HL-1000WP
HL-300WP
HL-3000WP
HL-3000LWP

1WMPD4000754C

COMPLIANCE

Compliance with FCC Rules
Please note that this equipment generates, uses and can radiate radio
frequency energy. This equipment has been tested and has been found tocomply with the limits of Class A digital devices pursuant to Part 15 of FCC
rules. These rules are designed to provide reasonable protection againstinterference when equipment is operated in a commercial environment. If thisunit is operated in a residential area, it may cause some interference andunder these circumstances the user would be required to take, at his own
expense, whatever measures are necessary to eliminate the interference.
(FCC = Federal Communications Commission in the U.S.A.)

SAFETY PRECAUTIONSAll safety messages are identified by the following, “WARNING” or“CAUTION”, of ANSI Z535.4 (American National Standard Institute:
Product Safety Signs and Labels). The meanings are as follows:
All safety messages are identified by the following, “WARNING” or“CAUTION”, of ANSI Z535.4 (American National Standard Institute:
Product Safety Signs and Labels). The meanings are as follows:
WARNINGCAUTION

A potentially hazardous situation which, if not avoided, could
result in death or serious injury.
A potentially hazardous situation which, if not avoided, may
result in minor or moderate injury.

 This manual is subject to change without notice at any time to improve the product.
 Product specifications are subject to change without any obligation on the part of themanufacturer.
 When using the HL-WP, the following safety precautions should always be followed.
WARNING Use only the specified AC adapter for the scale. Otheradapters may cause damage (The AC adapter is optional).
Internal service or adjustments to this product should beperformed by a qualified person.
CAUTION
Avoid installing the scale in direct sunlight, which may cause
discoloration or malfunctions.
Do not mix the type of batteries. Replace all the batteries at thesame time.
If the scale is not to be used for a long period of time, remove
all batteries from the battery compartment to avoid leakage.
Avoid overloading that could cause damage to the scale.
Avoid using the weighing pan to move the scale, as that could
cause damage to the scale.
When the scale is used with an AC adapter, the scale does notconform to IP-65 protection.
1

PARTS DESCRIPTION
Weighing Platform

Display

ON/OFF key

Turns the scale power on/off.

RE-ZERO key

Sets display to zero or
subtracts weight of container.

Spirit level

UNITS key

(U.S.A. models only)
Changes weighing units.

Four leveling feet
Rubber cap

Battery cover
AC adapter jackBattery cover lock(Batteries are not included.)Battery compartment

STABLE indicator

Indicates when the reading
is stable.

NET indicator

Indicates when NET weight
is displayed. (Tare functionis used.)

ZERO indicator

Indicates when the scale
zero is correct.
i t blPolarity

01200.0kg

Weighing unit, g
Weighing unit, lb

0-1.000klb

04-76.3koz

04l12.3koz

2Weighing unit, oz
U.S.A. models only

Weighing unit,lb-oz

PREPARATION
Installing Batteries / Connecting The AC adapter
Installing Batteries
Close the rubber cap
CAUTION
When replacing batteries, be careful
not to apply too much force to theweighing pan. An excessive force
could damage the scale.
Remove the battery cover and insert sixbatteries (R6P / LR6 / AA size) into thebattery compartment, taking extreme
care to that the polarities plus and minus
are observed. Be sure to lock thebattery cover again after installing thebatteries.
New six batteries (R6P/ LR6/ AA size)

Battery cover lock

Optional AC Adapter

AC adapter (optional)
CAUTION
When using an AC adapter, the scaledoes not comply with the IP65.
The AC input requirement could be 100,120, 220, 230 or 240 volts (50/60Hz)
depending on the area where used, so
please verify that the adapter is correct.
Plug the AC adapter to the AC adapter
socket inside the battery compartment.
Attach the battery cover and fit therubber cap.

AC Adapter jackSetting up the HL-WP

CAUTION
Avoid installing the scale in direct sunlight, as that may cause discoloration
or a malfunction. Place your HL-WP on a firm weighing table so that thescale is level. (The scale will not perform accurately when it is not level.)
Place the scale on the firm surface and adjust the feet so that the sprit levelshows the scale is level for an accurate weighing.
3

Before weighing

WEIGHING

Automatic Power-Off Function

If the scale is left ON and the stable indicator is displayed, the automaticpower-off function turns the power off after approximately 5 minutes(factory setting). To toggle the function disabled/enabled, press the[ON/OFF] key while pressing the [RE-ZERO] key. The scale stores thesetting and it will be shown by pressing both the [RE-ZERO] and[UNITS] keys in the weighing mode. (See “CALIBRATION” about the[UNITS] key.)
Function disabled.

Error message
ee eeeaaFunction enabled.

OverloadThe scale zero is out of range
Low battery

Calibration may be required before weighing

Read “CALIBRATION” first and if necessary, calibrate your scale for
accurate weighing.

Weighing procedures
1. Turn on the scale.

Press the [ON/OFF] key to turn on the scale. Then the scale displays all
segments for a few seconds and thenwill be displayed.

2. Start weighing

If you do not use a container for weighing,
Verify the reading is. If not, press the [RE-ZERO] key to display. Place objects on the weighing platform to weigh. When the reading
becomes stable, the stable indicator is displayed.
If you use a container for weighing,
Place an empty container on the platform.
Wait for the stability indicator to be displayed and press the [RE-ZERO]
key. Place the objects to be weighed in the container.
Select the weighing unit, --- Only for U.S.A. models --If you use the U.S.A. model, press the [UNITS] key to select weighing unit.
Normally scales power up in “lb” or “oz” when shipped from the factory. To
power up in other units, turn the scale off. Press and hold the [UNITS] key,then press the [ON/OFF].key The scale will start with the next unit.
Note) HL-300WP: ozgoz
HL-1000WP/3000WP/3000LWP: lbozlb-ozglb
4

CALIBRATION

When to calibrate

Calibration may be required when the scale initially installed, if it is
moved a substantial distance, or in accordance with local regulations.
This is necessary because the weight of a mass in one location is notnecessarily the same in another location. Also, with time and use,mechanical deviations can occur.

The position of the [UNITS] key for the Calibration
The “g” model has no[UNITS] key, but the key
used to enter calibration islocated under the overlay.

Location of the [UNITS] key

Calibration using a calibration weightCalibrate the scale with calibration weights at full capacity.
HL-1000WP :1000g ± 0.1g
HL-300WP:
300g ± 0.02g
HL-3000WP :3000g ± 0.2g
HL-3000LWP :3000g ± 0.2g

1. Enter calibration mode.

Press the [ON/OFF] key to turn the power off.
While pressing both the [RE-ZERO] key and the [UNITS] key, press the[ON/OFF] key.
will be displayed.

2. Zero calibration

Press the [RE-ZERO] key to calibrate zero.
will be displayed.
Wait for the stable indicator to be displayed and press the [RE-ZERO] key.
will be displayed in a few seconds. If only the zero calibration is
to be done, press the [UNITS] key. The scale will display
and automatically return to the weighing mode.

3. Span calibration

Place the weight at the center of the platform for accurate calibration. Wait
for the stable indicator to be displayed and then press the [RE-ZERO] key.
The display will show
and the scale will automatically return to theweighing mode.

5Gravity compensation
If you have calibration weights at the capacity of the scale, you can calibrate
the scale with the weights instead of compensating for acceleration due togravity.
The scale was calibrated at 9.798 m/sec2 , at the factory. If the acceleration dueto gravity at your location is not same, compensate the scale by adjusting theacceleration due to gravity. (Refer “The Value of Gravity at Various Locations”
and “World Map” and find your location value.)

1. Enter calibration mode.

Press the [ON/OFF] key to turn the power off. While pressing both the[RE-ZERO] key and the [UNITS] key, press the [ON/OFF] key.
will be displayed.

2. Select a digit to be changed.

Press the [UNITS] key. The previously set value, for example, willbe displayed. A digit after the comma “,” will be changed by pressing the[RE-ZERO] key. Press the [UNITS] key to move the comma “,” and selecta digit to be changed.

3. Set the new value

Change the value of the gravity acceleration using the [RE-ZERO] and[UNITS] keys.

4. Save the value in memory.

Press and hold the [UNITS] key until
automatically return to the weighing mode.

6is displayed. The scale willSPECIFICATIONSMODELCapacityxResolution

HL-1000WP
HL-300WP
HL-3000WP
HL-3000LWP
1000 x 0.5 g
300 x 0.1 g
3000 x 1 g
2.2 x 0.001 lb
6.6 x 0.002 lb
35.28 x 0.02 oz 10.58 x 0.01 oz
105.8 x 0.1 oz
2 lb 3.3 x 0.1 oz
6 lb 9.8 x 0.1 oz
Non-linearity
±1g
±0.2g
±2g
Repeatability
0.5g
0.1g
1g
Span drift±0.015% / °C TYP (5°C~35°C / 41°F~95°F)
Operating temp.
-10°C~40°C / 14°F~104°F, Less than 85%RH
Display
13 mm / 0.51 inches, 7segment liquid crystal display
Display update
Approximately 10 times per secondPower
6 x R6P / LR6 / “AA” size batteries or AC adapter
Battery life
Approximately 100 hours with manganese type
200 hours with alkaline cells at 20°C / 68°F
Platform size
128 (W) x 128 (D) mm
174(W)x137(D) mm
5.0 (W) x 5.0 (D) in.
6.8(W)x5.4(D) in.
Dimensions
170 (W) x 219 (D) x 63 (H) mm
174(W)x219(D)x63(H) mm
6.8(W)x8.6(D)x2.5(H) in.
6.7 (W) x 8.6 (D) x 2.5 (H) in.
Weight (approx.)
870 g / 1.9 lb
950 g / 2.1 lb
Calibration weight 1000g ± 0.1g
300g ± 0.02g
3000g±0.2g
Accessories
This manualOptions
AC adapter

The Value of Gravity at Various Locations
Amsterdam
AthensAuckland NZ
BangkokBirmingham
Brussels
Buenos Aires
Calcutta
Cape Town
ChicagoCopenhagen
CyprusDjakarta
Frankfurt
Glasgow
HavanaHelsinki
KuwaitLisbonLondon (Greenwich)
Los Angeles
Madrid

9.813 m/s2
9.807 m/s2
9.799 m/s2
9.783 m/s2
9.813 m/s2
9.811 m/s2
9.797 m/s2
9.788 m/s2
9.796 m/s2
9.803 m/s2
9.815 m/s2
9.797 m/s2
9.781 m/s2
9.810 m/s2
9.816 m/s2
9.788 m/s2
9.819 m/s2
9.793 m/s2
9.801 m/s2
9.812 m/s2
9.796 m/s2
9.800 m/s2

ManilaMelbourne
Mexico CityMilan
New York
Oslo
OttawaParis
Rio de Janeiro
Rome
San Francisco
Singapore
Stockholm
SydneyTaichung
TaiwanTaipeiTokyo
Vancouver, BC
Washington DC
Wellington NZ
Zurich79.784 m/s2
9.800 m/s2
9.779 m/s2
9.806 m/s2
9.802 m/s2
9.819 m/s2
9.806 m/s2
9.809 m/s2
9.788 m/s2
9.803 m/s2
9.800 m/s2
9.781 m/s2
9.818 m/s2
9.797 m/s2
9.789 m/s2
9.788 m/s2
9.790 m/s2
9.798 m/s2
9.809 m/s2
9.801 m/s2
9.803 m/s2
9.807 m/s2

World Map

8

Back cover: A&D Global

A&D Company, Limited

3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-1566

A&D ENGINEERING, INC.

47747 Warm Springs Blvd, Fremont, California 94539, U.S.A.
Tel: [1] (800) 726-3364
Weighing Support:[1] (888) 726-5931

Inspection Support:[1] (855) 332-8815

A&D INSTRUMENTS LIMITED

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

A&D AUSTRALASIA PTY LTD

32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 07331( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4264

Почтовый адрес:121357, Российская Федерация, г.Москва, ул. Верейская, дом 17Юридический адрес: 117545, Российская Федерация, г. Москва, ул. Дорожная, д.3, корп.6, комн. 8б
( 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Instruments India Private Limited
D-48( D-48, Udyog Vihar, Phase- , Gurgaon - 122016, Haryana, India ): [91] (124) 4715555: [91] (124) 4715599

A&D SCIENTECH TAIWAN LIMITED.A&D台灣分公司 艾安得股份有限公司
台湾台北市中正區青島東路 5 號 4 樓
( 4F No.5 Ching Tao East Road, Taipei Taiwan R.O.C. )Tel : [886](02) 2322-4722
Fax : [886](02) 2392-1794

A&D INSTRUMENTS (THAILAND) LIMITEDบริษทั เอ แอนด์ ดี อินสทรูเม้นท์ (ไทยแลนด์) จำกัด168/16 หมูท่ ี่ 1 ตำบลรังสิต อำเภอธัญบุรี จังหวัดปทุมธานี 12110 ประเทศไทย
( 168/16 Moo 1, Rangsit, Thanyaburi, Pathumthani 12110 Thailand )Tel : [66] 20038911

9