Digital Blood Pressure Monitor
Model UA-1200BLE
Instruction Manual

EnglishManuel d’instructions

Français

Manual de Instrucciones

EspañolManuale di Istruzioni

Italiano

Bedienungsanleitung

DeutschOriginal

Traduction

Traducción

TraduzioneÜbersetzung

使用手冊 中文翻譯

1WMPD4003740B‫دﻟﯾل اﻹرﺷﺎدات‬‫اﻟﺗرﺟﻣﺔ‬‫اﻟﻌرﺑﯾﺔ‬Contents
Dear Customers .......................................................................2

Preliminary Remarks .................................................................2
Precautions ..............................................................................3

Parts Identification ...................................................................6
Contents ..................................................................................6
Symbols ...................................................................................7

Using the Monitor...................................................................10
Measurements ........................................................................16
What is an The IHB Indicator? ..................................................20

What Is The IHB? .....................................................................20

WHO Classification Indicator ...................................................20
About Blood Pressure .............................................................21
Settings by App ......................................................................23
Setting by Monitor ..................................................................24
Troubleshooting .....................................................................25

Maintenance ...........................................................................26
Technical Data........................................................................26

English 1

Dear Customers
Thank you for purchasing a state-of-the-art A&D blood pressure monitor,
one of the most advanced monitors available today. Designed for ease ofuse and accuracy, this monitor will facilitate your daily blood pressure
regimen.
We recommend that you read through this manual carefully before
using the device for the first time.
The Bluetooth® word mark and logos are registered trademarks owned byBluetooth SIG, Inc. and any use of such marks by A&D is under license.
Other trademarks and trade names are those of their respective owners.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
Products. This is made evident by themark of conformity. (0123:
The reference number to the involved notified body)
Hereby, A&D Company, Limited declares that the radio equipment typeUA-1200BLE is in compliance with Directive 2014/53/EU. The full text of
the EU declaration is available at the following internet address:
http://www.aandd.jp/products/manual/manual_medical.html
The device is designed for use 13 years old and older, not newborns orinfants.
Environment for use. The device is for use to operate by yourself
in the home healthcare environment.
This device is designed to measure blood pressure and pulse rate of people
for diagnosis.

English 2

Precautions
Precision components are used in the construction of this device.
Extremes in temperature, humidity, direct sunlight, shock or dust should
be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened withwater and a neutral detergent. Never use alcohol, benzene, thinner orother harsh chemicals to clean the device or cuff.
The device and cuff are not water resistant. Prevent rain, sweat andwater from soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions,microwave ovens, cellular telephones, X-ray or other devices with strong
electrical fields.
When reusing the device, confirm that it is clean.
Used equipment, parts and batteries are not treated as ordinary
household waste, and must be disposed of according to the applicable
local regulations.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the upper arm must be squeezed by the cuff
hard enough to temporarily stop blood flow through the artery. This may
cause pain, numbness or a temporary red mark to the upper arm. Thiscondition will appear especially when measurement is repeatedsuccessively.
Any pain, numbness, or red marks will disappear with time.
If you feel strong pain during the measurement, please remove the cuff.
Measuring blood pressure too frequently may cause harm due to bloodflow interference.
Avoid tightly folding the cuff for long periods, as such action may shorten
the life of the components.
When the AC adapter is used, make sure that it can be readily removed
from the electrical outlet when necessary.

English 3

Sometimes the temperature of the cuff can reach 44 °C in an ambient
temperature of 40 °C. If you feel the cuff is too hot, please remove thecuff immediately.
Please do not simultaneous use HF surgical equipment.
Clinical testing has not been conducted on newborn infants and pregnant
women. Do not use on newborn infants or pregnant women.
If you have had a mastectomy, please consult a doctor before using thedevice.
Do not let children use the device by themselves and do not use thedevice in a place within the reach of infants. It may cause accidents or
damage.
There are small parts that may cause a choking hazard if swallowed by
mistake by infants.
Do not touch the batteries and the patient at the same time. That may
result in electrical shock.
Wireless communication devices, such as home networking devices,mobile phones, cordless phones and their base stations, walkie-talkies
can affect this blood pressure monitor. Therefore, a minimum distance of
30 cm should be kept from such devices.
Unplug the AC adapter when not in use during the measurement.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially causeburn.
Allow the device to adapt to the surrounding environment before use(about one hour).
Do not inflate without wrapping the cuff around the upper arm.
The automatic blood pressure monitor's performance may be afficted byexcessive temperature or humidity, or altitude.

English 4

Contraindications

The following are precautions for proper use of the device.
Do not apply the cuff on upper arm in which another medical device is
attached. The device may not function properly.
People who have a severe circulatory deficit in the upper arm must
consult a doctor before using the device, to avoid medical problems.
Do not self-diagnose the measurement results and start treatment by yourself.
Always consult your doctor for evaluation of the results and treatment.
Do not apply the cuff on upper arm with an unhealed wound.
Do not apply the cuff on upper arm receiving an intravenous drip or bloodtransfusion. It may cause injury.
Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as ahigh-pressure oxygen chamber or an oxygen tent. It may cause a fire or explosion.

English 5

Parts Identification

LED

OLED

Start Switch

Reset Switch

Cuff

USB Connector

Contents
・1 Blood Pressure Monitor
・1 Instruction Manual
・1 Charging Cable
・1 QR code card
English 6

Symbols

Symbols that are printed on the device case and cuff
Symbols

Function / MeaningStandby and turn the device on.
Direct current
Type BF: Device and cuff are designed to provide special
protection against electrical shocks.
CE marking
Authorized Representative
Manufacturer
Date of manufacture

IP

SN
BT

International protection symbol: IP classification indicates thedegrees of protection provided by enclosures in accordancewith IEC 60529. This device is protected against solid foreign
objects of 12mm diameter and greater such as a finger. This
device is not protected against water.
WEEE label
Serial number
Bluetooth address
Refer to instruction manual/booklet
To indicate generally elevated, potentially hazardous, levels of
non-ionizing radiation, or to indicate equipment or systems e.g.
in the medical electrical area that include RF transmitters orthat intentionally apply RF electromagnetic energy for diagnosis
or treatment.
Keep dry
Universal Serial Bus

English 7

Symbols

Symbols that appear on the display
Symbols
SYS
DIA

Function / Meaning
Systolic blood pressure in mmHg.
Diastolic blood pressure in mmHg

PUL

Pulse per minute
The battery power is low.

Charge the device

User Select mode

Select the user

Application software is connectedwith the device.
Application software has
disconnected from the device.
Appears while measurement is inprogress.
E PUL
/min
Err 1

Err 3

Err 9

Recommended Action

The pulse is not detected correctly.
Unstable blood pressure due tomovement during measurement. The
systolic and diastolic values arewithin 10mmHg from each other.
The pressure value did not increaseduring inflation. The cuff is notapplied correctly.

Measurement is inprogress. Remain as
still as possible.
Apply the cuff correctly,
and take anothermeasurement.
Take anothermeasurement. Remain
still during measurement.

Apply the cuff correctly,
and take anothermeasurement.
Press the reset button.
After reset, clock data
Blood pressure monitor internal error. will disappear. If theerror still appears,
contact A&D Medical.
Data is being transferred
Measurements can betaken but device will nottransmit data wirelessly.

Airplane mode

English 8

Symbols
Symbols

Function / Meaning

Recommended Action

Appears when a body or upper arm
movement is detected.

The reading may yield anincorrect value. Take
another measurement.
Remain still during
measurement.

Appears during measurement when
the cuff is attached loosely.
Updating firmware.
Appears when irregular heart beat is
detected.

English 9

The reading may yield
an incorrect value.
Apply the cuff correctly,
and take anothermeasurement.

Using the Monitor
UA-1200BLE Digital Blood Pressure Monitor
features 5 measurement modes.
1.

In-App Mode Measurement:
User can take measurements and view their trends in the A&D Medical
Connect App.

2.

Offline Mode Measurement:
Measurement can also be completed without the A&D Medical
Connect App.
Data is stored on the monitor. The data will be transferred to the App on
where connecting with it.

3.

Multiuser Mode Measurement: (In-App):Up to 6 users (5 accounts and 1 visitor) can use and store data on
the App. Each user must have a mobile device paired with the monitor.

4.

Guest Mode Measurement: (In-App):
User can take a measurement as Guest, without an account registrationin the A&D Medical Connect App. The data is saved on the mobile
device memory but not in your account.

5.

Visitor Mode Measurement (Offline):
Select “0 Visitor” to take a measurement in visitor mode fromthe monitor only. Data is not stored in visitor mode.

English 10

Using the Monitor
Charging the Device
Battery may be partially charged but be sure to charge the device
completely upon first use after purchase.
1. Remove the USB connector cover.

2. Connect the charging cable.

3. During charging, the LED be on.

4. When charging is complete, the LED will be off.

5. Once charging is complete, remove the charging
cable.

English 11

Using the Monitor
CAUTION!
 The battery life varies with the ambient temperature and may be
shorter at low temperatures.
 Please use the specified charging cable shipped with the device.
 Blood pressure measurement cannot be done during charging.
 When disconnecting the adapter from the electrical outlet,
hold and pull the adapter body out the outlet.
 When disconnecting the charging cable from the blood pressure
monitor, hold and pull the charging plug out of an outlet.
 Do not connect to device other than those specified.
 Do not change the battery.
 If the battery can not be charged any more, please contact theauthorized dealer.
 Overcharging the battery may reduce its lifetime.
 Lithium battery replacement by inadequately trained personnel
could result in a hazard such as a fire or explosion.

Wireless Function
In the unlikely event that this monitor causes radio wave interference to adifferent wireless station, change the location where this monitor is used orstop using immediately.

Be sure to use in a location where visibility between the two devices that you
want to connect is good. The connection distance is reduced by the structureof buildings or other obstructions. In particular, connection may be
impossible when devices are used on either side of reinforced concrete.



Do not use Bluetooth® connection in the range of a wireless LAN or otherwireless devices, near devices that emit radio waves such as microwaves, inlocations where there are many obstructions, or in other locations where
signal strength is weak. Doing so may result in frequent loss of connection,very slow communication speeds and errors.

English 12

Using the Monitor


Using close to an IEEE802.11g/b/n wireless LAN device may cause mutualinterference to occur, which may result in reduced communication speeds or
which may prevent connection.
In this case, switch off the power supply to the device that is not being used,
or use the monitor in a different location.



If the monitor does not connect normally when used near a wireless stationor broadcast station, use the monitor in a different location.



A&D Company, Limited cannot accept liability for any damages incurred due toimpaired operation or data loss, etc. that occur through the use of this product.

This product is not guaranteed to connect to all Bluetooth® compatible devices.

Bluetooth® Transmission
This product is equipped with a Bluetooth® 4.1 wireless function and canconnect to the following Bluetooth® devices.
Apple iOS v.8.0 or newer
iPhone 4s or newer,
iPad touch 5th gen;
iPad 3,4; iPad Air; iPad mini

Android OS 4.4.4,5.1,6.0, and 7.0
Bluetooth® devices Selected
Android phones only.

The latest compatibility list

https://www.aandd.co.jp/products/hhc/hhc-ict_compatibility.html
To connect with your mobile device – download and install the “A&D
Medical Connect application” onto your mobile device.

English 13

Using the Monitor
Device Setup (Pairing)
1.Ensure the Bluetooth® function of themobile device is ON.
2.Launch up A&D Medical CONNECT App.
The icon to the left will
blink on the monitor.
3.Accept the pairing request from the
pop-up window.

Multiuser Pairing
The monitor can be registered for up to 5 users. Follow the above
instructions to pair the mobile device for each user.
New users can be added anytime. The user who has the lowest frequencyof measurement times will be cleared once the device has reached the 5user max.
Note: If new user uses the same mobile device, then re-pairing is not required as thishas been completed.
Note: Default is in visitor mode.

CAUTION!
The communication distance between this device and the mobile
device is about 33 feet (10m). This distance may vary depending on
the surrounding environment so try to get closer to the receiving
device if the data doesn't get transferred.

English 14

Using the Monitor
Applying the Arm Cuff

Wrap the cuff around the upper arm, about 0.5”(1-2cm) above the insideof the elbow, as shown. Sit comfortably to ensure proper fit/orientation on
the arm.

Note: White tab should be closer to the artery.

How to Take Accurate Measurements

For the most accurate blood pressure measurement:
Sit comfortably on a chair. Place your arm on the table. Do not cross yourlegs. Keep your feet on the floor and straighten your back.
Relax for about five to ten minutes before the measurement.
Place the center of the cuff at the same height as your heart.
Remain still and keep quiet during the measurement.
Do not measure immediately after physical exercise or a bath. Rest fortwenty or thirty minutes before taking the measurement.
Try to measure your blood pressure at the same time every day.

English 15

Measurements
In-App Mode Measurement
1.

Ensure the Bluetooth® function of the mobile device is ON.

2.

Open the A&D Medical Connect App. The App will automatically detect
the presence of the device.

3.

Tap the Take Measurement
button, followed by the Start buttonon the App screen.
User number

Monitor display

Username

Take measurement
App display

Start

4.

The cuff starts to inflate. It is normal for the cuff to get tight. The App displays
the pressure value once inflation starts.

5.

When the measurement iscomplete, the App and themonitor will display results
similar to the displays on the right.
(Systolic bp diastolic bp and pulse rate
will be displayed in order.)Monitor display

Result notification
Icons(Refer to pg.9,20
for details)

App display

Note: Press the start button again to make the device standby mode. The device is also
provided with an automatic power shut-off function. Allow at least one minutebetween measurements on the same person.

English 16

Measurements
Offline Mode Measurement
1.

Press the Start/Stop button on the monitor. After
the A&D logo appears for 1 second, the User
Select screen will be displayed.

2.

Press the Start/Stop button again to select a user or switch between
username from User Select screen.
User name can be changed in the App under Settings. When this is not set,
the display name on App will be displayed as the user name.
User number

User name

Select User

3.

Measurement starts automatically after 3 seconds.

4.

Zero(0) is displayed blinking briefly. Then thedisplay changes, as measurement begins.

5.

The cuff starts to inflate. It is normal for the
cuff to get tight. Monitor displays the pressure
value once inflation starts.

6.

When the measurement is complete, themonitor will display result similar to the display
on the right. (Systolic bp, diastolic bp andpulse rate will be displayed in order.)Note: Press the start/stop button again to make the monitor standby mode. The device is also
provided with an automatic power shut-off function. Allow at least one minutebetween measurements on the same person.

English 17

Measurements
Upload Stored Measurement
The device can store up to 100 readings and can betransferred to the App on next connection.
To transfer data, open the A&D Medical Connect App andselect Upload Measurement from the dashboard pop-up
window.

Read stored readings

English 18

App display

Measurements
Visitor Mode Measurement
1. Press the START button on the monitor.
2. Press the START button again to switch between username and select “0Visitor”.
3. Measurement starts automatically after 3 seconds.
4. When the measurement is complete, the monitor
will display results similar to the display on the right.
(Systolic, diastolic and pulse rate will be
displayed in rotation.)
Visitor Mode is only available through the monitor. Data is not stored invisitor mode.

Notes for Accurate Measurement

Sit down in a comfortable position. Place your arm on a table with yourpalm facing upward and the cuff at the same level as your heart.
Relax for about five or ten minutes before taking a measurement. If you
are excited or depressed by emotional stress, the measurement willreflect this stress as a higher (or lower) than normal blood pressure
reading and the pulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you
are doing and what you have eaten. What you drink can have a verystrong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very
weak or irregular heartbeat, the device may have difficulty determining
your blood pressure.
If the device detect a condition that is abnormal, it will stop themeasurement and display an error symbol. See page 8 for thedescription of the symbols.
This blood pressure monitor is intended for use by 13 years and older.
Consult with your physician before using this device on a child. A childshould not use this device unattended.
This automatic blood pressure monitor’s performance may be affected byexcessive temperature or humidity, or altitude.

English 19

What is an The IHB Indicator?
When the monitor detects an irregular rhythm during the measurements,
the IHB indicator will appear on the display with the measurement
values.
Note: We recommend contacting your physician if you see thisIHB indicator frequently.

What Is The IHB?
The UA-1200BLE blood pressure monitor provides blood pressure and apulse rate measurement even when an irregular heartbeat occurs. An
irregular heartbeat is defined as a heartbeat that varies from the averageof all heartbeats during the blood pressure measurement.

WHO Classification Indicator
This Indicator’s by App:
Each segment of the bar
indicator corresponds to the
WHO blood pressureclassification described on thenext page.

English 20

About Blood Pressure
What is Blood Pressure?

Blood pressure is the force exerted by blood against the walls of the arteries.
Systolic pressure occurs when the heart contracts. Diastolic pressure occurs
when the heart expands. Blood pressure is measured in millimeters of
mercury (mmHg). One's natural blood pressure is represented by thefundamental pressure, which is measured first thing in the morning while one
is still at rest and before eating.

What is Hypertension and How is it Controlled?

Hypertension, an abnormally high arterial blood pressure, if left untreated,
may cause many health problems including stroke and heart attack.
Hypertension can be controlled by altering lifestyle, avoiding stress, and withmedication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?

Blood pressure measured at a clinic or doctor's office may cause
apprehension and may lead to elevate reading by 25 to 30 mmHg higher
than that measured at home. Home measurement reduces the effects ofoutside influences on blood pressure readings, supplements the doctor's
readings and provides a more accurate, complete blood pressure history.

English 21

About Blood Pressure
WHO Blood Pressure Classification
Reference Material: Journal of Hypertension 1999, Vol 17 No.2

mmHg
Diastolic blood pressureStandards to assess highblood pressure, without
regard to age, have been
established by the World
Health Organization (WHO),as shown in the chart.

Grade 3 hypertension (severe)

110

Grade 2 hypertension (moderate)

105
100

Grade 1 hypertension (mild)

95
90

High-normal

85
80

NormalOptimal120

130 140 150 160 170 180
Systolic blood pressure
mmHg

Blood Pressure Variations
An individual’s blood pressure varies greatly on a daily and seasonal basis.
It may vary by 30 to 50 mmHg due to various conditions during the day. In
hypertensive individuals, variations are even more pronounced. Normally,
the blood pressure rises during at work or play and falls to its lowest levels
during sleep. So, do not be overly concerned by the results of one
measurement.

English 22

Typical fluctuation within a day(Measured every five minutes)
Systolic
Blood pressure

Take measurements at the same time every
day using the procedure described in thismanual to get to know your normal bloodpressure. Regular readings give a morecomprehensive blood pressure history. Be
sure to note the date and time whenrecording your blood pressure. Consult yourdoctor to interpret your blood pressure data.

Sleep

Diastolic
PM

AM

Time

Settings by App
Setting
Nickname

Start Time

Remove User
Check Software
UpdateUnpairReset

Description
User can overwrite the default monitor
display user name and assign a personal
username (up to 10 characters).
Enables a user to choose how long of an interval
between pressing “Start” button and when themonitor will start measurement. For multiple user, alonger interval is useful. For single users, a shorter
interval is useful. Default for single user is set at 1
second and for multiple user is set at 5 seconds.
Allows user to remove himself or herself as a user inthe BP monitor.
Enables users to update the monitor with the mostrecent Bluetooth® software.
Allows user to unpair their mobile device with the
BP monitor. This can also be accomplished via theBluetooth® setting in the mobile device.
This function will reset the monitor to factory settings.
All date in the monitor will be deleted.
This will not affect any data stored in the mobile
device.

English 23

Setting by Monitor
Monitor Setting
The airplane mode feature can be used in an environment than does
not allow to emit radio frequency. Measurement can be done in
airplane mode and data will be transfered to the App on next
connection.
1. Hold the Start/Stop button and the airplane mode
select screen will be displayed. Press the Start/Stopbutton to switch between ON and OFF.
2. Select ON and wait 3 seconds to make it airplanemode.
In airplane mode, the airplane icon will be shown at thebottom left corner of the screen.

English 24

Troubleshooting
ProblemPossible Cause
Low battery

Suggestion
Charge the battery.

Powerdoes notturn on.

Blood Pressure monitor
internal error.

Press the Reset Switch.

Battery charging cable isconnected.

Take off the charging cable.

Low battery

Charge the battery.

The cuff
does notinflate.
The device
does notmeasure.
Readings
are toohigh or too
low.

Movement is detected.

The cuff position is notcorrect.

Make sure to remain still and quiet
during measurement.
If you have a very weak or irregularheartbeat, the device may havedifficulties in determining your bloodpressure.
Sit comfortably and still.
Place your arm on a table with yourpalm facing upward and the upperarm cuff at the same level as yourheart.

The value At a clinic or doctor’s office Home measurement reduces theis defferent apprehension may cause
effects of outside influences on
from that
an elevated reading.
blood pressure readings,
measured
supplementing the doctor’s
at a clinic
readings.
or doctor’s
office.
Note: If the solution described above do not solve the problem, contact theauthorized dealer. Do not attempt to open or repair this product, otherwise
your warranty may be invalid.

English 25

MaintenanceDo not open the device. It uses delicate electrical components and an
intricate air unit that could be damaged. If you cannot fix the problem
using the troubleshooting instructions, contact the authorized dealer inyour area or our customer service department. The A&D customer
service will provide technical information, spare parts and units toauthorized dealers.
The device is designed and manufactured for a long service life.
However it is generally recommended to have the device inspected
every 2 years, to ensure proper functioning and accuracy. Please contact
the authorized dealer in your area or A&D for maintenance.

Technical Data
Type

UA-1200BLE

Measurement method

Oscillometric measurement

Measurement range

Pressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 - 180 beats / min

Measurement accuracy

Pressure: ±3 mmHg
Pulse:±5 %Arm circumference

22 cm - 42 cm

Power supply

3.7 V Li-ion 325 mAh

Number of measurements Approx. 100 times
With pressure value 170 mmHg,
room temperature 23 °C.
Classification

Internally powered ME equipment(Supplied by batteries) /Class II (Supplied by adapter)
Continuous Operation mode

Clinical test

According to ISO81060-2

English 26

Technical Data
EMD

IEC 60601-1-2: 2014

Wireless communication

ZY (MURATA Manufacturing Co. Ltd.)
Bluetooth® Ver.4.1 Low Energy
Technology BLP
Frequency band: 2402 MHz to 2480 MHz
Maximum RF output power: 2.1 dBm
Modulation:GFSK

Operating conditions+10 °C to +40 °C / 15 %RH to 85 %RH/
800 hPa to 1060 hPa
Transport / Storage conditions -20 °C to +60 °C / 10 %RH to 95 %RH/
700 hPa to 1060 hPa
Dimensions

Approx. 44 [W] x 126 [H] x 26 [D] mm

WeightApprox. 205 g

Ingress protection

Device: IP20

Applied part

Cuff Type BF

Useful life

5 years (when used six times a day)Included items

UA-1200BLE - 1PC
Instruction Manual - 1PC
Charging cable (KSUSB-2C) - 1PC
QR code card - 1PC

OptionAC adapter: AX-TB283

English 27

MEMO

English 28

A&D Company, Ltd.

1-243 Asahi, Kitamoto-shi, Saitama-ken 364-8585 JAPAN
Telephone: [81] (48) 593-1111
Fax: [81] (48) 593-1119

Emergo Europe B.V.

Prinsessegracht 20, 2514 AP The Hague, THE NETHERLANDS

A&D INSTRUMENTS LIMITED

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY
United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

A&D ENGINEERING, INC.

1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119

A&D AUSTRALASIA PTY LTD

32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd 爱安德技研贸易（上海）有限公司

中国 上海市自由贸易试验区浦东南路855号世界广场32楼C,D室 邮编200120(32CD, World Plaza, No.855 South Pudong Road,China (Shanghai) Pilot Free Trade Zone,
200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599