Advanced-Level Analytical/Precision Balances

A&D Apollo

GX-AE /GX-A /GF-A Series

GF-3002A

GX-324AE

GX-1003A

Taking Weighing Precision
and Protection to New Heights

www.aandd.jp

Simply Created for Practical Purposes
All the innovative technologies of A&D Apollo are designed to provide functional
solutions for real world problems that users face in laboratories or in factories.
Now discover what they are, and what precision, protection, performance andproductivity really mean to you!

Smart SHS®
Smart super hybrid sensor (Smart SHS) realizing wide lineup from analytical tohigh-resolution, high-capacity precision models
Select a balance that best suits your applications and budgets from either the GX-A series of 16
models with internal calibration function (three 0.0001 g readability models, six 0.001 g
readability models, ﬁve 0.01 g readability models and two 0.1 g readability models) or the GF-A
series of 18 models without internal calibration function (three 0.0001 g readability models,seven 0.001 g readability models, six 0.01 g readability models and two 0.1 g readability models).
In addition, the GX-AE series, which consists of three 0.0001 g readability models with internal
calibration function, has an ionizer embedded in its breeze break for quickly eliminating staticfrom the sample.

Analytical
GX-AE series (d = 0.0001 g)
Built-in fanless ionizer••

Adopting A&D's original "Quick
Ion" technology,◆ it removes
static at twice the speed of theconventional ionizer (approx.
0.3 seconds*1 when the target
object is within a distance of 10cm) without any disturbance.

••••••••••••••
••

••

••

••

••

••

••

••

••

••

••

••

••

••

••

••

Large and detachable
breeze break
The breeze break provides
suﬃcient room for various
kinds of containers/ﬁxturesand can be easily removed forcleaning as well as use in
conﬁned space.

•◆ Patent pending*

1 Subject to the ambient environment,
shape of the target object andamount of static charge.

•••••••••••

Retractable doors

••
••
••
••
••

Antistatic glass panes

The glass panes of the breeze break arecoated with transparent evaporated
metal to block static from outside.
1••••••

••

••

••

••

••

••

••

••

••

••

••

••

••

••

••

••

••

•

The balance requires no extra
space at the rear for accessing
the weighing chamber, as thedoors disappear into the back.

External IR switch
The GX-AE series comes standard witha touchless switch to activate theionizer for a set duration.

GX-A series (d = 0.0001 g)

GF-A series (d = 0.0001 g)

Precision
GX-A series (d = 0.001 g)

GF-A series (d = 0.001 g)

GX-A series (d = 0.01/0.1 g)

GF-A series (d = 0.01/0.1 g)

Fast stabilization
A combination of a high-stiﬀness, Roberval-structure spring material and an electromagnetic
force restoration mechanism simultaneously realizes fast stabilization and high resolution foreﬃcient, precise weighing. Typical stabilization time is from 1.5 to 2 seconds with the 0.0001 g
readability models and 1 second with the 0.001/0.01/0.1 g readability models.*2*2 Excludes the 1,620 g × 0.001 g and 10,200 g × 0.01 g models, with a typical stabilization time of 1.5 seconds.
Long-term durability and low cost of ownership
A&D's internal experiment veriﬁed that Smart SHS withstood a durability test of 10 million
repeated loadings without issue. With proper usage and care it can maintain a long andfunctional life. In case damage is caused, only the defective parts, not the whole unit, need to be
replaced, incurring a minimal cost.
2

Accuracy and precision management
Internal weight (for the GX-AE/GX-A series)
Automatic self calibration (ASC)To ensure accuracy all the time, the balance can be set to calibrate itself automatically using its
internal weight either (1) in response to change in ambient temperature to prevent error due tosensitivity drift, (2) at a set interval time, or (3) at predetermined (up to three) times of the day.
Internal calibration can also be performed any time with one key press.

Internal weight value correction
In cases such as where the internal weight value varied over time, it can be corrected either byloading an external weight whose value the balance uses as a reference or by manually inputting
a correction value.

ECL
Electronically controlled load (ECL) that enables precision management without
using an external weight
ECL is A&D's unique technology of generating a minute load (0.3 to 3% of the balance capacity) by
purposely altering the equilibrium state of the electromagnetic force restoration sensor.♦ Equippedwith this ECL, A&D Apollo can automatically assess its precision and provide a number of valuable
indexes. The entire process takes only 1.5 minutes and can be readily included in a daily check SOP.
◆ Patented

Auto precision assessment (APA)
Dia-Check

The balance ﬁrst makes quick diagnoses on whether there is a critical
failure to give users peace of mind to proceed with daily weighing
routines.
Dia-Check

Standard Deviation
Following Dia-Check, the balance goes on to perform 10 repeatedmeasurements with ECL to calculate a standard deviation,*3 allowing users
to quickly assess the balance repeatability under any given environment.
Standard Deviation

QuickMin-S

QuickMin-S

Based on the standard deviation obtained through ECL, the balance
quickly indicates the minimum weight*4 by a tolerance of either 0.1% inaccordance with the United States Pharmacopeia (USP) Chapter 41, or 1%,
depending on the laboratory requirements.
unit of measure will be mg only.
*34 The
The unit of measure will be g only*

Adjustable response characteristics
Depending on the APA results above, you can trade weighing speed for higher display stability*5
or vice versa by selecting one of three preset response characteristics (FAST, MID or SLOW).

*5 This is not to improve repeatability.
3

Features eﬀective for speciﬁc applications
Use in production lines
The weight sensor of a high-precision balance will
be damaged in a short period if it is continuously
aﬀected by loads with acceleration (impact shocks).
This typically occurs in factories where materials areplaced roughly by machines or thrown by operators
onto balances. In addition, impact shocks can easily
shift values and cause errors.
The diﬃculty is that even if you know the above, you
may not know what degree of impact shock is safe orunsafe for the balance to take preventive measures
as necessary.

ISD
Impact shock detection (ISD) providing feedback to prevent damage to the weight sensor
Impact-Alert
A&D Apollo visualizes the magnitudes of impact shocks that its Smart SHS receives at four levels. It beeps once at Level
3 and twice at Level 4 (both of which should be avoided).

No impact shock

Level 1

Level 2

Level 3

Level 4

Shock-Log
While the power is on, up to 50 impacts of Level 3 and Level 4 will be stored in the balance with timestamp (and log-in
user information when the password lock function is used). Data will then be overwritten in order from the weakest
impact. This function is especially beneﬁcial for quality managers as it enables them to check and show how thebalance had been treated when returned for repair.

Cushioning and overload protection mechanism
The two leaf springs (or one cruciform spring for the
0.0001 g readability models) that support the weighing
pan mitigate impact shocks signiﬁcantly. Meanwhile,
Smart SHS is well protected against vertical andtransverse static overloading. It won't break due tomalfunction of the weighing system's actuator and theresulting excess load. A&D's testing demonstrated thateven a human did no damage to the balance bystepping on it.

Leaf springs

4Flow rate measurement
Flow rate is determined from the weight variation that occurs during a given interval. Many
operators time this interval by using a stopwatch. Besides being troublesome and error-prone,
this method does not allow precise recording of ﬂow rates that vary from moment to moment
because the interval used to divide the weight variation (calculation time) is too long.
Alternatively, a PC can be connected to the balance to take the weighing data and perform thecalculation. This method enables measurement to be more responsive to ﬂow rate changes by
shortening the calculation time. However, due to the syncing problems that inevitably occur
between the internal clocks of the balance and the PC, the timing at which the balance receives
the command from the PC to send the weight value and its display refresh cycle gradually go outof alignment. In consequence, the PC intermittently produces irregular values even when theactual ﬂow rate is constant.
1 sec

2 sec

3 sec

4 sec

5 sec

6 sec

Balance
0

1

2

3

4

5

1

2

1 sec

PC

3

4

5

1

2 sec

2

3

4

5

1

3 sec

2

3

4

5

1

4 sec

2

3

4

5

1

2

3

4

5 sec

5

6
6 sec

Time disparity of two diﬀerent internal clocks (when the balance display refresh rate

is 5 times/sec, the calculation time is 1 sec, and the balance's internal clock goes slightly ahead of the PC's)

FRD
Flow rate display (FRD) for calculating ﬂow rate with the balance itself
FR-CalA&D Apollo can calculate and display/output the mass ﬂow rate (the unit of measure can be selected from g/s, g/m org/h), or the volume ﬂow rate (mL/s, mL/m or mL/h) by entering the density of the material—up to 10 densities can besaved to the device. Both the display refresh rate and calculation time (set either manually between 1 second and 1
hour or automatically according to the selected level of responsiveness to weight/ﬂow rate variation♦) are regulated by
the same internal clock of the balance, allowing measurements to be precisely performed without the time disparity
problem described above.
♦ Patent pending
0.45 g/s

Conventional method
FR-Cal

0.4 g/s
0.35 g/s

Flow rate measurement using FR-Cal as opposed to the conventional method

FR-Compare
The balance indicates whether the ﬂow rate is maintained within the designated limits (either 3 levels or 5 levels). The
results can be output using the optional GXA-04 interface.

WinCT-FRD (freeware)This software enables a PC to simultaneously display the weight and ﬂow rate values transmitted from the balance on
a two axis graph so that you can see their changes in real time. The graph can be printed out using a printer or saved
as an image ﬁle, whereas the recorded data can be saved in CSV or TXT format.

Tare memory
The tare value can be saved in non-volatile memory. When this function is activated, thepower-on re-zero/tare is turned oﬀ. Therefore, even if the power supply is cut oﬀ while
performing, for example, ﬂow rate measurement, the weight value displayed before the poweroﬀ will be displayed as the balance is turned on again.
5

Data management, documentation and compliance
User access control (UAC) and key lock to prevent misusage
The balance can be password-protected in two ways: The ﬁrst way is to limit the use toauthorized individuals (up to 11 including one administrator—the administrator can perform all
operations while other users are limited to measurements and calibration*6 only) by setting apassword for each user. The second way is to set a password just for the administrator andanyone else can use the balance without entering a password but for measurements andcalibration*6 only.
Moreover, upon receiving a command to disable its keys, the balance becomes operable only bysending commands from an external device such as a PC.

*6 The administrator can inhibit calibration also so that others can perform measurements only.
Automatic minimum weight calculation and implementation
You can enter the minimum weight into the balance by either direct key input, using the result of
QuickMin-S described earlier, or having the balance calculate the minimum weight from 10
repeated measurements of an actual, external weight (as with QuickMin-S, the tolerance can be
set to either 0.1% pursuant to USP Chapter 41 or 1%) . The stored value can always be called up to
see/change or output together with timestamp. When QuickMin-S or an external weight wasused, the calculation data is also output.
Min-S Alert
To ensure that the measured sample amount meets theminimum weight requirement, the balance can display analert until the sample amount reaches the value entered as theminimum weight.*7*7 When g is selected as the unit of measure only

Blinking minimum weight alert

Universal Flexi Coms (UFC)This function enables you to customize printout content and layout by editing and sending
commands (UFC setting commands) to the balance. Following the commands, the balance createsdata to be output to either a printer capable of dump printing such as the AD-8127 compact
printer or a commercially-available label printer*8 for barcode printing.
WinCT-UFC (freeware)*8 A label printer that supports ZPL/ZPL II

With this software, you can easily edit UFC setting commands on a PC and send them to a balance/scale that has a
UFC function, such as A&D Apollo. All that is required is to select data (e.g. weighing result, date, time, ID number,etc.), enter text and adjust the layout while checking the preview screen, which the software automatically converts
to UFC setting commands. For label printing, users can choose one of three kinds of barcodes; namely, QR code,
Data Matrix code and Code 39.

Gross/Net/Tare output
On receiving a command or with key operation, the balance can also output the gross, net, and tare values.

Statistical calculation function (SCF)
A&D Apollo can display/output statistical calculation data including the number of data, sum,maximum, minimum, range (maximum-minimum), average, standard deviation, coeﬃcient ofvariation and relative error to facilitate the analysis of measurements.
6

GLP/GMP/GCP/ISO compliant output
For documentation requirements, the balance manufacturer, model, serial number, ID number,date + time,*9 space for signature for calibration report, calibration test report, and title & end
blocks for a series of weighing results can be output.
the AD-8127 compact printer is used, it is possible to use its clock & calendar function instead of the balance's to print date +*9 Whentime. This allows you to prevent falsiﬁcation of the timestamp using the password lock function on the printer side as you prefer.

Data memory
A&D Apollo has a capacity to store up to 200 weighing results + 50 calibration results (all withtimestamp), and 50 unit masses to be used in counting mode for cases when no external memory
device is available. The stored weighing results or calibration results can be output to a printeror PC in one batch.

Other advanced features

RS-232C and USB interfaces as standard

Security slot

For the USB interface, you can toggle between the Quick
USB mode (plug-and-play with weighing data output toa PC only) and the Virtual COM mode (for bi-directional
communication*10) with internal settings. A USB cableis provided as standard.

An oﬀ-the-shelf (such as Kensington)anti-theft lock can be used to preventthe balance from being lost.

special driver downloaded from the A&D website needs to be*10 Ainstalledon the PC (except for those with Windows 10).

Reverse backlit LCD display
The contrast of black and white provides
excellent visibility even in poorly-lit areas
and also prevents eye fatigue.

7

Bright Bubble

Fly-Thumbwheel

The large, front-mounted spirit level is easy to see with
LED illumination, which blinks for 5 seconds after the
display is turned on to call for attention to make sure thebalance is level.

The leveling feet can be adjusted up or
down smoothly using large thumbwheels
set in higher position.

Options*11

|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|

2nd RS-232C interface
Comparator relay output/buzzer/external key input interface

GXA-03
GXA-04

Enables signaling check weighing results (3 or 5 levels) by buzzer and/or external comparator (traﬃc) lights.
It also has two jacks for separately-sold foot switches.

GXA-06

Analog output interface (factory-installed/dealer option)

FXi-08GXA-09Ethernet interface
Built-in rechargeable battery (factory-installed/dealer option)

GXA-10*12
GXA-12Large glass breeze break
Animal weighing pan (0.001/0.01/0.1 g readability models of 320 g capacity or higher)

GXA-13
GXA-14GXA-17*13

GXA-23-PLUG
GXA-24

Density determination kit for the 0.001 g readability models
Density determination kit for the 0.0001 g readability models
Large glass breeze break with built-in fanless ionizer andexternal IR switch
External key input interface + the AX-SW137-PRINT
foot switch
External key input interface + the AX-SW137-REZERO
foot switch
External key input interface + the AX-T-314A-S plug
USB host interface (factory-installed/dealer option)

GXA-25

Fanless ionizer of Quick Ion technology

GXA-26External IR switch

0-1 V, or 0.2-1 V for conversion to 4-20 mA.

10 hours of charging for 14 hours of operation (the remaining battery level will be indicated on the display).
The balance can be used while recharging the battery.

Use together with the animal weighing (average & hold) mode.

GXA-23-PRINT
GXA-23-REZERO

Accepts an oﬀ-the-shelf USB ﬂash drive to save weighing results in CSV format.

Density determination kits
Make weighing the sample in air andin water quicker, easier and more
precise. A&D Apollo can then performa density calculation automatically.

Power is supplied from the balance. Activated for a set duration with anembedded IR sensor.
For touchless operation of print or re-zero (tare).

one of GXA-03, GXA-04, GXA-06, FXi-08, GXA-09, GXA-17, GXA-23-PRINT/REZERO/PLUG, GXA-24, GXA-25 or*11 Only
GXA-26 can be installed (meaning that GXA-03, GXA-04, GXA-06, FXi-08, GXA-09, GXA-23-PRINT/REZERO/PLUG, or
GXA-24 cannot be installed in the GX-AE series without ﬁrst removing its interface for the built-in fanless ionizer.).
12 Provided as standard for the 0.0001 readability models of the GX-A/GF-A series.
13 Provided as standard for the GX-AE series.

Accessories

*
*

|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|
|

AD-1641
AD-1682
AD-1683
AD-1684A
AD-1687
AD-1688
AD-1689
AD-1691
AD-8127
AD-8526
AD-8920A
AD-8922A
AX-BM-NEEDLESETAX-GXA-31
AX-SW137-PRINT
AX-SW137-REZERO
AX-USB-9P

Air ﬂow logger
Rechargeable battery unit
Static eliminator*14
Electrostatic ﬁeld meter
Weighing environment logger
Weighing data logger
Tweezers for calibration weight
Weighing environment analyzer
Compact printerSerial/Ethernet converter
Remote display
Remote controller
Electrode unit for ionizer (4 pcs)
Main unit cover (5 pcs)
Foot switch for print
Foot switch for re-zeroSerial/USB converter with cable

AD-8127

AX-SW137-PRINT

AX-SW137-REZERO

Power needs to be supplied from an AC adapter. Recommended when A&D Apollo has another option than
*14 GXA-17
or GXA-25 installed.

8

AD-8529PC/PR-W Bluetooth® converters

Enable wireless communication between a balance/scale and a PC/printer using Bluetooth® technology*15(line-of-sight distance approx. 10 m when there is no radio wave interference).
AD-8529PC-W for communication with a PC

USB connector

Approx. 10 m

D-Sub connector

AD-8529PR-W for communication with a printercontact your local A&D representative to ﬁnd out whether the AD-8529PC/PR-W*15 Pleaseis certiﬁed for compliance with Bluetooth communication laws in your country.

D-Sub connector®

Speciﬁcations
GX-AE/GX-A/GF-A (0.0001g)
ModelsCapacity

GX-124AE

GX-224AE

GX-324AE

GX-124A

GX-224A

GX-324A

GF-124A

GF-224A

GF-324A

122 g

220 g

Readability

320 g

0.0001 g

Repeatability (standard deviation)

0.0002 g (300 g)
0.0001 g (200 g)

0.0001 g

Minimum weight* (typical)

120 mgiLinearity±0.0002 g

Stabilization time
(when set to FAST under a good environment)±0.0003 g

Sensitivity drift

Approx. 2 sec (300 g)
Approx. 1.5 sec (100 g)

Approx. 2 sec (200 g)
Approx. 1.5 sec (100 g)

Approx. 1.5 sec (100 g)±2 ppm/°C (10 to 30 °C/50 to 86 °F, when automatic self calibration is OFF)

Operating environment

5 to 40 °C (41 to 104 °F), 85%RH or less (no condensation)

Display refresh rate

5 times/sec, 10 times/sec or 20 times/sec

Units of measure*ii

mg (milligram), g (gram), oz (ounce), ozt (troy ounce), ct (metric carat), mom (momme), dwt (pennyweight), gr (grain),pcs (counting mode), % (percent mode), SG (density mode), and a user-programmable unit

Counting mode
Percent mode

Minimum unit mass

0.0001 g

Number of samples

5, 10, 25, 50 or 100 pieces

Minimum 100% reference mass

0.0100 g% readability

0.01%, 0.1% or 1% (depends on the reference mass stored)

Communication interface

RS-232C and USB
50 g
100 g

50 g

50 g

Applicable calibration weight value

100 g

100 g

200 g

200 g

Weighing pan size

300 g

Ø90 mmExternal dimensions

259 (W) × 358 (D) × 332 (H) mm (including the large glass breeze break)

Net weight

Approx. 7 kg

Power supply / consumption

AC adapter / approx. 30 VA

GX-A/GF-A (0.001g)
ModelsCapacity

GX-203A

GX-303A

GX-403A

GX-603A

GX-1003A

GX-1603A

GF-123A

GF-203A

GF-303A

GF-403A

GF-603A

GF-1003A

GF-1603A

122 g

220 g

320 g

420 g

620 g

1100 g1620 gReadability

0.001 gRepeatability (standard deviation)

0.002 g (1600 g)
0.001 g (1000 g)

0.001 gMinimum weight*i (typical)

1.4 g

Linearity±0.003 g±0.002 g
Approx. 1 sec
Approx. 0.8 sec (5 g)

Stabilization time
(when set to FAST under a good environment)
Sensitivity drift

Approx. 1.5 sec
Approx. 0.8 sec (5 g)±2 ppm/°C (10 to 30 °C/50 to 86 °F, when automatic self calibration is OFF)

Accuracy immediately after internal
calibration (for the GX-A series)*iii±0.010 g (1000 g)±0.010 g

Operating environment

5 to 40 °C (41 to 104 °F), 85%RH or less (no condensation)

Display refresh rate

5 times/sec, 10 times/sec or 20 times/sec

Units of measure*ii

g (gram), oz (ounce), lb (pound), lb-oz (pound-ounce), ozt (troy ounce), ct (metric carat), mom (momme), dwt (pennyweight),
gr (grain), pcs (counting mode), % (percent mode), SG (density mode), and a user-programmable unit

Counting mode
Percent mode

Minimum unit mass

0.001 gNumber of samples

5, 10, 25, 50 or 100 pieces

Minimum 100% reference mass

0.100 g% readability

0.01%, 0.1% or 1% (depends on the reference mass stored)

Communication interface

Applicable calibration weight value
Weighing pan size
External dimensions
Net weight
Power supply / consumption

9

RS-232C and USB
50 g
100 g

50 g
100 g
200 g

50 g
100 g
200 g
300 g

50 g
100 g

50 g
100 g

50 g
100 g

50 g
100 g

(100 g interval) (100 g interval) (100 g interval) (100 g interval)
400 g

600 g

1000 g128 × 128 mm
212 (W) × 317 (D) × 171 (H) mm (including the small plastic breeze break)Approx. 5 kg
AC adapter / approx. 30 VA

1600 gGX-A/GF-A (0.01/0.1g)

GX-2002A GX-3002A GX-4002A GX-6002A GX-10002A GX-6001A GX-10001A
GF-1202A GF-2002A

ModelsCapacity

1220 g

GF-3002A GF-4002A GF-6002A GF-10002A GF-6001A GF-10001A

2200 g3200 gReadability

4200 g0.02 g (10000 g)
0.01 g (5000 g)

Minimum weight*i (typical)

100 g±0.02 gApprox. 1 sec
Approx. 0.8 sec (50 g)

Stabilization time
(when set to FAST under a good environment)

10200 g

0.1 g

14 g

Linearity±0.03 gApprox. 1.5 sec
Approx. 0.8 sec (50 g)±0.1 gApprox. 1 sec
Approx. 0.8 sec (500 g)±2 ppm/°C (10 to 30 °C/50 to 86 °F, when automatic self calibration is OFF)

Sensitivity drift
Accuracy immediately after internal
calibration (for the GX-A series)*iii±0.10 g±0.15 g (5000 g)±0.15 g±0.5 g (5000 g)

5 to 40 °C (41 to 104 °F), 85%RH or less (no condensation)
5 times/sec, 10 times/sec or 20 times/sec

Operating environment
Display refresh rate

g (gram), oz (ounce), lb (pound), lb-oz (pound-ounce), ozt (troy ounce), ct (metric carat), mom (momme), dwt (pennyweight),
gr (grain), pcs (counting mode), % (percent mode), SG (density mode), and a user-programmable unit

Units of measure*ii

Percent mode

6200 g0.1 g0.01 gRepeatability (standard deviation)

Counting mode

10200 g6200 g0.01 gMinimum unit mass

0.01 g0.1 g

Number of samples

5, 10, 25, 50 or 100 pieces
1.00 g10.0 gMinimum 100% reference mass

0.01%, 0.1% or 1% (depends on the reference mass stored)
RS-232C and USB% readability

Communication interface

Applicable calibration weight value

500 g1000 g

500 g

500 g

1000 g1000 g2000 g2000 g3000 g500 g1000 g500 g1000 g500 g1000 g500 g1000 g500 g1000 g(1000 g interval) (1000 g interval) (1000 g interval) (1000 g interval) (1000 g interval)
4000 g6000 g

10000 g6000 g

10000 g165 × 165 mm
212(W) × 317(D) × 93(H) mm
Approx. 5 kg
AC adapter / approx. 30 VA

Weighing pan size
External dimensions
Net weight
Power supply / consumption

Pursuant to the United States Pharmacopeia (USP), Chapter 41*iii One
unit from tael (Singapore/HK jewelry/Taiwan/China), tola or Newton can be added upon request.
*iii Underadditional
stable environment (no rapid temperature/humidity change, vibration, draft, magnetism, static, etc). The mass of the internal weight may vary with age.
*

Dimensions (mm/inches)
GX-124AE / GX-224AE / GX-324AE
GX-124A / GX-224A / GX-324A
GF-124A / GF-224A / GF-324A

358/14.09
190.5/7.5

90.5/3.56

332/13.07

223/8.78

259/10.2

94.7/3.73

217/8.54

45/1.77

224/8.82
274/10.79
190/7.48

188/7.4

172/6.77

207.5/8.17

268/10.55

141/5.55

161/6.34

259/10.2

10Dimensions (mm/inches)

317/12.48

40/1.58

GX-203A / GX-303A / GX-403A / GX-603A /GX-1003A / GX-1603A

160/6.3
144/5.67
124/4.88

210/8.27

212/8.35

160/6.3
144/5.67
124/4.88

74.5/2.93

77/3.03

GF-123A / GF-203A / GF-303A / GF-403A /GF-603A / GF-1003A / GF-1603A

190/7.48

175/6.7

128/5.04

90/3.54

93/3.66

128/5.04

94.7/3.73

181/7.13

217/8.54

224/8.82

317/12.48

GX-2002A / GX-3002A / GX-4002A / GX-6002A /GX-10002A / GX-6001A / GX-10001A

40/1.58

210/8.27

212/8.35

GF-1202A / GF-2002A / GF-3002A / GF-4002A /GF-6002A / GF-10002A / GF-6001A / GF-10001A

165/6.5
190/7.48

93/3.66

165/6.5

181/7.13

94.7/3.73

217/8.54

224/8.82

GX-AE/A

GF-A

Optional

A&D Company, Ltd.

3-23-14 Higashi-Ikebukuro, Toshima-Ku, Tokyo, 170-0013, Japan Tel: +81 3-5391-6132 Fax: +81 3-5391-1566 www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131, U.S.A. Tel: +1 408-263-5333 Fax: +1 408-263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031, Australia Tel: +61 8-8301-8100 Fax: +61 8-8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire, OX14 1DY, United Kingdom Tel: +44 1235-550420 Fax: +44 1235-550485<German Sales Office>Hamburger Straße 30, D-22926, Ahrensburg, Germany Tel: +49 4102-459230 Fax: +49 4102-459231

A&D Korea Ltd.

8F Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331, Korea Tel: +82 2-780-4101 Fax: +82 2-782-4280

A&D Rus Co., Ltd.

Vereyskaya Str. 17, 121357, Moscow, Russia Tel: +7 495-937-33-44 Fax: +7 495-937-55-66

A&D Instruments India (P) Ltd.

509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India Tel: +91 (124) 471-5555 Fax: +91 (124) 471-5599* GXGF-A-ADCC-06-PR6-206xxGP