MEDICAL SENSORSMEDICAL SENSORSTE Connectivity (TE) is one of the largest sensor companies in the world, with innovative sensor
solutions that help customers transform concepts into smart, connected creations. Electronic systems in
medical equipment, devices and probes rely on sensor signals as a basis for control activities, accurate
diagnosis and treatment. TE designs and manufactures sensors to exacting specifications for the rigors
of medical applications, with ISO 13485 certification and FDA registration for various products. Ourengineers provide full support of application-specific, standard and custom requirements, from productconcept through manufacturing.

SENSOR TECHNOLOGIES• AIR BUBBLE• FORCE• HUMIDITY• LIQUID LEVEL• PIEZO FILM
• POSITION• PRESSURE• PULSE OXIMETRY• TEMPERATURE• VIBRATIONQUALITYCERTIFICATION AND
AUDITED PROCESSES• ISO 13485• ISO 9001
• CE-MDD• FDA• CMDR-Health Canada

TE CONNECTIVITY SENSORS /// MEDICALMEDICAL APPLICATION SOLUTIONSMEDICAL PUMP TECHNOLOGY
The body and the medical devices that support them rely on liquids to flow continuously without interruption.
Infusion pumps, hemodialysis and blood flow monitoring applications are vital technologies for surviving certain
medical conditions. Sensors are embedded in various pump and flow applications to confirm the continuous, accurate
flow, detect occlusion, externally detect bubbles in lines, and measure liquid levels. These robust sensors fromTE Connectivity (TE) are easy to integrate, provide superior reliability and deliver confidence and trust in system
performance.

BLOOD PRESSURE MONITORING
Blood pressure is a key indicator of health and can provide insight into future health problems. There are two ways inwhich blood pressure can be measured: non-invasive blood pressure (NIBP) monitoring and invasive blood pressure(IBP) monitoring. TE's 1620 and 1630 series pressure sensors are fully piezoresistive pressure sensors for use ininvasive blood pressure monitoring. These sensors are designed to be used with automated assembly equipment andcan be dropped directly into a customer's disposable blood pressure housing.

BODY TEMPERATURE MEASUREMENT
External temperature measurement has advanced in technology with the addition of sensor technologies, increasing
patient comfort, improving accuracy, and creating better tools for monitoring. Its data can be crucial to patients
suffering from various conditions, from infections to hypothermia. The temperature measurement of the surfaceof the body can be accomplished using different sensor technologies. TE manufactures NTC (negative thermalcoefficient) thermistors, thermopiles, and digital temperature sensors to support the wide range of accuracy,
packaging, and performance conditions amongst the different applications.

WEARABLE TECHNOLOGY
The expectation of connectivity anytime, anywhere drives the need for wearable products. And as TE sensor
platforms bring wearable technologies to life, users become safer and healthier. To get powerful functionality to fit
inside wearable health monitoring devices, TE relies on cross-industry experience and an ongoing commitment toresearch and development. From heart pacemaker and prosthetic sensor technologies to wearable fitness bands, TE
can help you develop solutions for your wearable ideas.

PAGE 3

MEDICAL SENSORSFORCE SENSORSFS19FS20

FC22Sensor TypeOEM compression load cell

Miniature force sensor

Miniature force sensor

Dimensions (mm)ø9.50 x 3.45

30.708 x 17.272 x 8.255

Ø 26.00 x 42.00 x 19.50

Accuracy±1% FSO (CNL&H)±1% FSO (CNL&H)±1% FSO (CNL&H)

Range

1 - 7 Lbf

1 - 7 Lbf

10 - 100 Lbf

Unique Features• Low range• High overload protection• Ultra high cycle life
• Analog output• Low range• High overload protection• Ultra high cycle life
• Analog output• Low range• High overload protection• Ultra high cycle life
• Analog output

TypicalApplications

Medical devices, physical therapy,
oxygen tank, infusion pumps

Medical devices, physical therapy,
oxygen tank, infusion pumps

Medical devices, physical therapy,
oxygen tank, infusion pumps

HUMIDITY SENSORSHTU3500

HTU2xHS1101LF

Analog voltage RH and NTC temperature

Digital miniature humidity andtemperature sensor

Analog RHS humidity sensor

Dimensions (mm)27 x 11.9 x YY (Depending on theconnector, from 6 to 10.8 length)

3.0 x 3.0 x 1.0

ø10 x 6.0

Accuracy

20% to 80%RH±3% RH at 25°C
(10 to 95% RH)±0.3°C at 25°C

±2% (10%RH to 90% RH)fSensor TypeRange

0 to 100% RH

0 to 100% RH

0 to 100% RH

Unique Features• PTFE filter (Optional)• Electronics fully protected (5 V)• Multiple connector choices (JST,Samtec board through hole)• Based on HTU21• Low power consumption• Fast response time
• Very low temperature coefficient• I²C interface or PWM interface
or SDM interface• High reliability and long term stability• Fast response time
• Lead free component• Very low temperature coefficient

TypicalApplications

Plug and play transducers for
OEM medical devices

Humidifier for medical ventilator

Sleep apnea, respirator

TE CONNECTIVITY SENSORS /// MEDICALMEDICAL SENSORSPHOTO OPTIC SENSORSELM 4000

EPM 4001

Finger Clip, Disposable SpO2

Photo optic lead frame emitter

Photo optic lead frame detector

Biocompatible SpO2 sensor

Dimensions (mm)4.4 x 5.1 x 1.9

4.4 x 5.1 x 1.8

Application dependent

Accuracy

Sensor dependent

Sensor dependent

Sensor dependent

Range

660 - 940 nm

660 - 940 nm

Adult / neonatal

Unique Features• Low cost
• Dual drive• Clear epoxy lens

• Low cost
• High efficiency• Clear epoxy lens

• Soft pads
• Lightweight• Easily cleaned

TypicalApplications

Pulse oximetry, finger andear probes, disposable

Pulse oximetry, finger andear probes, disposable

Pulse oximetry

Sensor TypeTEMPERATURE SENSORSTSYS03Model 600 / G22K7MCD8TS / TSD Series

I2C, SPI, PWM, SDM(Convertable to analog voltage)

Micro-thermocouple /Micro-thermistors

Thermopiles / Single thermopile
digital output series

Dimensions (mm)

QFN16: 4 x 4 x 0.85
TDFN: 2.5 x 2.5 x 0.75

From 0.23 OD

Dia. 9.15mm x 4.3mm (body)Accuracy

Up to ±0.1°C at -5°C to 50°C

From ±0.1°C

Application dependent
Typical 1% full range (TSD only)

Range-Thermocouple type T, K /NTC from 1K to 100KΩ

-20°C to +85°C (Permanent)
20°C to +100°C (Non-permanent)
0°C to +300°C (TSD only)

Unique Features• Low power• Small size
• Calibrated and ready to use• 16-bit resolution• Welded or soldered junction(Thermocouple)• Low profile, fast response• Polyesterimide wire insulation• High signal output• Accurate reference sensors• Calibrated and ready to use,
I2C interface (TSD only)

Typical Applications

Patient monitoring, temperature logging,fluid temperature, warming blanket

Medical catheters

Medical thermometer (ear andforehead), pyrometerSensor TypePAGE 5

MEDICAL SENSORSPRESSURE SENSORS

1620, 1630

85 Flush Mount

Sensor TypeInvasive blood pressure monitoring

Media isolated pressure sensor for aggressive fluids

Dimensions (mm)1620: 11.43 x 8.13 x 4.20
1630: 12.7 x 5.08 x 3.94

Ø 17.2 x 11.4

Accuracy

1.0% FSO±0.1% FSO non-linearity

Range-30 to 300 mmHg

0 - 1, 2, 3, 7, 21, 34 bar /0 - 15, 30, 50, 100, 300, 500 psi

Unique Features• Low cost, disposable design• Supplied in tape and reel
• Compliant to AAMI spec

• High performance• High stability• Minimizes trapped volume

TypicalApplications

Disposable blood pressure, surgical procedures, ICU,
kidney dialysis machines, medical instrumentation

Dialysis machines, infusion pumps, medical systems

MS5805MS5637

MS45XX, MS55XX

Sensor Type

Miniature board mounted pressure sensors

Miniature board mounted pressure sensors

Miniature board mounted pressure sensor

Dimensions (mm)4.5 x 4.5 x 3.5

3 x 3 x 0.9

12.5 x 9.9

Accuracy±2.0 mbar at 25°C

±2.0 mbar / ±0.03 psi at 25°C

0.25% / 1% TEBRange

10 to 2K mbar

10 to 2000 mbar abs. / 0.15 to 29 psi abs.

0 – 2, 4, 5, 10, 20, 30, H2O(MS4515/DO)
0 – 1, 2, 4, 5, 10, 30, 50, 150 psi(MS4525/DO)

Unique Features• 24-bit digital sensor• 20 cm resolution• Supply voltage: 1.8 to 3.6 V• Sealing designed for 2.5 x 1 mm o-ring• Silicone gel protection• Waterproof• 24-bit digital sensor• Altitude resolution of less than 15 cm• Supply voltage: 1.5 to 3.6 V• Low power, 0.6 μA (Standby≤ 0.1 μA at 25°C)• Digital temperature readout• MS4515/25(12-bit DAC analog)• MS4515DO/25DO(14-bit digital SPI or I2C)
• MS5525DSO(24-bit digital SPI or I2C)
• Wide supply voltage: 1.8 to 5.7 VDC• Small package footprint• Varied port configurations

TypicalApplications

Fall detection, pneumatic handheld drills,
respirators / ventilators

Fall detection, pneumatic handheld drills,respirators / ventilators

Medical instruments, respirators / ventilators

TE CONNECTIVITY SENSORS /// MEDICALMEDICAL SENSORSPOSITION SENSORSKMT32B

MS32, KMA36

KMXP Series

SM, SPSensor Type

Angular sensor

Magnetoresistive linear
and angular sensors

Magnetic linear position sensor

Cable extension transducer

Dimensions (mm)

TDFN: 2.5 x 2.5 x 0.8 /TSSOP20: 5 x 4 x 1.75

TDFN: 2.5 x 2.5 x 0.8 /TSSOP20: 6.5 x 6.4 x 1.2

TDLMP12 6.0 x 2.0 x 0.7543 x 45 x 68

Accuracy

Typ. 0.1° to 1.0°

Typ. 0.1 kA/m / typ. 0.3°

10-50 μm±0.25% to ±1%

Range

180° angle

1 to 3 kA/m magnetic
switching field / 360° angle

1-5mm

0 - 2.5 to 0 - 50 inches

Unique Features• High accuracy• High resolution• Ultra low cost
• Ultra small size
• High accuracy• Digital / analog output• Magnetoresistive (MR)• High precision and resolution• Easy assembly• Superior performance• Compact design• Low cost, high value stringpot• Custom configurations
available for
OEM customersTypicalApplications

Various position control
applications

Various position control
applications

Surgical robots, prosthetics,
syringe pumps

Medical imaging systems,
surgical robots

ULTRASONIC SENSORVIBRATION SENSOR

AD-101MiniSense 100

Sensor TypeUltrasonic
air-in-liquid detectors

Piezoelectrical film sensors

Dimensions (mm)Application dependent

19.05 x 6.35 x 6.35

Accuracy

Application dependent±20.0% (Typical)

Range

Detects bubbles 4 μl and larger (Standard; consult
factory for 1 μl and smaller bubble size)±10 g (Typical)

Unique Features• Bubble detection from 1 mm tube
• Integral electronics• Occlusion option• Fluid differentiation• 3.3 and 5 V input option• Very low cost
• High sensitivity (1 V/g)
• Ultra low power (Self generating)

TypicalApplications

Infusion pumps, dialysis machines, apheresis,
auto-transfusion, 3D printing

Wake-up switch, impact sensing, vital signs monitoring

PAGE 7

te.com/MedicalSensors© 2023 TE Connectivity. All Rights Reserved.
TE Connectivity, TE, and the TE connectivity (logo) are trademarks of the TE Connectivity Ltd. family of companies. Other logos, product and company names mentioned herein
may be trademarks of their respective owners.
While TE has made every reasonable effort to ensure the accuracy of the information in this brochure, TE does not guarantee that it is error-free, nor does TE make any otherrepresentation, warranty or guarantee that the information is accurate, correct, reliable or current. TE reserves the right to make any adjustments to the information contained
herein at any time without notice. TE expressly disclaims all implied warranties regarding the information contained herein, including, but not limited to, any implied warranties
of merchantability or fitness for a particular purpose. The dimensions in this brochure are for reference purposes only and are subject to change without notice. Specifications
are subject to change without notice. Consult TE for the latest dimensions and design specifications.
TE-SEN-MED102 03/2023

TE CONNECTIVITY SENSORS /// MEDICAL