SPEC SHEETSGSDC Signal Transmitter (1-output)
Features:

SGS (1-input 1-output) converts DC voltage/current into isolated DC
voltage/current. Indication displays are equipped.

Functions:Sensor correction (Input value correction)
Normal/Reverse modeFilter time constantManual mode
Input and output types can be changed.

Model: SGS - ① ② - ③ - ④

Model name
Input 1Output 1Power supply
Option③ Power supply

DC voltage

DC current① INPUT 1A0A1A2A3A4A5A6A7A8A9V0V1V2V3V4V5V6V7V8V9

4 to 20mA (Built-in 50Ω shunt resistor)
4 to 20mA (250Ω shunt resistor)
4 to 20mA (50Ω shunt resistor)
0 to 20mA (250Ω shunt resistor)
0 to 16mA (62.5Ω shunt resistor)
2 to 10mA (250Ω shunt resistor)
0 to 10mA (100Ω shunt resistor)
1 to 5mA (100Ω shunt resistor)
0 to 1mA (1000Ω shunt resistor)
10 to 50mA (10Ω shunt resistor)
0 to 10mV (Input resistance 1MΩ)
0 to 50mV (Input resistance 1MΩ)
0 to 60mV (Input resistance 1MΩ)
0 to 100mV (Input resistance 1MΩ)
0 to 1V (Input resistance 1MΩ)
0 to 5V (Input resistance 1MΩ)
1 to 5V (Input resistance 1MΩ)
-5 to 5V (Input resistance 1MΩ)
0 to 10V (Input resistance 1MΩ)
-10 to 10V (Input resistance 1MΩ)01

100 to 240V AC 50/60Hz
24V AC/DC 50/60Hz
For SGSL (with serial communication function),
only [0: 100 to 240V AC] is selectable.

④ Option
0No option needed

1

Multi-rotation trimmer

2Moisture-proof treatment

3

Multi-rotation trimmer + Moisture-proof treatment

DC Voltage

DC Current② OUTPUT 112345ABCDEFG

4 to 20mA (allowable load resistance 750Ω max.)
0 to 20mA (allowable load resistance 750Ω max.)※1
0 to 16mA (allowable load resistance 900Ω max.)※1
2 to 10mA (allowable load resistance 1500Ω max.)
0 to 10mA (allowable load resistance 1500Ω max.)※1
0 to 10mV (allowable load resistance 10kΩ min.)※2
0 to 100mV (allowable load resistance 100kΩ min.)※2
0 to 1V (allowable load resistance 1000Ω min.)※2
0 to 5V (allowable load resistance 5000Ω min.)※2
1 to 5V (allowable load resistance 5000Ω min.)0 to 10V (allowable load resistance 10kΩ min.)※2-5 to 5V (allowable load resistance 10kΩ min.)※1: 0mA or less: Out of base accuracy.
※2: 0V or less: Out of base accuracy.

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

SPEC SHEET
Performance

Base accuracy (at 25℃)
Display accuracy
Temperature coefficient
Response time
Insulation resistanceDielectric strength±0.1% of each input span
Base accuracy ±1 digit± 0.015%/℃ (0 to 10mV output:±0.02%/℃)
0.5 sec. max. (0→90%)
100MΩ minimum, at 500V DC
2.0kV AC for 1 minute

General specifications

4 to 20mA, 0 to 20mA, 0 to 16mA, 2 to 10mA, 0 to 10mA, 1 to 5mA, 0 to 1mA,10 to 50mA DC
0 to 10mV, 0 to 50mV, 0 to 60mV, 0 to 100mV, 0 to 1V, 0 to 5V, 1 to 5V, -5 to 5V,
DC voltage
0 to 10V, -10 to 10V DC
Span adjustment range
95 to 105% (Adjustable from front panel.)-5 to 5% (Adjustable from front panel.)
100 to 240V AC 50/60Hz
85 to 264V AC
Allowable voltage range
24V AC/DC 50/60Hz
20 to 28V AC/DC
100 to 240 V AC: Approx. 9 VA max. (SGSL: Approx. 10 VA max.)
24 V AC: Approx. 6 VA max.
24 V DC: Approx. 3 W max.
-10 to 55℃ (Non-condensing, No icing), 35 to 85%RH (Non-condensing)-10 to 60℃
Case: Flame-resistant resin, Black color, Panel: Polycarbonate
DIN rail22.5(W) x 89(H) x 70(D)mm (without socket)
Approx. 77g (without socket)
DC current

Input
Zero adjustment range
Power supply
Power consumption
Operating temperature, humidity
Storage temperature
Material
Mounting methodExternal dimensions
WeightMeasurement range
Input code
A0A1A2A3A4A5A6A7A8A9V0V1V2V3V4V5V6V7V8V9Terminal arrangement

Measurement range-1999 to 9999 ※Resolution
1111111111111111111
1

PWR ⑬ - ⑭
OUT ⑨ - ⑫
A ①-④
V ①-④
RS-485

Power supply voltage

100 to 240V AC, 24V AC/DCOutput
DC current input
DC voltage input
Serial communication RS-485(For SGSL)※ Decimal point position change and scaling are possible.

Circuit configuration(8P Socket)

SHINKO TECHNOS CO., LTD. Tel: +81-72-727-6100
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technos.co.jp

20190507