PRODUCT FLYER

Multifunction Input
and Output Devices

CONTENTS
Multifunction I/O Device

PXI Multifunction I/O Module
Platform-Based Approach to Test and Measurement
Hardware Services

Multifunction I/O Device

 Software: Includes DAQExpress™ interactive
measurement software as well as API support for
LabVIEW and text-based languages, shipping examples,and detailed help files
 Analog, Digital and Counter/Timer I/O in one device
 USB or PCI/PCIe bus connectivity

 Voltage measurements up to 10 MS/s per channel
 Multiplexed or simultaneous analog architectures
 Software-selectable input ranges and input channel
isolation available Up to four analog output channels and four counters

Built for Accuracy and Reliability
NI Multifunction I/O Devices provide a combination of analog I/O, digital I/O, and counter/timer
functionality in a single device for computer-based systems.
Multifunction I/O devices offer a mix of I/O with varying channel counts, sample rates, output rates, andother features to meet many common measurement requirements. These devices are ideal for a wide
variety of industry applications such as laboratory automation, research, and design verification. The
included DAQExpress interactive measurement software enables quick hardware setup and datavisualization, while the included NI-DAQmx driver enables complete customization of measurement andautomation applications from a variety of supported programming languages.

Page 2| ni.com | Multifunction I/O

Table 1. NI USB multifunction I/O devices provide a simple plug-in interface with many specification options.
ANALOG INPUTNO.

SIMUL-SAMPLECHANNELS TANEOUS

RATE

RESOLUTION

ABSOLUTE
ACCURACYANALOG OUTPUTDIO

COUNTERSUPDATENO.

NO.

NUM CH.

USB-6366

8

yes

2 MS/s

16 bits

2.69 mV2

USB-6363

16

no2 MS/s

16 bits

1.66 mV4

USB-6361

16

no2 MS/s

16 bits

1.66 mV2

USB-6356

8

yes

16 bits

2.69 mV2

USB-6353

32

no16 bits

1.52 mV4

USB-6351

16

no16 bits

1.52 mV2

USB-6343

32

no500 kS/s

16 bits

2.19 mV4

USB-6341

16

no500 kS/s

16 bits

2.19 mV2

USB-628932

no625 kS/s

18 bits

0.98 mV4

USB-628116

no625 kS/s

18 bits

0.98 mV2

USB-625580

no16 bits

1.92 mV2

USB-622580

no16 bits

3.1 mV

2

Page 3

1.25
MS/s
1.25
MS/s
1.25
MS/s

1.25
MS/s
250 kS/s| ni.com | Multifunction I/O

RATE

3.33

CONNECTIVITY*

CHANNELS CHANNELS

24

4ST,MT,BNC

48

4

ST,MT,BNC

24

4

ST,MT,BNC

24

4

ST,BNC

48

4

ST

24

4

ST

900 kS/s

48

4

ST,BNC

900 kS/s

24

4

ST,BNC

48

2

ST,MT

24

2

ST,MT

24

2

ST,MT

24

2

ST,MT

MS/s
2.86
MS/s
2.86
MS/s
3.33
MS/s
2.86
MS/s
2.86
MS/s

2.86
MS/s
2.86
MS/s
2.86
MS/s
833 kS/sUSB-6218

32

no250 kS/s

16 bits

2.69 mV2

250 kS/s

USB-6216

16

no400 kS/s

16 bits

2.71 mV2

250 kS/s

USB-6215

16

no250 kS/s

16 bits

2.69 mV2

250 kS/s

USB-621216

no400 kS/s

16 bits

2.71 mV2

250 kS/s

USB-621116

no250 kS/s

16 bits

2.69 mV2

250 kS/s

USB-621016

no250 kS/s

16 bits

2.69 mV

0-

USB-6003

8

no100 kS/s

16 bits

26 mV

2

5 kS/sUSB-6002

8

no50 kS/s

16 bits

26 mV

2

USB-6001

8

no

20 kS/s

14 bits

26 mV

USB-6000

8no10 kS/s

12 bits

26 mV

8 in/8

2

ST,BNC

2

ST,MT,BNC

2

ST

2

ST,MT,BNC

2

ST

2

ST

13

1

ST

5 kS/s

13

1

ST

2

5 kS/s

13

1

ST

0

-

4

1

ST

out
32
4 in/4out
24
4 in/4out
4 in/4out*ST=Screw Terminals | MT=Mass Termination | BNC=BNC Terminals**Has only dedicated digital input and/or digital output lines
Table 2. NI PCI(e) multifunction I/O devices add high quality I/O to a desktop PC.
ANALOG INPUTNO.

SIMUL-SAMPLECHANNELSTANEOUSRATE

no2 MS/s

PCIe-6363 32

RESOLUTION

16 bits

ANALOG OUTPUTDIO

COUNTERSABSOLUTENO.

UPDATENO.

NO.

ACCURACY

CHANNELSRATE

CHANNELSCHANNELS

1.66 mV

4

2.86

48

4

24

4

MS/s

PCIe-6361 16

no2 MS/s

16 bits

1.66 mV

2

2.86
MS/s

Page 4| ni.com | Multifunction I/O

PCIe-6353 32

no

1.25

16 bits

1.52 mV

4

MS/s

PCIe-6351 16

no

1.25

no

500

16 bits

1.52 mV

2

no

500

16 bits

2.19 mV

4

no

250

16 bits

2.19 mV

2

no

250

16 bits

2.2 mV

4

no

260

4

900

48

4

900

24

4

900

48

4

24

4

kS/s

16 bits

2.2 mV

2

kS/s

PCIe-6320 16

24

kS/s

kS/s

PCIe-6321 16

2.86

kS/s

kS/s

PCIe-6323 32

4

MS/s

kS/s

PCIe-6341 16

48

MS/s

MS/s

PCIe-6343 32

2.86

900
kS/s

16 bits

2.2 mV

0

-

24

4

16 bits

0.98 mV

4

2.86

48

2

kS/s

PCI-6289

32

no

625
kS/s

PCI-6284

32

no

625

MS/s

18 bits

0.98 mV

0

-

48

2

18 bits

0.98 mV

2

2.86

24

2

kS/s

PCI-6281

16

no

625
kS/s

Page 5| ni.com | Multifunction I/O

MS/s

PCI-6280

16

no

625

18 bits

0.98 mV

0

-

24

2

16 bits

1.92 mV

2

2.86

24

2

48

2

24

2

24

2

0

2

kS/s

PCI-6255

80

no

1.25
MS/s

PCI-6229

32

no

250

MS/s

16 bits

3.1 mV4

kS/s

PCI-6225

80

no

250

kS/s

16 bits

3.1 mV2

kS/s

PCI-6221

16

no

250

0

yes

250

16 bits

3.1 mV

2

0

yes

250

833
kS/s

16 bits

5.28 mV4

kS/s

PCI-6143

833
kS/s

kS/s

PCI-6154

833

250
kS/s*

16 bits

3.61 mV

0

-

8

2

14 bits

4.66 mV

0

-

8

2

14 bits

4.66 mV

0

-

8

2

16 bits

4.96 mV

0

-

8

2

kS/s

PCI-6133

8

yes

2.5
MS/s

PCI-6132

4

yes

2.5
MS/s

PCI-6123

8

yes

500
kS/s

Page 6| ni.com | Multifunction I/O

PCI-6122

4

yes

500

16 bits

4.96 mV

0

-

8

2

kS/s

PCI-61204yes

1 MS/s

16 bits

76 mV

2

4 MS/s8

2

PCI-6115

4

yes

10 MS/s

12 bits

184 mV2

4 MS/s8

2

PCI-61112

yes

5 MS/s

12 bits

252 mV2

4 MS/s8

2

PCI-61104yes

5 MS/s

12 bits

252 mV2

4 MS/s8

2

PCI-6010

16

no

200

16 bits

5.08 mV

2

-

0

2

kS/s

*Simultaneous – sample rate is per output channel

Page 7| ni.com | Multifunction I/ODetailed Views of Multifunction I/O Devices
USB Multifunction I/O Device, Mass Termination Variant**Screw terminal and BNC terminal variants also available

Bus-Powered USB Multifunction I/O Device

Page 8| ni.com | Multifunction I/O

PCI Express Multifunction I/O Device

Page 9| ni.com | Multifunction I/OKey Features

High Resolution, High Accuracy Analog Measurements
NI Multifunction I/O Devices have analog signal paths that have been meticulously designed, tested,
and calibrated to ensure the highest possible accuracy is achieved across all input channels. Becauseof this thorough design-and-test philosophy, NI is able to provide thoroughly documented accuracy
specifications to allow a complete understanding of device performance under a range of possible
operating conditions. The specifications documentation for each device provides a section dedicated tounderstanding the calculation of AI absolute accuracy, alongside a wealth of other performance details.

Advanced Timing Technology
All multifunction data acquisition hardware requires onboard timing circuitry to control analog, digital,and counter I/O lines. Much of the Multifunction I/O Device family features the NI-STC3 chipset toprovide up to four enhanced counters, a 100 MHz timebase, and additional options for I/O timing andtriggering.

Figure 1. Modern NI DAQ devices feature NI-STC3 timing and synchronization technology – a USB Multifunction I/O
Device is shown here as an example.

100 MHz Timebase
The onboard timebase of any data acquisition device acts as the internal heartbeat that drives all digital
circuitry. Everything from sample clocks to trigger lines uses the timebase as an onboard reference togenerate clock frequencies and latch digital edges. Modern devices use a 100 MHz timebase for all
analog and digital timing, which is up to five times faster than their predecessors. This means thatsampling frequencies are up to five times more accurate, and analog triggers can respond within 10 ns
of a trigger condition being met.

Page 10| ni.com | Multifunction I/OIndependent Timing Engines for Analog and Digital I/OAdvanced timing and triggering functionality on data acquisition devices has often relied on onboard
counters and complex signal routing to achieve specialized hardware-timed performance. NI’s
multifunction I/O technology offers completely independent sample clocks and triggers for each different
group of I/O on a multifunction device. Retriggerable acquisitions, for example, involve waiting for a
trigger condition to be met, taking a finite number of samples, and then immediately rearming the trigger
for the next acquisition. Using driver software function calls to rearm the trigger risks missing the next
trigger due to software latency; therefore, the best possible performance requires a hardware-timed
approach. In the past, counters were the only way to implement hardware-timed retriggering, and socounters would be used to generate a retriggerable pulse train, which was then internally routed to act
as the analog input sample clock. Devices with advanced timing capabilities can perform this type ofaction natively, which greatly simplifies the hardware programming and measurement configuration
process.

Self-Calibration and Two-Year Guaranteed Specifications
Self-calibration provides the ability to characterize nonlinearity, gain, and offset errors in Multifunction
I/O Devices. These changes are caused by deviations in the operating environment as well as
manufacturing variations in the integrated circuitry that may have shifted since the last external
calibration. Supported devices use an integrated self-calibration algorithm called NI-MCal tocharacterize and save the correction polynomials to an onboard EEPROM, and does so in a matter of
seconds. This allows subsequent measurements to be scaled automatically by the device driver
software before being returned to the user through application software. NI-MCal has the unique ability
to return calibrated data from every channel in a scan, even if they are at different input ranges. Thismeans a device can easily load and apply channel-specific correction functions even while scanning at
maximum device rates without impacting performance.

By eliminating the limitations of hardware components traditionally used for device error correction andusing the power and speed of software and PC processing, NI-MCal raises the bar for measurement
accuracy by redefining device self-calibration. Most Multifunction I/O Device models have a two-year
external calibration cycle thanks to the self-calibration precision circuitry that minimizes the maintenanceburden of deployed systems, while also maintaining tight measurement tolerances. Visit ni.com to learn
more about NI’s calibration services.

Page 11| ni.com | Multifunction I/ONI-DAQmx Application Programming Interface (API)The NI-DAQmx driver includes a best-in-class API that works directly with a variety of development
options including LabVIEW, DAQExpress, C, C#, Python, and others. The native integration provides
exceptional performance and a seamless experience without the need for manual wrapping of functions.
To ensure long-term interoperability of DAQ devices, the NI-DAQmx driver API is the same API used for
all NI DAQ products – meaning re-development efforts can be minimized regardless of hardware
changes or upgrades. Additionally, the driver provides access to help files, documentation, and dozens
of ready-to-run shipping examples you can use as a starting point for your application.

DAQExpress Companion Software
DAQExpress is interactive companion software included with the purchase of a supported hardware
product. It provides quick, clear access to all the measurements supported by a DAQ device as soon as
it is plugged it, and allows you to get instant access to the measurement data and apply analysis
functions without writing any code. All USB and PCI(e) multifunction I/O devices are supported by, andship with, DAQExpress and the NI-DAQmx driver.

Page 12| ni.com | Multifunction I/OPXI Multifunction I/O Module

 Software: Includes API support for LabVIEW
and text-based languages, complete withshipping examples and detailed help files.
 Voltage measurements up to 10 MS/s perchannel Analog, Digital and Counter/Timer I/O in one
devicePage 13| ni.com | Multifunction I/O Multiplexed or simultaneous analog
architectures
 Software-selectable input ranges and input
channel isolation available Up to four analog output channels and four
counters/timers
 PXI and PXI Express platform compatibility

Built for Accuracy and Reliability
NI PXI Multifunction I/O Modules provide a combination of analog I/O, digital I/O, and counter/timer
functionality in a single device for PXI-based systems.
PXI Multifunction I/O Modules offer a mix of I/O with varying channel counts, sample rates, output rates,and other features to meet many common measurement requirements. These devices are ideal for awide variety of industry applications such as laboratory automation, research, and design verification.
The included NI-DAQmx driver enables complete customization of measurement and automation
applications from a variety of supported programming languages.
Table 2. NI PXI(e) multifunction I/O devices add high quality I/O to a PXI System.
ANALOG INPUTNO.

SIMUL-ANALOG OUTPUTSAMPLECHANNELS TANEOUS RATE
PXIe-

16

Yes

3.57 MS/s

RESOLUTION

16 bitsNO.

ACCURACY CH.

RATE

CHANNELS CHANNELS

2.69 mV

3.3

48

4

24

4

24

4

48

4

24

4

24

4

48

4

4

MS/s*

8

Yes

3.57 MS/s

16 bits

2.69 mV

2

208

No

3.86 MS/s

16 bits

1.66 mV

2

16

Yes

2 MS/s

16 bits

2.69 mV

48

Yes

2 MS/s

16 bits

2.69 mV

2

3.3
MS/s*

144

No

2 MS/s

16 bits

1.52 mV

2

6365

PXIe-

3.3
MS/s*

6366

PXIe-

2.86
MS/s

6368

PXIe-

3.3
MS/s*

6375

PXIe-

NO.

UPDATE

6376

PXIe-

COUNTERSABSOLUTE NO.

6378

PXIe-

DIO

2.86
MS/s

32

No

2 MS/s6363

Page 14 | ni.com | Multifunction I/O Flyer

16 bits

1.66 mV

4

2.86
MS/s

PXIe-

16

No

2 MS/s

16 bits

1.66 mV

2

6361

PXIe-

16

Yes

1.25 MS/s

16 bits

2.69 mV

4

8

Yes

1.25 MS/s

16 bits

2.69 mV

2

80

No1.25 MS/s

16 bits

1.52 mV

2

32

Yes

500 kS/s

16 bits

3.26

2

mV

80

No

500 kS/s

16 bits

1.52 mV

2

3.3

24

4

2.86

24

4

900

24

4

2.86

24

4

24

4

MS/s

16

No

500 kS/s

16 bits

2.19 mV

2

6341

PXIe-

4

kS/s

6345

PXIe-

48

MS/s

6349

PXIe-

3.3

MS/s*

6355

PXIe-

4

MS/s*

6356

PXIe-

24

MS/s

6358

PXIe-

2.86

900
kS/s

4

Yes

4 MS/s

16 bits

3.15 mV

2

4 MS/s*

24

2

32

No

625 kS/s

18 bits

0.98 mV

4

2.86

48

2

48

2

24

2

24

2

6124

PXI6289

PXI-

MS/s

32

No

625 kS/s

18 bits

0.98 mV

0

6284

PXI-

MS/s

16

No

625 kS/s

18 bits

0.98 mV

2

6281

PXI-

2.86

2.86
MS/s

16

No

625 kS/s

6280

Page 15 | ni.com | Multifunction I/O Flyer

18 bits

0.98 mV

0

2.86
MS/s

PXI-

8

No

250 kS/s

16 bits

-

2

6239

PXI-

8

No

250 kS/s

16 bits

-

2

4

No

250 kS/s

16 bits

-

4

16

No

250 kS/s

16 bits

3.1 mV2

16

No

250 kS/s

16 bits

3.1 mV2

8No

250 kS/s

16 bits

3.1 mV

4

32

No

250 kS/s

16 bits

3.1 mV

4

0

2

500

0

2

500

0

2

500

0

2

833

48

2

48

2

kS/s

80

No

250 kS/s

16 bits

3.1 mV

4

6225

PXI-

500

kS/s

6229

PXI-

2

kS/s

6230

PXI-

0

kS/s

6232

PXI-

500

kS/s

6233

PXI-

2

kS/s

6236

PXI-

0

kS/s

6238

PXI-

500

833
kS/s

32

No

250 kS/s

16 bits

3.1 mV

0

-

48

2

8

Yes

250 kS/s

16 bits

3.61 mV

0

-

8

2

8

Yes

2.5 MS/s

14 bits

4.66 mV

0

-

8

2

4

Yes

2.5 MS/s

14 bits

4.66 mV

0

-

8

2

6224

PXI6143

PXI6133

PXI6132

Page 16 | ni.com | Multifunction I/O Flyer

PXI-

8

Yes

500 kS/s

16 bits

4.96 mV

0

-

8

2

4

Yes

500 kS/s

16 bits

4.96 mV

0

-

8

2

4

Yes

1 MS/s

16 bits

76 mV

2

4 MS/s

8

2

4

Yes

10 MS/s

12 bits

185 mV2

4 MS/s

8

2

6123

PXI6122

PXI6120

PXI6115*Simultaneous – sample rate is per output channel

Detailed View of PXI Express Multifunction I/O Module

Page 17 | ni.com | Multifunction I/O Flyer

Key Features
High Resolution, High Accuracy Analog Measurements
NI PXI Multifunction I/O Modules have analog signal paths that have been meticulously designed,
tested, and calibrated to ensure the highest possible accuracy is achieved across all input channels.
Because of this thorough design-and-test philosophy, NI is able to provide thoroughly documented
accuracy specifications to allow a complete understanding of device performance under a range ofpossible operating conditions. The specifications documentation for each Device provides a section
dedicated to understanding the calculation of AI absolute accuracy, alongside a wealth of otherperformance details.

Advanced Timing Technology
All multifunction data acquisition hardware requires onboard timing circuitry to control analog, digital,and counter I/O lines. Much of the PXI Multifunction I/O Module family features the NI-STC3 chipset toprovide up to four enhanced counters, a 100 MHz timebase, and additional options for I/O timing andtriggering.

Figure 1. Modern NI DAQ modules feature NI-STC3 timing and synchronization technology.

100 MHz Timebase
The onboard timebase of any data acquisition device acts as the internal heartbeat that drives all digital
circuitry. Everything from sample clocks to trigger lines uses the timebase as an onboard reference togenerate clock frequencies and latch digital edges. Modern devices use a 100 MHz timebase for all
analog and digital timing, which is up to five times faster than their predecessors. This means thatsampling frequencies are up to five times more accurate, and analog triggers can respond within 10 ns
of a trigger condition being met.

Independent Timing Engines for Analog and Digital I/OAdvanced timing and triggering functionality on data acquisition devices has often relied on onboard
counters and complex signal routing to achieve specialized hardware-timed performance. NI’s
multifunction I/O technology offers completely independent sample clocks and triggers for each different

Page 18 | ni.com | Multifunction I/O Flyergroup of I/O on a multifunction device. Retriggerable acquisitions, for example, involve waiting for a
trigger condition to be met, taking a finite number of samples, and then immediately rearming the trigger
for the next acquisition. Using driver software function calls to rearm the trigger risks missing the next
trigger due to software latency; therefore, the best possible performance requires a hardware-timed
approach. In the past, counters were the only way to implement hardware-timed retriggering, and socounters would be used to generate a retriggerable pulse train, which was then internally routed to act
as the analog input sample clock. Devices with advanced timing capabilities can perform this type ofaction natively, which greatly simplifies the hardware programming and measurement configuration
process.

Page 19 | ni.com | Multifunction I/O Flyer

Synchronization and Integration
PXI Multifunction I/O Modules use the inherent timing and synchronization capabilities of the PXI
platform to synchronize measurements between multiple modules, which is ideal for high-channel-countapplications. PXI builds on its CompactPCI architecture base by adding integrated timing andsynchronization that is used to route synchronization clocks and triggers internally. A PXI chassis
incorporates a dedicated 10 MHz system reference clock, PXI trigger bus, star trigger bus, and slot-toslot local bus, while a PXI Express chassis adds a 100 MHz differential system clock, differential
signaling, and differential star triggers to address the need for advanced timing and synchronization.

Self-Calibration and Two-Year Guaranteed Specifications
Self-calibration provides the ability to characterize nonlinearity, gain, and offset errors in PXIMultifunction I/O Modules. These changes are caused by deviations in the operating environment aswell as manufacturing variations in the integrated circuitry that may have shifted since the last external
calibration. Supported modules use an integrated self-calibration algorithm called NI-MCal tocharacterize and save the correction polynomials to an onboard EEPROM, and does so in a matter of
seconds. This allows subsequent measurements to be scaled automatically by the device driver
software before being returned to the user through application software. NI-MCal has the unique ability
to return calibrated data from every channel in a scan, even if they are at different input ranges. Thismeans a device can easily load and apply channel-specific correction functions even while scanning at
maximum device rates without impacting performance.

Page 20 | ni.com | Multifunction I/O FlyerBy eliminating the limitations of hardware components traditionally used for device error correction andusing the power and speed of software and PC processing, NI-MCal raises the bar for measurement
accuracy by redefining device self-calibration. Most PXI Multifunction I/O Module models have a twoyear external calibration cycle thanks to the self-calibration precision circuitry that minimizes themaintenance burden of deployed systems, while also maintaining tight measurement tolerances. Visitni.com to learn more about NI’s calibration services.

NI-DAQmx Application Programming Interface (API)The NI-DAQmx driver includes a best-in-class API that works directly with a variety of development
options including LabVIEW, DAQExpress, C, C#, Python, and others. The native integration provides
exceptional performance and a seamless experience without the need for manual wrapping of functions.
To ensure long-term interoperability of DAQ devices, the NI-DAQmx driver API is the same API used for
all NI DAQ products – meaning re-development efforts can be minimized regardless of hardware
changes or upgrades. Additionally, the driver provides access to help files, documentation, and dozens
of ready-to-run shipping examples you can use as a starting point for your application.

What Is PXI?
Platform-Based Approach to Test and Measurement
Powered by software, PXI is a rugged PC-based platform for measurement and automation systems.
PXI combines PCI electrical-bus features with the modular, Eurocard packaging of CompactPCI andthen adds specialized synchronization buses and key software features. PXI is both a high-performanceand low-cost deployment platform for applications such as manufacturing test, military and aerospace,machine monitoring, automotive, and industrial test. Developed in 1997 and launched in 1998, PXI is an
open industry standard governed by the PXI Systems Alliance (PXISA), a group of more than 70
companies chartered to promote the PXI standard, ensure interoperability, and maintain the PXI
specification.

Page 21 | ni.com | Multifunction I/O Flyer

Integrating the Latest Commercial Technology
By leveraging the latest commercial technology for our products, we can continually deliver highperformance and high-quality products to our users at a competitive price. The latest PCI Express Gen
3 switches deliver higher data throughput, the latest Intel multicore processors facilitate faster and more
efficient parallel (multisite) testing, the latest FPGAs from Xilinx help to push signal processing
algorithms to the edge to accelerate measurements, and the latest data converters from TI and ADI
continually increase the measurement range and performance of our instrumentation.

Page 22 | ni.com | Multifunction I/O Flyer

Hardware Services

All NI hardware includes a one-year warranty for basic repair coverage, and calibration in adherence to
NI specifications prior to shipment. PXI systems also include basic assembly and a functional test. NI
offers additional entitlements to improve uptime and lower maintenance costs with service programs for
hardware. Learn more at ni.com/services/hardware.
STANDARD

PREMIUM

Program3 or 53 or 5

Duration

years

years

●

●

Extended Repair
Coverage

SystemConfiguration,
Assembly, and

DESCRIPTION

Length of service program

NI restores your device’s functionality and includes
firmware updates and factory calibration.

NI technicians assemble, install software in, and test●●

your system per your custom configuration prior to
shipment.

Test1

Advanced●

Replacement

2

NI stocks replacement hardware that can be shipped
immediately if a repair is needed.

System Return
Material●

Authorization

NI accepts the delivery of fully assembled systems
when performing repair services.

(RMA)

1

Calibration
Plan (Optional)

NI performs the requested level of calibration at theStandard

Expedited3

specified calibration interval for the duration of theservice program.

1This option is only available for PXI, CompactRIO, and CompactDAQ systems.
2This option is not available for all products in all countries. Contact your local NI sales engineer to confirm

availability.

3Expedited calibration only includes traceable levels.

Page 23 | ni.com | Multifunction I/O Flyer

PremiumPlus Service Program
NI can customize the offerings listed above, or offer additional entitlements such as on-site calibration,custom sparing, and life-cycle services through a PremiumPlus Service Program. Contact your NI sales
representative to learn more.

Technical Support
Every NI system includes a 30-day trial for phone and e-mail support from NI engineers, which can beextended through a Software Service Program (SSP) membership. NI has more than 400 support
engineers available around the globe to provide local support in more than 30 languages. Additionally,take advantage of NI’s award winning online resources and communities.

Page 24 | ni.com | Multifunction I/O Flyer