120 PERISTALTIC PUMPS
NEWSmall, Simple, Precise• Biopharm• Science• Research• OEMThe new 120 peristaltic pumps• Compact footprint saves bench space• Exceptional speed control and accuracy• New ﬂip-top 114 pumphead up to 190 ml/min• Simple: minimum key presses, intuitive operation

Smooth, wipe-down
enclosure for ultimate
cleanability

Fewer key presses, fewer mistakes
and minimal setup time. The 120
pump is efficient and intuitive to use

The 114DV pumphead offers
trouble-free tube loading
in seconds, for high accuracy
and repeatable flow2

For biopharm, science, research and OEM
Fermentation

Single-use systems

The 120 is theperfect pump for
maximizing yield infermentation and cell
culture applications.
Ultra-compact andstackable formultiple feeds likepH control, antifoam, nutrient andbuffer addition, the120 offers the ideal combination of size andtechnology to optimize your process.

The new 120 pump is perfectly suited to singleuse systems where no contamination can beaccepted. Error-free tube change takes only
seconds, with tubing automatically clamped andtensioned to ensure repeatable performance.

Process validation
The biopharm process requires accurate andrepeatable pumping which is easy to validate.
With an exceptional 2,000:1 speed range and aselection of USP Class VI tubes, the 120 delivers
the flexibility and purity so vital to drug trials. A
keypad lock secures your pumping parameters
for constant and repeatable performance.

3

Small, simple, precise

Compact footprint

Flexible control

The compact, stackable
120 pump saves valuable
space in cleanroom, LAF
and biosafety cabinet
installations.

The 120 range of fixed
speed, manual control
or auto control pumps
all guarantee ±1%
accuracy, considerablybetter than the nearest
competitive peristaltic
pumps. For ultimate
flexibility, the 120U
delivers 2,000:1
speed control.

High accuracy flows

Intuitive use

The flip-top 114DV
pumphead assurestrouble-free tube loading
in seconds for highaccuracy and repeatable
flow, and long tube life.

The 120 pump’s
intuitive interface
makes the pump easyand quick to setup,with fewer key presses
for fewer mistakes.
The pump brings
maximum efficiency
to your process.

Hygienic

Process security

The pump case has nosharp edges to tear gloves
and no bug traps. Rated toNEMA 2, it is ideally suitedfor ethanol wipe-down.

Two of the 120 models
feature a keypad lock.
The security of yourprocess and product issafe with a 120 pump.

4

The Watson-Marlow family of pumps
Watson-Marlow value
Our pumps combine simple operation with sophisticated control and flow accuracy – available next day
and backed by an outstanding warranty and knowledgeable global support. Our products deliver realvalue with minimum maintenance, maximum performance and the lowest lifecycle cost.

Three-year warranty
Demonstrating total confidence in our reliability and our commitment to customer satisfaction,
Watson-Marlow 120 pumps carry a three-year warranty. Your production is safe with us.

The perfect pumpLow-shear peristaltic pumping is created bycompressing a tube with rollers, which
progress the fluid forward. This simple
dynamic effect requires no seals or valves toleak, clog or replace. The fluid is totallycontained within the tube, eliminating
contamination risks and simplifying validation.
The pump handles delicate or viscous fluidswith precision, and pumps suspended solids
and live cells with integrity.

The 120 is the latest model inLING
the world’s most widely
ACUPS
respected biopharm
SSCE
pumps range
OPR
Full scale production

Pilot / lab scale

520 for flows up to120 for flows up to190 ml/min, pressures
up to 30 psi

323 for flows up to
2,000 ml/min, pressures
up to 30 psi

3,500 ml/min, pressures
up to 100 psi

620 for flows up to18 liter/min, pressures
up to 60 psi

720 for flows up to66 liter/min, pressures
up to 30 psi
5Choose your pumpThree drives
Fixed speed
The 120F is offered in five
fixed flow rate versions: 10,17, 31, 52 and 220 rpm.
Pumphead options are the114DV and the 120R.

Manual variable
speed
The manual control, variable
speed 120S offers 1-200 rpm
operation in 1 rpm
increments. It features speed scrolling, priming
and keypad lock. All pumpheads are available.

Automatic control
The 0.1 to 200 rpm 120U
offers manual remote controlof the pump in 0.1 rpm
increments or automatic
control via 4-20mA or 0-10V
inputs. All pumpheads are available.

Five pumpheads
114DV
The flip-top, 114DV singlechannel pumphead assurestrouble-free continuous tube
loading in seconds, highaccuracy and repeatable flow.
It accepts tubing in bores sizes 0.5mm to 4.8mm.

102R

400 series

The 102R single-channel
pumphead with two-roller
rotor is suitable for silicone
tubing in bore sizes0.5mm to 4.8mm.

The single-channel
400D1 accepts tubingfrom 0.5mm to 3.2mm.
The multi-channel
400DM2 and 400DM3
pumpheads use threebridge manifold tubing.

TubingPumpsil

Bioprene

PureWeld XL

Pumpsil platinum-cured
silicone tubing is Animal
Derived Content Free (ADCF)
and laser-etched to make it lot
traceable, and is post-cured toreduce leachables to near zero. It is manufactured
in our own dedicated tubing plant, and carries full
USP Class VI and ISO 10993 certification, is FDA
compliant and is fully validated.

Fully weldable ADCF andheat-sealable PureWeld XL
thermoplastic elastomer
gives excellent pumping
performance and produces
virtually no internal spallation. It is USP Class VI
and FDA compliant, and is fully validated.

Bioprene thermoplastic
elastomer offers the best
combination of purity,
chemical compatibility andlong pumping life. It is
manufactured in our own dedicated tubingplant. It is USP Class VI certified and FDA
compliant, and is fully validated.

STA-PURE PCSSTA-PURE PFLGORE STA-PURE PCS is a composite of silicone in an ePTFE
lattice. It offers 18 times the pumping life of silicone. It is USPClass VI approved.

GORE STA-PURE PFL is a composite of ePTFE and a highgrade fluoroelastomer, offering extraordinary chemical resistanceand long life. It is USP Class VI approved.

Performance information
Tube bore and flow rates 114DV, 102R and 400D1 (ml/min)
1.6mm

2.4mm

3.2mm

4.0mm ※1

4.8mm

120F/DV 10 rpm
0.20.4120F/DV 17 rpm
0.30.7120F/DV 31 rpm
0.61.2120F/DV 52 rpm
1.02.1120F/DV 220 rpm
4.48.8120S/DV 1-200 rpm
0.02-4.00.04-8.0
120U/DV 0.1-200 rpm
0.002-4.00.004-8.0
120F/R 10 rpm
0.30.5120F/R 17 rpm
0.50.9120F/R 31 rpm
0.91.6120S/R 1 - 32 rpm
0.03-0.90.05-1.6
120U/R 0.1 - 32 rpm
0.003-0.9
0.005-1.6
120S/D1 1-200 rpm
0.01-2.2
0.03-5.8
120U/D1 0.1 - 200 rpm
0.001-2.2
0.003-5.8
Tube bore and flow rates 400DM2 and 400DM3 (ml/min)
Model and Speed
0.13mm0.19mm

1.4
2.4
4.3
7.3
31
0.14-280.014-28
2.1
3.6
6.5
0.21-6.7
0.02-6.7
0.11-230.011-23

2.9
4.9
9.0
15
64
0.29-58
0.029-58–––––
0.24-49
0.024-49

4.7
8.0
15
24
100
0.47-94
0.047-94
8.5
14
26
0.85-27
0.09-27
0.41-810.041-81

6.7
1121
35
150
0.67-1300.067-130–––––
0.59-1200.059-120

8.5
14
26
44
190
0.85-170
0.085-170
1729
52
1.6-540.16-54––0.25mm0.38mm

0.50mm0.63mm

0.76mm

120S/DM2 1-100 rpm
120U/DM3 0.1-100 rpm

0.001-0.1
0.0001-0.1
0.88mm

0.002-0.2
0.0002-0.2
1.02mm

0.004-0.4
0.0004-0.4
1.14mm

0.008-0.8
0.0008-0.8
129mm

0.014-1.4
0.001-1.4
1.42mm

0.022-2.2
0.002-2.2
1.52mm

0.031-3.1
0.003-3.1
1.65mm

120S/DM2 1-100 rpm
120U/DM3 0.1-100 rpm

0.043-4.3
0.004-4.3
1.85mm

0.055-5.5
0.006-5.5
2.05mm

0.070-7.0
0.007-7.0
2.29mm0.089-8.9
0.009-8.9
2.54mm0.110-110.011-11
2.79mm0.120-12
0.012-120.140-14
0.014-14

120S/DM2 1-100 rpm
120U/DM3 0.1-100 rpm

0.180-18
0.018-180.210-21
0.021-210.260-26
0.026-26

0.31-310.031-31

0.360-36
0.036-36––––

Model and Speed

60.5mm0.8mm※1．内径4.0mmのチューブは廃番になりましたProduct ordering
Pump product codes
PumpSpecification

Pumphead

Product code

120F/DV10 rpm17 rpm31 rpm52 rpm
220 rpm
10 rpm17 rpm31 rpm
1-200 rpm
1-32 rpm
1-200 rpm
1-100 rpm
1-100 rpm
0.1-200 rpm
0.1-32 rpm
0.1-200 rpm
0.1-100 rpm
0.1-100 rpm

114DV pumphead, continuous tubing, max 22.5 psi
114DV pumphead, continuous tubing, max 22.5 psi
114DV pumphead, continuous tubing, max 22.5 psi
114DV pumphead, continuous tubing, max 22.5 psi
114DV pumphead, continuous tubing, max 22.5 psi
102R pumphead, continuous tubing, max 7.5 psi
102R pumphead, continuous tubing, max 7.5 psi
102R pumphead, continuous tubing, max 7.5 psi
114DV pumphead, continuous tubing, max 30 psi
102R pumphead, continuous tubing, max 30 psi
400D1 pumphead, continuous tubing, max 30 psi
400DM2 pumphead, manifold tubing, max 30 psi
400DM3 pumphead, manifold tubing, max 30 psi
114DV pumphead, continuous tubing, max 30 psi
102R pumphead, continuous tubing, max 30 psi
400D1 pumphead, continuous tubing, max 30 psi
400DM2 pumphead, manifold tubing, max 30 psi
400DM3 pumphead, manifold tubing, max 30 psi

010.6111.DA0010.6112.DA0010.6113.DA0010.6114.DA0010.6115.DA0
010.6111.R00
010.6112.R00
010.6113.R00
010.6131.DA0
010.6131.R00
010.6131.010
010.6131.M20
010.6131.M30
010.6141.DA0010.6141.R00
010.6141.010
010.6141.M20
010.6141.M30

120F/R

120S/DV
120S/R
120S/D1
120S/DM2
120S/DM3
120U/DV
120U/R
120U/D1120U/DM2
120U/DM3
Pumpheads
Pumphead

Specification

Product code

114DV
102R
400D1400DM2
400DM34 roller pumphead. Accepts continuous tubing only
2 roller pumphead. Accepts continuous Pumpsil silicone tubing only
Single-channel pumphead. Accepts continuous tubing only, 3.2mm (1/8in) bore maximum
2-channel pumphead. Accepts manifold tubing only
3-channel pumphead. Accepts manifold tubing only

013.6000.00A
013.7101.000
043.001D.D1C
043.001D.D2C
043.001D.D3C

Continuous tubing for 114DV, 102R and 400D1 pumpheads 1.6mm (1/16in) thin wall tubing※１Bore0.5mm(#112, 1/50in)

0.8mm(#13 1/32in)

1.6mm(#14, 1/16in)

2.4mm(3/32in)

3.2mm(#16, 1/8in)

4.0mm(5/32in)Pumpsil
Bioprene
STA-PURE PCSSTA-PURE PFL
PureWeld XL

913.A005.016
933.0005.016––
941.0005.016

913.A008.016
933.0008.016–––

913.A016.016
933.0016.016
961.0016.016
966.0016.016
941.0016.016

913.A024.016
933.0024.016–––

913.A032.016
933.0032.016
961.0032.016
966.0032.016
941.0032.016

913.A040.016
903.0040.016
–
–
–

4.8mm
(#25, 3/16in)

913.A048.016
933.0048.016
961.0048.016
966.0048.016
941.0048.016

Manifold tubing for 400DM2 and 400DM3 pumpheads
Color code

Wall mmBore mm

Autoclavable Marprene

Marprene

PVC

Silicone

Orange/black
Orange/red
Orange/blue
Orange/green
Orange/yellow
Orange/white
Black/black
Orange/orange
White/white
Red/redGrey/grey
Yellow/yellow
Yellow/blue
Blue/blue
Green/greenPurple/purple
Purple/black
Purple/orange
Purple/white

0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8
0.8

0.13
0.19
0.25
0.38
0.50
0.63
0.76
0.88
1.02
1.14
1.29
1.42
1.52
1.65
1.85
2.05
2.29
2.54
2.79

–
–
979.0025.00+
979.0038.00+
979.0050.00+
979.0063.00+
979.0076.00+
979.0088.00+
979.0102.00+
979.0114.00+
979.0129.00+
979.0142.00+
979.0152.00+
979.0165.00+
979.0185.00+
979.0205.00+
979.0229.00+
979.0254.00+
979.0279.00+––
979.0025.000
979.0038.000
979.0050.000
979.0063.000
979.0076.000
979.0088.000
979.0102.000
979.0114.000
979.0129.000
979.0142.000
979.0152.000
979.0165.000
979.0185.000
979.0205.000
979.0229.000
979.0254.000
979.0279.000

981.0013.000
981.0019.000
981.0025.000
981.0038.000
981.0050.000
981.0063.000
981.0076.000
981.0088.000
981.0102.000
981.0114.000
981.0129.000
981.0142.000
981.0152.000
981.0165.000
981.0185.000
981.0205.000
981.0229.000
981.0254.000
981.0279.000–––––
983.0063.000
983.0076.000
983.0088.000
983.0102.000
983.0114.000
983.0129.000
983.0142.000
983.0152.000
983.0165.000
983.0185.000
983.0205.000
983.0229.000
983.0254.000
983.0279.000

Weights
120 cased pump weights
120/DV

120/R

120/D1120/DM2

120/DM3

3lbs 4oz

3lbs 4oz

3lbs 9oz

3lbs 11oz

3lbs 11oz

120/DV

120/R

1.6 in1.7 in7.6 in120/DV

7.7 in4.4 in120/DM2

120/D1120/DM3

4.3 in1.3 in1.5 in7.3 in2.1 in7.5 in8.1 in7

Watson-Marlow Pumps Group has five world-classfactories supported by direct sales operations in 20
countries and distributors in more than 50 countries.
For contact details visit our website:www.wmpg.com
Watson-Marlow

BredelAliteaFlexicon

MasoSine

Watson-Marlow online
Our engineers around the world can help you
choose the perfect pump and tubing for your needs.

Copyright © 2010 Watson-Marlow Pumps Group

AUKETT BROCKLISS GUY HB0361

More information? Our brochures are on our website- www.wmpg.com

Watson-Marlow...Innovation in Full Flow
Tel: 800-282-8823
support@wmpg.us

Watson-Marlow Pumps Group
37 Upton Technology Park, Wilmington, MA 01887

www.wmpg.com

The information in this document is believed to be correct, but Watson-Marlow
Pumps Group accepts no liability for any errors it contains, and reserves the right toalter specifications without notice. WARNING: These products are not designed foruse in, and should not be used for, patient-connected applications. Watson-Marlow,Pumpsil, PureWeld, LoadSure, LaserTraceability, Bioprene and Marprene areregistered trademarks of Watson-Marlow Limited. STA-PURE PCS and STA-PURE
PFL are trademarks of WL Gore and Associates Inc.