconds. Automatic adjustment will
begin, and Span position will be registered.

Inquiry(Fig. 2.2-2)(Fig. 2.2-3)(3) Be sure to use fastening plates at both ends of the unit.

3. Wiring

When using ferrules, use the following recommended ferrules and crimping pliers
made by Phoenix Contact GMBH &CO. (Table 3-1)

For any inquiry about this unit, please contact the vendor where you purchased theunit or our agency after checking the following.
(e.g.)• Model -------------------- SBA-0101-0• Serial number ---------- No.22AF05000
In addition to the above, please let us know the details of malfunction, if any, andthe operating conditions.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION

Head Office : 2-5-1, Senbahigashi, Minoo, Osaka, Japan[URL]: https://shinko-technos.co.jp/e/Tel : 81-72-727-6100[E-mail]: overseas@shinko-technos.co.jp
Fax: 81-72-727-7006