TOOL CABLE

Instruction Manual

CME-001

No. CMEJE1 2018.06
This manual contains instructions for functions and usage of the CME-001. To
ensure safe and correct use, thoroughly read and understand this manual before
using this cable. To prevent accidents arising from the misuse of the cable, please
ensure the operator receives this manual.

(1) Connect a USB cable (commonly used in PC USB ports) to this cable.
USB cable CUS-100 (sold separately) is recommended.

USB cable CUS-100 (sold separately)

Safety Precautions(Be sure to read these precautions before using cable.)
CautionIndicates procedures which may lead to dangerous
conditions and cause superficial to medium injury or
physical damage or may degrade or damage theproduct, if not carried out properly(Fig. 3-1)
(2) Connect this cable to the Console connector at the top of the instrument.
SG series is used as an example. (Fig. 3-2)

Safety Precautions(1) Do not confuse the direction of the USB plug when connecting to USB portof a PC. If the plug cannot be inserted smoothly, do not force it as this may
damage the plug or port.
(2) Do not bend or pull forcibly or put any heavy object on the cable.
(3) Do not connect or remove this cable during communication. This may
cause damage or malfunction.
(4) Do not allow the metallic section of this cable to come into contact withexternal power terminals.
(5) Do not handle the cable plug with wet hands. This may result in electrical shock.

Top of SG series

Connect this cable withof SG series.

side facing backward(Fig. 3-2)
(3) Connect the USB plug of USB cable to the PC USB port.

Precautions
Caution(1) Ensure the cable is used under the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of 0 to 50 (32 to122 ) that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85 %RH• No large capacity electromagnetic switches or cables through which large
current is flowing• No water, oil, chemicals or where the vapors of these substances can comeinto direct contact with the cable• Ambient temperature of this cable – not the ambient temperature of the control
panel – must not exceed 50 (122 ) if mounted within the control panel.
(2) Ensure the USB cable matches the USB port of your PC.
USB cable CUS-100 (sold separately) is recommended.
(3) It sometimes takes time for the PC to identify the cable when this cable isconnected to the PC. However, this is not a malfunction.
Be sure to confirm the COM port number before starting communication.
(4) Use a soft, dry cloth when cleaning this cable.
(Alcohol based substances may tarnish or deface the cable.)1. Model
1.1 Model
CME-001

Serial number

2. Name and External Dimensions (Scale: mm)
Communication indicator

Communication indicator (Yellow) : Lights when sending data to the instrument.
(Fig. 2-1)
When connecting the USB cable CUS-100 (sold separately)
2000

Tool cable CME-001(Fig. 3-4)
(5) Close [Device Manager], [Hardware and Sound], and [Control Panel].
(6) Start the Console software of the instrument.
For the Console software, download it from Shinko website:(http://www.shinko-technos.co.jp/e/)
For details, consult our overseas department.

RatingUSB I/F standard
DTE speed
Socket specification
Plug specification(Fig. 1.2-1)

60(4) Confirm the COM port number as follows.
1 Click [Start] – [Control Panel].
2 Click [Hardware and Sound].
3 Click [Device Manager].
4 If [USB Serial Port (COM3)] is indicated in the [Ports (COM & LPT)],
COM port has been allotted No.3.

4. Specifications

1.2 How to Read the Model Label
The model label is attached to the case.
Model200(Fig. 3-3)

Corresponds to USB Specification 2.0.
9600/19200/38400 bps
microUSB Type B
Specific interface connector for instrument

Operating Environment
Operating System
Windows 7/8 (Japanese/English))
Personal computer
OS certified PC (Intel Pentium processor orcompatible microprocessor)
USB port equipped
General Structure
Case material
Case color
Cable color
Cable length

PCBlackBlack
Standard length: 20 cm

Other
Power source
Supply voltage
Current consumption
Ambient temperature
Ambient humidity
WeightAccessories included
Accessories soldseparately

Bus power (Supplied from PC via USB)
5 V DC
200 mA max.
0 to 50
35 to 85 %RH (Non-condensing)
Approx. 20 g
Instruction manual: 1 copy
USB cable CUS-100(microUSB Type B – USB Type A, Total length: 2 m)

USB cable CUS-100 (sold separately)(Fig. 2-2)

3. Wiring
CautionFor SGD, SGDW, SGDL, SGM, SGMW, SGML, SHD, and SHDW, if a 2-wiretransmitter is connected, the power cannot be supplied by this cable.
Be sure to supply power to the instruments.

SHINKO TECHNOS CO., LTD.
Head office: 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, Japan
TEL: +81-72-727-6100
FAX: +81-72-727-7006
URL: http://www.shinko-technos.co.jp/e/E-mail: overseas@shinko-technso.co.jp

CME-001()(1)

USBUSBNo. CMEJE2 2018.06)

CME-001(USB
CUS-100

USB(CUS-100(3-1)
(2)
SG(1)

USBUSB(3-2)

USBUSBUSBSG(2)
(3)
(4)
(5)

SG

3-2(3) USB

USB

USB

(1)

03550

3-385 RH

(4)

50(2)

USBUSB(3)

COM-USB
CUS-100(COMCOM

USB
COMLPT)USB Serial Port (COM3)
3(4)
()

1.
1.1
CME-0013-4

1.2(5)(6)(http://www.shinko-technos.co.jp)
1.2-1

2.

(200: mm)

4.
USB I/F
DTE

60OS(Windows 7/8 (OS/)(IntelPentium)USB):2-1

USB

USB Specification 2.0
9600/19200/38400 bps
microUSB
BPC

CUS-100
20 cm
2000

USB

CME-001

CUS-100((USB5 V DC
200 mA
0 50
35 85 %RH(20 g1USB
CUS-100(microUSB
B - USB)

2-2

3.
SGD, SGDW, SGDL, SGM, SGMW, SGML, SHD

SHDW)

)

A2 m)

2562-0035

251URL http://www.shinko-technos.co.jp
TEL(072)727-3991 FAX(072)727-2991)