KP3000 SERIESDIGITAL PROGRAM SETTERThe KP3000 series is a 96x96mm digital program
setter with the analog output accuracy of ±0.1% andmaximum 30 program patterns (maximum 19 steps/pattern).
Output signal can be specified from analog output typeand digital output type. By combination with a digitalindication controller with digital input, the configurationof low cost program control system is enabled.

■ FEATURES■ MODELS●Program pattern
Settings of maximum 19 steps per pattern and maximum
30 sets of patterns are enabled. Repeating of a whole program pattern, linking of program patterns and repeating ofa specific step in a program pattern are enabled, too.

KP3−□0C□□□−□□□
Output signal
1: Digital output (RS422A)
2: Analog output (4 to 20mA)
4: Analog output (0 to 10V)
5: Analog output (0 to 1V)
6: Analog output (Others)
7: Digital output (RS485)
1st zone*
0: NoneP: 6 Digital inputs
T: 6 Digital outputs
2nd zone*
0: NoneP: 6 Digital inputs*1
T: 6 Digital outputs*1
3rd zone*
0: None
R: Communications 1 port (RS232C)+ 3 Digital inputs*2
A: Communications 1 port (RS422A)+ 1 Digital input*3
S: Communications 1 port (RS485)+ 3 Digital input*2
B: Communications 2 ports (RS232C + RS232C)+ 1 Digital input*4
C: Communications 2 ports (RS232C + RS422A)+ 1 Digital input*4
D: Communications 2 ports (RS232C + RS485)+ 1 Digital input*4
E: Communications 2 ports (RS485 + RS232C)+ 1 Digital input*4
F: Communications 2 ports (RS485 + RS422A)+ 1 Digital input*4
G: Communications 2 ports (RS485 + RS485)+ 1 Digital input*4
P: 6 Digital inputs*4
T: 6 Digital outputs*4
U: 8 Digital inputs*4
W: 8 Digital outputs*4
Y: 3 Digital inputs+ 5 Digital outputs*4
Z: 4 Digital inputs+ 4 Digital outputs*4
Case color
G: Gray
B: Black*
Panel sealing and terminal cover*
0: None1: Terminal cover
2: IP54 panel sealing
3: IP54 panel sealing + Terminal cover
Power supply voltage
A: 100 to 240V (AC)D: 24VAC/24VDC●Analog output type and digital output typeOutput of setting unit is selected from high accuracy(0.1%FS) analog output type and digital output type which
has no setting error by using communications function.
●Communications 2-port type provided
Models with 2 communications ports are available. In addition, speeding up and highly-functionalization of communications have been realized. For example, you can use 1
port for high order communications with a personal computer and another port for the communications remote (digitalremote) function. The communications protocol can be arbitrarily selected from [MODBUS] and [PRIVATE].
In the digital output type, however, 1-port type of communications is only available.
●DI/DO arbitrarily-allocation
When the digital input (DI) or the digital output (DO) is added, arbitrarily-allocation for assigning functions to those
DI/DO's is enabled. It is the function enabling allocations
such as [External drive input] to DI1 to DI3 and [Pattern selecting input] to DI4 to DI6.
●Conforming to international safety standards and European directives (CE)The controller is conformity with European directives (CE),and is UL and c-UL approved.
●Conforming to RoHS
The controller is an environmental consideration productwhich does not contain directed hazardous substancessuch as lead, etc.

* Option*1 It can be selected when the control signal is 1 or 7 only.
*2 The digital input is 1 point for the control signal of 1 or 7.
*3 It can be selected when the control signal is 2, 4, 5, 6 or 7 only.
*4 It can be selected when the control signal is 2, 4, 5 or 6 only.
Note: For options common to 1st zone, 2nd zone and 3rd zone, assign them inorder of [P] and [T] from 3rd zone first

1PSE-360B

KP3000 SERIES■ SCREENS
12
315

4
5
616
17
718

8910

Upper display

11121314

Function keys

1.

Operation status (RUN) indication
Lights in operation.

2.

Operation stop (STOP) indication
Lights in the state of operation stop.

3.

RESET indication
Lights when operation is cancelled and returns tothe start point.

4.

Pattern No. (PTN) indication

5.

Program remote (REM) indication
Lights when operation is executed by digital input.

6.

Executing step number (STP) indication
The step No. being executed is indicated.

7.

Function (FNC) operation indication
Lights when the function key is operated.

8.

9.

10.

11.

12.

15. Set value (SV) indication
16. Time signal (TS1 to TS8) indication

13.

14.

FNC keyWith the operation screen displayed, pressing it puts the controller in theoperation key mode. With the settings screen displayed, pressing it puts
the controller in the setting key mode and it operates to move the cursor
backwards.
RUN keyIn the operation key mode, it operates as Run key. With the settings
screen displayed, pressing it puts the controller in the setting key mode
and it is used for switching between the operation screen and the mode
screen of Mode 0, or for switching from the settings screen to the mode
screen.
STOP key
In the operation key mode, it operates as Stop key. With the settings
screen displayed, pressing it puts the controller in the setting key mode
and it is used to switch the settings screen.
ADV keyIn the operation key mode, it operates as Advance key. With the settings
screen displayed, pressing it puts the controller in the setting key mode
and it is used for moving the cursor and for selecting a parameter.
RESET key
In the operation key mode, it operates as Reset key. With the settings
screen displayed, pressing it puts the controller in the setting key mode
and it is used for changing a setting value (or selecting a parameter) indescending order.
PTN keyIn the operation key mode, it operates as Pattern key. With the settings
screen displayed, pressing it puts the controller in the setting key mode
and it is used for changing a setting value (or selecting a parameter) inascending order.
ENT key
It is used for registering the settings.

18. Engineering port

Lower display
17. A wide variety of operation screens are prepared and arbitrarily-selection is enabled.
On the whole program pattern display screen, the simultaneous display of the shape of whole program pattern and the progressed
pattern position has been realized.

Time screen

Pattern screen

2

■ OUTPUT SPECIFICATIONS■ SAFTY STANDARD

Output signal:

CE directives:

Analog output 4 to 20mA, 0 to 1V, 0 to 10V
Digital output RS422A, RS485
Accuracy rating: ±0.1% of full scale
Output updating cycle:
Analog output Approximately 0.1 seconds
Digital output Approximately 1 second
Resolution:
Approximately 1/30000
Output impedance:Voltage output Approximately 10Ω
Load resistance: Current output 400Ω or less
Voltage output 50kΩ or more

UL:
c–UL:

■ REFERENCE OPERATION CONDITIONS■ DISPLAY SPECIFICATIONSUpper display:
Lower display:

EN61326: 1997 +A1+A2+A3
EN61010-1: 2001 (Overvoltage category II, pollution
degree 2)* Under the test conditions of EMC directives, there may bevariation of indication value or output value which isequivalent to maximum ±10% or maximum 2mV, whicheveris greater.
UL61010-1 2nd edition
CAN/CSA C22.2 No.61010-1-04

Ambient temperature:
23°C ± 2°C
Ambient humidity: 55%RH ± 5% (No condensation)
Power voltage:
General power supply specifications
100VAC ±1%
24V power supply specifications
24VDC ±1%
Power supply frequency:
General power supply specifications50/60Hz ±0.5%
24V power supply specifications
DCMounting angle: Forward or backward ±3°, lateral ±3°
Installation height: Altitude 2000m or below
Vibration:
0m/s2
Impact:0m/s2
Mounting condition:
Single-unit panel mounting (Space above, below, right andleft of unit is needed.)
Wind:
None
External noise:
None
Warm up time:
30 minutes or longer

LED
LCD (with back light) 108 x 24 dots

■ GENERAL SPECIFICATIONS
No. of program patterns:
30 patterns
Pattern repetition … Max.9999 times
No. of program step:
19 step/pattern
Step repetition … Max.99 times
Rated power voltage:
General power supply specifications 100 to 240VAC
24V Power supply specifications 24VAC/24VDC
Rated power supply frequency:
General power supply specifications 50/60Hz
24V Power supply specification 50/60Hz (24VAC)
Maximum power consumption:
General power supply specifications
Without options 100VAC 10VA
240VAC 15VA
With options
100VAC 15VA
240VAC 20VA
24V Power supply specifications
Without options 24VAC 10VA
24VDC 5W
With options
24VAC 15VA
24VDC 10W
Power failure countermeasures:
Settings stored in EEPROM (Rewrite count: One milliontimes or less) and stored by a lithium battery for 5 years ormoreTerminal screws: M3.5Insulation resistance:
Between primary terminals and secondary terminals
20MΩ or more (500VDC)
Between primary terminals and protective conductor
terminal
20MΩ or more (500VDC)
Between secondary terminals and protective conductor
terminal
20MΩ or more (500VDC)
Withstand voltage: Between primary terminals and secondary terminals
1500VAC (For 1 minute)
Between primary terminals and protective conductor
terminal
1500VAC (For 1 minute)
Between secondary terminals and protective conductor
terminal
500VAC (For 1 minute)*Primary terminal: Terminals for power supply (100to 240VAC)
Casing:Fire-retardant polycarbonate
Color:
Gray or blackMounting:
Panel mounting
External dimensions:
96 (H) x 96 (W) x 127 (D) mm*The depth from the front panel is 120mm.
Weight:Without options Approximately 450g
With options Approximately 580g

■ NORMAL OPERATION CONDITIONS
Ambient temperature:-10°C to 50°C (-10°C to 40°C for closed installation)
Ambient humidity: 10 to 90%RH (no condensation)
Power voltage:
General power supply specifications 90 to 264VAC
24V Power supply specifications 21.6 to 26.4VDC/AC
Power supply frequency:
General power supply specifications 50/60Hz ± 2%
24V Power supply specifications DC, 50/60Hz ± 2%
Mounting angle: Forward or backward ±10°, lateral ±10°Installation height: Altitude 2000m or below
Vibration:
2m/s2
Shock:
0m/s2
Mounting condition:
Single-unit panel mounting (Space above and below of theunit is needed.)External noise:
None
Rate of ambient temperature change:
10°C/hour or less

■ TRANSPORT CONDITIONS
Ambient temperature:-20°C to 60°C
Ambient humidity: 5 to 90%RH (no condensation)
Vibration:4.9m/s2 (10 to 60Hz)
Impact:392m/s2
Under the condition that the unit is packed for shipment by
the factory■ STORAGE CONDITIONS
Ambient temperature:-20°C to 60°C
For long term storage, the temperature should be 10°C to30°C.
Ambient humidity: 5 to 90%RH (no condensation)
Vibration:
0m/s2
Impact:0m/s2
Under the condition that the unit is packed for shipment bythe factory3

KP3000 SERIES■ OPTIONS●Communications interface
With RS232C, RS422A or RS485, the setting and measured values of thecontroller can be transmitted to a master CPU and various parameters can be
set by the master CPU.
Number of communications points:
2 points
Communications type: RS232C, RS422A, RS485
Communication speed: 2400/4800/9600/19200/38400 bps
Protocol:MODBUS (RTU), MODBUS (ASCII), PRIVATE●Digital signal input
The following switching is enabled by digital input signal.
Input signal:
No-voltage contact, open-collector signal
External contact capacity:
5VDC 2mA
Functions:
1. Selection of pattern No. (6 points)
2. Run/stop
3. Advance
4. Reset
5. Wait
6. Fast●Digital signal output
Time signal or status signal can be outputted externally open-collector signal.
Output signal:
Open-collector signal
Capacity:
24VDC, Maximum 50mA
Functions:
1. Time signal (Maximum 8 points)
2. Run/stop
3. Advance
4. Reset
5. Wait
6. End●Panel sealing
By mounting the controller to a panel, it has the panel sealing equivalent to[IP54 compliance].

●Terminal cover
It covers the terminals for safe. The cover is transparent.

4

■ TERMINAL ARRANGEMENT
Digital output

Analog output(Voltage/current)

Option terminals

Communications
RS422A

Communications
RS485

112192635192132027362014212837＋21SG

315222938−

22RDASA

2941623303923RDB

SB30

517243140

24SDASG3161825324125SDB

334233344334710911826272832100-240V AC
24V AC/DC●Digital output + Communications
Communications RS232C＋
Digital output RS422A＋
1 digital input

Communications RS485＋
Digital output RS422A＋
1 digital input

Communications RS232C＋
Digital output RS485＋
1 digital input

Communications RS485＋
Digital output RS485＋
1 digital input

Digital output RS485＋
Communications RS422A＋
1 digital input

26RDSARDSASA27SDSB

SD

SB

SB

28

SG

SG

SG

SG

SG

29

RDA

RDA

SA

SA

RDA

30

RDB

RDB

SB

SB

RDB

31

SDA

SDA

SG

SG

SDA

32

SDB

SDB

33

DI

DI

DI

DI

DI

34

COM

COM

COM

COM

COM

SDB

●Option terminals
PT

1st2nd

3rd zone

DI

DO

1219

26

DI

DO

13

20

27

DI

DO

1421

28

DI

DO

15

22

29

DI

DO

1623

30

DI

DO

1724

31

COM

COM

1825

32

Options common to each zone

P: 6 digital inputs

33

T: 6 digital output

Based on combination with other options,assign the zone in the above order.

34

3rd zone
Terminals

R

A

S

B

C

DE

F

GUWYZ

26

RDRDA

SA

RD1

RD1

RD1

SA1

SA1

SA1

DI

DO

DO

DO

27

SD

SDB

SB

SD1

SD1

SD1

SB1

SB1

SB1

DI

DO

DO

DO

28

SG

SDA

SG

SG1

SG1

SG1

SG1

SG1

SG1

DI

DO

DO

DO

29

DI

SDB

DI

RD2

RDA2

SA2

RD2

RDA2

SA2

DI

DO

DO

DO

30

DI

SG

DI

SD2

RDB2

SB2

SD2

RDB2

SB2

DI

DO

DO

DI

31

DI

DI

DI

SG2

SDA2

SG2

SG2

SDA2

SG2

DI

DO

DI

DI

32

COM

COM

COM

DI

DO

DI

DI

SDB2

SDB2

33

DI

DI

DI

DI

DI

DI

DI

DO

DI

DI

34

COM

COM

COM

COM

COM

COM

COM

COM

COM

COM

R: Communications RS232C + 3 Digital inputs
A: Communications RS422A + 1 Digital input
S: Communications RS485 + 3 Digital inputs
B: Communications RS232C + Communications RS232C+ 1 Digital input
C: Communications RS232C + Communications RS422A+ 1 Digital input

D: Communications RS232C + Communications RS485+ 1 Digital input
E: Communications RS485 + Communications RS232C+ 1 Digital input
F: Communications RS485 + Communications RS422A+ 1 Digital input

5

G: Communications R485 + Communications RS485+ 1 Digital input
U: 8 Digital inputs
W: 8 Digital outputs
Y: 3 Digital inputs + 5 Digital outputs
Z: 4 Digital inputs + 4 Digital outputs

●ABOUT CRIMP STYLE TERMINALS
7 or lessφ3.7 or less7 or less
φ3.7 or less●Ring type

●Spade type7 or less7 or less(in pressed condition)(in pressed condition)*Use terminal with insulation■EXTERNAL DIMENSIONS

796

7120

115

91×91

88×88

Terminal cover

115

96

91×91

Mounting metal

120147●Closed mounting panel dimensions●PANEL CUTOUT

92＋0.8092＋0.80120

120

96×Nー4 ＋20
N: Number of mounted instruments

Unit: mm92＋0.80Specifications subject to change without notice. Printed in Japan (I) 2018. 832-8 KUMANO-CHO,ITABASHI-KU,TOKYO 173-8632
Telephone : +81-3-3956-2171
Facsimile : +81-3-3956-0915
E-mail : inter@chino.co.jp
Website : www.chino.co.jp/$