ENA

ACCURATE
AND POWERFUL
Passionately enabling chemists to create a better world since 1910.
From A to Z.

TEMPERATURE CONTROL PRODUCTS MADE BY IKA.

Temperature control products

Temperature control products

Heating and CoolingTemperature Control Instruments/// Highly precise and full of power
All from one source: With our temperature control products we offer a wide range for all temperature control
applications with the highest precision and with full power.
We promise that you will not only be impressed by theabove-average pressure and suction power of the pump.
From - 30 °C to + 250 °C: The temperature range of our temperature control products is meeting all challenges. And also yourbudgets: We offer affordable entry level devices as well as highend products for the most demanding requirements.
In addition to the above-average and industry-inspiring pressure
and suction power, our devices are intent on sustainability. For
example, our topseller RC 2 basic/control is equipped with acompressor, which only runs if cooling is necessary.
Another highlight, besides the outstanding compatibility to many
applications, is our Wireless Controller: It enables safe and remotecontrol. A safety factor, but at the same time a very convenient
way to control the IKA tempering systems in any position.

4

14/// COMPARISON OF ALL
TEMPERATURE CONTROL PRODUCTS/// HEATING

28

38/// HEATING & COOLING/// COOLING

50

54/// TECHNICAL DATA/// SCOPE OF DELIVERY

56

66/// ACCESSORIES/// PACKAGES

70

71/// SOFTWARE/// IKA SERVICE

3-YEAR WARRANTY** 2 years + 1 year after registering on www.ika.com/register, excludes wear parts

Find out more about IKA and order our fascinating products online: www.ika.com.

2

3

Temperature control products

Temperature control products

The right temperature
control product for every application/// Comparison of all temperature control products

Page

ICC basic | control

ICC eco Package*
basic | control

ICC pro Package**
basic | control

IC basic | control

IC pro Package**
basic | control

HBC 5 basic | control

HBC 10 basic | control

CBC 5 basic | control

HRC 2 basic | control

16 – 19

66 – 67

66 – 67

24 – 2569

20 – 23, 26 – 27

20 – 23, 26 – 27

30 – 33± 0.02 K | ± 0.01 K± 0.02 K | ± 0.01 K± 0.02 K | ± 0.01 K± 0.02 K | ± 0.01 K± 0.02 K | ± 0.01 K± 0.1 K | ± 0.05 K± 0.1 K | ± 0.05 K± 0.2 K | ± 0.1 K

2,500 W

2,500 W

2,500 W

2,500 W

2,500 W

1,500 W

–

–

–

–

–

350 W (at + 20 °C)

400 W (at + 20 °C)

400 W (at + 20 °C)

1,400 W (at + 20 °C)

0.61 bar0.61 bar0.61 bar0.61 bar

0.5 bar

0.3 bar0.61 bar0.45 bar0.45 bar0.45 bar

0.2 bar

0.2 bar0.45 bar

31 l/min

31 l/min31 l / min

21 l/min18 l/min

31 l/min

yes

yes

yes

yes

no | yesno | yesno | yesno | yesno | yesno | yesno | yesnono› Immersion circulator for
demanding internal andexternal applications.

› Heating bath circulator
for demanding internaland external applications.

› Powerful circulators for tempering external applications,e.g. for tempering double-walled lab reactors or
distillation equipment.

› P owerful refrigerated
circulators for external
use.

› R
 ecirculating chiller for mainly external uses.

› Can be placed in different
baths using removablebath bridge, e.g. formaterial testing in large,
open baths or for
external high-performance
tempering of analyticaldevices.

› IKA immersion racks canbe used for tempering
test tubes.

› With IKA accessories, the HBC series circulators arealso suitable for tempering large, external, open baths.

› Ideal for tempering
double-walled reaction
vessels, reaction systems
and autoclaves.

› Compact refrigerated
and heating circulator
for tempering external
applications, such asbioreactors.
› Usable in life science,
medical, chemical, cosmetics and food industry
labs (and many others).

› IKA accessories are also suitable for external, open baths.

Temperature stability± 0.02 K | ± 0.01 K± 0.02 K | ± 0.01 K± 0.02 K | ± 0.01 K

Heating power

2,000 W

2,000 W

2,000 W

Cooling power

–

–

–

–

Pump power pressure side

0.3 bar

0.3 bar

0.3 bar0.61 bar

Pump power suction side

0.2 bar

0.2 bar

0.2 bar0.45 bar0.45 barMax. flow rate

18 l/min18 l/min18 l/min31 l / min

31 l/minExternal temperingno | yesno | yesno | yes

yes

yes

Solenoid valve control

no

no

no

no | yes

Applications› Compact immersion
circulator primarily forinternal applications.

› H eating bath circulator for internal or
simple external applications.

+ 100 °C

+ 150 °C

+ 200 °C

- 30 °C- 0.4+ 250 °C////////////////////////- 0.3› Suitable for external tempering of double-walled
vessels (e.g. lab reactors)
with usable volumes
greater than 3 liters.

- 0.2

- 0.1

0

› For the determination of temperature-dependent
material constants, e.g. viscosity or thermal conductivity
in liquid-tempered test apparatuses.

+ 0.1

+ 0.2

+ 0.3

///////////+ 0.5

IC, HBC 5 | 10, CBC 5, RC 5
HRC 2
ICC, RC 2

ICC basic and control, ICC pro Package**

RELATIVE PRESSURE AND VOLUME FLOWICC eco Package*

///

› B
 road application
options due to wide
temperature range, e.g.
in semi-conductors,
packaging and plastics
industries.

USB› F or fast and efficient cooling of external devices such as
rotary evaporators, soxhlet apparatuses, calorimeters andincubating shakers.

› For tempering analyticaldevices such as viscometers, rheometers andpolarimeters.

RS 232PT 100

MULTI I/O-PORT+ 0.6 bar

IC pro Package**////////////////////////+ 0.4- 0.5 bar

IC basic and control, HBC 5 | 10 basic and control////////////////////////- 30 °C to RT- 25 °C to + 200 °C+ 20 °C to + 250 °C- 20 °C 0 °C + 20 °C

40 – 49

RT +10 °C to + 250 °C

RT + 10 °C to + 150 °C› W ith pump connection set as well as suitable fortempering small analytical devices or reaction systems.

40 – 49- 20 °C to RT |- 30 °C to RT

RT + 10 °C to + 250 °C

RT + 10 °C to + 100 °C› For tempering diverse
samples, e.g. foranalysis, material
and food testing.

34 – 37

RT + 10 °C to + 250 °C

RT + 10 °C to + 150 °C› Usable in different bathvessels.

RC 5 basic | control- 20 °C to + 100 °C |- 30 °C to + 100 °C

Working temperature range› F or tempering various samples, e.g. in test tubes
with precise-fitting IKA immersion racks.

RC 2 basic | control

IC control, HBC 5 | 10 control, CBC 5 control, HRC 2 control, RC 5 control

31 L/MIN
21 L/MIN
18 L/MIN

ICC control, IC basic, HBC 5 | 10 basic, CBC 5 basic, RC 2 control
ICC basic, HRC 2 basic, RC 2 | 5 basic

INTERFACES

CBC 5 basic and control°
HRC 2 basic and control
RC 2 basic/////////////////////////////////RC 2 control° | RC 5/////////////////////////////////TEMPERATURE RANGE
4

Pump connection set required for external applications. Find out more on our "Accessories" page.
* Plastic baths (eco packages) can be used at temperatures of up to + 100 °C (H2O only).
* Stainless steel baths (pro packages) can be used at temperatures of up to + 200 °C.
° At 2,000 rpm up to - 30 °C are possible.
/// = Operating temperature range (with external coolant). Advanced recirculating temperature (with external heating).

6Temperature control products

Temperature control products

SafetyPowerIntelligence/// All IKA tempering products meet the highest safety standards/// Superior power/// Intuitive operation and automatic tempering

All of the IKA temperature control devices that are equipped with aheating function meet the highest safety classification III (FL) for use
with combustible fluids pursuant to DIN 12876.

Whether infinitely adjustable pressure/suction pump or premium
grade stainless steel: Rely on a strong partner.

Smart technology as well as user-friendly menu navigation simplify
temperature control for any application.

SAFE HANDLING DUE TO ERGONOMIC AND WELL THOUGHT-OUT DESIGNS

Carrying handle
Safe carrying and positioning (ICC).

Recessed handles
Ergonomic handling (HBC and RC 2).

BracketSecures the base and protects thefloats and tubular heater (ICC).

Transport handle
Easy and safe handling (HBC).

Safe and complete drainage of baths
The bath can be fully emptied of thermal
fluids, in a simple and clean manner. The
physical separation of the drain valve andthe opening screw ensures that the user
does not come into contact with the fluid.

›› Thermofluid temperature threshold
The thermal fluid that is being used canbe selected in the menu. This ensuresthat the temperature remains outside thecritical values for that particular fluid.
Minimum and maximum temperature
values can be adjusted manually withinthese parameters.
›› Speed
The speed can be restricted; which allows
for determining the maximum pumppressure.
7›› For decades, temperature
control has been one of IKA's
core competencies
IKA heating circulators reach a temperature constancy of up to ± 0.01 K.
The power-regulated compressor of therecirculating chiller allows a constanttemperature of up to 0.05 K.
The large heating surfaces gently control
the temperature of the thermal fluids
and ensure outstanding heat transfer.
The strong heat output of the circulators
ensures short heat-up times.

Casters
Facilitate easy positioning of the device(RC, HBC, CBC).

SAFE OPERATION DUE TO
ADJUSTABLE THRESHOLD VALUES›› Safety temperature
The safety temperature can be adjusted
using tools or through the display. The
temperature is monitored by an independent temperature sensor.

›› Tempering
The IKA heating and refrigerating circulators temper fluids ranging from - 30 °C
to + 250 °C with a heating and coolingpower of up to 2.5 kW and 1.4 kW
respectively.

VISUAL AND AUDIBLE ALARM
Indication of critical fluid level, criticaltemperature or blocked pump.

LOCK FUNCTION
Locks set parameters to prevent accidental
adjustments on the WiCo.

FLUID LEVEL DETECTION
Critical minimum or maximum levels are recognized both mechanically (by the floater)
as well as electronically (by a temperature
sensor).

All IKA heating circulators have the optionof adding a cooling coil with large coolingsurface for operating in or below ambient
temperature. By connecting a chiller, it ispossible to reach an operating temperature down to - 20 °C.
›› Pressure/suction pump
All IKA circulators use a powerful, infinitely adjustable pressure/suction pump madeof high-quality PEEK plastic. It allows
for flexible use of the devices in open or
closed system applications. The pump ensures homogenous mixing inside the bathand provides a high flow rate for external
applications.

›› Pump curve
The actual maximum flow rate can be
determined under known pressure loss ofthe experiment based on a pump curve.
Pressure losses occur, for instance, fromheight differences, narrow and long hoses
as well as the high viscosity of a bathfluid. The maximum viscosity of the bathfluid is 50 mPas.
›› Energy efficiency
The excellent insulation and the demanddriven output control ensure that IKA
temperature control instruments arehighly energy-efficient.
As such, IKA recirculating chillers use up
to 60 percent less energy than comparable competitors' devices in standard
applications.
›› Robust and durable
IKA temperature control products are
made of high-quality materials anddesigned for a long service life.
Parts that come into contact with products are made of premium grade stainlesssteel (V4A) and highly durable PEEK, FKM
and PTFE; meeting the basic requirements
for use in the food industry.

›› Connectivity
USB and RS 232 are standard interfaces.
Software programs are used to gather themeasured data and control the devices,e.g. labworldsoft® by IKA. Following
registration, the Firmware Update Tool
ensures that users always have the latest
version of the software.

›› Operating modes
There are four options after a restart or
power failure – simply select the modeof operation.

Most IKA circulators have a PT 100
interface for the display and controlof the external temperature (except for
the ICC basic, RC 2 | 5 basic and HRC 2
basic ).

Operating mode B:
After switching on/power interruption,
automatic start of functions dependingon the previous settings. Ideal whenworking via switchable sockets.

›› Calibration and adjustment
The internal (and, if available external)
temperature sensor can be adjusted
either via a two- or three-point calibration
process.

Operating mode C:
Set values (set in A or B) cannot be
modified. After switching on/powerinterruption, automatic start of functions
depending on the previous settings.

›› Automatic tempering
Before starting the tempering process,the control parameters of the bath fluid
as well as the amount of the bath fluid
are measured in order to prevent thetemperature from being exceeded. There
are also options to manually set the PID
control parameters.

Operating mode D:
Confirmation query for set value changes;
if functions are active. After switching on/
power interruption, no automatic restartof functions.

›› Software control/specificationof heating rates
Labworldsoft® software can be used toprecisely specify temperature ramps andheat-up times/heating rates.

Operating mode A:
After switching on/power interruption,no automatic restart of functions.

›› Intuitive operation
User-friendly menu navigation,
push buttons and knobs enablesimple operation.

All circulators are equipped or can be
retrofitted using pump connections(M 16 × 1).
Pressure/suction pump9

Temperature control products

Temperature control products

IKA control models/// The added value of the control models
DISCOVER THE BENEFITSOF THE CONTROL MODELSSAFETY

POWER›› Monitoring the temperature
Additionally adjustable range forexternal temperature. Checking the
gap between internal and external
temperature (adjustable).

›› Increased precision
IKA control devices reach a temperature
constancy of up to ± 0.01 K. The powerregulated compressor of the circulating
chiller allows for a constant temperatureof up to 0.05 K.

›› Monitoring the pump pressure
Easy adjustment/selection of max.
pressure.
›› Wireless Controller (WiCo)(excl. ICC control):
Safe remote control of the devices, e.g.
inside the fume hood. Wireless bluetooth
control of the devices with a range of up
to 10 meters (30 ft.) or via USB cable.
›› Audible signals e.g. reaching
set temperature›› Alarm-controlled shut-off(via multi-IO port)›› Output performance can be reduced
The option to reduce the heat performance down to 50 % of the rated output
for moderate heating times in order toadjust to previous systems or as overload
protection.
›› Control of a cooling coil(via multi-IO port)
Using a cooling coil as an extension to
the heating circulators for quick cooling,
tempering at room temperature or toabsorb moderate temperature increases(e.g. via exothermal reaction).
With a controlled solenoid valve, the tap
or cooling water is purposefully regulated,thus reducing the water consumption toa minimum.

›› Stand-by operation(via multi-IO port)
By connecting current to switch input,
the device can be turned off.

Option to connect externalsolenoid valves via multi I/O port
(not ICC/RC 2 control)

HBC control
Heating bath circulator

Controlling solenoid valves- automatic refills- turning the coolingwater circuit on/off- monitoring the fluid level- electronic shut-off valve› Output for alarm signals›	
Input for standby mode
(for turning off the device)
›

10

11

Temperature control products

Temperature control products

INTELLIGENCE›› Clear and user-friendly display
All important process parameters are clearly arranged and easy to read. Actual andset temperature, fill level and safety temperature readout (among other parameters).
Quick access to all relevant control parameters.
›› Programming function
10 individual programs with 10 steps each can be operated via time, set temperature
and/or a temperature gradient. Additional features such as integrating solenoid valves
are possible.
›› Measuring graph
The main screen can display either the process parameters (standard) or a temperature/
time graph. The user can switch between these options using a quick-access key.
›› Timer/counter function›› Switch from external to internal temperature control at the press of a button
All control units access one PT 100 interface for an external temperature sensor.
Thus, the internal or external temperature can be regulated at any time.

External probe connected/
not connected

Operating mode

Lock key

USB port enabled
WiCo is docked on the station
Battery status display
and charging status

Bluetooth enabled

Heating on/off

Actual internal
temperature

Fluid level display
Connection made to the PC

Pump on/off
Actual external
temperature

Safety temperature

Set temperature

Run time display
HH:MM:SS›› Degassing function
For reducing air pockets in oils.

HRC 2 control
Refrigerated andheating circulators

Programming function

12

Measuring graph13

Temperature control products

Temperature control products

Heating/// Optimal heat exchange within atemperature range of RT + 10 °C to + 250 °C.
Regardless of our circulators ICC and IC or our heating bath circulators
HBC 5 & 10 – IKA temperature control products are reliable, safe andwork under full pressure of the powerful pressure and suction pump.
Large heating surfaces provide for optimal heat exchange of temperatures up to + 250 °C.

+ 100 °C

+ 150 °C

HBC 5 | 10 basic and control
Heating bath circulators+ 200 °C

RT + 10 °C+ 250 °C

The colorful TFT graphic
display of the HBC 10
control displays all relevant
process data.

IC basic and control, HBC 5 | 10 basic and control
IC pro Package
ICC basic and control, ICC pro Package
ICCICCecoecoPackage套装

ICC basic and control
Compact immersion circulators

14

IC basic and control
Immersion circulators

15

Temperature control products

Temperature control products

ICC basic and control/// Compact immersion circulators
The compact immersion circulator ICC basic and the ICC control are designed for tempering fluids up to + 150 °C and thus provide an economic and attractive solution forstandard applications, such as the tempering of samples. The ergonomic handle andthe compact design allow for safe transport as well as convenient use. The integrated
base provides a secure stand while protecting floaters and tubular heating elements. A
bracket for mounting the bath vessel is included in the scope of delivery. The compact
ICC immersion circulators allow for simple and flexible exchange of different baths.

Convenient carrying handle

The ICC circulators from IKA are ready to connect to a pump connection set (PCS.ICC)
for external tempering as well as to a cooling coil (CC2) for operation at and below
ambient temperature.

ICC control – graphic display showing
parameters such as temperature, pumpspeed, etc.

USB/RS 232 interfaces for connecting
a PC, using labworldsoft and enabling
online updates of device software.

ICC control – integrated
PT 100 temperature probe interface.

Integrated base protects tubular
heating elements and floaters.
Integrated base protects
tubular heating elements
and floaters.

16

ICC control
Compact immersion circulator

17Temperature control products

Temperature control products

APPLICATION EXAMPLEAs shown in the application, the ICC circulators from IKA can be mounted with a bathbridge or connected to various sized baths
using the mounting bracket included in thescope of delivery.
An additional open bath can be temperedby connecting the pump connection set.
The ICC is connected to the external plastic
bath via a level controller.
These features show that, the ICC circulators are very flexible and space-saving.
One example is to use the test tube inserts,
making the ICCs suitable for temperinglarge number of samples.

ICC with bracket (includedin scope of delivery)
and pump connection set
PCS.ICC (accessory).

0.3

Pressure / suction [bar]0.2
0.1
1,000 rpm
02,000 rpm

3,200 rpm

- 0.1
- 0.2
- 0.3

Pump characteristic curves: ICC basic & control
Measurement in accordance with DIN 12876-2 withwater at + 20 °C, closed pump circuit

0

5

10

15

20

Flow rate [l / min]100

TECHNICAL DATA/// ICC basic, Ident. No. 0004134400 |ICC control, Ident. No. 0004136600

Temperature [°C]

80
60
40

7 l water
10 l water20

14 l water
18 l water

0

0

20

40

Heating time curves: ICC basic & control
Heating time varies based on bath sizes.
The information is based on water volume inan open bath at room temperature.

Heat output (230 V)

2,000 W

Working temperature range

RT + 10 °C to + 150 °C

Max. flow rate (at 0 bar)

18 l/mPump power (pressure)

0.3 bar

Pump power (suction)

0.2 bar60

Time [ min]18

19

Temperature control products

Temperature control products

IC and HBC/// Powerful heating circulators
The IKA IC and HBC circulators are following a modular design principle. The IC
head is its core. When combined with a high-quality insulated bath, the IC turns
into an HBC heating bath circulator. The high performance pump reaches a highflow rate and, together with the large heating body surface, provides for optimalheat exchange between the application and circulator.

IC basic and control

NEW: BASIC DEVICESWITH ADDED POWER.

The IC and HBC basic circulators were optimized
and come with improved performance data:
Tempering up to + 250 °C with the same pump
power and control units.

The IC immersion circulators
are designed for temperingliquids up to + 250 °C.

HBC 5 basic and control

The excellent insulation of theheating bath circulators HBC
offers short heat-up times.

IC basic
Immersion circulator

20

21Temperature control products

Temperature control products

9080

Temperature [°C]

70
60
50
40
7 l water

30

10 l water20

14 l water

10BASIC AND CONTROLINTERFACES

0

18 l water
0

10

20

30

40

Heating curves: IC basic & control
The heating curves of the IC basic/control show theheating time dependent on different bath sizes.
The information is based on water volume in an
open bath at room temperature.

50

Time [ min]90
80

Temperature [°C]

70
60
50
40
30

HBC 5:
5.5 l water20

RS 232/USB
interfaces

100

HBC 10:10.0 l water
0

10

0.7

Multi-IO port for externalsolenoid valves (control
only)

0.5

Pressure / suction [bar]Cooling coil

30

Time [ min]

PT 100 external
temperature probe

Pump input/output

20

Heating curves: HBC 5 | 10 basic & control
The heating curves show the heating times of the HBC 5 with
5.5 l water volume and the HBC 10 with 10 l water volume.

0.3
3,200 rpm

4,600 rpm

2,000 rpm

0.1
0
- 0.1

- 0.3
- 0.5

Pump curves: IC | HBC basic & control
Measurement in accordance with DIN 12876-2
with water at + 20 °C, closed pump circuit.

0

5

10

15

20

25

30

35

Flow rate [l / min]TECHNICAL DATA/// IC basic, Ident. No. 0003861000 | IC control, Ident. No. 0003863000
HBC 5 basic, Ident. No. 0004125000 | HBC 5 control, Ident. No. 0004127000

22

Heat output (230 V)

2,500 W

Max. flow rate (at 0 bar)

31 l/mWorking temperature range

RT + 10 °C to + 250 °C

Pump power (pressure)0.61 bar

Pump power (suction)

0.45 bar23

Temperature control products

Temperature control products

IC basic and control/// Universal immersion circulators
The IC immersion circulators are designed for tempering liquids up to + 250 °C.
Due to the flexible bath bridge, the device can be mounted on various size baths.
The control version features a removable WiCo (wireless controller), which allows
for working in a fume hood, for example. The advanced features enable the device
to be used in demanding internal and external applications, such as analysis andmaterial testing.

APPLICATION EXAMPLE A
The IKA IC bridge circulator can be used
for internal and external tempering
applications.

The example shows the IC control in an
IB pro 20 stainless steel bath, which cantemper samples that are attached to aheight adjustable base of up to + 200 °C.

APPLICATION EXAMPLE B
The IKA IC immersion circulators are
exceptionally well suited for external
applications. The setup shows the IC control
with stainless steel bath, bath bridge andcover (IC control pro 20 c package) under
the hood; all connected to a metal doublewalled reactor. With the large stainless steelbath, the usable volume is approx. 5.5 l.

IC basic
Universal immersion circulators

By connecting the PT 100 sensor, thetemperature of the medium in the reactor
can be measured and controlled.
Using the detachable WiCo, the immersion
circulator IC control can be safely operated
from up to 10 meters (30 ft.) away.

IC control
Universal immersion circulators

SAFETY ANDCONVENIENCE FEATURES
USB/RS232 interfaces for connecting aPC, using labworldsoft and enabling online
updates of device software.

Due to the flexible bath bridge, the IC
immersion circulator can be mounted onbaths of various sizes (285 – 400 mm).

Integrated pressure/suction pump forinternal and external temperature control.

24

Connector for
PT 100 temperature probe.

IC control – connection option of externalsolenoid valves over multi-IO port.

Detachable WiCo (wireless controller)
for simple and safe remote access from up
to 10 m (30 ft.).

›› Adjustable safety circuit for temperature›› Mechanical and electronic
fluid level detection›› Visual and audible alarm›› Switch from external to internal
temperature control at the pushof a button (control model)›› Universal use for internal andexternal applications›› Cooling coil included in the scopeof delivery (control model)

25Temperature control products

Temperature control products

HBC 5 | 10 basic and control/// Heated bath circulators for external tempering
applications
The well-insulated stainless steel heating bath and powerful PEEK pressure and suction
pump are two of the key features of HBC heated bath circulators. The maximum
temperature of the HBC heating bath circulator is + 250 °C. The large surface of thetubular heating elements provides for optimal heat exchange. The bath fluid is heatedgently and promptly.
Due to its high temperature consistency of up to ± 0.01 K, short heat-up times andadvanced features of the high-tech TFT display with WiCo (wireless controller), the
HBC control heating bath circulator is the ideal solution for demanding and complex
tempering processes.

Visual and audible alarm.

USB/RS 232 for connecting a PC, using
labworldsoft and enabling online updates ofdevice software.

Integrated pressure/suction pump forinternal and external temperature control.

HBC 5 basic and control

The HBC 5 has a bathcapacity of 5.5 up to
7.5 liters. This adds up toa usable volume of 2 liters.

SAFETY ANDCONVENIENCE FEATURES›› Ergonomic design›› Excellent insulation for short heat-up
times and improved heat transfer›› Safety drain valve›› Adjustable safety circuit›› Switch from external to internal
temperature control at the pushof a button (control model)
HBC 10 basic and control

Integrated transport handle on theback of the device, recessed handles forergonomic transport.

26

BIG EXPANSION VOLUMEThe HBC 10 has a bathcapacity of 7.5 up to
10.5 liters. This adds up toa usable volume of 3 liters.

HBC control – connection option of externalsolenoid valves over multi-IO port.

APPLICATION EXAMPLE

Detachable WiCo (wireless controller)
for simple and safe remote access from up
to 10 m (30 ft.).

The HBC heating bath circulator is ideal
for external applications, for example
the heating of double-walled laboratory
reactors, such as the LR-2.ST from IKA.

Bath opening 160 × 90 mm

27Temperature control products

Temperature control products

Heating and cooling/// Heating and cooling in one
IKA has expanded its circulator portfolio by four new devices, adding the combined
refrigerated and heating circulators CBC 5 and HRC 2 devices (each in basic andcontrol variants). The temperature range is from - 30 °C to + 200 °C.

CBC 5 basic
Refrigerated andheating circulator

CBC 5 control
Refrigerated andheating circulator

NEW!

HRC 2 basic
Refrigerated andheating circulator

HRC 2 control
Refrigerated andheating circulator- 20 °C 0 °C

- 30 °C+ 200 °C///

CBC 5 basic and control
HRC 2 basic
HRC 2 control+ 100 °C

28

29

Temperature control products

Temperature control products

CBC 5 basic and control/// Powerful refrigerated and heating circulators
The CBC 5 is a powerful refrigerated and heating circulator with 2,500 W heating
power and 350 W cooling power. It is best suited for external applications as it
uses the proven pressure and suction pump of the HBC and IC series. High-quality
insulation of the device allows for fast heat-up times and reduces heat input at low
temperatures.
The basic as well as the control models can temper the medium by using an external
temperature sensor. All parameters can be read out, monitored and completely documented by software (e.g., labworldsoft® or NAMUR commands), using theRS 232 or USB interfaces.

CBC 5 control
Refrigerated andheating circulator

SAFETY ANDCONVENIENCE FEATURES›› Safety and convenience features›› Adjustable safety circuit›› Filling level detection›› Visual and audible alarm›› RS 232 and USB interfaces›› Multi IO-port (control version only)›› Switching from external to internal
temperature control at the push ofa button (control model)›› Excellent insulation›› Safety drain valve

30

31Temperature control products

Temperature control products

0.7

APPLICATION EXAMPLE

0.5

Pressure / suction [bar]A typical field of application of the
combined heating and cooling circulators istempering process systems on laboratory or
pilot scales. In our example, the IKA magic
PLANT is combined with the inline disperser,
IKA magic LAB. First, the product is being
heated by the CBC 5 using the doublewalled IKA magic PLANT, then held at theset temperature in the subsequent disperser
process with cooling function. The productis then cooled to room temperature. The
external temperature sensor directly impacts
the temperature of the final product.

0.3
3,200 rpm

4,600 rpm

2,000 rpm

0.1
0
- 0.1

- 0.3
- 0.5

Pump curve: CBC 5 control & basic
Measurement in accordance with DIN 12876-2
with water at + 20 °C, closed pump circuit.

0

5

10

15

20

25

30

35

Flow rate [l / min]TECHNICAL DATA/// CBC 5 basic, Ident. No. 0004165000 | CBC 5 control, Ident. No. 0004167000
Heat output (230 V)

2,500 W

Max. flow rate (at 0 bar)

31 l/m- 25 °C to + 200 °C(- 30 °C possible
at 2,000 rpm)

Pump power (pressure)0.61 bar

Working temperature range

Pump power (suction)

0.45 bar

COOLING POWER/// CBC 5 basic, Ident. No. 0004165000 | CBC 5 control, Ident. No. 0004167000

32

Temperature

Cooling power at max. speed

Cooling power at 3,200 rpm+ 20 °C

350 W

400 W

+ 10 °C

320 W

370 W

0 °C

270 W

320 W

- 10 °C

190 W

240 W

- 20 °C

80 W

130 W

USB/RS 232 interfaces for connecting aPC, using labworldsoft and enabling online
updates of device software.

CBC control – detachable WiCo(wireless controller) for simple and safe
remote access from up to 10 m (30 ft.).

Silent mode –the fan only runs as needed.

CBC control – connection option of externalsolenoid valves via multi-IO port.

Connector for external
PT 100 temperature probe.

CBC control – control accuracy. The
speed-regulated compressor provides atemperature stability of up to ± 0.01 K.

33

Temperature control products

Temperature control products

HRC 2 basic and control/// Compact refrigerated and heating circulators

NEW!

Launching the HRC 2, IKA is offering a compact refrigerated and heating circulator
with 400 W cooling power and 1,500 W heating power. It is best suited for tempering
small, external applications. Applications with larger pressure losses, e.g. lab reactors
or viscometers can be supplied with appropriate flow volume of bath fluid throughthe more powerful pressure and suction pump (0.5 bar). With a level controller, it ispossible to temper samples in open baths.
The on-demand cooling unit runs quietly and with utmost efficiency. Maximum cooling
performance is achieved if the easy-to-reach air filter is replaced routinely.

USB/RS 232 interfaces for connecting aPC, using labworldsoft and enabling online
updates of device software.

Silent mode –the fan only runs as needed.

Integrated transport handle on theback of the device, recessed handles forergonomic transport.

SAFETY ANDCONVENIENCE FEATURES›› Adjustable safety circuit›› Filling level detection›› Visual and audible alarm›› RS 232 and USB interfaces›› Multi IO-port (control version only)

HRC 2 control – connector for external
PT 100 temperature probe.

HRC 2 control – connection option of external solenoid valves via multi-IO port.

HRC 2 control – detachable WiCo(wireless controller) for simple and safe
remote access from up to 10 m (30 ft.).

34

35

Temperature control products

Temperature control products

500
400

Pressure / suction [mbar]

300
4,000 rpm
200
2,000 rpm

APPLICATION EXAMPLE

3,200 rpm

100

The example shows a double-walled 10 l glass reactor for growing algae.

0
- 100
- 200
- 300

Pump curve: HRC 2 control & basic
Measurement in accordance with DIN 12876-2
with water at + 20 °C, closed pump circuit.

0

5

10

15

The HRC 2 is well suited for achieving constant temperature conditions inside the reactor in order toachieve a high/controlled growth rate of the algae formation. Applications requiring critical temperature control, the temperature can be directly controlled in the medium by using the HRC 2 control with an
external temperature measurement sensor.

20

Flow rate [l / min]

COOLING POWER/// HRC 2 basic, Ident. No. 0025003742 |HRC 2 control, Ident. No. 0025004524

TECHNICAL DATA/// HRC 2 basic, Ident. No. 0025003742 |HRC 2 control, Ident. No. 0025004524

Heat output (230 V)

1,500 W

Working temperature range- 20 °C to +100 °C |- 30 °C* to +100 °C

Max. flow rate (at 0 bar)

21 l/mPump power (pressure)

0.5 bar

Pump power (suction)

0.2 barTemperature

Cooling power HRC 2+ 20 °C

400 W+ 10 °C

370 W

0 °C

320 W

- 10 °C

240 W- 20 °C

130 W* - 30 °C possible at 2,000 rpm.

3637

Temperature control products

Temperature control products

Cooling/// Compact cooling power of 1,400 watts- 20 °C
IKA is expanding its product offering of highly efficient recirculating chillers by two
new devices. With the RC 5 basic and control, there are now two devices available
with a cooling power of 1,400 W, complementing the line by extremely compact
and powerful chillers.

- 30 °C+ 20 °C
RC 2 basic
RC 2 control | RC 5 basic and control

Think smart – not only save valuable drinking water, but also energy and lower youroperating costs.

RC 2 basic
Recirculating chiller

38

RC 2 control
Recirculating chiller

RC 5 basic
Recirculating chiller

RC 5 control
Recirculating chiller

39Temperature control products

Temperature control products

RC 2 | RC 5 basic and control/// Energy-efficient recirculating chillers
Due to the high energy efficiency of our R134a-based refrigeration machine, asignificantly lower amount of greenhouse gases is emitted over the life span of the
product compared to unregulated chillers using natural coolants.

IKA RECIRCULATING CHILLER –ENVIRONMENTALLY FRIENDLY

Even in the worst-case scenario, IKA devices have the advantage of saving
500 g/kWh CO2 in the current energy mix after just 1.5 years, compared tounregulated devices with natural coolants. (Assumptions: coolant has dissipatedcompletely at the end of the product life cycle; GWP for R134a = 1,300,
according to IPCC AR5 100 years, 250 work days/year; 2 kWh savings/day).
With the launch of the RC 5, IKA is offering, for the first time, a recirculating chiller
with natural coolant R 290. Adding the innovative on-demand control, IKA iscontributing to a green future.

Simple handling of cooling
fluids due to the large opening
and the integrated funnel

DURING THE DEVELOPMENT OF THE RC RECIRCULATING CHILLERS, IKA ENGINEERSSTRONGLY FOCUSED ON ENERGY EFFICIENCY AND DEVELOPING UNIQUE SOLUTIONS.
›› The heart of the RC 2 and RC 5 is a speed-controlled compressor, which is used to respond to the actual
cooling need. The energy consumption can be reduced significantly as well as the compressor service life
increased.

Safety drain at the front›› The high-quality foam insulation, surrounding the storage tank, minimizes the energy input and keeps
the bath fluid cool.
›› The air-cooled microchannel condenser ensures optimal heat dissipation. The air flow, required for the
micro-channel condenser, is generated by a speed-controlled fan. It reduces the noise level and lowers
energy consumption.

Easy-to-clean air filter›› The electronically controlled expansion valve contributes to achieving an excellent temperature stabilityof up to ± 0.05 K.

40

41

Temperature control products

Temperature control products

Economic features/// Save money

SAVE WATER›› Calculated at an assumed average of six operating hours a day on 200 business days a year, a rotary
evaporator (50 l/h) cooled with tap water consumes 60,000 liters of water per year. That amount canbe saved by using a recirculating chiller, not only protecting the environment, but also reducing
operating costs.

SAVE ENERGY›› IKA has succeeded in achieving energy savings of up to 60 % in comparison to competitors' devices,
simply by the afore-mentioned innovations of the recirculating chiller, particularly the speed-controlled
compressor.

400
Power input [W]350

POWER CONSUMPTION – A COMPETITIVE ANALYSIS›› Example: Energy consumption of a chiller in a standard distillation of a rotary evaporator (500 ml water inan 1 liter evaporating flask). Water bath temperature + 60 °C, pressure 70 mbar.
When comparing the average power consumption of the chiller in neutral/idle mode (rotary evaporator isturned off), the RC 2 from IKA clearly runs more efficiently in comparison to the competition (Figure 1).
When considering energy consumption over an entire day, the efficiency has an even greater impact. The
on-demand circulation chiller from IKA reduces its power to a minimum, adapting to the actually required
cooling performance. Figure 2 shows that the IKA RC 2 in comparison to competitor 1 consumes less than
one-third of the energy.

Competitor 2(600 W cooling power)

Competitor 2(600 W cooling power)

Competitor 1(350 W cooling power)

Competitor 1(350 W cooling power)

IKA RC 2(400W cooling power)

IKA RC 2(400W cooling power)

0100

Zero load power (W)Figure 1

42

200

300

400

0

2

300
250
200
150
100

Distillation 1

50

Distillation 2

Distillation 3

Distillation 4

0
0

1

2

3

4

5

6

7

8

Time [h]Power input (W) and energy consumption (kWh) over 8 h with 4 distillation runs Picture 3
Total energy consumptioncompetitor 1: 1.85 kWh

Total energy consumption
IKA RC 2: 0.81 kWh›› Figure 3 shows the mechanism of our on-demand recirculating chiller (red) in comparison to an unregulated chiller (blue). Depicted is an 8-hour work day, with four 500 ml (H2O) distillation runs. It's easy torecognize the overall lower power demand when in idle mode, as well as the adjustment of the cooling
performance during the distillation process.

4

Daily energy consumption
Figure 243Temperature control products

Temperature control products

RC 2 | RC 5 basic and control/// Energy-efficient recirculating chillers
The RC cooling circulators are designed for fast and efficient cooling of external
devices. The chillers distinguish themselves by short cooling times with a temperaturestability of up to ± 0.05 K. The operating temperature range is from room temperature
down to - 30 °C. The maximum recirculation temperature is + 80 °C. You can adjust
the pump performance as needed through the speed-controlled pressure and suction
pump; hence exceeding the cooling power according to DIN.

NEW!
THE IKA RECIRCULATING CHILLER HAS A BROADEROPERATING TEMPERATURE RANGE OF UP TO + 80 °C
THE RC 2 CONTROL COOLS DOWN TO - 30 °C(50 W COOLING POWER AT 3,200 RPM)

The control devices are equipped with a connector for an external temperature sensor.
Using the PT 100 temperature probe (within scope of delivery), you can control thetemperature directly in your target medium.
The device can be operated conveniently via the RC 2 | 5 control WiCo (wirelesscontroller). It makes the chiller space-saving and allowing for it to be placed in evenhard-to-reach areas of the lab.

Energy efficiency – up to 60 % lower
energy consumption during standard operation (compared to devices of competitors).

Control accuracy – the speed-regulatedcompressor provides a ten-times higher
temperature stability of up to ± 0.05 K.

Silent mode –the fan only runs as needed.

Handling – safe and ergonomic handlingdue to a well thought-out design. Casters
on the back of the device enable easytransport and set-up.

44

RC 2 basic and control
Recirculating chillers

Large operating volume – the large
difference between the maximum andminimum volume can be used as theoperating volume for external tempering.

RC 2 | 5 control – connection for PT 100
temperature probe.

RC 2 | 5 control – detachable WiCo(wireless controller) for simple and safe
remote access from up to 10 m (30 ft.).
RC 5 basic and control
Recirculating chillers

45Temperature control products

Temperature control products

IKA RECIRCULATING CHILLERS CAN ALSO
COOL OPEN BATHS WITHOUT PROBLEMSDOWN TO LOW TEMPERATURES DUE TO
A POWERFUL PRESSURE/SUCTION PUMP.
THE EXAMPLE SHOWS A BATH THAT HAS
BEEN COOLED TO - 17 °C WITH THE RC 2.

0.3

Pressure / suction [bar]0.2
0.1
1,000 rpm
02,000 rpm

3,200 rpm

- 0.1
- 0.2
- 0.3

Pump curve: RC 2 basic & control
Measurement in accordance with DIN 12876-2 withwater at + 20 °C, closed pump circuit

0

5

10

15

20

Flow rate [l / min]TECHNICAL DATA/// RC 2 basic, Ident. No. 0004171000 | RC 2 control, Ident. No. 0004173000
RC 5 basic, Ident. No. 0004181000 | RC 5 control, Ident. No. 0004183000
RC 5 basic | control

0.7

Cooling power (at + 20 °C)

400 W

1,400 W

0.5

Working temperature range- 20 °C to RT | - 30 °C to RT- 30 °C to RT

Max. flow rate (at 0 bar)

18 l/min

31 l/minPump power (pressure)

0.3 bar0.61 bar

Pump power (suction)

0.2 bar0.45 bar

COOLING POWER/// RC 2 basic, Ident. No. 0004171000 | RC 2 control, Ident. No. 0004173000
RC 5 basic, Ident. No. 0004181000 | RC 5 control, Ident. No. 0004183000

46

Temperature

RC 2 basic | control

RC 5 basic | control+ 40 °C–

1,700 W

+ 20 °C

400 W

1,400 W

+ 10 °C

370 W

1,100 W

0 °C

320 W

950 W

- 10 °C

240 W

600 W

- 20 °C

130 W

350 W

- 30 °C–

200 W

Pressure / suction [bar]RC 2 basic | control

0.3
3,200 rpm

4,600 rpm

2,000 rpm

0.1
0
- 0.1

- 0.3
- 0.5

Pump curve: RC 5 basic & control
Measurement in accordance with DIN 12876-2 withwater at + 20 °C, closed pump circuit

0

5

10

15

20

25

30

35

Flow rate [l / min]47

Temperature control products

Temperature control products

RC 2 | RC 5 basic and control/// Application information
APPLICATION EXAMPLE A
EFFICIENT USE OF THE RECIRCULATING CHILLER FOR EVAPORATION AND EXTRACTION

Using rotary evaporators for distillation or e.g. conducting Soxhlet extractions, therequired cooling power is contingent upon the flask size and the number of extraction
units.

The IKA RC 5 is ideal for the cooling of multiple rotary evaporators, e.g. the IKA rotary evaporators
RV 10 and RV 8. When using a 1-liter evaporation flask (depending on the solvent to be distilled), it ispossible to cool up to four rotary evaporators at the same time. This results in essential space, energy
and cost savings.

The following table shows how many rotary evaporators and extraction units can becooled with an IKA recirculating chiller. It is also a guide to finding the ideal recirculating chiller for your specific application.

COOLING POWER AT 20 °C COOLINGTEMPERATURE DEPENDING ON THE FLASK SIZE*
600

600

3,000 ml

400
Cooling power demand [W]Cooling power demand [W]

500

300

200

100

500

APPLICATION EXAMPLE B

2,000 ml

The RC 2 recirculating chillers are suitable for cooling external analytical equipment such as laboratory
reactors, calorimeters, incubating shakers or rotary evaporators. The set up below shows the RC 2 basic
recirculating chiller connected to the IKA C 1 calorimeter.

400
1,000 ml
300
500 ml200
100

100 ml
0

50

100

250

500

1,000

2,000

3,000

0
0

Flask size [ml]

200

400

600

800

1,000

Flask surface [mm2]* Water was used as reference solvent. When using other solvents, the cooling power demand could be lower.

RECIRCULATING CHILLERS/// RC 2 basic, Ident. No. 0004171000 | RC 2 control, Ident. No. 0004173000
RC 5 basic, Ident. No. 0004181000 | RC 5 control, Ident. No. 0004183000

48

RC 2 basic | control

RC 5 basic | control

Flask size

50 – 100 ml | 250 – 500 ml | 1000 ml

1000 ml | 2000 ml | 3000 ml | 5000 ml

Number of rotary evaporators (Pic. 1)

4|2|1

4|3|2|1

Extraction units(Soxhlet apparatus – Pic. 2)

up to 4up to 14

Rotary evaporators

Soxhlet apparatus

Picture 1

Picture 249

Temperature control products

Temperature control productsTechnical data/// Temperature control instruments

TECHNICAL DATA/// Comparison
ICC basic | control
Ident. No. 0004134400 |
0004136600

IC basic | control
Ident. No. 0003861000 |
0003863000

Instrument typeCompact immersion circulator

Immersion circulator

Safety class

III (FL)

III (FL)HBC 5 basic | control
Ident. No. 0004125000 |
0004127000

HBC 10 basic | control
Ident. No. 0004135000 |
0004137000

Instrument typeHeated bath circulator

Heated bath circulator

Safety class

III (FL)

III (FL)Heat output (230 V)

2000 W

2500 W

Heat output (230 V)

2500 W

2500 W

Cooling capacity––

Cooling capacity––

Coolant––

Coolant––

Working temperature range

RT + 10 °C to + 150 °C

RT + 10 °C to + 250 °C

Working temperature range

RT + 10 °C to + 250 °C

RT + 10 °C to + 250 °C

Operating temperature range (with outside coolant)- 20 °C to + 150 °C- 20 °C to + 250 °C

Operating temperature range (with outside coolant)- 20 °C to + 250 °C- 20 °C to + 250 °C

Temperature display

LED | TFT

LED | TFT

Temperature display

LED | TFT

LED | TFT

Display resolution

0,1 °C | 0,01 °C

0,1 °C | 0,01 °C

Display resolution

0,1 °C | 0,01 °C

0,1 °C | 0,01 °C

Setting resolution

0,1 °C

0,1 °C

Setting resolution

0,1 °C

0,1 °C± 0,02 K | ± 0,01 K± 0,02 K | ± 0,01 K

Temperature consistency inaccordance with DIN 12876± 0,02 K | ± 0,01 K± 0,02 K | ± 0,01 K

Bath capacity (liters)

dependent on the bath used

dependent on the bath used

Bath capacity (liters)

5,5 – 7,5 l

7,5 – 10,5 lUseable volume (liters)

dependent on the bath used

dependent on the bath used

Useable volume (liters)

2l

3l

Temperature consistency inaccordance with DIN 12876

50

TECHNICAL DATA/// Comparison

Pump power (pressure)

0,3 bar0,61 bar

Pump power (pressure)

0,61 bar

0,61 bar

Pump power (suction)

0,2 bar

0,45 bar

Pump power (suction)

0,45 bar

0,45 barMax. flow rate

18 l/min

31 l/minMax. flow rate

31 l/min

31 l/minDimensions (W × H × D)

145 × 340 × 200 mm

285 × 313 × 291 mm

Dimensions (W × H × D)

275 × 406 × 500 mm

275 × 456 × 506 mm

Weight3,75 kg

8,5 kg | 8,8 kg

Weight17,0 kg | 17,3 kg

18,0 kg | 18,3 kg

Permissible ambient temperature5 – 40 °C

5 – 40 °C

Permissible ambient temperature5 – 40 °C

5 – 40 °C

Permissible relative humidity

80 %

80 %

Permissible relative humidity

80 %

80 %

USB/RS 232 interface

yes

yes

USB/RS 232 interface

yes

yes

Connection for external PT 100 probeno | yes

yes

Connection for external PT 100 probe

yes

yes

Threaded connection

optional

M 16 × 1

Threaded connection

M 16 × 1

M 16 × 1

Cooling coil includednono | yes

Cooling coil included

yes

yes

Multi IO port includednono | yes

Multi IO port included

no | yesno | yes€ 910,00 | € 1.499,00€ 2.299,00 | € 2.949,00€ 2.899,00 | € 3.549,00€ 3.099,00 | € 3.749,00

51

Temperature control products

Temperature control products

TECHNICAL DATA/// Comparison

TECHNICAL DATA/// Comparison
CBC 5 basic | control
Ident. No. 0004165000 |
0004167000

HRC 2 basic | control
Ident. No. 0025003742 |
0025004524

Instrument typeHeated bath circulator

Heated bath circulator

Safety class

III (FL)

III (FL)RC 2 basic | control
Ident. No. 0004171000 |
0004173000

RC 5 basic | control
Ident. No. 0004181000 |
0004183000

Instrument type

Recirculating chiller

Recirculating chiller

Safety class––Heat output (230 V)

2500 W

1500 W

Cooling capacity

400 W

1400 W

Cooling capacity

350 W I 400 W at 3200 rpm

400 W

Coolant

R134a

R290

Coolant

R134a

R134a

Working temperature range- 20 °C to RT | - 30 °C to RT- 30 °C to RT

Working temperature range- 25 °C to + 200 °C- 20 °C to + 100 °C |- 30 °C** to + 100 °C

Operating temperature range(with outside heating)- 20 °C to + 80 °C |- 30 °C to + 80 °C- 30 °C to + 80 °C

Operating temperature range (with outside coolant)- 30 °C** to + 200 °C- 20 °C to + 100 °C |- 30 °C to + 100 °C

Temperature display

LED | TFT

LED | TFT

Display resolution

0,1 °C | 0,01 °C

0,1 °C | 0,01 °C

Temperature display

LED | TFT

LED | TFT

Setting resolution

0,1 °C

0,1 °C

Display resolution

0,1 °C | 0,01 °C

0,1 °C | 0,01 °C

Setting resolution

0,1 °C

0,1 °C

Temperature consistency inaccordance with DIN 12876± 0.1 K | ± 0.05 K± 0,1 K

Temperature consistency inaccordance with DIN 12876± 0,02 K | ± 0,01 K± 0,05 K

Bath capacity (liters)

1,5 – 4 l

5,2 – 8 l

Useable volume (liters)

2,5 l

2,8 l

Bath capacity (liters)

5–7l

1,5 – 4 l

Pump power (pressure)

0,3 bar0,61 bar

Useable volume (liters)

2l

2,5 l

Pump power (suction)

0,2 bar

0,45 bar

Pump power (pressure)

0,61 bar

0,5 bar

Pump power (suction)

0,45 bar

0,2 barMax. flow rate

31 l/min21 l/minDimensions (W × H × D)

275 × 490 × 690 mm

220 × 525 × 475 mm

Weight39,5 kg

28,5 kg

Permissible ambient temperature5 – 32 °C

5 – 32 °C

Permissible relative humidity

80 %

80 %

USB/RS 232 interface

yes

yes

Connection for external PT 100 probeyesno | yes

Threaded connection

M 16 × 1

M 16 × 1

Cooling coil includednonoMulti IO port included

no | yesMax. flow rate

18 l/min

31 l/minDimensions (W × H × D)

220 × 475 × 525 mm

310 × 490 × 546 mm

Weight28,0 kg | 28,5 kg

37,5 kg

Permissible ambient temperature5 – 32 °C

5 – 32 °C

Permissible relative humidity

80 %

80 %

USB/RS 232 interface

yes

yes

Connection for external PT 100 probeno | yes*no | yes*

Threaded connection

M 16 × 1

M 16 × 1

Multi IO port includednono | yes€ 3.099,00 | € 3.999,00€ 4.599,00 | € 5.249,00

no | yes* PT 100 Fühler im Lieferumfang enthalten.

€ 4.600,00 | € 5.304,00€ 3.749,00 | € 4.590,00* PT 100 temperature probe included.

** - 30 °C possible at 2000 rpm.

52

53

Temperature control products

Temperature control products

Scope of delivery/// Temperature control instruments

SCOPE OF DELIVERY/// Comparison
ICC
basic

ICC
control

Pump connection set

IC
basic

IC
control×

Cooling coil CC 1
PT 100 interface

××

HBC 5 | 10
basic

HBC 5 | 10
control

CBC 5
basic

×

×

×

Pump connection set

×

×

×

Cooling coil CC 1

×

×

×

PT 100 interface

External PT 100 probe

CBC 5
control

×

×

×

×

HRC 2
basic
×

HRC 2
control

RC 2 | 5
basic

RC 2
control

RC 5control

×

×

×

×

×

×

×

×

×

External PT 100 probe

USB interface

×

×

×

×

×

×

USB interface

×

×

×

×

×

×

×

RS 232 interface

×

×

×

×

×

×

RS 232 interface

×

×

×

×

×

×

×

×

Multi IO interface

×

×

×

×

×

1 x USB cable (station)

×

×

×

Multi IO interface
1 x USB cable (station)

×
×

×
×

×

×
×

×

×

1 x USB cable (WiCo)××

1 x USB cable (WiCo)

×

×

×

×

Charger for WiCo××

Charger for WiCo

×

×

×

×

Power cable
Barb fittings for DN 12 hoses (2 ×)

×

×

×

×

×

×

Power cable

×

×

×

×

×

×

×

×

×

×

×

Barb fittings for DN 12 hoses (2 ×)

×

×

×

×

×

×

×

×

×

×

×

×

×

×

Barb fittings for DN 8 hoses (2 ×)

Barb fittings for DN 8 hoses (2 ×)WiCo wall mount

WiCo wall mount

HeatingHeating and Cooling

Cooling

× = included with delivery

54

55

Temperature control products

Temperature control products

Accessories/// Baths and covers

BATH VESSELS/// Comparison

BRIDGE AND COVER/// Comparison

Bath typeOuter
dimensions [mm]Innerdimensions [mm]Bath size S

Price

ICC

IB 8 eco,
Ident. No. 0004248100

Plastic bath,
8l

IB R 9 eco,
Ident. No. 0020004382

Plastic bath,
9 l, rectangular

356 × 287 × 167

307 × 239 × 151

IB 9 pro,
Ident. No. 0004248500

Stainless steelbath, 9 l

377 × 374 × 195

292 × 230 × 150

335 × 320 × 155

286 × 227 × 150€ 330,00

Stainless steelbath, 12 l

461 × 354 × 195

317 × 292 × 150

Bridge BS.ICC,
Ident. No. 0020003077

S€ 120,00€ 350,00

Bridge BL.ICC
Ident. No. 0020003078

M+L€ 125,00€ 470,00

Cover CS.ICC
Ident. No. 0004471500

S€ 120,00

Cover CM.ICC
Ident. No. 0025000290

M€ 150,00

Cover CL.ICC
Ident. No. 0004471600

L€ 170,00

M+L€ 123,00

Cover CM.IC
Ident. No. 0004577600

M€ 123,00

Cover CL.IC
Ident. No. 0004471800

L€ 164,00

Bath size M
IB 12 pro,
Ident. No. 0004577500

Suitable forthe following sizesPrice€ 530,00
IC

Bath size L
IB 18 eco,
Ident. No. 0004248200

Plastic bath,18 l

IB 24 HF eco,
Ident. No. 0020006884

Bridge BS.IC,
Ident. No. 0004472800
584 × 338 × 155

490 × 286 × 150€ 460,00

Plastic bath,
24 l, tall form

584 × 338 × 205

490 × 286 × 200€ 520,00

IB R 20 eco,
Ident. No. 0020004383

Plastic bath,
20 l, rectangular

552 × 365 × 167

504 × 317 × 151€ 460,00

IB R 52 eco,
Ident. No. 0020007044

Plastic bath,
52 l, rectangular

648 × 365 × 316

600 × 317 × 300€ 650,00

IB R RO 15 eco,
Ident. No. 0020006341

Plastic bath,for magnetic
stirrer RO 15

722 × 365 × 165

674 × 317 × 98

€ 790,00

IB 20 pro,
Ident. No. 0004248600

Stainless steelbath, 20 l

641 × 354 × 195

495 × 292 × 150€ 630,00

Floating globes, PP›› out of Polypropylene›› for covering
open water baths›› only for water›› ø 20 mm›› 500 pcs.

Information: eco - water, + 100 °C | pro - water, oil, + 200 °C

IB 18 eco
Plastic bath, 18 l

56

IB HF 24 eco
Plastic bath,tall form, 24 l

IB 9 pro
Stainless steel bath, 9 l

IB 20 pro
Stainless steel bath, 20 l

IB R RO15 eco
for magnetic stirrer RO 15

BS.ICC
BridgeCS.ICC
Cover

Ident. No. 0020003666€ 105,00

57

Temperature control products

Temperature control products

Accessories/// Immersion racks

IMMERSION RACKS/// Comparison

FLOATING RACKS/// Comparison

ø Tubes
[mm]

Depth
[mm]

Immersion
depth
[mm]

Numberof tubes

Floating tube rack 1
Ident. No. 0020003667

13

100

70

57

€ 80,00

Tube rack, 17 mm, stainlessIdent. No. 0020004027

17

100

100

37

€ 80,00

Tube rack, 22 mm, stainlessIdent. No. 0020004028

22

100

50

22

€ 80,00

13

100

70

73

€ 135,00

Tube rack, 17 mm, ML, stainlessIdent. No. 0020004030

17

100

100

47

€ 135,00

Tube rack, 22 mm, ML, stainlessIdent. No. 0020004031

22

100

50

30

€ 135,00

Variable rack for bath sizes M/L
Number of racks with ICC: 1/2 | Number of racks with IC: 1 (only L)
Variable rack, ICC, ML, stainlessIdent. No. 0020004032

Unit ofmeasurePrice

1.5/2.0 ml

24

5 pcs.

€ 31,00

Floating tube rack 2
Ident. No. 0020003668

15 ml

8

5 pcs.

€ 31,00

Floating tube rack 3
Ident. No. 0020003669

50 ml

4

5 pcs.

€ 31,00

FIXING CLIPS/// for Ident. No. 0020004032, 0020004614, 0020006212, 0020007353

Racks for bath sizes M/L
Number of racks with ICC: 1/3 | Number of racks with IC: 2 (only L)
Tube rack, 13 mm, ML, stainlessIdent. No. 0020004029

Max. numberof samples

Price

Racks for bath size S
Number of racks with ICC: 1
Tube rack, 13 mm, stainlessIdent. No. 0020004026

Suitable
vessels

–

132

–

–

€ 175,00

Inlay 1, variable rack, ICC, 2 pcs.
Ident. No. 0020004033

13–

0 – 120*84€ 55,00

Inlay 2, variable rack, ICC, 2 pcs.
Ident. No. 0020004034

17–

0 – 120*51€ 55,00

Inlay 3, variable rack, ICC, 2 pcs.
Ident. No. 0020004035

22–

0 – 120*33€ 55,00

AS 2.1 Fixing clips
Ident. No. 0001234300

Suitable sample vessels

Price

25 ml

€ 22,50

AS 2.2 Fixing clips
Ident. No. 0001234400

50 ml€ 22,50

AS 2.3 Fixing clips
Ident. No. 0001234500

100 ml€ 22,50

AS 2.4 Fixing clips
Ident. No. 0001234600

200/250 ml€ 44,00

AS 2.5 Fixing clips
Ident. No. 0001234700

500 ml€ 44,00

Bottom plates size L
Number of racks with ICC: 1; except for bottom plate L: 0 | Number of racks with IC: with variable bottom: 1; else 0
Variable bottom, ICC, stainlessIdent. No. 0020004614–

240

0 – 115–€ 385,00

Bottom plate L
Ident. No. 0020006212–

460

50 – 110–€ 420,00

Bottom plate L.ICC
Ident. No. 0020007353–

290

50 – 110–€ 350,00

Tube rack
13 mm, S, stainless

Variable rack ICC
ML, stainlessin combination with inlay 1

Variable bottom ICC
L, stainless

Bottom plate L* in 15 mm increments

58

59

Temperature control products

Temperature control products

Accessories/// Bath fluids
60

50

Viscosity [mm2/s]

40

Chemical basis›› Silicon (Si)Bath fluid type›› Heating fluid (HF)›› Universal fluid (UF)Viscosity

UF.Si.150.N30.10 LV

20

HF.Si.20.200.50

10

UF. Si. N20.150. 20 LV
Minimum/
maximum temperature

30

Additional information›› Low viscosity (LV)›› Contains additives (A)0

HF.Si.20.250.50A
-30

10

50

90

130

170

210

250

290

Temperature [°C]

SILICON-BASED TEMPERATURE CONTROL FLUIDS/// Comparison
Temperature range

Viscosity
at + 25 °C

Color

QuantityPrice

HF.Si.20.250.50 A
Ident. No. 0020003521+ 20 °C to+ 250 °C*

50 mm²/s

reddishtranslucent

10 kg€ 480,00

HF.Si.20.200.50
Ident. No. 0020003520+ 20 °C to+ 200 °C**

50 mm²/s

clear

10 kg

€ 430,00

10 mm²/s

clear

9 kg

€ 450,00

Heating Fluids

Universal Fluid
UF.Si.N30.150.10 LV
Ident. No. 0020003518- 30 °C to+ 150 °C**** + 250 °C only in enclosed baths (HBC), otherwise + 200 °C** + 250 °C only for a short time in enclosed baths*** + 130 °C in open baths

60

61Temperature control products

Temperature control products

Accessories/// Temperature control hoses

TECHNICAL DATA/// Comparison

TECHNICAL DATA/// Comparison
H.PVC.8 | H.PVC.12
Ident. No. 0004568800 |
0004568900

H.SI.8 | H.SI.12
Ident. No. 0004569000 |
0004569100

H.PUR.8.R | H.PUR.12
Ident. No. 0020004612 |0020004613

Unit of measure22

2, incl. 4 hose clamps

Length

1,5 m

1,5 m

H.FKM.8 | H.FKM.12
Ident. No. 0004569200 |
0004569300

LT 5.20 | LT 5.21
Ident. No. 0002606700 |
0020000988

Unit of measure

2, incl. 4 hose clamps (only H.FKM.8)

2

1,5 m

Length

1,5 m

1,5 m

Material

Viton (FKM/FPM)

Stainless steel | PTFE

Material

PVC

SiliconPUR clear, reinforced |PUR clearØ internal [mm]8 | 12

8 | 12

8 | 12

Ø internal [mm]8 | 12

10 | 13

Ø external [mm]12 | 16

12 | 16

12 | 16

Ø external [mm]12 | 1645 | 38

Connection

for hose barb fitting

for hose barb fitting

for hose barb fitting

Connection

for hose barb fitting/M 16 × 1

M 16 × 1

Temperature range- 20 °C to + 60 °C- 30 °C to + 180 °C- 30 °C to + 90 °C

Temperature range- 30 °C to + 180 °C- 30 °C to + 300 °C | - 30 °C to + 260 °C

Max. operating pressure (+ 20 °C)

depressurized operation

depressurized operation

8 bar | 3 bar

Max. operating pressure (+ 20 °C)

6 bar/1 bar (>1 bar on request)

6 bar

transparent

transparent frosted

transparent frosted |transparent

Color

black (additional
stainless steel sheathing)red€ 12,00 | € 15,00€ 30,00 | € 36,00€ 55,00 | € 68,00€ 280,00 | € 370,00€ 999,00 | € 849,00

Color

H.PVC.8

H.PVC.12

H.SI.8

H.SI.12

Water
Silicon oil
Ethylene glycol
Mineral oil

H.PUR.8.R

H.PUR.12

H.FKM.8

H.FKM.12

LT 5.20

LT 5.21
ISO.8›› Hose insulation
for DN 8 hoses›› 1.5 m›› 2 pcs.

ISO.12›› Hose insulation
for DN 12 hoses›› 1.5 m›› 2 pcs.

Ident. No. 0004569400

Ident. No. 0004569500€ 25,00€ 32,00

Ethan oil

RESISTANCE
very good resistance at + 80 °C
not resistant
do not operate at › + 30 °C

6263

Temperature control products

Temperature control products

Accessories/// Temperature control instruments

SOLENOID VALVES/// Comparison

HOSE BARB FITTINGS AND ADAPTERS/// Comparison
Description

Connection

Price

MV 1
Ident. No. 0020003763

Solenoid valve for cooling water
regulation, 0 to + 90 °C, max. 10 bar

2 hose olives DN8 included€ 380,00

CO V 1
Ident. No. 0020000249

Shut-off valve for external
tempering, - 40 °C to - 180 °C,
max. 1 bar

Directly on circulator, second
page for hose olives M 16 × 1€ 560,00

Manually operated ball valve

With union nut on one side formounting on M 16 × 1 thread.
Second connection M 16 × 1

Ball valve M 16 × 1
Ident. No. 0020004620€ 80,00

FURTHER ACCESSORIES/// for temperature control instruments

CO V 1
64

Unit of measure

Price

Barb fitting for DN 6 hoses
Ident. No. 0020004667

Barb fitting adapter forDN 6 hose2€ 26,00

Barb fitting for DN 8 hoses
Ident. No. 0020004566

Barb fitting adapter for
DN 8 hose2€ 26,00

Barb fitting for DN 10 hoses
Ident. No. 0020004568

Barb fitting adapter for
DN 10 hose2

€ 26,00

Barb fitting for DN 12 hoses
Ident. No. 0020004889

Barb fitting adapter for
DN 12 hose2

€ 26,00

Adapter NPT 1/4
Ident. No. 0020004569

Adapter M 16 × 1 to
NPT 1/4 (male)2€ 40,00

Adapter NPT 1/2
Ident. No. 0020004570

Adapter M 16 × 1 to
NPT 1/2 (male)2€ 125,00

Adapter NPT 3/4
Ident. No. 0020004571

Adapter M 16 × 1 to
NPT 3/4

2€ 150,00

Description

Price

Pump connection set for ICC€ 140,00

Lock nut M 16 × 1
Ident. No. 0020004583

Nut for mounting hose barb fitting
adapters, stoppers, NPT adapters2€ 30,00

PT 100.30
Ident. No. 0004284700

Temperature probe, stainless steel, 250 mm€ 285,00

Stopper
Ident. No. 0020004584

For sealing purposes, used in
combination with a lock nut

2€ 30,00

PT 100 extension (Lemo)
Ident. No. 0020004629

Extension cable PT 100, lemo connector, 3 m€ 175,00€ 74,00

WiCo (wireless controller) wall mount€ 20,00

90° tube adapter, e.g. used forconnecting hoses without creatinga bend1

WH 10
Ident. No. 0020000984

Elbow tube 90°
Ident. No. 0025001212

PC 1.1
Ident. No. 0002616700

RS 232 cable, 3 m (10 ft)€ 33,50

Labworldsoft 6 Pro
Ident. No. 0020017366

Laboratory software€ 9.167,00

Labworldsoft 6 Starter
Ident.No. 0020019397

Laboratory software€ 4.017,00

Barb fitting for
6 mm/8 mm
Barb fitting adapter

Adapter NPT 1/4
Adapter M 16 × 1 to
NPT 1/4 (male)

Adapter NPT 1/2
Adapter M 16 × 1 to
NPT 1/2 (male)

Stopperfor sealing purposes

Lock nut M 16 × 1
for mounting hosebarb fitting adapters

Elbow tube 90°
90° tube adapter

PCS.ICC
Ident. No. 0004471900

MV 1

Description

CC1›› Cooling coil for IC basic

CC2›› Cooling coil for ICC

Mechanic fluid
level controller›› Fluid level monitor for operating heating bath circulators orcoolers on open baths.

Ident. No. 0020005116

Ident. No. 0025001061

Ident. No. 0020004618€ 128,00€ 120,00€ 310,00

Adapter NPT 3/4
Adapter M 16 × 1 to
NPT 3/4 (male)65Temperature control products

Temperature control products

ICC Packages/// Heating bath circulators and combined
heating bath circulators

TECHNICAL DATA/// ICC Packages
Bath size

ECO BATHS MADE OF PLASTIC CAN BE USEDAT TEMPERATURES OF UP TO + 100 °C(H2O ONLY). PRO BATHS MADE OF STAINLESSSTEEL CAN BE USED AT TEMPERATURESOF UP TO + 200 °C.

ICC basic eco 8
Ident. No. 0008034900
ICC control eco 8
Ident. No. 0008035300

total: 320 × 335 × 349 mmbath opening: 188 × 105 mm*
max. bath opening: 227 × 118 mm**

ICC basic pro 12
Ident. No. 0010000414
ICC control pro 12
Ident. No. 0010000415

Cooling coilM

ICC basic eco 18
Ident. No. 0008035000
ICC control eco 18
Ident. No. 0008035400
ICC basic pro 20
Ident. No. 0008035200

Cover€ 1.299,00€ 1.899,00

S

ICC control pro 9
Ident. No. 0008035500

Bath bridge

Price

ICC Package 1 - Heating bath circulators | includes: ICC basic/control head, bath bridge, bath vessel

ICC basic pro 9
Ident. No. 0008035100

ICC basic or control head

Dimensions (W × D × H)*

total: 374 × 377 × 388 mmbath opening: 195 × 100 mm*
max. bath opening: 230 × 121 mm**

€ 1.439,00

total: 354 × 461 × 388 mmbath opening: 255 × 127 mm*
max. bath opening: 292 × 147**€ 1.499,00

total: 338 × 584 × 349 mmbath opening: 245 × 305 mm*
max. bath opening: 286 × 325 mm**

€ 1.449,00

total: 354 × 641 × 388 mmbath opening: 255 × 309 mm*
max. bath opening: 292 × 325 mm**

€ 1.599,00€ 1.999,00€ 2.099,00€ 2.029,00

L

ICC control pro 20
Ident. No. 0008035600€ 2.199,00

ICC Package 2 - Combined heating bath circulators | includes: ICC basic/control head, bath bridge,
bath vessel, cover, cooling coil, pump connection set, PT 100.30 temperature probe (control version only)
ICC basic eco 8 c
Ident. No. 0008035700
ICC control eco 8 c
Ident. No. 0008036100

Bath vessel
Pump connection set
PT 100 temperature probe(control version only)

ICC control pro 9 c
Ident. No. 0008036300

ICC Package 1›› ICC basic/control head›› Bath bridge›› Bath vessel

ICC Package 2›› ICC basic/control head›› Bath bridge›› Bath vessel›› Cover›› Cooling coil›› Pump connection set›› PT 100 temperature probe(control version only)

from € 1.299,00

from € 1.699,00

ICC control pro 12 c
Ident. No. 0010000417

M

ICC basic eco 18 c
Ident. No. 0008035800
ICC control eco 18 c
Ident. No. 0008036200
ICC basic pro 20 c
Ident. No. 0008036000
ICC control pro 20 c
Ident. No. 0008036400€ 1.699,00€ 2.499,00

S

ICC basic pro 9 c
Ident. No. 0008035900

ICC basic pro 12 c
Ident. No. 0010000416

66

total: 320 × 335 × 349 mmbath opening: 188 × 105 mm*
max. bath opening: 227 × 118 mm**

total: 374 × 377 × 388 mmbath opening: 195 × 100 mm*
max. bath opening: 230 × 121 mm**

€ 1.799,00

total: 354 × 461 × 388 mmbath opening: 255 × 127 mm*
max. bath opening: 292 × 147**€ 1.899,00

total: 338 × 584 × 349 mmbath opening: 245 × 305 mm*
max. bath opening: 286 × 325 mm**

€ 1.849,00

total: 354 × 641 × 388 mmbath opening: 255 × 309 mm*
max. bath opening: 292 × 325 mm**

€ 2.019,00€ 2.649,00€ 2.749,00€ 2.699,00

L€ 2.849,00* Dimensions to set plate at the bottom of bath.
Immersion depth for all baths with ICC: 85 mm to 125 mm.
** Maximum bath opening dimension at the upper edge of the bath.

67

Temperature control products

Temperature control products

IC Packages/// Combined heating bath circulators

ICC RO 15 Package›› ICC basic/control head›› Magnetic Stirrer RO 15›› Bath vessel IB R RO 15 eco›› Bath bridge BL.ICC

IC Package›› IC basic/control head›› Bath bridge›› Bath vessel›› Cover›› PT 100 temperature probe›› Cooling coil**from € 3.399,00

from € 3.349,00

TECHNICAL DATA/// ICC RO 15 Package

TECHNICAL DATA/// IC Package
Bath size

Dimensions (W × D × H)*

Price

Bath size

Dimensions (W × D × H)*

Price

ICC RO 15 Package | includes: ICC circulator, bath bridge BL.ICC, bath vessel IB R RO 15 eco, multiposition magnetic stirrer RO 15

IC Package - Combined heating bath circulators | includes: IC circulator, bath, bath bridge, cover, cooling coil**,
PT 100.30 temperature probe

ICC basic IB R RO 15
Ident. No. 0010002471

IC basic pro 12 c
Ident. No. 0008039900

L

ICC control IB R RO 15
Ident. No. 0010002474

total: 365 × 722 × 388 mmbath opening: 317 × 492 mm€ 3.399,00€ 3.999,00

IC control pro 12 c
Ident. No. 0008040000
IC basic pro 20 c
Ident. No. 0008036800

Immersion depth for all baths with IB R RO 15: 35 to 75 mm

M

IC control pro 20 c
Ident. No. 0008037200

L

total: 354 × 461 × 359 mm
bath opening: 255 × 80 mm*
max. bath opening: 292 × 105 mm***

total: 354 × 641 × 359 mm
bath opening: 255 × 262 mm*
max. bath opening: 292 × 278 mm***€ 3.349,00€ 3.949,00€ 3.499,00€ 4.099,00* Dimensions to set plate at the bottom of bath. | ** Already included in scope of delivery for IC control device.
Immersion depth for all baths with IC: 95 mm to 135 mm.
*** Maximum bath opening dimension at the upper edge of the bath.

6869

Temperature control products

Temperature control products

labworldsoft® software

Calibration and adjustment/// Advanced software with an innovative visualapproach to lab automation/// Two-point or three-point calibration

IKA‘s software labworldsoft® enables the networking of up to
64 laboratory devices which can be controlled simultaneously
via one PC. Not only IKA products, but also other manufacturers
laboratory devices can be integrated using labworldsoft®.
This makes the automation of your laboratory experiments andprocesses much more easy and efficient.

The internal (and external, if used) temperature sensor can be adjusted via
either a two-point or three-point calibration process. On request, calibration can also be performed in the plant by the IKA service team or by an
external, ISO- and DAkks certified service provider.
If you would like to request this service, please contact our service department by telephone at 00 8000 4524357 (00 8000 IKAHELP) or by email
service@ika.de.

Labworldsoft 6 Pro

Ident. No. 0020017366€ 9.167,00

Labworldsoft 6 Starter

Ident. No. 0020019397€ 4.017,00

HARDWARE AND SOFTWARE REQUIREMENTS/// labworldsoft® Software

TRY IT FOR YOURSELF
Get your free, fully functional 30 day trial
version download at: www.labworldsoft.com

70

Windows 7/8.1/10 (64-bit operating system) with at least 2 GB
RAM and 100 MB free hard disc memory. Some devices require
the installation of a specific device driver. Communications
interfaces can be retrofitted using an adapter at the PC, if
necessary.

71Temperature control products

Temperature control products

Customizing Center

IKA Application Support

It is important that IKA products perform in real laboratory applications.
We have a special program of product solutions that are customized toyour individual needs. If you cannot find the right device in our standard
product range, please send us the details of the specification you needusing the online form. Our team will check the feasibility of the specification and offer you a solution.

Our Application Center spans 400 m² and is equipped with the mostmodern facilities for presenting and testing laboratory equipment andprocesses. The Center brings us even closer to our customers and improves our service. If you are interested, you can use our facilities to test
processes that include stirring, shaking, dispersing, grinding, heating,analysis and distillation.

Please visit www.ika.com to have look at the product modification
requests that we have already implemented.

Please call us at 00 8000 4522777 (00 8000 IKAAPPS)
or send us an email applicationsupport@ika.de.

Worldwide service network/// Direct contacts in your region
Our dedicated team of engineers provides comprehensive technical service on a global level. If you have any questions, please do not hesitate tocontact IKA directly. Alternatively, you can get in touch with your dealer.

Our Application Center covers 400 m²
and offers modern equipment for
demonstrating and testing laboratorydevices and processes.

We would be happy to help you findthe perfect device for your application.

Send us your sample. We will
run a test with the suitable device within 48 hours.

Interested individuals and customers can test processes including
stirring, shaking, dispersing, milling,heating, analyzing and distilling.

IKA guarantees that spare parts will be available for 10 years. In theevent of any faults with a device, or if you have any technical questions
regarding our products, their maintenance or replacement parts, pleasecall us at 00 8000 4524357 (00 8000 IKAHELP) or send us an email
service@ika.de.

72

73

EN

USA
IKA Works, Inc.
Phone: +1 910 452-7059
eMail: sales@ika.net

KOREA
IKA Korea Ltd.
Phone: +82 2 2136 6800
eMail: info@ika.kr

BRAZILIKA Brasil
Phone: +55 19 3772-9600
eMail: sales@ika.net.br

MALAYSIA
IKA Works (Asia) Sdn Bhd
Phone: +60 3 6099-5666
eMail: sales.lab@ika.my

INDIA
IKA India Private Limited
Phone: +91 80 26253 900
eMail: info@ika.in

CHINA
IKA Works Guangzhou
Phone: +86 20 8222 6771
eMail: info@ika.cn

JAPAN
IKA Japan K.K.
Phone: +81 6 6730 6781
eMail: info_japan@ika.ne.jp

POLANDIKA Poland Sp. z o.o.
Phone: +48 22 201 99 79
eMail: sales.poland@ika.com

ENGLAND
IKA England LTD.
Phone: +44 1865 986 162
eMail: sales.england@ika.com

201912_ThermostateLab_EN_IWS_EUR_ 94000320/// WORLDWIDETEMPERATURE CONTROL PRODUCTS MADE BY IKA.

IKA-Werke GmbH & Co. KG
Janke & Kunkel-Straße 10, 79219 Staufen, Germany
Phone: +49 7633 831-0, Fax: +49 7633 831-98
eMail: sales@ika.de

VIETNAMIKA Vietnam Company Limited
Phone: +84 28 38202142
eMail: sales.lab-vietnam@ika.com/// SOCIAL MEDIA

For more information on our products or to order online, please visit:www.ika.com

IKAworldwide

Prices are valid until October 31, 2020.
All prices exclude VAT.
IKA reserves the right to modify products,
part numbers, and pricing.

IKAworldwide /// #lookattheblue@IKAworldwide

www.ika.com/// ONLINESHOP