• Never disassemble, remodel, modify, or repair this
instrument. Otherwise malfunction, electric shock,or failure may result.
• Never touch the terminal while the instrument is being energized. Otherwise electric shock or malfunction may result.

Please Read First
Caution on Safety
Read this “Caution on Safety” carefully before
using the instrument.
• Be sure to observe the instructions shown below,
because they describe important information on
safety. The degree of danger is classified into thefollowing two levels: “WARNING” and “CAUTION.”

1.2• Turn off the power before attaching/detaching themodule/unit. Otherwise electric shock, malfunction
or failure may result.
• We recommend you to perform periodic maintenance for the safe and continuous use of this instrument, because consumable parts or those which
deteriorate with time are mounted in this instrument.
• Do not block the ventilation holes at the top and thebottom of this instrument. Otherwise a failure, malfunction, shortened service life, or fire may result.

The signs and their meanings are as follows:
Improper handling may cause danger-

WARNING ous situations that may result in deathor severe injury.

Improper handling may cause danger-

CAUTION ous situations that may result in moderate or light injuries or property damage.

1.

WARNING

1.1

MaintenanceInstallation• This equipment is intended to be used under thefollowing conditions.
Ambient temperature 0 to 50°C (panel mounted type)
0 to 40°C (portable type or with Ethernetfunction)
Ambient humidity
20 to 80% RH (with no condensation)
Installation category II
by IEC1010-1
Pollution degree2• IP : 50 (Front) / 20 (Terminal)• When there is a possibility that the abnormalityof this instrument may cause a major accident or
damage to other instruments, externally install anadequate emergency stop circuit or a protection
circuit to prevent accidents.
• This product is provided with a built-in fuse thatcannot be replaced by the customer. Therefore,
we recommend you to separately provide adequate
fuses externally. (Rating: 250V, 1A)
The details of the built-in fuse are as follows.
Type : TR-5 19372, 3.15A (Manufactured by
Wickmann-Werke GmbH)
Rating : 250V, 3.15A, Type: T (Slow-blow type)• Feed the power-supply voltage to specifications toprevent damages to and breakdown of the instrument.
• Never turn on the power before all the mounting
and wiring work are finished to prevent electric
shock, malfunction or failure of the instrument.
• Never use this instrument in an environment where
flammable or explosive gases exist, since this is notof intrinsically safe construction.

–2–

2.
2.1

CautionCautions when installing• Never use the instrument if it is found damaged
or deformed when unpacked. Otherwise a fire, malfunction, or failure may result.
• Check that the instrument is to the proper specifications. Otherwise damage or failure may result.
• Do not give a shock to the instrument by falling or
toppling it. Otherwise damage or failure may result.
• Avoid installing the instrument in the following
places.
• A place where the ambient temperature goesbeyond the range from 0 to 50°C (0 to 40°C
when the instrument is mounted with its side faceclosely contacted, or in the case of portable type,or provided with Ethernet function)• A place where the ambient humidity goes beyond
the range from 20 to 80% RH• A place where condensation occurs• A place where corrosive gases (sulfuric gases orammonia, etc., in particular) or flammable gasesexist• A place where vibration or impact may be applied
to the instrument (permissible continuous vibration condition: 4.9 m/s2 or lower)• A place subjected to water, oil, chemicals, vapor,or steam• A place subjected to dust and high in salt or ironcontent• A place where inductive interference may have agreat effect, thus causing static electricity, magnetism, or noises• A place subjected to heat accumulation by radiantheat or the like
• A place that is affected by electromagnetic wave
from wireless machine or portable telephones.
• If the instrument is installed near other electronics
instruments, such as TV in particular, noises may
be caused. Take the following measures in thesecases.
• Place the instrument as far from the TV or theradio as possible (1m or more)• Change the orientation of the antenna of the TV
or the radio.
• Use separate receptacles.

2.2

Cautions when installing to the panel• When mounting this instrument against the panel,
pay attention not to apply stress to the case. Otherwise the case may be damaged.
• Fix the instrument by fastening the screws at the top
and the bottom of the supplied mounting bracket.
• The panel to be used should be more than 2 mm
thick.
• Excessive torque will cause damage to front panel
frame or result in case deformation.
Tightening torque: 0.2 N·m• If the panel front is subject to water splashes, usepanel packing between the unit and panel.

–3–• Mount the unit horizontally, withno tilt to the left or right (The
forward tilt should be 0° but theunit may be inclined 0 to 30°
rearwards).

2.3∠α∠α = 60 to 90°

Cautions for wiring connection• Only authorized workers should perform wiring.
Improper wiring may cause fire, failure, or electric
shock.
• Be sure to ground the instrument. Otherwise electric shock or malfunction may result.
• When cables are connected to terminals of the recorder unit, don’t apply pulling force to them excessively. Excessive force to the terminal may result indamage to the terminal or cable.
• Use the power cable that has the performanceequivalent to or higher than 600-V vinyl insulated
power cable (JIS C3307). Install the attached noise
filter within approximately 20cm from the power
terminal of this instrument. (Wind the power cable 1
to 2 turns. The internal diameter of the ferrite core is8mm.) Do not connect two or more instruments inparallel.
• Up to 2 solderless terminals should be used whenconnecting cables to terminals. Be sure to use an
insulation cap.
(Note)
1) At the completion of wiring of the input terminals, be sure to close the rear cover to ensure
the compensation of reference contact whenthermocouple input is used.
In case of thermocouple input, follow the steps
to stabilize temperature at the terminal.
• Be sure to attach input terminal cover.
• Don’t use a thick cable to prevent the effectof radiation. The cable with a diameter of 0.5
mm or less is recommended.
• Don’t mount other instruments near a fan tokeep temperature stable.
2) For connection of lead wires to terminals, useof sleeve-insulated clamping terminals (for M3
screw) is recommended.
3) This unit has no power fuse which is removable. Mount a power fuse outside the unit asneeded.
Recommended fuse rating: 250V AC, 1A
4) Don’t loosen screws that are secured to theterminal case and power terminal.
• Input signal cables should be wired separately as
far as possible (30 cm or more) from power lines
and high-voltage lines to minimize the effect of
inductive noise. Shielded cables should preferably
be used. In this case, the shield braids should beearthed at one point.
• If a large noise is generated from the power supply,
provide an isolating transformer and use a noise
filter.
• Install the instrument paying attention to prevent
foreign matters such as scraps, electric wire chips,
and iron powder from entering in the instrument.

• Do not use the wires other than the specified compensation conducting wires for the thermocouple
input connection. Otherwise improper indication or
malfunction may result.
• Use a wiring cable with low resistance and withsmall resistance difference among the three wires
for the resistance bulb input connection. Otherwise
improper indication or malfunction may result.

2.44) We recommend you to back up the data contained in the card once a month. If the CF cardis broken, the important log data contained in CF
card will be lost. Be sure to back up the data.
Before using a compact flash card adaptor, checkthe capacity of the adaptor. If the capacity of thememory card to be formatted is larger than thatof the adaptor, do not format the card. If the card
is formatted, a display may appear on Windows
indicating that formatting is completed, but this
instrument does not recognize the completion offormatting.
5) Compact flash in the capacity range from 8MB to
1GB can be used.
6) The data recorded in the compact flash can beregenerated on the PC by using the data viewer(contained in the attached CD-ROM).
If the data is recorded in ASCII format, it can bedirectly opened in a spreadsheet such as EXCEL.
However, large-amount data cannot be opened(about 10MB or larger in the case of 9-point input,
and about 5MB or larger in the case of 18-point
input).
In those cases, read in data with the data viewer(contained in the attached CD-ROM), and perform CSV conversion to divide the file, whichallows the data to be read in.
The data recorded in binary format cannot be directly opened in a spreadsheet such as EXCEL.
Be careful not to make the size of a file too large
even if a large-capacity CF card is used. (Keep it
to 10MB or less.)

Cautions in Operation• Check every six months that the terminal screws
and mounting screws are securely fastened. Loosescrews may cause fire or malfunction.
• When changing the setting during the operation or
forcibly outputting, starting or stopping the instrument, be sure to check that safety is ensured.
Improper operation may result in damage or failureof the instrument.
• Be sure to keep the attached terminal cover
mounted on the terminal block during the operation.
Otherwise electric shock or fire may result.
• Stop using the instrument if it is immersed in water.
Otherwise electric leak, electric shock, or fire may
result.
• Never use organic solvents such as alcohol or benzene when cleaning this instrument. Do not directly
water the main unit. Otherwise deterioration, failure, electric leak, electric shock, or fire may result.
When cleaning the main unit, wipe with a dry cloth.
• Dispose the instrument as an industrial waste.

2.5

Handling memory card (Compact Flash)• For the memory card, use Sandisk’s compact flash
memory (URL: http://www.sandisk.com). A compact
flash manufactured by other companies may not be
used or cause other problems.
1) Be sure to format the memory card with the PC
you use.
Format it as FAT16 or FAT. If it is formatted as
NTFS, for example, it cannot be used because
this instrument does not recognize it.
2) The memory card should be inserted in theproper direction and fixed securely to the slot.
3) Don’t turn OFF the power or remove the cardfrom the slot while data is being written in or readfrom the card, or recorded data may be damaged
or lost.

–4–

For Proper Usage<Reference Item>

1 Installation
Dimensions/Panel Cutout<Details>(unit : mm)

Installation

2

Wiring• Terminal connection diagramPanel

2 ≤ T ≤ 26
T

Mounting bracket

144

1• Dimensions• Panel cutout• How to install on the panel

Dimensions

136.5
161

Model Designation Confirmation• Please confirm that the model
delivered matches your order.

Turn Power On

GR
REC

DISP

SEL

ENT
ACTIVE• How to handle the memorycard
3 Usage (Read before using)• How to change the data
display screen
4 Display and Operation Methods • How to change the parametersetting screen• Setting of parameters

21.6

Panel cutout
Mounting n units

137 +1.50(160 × n–22) +20

137 0• Stopping and starting therecordingNote2)+1.5

Mounting 1 unit

+1.5

List of parameters

Operation

185
24.6 when panel packing is used

137 0

5

140

160

6 When the display does not make sense. • Displays during abnormalities.
Since about 60 minutes is needed until the unit becomes
thermally stable, wait for 60 minutes or more after turning thepower on before making measurements, etc.

Note1) If other instruments are placed under the recorder unit,
provide a space of 100 mm or more from the instrument or floor.
Note2) When unit is mounted with its side face closely contacted,
don’t use the packing.

Cautions on wiring• Use crimped terminals matched to the screw size.
Tightening torque should be about 0.2 N·m.
• Do not connect anything to terminals not used.
AC power source
Protective grounding
Power ON
Power OFF
Double insulation
Electric shock hazard–5–

2 Wiring(3) Power supply terminal(4) Grounding terminalN

100 240VAC

Screwed (panel mounted)
L

100 to 240V AC
50/60Hz

DIOSHLD A6 DO11 COM 24VDC
IN

Inlet (portable)
100 to 240V AC
50/60HzLN(7) Ethernet
RJ45 connector

TRX1 TRX2

(5)

211

231

212

232

213

233

214

234

215

235

216

236

217

237

218

238

219

239

220

240

221

241

222

242

223

243

224

244

225

245

(1)

DI1
DI2
DI3
DI4
DI5
DO1
DO2
DO3
DO4
DO5
DO6
DO7
DO8
DO9
DO10

Thermocouple
Voltage
Resistance
bulb
+
+

61

CH6

CH7

161

CH15

CH16

–

–

+

+

–

–

162
163
171
172
173

CH17

CH18

+

+

–

–

+

+

–

–

182
183
191
192
193

+

+

–

–

62
63
71
72

CH8

CH9

+

+

–

–

+

+

–

–

11

12
13
21
22
23
31
32

RCJ

33
81
82
83
91
92
93

DIO(+24V)
PCD

41
42
43
51
52

–
+

–

PCD

–

+

+

–

–

+

+

–

–

+

+

–

–

113

TRX2
–
TRX1
+

CH2

121
122
123
131

141
142
143
151
152

–

+

+

–

–

+

+

–

–

+

+

–

–

+

+

–

CH4

CH5

(6)

–

(+)

53

CH11

CH12

CH13

CH14

PCD

Note) For current input, connect optional shunt resitors to thevoltage input terminals.

(-)

Alarm (open collector)
output, and DI terminal(D-Sub 25-pin)

CH10

153

(Shield)

CH3Note) For current input, connect optional shunt resitors to thevoltage input terminals.

–

DI6
SHLD

+

–

DIO(0V)
DO11

CH1

111

112

133
181

–

VPD

Thermocouple
Resistance
Voltagebulb++

Thermocouple
Resistance
Voltage
bulb
+
+

132

RCJ

–

73

(1) 18 points Input terminal
Thermocouple
Voltage
Resistance
bulb
+
+

Input terminal

24VDC(2) Alarm output (DO)/DI input terminal

Communication,
Alarm (open collector)
output, and DI terminal

MODBUS
communication

DIO-ON

1
VPD
14
13
PCD
25
6 DO12
7 DO13
8 DO14
9 DO15
10 DO16
11 DO17
12 DO18
15 DO19
16 DO20
17 DO21
18 DO22
19 DO23
20 DO24
21 DO25
22 DO26
23 DO27
24 DO28
2 DI7
3 DI8
4 DI9
5 DI10VPD

PCD(1) Input terminal: Connect signal cable for each channel.
(2) Alarm (DO)/DI (External control unit):
Connect the output of alarm signals and the input of DI (external control) signals.
(for alarm [DO] 1 to 10 and DI [external control] 1 to 5)(3) Power terminal: Connect power cable to L/N terminals. Power source to be connected should be free from noise.
(4) Grounding terminal : Connect to “G” terminal (Class-D, 100Ω or less).
(5) Communication, alarm (open collector) output and DI terminal:
Connect the input of alarm output (DO11) and DI (external control: D16) signals.
Connect communication signal cable to TRX (+) and TRX (-).
(6) Alarm (open collector) output and DI input:
Connect the input of alarm output (DO12 to 28) and DI (external control: DI7 to 10) signals.
(7) Ethernet terminal–6–

3 Usage (Read before using)Operation parts and their functions

(1)

(5)

(2)

ON
OFF
o
OWER

(3)

GR
REC

DISP

SEL

ENT
ACTIVES1

Name(1) Display unit

(2) Power switch

S2

S3

S4

S5

(7)

(6)

(4)

Function

Name

Allows the Real time trend screen, Bar Graph
Display screen, Analog Meter Display screen,
Digital Display screen, Totalized Value Display
screen, Historical trend screen and other various
Parameter Set screens to be displayed.

• Used to start or stop recording.
Pressing once starts recording, and pressing once
again stops recording.

S2 DISP key• Used to switch displayed contents. Every time it ispressed, the display is switched in the following
order: (1) (2) (3) (4) (5) (6) (7) and backto (1).
(1) Real time trend display(2) Key guidance display(3) Bar graph/analog meter display(4) Digital display.
(5) Totalizing data display(6) Event summary display(7) Ethernet log display• Used to switch the parameter setting screen to
the data display screen.

S3 SEL key• Used to switch from the data display screen to theparameter setting screen.
• Pressing the key on the parameter setting screen
switches to the screen one step up.
Note, however, that pressing the key on the menu
screen does not change screens.

S4 Cursor key(1) Used to select setting items.
(2) Used to increase or decrease numerical values.
(3) Pressing the ▼ key on the real time trend
displays the historical trend screen (*1).
At this time, the window can be scrolled using
the cursor key.
(4) Pressing the ◄ or the ► key on the real timetrend display, bar graph/analog meter display,
digital display, or totalized value display screen
switches group screens as follows.
► key: Group 1 → 2 → 3 → 4 → 1 → ...
◄ key: Group ... ← 1 ← 4 ← 3 ← 2 ← 1

Used to turn the power ON or OFF.

(3) Memory card slot Used for inserting the memory card.
Insert the card straight into the slot as shown inupper right figure.
Be careful not to forcibly press the card if it isinserted obliquely. Otherwise the pin on the
GR200 may be damaged.
(4) Memory card
ejection button

To remove the memory card from the slot, Cautionpress this button.

(5) Connector to
When changing parameters by using a loader,parameter loader connect the exclusive cable (optional cable: Type;CGL or USB cable [USB (A) plug – USB (mini B)plug] to the connector.
(6) Status indicatorlamp(7) Memory card

Displays power ON/OFF, LCD (screen) ON/OFF ,and recording status.
Lamp ON (lighting): Power: ON,
LCD: ON (recording in progress or suspended)
Lamp blinking(ON for 2 seconds / OFF for 2 seconds): Power: ON,
LCD: OFF (recording in progress)
Lamp blinking(ON for 1 second / OFF for 1 second): Power: ON,
LCD: OFF (recording suspended)
Lamp OFF (extinction): Power: OFF

Function

S1 REC key*1: The screen in the past of the data
currently recorded

For the memory card, use Sandisk’s compact flash
memory (URL: http://www.sandisk.com). Other
memory cards may cause trouble to the recorder.

S5 ENT key

Caution
1. Do not remove the memory card during recording. Refer toanother Manual in the CD-ROM “Removing memory card
(compact flash)” for the removal of the memory card while
recording is in progress.
2. After inserting the memory card into the slot, don’t remove thecard until the recorder can acknowledge it.
3. Be careful with static electricity when removing the memory card.

(1) Used for selection on the setting screen or
registration of the set data.
(2) If the key is pressed while the scales are
displayed on the real time trend display screen,
historical trend display screen (*1), or recorded
data display screen, the channels for whichscales are to be displayed can be switched.
(Scale of ch1
scale of ch2
…..
scale
of ch9scale of ch1
scale of ch2…..)*1: The screen in the past of the data
currently recorded–7–

4 Displays and Operation Methods
Data display screensReal timeKey

Bar gragh

Digital

Totalizing

Event

Ethernettrend

operationordisplaydatasummarylog(Group1)

guideAnalog meter(Group2)(Group2)(Group2)

Historical(Group2)(Group3)

trend(Group2)(Group3)(Group4)(Group3)(Group4)(Group4)(Group3)(Group3)(Group4)(Group4)

Go to Real time trend, group1.

Go to Parameter setting screen.

(To real time trend group1)
Parameter setting

Parameter setting screen

Each parameter screen(Example: Range setting screen)
Parameter setting
(1)

TAG

(1)
Display color(2)

Display range(3)

Recording type

(4)

Recording mode

(4)(1) Enter characters.
Select characters using thekeys, and press the
key to confirm the selection.
Repeat until all the characters are
input, then move the cursor to[Register], and press thekey.
(2) Select from menu.
Select using theand press thekey.

(3) Enter numeric characters.
Press thekeys to select digits,press thekeys to change
numerals, and then press thekey.

(4) Select using the
There are two types of parameters. In the case of (4),Select using theselect the parameter using thekey, and then changepress thekey.
it using the
key. In the case of the other type, press
thekey to enter the setting screen, where the parametercan be changed (parameters from (1) to (3)).

–8–

keys,

keys
keys, and

5 List of Parameters

Parameter names
Input setting

Factory setting contents (initial value)
Input type : K thermocouple

Unit : °C
Input filter : 3 s
Subtraction channel : 0
PV shift : 0
PV Gain : 100%
F value calculation function : OFF
F value
Reference temperature : 0.0°C
calculation setting Z value : 0.0°C
Decimal point : 1
Reset temperature : 0.0°C
Alarm setting
Alarm No : 1
Alarm type : OFF
DO relay No. : None
Alarm set value : 0.0°C
Display recording Display refresh cycle : 1 sec
mode setting
File division cycle : No division
File overwrite function : OFF
Trend display compression : 1/1
Range setting
Tag (upper) : TAG01 to TAG36
Tag (lower) : Blank
Display color : depends on channel No.
Display range : 0.0 to 1200.0°C
Recording type : Max/Min recording

Main Unit settings

Display record settings

Channel settings

Calculation
setting

Message setting

Recording mode : With Record
Message : Blank
Timing : None

Setting range
Skip, K, E, J, T, R, S, B, N, W, L, U, PN
thermocouple, Pt100, JPt100, 50mV, 500mV,
1 to 5V, 0 to 5V range, Other channels°C, °F, and engineering unit for voltage input
0 to 900 sec (1 sec step)
0 to 30 (0: No subtraction)
Engineering unit
0.00 to 327.67%
OFF, ON−230.0 to 3276.7°C−230.0 to 3276.7°C
0 to 4−230.0 to 3276.7°C
1 to 4
OFF, H, L
None, 1 to 28
Engineering unit
1 sec to 12 hours
No division, 1 hour, 1 day, 1 week, 1 monthOFF, ON
1/1, 1/10, 1/30, 1/60
Up to 8 characters
Up to 8 characters
14 colors
Varies depending on input type.
Max/Min recording, Instantaneous value recording,
Average value recording
With Record/ Display Only
Up to 32 characters
No, DI ON, DI OFF, Alarm ON, Alarm OFF

Display Name : Display group 1 to 4
Up to 16 characters
Screen structure : No.1-9=ch1-9
From ch1 to 30 for each of No.1 to 10
Trend direction : Vertical
Horizontal, Vertical
Display divided to : 10
1 to 20
Trend scale display : OFF
OFF, ONBar graph/Analog meter selection : Bar graph Bar graph, Analog meter
Color bar display selection :Channel No., Tag No., Unit
Channel No. display
Background color Trend screen background color : white
White, black
setting
Historical screen background color : black
White, black
Clock adjustment Current timeLCD OFF timeLCD OFF time : 0 min
0 to 60 min (0 : Kept lit at all times.)(Kept lit at all times.)
Unit define
Blank
Up to 7 characters × 12
DI setting
Function Invalid
Function Invalid, Recording Start/Stop,F value calculation reset, Totalize start/stop,
Totalize reset, LCD ON
Alarm Setting
Alarm hysteresis : 0.20%
0.00 to 100.00%
Alarm latch : OFF
OFF, ON
Memory FULL alarm : None
None, 1 to 28
Battery alarm : None
None, 1 to 28
Record data
Recorded data format : ASCIIASCII, Binary
formatMODBUS station No. : 1
0 to 255 (0 : No communication)
RS-485
communication
MODBUS communication speed : 19200bps 19200bps, 9600bps
setting
MODBUS parity : Odd
None, Odd, Even
Front communication function : ON
OFF, ON½
Copy
setting
{
{
{
{
U
U
{

Remarks
Set the same
input types
for every 2channels.

Setting iscommon to allchannels.
{
{
{
{

{
{
{
{
{
Messagedata is only
recorded inevent file.

Display setting¾) When parameter copy is selected, the items marked with { are copied.
Initial value is restored with the items marked with U .

–9–Be sure to set
it to ON.

Parameter names
Ethernetsetting 1

Math Channel setting

Totalized setting

Ethernet setting

Main Unit settings

Ethernetsetting 2

E-mailsetting 1

Factory setting contents (initial value)
IP address : 192.168.1.1
Subnet mask : 255.255.255.0
Default gateway : 0.0.0.0
FTP server function : OFF
FTP access control : OFF
Web server function : OFF
E-mail function : OFF
MODBUS TCP/IP function : OFF
SMTP IP address : 0.0.0.0
Sender’s mail address : Blank
Sender's name : Blank
Receiver’s mail address 1 to 8 : 0.0.0.0

E-mailsetting 2
E-mail trigger Title : Blank
setting
Trigger timing : None

Text 1 : Blank
Text 2 : Blank
PV value affixation : OFF
Receiver’s address No. : None
Mail send test :User account User name : Blank
setting
Password : Blank
User Level : Guest
Password setting Record start/stop : 0000
Parameter setting : 0000
CF manager : 0000
Daily report
Totalize cycle : 1 hour
Annual report
Base date : 1
External input
External input : DI1
Channel setting 1 Totalize calculation : Input value totalizer
External input : Dl1
Totalize base time : /hTotalize reset operation : ON
Totalize reset input : None
Channel setting 2 Totalize TAG : STAG01 to STAG72
Totalize unit : Blank
Totalize lower limit cut value : 0.0°C
Totalize scaling value : 1
Arithmetic
Arithmetic operation : No setting
operation setting
Input setting
Unit : °C
Measurement range : 0.0 to 500.0
Engineering value : 0.0 to 500.0
Square rooter : OFF
Logarithmic calculation : OFF
Math setting
Input filter : 3 sec
Subtraction channel : 0
PV shift : 0
PV gain : 100%
F value calculation function : OFF
Alarm setting
Alarm No. : 1
Alarm operation : OFF
DO relay No. : None
Alarm set value : 0
Range setting
Tag (upper) : TAG37 to TAG72
Tag (lower) : Blank
Display color
Display range : 0.0 to 1200.0°C
Recording type : Max/Min recording
Recording mode : yes
Math timer setting H-P, L-P timer cycle setting : 1
Average value timer cycle setting : 1
Total value timer cycle setting : 1
Constant setting Constant : 0

Setting range½
Copy
setting

0 to 255 (each digit)
0 to 255 (each digit)
0 to 255 (each digit)
OFF, ON
OFF, ON
OFF, ON
OFF, ON
OFF, ON
0 to 255 (each digit)
Up to 64 characters
Up to 32 characters
Up to 64 characters
Up to 32 characters
None, DI ON, DI OFF, Alarm ON, Alarm OFF,
Warning, Timer cycle
Up to 32 characters
Up to 32 characters
OFF, ON
1 to 8 (selectable)
Up to 16 characters
Up to 8 characters
Guest, Administrator
0000 to 9999 (no password at 0000)
0000 to 9999 (no password at 0000)
0000 to 9999 (no password at 0000)
10, 20, and 30 min., 1, 2, 3, 4, 6, 12, and 24 hours
1 to 31
Dl1 to 10, Channel 1 to 30, Alarm No. 1 to 4
Input value totalizer, Counter, Time measurement
Dl1 to 10, Channel 1 to 30, Alarm No. 1 to 4/s., /min, /h., /day
OFF, ON
None, DI 1 to 10, Channel 1 to 30,
Alarm No. 1 to 4
Up to 8 characters
Can be selected arbitrarily.
Engineering unit
1 to 32767{
Engineering unit
−3276.7 to 3276.7−3276.7 to 3276.7
OFF, ON
OFF, ON
0 to 900 sec (1 sec step)
0 to 30 (0: no subtraction)
Engineering value
0.00 to 327.67
OFF, ON
1 to 4
OFF, H, L
None, 1 to 28
Engineering value
Up to 8 characters
Up to 8 characters
14 colors
Engineering value
Max/Min recording, Instantaneous value recording,
Average value recording
With record/ display only
1 to 32767 min
1 to 32767 min
1 to 32767 min−32767 to 32767¾) When parameter copy is selected, the items marked with { are copied.
Initial value is restored with the items marked with U .

– 10 –

{
{
{
{
{
{
{
U
U
{
{
{
{
{
{
{
{
{
{

Remarks

Communication
function

6 Troubleshooting
Error indications
This recorder is provided with an error display function.
Modify the abnormal conditions as soon as an error
display appears. Make sure to turn off the power before checking the wire connection to modify the abnormal conditions. Otherwise electric shock or a serious
accident may result.
Display

Cause

OVER

Displayed when the measurement signal inputs
exceeding the settable recording range (Table 1).

UNDER

Displayed when the measurement signal inputs lower
than the settable recording range.

ERROR

Displayed in the following cases with 50mV, 500mV,
and 5V input.
1) When the measurement signal inputs exceeding
OVER.
2) When the measurement signal inputs lower than
UNDER.

BURNOUT

Displayed in the following cases with TC and PT input.
1) When a break or disconnection of a cable occurs
within the sensor
2) When the measurement signal inputs exceeding
OVER.
3) When the measurement signal inputs lower than
UNDER.

7 Specifications
Specification
Power voltage
Power consumption
(VA)
Input
measurement
range
Record range
Input circuit
Measurement
cycle
Record cycle
Record type

: 100 to 240 V AC (10%), 50/60 Hz: 47 VA max. (240 V AC): Range where input reading accuracy is assured.

: Settable recording range. (See Table 1.): Insulated between all channels(functional insulation).
: 100 msec

Operatingenvironment
Operatingtemperature: RS-485 Interface (*2)
Communication method / 2-wire, half-duplex
communication
Synchronous system / Start-stop synchronization
Baud rate / 9,600bps, 19,200bps
Protocol / compliant to Modbus RTU
Communication distance / 500 m max. (total
extension of communication)
Number of connectable units / 31 (Masterexcluded): Indoor: 0 to 50ºC (panel mounted): 0 to 40ºC (portable or with Ethernet function)
Operating: 20 to 80% RH (No condensation allowed.)
humidity
Storage: −10 to 60ºC
temperature
Storage humidity : 5 to 90% RH (No condensation allowed.)
Installation category : II
Pollution degree : 2
Operating altitude : 2000m max.
Calibration: Calibrate at least once a year in order to maintainhigh measurement accuracy. For calibration,refer to the instruction manual in the CD-ROM, orcontact us.
Ethernet: 10 Base-T- protocol: Web (server)
FTP (server)
SMTP (client)
MODBUS-TCP (slave)*1 Usage size of measurement data
Usage size for each sampling is as follows:
ASCII (max.-min. record)
Time : 21 bytes
Measurement data: 16 bytes × number of input points
From the above, if the number of input point is 9ch,
21 + 16 × 9 = 165 bytes
ASCII (instantaneous or average value)
Time : 21 bytes
Measurement data: 8 bytes × number of input points
From the above, if the number of input point is 9ch,
21 + 8 × 9 = 93 bytes
Binary (max.-min. record)
Time : 4 bytes
Measurement data: 4 bytes × number of input points
From the above, if the number of input point is 9ch,
4 + 4 × 9 = 40 bytes
Binary (instantaneous or average value)
Time : 4 bytes
Measurement data: 2 bytes × number of input points
From the above, if the number of input point is 9ch,
4 + 2 × 9 = 22 bytes*2 Communication converter is necessary for the connection with acomputer.
Recommendation: Model IF-400 (manufactured by
Shinko Technos): 1 sec to 12 hours/one time
: Select from max.-min. value, instantaneous
value, or average value.
Data storage: 1 min. (when record cycle is less than 1 min.)cycle
or synchronized with record cycle (when record
cycle is more than 2 min.).
Record media: Compact Flash (by Sandisk), 1GB max.
Only FAT16 or FAT is supported.
Data format: Select from ASCII or binary. Refer to *1 forusage size of measured data.
Alarm type
: Upper limit alarm, lower limit alarm
Alarm Output (relay) : (DO1) 150Vac/3A, 30Vdc/3A: (DO2 to10) 240Vac/3A, 30Vdc/3A
Alarm output: 30Vdc/0.1A(transistor)
Digital input: No-voltage contact input.

– 11 –

8 Confirming the Code Symbols
Digit

Type
GR2
Specifications
9 points
Input point
18 points
Power supply voltage 100 to 240V AC
Panel mounting / Japanese
Model name 1
Panel mounting / English(Mounting
instruction/Language Portable / Japanesefor display)
Portable / English
Model name 2
Multi-range
Model name 3
None
Model name 4
None
None
DI/DO(10)DI/DO(18), C5
DI/DO(10), DI/DO(18), C5
Option (*)
E-net
DI/DO(10), E-net
DI/DO(18), C5, E-net
DI/DO(10), DI/DO(18), C5,
E-net
None (Providing the No. for aSpecial order Number
special order product)
Instruction manual
Japanese/English (CD-ROM)
Shinko ID

4 5

6

7

- 0

8 9

10 11 12 13 14 15 16 17

M0

00

00

0- 2 0

09
18
0
A
B
C
D
M0
00
00
0
1
2
3
4
5
6
7
0
2
0

(*) Description of option codes
Option codesDI
DO(10)
DO(18)
C5
E-net

Description
5 points (No-voltage contact input)
10 points (1a contact output)
18 points (Open collector output)
Communication RS-485
Ethernet

Table 1 Display Range Set Range

9 MaintenanceInput range

Type
Thermocouple

Recommended parts replacement cycle
Front panel: 5 years (degradation of brightness of LCD back light.
Pickup repair: including packing
and battery)
Panel packing : 5 years (pickup repair)
Lithium battery : 5 years (pickup repair)
Fuse (external) : 2 years (for preventive maintenance): rating: 250V AC 1A
Memory card
: 6 months: Back up the data more frequently
than 6 months.

– 12 –

Resistancebulb

Record range set range

400 to 1760°C
0 to 1760°C
0 to 1760°C–200 to 1370°C–200 to 800°C–200 to 1100°C–200 to 400°C
0 to 1300°C
0 to 1760°C–200 to 900°C–200 to 400°C
0 to 1300°C

370.0 to 1790.0°C– 30.0 to 1790.0°C– 30.0 to 1790.0°C–230.0 to 1400.0°C–230.0 to 830.0°C–230.0 to 1130.0°C–230.0 to 430.0°C– 30.0 to 1330.0°C– 30.0 to 1790.0°C–230.0 to 930.0°C–230.0 to 430.0°C– 30.0 to 1330.0°C

JPt100 –200 to 600°C
Pt100 –200 to 600°C–230.0 to 630.0°C–230.0 to 630.0°C

B
R
S
K
E
J
T
N
W
L
U
PN

DC voltage

0 to + 50mV
0 to +500mV+ 1 to + 5V
0 to + 5V– 10.00 to + 55.00mV– 10.0 to + 550.0mV+ 0.500 to + 5.500V– 0.100 to + 5.500V