UV Sensor "UV-Cure"UV sensor for high UV irradiance (>100mW/cm2)
1/4

General Features

The “UV-Cure” is a UV sensor for high UV irradiance (>100mW/cm2) and moderate
operating temperature (<80°C). Typically this sensor is used as a duty sensor in LED andcooled medium pressure lamp based UV curing systems. It will be configured upon
individual customer’s requirements which are clarified within the order process.
Configurable parameters are the signal output type, the measurement range and thespectral responsivity. The signal output is configurable as a 0…5V or 0…10V voltage
output or a 4…20mA current loop. Digital output sensors are available with a MOD bus, aCAN bus or a USB interface.
The determination of the individual dynamic range needs customer’s assistance, e.g. information about the source tobe measured and a typical distance between the sensor and the source. A PTB traceable calibration can be ordered.
Figure 1 shows the different options regarding the spectral responsivity of the sensor. Our sales team is happy toassist our customers selecting the best suitable responsivity for the specific application. Alternatively, technicalreports and selection guides are available on our website providing further assistance.

spectral responsivity selection options
Figure 1 shows the available spectral responsivites. Table 1 shows the position of the peak and the 10% of maximun
margins. For UV measurement, by default, unfiltered broadband SiC is applied. If a UV source also emits radiationthat must not contribute to the sensor’s signal a filtered SiC sensor (UVC, UVB or UVA only) is to be selected. For
measurement of radiation above 390nm GaP based detectors are used.

Table 1: position of peak responsivity and10% of maximum margins, values in nm
l_Slowl_Shigh

280

221

358

331

309

367

280

231

309

275

225

287

UV+VIS

445

240

560

BLUE

445

390515

SRPeakBroadBUVAUVBUVC

Figure 1: available spectral responsivities

sglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.deRev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.

UV Sensor "UV-Cure"UV sensor for high UV irradiance (>100mW/cm2)
general Specifications

2/4

Fixed Specifications Parameter

ValueDimensions

Please refer to drawing on page 4.

Field of view

Please refer to graph on page 4.

WeightTemperature coefficient (30 to 65°C)

90 g
0.05 to 0.075%/K

Operating temperature-20 to +80˚C

Storage temperature-40 to +80˚CHumidity
Time constantConfigurable Specifications Parameter
Spectral sensitivity
Measurement range< 80%, non condensing
0.1s +/-20% - other time constants on request, device has 1st order low
pass characteristics
Value
Broadband UV, UVA, UVB, UVC, Bluelight or UV+VIS (see Fig. 1 at page 1)
This sensor is designed for measurement of high irradiance.
Any range between the lowest range of 10µW to 10mW/cm2 and thehighest range 20mW/cm2 to 20W/cm2 is configurable for analog sensors
Any range between the lowest range of 1µW to 10mW/cm2 and thehighest range 2mW/cm2 to 20W/cm2 is configurable for digital sensors

signal output Specifications
Signal Output 0 to 5 V
Supply voltage
Current consumption
Connections

0 to 5V voltage output proportional to the irradiance
7,5 to 24 VDC (0 to 5V output) and 12 to 24 VDC (0 to 10V)
< 30mA2m cable version: V-=brown, V+=white, Vout=green, shield=black
plug version 0-5V: GND=1(brown), V+=4(black), Vout=3(blue)
plug version 0-10V: GND = 2(white), V+=4(black), Vout=1(brown)

Dark offset voltage< 3 mV

Measurement range

3 orders of magnitude

Signal Output 0 to 10 V

available on request

sglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.de
Rev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.

UV Sensor "UV-Cure"UV sensor for high UV irradiance (>100mW/cm2)
3/4
Signal Output 4 to 20 mA
Supply voltage
Current consumption
Connections

Measurement range
Sensor compliance voltage
Max. loop resistanceoffsetSignal Output USB
Supply voltage
Current consumption
Connections

Measurement range
Signal Output CAN bus
Supply voltage, current consumption
Connections

4 to 20mA current loop for PLC controllers - The current is proportional tothe irradiance.
24 VDC +-10% (down to 12V possible if compliance voltage and loop
resistance is considered)=signal out
cable version: Iout=brown, V+=white, shield=black
2 m cable length, other lengths available (max.20 m)
plug version: Iout=1(brown), V+=4(black)3 orders of magnitude
8,5 V
645 Ohm @ 24V and 145 Ohm @12V
4 mA +- 0,01 mA
USB output with USB-A (to computer) or µUSB connector (to smartphone)
5V (USB powered)< 17 mA
USB2.0-A connector (to computer, free software "UVPLOT" is available)or USB2.0-micro-B connector (to a smartphone device like the Radiometer SXL55) 2m cable length.
4 orders of magnitude
CAN Bus with VSCP protocol for integration into a bus system or to be
used with the sglux UVTOUCH or the sglux Digibox
5 to 24 V +- 10%
8-pin M16 x 0.75 connector: Pins 1&7 = CAN low, Pins 3&8 = CAN high,Pin 6=V+, Pins 2&4&5 = GND, 2m cable length, other lengths available

Measurement range

4 orders of magnitude

Available displays and converters

UVTOUCH and Digibox

Signal Out MOD bus
Supply voltage, current consumption
Connections

MOD bus RTU over RS-485 (connection parameters programmable)
5 to 24V +-10%, typ. 20mA, max. 25mA
5-pin M12 connector at sensor side and Binder cable M12-A Series 763
with open wires, Shield =1 (shield), V+ = 2 (red), GND = 3 (black), B = 4(white), A = 5 (blue)

sglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.de
Rev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.

UV Sensor "UV-Cure"UV sensor for high UV irradiance (>100mW/cm2)
4/4

Field of  View

Drawing (values in mm)
77.5

ANALOG CABLE

53
9.538

Ø10

Ø22

Ø22

wrenchwidth 19

window view
66

ANALOG PLUG

53
9.538

Ø10

Ø22

Ø22

wrenchwidth 190.8
window view
212453connector view
5 pin M 12 x 1
RSFM5

plug connection
5 pin M 12 x 1e.g. Lumberg PRSFM 5487.5

DIGITAL

6348

9.50.8

CANØ10Ø22

Ø22wrenchwidth 19

window viewUSB

M 16 x 0.75
KFV 80 plug

78631542pin layout

USB Type A

Micro USBsglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.deRev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.

Sensor Probes Overview
laboratory & experiments
UV-SurfaceUniversal radiometric UV sensor
for calibration and reference
measurements, cosine correction.
Often used with radiometer SXL55.
UV-Cosine
Waterproof dirt repellent UV sensor
for outdoor measurement, cosine
field of view. Also available as UVI
sensor (ERYCA), M20x1,5 thread
UV-AirAxial measuring screw-in UV sensor
very good EMC properties, M22x1.5thread,
TOCON-Probe
Miniature UV sensor with0 to 5 V voltage output,
M12x1 thread

special applications
UV-ArcWaterproof UV sensor for measurement ofelectric arcs between overhead contact wiresand pantograph, complies with EN 50317,
G3/4" threadsglux ERYCA
highaccuracy
UV-Indexsensor,
measurement uncertainty is < 5%. Thesensor complies with ISO 17166, M20x1,5
threadUVI-Solo
like sglux ERYCA but configured as aready-to-mount system (avaliable for pole orrailings assembly)

UV-Wirelesswireless UV sensor with a display unit for
intensity and dose measurement

DUty Sensors monitoring
UV disinfection of Air,surfaces and water
UV-Sanitize
UV sensor for monitoring of air and surface UV
disinfection systems, configurable formonitoring of Hg low pressure lamps, excimer
lamps or xenon flash lamps, M20x1,5 thread
UV-Water-G3/4
UV sensor for operation in pressurized water(10 bar), for Hg medium and low pressure
lamps. Also available as UV-Water-G1/4,
G3/4" thread
UV-Water-PTFE
PTFE UV sensor for operation in pressurizedwater (10 bar), only for Hg low pressure
lamps or LEDs, G1/4" thread

UV-ÖNORM / UV-DVGW
UV sensor for DVGW(160°) and ÖNORMcertified water purifiers, also avaliable asUV-DVGW (40°). The sensors comply withÖNORM M5873, DVGW W294(06), DIN19294
UV-Radial
Waterproof side looking UV sensor formonitoring of lamp bundles, for operation ina cladding tube or directly in water, M20x1,5thread

HIGH UV radiation
UV-CureUV sensor for high irradiance (>100mW/cm2) for LED curing or cooled
medium pressure lamps, M22x1,5thread (temperature sensor available).
UV-Cure_HT
Like UV-Cure but for temperatures up to
170°C, e.g. for uncooled medium pressure
systems, M22x1,5 threadsglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.deRev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.