GXA-03
GXA-04GXA-06

1WMPD4003498

RS-232C Isolated Interface
Comparator output
Analog voltage output

© 2017 A&D Company, Limited. All rights reserved.
No part of this publication may be reproduced, transmitted, transcribed, or translated
into any language in any form by any means without the written permission of A&D
Company, Limited.

The contents of this manual and the specifications of the instrument covered by thismanual are subject to change for improvement without any notice and obligation on
the part of the manufacturer.

Contents
1.

Introduction................................................................................................................................... 2

2.

Features ....................................................................................................................................... 2

3.

Packing List.................................................................................................................................. 3

4.

Installing the Option ..................................................................................................................... 3

5.

Function Table of Balance........................................................................................................... 4
5.1.
Operation Keys and Display of Function Table................................................................... 4
5.2.
Items List of the Function Table ........................................................................................... 5

6.

GXA-03 Isolated RS-232C interface .......................................................................................... 8
6.1.
Specifications of Serial Interface.......................................................................................... 87.

GXA-04 Comparator Output.......................................................................................................9
7.1.
Specifications of Comparator Output................................................................................... 9
7.2.
Using the Comparator Output............................................................................................10
7.3.
Comparison Method Selection (Tolerance Range and Center Value or Threshold Values) .............. 11
7.4.
Setting Threshold Values ...................................................................................................12
7.5.
Setting Center Value and Tolerance Range......................................................................13

8.

GXA-06 Analog Voltage Output................................................................................................15
8.1.
Specifications of Analog Voltage Output............................................................................15
8.2.
Function Table of Analog Voltage Output..........................................................................16
8.3.
Voltage Output Selection....................................................................................................18
8.4.
Voltage Output Fine Adjustment ........................................................................................18
8.5.
Fixed Voltage Output ........................................................................................................19

1. Introduction
This manual describes options for the GX-A series and GF-A series and how to get the mostout of them in terms of performance. Read this manual thoroughly before using the options
and keep it at hand for future reference.

2. Features
The following options can be used for the GX-A series and GF-A series.
GXA-03Isolated RS-232C interface
GXA-04Comparator output (with buzzer) / External contact input x 2
GXA-06
Analog voltage output

GXA-03: Isolated RS-232C interface
Commands from the PLC or other devices installed in the factory environment can be used tomake the following operations : output measurement data from the balance, input parameters
into the balance, control the balance, reference parameters of the balance, etc.
The RS-232C is equipped with a data format to manage the balance in accordance with GLP.
( Refer to the instruction manual of the balance. )GXA-04: Comparator output / External contact input
The GXA-04 can compare the weighing value and preset threshold values and can output
the result to the contact output.
The GXA-04 is equipped with six contact outputs: "HH", "HI", "OK", "LO", "LL" of thecomparison output and "READY" output to indicate the status of the balance.
3-stage comparison or 5-stage comparison can be selected.
The GXA-04 can sound a buzzer according to the comparison result.
The GXA-04 is equipped with external control input terminals that can make the RE-ZEROand PRINT key operations. An optional foot switch (AX-SW137-PRINT,
AX-SW137-RE-ZERO ) can be used.

GXA-06: Analog voltage output
The GXA-06 can output analog voltage in the following modes :"mode where the specified digits of the weighing value are converted to voltage" and"mode where weighing value is converted to voltage in range between gross zero or net zeroand full scale".
The voltage output range can be selected using the "0V /0.2V " switch on the panel.
It can be selected "0 to 1V" range and "0.2V to 1V" range. The factory setting is "0 to 1V" range.
2

3. Packing List
GXA-03 ...............This manual
GXA-04 ...............Mini DIN connector ( 8 pin, plug )Cable with connector ( AX-KO5692 )This manual
GXA-06 ...............Screwdriver for adjustment
Mini DIN connector ( 4 pin, plug )Cable with connector ( AX-KO5692 )This manual

1
1
1
1
1
1
1
1

4. Installing the Option
Caution Disconnect the AC adaptor before installing the option to the balance.

Installing the option board to the balance
Step1

Step2

Step3

Remove the two screws from the panel.
Pull gently the cable and option board
from the balance.
Peel off the connector that is affixed
to the panel with double-sided tapeand insert it into the socket of theoption board.

Panel
ScrewsConnector

Option panel: Confirm
the layout of terminal,
etc. on the option.
Socket : Confirm the positionof the socket on the option.

Attach the option board to the balance.
Secure with the two screws removed in step 1.

Assembling the mini-DIN connector
You can use the GXA-04 ( AX-K0592 ) and GXA-06 ( AX-K0593 )of the option cable assembled with the mini-DIN connector.
Step 1 Pass the cable through the rubber sleeve.
Step 2 Solder wires and terminal.

Top metal case

Step 3 Cover the terminal with the top
metal case and bottom metal case.
Step 4 Fasten wires with the top metal case.

Part to secure wires

Cable
Terminal

Step 5 Insert the assembled parts into therubber sleeve.

Rubber sleeve
Fasten wires tightly
Solder wires and terminal
Bottom metal case3

5. Function Table of Balance
The function table stores parameters that specify how to operate the balance, and it is used tomointor or update those parameters. These parameters are stored in memory even if the AC
adaptor is removed and are retained until overwritten. Balance operations are specified bythis table when using the GXA-03, GXA-04 and GXA-06. The table menu consists of twolayers as shown in the diagram below. The first layer is the “Class” and the second layer is the“Item”. Each item stores a "parameter". The last displayed parameter is in effect. After the
PRINT key is pressed, these updated parameters are applied to operations of the balance.

Example of settings and menu structure
Example : Set to "Center value using sample weight input" of "Input method" in "Comparator".
Start from weighing mode
Press and holdPress several times

Press several times

Press several times

Parameter
Center value using numerical input
Parameter
Center value using sample weight

Press

Press

Item
Input method

Class

Finish to weighing mode
CautionThe balance may not function properly, depending on the settings and the operating
environment. Be sure to set parameters correctly.

5.1.

Operation Keys and Display of Function Table
The 〇 indicates that the parameter displayed is in effect.
Enters the function table when pressed and held in the weighing mode.
Selects the class or item in the function table.
Changes the parameter.
Moves to an item in the class when a class is displayed.
Stores the new parameter and displays the next class when an item is displayed.
Cancels new parameter and displays the next class when an item is displayed.
Exits the function table and returns to the weighing mode when a class is displayed.
4

5.2.
Note

Items List of the Function Table

Functions available differ by model of the balance.

Class

Item

Param
eter

Description

Environment, display( Refer to the instruction manual of main unit )Clock adjustment
No comparison (Comparator is not used)
Comparison when stable value or overloaded
Comparison always
3-stage comparison
HI, OK, LO
5-stage comparison
HH, HI, OK, LO, LL
Comparison including near zeroComparison excluding ±5 digits (counts) from zeroComparison excluding ±10 digits (counts) from zeroComparison excluding ±20 digits (counts) from zeroComparison excluding ±50 digits (counts) from zeroComparison excluding ±100 digits (counts) from zeroComparison of positive values only
Comparison of negative values only
Comparison of both polarities
Threshold value using numerical input,,
are usedThreshold value using sample weight andCenter value using numerical input
,
and
are
used
Center value using sample weight

Comparator mode
Comparison method

Near zeroComparator

Polarity

Input method

Second upper limit
Upper limit

Refer to explanation of "7. GXA-04 Comparator output".
Input number when numerical input(or) is selected.

Lower limit,

are displayed when 5-stage comparison) is selected.

(
Comparator
thresholdvalue

Second lower limit
Center value
Tolerance range
from center value
Second tolerance
range from center
value

Refer to explanation of "7. GXA-04 Comparator output".
Input sample weight when sample weight input(or) is selected.

(

is displayed when 5-stage comparison) is selected.

indicates factory settings. Digit is the unit of minimum division (minimum weighing value).
5ClassItem
HH buzzer
Hi buzzer

Comparator
buzzerOK buzzer
Lo buzzer
LL buzzer

Param
eter

Description
Off
On
Off
On
Off
On
Off
On
Off
On

Displayed only when 5-stage comparison() is selected.

Displayed only when 5-stage comparison) is selected.
(

Data output
Serial interface( Refer to instruction manual of main unit )USB interface
Connection

Baud rate

Option serialinterface

Data bit, parity bit
Terminator

Data format

Timeout
AK, Error code

Computer
Printer
External display
600 bps
1200 bps
2400 bps
4800 bps
9600 bps
19200 bps
38400 bps
7 bits, Even
7 bits, Odd
8 bits, None
CR LF
CR: ASCII code 0Dh
Do not use
LF: ASCII code 0Ah
A&D standard format
DP format
KF format
MT format
NU format
CSV format
No limit
Select wait time to receive a command
1 second
No output
AK: ASCII code 06h
Outputindicates factory settings.
6

ClassItem

Analog output mode

Analog output
Output digit
selection

Param
eter

Description
2-digit output mode
3-digit output mode
Net full scale output mode
Gross full scale output modeFirst digit
Second digit
Third digit
4th digit
5th digit
6th digit
7th digit

Displayed only when
the GXA-06 isconnected.

Displayed only when
the GXA-06 isconnected.

Application
Alarm function for minimum
weighing value
Unit( Refer to instruction manual of main unit )Density function
ID number
Password lock
Automatic calibration##
Internal mass value correction( Refer to instruction manual of main unit )indicates factory settings. # : Settings for GX-A series.

76. GXA-03 Isolated RS-232C interface
This isolated RS-232C interface is used to communicate with peripherals.
Connect the AC adaptor to this option (GXA-03) when AD-8529R-W or AD-8529PC-W is used.
Refer to the instruction manual of main unit concerning function of RS-232C

6.1.

Specifications of Serial Interface

This interface is DCE. Use straight cable to connect to a computer ( DTE ).
Transmission system EIA RS-232C
Transmission
Asynchronous, bi-directional, half-duplex
Data format
Baud rate:
600, 1200, 2400, 4800, 9600, 19200, 38400 bps
Data bits:
7 bits or 8 bits
Parity bit:
Even, Odd
(Data length 7 bits)
None(Data length 8 bits)
Stop bit:
1 bit
Code:
ASCII
RS-232C
Connector
D-Sub9 pin [male]-5 V to -15 V
Data bits
Start bit

+5 V to +15 V
Stop bit
Parity bit

Pin assignments

Pin No. Signal Direction
Description
1
N.C.
No connection
2
RXD
Output Transmit data
3
TXD
Input Receive data
4
N.C.
No connection
5
SG
Signal ground
6
DSR
Output Data set ready
7
RTS
Input Request to send
8
CTS
Output Clear to send
9N.C.
No connection
Signal names are those of the DTE side.

Balance (DCE)
Inside balance

Straight cableComputer (DTE)
Terminal example

Approx. 5 V

8Inch screw#4-40UNC

7. GXA-04 Comparator Output
The comparator output (GXA-04) has the function to compare between the weighing value
and preset second upper limit, upper limit, lower limit or second lower limit. The result ofcomparison can be output to the contact. When the contact is shorted, this option can selectwhether or not to sound the buzzer.
Optional foot switch ( AX-SW137-PRINT, AX-SW137-RE-ZERO ) or other external switches
can be used.

7.1.

Specifications of Comparator Output

50 V DC
100 mA DC
20 ΩCompatible plug : 8 pin mini DIN connector [Female](Accessory)
E8-200J-100 or equivalent
External contact input connector Compatible plug :φ3.5 mm 3 pole mini stereo jack
MP-013LC or equivalent parts

Maximum contact voltage:
Maximum contact current:
Maximum contact resistance:
Output connector

Panel External View
8

7

6

3

Comparator output connector

4
5
2
1

Comparator output circuit

External contact input circuit

External contact input x2
2

3

1

Example of contact input terminal

Pin assignments and wire colors ofaccessory cable
Pin No.
Description
1
READY Output
2
LL
Output3
LO
Output4
N.C.
No connection
5
OK
Output6
HI
Output7
HH
Output8
COM
Housing Case
Pin No.
1
GND
2
RE-ZERO
3
PRINT

Wire color
Red
White
Black
Yellow
Blue
Green
Brown
Gray
ShieldDescription
External contact input
External contact input

Optional foot switch can connect to either input.
9Using External Contact Input
When 1 pin and 3 pin or 1 pin and 2 pin of lead wire of optional foot switch( AX-SW137-PRINT, AX-SW137-RE-ZERO ) is shorted 100 ms or longer, it can be
operated the same as the PRINT key and RE-ZERO key on the panel.

7.2.

Using the Comparator Output

This option can output results of the 3-stage comparison or 5-stage comparison that is
specified at "Comparison method ()" in the function table of the balance.
Comparison results are displayed using HI , OK , LO and are output using contacts ofI/O terminals. Comparison tolerance ranges are the following 3 types and they can be
specified at "Comparator mode ()" in the function table.
"No comparison", "Comparison when stable value or overloaded", or "Comparison always"
The range of near zero can is specified at "Near zero ()" in the function table.
There are two types of comparison method. One is the method using second upper limit,
upper limit, lower limit and second lower limit. Another is the method using center value andtolerance range. There two types of input methods for second upper limit, upper limit, lower
limit and second lower limit. One is the numerical input. Another is the input using sample
weight. Balance can sound a buzzer according to the comparison result. That can be)" in the function table.
specified at "Comparator buzzer (When weighing value is not displayed in performing RE-ZERO, etc., READY contact
becomes "open". When weighing value is displayed, READY contact becomes "short".

3-stage comparison result

3-stage comparison - display, contact output

Weighing value
Threshold value Formula
UL < WVUL
LL ≦ WV ≦ UL
LL
WV < LL

Contact output
BuzzerHI OK LO
HI illuminated ShortOpen
OK illuminatedShort
LO illuminated Open
Short

ResultHI
OK
LO

Display

Weighing value: WV, Upper limit: UL, Lower limit: LL

5-stage comparison result5-stage comparison - display, contact output

Weighing value
Contact output
ResultDisplay
BuzzerThreshold value Formula
HH
HI
OK
LO
LL
HI blinking
SUL < WV
HH
Short
SUL
HI illuminated
UL < WV ≦ SUL
HI
Short
Open
UL
OK illuminated
LL ≦ WV ≦ UL
OK
Short
LL
LO illuminated
SLL ≦ WV < LL
LO
Open
Short
SLL
LO blinking
WV < SLL
LL
Short
Weighing value: WV, Second upper limit: SUL, Upper limit: UL, Lower limit: LL, Second lower limit: SLL
10

7.3.

Comparison Method Selection (Tolerance Range and Center Value or Threshold Values)

Setting Example

Comparison always ......................................................................
5-stage comparison ......................................................................
Comparison excluding ±5 digits (counts) from zero ...................
Comparison of both polarities ......................................................
Center value using numerical input ...................................................
Center value using sample weight ....................................................

Factory settingorProcedure
Step 1 Press and hold the SAMPLE key to enter the function table.
is displayed.
Step 2 Press the SAMPLE key to display(Comparator).
(Comparator).

Step 3 Press the PRINT key to enter

Step 4 Press the RE-ZERO key several times to display(Comparison always).
Step 5 Press the SAMPLE key to display
method).

(Comparison

Step 6 Press the RE-ZERO key to display
comparison).

(5-stage

Step 7 Press the SAMPLE key to display(Near zero).

Step 8 Press the RE-ZERO key several times to display(Comparison excluding ±5 digits (counts) from zero).
Step 9 Press the SAMPLE key several times to display(Input method).
Step 10 Press the RE-ZERO key several times to select input method.
(Center value using numerical input) or,(Center value using sample weight)
Step 11 Press the PRINT key to store new settings and use them.
or

11

7.4.

Setting Threshold Values

This explanation uses
(5-stage comparison) and specifies(Threshold value using numerical input) or(Threshold value using sample
weight) to(input method) in the function table and inputs the threshold value.

Selecting a input method

(If

or

Step 1 Press and hold the SAMPLE key to enter the function table.
Step 2 Press the SAMPLE key several times to display
Step 3 Press the PRINT key to enter)

is displayed.

(Comparator threshold value).

(Comparator threshold value).

Step 4 Press the SAMPLE key several times to input threshold value. Refer to the following"Threshold value using numerical input" or "Threshold value using sample weight".
Step 5 If the PRINT key is pressed when threshold value is displayed, the current parameters
are displayed ( all digits blinks ).

Numerical input of threshold values(If)

Step 6 If changing the parameters, press the RE-ZERO key to store the following keys.
SAMPLE key
RE-ZERO key
MODE key
PRINT key
CAL key

Moves blinking digit.
Changes value of blinking digit.
Changes polarity.
Stores and proceeds to step 7.
Cancels and proceeds to step 7.

If the current parameters will be maintained, press the PRINT or CAL key toproceed to step 7.
Step 7 Repeat step 4, 5 and 6 for other threshold values.

Storing threshold values using sample weight(If

Step 8 Press the RE-ZERO key to display 0.00 g .
Put the sample on the weighing pan and press the PRINT key to store weighing value
as the threshold value.
RE-ZERO key
PRINT key

Resets display of the balance to zero .
Stores weighing value and proceeds to step 9.

Step 9 Repeat step 4, 5 and 6 for other threshold values.

12)Storing threshold values using RS-232C commands from peripherals#

The second upper limit, upper limit, lower limit and second lower limit can be specified using
RS-232C commands from peripherals.
Example command
HH: +1000.000g
second upper limit
HI: +0800.000g
LO: +0600.000g
LL: +0400.000g: space of ASCII code 20 h

7.5.

upper limit
lower limit
second lower limit

Setting Center Value and Tolerance Range

This explanation uses
(5-stage comparison), specifies(Center value using numerical input) or(Center value using sample
weight input) to(input method) in the function table, and compares using thecenter value and numerical input of tolerance range.

Selecting a input method

(If

or

)

Step 1 Press and hold the SAMPLE key to enter the function table.
is displayed.
Step 2 Press the SAMPLE key several times to display
value).
Step 3 Press the PRINT key to display(comparator threshold(comparator threshold value).

Step 4 Press the SAMPLE key several times to display "Center value using numerical input" or"Center value using sample weight". Specify the center value, tolerance range and second
tolerance range.

Numerical input of center value(If

Step 5 Press the PRINT key while displaying. The current parameter is displayed (alldigits blinking). If parameter does not need to be changed, press the PRINT key or CAL
key and proceed to step 6. If the parameter is to be changed, press the RE-ZERO key andchange it using the following keys.
SAMPLE key
Moves blinking digit.
RE-ZERO key
Changes value of blinking digit.
MODE key
Changes polarity.
PRINT key
Stores and proceeds to step 6.
CAL key
Cancels and proceeds to step 6.
Step 6 Proceed to "Storing tolerance range from center value using numerical input".
13)

Storing center value using sample weight(If

Step 7 Press the PRINT key while displaying. The current parameter is displayed(all digits blinks). Press the RE-ZERO key to enter to the sample weight input mode.
Press the RE-ZERO key to display 0.00 g . Put sample of center value on the pan
and press the PRINT key. The current parameter can be confirmed (all digits blinking).
Press the PRINT key to store the center value.
Step 8 Next,

is displayed.

Step 9 Proceed to "Storing tolerance range from center value using numerical input".

Storing tolerance range from center value using numerical input
Step 10 Press the PRINT key while displaying. The current parameter is displayed.
If the tolerance range is to be changed, change it using the following keys.
Input the tolerance range with center value assumed to be 100 %.
SAMPLE key
RE-ZERO key
PRINT key
CAL key

Moves blinking digit.
Changes value of blinking digit.
Stores and proceeds to step 11.
Cancels and proceeds to step 11.

Step 11 Press the PRINT key while displaying. The current parameter is displayed.
If the second tolerance range is to be changed, change it using the following keys.
Input the second tolerance range with center value assumed to be 100 %.
SAMPLE key
RE-ZERO key
PRINT key
CAL key

Moves blinking digit.
Changes value of blinking digit.
Stores and proceeds to step 12.
Cancels and proceeds to step 12.

Step 12 Press the CAL key to return to weighing mode.

14)

8. GXA-06 Analog Voltage Output
The analog voltage output (GXA-04) can convert the specified weighing value to analog
voltage between 0 and 1 V and output it.

8.1.

Specifications of Analog Voltage Output

Analog Voltage Output
Output impedanceLinearity
Output connector

100  or less±0.3 % or lessCompatible plug:

Pin connections

Output
GND
0 V to 1 V

Output range
Input impedance ofthe device connected

4 pin mini DIN connector [Female](Accessory)
E4-200J-100 or equivalent
4 pin
3 pin
With the slide switch set to "0V "0.2 V to 1 V
10 k or greater

With the slide switch set to "0.2V "Panel External View
Analog output connector
4 Analog voltage output

3 Analog GND

Slide switch: Output range
Zero volume for fine adjustment
Span volume for fine adjustment21

Output circuit
Analog voltage output

Pin assignments andwire colors of accessory cable
Pin No.
12
34
Housing

15

Description
N.C.
No connection
N.C.
No connection
Analog GND(0 V)
Analog voltage output
Case

Wire color
Black
Blue
Red
White
Shield8.2.

Function Table of Analog Voltage Outputof the function table can be selected when this option is connected.

Analog output
Item

Analogoutput mode

Output digit
selection

Parameter

Description
2-digit output mode Converts 2 consecutive digits with minimum digit
selected by
to voltage.
3-digit output mode Converts 3 consecutive digits with minimum digit
to voltage.
selected by
Net full scale output mode
Outputs 0.000 V when the net weight is zero.
Outputs 1.000 V when the net weight is full scale.
Output is 0.000 V when zero is set using the RE-ZERO key.
Gross full scale output mode
Outputs 0.000 V when the gross weight is zero.
Outputs 1.000 V when the gross weight is full scale.
Tare operation using the RE-ZERO key will not affect the output.
(Note: If tare is extremely light, tare operation might change the zero
point. Thus affecting the output.)
Specify minimum digit to be output at
mode.
When
is set toor, the setting is in effect.
Specify first digit as minimum digits.
Specify second digit as minimum digits.
Specify third digit as minimum digits.
Specify 4th digit as minimum digits.
Specify 5th digit as minimum digits.
Specify 6th digit as minimum digits.
Specify 7th digit as minimum digits.

indicates factory settings.

ExampleIf.

Analog voltage output ( 0 V to 1 V )0.67 V
0.56 V
0.45 V
0.34 V
0.23 V
0.12 V
0.01 V

Caution The hidden high-order digits are regarded as zero.
The hidden least significant digit is regarded as zero (when the least significant digit ishidden using the SAMPLE key).

16

.

If

If

or

Model
GX-203A
GX-303A
GX-403A
GX-603A
GX-1003A
GX-1603A
GX-2002A
GX-3002A
GX-4002A
GX-6002A
GX-6001A
GX-10001A
ExampleAnalog voltage output ( 0 V to 1 V )0.567 V
0.456 V
0.345 V
0.234 V
0.123 V
0.012 V
0.001 V

Full scale
220 g
320 g
420 g
620 g
1100 g1620 g
2200 g3200 g4200 g6200 g6200 g10200 gModel
GF-203A
GF-303A
GF-403A
GF-603A
GF-1003A
GF-1603A
GF-2002A
GF-3002A
GF-4002A
GF-6002A
GF-6001A
GF-10001A

Full scale
220 g
320 g
420 g
620 g
1100 g1620 g
2200 g3200 g4200 g6200 g6200 g10200 gWhen the GX-2002A displays 200 g, the voltage output of( Using "0V " of the slide switch )1.000 V ×

is 0.1 V.

200 g= 0.100 V
2000 gCaution "Full scale" of the full scale output mode indicates the full scale values shown in thetables above. The output voltage may exceed 1.000 V, depending on the weighing value.
ExampleWhen the GX-2002A displays 2200 g, the voltage output is 1.1 V.
( Using "0V " of the slide switch )1.000 V ×

2200 g= 1.100 V
2000 g178.3.

Voltage Output Selection

The range of output voltage can be change using the slide switch on the panel.
Factory settings is "0V ".
"0V "(0 to 1 V)
: Zero 0.000 V, full scale 1.000 V"0.2V " (0.2 to 1 V)
: Zero 0.200 V, full scale 1.000 V

Panel External View
Analog output connector
4 Analog voltage output

3 Analog GND

Slide switch: Output range
Zero volume for fine adjustment
Span volume for fine adjustment

2
1

8.4.

Voltage Output Fine Adjustment

The voltage output has been adjusted at thefactory before shipment.
Using the ZERO volume and SPAN for fineadjustment and a voltmeter, voltage output canbe finely adjusted.

0V～/ 0.2V～ ZERO SPAN

Procedure
Step 1 Turn off the display using the ON:OFF key.
Voltage output becomes zero point status.
Adjust the ZERO volume so that the voltmeter indicates
0.000 V ( when the slide switch is set to "0V ").
Adjust the ZERO volume so that the voltmeter indicates
0.200 V ( when the slide switch is set to "0.2V ").

The display to output 0 V ( 0.2 V).

Step 2 While pressing and holding the PRINT and
SAMPLE keys, press the ON:OFF key to enter
the initializing mode. Then 1 V is output. Adjust the SPAN
volume so that the voltmeter indicates 1.000 V

The display to output 1 V.

Step 3 Repeat step 1 and step 2 until the correct voltage output
is obtained.
18

8.5.

Fixed Voltage Output

The voltage output is fixed under the following conditions:With the display off, 0 V is output ( or 0.2 V if the slide switch is set to "0.2V ") when notweighing during calibration, etc..
During the re-zeroing operation in gross full scale output mode (the previous output value is retained.

),

During the re-zeroing operation in 2-digit output mode (0 V is output ( or 0.2 V if the slide switch is set to "0.2V ").

),During the re-zeroing operation in 3-digit output mode (0 V is output ( or 0.2 V if the slide switch is set to "0.2V ").

),During the re-zeroing operation in Net full scale output mode (0 V is output ( or 0.2 V if the slide switch is set to "0.2V ").

),When -E (weighing value is negative overload) is displayed,
0 V is output ( or 0.2 V if the slide switch is set to "0.2V ").
When E (weighing value is positive overload) is displayed,
the following voltages are output.
Example Voltage output when E is displayed if the slide switch is set to "0V ".
Model
GX-203A
GX-303A
GX-403A
GX-603A
GX-1003A
GX-1603A
GX-2002A
GX-3002A
GX-4002A
GX-6002A
GX-6001A
GX-10001A.,.
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V.,.
1.100 V
1.067 V
1.050 V
1.033 V
1.100 V
1.013 V
1.100 V
1.067 V
1.050 V
1.033 V
1.033 V1.020 VModel
GF-203A
GF-303A
GF-403A
GF-603A
GF-1003A
GF-1603A
GF-2002A
GF-3002A
GF-4002A
GF-6002A
GF-6001A
GF-10001A

19.,.
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V
1.000 V.,.
1.100 V
1.067 V
1.050 V
1.033 V
1.100 V
1.013 V
1.100 V
1.067 V
1.050 V
1.033 V
1.033 V1.020 V

MEMO

20

3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-6148

A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119

A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

OX14 1DY United Kingdom

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 150-749( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 150-749 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4280

OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599