X-Y Force Type Load Cells

X-Y Series

Ideal for tire uniformity measurement, etc.
Simultaneous measurement in the X and Y directions with one unit
Structure designed for excellent toughness and durability
Minimal mutual interference with simultaneous loading on the X and Y axes
Models & Capacities
Model No.
XY-500L/250L
XY-750L/250L
XY-1000L/500L

Capacity
X-500 kg (5 kN)
Y-250 kg (2.5 kN)
X-750 kg (7.5 kN)
Y-250 kg (2.5 kN)
X-1000 kg (10 kN)
Y-500 kg (5 kN)

Note: XY-500L/500L to be specially designed and re-quoted

Specifications
Rated capacity (R.C.)

X-Y Load Cells Dimensions〜42105〜30 25.4

□32

Unit = mm

28.6
1.6

Y-Load direction

15.75

ø19
ø63.5(70)

8.15

3.5

6.35

.D
P.C

Y

X-Load direction

ø143

6.35

X

WiringInformation

25.5
6.35

69.85

ø177.845°

15X-500 kg Y-250 kg (X:5 kN, Y:2.5 kN )X-750 kg Y-250 kg (X:7.5 kN, Y:2.5 kN)
X-1000 kg Y-500 kg (X:10 kN, Y:5 kN)
Rated output (R.O.)
2 mV/V ±1% (for X, Y)
Non linearity
0.1% R.O.
Hysteresis
0.05% R.O.
Repeatability
0.05% R.O.
Recommended excitation voltage 5 V to 10 V AC/DCMax. excitation voltage
20 V AC/DC
Zero balance±1% R.O.
Input terminals bridge resistance 350 Ω ± 3.5 Ω
Output terminals bridge resistance 350 Ω ± 3.5 Ω
Insulation resistance2000 MΩ (B/E DC 50 V)
Cable diameter/length
ø7, 4-core shielded cable,
10 m, connector end
Zero point temperature effect
0.05% R.O./10 ˚CTemperature effect on output
0.05% Load./10 ˚CTemperature compensated range –10 ˚C to +70 ˚C
Temperature usable range–10 ˚C to +70 ˚C
Safe overload
150% R.L.
Limited overload
200% R.L.
Dustproof/Waterproof
IP54
Natural frequency
More than 1 kHzCrosstalk
2.5 N or less (5 kN / 2.5 kN)
5.0 N or less (7.5 kN /5 kN, 10 kN, /2.5 kN, 5 kN / 5 kN)

6.35

GREEN : EXC (+)
BLUE : EXC (–)
RED SIG : (+)
WHITE : SIG (–)
YELLOW : GND

Appearance and/or specifications subject to change for improvement without notice.
…Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku, Tokyo 170-0013 JAPANTelephone:［81］(3) 5391-6132 Fax:［81］(3) 5391-6148
http://www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:［1］(408) 263-5333 Fax:
［1］(408) 263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:［61］(8) 8301-8100 Fax:［61］(8) 8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:［44］(1235) 550420 Fax:［44］(1235) 550485

A&D Korea LimitedManhattan Bldg. 8F, 36-2 Yoido-dong, Youngdeungpo-gu,Seoul, KOREA
Telephone:［82］(2) 780-4101 Fax:［82］(2) 782-4280

A&D RUS Co., Ltd.

Vereyskaya str.17, Moscow, 121357 RUSSIA,
Telephone:［7］(495) 937-33-44 Fax:
［7］(495) 937-55-66

A&D Technology Trading (Shanghai) Co., Ltd.

Room 101, No.1 Fu Hai Business Building, No. 289,
Zhang Jiang Bi Sheng Road, Shanghai 201204, CHINA
Telephone:［86］(21) 3393-2340 Fax:
［86］(21) 3393-2347

A&D Europe GmbHIm Leuschnerpark 4, D-64347 Griesheim, GERMANY
Telephone:［49］(6155) 605-227 Fax:［49］(6155) 605-100

A&D Instruments India Private Limited

509 Udyog Vihar Phase V Gurgaon-122 016,Haryana, INDIA
Telephone:［91］(124) 471-5555 Fax:［91］(124) 471-5599

A&D Technology, Inc.

4622 Runway Blvd. Ann Arbor, MI 48108 U.S.A.
Telephone:［1］(734) 973-1111 Fax:
［1］(734) 973-1103✽ XY-ADCC-01-NA1-11103