Test Equipment andSeismometers

Rockwell Hardness Testing Machines

HR-200/300/400 Series

Catalog No.E17000(2)Rockwell hardness testing machine series Versatile smart models

4 models to suit practically every

application

HR-210MR

HR-320MS

HR-430MR

HR-430MS

Rockwell hardness testing machine

Dual type (Rockwell/Rockwell superficial)hardness testing machine

Rockwell hardness testing machine

Dual type (Rockwell/Rockwell superficial combined use)
hardness test equipment

Manually handles test force and preliminary test force
selection. Motor drive controls loading sequence.
Preliminary test force (handling support):
Automatic pre-setting dial gage

Manually handles test force and preliminary test force
selection. Motor drive controls loading sequence.
Preliminary test force (handling support): Loading
navigator indication

Smart type, but equipped with switching power steeringand automatic brake handle for auto start feature.
Motor drive controls loading sequence.

Smart type, but equipped with switching power steeringand automatic brake handle for auto start feature.
Motor drive controls loading sequence.

Features
The newly designed frame provides maximum clearance for positioning
the workpiece. A flat table is all that is needed for mounting these testingmachines.

The digital type (HR-430MR and HR-430MS), withthe adoption of the automatic brake handle and load
sequence, with only simple handling test possibility.

The digital type (HR-320MS, HR-430MR and HR-430MS) has
digimatic output and our Digimatic Mini-Processor(DP-1VA LOGGER) to print and input tools (USB-ITN-E) toconnect to a PC and use for data transfer.

Simple to operateWith analogue type (HR-210MR), the gauge presetting operation is unnecessary by the adoptionof an automatic presetting dial gauge.

Supports Brinell hardness tests. Brinell hardness tests can be
performed using the optional Brinell weight set and ball indenter.
The microscope used for measurement must be prepared bythe user.
23

Specifications
Model No.

Optional accessories
HR-210MR

HR-320MS

HR-430MR

HR-430MS

RockwellType of hardness test—
98.07—

Rockwell superficial—
29.42 98.07
98.07
147.1 294.2 441.3—
588.4 980.7 1471
IS B 7726 ISO6508-2 ASTM E18-20
Digital0.1 HR graduation

Rockwell superficial
29.42 98.07
147.1 294.2 441.3

Preliminary test force (N)Test force (N) Superficial
RockwellStandard
JIS B 7726 ISO6508-2
Display
AnalogMinimum reading
0.5 HR graduation
Preliminary test force
Automatic pre-setting dial gauge Loading navigator indication
Automatic brake handle(handling support)
Preliminary test force setting—
Dial switching—
Dial switching
Total test force setting
Weight change
Dial switching
Motor drive
Motor drive
Total test force control
Button start
Automatic start
3-5.5 s setting
1-99 s setting
Test force duration
ManualManualMaximum specimen dimensions
Height 180 mm (100 mm if the cover is attached), depth 165 mm (from indenter axis to the frame)—
Success or failure decision function
Function—
Offset revision function—Hardness conversion function
Data output—
Digimatic RS-232C
Power supply
AC Input : 100-240 V, 50-60 Hz, 1.8 A, DC Output: 12 V – 4.17 A
Dimensions
214 (W) × 512 (D) × 780 (H) mm
Mass
Approximately 47 kg
Approximately 47 kg
Approximately 50 kg
Note1: Plastic tests may not be supported depending on the plastic material.
Note2: Brinell hardness tests can be performed by using a Brinell weight set, Brinell indenter, and a measurement microscope.
Note3: The testing machines are supplied without indenters. Please prepare optional indenters.

Standard accessories
Order No.
810-039810-040

Item
Flat anvil
V-anvil (large)

357651

AC adapter

Specifications
ø64 mm
ø40 mm,120 º V-groove width 30 mm
AC100-240 V 1.2 A
DC12 V 3.5 A

Order No.
—383876——

Item
User's manual
Vinyl cover
Accessory box
Level

Specifications(Depends on the model)———

External dimensions

Spare ball for indenters
Order No.
19BAA082
19BAA083
19BAA084
19BAA085
19BAA507
19BAA508
19BAA509
19BAA510
11AAE323
11AAE324
11AAE325
11AAE326

Itemø1/16” Steel ball (spare)ø1/8” Steel ball (spare)ø1/4” Steel ball (spare)ø1/2” Steel ball (spare)ø1/16” Spare carbide ball
ø1/8” Spare carbide ball
ø1/4” Spare carbide ball
ø1/2” Spare carbide ball
ø1/16” Spare carbide ball ASTM
ø1/8” Spare carbide ball ASTM
ø1/4” Spare carbide ball ASTM
ø1/2” Spare carbide ball ASTM

Order No. 810-038

HR-320MS
474.5

Order No.
Item
JIS JIS/ISO ASTM/ISO
Remark19BAA072
Diamond indenter
Compliant with ISO/JIS standards, only for Rockwell hardness tests
19BAA073
Diamond indenter
Compliant with ISO/JIS standards, also for superficial hardness tests
11AAD461
ø1/16” Steel ball indenter
11AAD462
ø1/8” Steel ball indenter
Compliant with JIS standards
11AAD463
ø1/4”Steel ball indenter
11AAD464
ø1/2” Steel ball indenter
11AAD465
ø1/16” Carbide ball indenter
11AAD466
ø1/8” Carbide ball indenter
Compliant with ISO standards
11AAD467
ø1/4” Carbide ball indenter
11AAD468
ø1/2” Carbide ball indenter
11AAE318
Rockwell diamond indenter ASTM
11AAE319
ø1/16” Carbide ball indenter ASTM
Compliant with ASTM/ISO standards11AAE320
ø1/8” Carbide ball indenter ASTM
With class B calibration certificate and inspection certificate
11AAE321
ø1/4” Carbide ball indenter ASTM
11AAE322
ø1/2” Carbide ball indenter ASTM
Note1: indicates indenters for HR-320MS and HR-430MS.
Note2: Indenters compliant with ASTM/ISO standards are for HR-320MS, HR-430MR, and HR-430MS.

13502.5

16215013Round table ø250 mm

780

212.8

180352.4

235

10 pcs./set

1 pc./set

1 pc./setCompliant with ASTM/ISO standards
With class B calibration certificate and inspection certificate

Order No. 810-037
Round table ø180 mm

Order No. 810-043(ø40 mm, groove width 30 mm)
For shaft material (ø9 to 58 mm)(ø12 mm)

V anvil (large)

Spot anvil

For large specimens

Order No. 810-041

Order No. 810-044(ø40 mm, groove width 6 mm)
For shaft material (ø3 to 7 mm)(ø5.5 mm)
For sheet specimens

V anvil (small)

Order No. 810-040

Spot anvil

Order No. 810-028

Jack rest

For testing long specimens
using round table or anvil

(30)

(30)

218

218

113

113

780

180

352

Maximum specimen
heightFor large specimens
235

RemarkSpecimen fixturesHR-210MR
162150Indenters for Rockwell hardness testing434(512)214

65

214HR-430MR

434(512)65

Order No. 810-027

HR-430MS502.5

16215013

VARI-REST
13502.5

162
150

Order No. 810-029

Special V anvil(Length 400 mm, groove width 50 mm)
For shaft material (Max. ø100 mm)

Order No. 810-026

Fine adjustment table for Jominy test
JIS G 0561 For steel
hardenability testing

(30)

(30)

218

218

113

780

352.4
180780

235

113

235

180352.4

For testing long specimens using anvil

2144434(512)65214434(512)

Order No. 810-030

Order No. 810-042(ø10 mm)(ø10 mm)
For shaft material (ø3 to 14 mm)

Diamond spot anvil
For sheet specimens*For Rockwell superficial
hardness testing

Small V anvil

Order No. 810-048

Console table

300700400

554

655Optional accessories

Options for data output * Cannot be used with HR-210.

Hardness reference blocks
Order No.
19BAA035
19BAA036
19BAA037
19BAA038
19BAA039
19BAA040
19BAA041
19BAA042
19BAA043
19BAA044
19BAA045
19BAA046
19BAA047
19BAA048
19BAA049
11AAD474
11AAD475
11AAD476
11AAD477
11AAD478
11AAD47911AAD480
11AAD481
11AAD482
11AAD483
11AAD484
11AAD485
11AAD486
11AAD487
11AAD488

ItemRemark

Hardness reference block 10HRC
Hardness reference block 20HRC
Hardness reference block 30HRC
Hardness reference block 40HRC
Hardness reference block 50HRC
Hardness reference block 60HRC
Hardness reference block 70HRC
Hardness reference block 41HR30N*
Hardness reference block 50HR30N*
Hardness reference block 60HR30N*
Hardness reference block 73HR30N*
Hardness reference block 83HR30N*
Hardness reference block 75HR15N*
Hardness reference block 85HR15N*
Hardness reference block 90HR15N*
Hardness reference block 32HRBW
Hardness reference block 42HRBW
Hardness reference block 52HRBW
Hardness reference block 62HRBW
Hardness reference block 72HRBW
Hardness reference block 82HRBW
Hardness reference block 90HRBW
Hardness reference block 32HR30TW*
Hardness reference block 42HR30TW*
Hardness reference block 52HR30TW*
Hardness reference block 62HR30TW*
Hardness reference block 72HR30TW*
Hardness reference block 78HR15TW*
Hardness reference block 82HR15TW*
Hardness reference block 87HR15TW*

Compliant with ISO/JIS standards
With an inspection certificate
from the standard block
manufacturer.

Order No.

Item

11AAE327
11AAE328
11AAE329
11AAE330
11AAE331
11AAE332
11AAE333
11AAE334
11AAE335
11AAE336
11AAE337
11AAE338
11AAE339
11AAE340
11AAE341
11AAE342
11AAE343
11AAE344
11AAE345
11AAE346
11AAE347
11AAE348
11AAE349
11AAE350
11AAE360
11AAE361
11AAE362

Hardness reference block 30HRC ASTM
Hardness reference block 45HRC ASTM
Hardness reference block 63HRC ASTM
Hardness reference block 30HRBW ASTM
Hardness reference block 70HRBW ASTM
Hardness reference block 90HRBW ASTM
Hardness reference block 65HRA ASTM
Hardness reference block 76HRA ASTM
Hardness reference block 85HRA ASTM
Hardness reference block 75HR15N ASTM*
Hardness reference block 85HR15N ASTM*
Hardness reference block 92HR15N ASTM*
Hardness reference block 50HR30N ASTM*
Hardness reference block 68HR30N ASTM*
Hardness reference block 83HR30N ASTM*
Hardness reference block 25HR45N ASTM*
Hardness reference block 43HR45N ASTM*
Hardness reference block 72HR45N ASTM*
Hardness reference block 67HR15TW ASTM*
Hardness reference block 83HR15TW ASTM*
Hardness reference block 91HR15TW ASTM*
Hardness reference block 36HR30TW ASTM*
Hardness reference block 63HR30TW ASTM*
Hardness reference block 76HR30TW ASTM*
Hardness reference block 75HREW ASTM
Hardness reference block 87HREW ASTM
Hardness reference block 100HREW ASTM

Remark

11AAD195 Hardness reference block 90HREW
19BAA028 Hardness reference block 32HRBS
19BAA029 Hardness reference block 42HRBS
19BAA030 Hardness reference block 52HRBS
19BAA031 Hardness reference block 62HRBS
19BAA032 Hardness reference block 72HRBS
19BAA033 Hardness reference block 82HRBS
19BAA034 Hardness reference block 90HRBS
19BAA050 Hardness reference block 32HR30TS*
19BAA051 Hardness reference block 42HR30TS*
19BAA052 Hardness reference block 52HR30TS*
19BAA053 Hardness reference block 62HR30TS*
19BAA054 Hardness reference block 72HR30TS*
19BAA055 Hardness reference block 78HR15TS*
19BAA056 Hardness reference block 82HR15TS*
19BAA057 Hardness reference block 87HR15TS** Reference block for superficial hardness testing

required for X-R control charts.

• Up to 1,000 data items can be saved in the machine using the data logger function. All data can be transferred in one operation by connecting theprinter to a computer using a USB cable (sold separately).

•The connection cable to the hardness testing machine is not supplied. A connection cable sold separately is required.
Compliant with ASTM/ISO
standards
With class B calibration certificate
and inspection certificate

CablesOrder No. 06AFZ050
USB cable (Type A - micro B)

HR-210MR

Measurement data wireless communication system

U-WAVE• Used to import data from hardness testing machines with Digimatic output function to a computer using wireless communication.
• Easy installation without cables that may get in the way thanks to wireless communication (up to 20 m).
• Data can be imported to common software with keyboard input (Excel, notepad, etc.) using the data interface function of the software supplied with U-WAVE-R.
• Test results from several hardness testing machines can be imported to a single computer using simultaneous transfers from multiple U-WAVE devices.

Order No. 02AZD810D
U-WAVE-R
With USB 2.0 cable (1 m)

Compliant with JIS standards
With an inspection certificate
from the standard block
manufacturer.

U-WAVE-T connection cable

Weight set

11AAD469
ø1 mm Brinell indenter

Item
Brinell weight set
62.5 125 187.5—HR-320MS

Brinell weight set
31.25 62.5 125 187.5

HR-430MR

Brinell weight set
62.5 125 187.5

HR-430MS

Brinell weight set
31.25 62.5 125 187.5

Order No. 937387
Order No. 965013
Connection cable (1 m) Type E Connection cable (2 m) Type E

Order No. 264-505
DP-1VA LOGGER

Order No. 02AZD790E
U-WAVE-T dedicated connection cable Type E

Optional accessories for Brinell testing (test conditions, weight set, indenter)
Hardnesstestingmachine

Digimatic Mini-Processor DP-1VA LOGGER• A compact printer for statistical calculation and printing data from hardness testing machines with Digimatic output function.
•High performance with support of measurement data printing, statistical calculation, histograms, and D chart creation, as well as complex calculations

To confirm operation with plastic tests.
With an inspection certificate fromthe standard block manufacturer.

11AAD194 Hardness reference block 90HRES

Compact printer with data logger function(HBW1/10*1)(HBW1/30*1)—(HBW1/10*1)(HBW1/30*1)
Order No.
Item

Size (Quantity)

19BAA281
Spare carbide ball
1 mm
ø1 mm (1 pc.)

Carbide ball indenter for Brinell
11AAD470
11AAD471
ø2.5 mm Brinell indenter ø5 mm Brinell indenter
HBW2.5/62.5
HBW5/62.5
HBW2.5/187.5
HBW5/125
HBW2.5/31.25
HBW5/62.5
HBW2.5/62.5
HBW5/125
HBW2.5/187.5
HBW2.5/62.5
HBW5/62.5
HBW2.5/187.5
HBW5/125
HBW2.5/31.25
HBW5/62.5
HBW2.5/62.5
HBW5/125
HBW2.5/187.5
Spare Carbide ball for Brinell
19BAA283
19BAA162
Spare carbide ball
Spare carbide ball
2.5 mm
5 mm
ø2.5 mm (1 pc.)
ø5 mm (1 pc.)

Order No. 02AZD880G
U-WAVE-T

11AAD472
ø10 mm Brinell indenter(HBW10/100*1)(HBW10/100*1)(HBW10/100*1)(HBW10/100*1)

19BAA163
Spare carbide ball
10 mm
ø10 mm (1 pc.)

Digimatic Gage/PC Data Input Device

USB Input Tool

• Interface used to export calculation results to spreadsheet software on a computer via a USB cable. Calculation results (values) can be exported in one
operation.
USB Direct Input Tool

USB keyboard signal converter type*

Order No. 06AFM380E
USB-ITN-E

Order No. 264-016-10
IT-016U* One of the cables below (sold separately) is required toconnect the testing machine to the IT-016U.

1 m: Order No. 937387
2 m: Order No. 965013*1: The built-in weights are used for this range. Only an indenter needs to be purchased.

67

106 2111(2)g-(TS)NE-(JP2), Printed in Japan

Coordinate Measuring Machines

Vision Measuring Systems

Form Measurement

Optical Measuring

Whatever your challenges are, Mitutoyo
supports you from start to finish.
Mitutoyo is not only a manufacturer of top quality
measuring products but one that also offers qualified
support for the lifetime of the equipment, backed up
by comprehensive services that ensure your staff canmake the very best use of the investment.
Sensor Systems

Test Equipment

Digital Scale and DRO Systems

Small Tool Instruments
and Data Management

Apart from the basics of calibration and repair,
Mitutoyo offers product and metrology training,as well as IT support for the sophisticated software
used in modern measuring technology. We can also
design, build, test and deliver measuring solutions
and even, if deemed cost-effective, take your criticalmeasurement challenges in-house on a sub-contract
basis.

Find additional product literatureand our product catalogue
https://www.mitutoyo.co.jp/global.html
Mitutoyo Corporation
Our products are classified as regulated items under Japanese Foreign Exchange and Foreign Trade Law.
Please consult us in advance if you wish to export our products to any other country.
If the purchased product is exported, even though it is not a regulated item (Catch-All controls item), the customer service
available for that product may be affected. If you have any questions, please consult your local Mitutoyo sales office.
Note: Product illustrations are without obligation. Product descriptions, in particular any and all technicalspecifications, are only binding when explicitly agreed upon.
MITUTOYO and MiCAT are either registered trademarks or trademarks of Mitutoyo Corp. in Japan and/or other countries/regions.
Other product, company and brand names mentioned herein are for identification purposes only and may be thetrademarks of their respective holders.

20-1, Sakado 1-Chome,
Takatsu-ku, Kawasaki-shi,
Kanagawa 213-8533, Japan
T +81 (0) 44 813-8230
F +81 (0) 44 813-8231
https://www.mitutoyo.co.jp