(This document is included in the packing contents for the balance.)

Quick Start Guide
BA Series
Analytical Balances

English (EN)Guide de démarrage rapide
Série BA
Balances analytiques

Français (FR)

Schnellstartanleitung
Serie BA
Analysenwaage

Deutsch (DE)

Guía de inicio rápido
Serie BA
Básculas analíticas

Español (ES)Guida introduttiva
Serie BA
Bilance analitiche

Italiano (IT)

快速入門指南BA 系列
分析天平

中文 (CT)

1WMPD4004294B

© 2021 A&D Company, Limited. All rights reserved.
No part of this publication may be reproduced, transmitted, transcribed, or translated into
any language in any form by any means without the written permission of A&D Company,
Limited.

The contents of this publication and the specifications of the instrument covered by thispublication are subject to change for improvement without notice.

Quick start guide – BA Series

EN

Analytical balances ..........................................................................4

Guide de démarrage rapide – Série BA

FR

Balances analytiques .....................................................................19

Schnellstartanleitung – Serie BA

DE

Analysenwaage .............................................................................35
Guía de inicio rápido – Serie BA

ES

Básculas analíticas ........................................................................53

Guida introduttiva – Serie BA

ITBilance analitiche ...........................................................................70

快速入門指南– BA 系列

CT

分析天平 ........................................................................................86

3

EN

1.

Quick Start Guide

Thank you for purchasing an A&D BA series analytical balance. This quick start guide describes how to
install the BA series balance, and it covers the basic functions and main operations of the balance.
For further information about the BA series balance, please refer to the separate instruction manuals
listed in "1.1. Detailed manuals".

1.1. Detailed manuals
Detailed manuals are available for download from the A&D website ( https://www.aandd.jp ):BA Series Instruction ManualThis is the manual to help you understand the functions and operations of the BA series in detail andmake full use of them.

Ionizer (static eliminator) Instruction ManualThis is the manual describing the functions of the AX-ION-25 ionizer and instructions on use.

2.

Assembly and Installation

CautionA BA series analytical balance is a precision instrument, and it should be carefully unpacked.
It is advisable to store the packing materials so that they can be used when transporting the balance.
The contents of the package vary depending on the product. Refer to the illustration of the packing
contents and make sure that everything is included.
Do not connect the AC adapter to the balance until the balance is assembled and installed.
Use the dedicated AC adapter specified for the balance.
Do not connect the included AC adapter to other devices.
If you use the wrong AC adapter, the balance and other equipment may not work properly.
Make sure that the AC adapter is unplugged before you connect the breeze break unit and the ionizer.
The BA series analytical balance is composed of the weighing unit and display unit. It is not possible
to replace just one of the units.

4

EN

2.1. Unpacking
Packing contents common to BA series
Weighing unit

Breeze breakbottom plate

Display unit protection coverBreeze break
top door

Tweezers
Display unit
Cable A (approx. 0.2 m)

Cable B(approx. 1.0 m)

USB cable(approx. 1.8 m)

AC adapter(approx. 1.5 m)

AC adapter ID labels

This document

Attach the AC adapter labels to the adapter.

Packing contents for BA-6E / BA-6DE
Pan A, Φ25 Pan support A Weighing pan for filters
Dust plate A

Breeze break unit
Breeze breakfront glass

Breeze break ring A

Small glass breeze break set

Side panes (3 pieces)

Top pane

Base standAluminum round pans for analysis (10 pieces each)
Breeze break doors
Large pan,
Φ15, 0.8 mL

Medium pan,Φ12, 0.3 mL

Small pan,
Φ8, 0.05 mL

Cable C(approx. 0.6 m)

IonizerPacking contents for BA-225 / BA-225D / BA-125D
Pan B, Φ85

Breeze break ring B

Dust plate B

Breeze break front glass

Breeze break doors

5

Breeze break unitEN

2.2. Assembly
The BA-6E is used here for the example of assembly. Other models can be assembled in the same way.

Breeze break unit

Step 1 Pull out the breeze
locking handles.

break

Step 1

Step 2 Place the breeze break unit on
the weighing unit.

Step 3

Step 2

Step 3 Push in the breeze breaklocking handles to secure theunits to each other.

Step 4 Assemble the weighing pan.
For BA-6E / BA-6DE, assemble
the five parts of the weighing
pan in the correct position byreferring to the cross-sectionalview.
In addition to the standard
weighing pan (Pan A), you canalso use the weighing pan forfilters. For details, refer to thedetailed instruction manual
listedin“1.1.
Detailed
manuals”.

Weighing unit

For BA-225 / BA-225D / BA-125D
For BA-6E / BA-6DE

Weighing pan
for filters

Pan A

Pan support A
Breeze breakring A
Dust plate A
Breeze breakbottom plate

For BA-225 / BA-225D / BA125D, assemble the four partsof the weighing pan in thecorrect position by referring to
the cross-sectional view.

Cross-sectional view
BA-6E / BA-6DE
Cross-sectional view
BA-225 / BA-225D
BA-125D

6

Pan B
Breeze breakring B

Dust plate B

EN
Step 5 Assemble the glass breeze
break.
Assemble the BA-6E / BA-6DE
glass breeze break in theweighing chamber. Align theside panes with the notch of thebase stand.

Top paneSide panes(3 pieces)

Step 5

Base stand

Example of the small breeze
break with the right side open.

Side pane

Notch
Base stand

Step 6 Insert the breeze break top door
from the handle side.
Step 7 Assemble the breeze break
doors on the left and right sides.
7a7bInsert the breeze break doors
on the left and right sides.
Secure the breeze break doors
with the latches.

Step 7a
Step 6
Step 8b
Step 7b
Step 7a

Step 8a

Step 8 Assemble the breeze breakfront glass.
8a

Insert the breeze break front
glass.

8b

Secure the breeze break front
glass with the latches.

Step 8b
Step 7b

7

EN
Step 9Using the included cable A and cable B, connect the "weighing unit" and "breeze break unit"
and the "display unit" and "weighing unit" respectively, paying attention to the direction of thearrow on each cable.
Caution Make sure to unplug the AC adapter before connecting.
Step 10 Connect the breeze break unit and ionizer with the included cable C.
Caution The ionizer operates when the AC adapter is connected to the balance.
Step 11 Connect the included AC adapter to the balance.
Grounding terminal

Display unit
Breeze break unit

Serial number
IonizerCable A
Step 9Cable B
Step 11

AC adapter
Weighing unit
Step 10

Cable C

BA-225 / BA-225D / BA-125D

BA-6E / BA-6DE

8

EN

3.

Installation Considerations, Preparation and Precautions

Prepare the following installation conditions in order to bring out the full performance of the balance.
Install the balance in an environment where the temperature and humidity are not excessive. The
best operating temperature is about 20°C ±2°C at about 45% to 60% RH relative humidity.
Install the balance where it is free of dust.
The weighing table should be solid. (An anti-vibration table or stone table is ideal)
Place the balance on a horizontal table, and make sure that it is not tilted.
Install the balance in a stable location, avoiding vibration and shock. Corners of rooms on the firstfloor are best, as they are less prone to vibration.
Install the balance where it is not affected by heaters or air conditioners.
Install the balance where it is not exposed to direct sunlight.
Leveling footInstall the balance away from equipment which produces
magnetic fields.
Bubble spirit level
Level the balance with the leveling feet and bubble spirit level.
Refer to “3.1. How to adjust the level of the balance”.
Be sure to warm up the balance before use for at least an hour, or at least four hours for BA-6E / BA6DE, with the AC adapter connected to the power supply.
Adjust the sensitivity of the balance before using it for the first time or after having moved it to anotherlocation so that accurate weighing can be performed. For details, refer to the detailed instruction
manual listed in “1.1. Detailed manuals”.
Caution Do not install the balance where flammable or corrosive gas is present.

3.1. How to adjust the level of the balance
Bubblespirit level
The balance is level(Correct weighing)

The balance is tilted(Inaccurate weighing).

Up

Down

Leveling
footLeveling
foot

Down

Red circle

BubbleUp

Level the balance by adjusting the leveling feet so that the bubble of the bubble spirit level is centeredin the red circle.

When the bubble is off to the left.

When the bubble is off to the right.

When the bubble is off to the backward
position.

When the bubble is off to the forward
position.

Turn the leveling foot on the front right in theclockwise direction.

Turn the leveling foot on the front left in theclockwise direction.

Turn both leveling feet on the front in theclockwise direction at the same time.

Turn both leveling feet on the front in the counterclockwise direction at the same time.

9

EN

4.

Display and Key Panel (Basic Operation)

Lit display
Data memory number
Response indicator (lights for 30 seconds after startof weighing)
Shock indicator

IR sensor indicator
Preset tare markGross markNet mark
Processing indicator

Gross zero markStabilization indicatorInterval output mode
standby indicator

USB connecting markStandby indicator

Unit display
Displays weighing value, stored data, and items

Blinking display
Data number being displayed
Processing indicator
Automatic sensitivity adjustment noticeInterval output mode active indicator

Keys
Right IR sensor

Left IR sensor

key

SAMPLE

key

MODE
CAL

key

ON/OFF

key

PRINT

key

TARE

key

RE-ZERO

10

key

EN

Key operation

Key operations affect how the balance functions. Normal
key operation during measurement is “Press and release
the key immediately” or “Press the key”. Please do not“Press and hold the key (for 2 seconds)” unless required.
Key

When pressed and released

Press the key(Press and release
the key immediately.)

Press and holdthe key for 2
seconds.

When pressed and held (for 2 seconds)

Turns the display on and off.
When the display is turned off, only thestandby indicator is displayed.
When the display is turned on, weighing ispossible.
If the password function is enabled, you will
be prompted to enter the password when thedisplay is turned on. For details, refer to theseparate instruction manual listed in “1.1.
Detailed manuals".
The ON:OFF key is active at any time,and pressing this key during operation
always turns off the display.

In weighing mode, press the key to turn thedigit for readability on and off.
In counting or percent mode, press the key
to enter the sample storing mode.

Switches the IR sensor on and off.

Enters the function table mode. Pleaserefer to the separate instruction manual
listed in “1.1. Detailed manuals".
Runs the repeatability check function whenpressed and held for another 2 seconds
after the function table menu is displayed.
For details, refer to the separate instruction
manual listed in “1.1. Detailed manuals".

Switches the weighing units stored in thefunction table.

Enters "Changing Weighing Speed" mode.

Performs sensitivity adjustment of thebalance using the internal weight.

Displays the menu related to sensitivity
adjustment.

Stores or outputs the weighing value whenstable according to the function tablesettings.
(Data is output at factory setting.)

Enters mode to change the unit weightregistration number in counting mode.
The following settings are possible with thefunction table:
Outputs "Title block" and "End block" for
GLP/GMP report.
Displays the data memory menu.

Performs tare operation.

-

Sets the displayed value to zero.

-

IR sensor (touchless sensor).
It reacts when you bring your hand close toit.
Opening and closing of the breeze breakdoor are assigned.
For details, refer to "5. IR Sensors andAuto Doors".

-

11

EN

5.

IR Sensors and Auto Doors

5.1. IR sensors
BA series analytical balances are equipped with IR sensors that allow operation without directly touching
the balance display. At factory setting, the IR sensors on the left and right of the display are assigned toopen and close the breeze break doors. By pressing and holding the ON:OFF key (for about 2
seconds), you can switch the IR sensors on and off with a single touch.

5.2. Auto doors
BA series analytical balances are equipped with auto doors that allow you to open/close the breeze breakwithout touching the doors.
The IR sensors on the left and right of the display are assigned settings to open/close the breeze break
doors on the right and left respectively.
At factory setting, each breeze break door opens to the position it was previously opened to.
It is also possible to change which sensors the breeze break doors are assigned to and for the doors to
be fully open or partially open in the function table of the balance.
For details, refer to the separate instruction manual listed in “1.1. Detailed manuals”.

Opening the breeze break door

Step 1. If you want to open the breeze break door on the left side (or on the right side) when it is closed,
hold your hand over the right IR sensor (or the left IR sensor).
Step 2. The detection buzzer sounds and the breeze break door on the left (or right) side opens.

Closing the breeze break door

Step 1. If you want to close the breeze break door on the left side (or on the right side) when it is open,
hold your hand over the right IR sensor (or the left IR sensor).
Step 2. The detection buzzer sounds and the breeze break door on the left (or right) side closes.

Operation example
Left IR sensor:

Opens/closes the right breeze
break door.

Example for BA-6E / BA-6DE

Right IR sensor: Opens/closes the left
breeze break door.

12

EN

6.

Sensitivity Adjustment

6.1. Automatic sensitivity adjustment
This function automatically adjusts the sensitivity of the balance according to ambient temperaturechange, set time or interval time using the internal weight. For details, refer to the separate instruction
manual listed in “1.1. Detailed manuals”.

6.2. Sensitivity adjustment using the internal weight
Sensitivity adjustment using the internal weight can be performed with one key press.
Step 1. Be sure to warm up the balance with nothing on the weighing pan for at least an hour, or at least
four hours for BA-6E/BA-6DE, with the AC adapter connected to the power supply.
Step 2. Press the

CAL key. The balance displays.

Step 3. The balance automatically performs sensitivity adjustment using the internal weight. Do not
apply vibration and the like to the balance.
Step 4. After sensitivity adjustment, the balance will output the “sensitivity adjustment report” and store
the data as set in the function table.
For details, refer to the separate instruction manual listed in “1.1. Detailed manuals”.
Step 5. The balance returns automatically to weighing mode.

6.3. Sensitivity adjustment using an external weightThis function performs sensitivity adjustment of the balance using your external weight.
For details, refer to the separate instruction manual listed in “1.1. Detailed manuals”.

13

EN

7.
Display

Error Displays (Error Codes)
Code

Description

Stability error
Weighing value is unstable and "zero display", "sensitivity adjustment", etc.
cannot be executed. Check around the pan. Refer to the separate
EC,E11
instruction manual listed in “1.1. Detailed manuals”. Improve theenvironment of the installation location (vibration, drafts, static electricity,etc.). To return to weighing mode, press the CAL key.

Out of the setting range
The value to be set exceeds the setting range.
Set again within the setting range.

Malfunction of the internal memory element of the balance
If this error continues to be displayed, repair is necessary. Please contact
your local dealer for repair.

Internal weight error
Applying the internal weight does not yield a change in the mass value as
EC,E16 specified.
Confirm that there is nothing on the pan and perform the operation from the
beginning.

Internal weight error

EC,E17 The internal weight application mechanism does not function properly.
Perform the operation from the beginning.

Abnormality in the internal memory data of the balance
If this error continues to be displayed, repair is necessary.
Please contact your local dealer for repair.

Weighing unit error
Data from the weighing unit has not been received. Check the connection
between the weighing unit and the display unit, and turn the power off andthen on again.

Breeze break unit error
Communication with the breeze break unit is not possible.
Unplug the power supply and reconnect the cable between the breeze
break unit and the weighing unit.

Serial number error
The serial numbers of the weighing unit and the display unit do not match.
Reconnect with the correct combination.

Sensitivity adjustment weight error (Positive value)
EC,E20The sensitivity adjustment weight is too heavy. Check around the pan.
Check the mass value of the weight.
To return to weighing mode, press the CAL key.

14

EN
Display

Code

Description

Sensitivity adjustment weight error (Negative value)
EC,E21The sensitivity adjustment weight is too light. Check around the pan. Checkthe mass value of the weight.
To return to weighing mode, press the CAL key.

Overload error
A sample beyond the balance weighing capacity has been placed on thepan. Remove the object from the pan.

Weighing pan error
The weighing value is too light. Check that the weighing pan is installed
correctly. Set the weighing pan correctly. Perform sensitivity adjustment.

Sample mass error
The sample is too light to be stored as a sample mass for the counting modeor percent mode. The sample cannot be used.

Unit weight error
The sample mass for the counting mode is too light. Storing and using it for
counting may cause a counting error. Add samples to reach the specified
number and press the PRINT key. Although pressing the PRINT key
without adding samples will put the balance in counting mode, make sureto add samples for accurate counting.

Clock battery error
The clock backup battery has been depleted. Press any key and set thetime and date. Even if the clock backup battery is depleted, the clock andcalendar function works normally as long as the balance is powered with
the AC adapter. If this error appears frequently, contact your local dealer for
repair.

Power supply voltage fault
The voltage supplied from the AC adapter is abnormal.
Check that the AC adapter is the one supplied with the balance.

Repeatability error
The standard deviation (SD) of repeatability exceeded 50 d. Review the
installation environment of the balance.
displayed in repeatability display.
displayed in minimum weighing value (reference value) display.
For details, refer to the separate instruction manual listed in”1.1. Detailed
manuals”.

Full memory
Blinking (alternately)

The number of stored weighing values has reached the upper limit. In orderto store a new weighing value, it is necessary to delete data. For details,
refer to the separate instruction manual "1.1. Detailed manuals”.

Full memory
Blinking (alternately)

The stored sensitivity adjustment / calibration test history has reached 50
results. In order to store a new result, the oldest history will be deleted. For
details, refer to the separate instruction manual "1.1. Detailed manuals”.

15

EN
Display

Code

Description

Communications error
EC,E00 A protocol error occurred in communications.
Check the format, baud rate, etc.

Undefined command error
EC,E01 An undefined command was found.
Check the transmitted command.

Not readyThe received command cannot be executed:
EC,E02 (e.g.) Q command was received when not in weighing mode.
(e.g.) Q command was received while re-zeroing.
Adjust the delay time to transmit a command.

Timeout error
EC,E03

With the timeout parameter set to, there was a wait time of
approximately 1 second or more while receiving command characters.
Check the communication.

Character length error
EC,E04 The number of characters in the received command has exceeded thelimit. Check the command to transmit.

Format error
The description of the received command is incorrect:
EC,E06 (e.g.) The number of digits of numerical values is incorrect.
(e.g.) There are alphabet characters among the numerical values.
Check the transmitted command.

Parameter setting error
EC,E07 The value of the received command has exceeded the allowed value.
Check the setting range of the numerical value of the command.

7.1. Other display
The automatic sensitivity adjustment notice (theindicator blinking) indicatesthat the automatic sensitivity adjustment will start.
If the balance is not used for a certain period of time with this indicator blinking, thebalance automatically performs sensitivity adjustment using the internal weight. (Theblinking period depends on the operating environment.)
Tips

Although it is possible to continue using the balance even while this indicator isblinking, use after sensitivity adjustment is completed is advisable in order to maintainthe weighing accuracy.

16

EN

8.

Ionizer (AX-ION-25)

The AX-ION-25 ionizer comes standard with the BA-6E / BA-6DE. The ionizer removes static electricity
by irradiating the target object with positive or negative ions, which are generated from four discharge
electrodes by DC corona discharge. Ordinarily, insulators such as powders, filters, and weighing paper
tend to be charged when the humidity is 45% RH or less, and an error of a few milligrams may occur
during weighing. By removing static from the weighing object with the ionizer, it is possible to eliminate
the error in the weighing value due to charging, and correct weighing can be performed. For details, referto the separate instruction manual in "1.1. Detailed manuals".

8.1. Using the ionizer
Step 1. Make sure that the AC adapter of the balance is not connected, and then connect the balance
and the ionizer (as noted in "2. Assembly and Installation").
Step 2. Connect the AC adapter, plug it in, and then turn on the power. The power lamp of the ionizer
lights up.
Step 3. Place the target object within the effective range of static elimination.
Step 4. Hold your hand over the IR sensor (touchless infrared proximity sensor) on the front of theionizer to start static elimination. The ION lamp indicating that static elimination is in progress
lights up.
Step 5. Refer to the figure below for the effective range of static elimination, and perform static
elimination. At factory setting, when 3 seconds have passed from the start of static elimination,
the ION lamp indicating that static elimination is in progress turns off and static elimination stops.

IR SENSOR (touchless infrared proximity sensor)

Ionizer(AX-ION-25)

Power lamp (lights up when the power is turned on)
ION lamp (lights up during static elimination)

Effective range of static elimination
Top view

Effective range
0 cm
Side view

5 cm
Distance

7 cm

10 cm

1 sec 1.5 sec
Static elimination time

3 sec

Effective range

17

EN

9.

Specifications
BA-6E

Weighing capacity

6.2 g

Maximum display

6.200084 g

BA-6DE
6.2 g
2.1 g
6.20008 g
2.100009 g
0.01 mg

Readability

0.001 mg

Repeatability(Standard deviation)

0.0030 mg (6 g)

0.01 mg (6 g)

0.0010 mg (1 g)

0.0025 mg (1 g)

Linearity±0.010 mg

0.001 mg±0.02 mg
±0.010 mg

Display

Liquid crystal display (monochrome)

Operation

Key switches

Communication

RS-232C (printer, PLC, etc.), USB Type Mini-B (PC)
Stereo jack (external switch)

BA-225Weighing capacity

220 g

Maximum display

220.00084 g

Readability
Repeatability(Standard deviation)
Linearity

0.01 mg
0.03 mg (200 g)
0.015 mg

BA-225D

BA-125D

220 g

120 g

51 g

51 g

220.0008 g

120.0008 g

51.00009 g

51.00009 g

0.1 mg

0.1 mg0.01 mg

0.01 mg

0.1 mg (200 g)(50 g)

0.025 mg

±0.15 mg

(50 g)±0.2 mg

0.1 mg (100 g)
0.030 mg±0.2 mg

Display

Liquid crystal display (monochrome)

Operation

Key switches

Communication

RS-232C (printer, PLC, etc.), USB Type Mini-B (PC)
Stereo jack (external switch)

18(50 g)

FR

1.

Guide de démarrage rapide

Nous vous remercions d'avoir acheté une balance analytique A&D de la série BA. Ce guide de démarrage
rapide décrit comment installer la balance de la série BA, et il couvre les fonctions basiques et lesopérations principales de la balance.
Pour d'informations sur la balance série BA, se référer aux manuels d'instructions séparés répertoriés
dans « 1.1. Manuels détaillés ».

1.1. Manuels détaillés
Les manuels détaillés peuvent être téléchargés depuis le site web d'A&D ( https://www.aandd.jp ) :BA Series Instruction Manual (Manuel d'instructions de la série BA)
Ce manuel vous aidera à comprendre en détail les fonctions et les opérations de la série BA et à lesutiliser intégralement.

Ionizer (static eliminator) Instruction Manual (Manuel d'instructions de l'ionisateur(éliminateur d'électricité statique))Il s'agit du manuel décrivant les fonctions de l'ionisateur AX-ION-25 et les instructions d'utilisation.

2.

Montage et installation

Attention
Une balance analytique de la série BA est un instrument de précision et elle doit être déballée avec
soin.
Il est conseillé de stocker le matériel d'emballage afin de pouvoir l'utiliser lors du transport de labalance.
Le contenu de l'emballage varie en fonction du produit. Se référer à l'illustration du contenu del'emballage et s'assurer que tout est inclus.
Ne pas connecter l'adaptateur CA à la balance avant que la balance ne soit montée et installée.
Utiliser l'adaptateur CA dédié spécifié pour la balance.
Ne pas connecter l'adaptateur CA fourni sur d'autres dispositifs.
Si un adaptateur CA incorrect est utilisé, la balance et les autres équipements peuvent ne pas
fonctionner correctement.
S'assurer que l'adaptateur CA est débranché avant de connecter l'unité de coupe-brise et l'ionisateur.
La balance analytique de la série BA est composée de l'unité de pesage et de l'unité d'affichage. Il
n'est pas possible de remplacer seulement une de ces unités.

19

FR

2.1. Déballage
Contenu de l'emballage ordinaire de la série BA
Couvercle de protection del'unité d'affichage

Unité de pesage

Plaqueinférieure ducoupe-brisePortesupérieure ducoupe-brise

Pince
Unité
d'affichage
Câble A (environ 0,2 m)

Câble B(environ 1,0 m)

Câble USB
(environ 1,8 m)

Adaptateur CA(environ 1,5 m)

Ce document

Contenu de l'emballage pour BA-6E / BA-6DE

Jeu de petites vitres de coupe-brise

Volet supérieur

Socle

Plateaux circulaires en aluminium (10 pièces de chaque)Portes du coupe-briseGrandplateau, Φ15,
0,8 mL

Plateaumoyen, Φ12,
0,3 mL

Petit plateau,
Φ8, 0,05 mL

Câble C
(environ 0,6 m)

Ionisateur

Contenu de l'emballage pour BA-225 / BA-225D / BA-125D
Plateau B, Φ25 Bague du coupe-brise B

Plaque antipoussière B

Vitre avant du coupe-brisePortes du coupe-brise

20

Unité de coupe-briseFR

2.2. Montage
Le modèle BA-6E est utilisé ici à titre d'exemple de montage. Les autres modèles peuvent être montésde la même manière.
Unité de coupe-brise

Étape 1 Sortir
lespoignées
deverrouillage du coupe-brise.

Étape 1

Étape 2 Placer l'unité de coupe-brise
sur l'unité de pesage.

Étape 3

Étape 2

Étape 3 Pousser les poignées deverrouillage du coupe-brisepour fixer solidement lesunités les unes aux autres.

Étape 4 Monter le plateau de pesage.
Pour le modèle BA-6E / BA-6DE,monter les cinq parties duplateau de pesage dans laposition correcte en se référant àla vue en coupe transversale.
En plus du plateau de pesage
standard (plateau A), il estégalement possible d'utiliser leplateau de pesage pour filtres.
Pour plus de détails, se référer
aumanueld'instructions
détaillées répertorié dans
« 1.1. Manuels détaillés ».
Pour le modèle BA-225 / BA225D / BA-125D, monter lesquatre parties du plateau depesage dans la position
correcte en se référant à la vue
en coupe transversale.

Unité de pesage

Pour BA-225 / BA-225D / BA-125D
Pour BA-6E / BA-6DE
Plateau depesage pourfiltres

Plateau A

Bague ducoupe-brise A
Plaqueantipoussière A
Plaqueinférieure ducoupe-brise

Vue en coupetransversale dumodèle BA-6E /BA-6DE

Vue en coupetransversale du modèle
BA-225 / BA-225D
BA-125D

21

Plateau B

Bague ducoupe-brise B

Plaqueantipoussière B

FR
Étape 5 Monter la vitre du coupe-brise.
Monter la vitre du coupe-brisedu modèle BA-6E / BA-6DE
dans la chambre de pesage.
Aligner les volets latéraux
avec l'encoche du socle.

Volet supérieur
Volets latéraux
(3 pièces)Étape 5

Socle

Exemple de petit coupe-brise
avec le côté droit ouvert.

Volet latéral

Encoche
Socle

Étape 6 Insérer la porte supérieure ducoupe-brise du côté de lapoignée.
Étape 7 Monter les portes du coupebrise sur les côtés gauche etdroit.
7a

7b

Insérer les portes du coupebrise sur les côtés gauche etdroit.

Étape 7a
Étape 6
Étape 8b
Étape 7b
Étape 7a

Étape 8aFixer solidement les portes ducoupe-brise avec les loquets.

Étape 8b

Étape 8 Monter la vitre avant du coupebrise.
8aInsérer la vitre avant du coupebrise.

8b

Fixer solidement la vitre avant
du coupe-brise avec lesloquets.

Étape 7b22

FR
Étape 9 À l'aide du câble A et du câble B fournis, connecter l'« unité de pesage » et l'« unité de coupebrise » et l'« unité d'affichage » et l'« unité de pesage » respectivement, en faisant attention ausens de la flèche sur chaque câble.
Attention S'assurer de débrancher l'adaptateur CA avant la connexion.
Étape 10 Connecter l'unité de coupe-brise et l'ionisateur avec le câble C fourni.
Attention L'ionisateur fonctionne lorsque l'adaptateur CA est connecté à la balance.
Étape 11 Connecter l'adaptateur CA fourni à la balance.
Borne de mise à la terre

Unité d'affichage
Unité de coupe-brise

Numéro de série
Ionisateur.

Câble A
Étape 9Câble B
Étape 11Adaptateur CA
Unité de pesage
Étape 10

Câble C

BA-225 / BA-225D / BA-125D

BA-6E / BA-6DE

23

FR

3. Considérations relatives à l'installation, préparation etprécautions
Préparer les conditions d'installation suivantes afin de tirer le meilleur rendement de la balance.
Installer la balance dans un environnement où la température et l'humidité ne sont pas excessives. La meilleuretempérature de fonctionnement est d'environ 20 °C±2 °C avec environ 45 à 60 % d'humidité relative.
Installer la balance dans un endroit exempt de poussière.
La table de pesage doit être robuste. (Une table anti-vibrations ou une table en pierre est idéale)Placer la balance sur une table horizontale et s'assurer qu'elle n'est pas inclinée.
Installer la balance dans un endroit stable, exempt de vibrations et de chocs. Les coins des pièces
au rez-de-chaussée sont idéaux car ils sont moins exposés aux vibrations.
Installer la balance dans un endroit non affecté par des systèmes de chauffage ou de climatisation.
Installer la balance dans un endroit non exposé à la lumière
Pied réglable
directe du soleil.
Installer la balance à l'écart d'équipements produisant deschamps magnétiques.
Niveau à bulle
Mettre la balance de niveau avec les pieds réglables et le niveau àbulle. Se reporter à « 3.1. Comment régler le niveau de la balance ».
Veiller à réchauffer la balance avant de l'utiliser pendant au moins une heure, ou au moins quatre
heures pour les modèles BA-6E / BA-6DE, avec l'adaptateur CA connecté à l'alimentation électrique.
Régler la sensibilité de la balance avant de l'utiliser pour la première fois ou après l'avoir déplacéevers un autre endroit afin de pouvoir effectuer un pesage précis. Pour plus de détails, se référer aumanuel d'instructions détaillées répertorié dans « 1.1. Manuels détaillés ».
Attention Ne pas installer la balance dans un environnement où des gaz inflammables ou corrosifs sont présents.

3.1. Comment régler le niveau de la balance
Niveau à
bulle

Haut

Bas

Pied
réglablePiedréglableBas

Cercle rouge

Bulle

Haut

Mettre la balance à niveau en réglant les pieds réglables de manière à ce que la bulle du niveau àbulle soit centrée dans le cercle rouge.

Quand la bulle dévie vers la gauche.

Quand la bulle dévie vers la droite.

Quand la bulle dévie vers l'arrière.

Quand la bulle dévie vers l'avant.

Tourner le pied réglable avant droit dans le sens
des aiguilles d'une montre.

Tourner le pied réglable avant gauche dans lesens des aiguilles d'une montre.

Tourner simultanément les deux pieds réglables
avant dans le sens des aiguilles d'une montre.

Tourner simultanément les deux pieds réglables avant
dans le sens contraire des aiguilles d'une montre.

24

FR

4.

Affichage et panneau à touches (Fonctionnement basique)Affichage allumé
Numéro de la mémoire de données
Indicateur de réponse (s'allume pendant 30
secondes après le début du pesage)
Indicateur de chocIndicateur de capteur IR
Indicateur de tare prérégléeMarque BrutMarque Net
Indicateur de traitement

Indicateur zéro brut

Indicateur de stabilisation

Indicateur de veille du mode
d'exportation de l'intervalle
Affichage des unités

Indicateur de connexion USB
Indicateur de veille

Affiche la valeur de pesage, les données
enregistrées et les éléments

Affichage clignotant
Numéro de données affiché
Indicateur de traitement
Avis de réglage automatique dela sensibilité
Indicateur de veille d'activité du mode d'exportation de l'intervalle

Touches
Capteur IR droit

Capteur IR gauche

SAMPLE
MODE

TouchePRINT

CALToucheTARE

ON/OFF

ToucheRE-ZERO

25

FR

Opération de touche

Les opérations de touche produisent des influences sur lesfonctions de la balance. « Appuyer et relâcher la touche
immédiatement » et « Appuyer sur la touche » sont desopérations de base normales pendant la mesure. Ne pas« Appuyer sur la touche et la maintenir enfoncée (pendant2 secondes) » sauf exigence spécifique.

Appuyer sur la touche(Appuyer sur latouche et la relâcher
immédiatement.)

ToucheEnfoncée et maintenue enfoncée(pendant 2 secondes)

Enfoncée puis relâchée
Met l'écran sous et hors tension.
Lorsque l'écran est mis hors tension, seul
l'indicateur de veille s'affiche.
Lorsque l'écran est mis sous tension, lepesage est possible.
Avec la fonction de mot de passe, « Saisir lemot de passe au début du pesage » s'affichelorsque l'écran est mis sous tension. Pourplus de détails, se référer au manueld'instructions séparé répertorié dans « 1.1.
Manuels détaillés ».
La touche ON:OFF est active à tout
moment, et appuyer sur cette touche
pendant l'opération met toujours horstension l'écran.

En mode de pesage, appuyer sur le touche
pour activer et désactiver le chiffre pour lalisibilité.
En mode de comptage ou de pourcentage,
appuyer sur le touche pour passer en moded'enregistrement de l'échantillon.

Appuyer sur latouche et lamaintenir
enfoncée pendant2 secondes.

Allume et éteint le capteur IR.

Passe en mode table de fonctions. Se
référer au manuel d'instructions séparé
répertorié dans « 1.1. Manuels
détaillés ».
Exécute la fonction de vérification de la
répétabilité en appuyant sur la touche eten la maintenant enfoncée pendant 2
secondes supplémentaires après
l'affichage du menu de la table defonctions. Pour plus de détails, se référer
au manuel d'instructions séparé répertorié
dans « 1.1. Manuels détaillés ».

Permute entre les unités de pesage
enregistrées dans la table de fonctions.

Passe en mode « Changement de la vitessede pesage ».

Effectue un réglage de la sensibilité de labalance à l'aide d'une poids interne.

Affiche d'autres éléments du menu deréglage de la sensibilité.

Enregistre ou exporte la valeur de pesage
lorsqu'elle est stable selon les réglages de latable de fonctions.
(Les données sont exportées lors du réglage
d'usine).

Passe au mode permettant de changer lenuméro d'enregistrement de la masse
unitaire en mode de comptage.
Les réglages suivants sont possibles avec latable de fonctions :Exporte le « Title block » (Bloc de titre) etle « End block » (Bloc de fin) pour établir
un rapport GLP/GMP.
Affiche le menu de mémoire des données.

Effectue une opération de tare.

26

FR
Règle la valeur affichée sur zéro.

-

Capteur IR (capteur sans contact).
Il réagit à l'approche d'une main.
L'ouverture et la fermeture de la porte ducoupe-brise sont attribuées.
Pour plus de détails, se référer à « 5.
Capteur IR et portes automatiques ».

-

27

FR

5.

Capteur IR et portes automatiques

5.1. Capteurs IR
Les balances d'analyse de la série BA sont équipées de capteurs IR qui permettent de les utiliser sans
toucher directement l'écran de la balance. Lors du réglage d'usine, les capteurs IR à gauche et à droite
de l'écran sont attribués à l'ouverture et à la fermeture des portes du coupe-brise. En appuyant et enmaintenant enfoncée la touche ON:OFF (pendant environ 2 secondes), les capteurs IR peuvents'allumer et s'éteindre par une simple pression.

5.2. Portes automatiques
Les balances analytiques de la série BA sont équipées de portes automatiques qui permettent d'ouvrir/fermer lecoupe-brise sans toucher les portes.
Les capteurs IR à gauche et à droite de l'écran sont des réglages attribués pour ouvrir/fermer les portes du coupebrise à droite et à gauche respectivement.
Lors du réglage d'usine, chaque porte du coupe-brise s'ouvre dans la position où elle a été ouverte précédemment.
Il est également possible de changer les capteurs auxquels les portes du coupe-brise sont attribuées et de changer
l'ouverture totale ou partielle des portes dans la table de fonctions de la balance.
Pour plus de détails, se référer au manuel d'instructions séparé répertorié dans « 1.1. Manuels détaillés ».

Ouverture de la porte du coupe-brise

Étape 1. Pour ouvrir la porte du coupe-brise du côté gauche (ou du côté droit) lorsque celle-ci est fermée,
passer la main sur le capteur IR droit (ou le capteur IR gauche).
Étape 2. Le signal sonore de détection retentit et la porte du coupe-brise du côté gauche (ou droit) s'ouvre.

Fermeture de la porte du coupe-brise

Étape 1. Pour fermer la porte du coupe-brise du côté gauche (ou du côté droit) lorsque celle-ci est
ouverte, passer la main sur le capteur IR droit (ou le capteur IR gauche).
Étape 2. Le signal sonore de détection retentit et la porte du coupe-brise du côté gauche (ou droit) se ferme.

Aperçu de l'opération
Le capteur IR de gauche ouvre/ferme la porte à droite.

Exemple pour BA-6E / BA-6DE

Le capteur IR de droite ouvre/ferme laporte à gauche.

28

FR

6.

Réglage de la sensibilité

6.1. Réglage automatique de la sensibilité
La sensibilité de la balance est automatiquement réglée à l'aide de la poids interne selon le changement
de température ambiante, du temps défini ou du temps d'intervalle. Pour plus de détails, se référer aumanuel d'instructions séparé répertorié dans « 1.1. Manuels détaillés ».

6.2. Réglage de la sensibilité à l'aide de la poids interne
Le réglage de la sensibilité à l'aide de la poids interne peut s'effectuer par une simple pression.
Étape 1. Sans rien sur le plateau de pesage, brancher la balance avec l'adaptateur CA afin de laréchauffer pendant au moins 1 heure (ou 4 heures pour BA-6E / BA-6DE).
Étape 2. Appuyer sur la touche

CAL pour afficher.

Étape 3. La balance effectue le réglage de la sensibilité à l'aide de la poids interne. Éviter d'exposer labalance à des vibrations ou des courants d'air.
Étape 4. Après le réglage de la sensibilité, le « relevé de réglage de la sensibilité » est exporté etenregistré comme un réglage défini dans la table de fonctions.
Pour plus de détails, se référer au manuel d'instructions séparé répertorié dans « 1.1. Manuels
détaillés »Étape 5. Une fois l'opération terminée, la balance revient automatiquement en mode de pesage.

6.3. Réglage de la sensibilité à l'aide d'un poids externe
À l'aide du poids externe, régler la sensibilité de la balance.
Pour plus de détails, se référer au manuel d'instructions séparé répertorié dans « 1.1. Manuels
détaillés ».

29

FR

7.
Affichage

Affichage d'erreur (Codes d'erreur)
Code

Description

Erreur de stabilité
La valeur de pesage est instable et les fonctions « affichage du zéro », « réglage
de la sensibilité », etc. ne peuvent pas être exécutées. Vérifier les abords duEC,E11 plateau. Se référer au manuel d'instructions séparé répertorié dans « 1.1.
Manuels détaillés ». Améliorer l'environnement de l'emplacementd'installation (vibrations, courants d'air, électricité statique, etc.). Pour revenir enmode de pesage, appuyer sur la touche CAL .

Hors de la plage de réglage
La valeur définie dépasse la plage de réglage.
Régler à nouveau dans la plage de réglage.

Défaillance d'un élément de mémoire interne de la balance
Si cette erreur continue à s'afficher, une réparation est nécessaire.
Contacter votre revendeur local pour procéder à la réparation.

Erreur de poids interne

EC,E16 L'application de la poids interne ne modifie pas la valeur de la masse spécifiée.
Confirmer que le plateau est vide et effectuer l'opération depuis le début.

Erreur de poids interne

EC,E17 Le mécanisme d'application de la poids interne ne fonctionne pas
correctement. Effectuer l'opération depuis le début.

Anomalies des données de mémoire interne de la balance
Si cette erreur continue à s'afficher, une réparation est nécessaire.
Contacter votre revendeur local pour procéder à la réparation.

Erreur de l'unité de pesage
Les données n'ont pas été reçues de l'unité de pesage. Vérifier la connexion
entre l'unité de pesage et l'unité d'affichage, puis mettre l'appareil horstension puis à nouveau sous tension.

Erreur de l'unité de coupe-brise
La communication avec le coupe-brise n'est pas connectée.
Débrancher l'alimentation électrique et reconnecter le câble entre l'unité decoupe-brise et l'unité de pesage.

Erreur de numéro de série
Les numéros de série de l'unité de pesage et de l'unité d'affichage necorrespondent pas.
Reconnecter avec la combinaison correcte.

Erreur de poids du réglage de la sensibilité (valeur positive)
EC,E20

Le poids du réglage de la sensibilité est trop lourd. Vérifier les abords duplateau. Vérifier la valeur de la masse du réglage de la sensibilité.
Pour revenir en mode de pesage, appuyer sur la touche CAL .

30

FR
Affichage

Code

Description

Erreur de poids du réglage de la sensibilité (valeur négative)
EC,E21

Le poids du réglage de la sensibilité est trop léger. Vérifier les abords duplateau. Vérifier la valeur de la masse du réglage de la sensibilité.
Pour revenir en mode de pesage, appuyer sur la touche CAL .

Erreur de surcharge
Un échantillon dépassant la capacité de pesage de la balance a été placé
sur le plateau. Retirer l'objet du plateau.

Erreur du plateau de pesage
La valeur de la masse est trop légère. Vérifier que le plateau de la balance
est correctement installé. Régler le plateau de la balance correctement.
Étalonner la balance.

Erreur de masse de l'échantillon
L'échantillon est trop léger pour être enregistré en tant que masse del'échantillon pour le mode de comptage ou de pourcentage. L'échantillon nepeut pas être utilisé.

Erreur de la masse unitaire
La masse de l'unité d'échantillonnage en mode de comptage est trop légère.
L'enregistrement et l'utilisation de l'échantillon peuvent entraîner une erreurde comptage. Ajouter des échantillons pour obtenir le nombre spécifié etappuyer sur la touche PRINT . Appuyer sur la touche PRINT sansajouter d'échantillon mettra la balance en mode de comptage. Toutefois, il
convient d'ajouter des échantillons pour obtenir un comptage précis.

Erreur de batterie de l'horloge
La batterie de secours de l'horloge est déchargée. Appuyer sur n'importequelle touche pour régler l'heure et la date. Même si la batterie de secours
de l'horloge est épuisée, la fonction horloge et calendrier fonctionne
normalement tant que la balance est alimentée par l'adaptateur CA. Si cetteerreur s'affiche de manière récurrente, contacter votre revendeur local.

Erreur de tension d'alimentation
L'alimentation fournie par l'adaptateur CA est anormale.
Vérifier si le problème provient de l'adaptateur CA fourni avec la balance.

Erreur de répétabilité
L'écart type (ET) de la répétabilité a dépassé 50 chiffres. Contrôlerl'environnement d'installation de la balance.
dans l'affichage de la répétabilité.
dans l'affichage de la valeur de pesage minimale (valeur de référence).
Pour plus de détails, se référer au manuel d'instructions séparé répertorié
dans « 1.1. Manuels détaillés ».

Mémoire pleine
Le nombre de valeurs de pesage enregistrées a atteint la limite supérieure.
Il est nécessaire de supprimer les données pour enregistrer de nouvelles
Clignotement (en alternance)
valeurs de pesage. Pour plus de détails, se référer au manuel d'instructions
séparé « 1.1. Manuels détaillés ».

31

FR
Affichage

Code

Description

Mémoire pleine
L'historique des tests d'étalonnage / de réglage de la sensibilité et a atteint
50 résultats. L'historique le plus ancien sera supprimé pour enregistrer un
Clignotement (en alternance)
nouveau résultat. Pour plus de détails, se référer au manuel d'instructions
séparé « 1.1. Manuels détaillés ».

Erreur de communications
EC,E00 Une erreur de protocole est survenue dans les communications.
Vérifier le format, la vitesse de transmission et la parité.

Erreur de commande non définie
EC,E01 Une commande indéfinie a été trouvée.
Vérifier la commande transmise.

Non prête
Une commande reçue ne peut pas être exécutée :(Par ex.) La commande Q a été reçue alors que le mode de pesage n'était
EC,E02pas réglé.
(Par ex.) La commande Q a été reçue pendant la remise à zéro.
Ajuster le temps de retard pour transmettre une commande.

Erreur de délai d'expiration
EC,E03

Avec le paramètre du délai d'expiration défini sur, il y a eu
un temps d'attente d'environ 1 seconde ou plus lors de la réception descaractères de commande. Vérifier la communication.

Erreur de longueur de caractères
EC,E04 Le nombre de caractères dans la commande reçue a dépassé la limite.
Vérifier la commande à transmettre.

Erreur de format
La description de la commande reçue est incorrecte :EC,E06 (Par ex.) Le nombre de chiffres des valeurs numériques est incorrect.
(Par ex.) Il y a des caractères alphabétiques parmi les valeurs numériques.
Vérifier la commande transmise.

Erreur de réglage des paramètres
EC,E07 La valeur de la commande reçue a dépassé la valeur autorisée.
Vérifier la plage de réglage de la valeur numérique de la commande.

7.1. Autre affichage
Il s'agit de l'avis de réglage automatique de la sensibilité (le repère
clignote).
Si la balance n'est pas utilisée pendant plusieurs minutes lorsque ce repère clignote,la balance effectue automatiquement un réglage de la sensibilité à l'aide de la poids
interne. (La période de clignotement dépend de l'environnement de fonctionnement.)
Conseil

Bien qu'il soit possible de continuer à utiliser la balance même lorsque ce repère
clignote, il est conseillé de l'utiliser une fois le réglage de la sensibilité terminé afin
de maintenir la précision du pesage.

32

FR

8.

Ionisateur (AX-ION-25)

L'ionisateur AX-ION-25 est livré en standard avec le modèle BA-6E / BA-6DE. L'ionisateur élimine
l'électricité statique en irradiant l'objet cible avec des ions positifs ou négatifs, qui sont générés à partirde quatre électrodes de décharge via une décharge corona DC. Habituellement, les isolants tels que lespoudres, les filtres et le papier de pesage ont tendance à être chargés lorsque l'humidité est de 45 % HR
ou moins, et une erreur de quelques milligrammes peut se produire lors du pesage. En éliminantl'électricité statique de l'objet à peser avec l'ionisateur, il est possible d'éliminer l'erreur de la valeur depesage due à la charge, et un pesage correct peut être effectué. Pour plus de détails, se référer aumanuel d'instructions séparé dans « 1.1. Manuels détaillés ».

8.1. Utilisation de l'ionisateur
Étape 1. S'assurer que l'adaptateur CA de la balance n'est pas connecté, puis connecter la balance etl'ionisateur (comme indiqué dans « 2. Montage et installation »).
Étape 2. Connecter l'adaptateur CA, le brancher, puis mettre l'appareil sous tension. Le voyant
d'alimentation de l'ionisateur s'allume.
Étape 3. Placer l'objet cible dans la plage effective d'élimination d'électricité statique.
Étape 4. Mettre la main au-dessus du capteur IR (capteur de proximité infrarouge sans contact) situé àl'avant de l'ionisateur pour commencer l'élimination de l'électricité statique. Le voyant ION
indiquant que l'élimination d'électricité statique est en cours s'allume.
Étape 5. Se reporter à la figure ci-dessous pour connaître la plage effective de l'élimination d'électricité
statique, et effectuer l'élimination de l'électricité statique. Lors du réglage d'usine, lorsque 3
secondes se sont écoulées depuis le début de l'élimination d'électricité statique, le voyant ION
indiquant que l'élimination de l'électricité statique est en cours s'éteint et l'élimination del'électricité statique s'arrête.
CAPTEUR IR (capteur de proximité infrarougesans contact)

Ionisateur(AX-ION-25)

Voyant d'alimentation (s'allume lorsque l'appareil
est mis sous tension)
Voyant ION (s'allume pendant l'élimination del'électricité statique)

Plage effective de l'élimination d'électricité statique
Vue de dessus
Plage effective
0 cm
Vue latérale

5 cm
Distance

7 cm

10 cm

Plage effective

1 sec 1,5 sec
3 sec
Durée de l'élimination d'électricité statique

33

FR

9.

Spécificités
BA-6E

Capacité de pesage

6,2 g

Affichage maximum

6,200084 g

BA-6DE
6,2 g
2,1 g
6,20008 g
2,100009 g
0,01 mg

Lisibilité

0,001 mg

Répétabilité(Écart type)0,0030 mg (6 g)

0,01 mg (6 g)

0,0010 mg (1 g)

0,0025 mg (1 g)

Linéarité±0,010 mg

0,001 mg

±0,02 mg
±0,010 mg

Affichage

Écran à cristaux liquides (monochrome)

Fonctionnement

Commutateurs touches

Communication

RS-232C (imprimante, PLC, etc.), USB type Mini-B (PC)
Prise stéréo (commutateur externe)

BA-225
Capacité de pesage

220 g

Affichage maximum

220,00084 g

Lisibilité
Répétabilité(Écart type)
Linéarité

0,01 mg
0,03 mg (200 g)
0,015 mg

BA-225D

BA-125D

220 g

120 g

51 g

51 g

220,0008 g

120,0008 g

51,00009 g

51,00009 g

0,1 mg0,1 mg0,01 mg

0,01 mg

0,1 mg (200 g)(50 g)

0,025 mg

±0,15 mg

(50 g)±0,2 mg

0,1 mg (100 g)
0,030 mg

(50 g)±0,2 mg

Affichage

Écran à cristaux liquides (monochrome)

Fonctionnement

Commutateurs touches

Communication

RS-232C (imprimante, PLC, etc.), USB type Mini-B (PC)
Prise stéréo (commutateur externe)

34

DE

1.

Schnellstartanleitung

Vielen Dank für den Kauf einer Analysenwaage der Serie A&D BA. In dieser Schnellstartanleitung werden
die Installation der Waage der Serie BA sowie die Grundfunktionen und die Hauptbedienungen der
Waage beschrieben.
Weitere Informationen zu der Waage der Serie BA finden Sie in den separaten Bedienungshandbüchern,aufgeführt unter „1.1. Detaillierte Handbücher“.

1.1. Detaillierte Handbücher
Detaillierte Handbücher stehen auf der A&D-Webseite ( https://www.aandd.jp ) zum Download zur
Verfügung:BA Series Instruction Manual (Bedienungshandbuch der Serie BA)
Dieses Bedienungshandbuch soll Ihnen dabei helfen, die Funktionen und Bedienungen der Serie BA
genau zu verstehen und sie vollständig zu nutzen.

Ionizer (static eliminator) Instruction Manual (Bedienungshandbuch des Ionisators(Vorrichtung zur Beseitigung statischer Ladung))
In diesem Handbuch werden die
Gebrauchsanweisungen beschrieben.

2.

Funktionen

desIonisators

AX-ION-25

sowie

die

Aufbau und Installation

Vorsicht
Die Analysenwaage der Serie BA ist ein Präzisionsinstrument und sollte vorsichtig ausgepackt
werden.
Es wird empfohlen, die Verpackungsmaterialien aufzubewahren, damit sie beim Transport der Waage
verwendet werden können.
Der Inhalt des Pakets variiert je nach Produkt. Beachten Sie auch die Darstellung auf den
Verpackungsinhalten und stellen Sie sicher, dass alle Teile enthalten sind.
Verbinden Sie den Wechselstromadapter erst mit der Waage, wenn die Waage zusammengebaut
und installiert ist.
Verwenden Sie den für die Waage angegebenen Wechselstromadapter.
Verbinden Sie den enthaltenen Wechselstromadapter nicht mit anderen Geräten.
Wenn Sie den falschen Adapter verwenden, funktionieren die Waage und andere Geräte eventuell
nicht richtig.
Stellen Sie sicher, dass der Wechselstromadapter getrennt ist, bevor Sie die Windschutzeinheit undden Ionisator anschließen.
Die Waage der Serie BA besteht aus der Wägeeinheit und der Anzeigeeinheit. Es ist nicht möglich,
nur eine der Einheiten zu ersetzen.

35

DE

2.1. Auspacken
Einheitliche Packungsinhalte der Serie BA

Schutzabdeckung derAnzeigeeinheit

Wägeeinheit

Bodenplatte desWindschutzes

Obere Tür desWindschutzes

Pinzette
Anzeigeeinheit
Kabel A (ca. 0,2 m)Kabel B(ca. 1,8 m)

USB-Kabel(ca. 1,8 m)

Wechselstromadapter(ca. 1,5 m)

Packungsinhalte für BA-6E / BA-6DE

Staubblech A

Windschutzring A

Kleines Glas-Windschutzset

Deckscheibe

Grundgestell

Rundschalen aus Aluminium zur Analyse (je 10 Stück)Windschutztüren
Große
Schale, Φ15,
0,8 mL

Mittlere
Schale, Φ12,
0,3 mL

Kleine Schale,
Φ8, 0,05 mL

Kabel C(ca. 0,6 m)

Ionisator

Packungsinhalte für BA-225 / BA-225D / BA-125D
Schale B, Φ25 Windschutzring B

Staubblech B

Frontglas des Windschutzes

Windschutztüren

36

Windschutzeinheit

DE

2.2. Zusammenbau
Das Modell BA-6E wird hier als Beispiel für den Zusammenbau verwendet. Die anderen Modelle können
auf gleiche Weise zusammengebaut werden.
Windschutzeinheit

Schritt 1

ZiehenSie
VerriegelungsgriffeWindschutzes heraus.

die
des

Schritt 1

Schritt 3

Schritt 2

Platzieren
Sie
dieWindschutzeinheit auf derWägeeinheit.

Schritt 3

DrückenSie
die
VerriegelungsgriffederWindschutzeinheit hinein, um
die Einheiten miteinander zu
verriegeln.

Schritt 4

Bauen Sie die Wagschalezusammen.
Bei den Modellen BA-6E/BA6DE bauen Sie die fünf Teile
der Wagschale in richtiger
Position,
wie
in
der
Querschnittansicht gezeigt,
zusammen.
Zusätzlich
zurStandardWaagschale(SchaleA)
könnenSie
auchdie
WaagschalefürFilterverwenden.
WeitereEinzelheiten finden Sie in dem
detaillierten
Bedienungshandbuch,aufgeführt
unter
„1.1.
Detaillierte Handbücher“.
Bei den Modellen BA-225/BA225D/BA-125D bauen Sie dievier Teile der Wagschale inrichtiger Position, wie in derQuerschnittansicht
gezeigt,
zusammen.

Schritt 2

Wägeeinheit

Für BA-225 / BA-225D / BA-125D
Für BA-6E / BA-6DE
Waagschale fürFilterSchale A

Windschutzring
A
Staubblech A

Bodenplatte desWindschutzes

Querschnittansicht
BA-6E/BA-6DE

Querschnittansicht
BA-225/BA-225D
BA-125D

37

Schale B

Windschutzring B

Staubblech B

DE
Schritt 5

Bauen
Sie
den
Glaswindschutz zusammen.
Bauen
Sie
den
Glaswindschutz BA-6E/BA6DE in der Wägekammer
zusammen. Richten Sie die
Seitenscheiben mit der Kerbean dem Grundgestell aus.

Deckscheibe
Seitenscheiben(3 Stück)Schritt 5

Grundgestell

Beispiel des kleinenWindschutzes mit offenerrechter Seite.

Kerbe
Grundgestell

Schritt 6

Schritt 77a7bFügen Sie die obere Tür desWindschutzes
vonderGriffseite aus ein.
Bauen
Sie
die
Windschutztüren an derlinken und rechten Seite
zusammen.
Fügen
Sie
die
Windschutztüren an derlinken und rechten Seite ein.
Sichern
Sie
Windschutztüren mit
Verriegelungen.

Schritt 7a
Schritt 6
Schritt 8b
Schritt 7b
Schritt 7a

Schritt 8a

die
den

Schritt 8

Bauen Sie das Frontglas desWindschutzes zusammen.

8a

Fügen Sie das Frontglas desWindschutzes ein.

8bSichern Sie das Frontglas
des Windschutzes mit den
Verriegelungen.

Schritt 8b
Schritt 7b

38

DE
Schritt 9

Verbinden Sie unter Verwendung des mitgelieferten Kabels A und Kabels B jeweils die„Wägeeinheit“ und „Windschutzeinheit“ sowie die „Anzeigeeinheit“ und die „Wägeeinheit“,
wobei Sie auf die Pfeilrichtung auf den einzelnen Kabeln achten sollten.
Vorsicht Achten Sie darauf, den Wechselstromadapter vor dem Verbinden abzuziehen.
Schritt 10 Verbinden Sie die Windschutzeinheit und den Ionisator mit dem mitgelieferten Kabel C.
Vorsicht Der Ionisator wird ausgeführt, wenn der Adapter mit der Waage verbunden ist.
Schritt 11 Verbinden Sie den enthaltenen Wechselstromadapter mit der Waage.
Masseanschluss

Anzeigeeinheit
Windschutzeinheit

Seriennummer
Ionisator

Kabel A
Schritt 9

Kabel B
Schritt 11

Wechselstromadapter
Wägeeinheit
Schritt 10

Kabel C

BA-225 / BA-225D / BA-125D

BA-6E / BA-6DE

39

DE

3.

Installationshinweise, Vorbereitung und Vorsichtshinweise

Stellen Sie die folgenden Installationsbedingungen her, um die volle Leistung der Waage nutzen zu können.
Stellen Sie die Waage in einer Umgebung ohne extreme Temperaturen und Feuchtigkeit auf. Die optimale
Betriebstemperatur beträgt etwa 20°C ± 2°C bei einer relativen Luftfeuchtigkeit von etwa 45 % bis 60 % RH.
Stellen Sie die Waage in einer staubfreien Umgebung auf.
Der Wägetisch sollte stabil sein. (Ideal ist ein Anti-Vibrationstisch oder ein Steintisch.)
Platzieren Sie die Waage auf einem horizontalen Tisch und stellen Sie sicher, dass er nicht geneigt ist.
Stellen Sie die Waage an einen stabilen Ort, der weder Vibrationen noch Stößen ausgesetzt ist. Am
besten geeignet sind die Eckräume im Erdgeschoss, weil sie weniger Vibrationen ausgesetzt sind.
Installieren Sie die Waage an einem Ort, wo sie nicht durch Heizkörper oder Klimaanlagen
beeinträchtigt wird.
Installieren Sie die Waage an einem Ort, wo sie keiner direkten
Sonneneinstrahlung ausgesetzt ist.
Stellen Sie die Waage nicht in der Nähe von Geräten auf, diemagnetische Felder erzeugen.
Wasserwaage
Nivellieren Sie die Waage mithilfe der Nivellierungsfüße undder Libelle in der Wasserwaage aus. Siehe „3.1.
Ausnivellieren der Waage“.
Wärmen Sie die Waage vor der Verwendung mindestens eine Stunde, bzw. mindestens vier Stundenbei BA-6E/BA-6DE lang auf, während der Wechselstromadapter mit der Stromquelle verbunden ist.
Stellen Sie die Empfindlichkeit der Waage vor dem ersten Gebrauch ein oder nachdem sie an einen anderen Ort gestellt
wurde, damit ein genaues Wiegen erzielt werden kann. Weitere Einzelheiten finden Sie in dem detaillierten
Bedienungshandbuch, aufgeführt unter „1.1. Detaillierte Handbücher“.
Vorsicht Stellen Sie die Waage nicht in einer Umgebung auf, in der entzündliche oder ätzende Gase auftreten.

3.1. Ausnivellieren der Waage
Nach unten

Nach oben

Die Waage ist
optimal ausnivelliert(Korrektes Wiegen)

Die Waage ist geneigt(Inkorrektes Wiegen). Nach unten

Roter Kreis

LibelleNach oben

Nivellieren Sie die Waage aus, indem Sie die Nivellierungsfüße anpassen, sodass sich die Blase derLibelle in der Wasserwaage im roten Kreis befindet.

Wenn sich die Libelle links befindet.

Wenn sich die Libelle rechts befindet.

Wenn sich die Libelle hinten befindet.

Wenn sich die Libelle vorn befindet.

Drehen Sie den Nivellierungsfuß vorn rechts im
Uhrzeigersinn.

Drehen Sie den Nivellierungsfuß vorn links imUhrzeigersinn.

Drehen Sie beide Nivellierungsfüße an der
Vorderseite gleichzeitig im Uhrzeigersinn.

Drehen Sie beide Nivellierungsfüße an der
Vorderseite gleichzeitig gegen den Uhrzeigersinn.

40

DE

4.

Anzeige und Tastenbetrieb (Grundlegende Bedienung)

Beleuchtete Anzeige
Datenspeichernummer
Reaktionsanzeige (leuchtet nach dem Start desWiegens 30 Sekunden lang auf)
Stoßanzeige

IR-Sensoranzeige
Voreingestellte Taramarke
Bruttomarke
Nettomarke
Verarbeitungsanzeige

Bruttonullmarke

Stabilisierungsanzeige

Intervall-AusgabemodusBereitschaftsanzeige
Einheitsanzeige

USB-Anschlussmarke
Bereitschaftsanzeige

Zeigt, gemessenen Wert, gespeicherte Daten,
Elementnamen an

Blinkende Anzeige
Angezeigte Datenmenge
Verarbeitungsanzeige
Hinweis zur automatischen
EmpfindlichkeitsanpassungIntervall-Ausgabemodus-Aktiv-Anzeige

Tasten
Rechter IR-Sensor

Linker IR-Sensor

SAMPLE -Taste
MODE -Taste

PRINT -Taste

CAL -Taste

TARE -Taste

ON/OFF -Taste

RE-ZERO -Taste

41

DE

Tastenbetrieb

Der Tastenbetrieb wirkt sich auf die Funktion der Waage aus.
Die normale Bedienung während der Messung erfolgt durch„Taste drücken und sofort wieder loslassen“ oder „Taste
drücken“. Verwenden Sie nicht die Variante „Taste drücken
und 2 Sekunden lang gedrückt halten“, wenn dies nicht
ausdrücklich erforderlich ist.
Taste

Gedrückt und losgelassen

Taste drücken
(Taste drücken undsofort wieder
loslassen.)

Taste drücken
und 2 Sekundenlang gedrückt
halten.

Gedrückt und gedrückt gehalten (für 2 Sekunden)

Schaltet die Anzeige ein und aus.
Wenn die Anzeige ausgeschaltet ist,erscheint die Bereitschaftsanzeige.
Wenn die Anzeige eingeschaltet ist, ist das
Wiegen möglich.
Bei der Kennwortfunktion wird „Eingabe desKennworts zu Beginn des
Wägevorgangs“ angezeigt, wenn die
Anzeige eingeschaltet wird. Weitere
Einzelheiten finden Sie in dem separaten
Bedienungshandbuch, aufgeführt unter „1.1.
Detaillierte Handbücher“.
Die ON:OFF -Taste ist zu jederzeit aktiv
und durch Drücken dieser Taste während
des Betriebs wird die Anzeige immer
ausgeschaltet.

Drücken Sie im Wägemodus die Taste, um
die Ziffer für die Lesbarkeit ein- undauszuschalten.
Drücken Sie im Zähl- und Prozentmodus die
Taste, um in den Probenspeichermodus zu
schalten.

Schaltet den IR-Sensor ein und aus.

Schaltet den Funktionstabellen-Modus ein.
Bitte lesen Sie auch das separate
Bedienungshandbuch, aufgeführt unter„1.1. Detaillierte Handbücher“.
Führt die
Wiederholbarkeitsprüfungsfunktion aus,
wenn die Taste nach der Anzeige desFunktionstabellenmenüs gedrückt und 2
Sekunden lang gedrückt gehalten wird.
Weitere Einzelheiten finden Sie in dem
separaten Bedienungshandbuch,aufgeführt unter „1.1. Detaillierte
Handbücher“.

Wechselt zwischen den in der
Funktionstabelle gespeicherten
Gewichtseinheiten.

Schaltet in den Modus „Ändern derWägegeschwindigkeit“.

Führt eine Empfindlichkeitsanpassung derWaage mit der internen Gewichts durch.

Zeigt weitere Elemente des Menüs zur
Empfindlichkeitsanpassung an.

42

DE

Speichert den Wägewert oder gibt ihn aus,
wenn er gemäß den
Funktionstabelleneinstellungen stabil ist.
(Daten werden in der Werkseinstellungausgegeben.)

Schaltet in den Modus zur Änderung derRegistrierungsnummer der Einheitsmasse im
Zählmodus.
Die folgenden Einstellungen sind mit der
Funktionstabelle möglich:
Gibt den „Title block (Titelblock)“ und den„End block (Endblock)“ für den GLP/GMPBericht aus.
Zeigt das Datenspeichermenü an.

Führt den Tara-Betrieb aus.

-

Stellt den Anzeigewert auf Null.

-

IR-Sensor (berührungsloser Sensor).
Er reagiert, wenn Sie Ihre Hand in seine
Nähe halten.
Das Öffnen und Schließen derWindschutztüren wird zugewiesen.
Für Einzelheiten siehe „5. IR-Sensoren und
Automatiktüren“.

-

43

DE

5.

IR-Sensoren und Automatiktüren

5.1. IR-Sensoren
Die Analysenwaagen der Serie BA sind mit IR-Sensoren ausgestattet, die einen Betrieb ermöglichen,ohne die Anzeige der Waage berühren zu müssen. In den Werkseinstellungen sind die IR-Sensoren links
und rechts der Anzeige zum Öffnen und Schließen der Windschutztüren zugewiesen. Durch Drücken und
Halten der ON:OFF -Taste (für etwa 2 Sekunden) können Sie die IR-Sensoren mit einer einzigen
Berührung ein- und ausschalten.

5.2. Automatiktüren
Die Analysenwaagen der Serie BA sind mit Automatiktüren ausgestattet, die es Ihnen ermöglichen, denWindschutz zu öffnen/schließen, ohne die Türen zu berühren.
Den IR-Sensoren links und rechts der Anzeige sind Einstellungen zum Öffnen/Schließen der jeweils
rechts und links befindlichen Windschutztüren zugewiesen.
In den Werkseinstellungen öffnet jede Windschutztür auf die Position, in der sie zuvor geöffnet wurde.
Es ist auch möglich, in der Funktionstabelle der Waage zu ändern, welchen Sensoren die
Windschutztüren zugewiesen sind, und dass die Türen vollständig oder teilweise geöffnet sein sollen.
Weitere Einzelheiten finden Sie in dem separaten Bedienungshandbuch, aufgeführt unter „1.1.
Detaillierte Handbücher“.

Öffnen der Windschutztür

Schritt 1. Wenn Sie die Windschutztür auf der linken Seite (oder der rechten Seite) öffnen möchten, wenn sie
geschlossen ist, dann halten Sie Ihre Hand über den rechten IR-Sensor (oder den linken IR-Sensor).
Schritt 2. Der Erkennungssummer ertönt und die Windschutztür auf der linken (oder rechten) Seite öffnet sich.

Schließen der Windschutztür

Schritt 1. Wenn Sie die Windschutztür auf der linken Seite (oder der rechten Seite) schließen möchten, wenn sie
geöffnet ist, dann halten Sie Ihre Hand über den rechten IR-Sensor (oder den linken IR-Sensor).
Schritt 2. Der Erkennungssummer ertönt und die Windschutztür auf der linken (oder rechten) Seite schließt sich.

Übersicht über die Funktionsweise
Der linke IR-Sensor öffnet/schließt die Tür auf der rechten Seite.

Beispiel von BA-6E/BA-6DE

Der rechte IR-Sensor öffnet/schließt
die Tür auf der linken Seite.

44

DE

6.

Empfindlichkeitsanpassung

6.1. Automatische Empfindlichkeitsanpassung
Die Empfindlichkeit der Waage wird automatisch mithilfe der internen Gewichts gemäß der Änderung der
Umgebungstemperatur, der eingestellten Zeit oder einer Intervallzeit angepasst. Weitere Einzelheiten
finden Sie in dem separaten Bedienungshandbuch, aufgeführt unter „1.1. Detaillierte Handbücher“.

6.2. Empfindlichkeitsanpassung mithilfe der internen Gewichts
Die Empfindlichkeitsanpassung mithilfe der internen Gewichts kann durch eine einzige Berührungausgeführt werden.
Schritt 1. Wenn sich nichts in der Wagschale befindet, stecken Sie den Wechselstromadapter der
Waage ein, um die Waage mindestens 1 Stunde lang (oder 4 Stunden bei BA-6E/BA-6DE)
aufzuwärmen.
Schritt 2. Drücken Sie die

CAL -Taste, um

anzuzeigen.

Schritt 3. Die Waage führt eine Empfindlichkeitsanpassung mithilfe der internen Gewichts durch.
Sorgen Sie dafür, dass die Waage weder Vibrationen noch Zugluft ausgesetzt ist.
Schritt 4. Nach der Empfindlichkeitsanpassung wird der „Empfindlichkeitsanpassungsbericht“ ausgegeben
und wie in der Funktionstabelle eingestellt gespeichert.
Weitere Einzelheiten finden Sie in dem separaten Bedienungshandbuch, aufgeführt unter „1.1.
Detaillierte Handbücher“.
Schritt 5. Nach dem Beenden kehrt die Waage automatisch in den Wägemodus zurück.

6.3. Empfindlichkeitsanpassung mithilfe eines externen Gewichts
Passen Sie die Empfindlichkeit der Waage mithilfe Ihres externen Gewichts an.
Weitere Einzelheiten finden Sie in dem separaten Bedienungshandbuch, aufgeführt unter „1.1.
Detaillierte Handbücher“.

45

DE

7.
AnzeigeFehleranzeigen (Fehlercodes)
CodeBeschreibung

Stabilitätsfehler
Der
Wägewert
ist
instabilund„Nullanzeige“,„Empfindlichkeitsanpassung“ usw. können nicht ausgeführt werden. Prüfen
Sie die Umgebung der Waagschale. Lesen Sie auch das separate
EC,E11
Bedienungshandbuch, aufgeführt unter „1.1. Detaillierte Handbücher“.
Verbessern Sie die Umgebung des Installationsortes (Vibration, Luftzüge,
statische Elektrizität usw.). Drücken Sie die CAL -Taste, um in den
Wägemodus zurückzukehren.

Außerhalb des Einstellbereichs
Der einzustellende Wert überschreitet den Einstellbereich.
Stellen Sie ihn erneut innerhalb des Einstellbereichs ein.

Fehlfunktion des internen Speicherelements der Waage
Wenn dieser Fehler weiterhin angezeigt wird, ist eine Reparatur
erforderlich. Bitte wenden Sie sich für die Reparatur an Ihren lokalen
Händler.

Fehler mit der internen Gewichts
Das Anwenden der internen Gewichts führt nicht zu einer Änderung desEC,E16 Massenwerts wie angegeben.
Vergewissern Sie sich, dass sich nichts auf der Waagschale befindet, undführen Sie den Vorgang von Anfang an durch.

Fehler mit der internen Gewichts

EC,E17 Der Mechanismus für die interne Masse funktioniert nicht richtig. Führen
Sie den Vorgang von Anfang an durch.

Anomalie der internen Speicherdaten der Waage
Wenn dieser Fehler weiterhin angezeigt wird, ist eine Reparatur
erforderlich.
Bitte wenden Sie sich für die Reparatur an Ihren lokalen Händler.

Wägeeinheit-Fehler
Die Daten der Wägeeinheit wurden nicht erhalten. Prüfen Sie die
Verbindung zwischen der Wägeeinheit und der Anzeigeeinheit und schalten
Sie den Strom dann aus und wieder ein.

Windschutzeinheit-Fehler
Die Kommunikation mit dem Windschutz ist nicht verbunden.
Ziehen Sie die Stromzufuhr ab und verbinden Sie das Kabel erneut
zwischen der Windschutzeinheit und der Wägeeinheit.

Seriennummer-Fehler
Die Seriennummern der Wägeeinheit und der Anzeigeeinheit stimmen nichtüberein.
Verbinden Sie mit der richtigen Kombination erneut.

46

DE
Anzeige

Code

Beschreibung

Fehler des Empfindlichkeitsanpassungswerts (Positiver Wert)
Das Empfindlichkeitsanpassungsgewicht ist zu schwer. Prüfen Sie die
EC,E20 Umgebung der Waagschale. Prüfen Sie den Massewert derEmpfindlichkeitsanpassung.
Drücken Sie die CAL -Taste, um in den Wägemodus zurückzukehren.

Fehler des Empfindlichkeitsanpassungswerts (Negativer Wert)
Das Empfindlichkeitsanpassungsgewicht ist zu leicht. Prüfen Sie die
EC,E21 Umgebung der Waagschale. Prüfen Sie den Massewert derEmpfindlichkeitsanpassung.
Drücken Sie die CAL -Taste, um in den Wägemodus zurückzukehren.

Überlastfehler
Eine Probe, die schwerer ist als die zulässige Kapazität der Waage, wurde
auf die Waagschale gelegt. Nehmen Sie den Gegenstand von der
Waagschale.

Waagschalen-Fehler
Die Masse ist zu leicht. Prüfen Sie, dass die Wagschale korrekt installiert
ist. Stellen Sie die Wagschale korrekt ein. Kalibrieren Sie die Waage.

Probenmassen-Fehler
Die Probe ist zu leicht, um als Probemasse für den Zähl- oder
Prozentmodus gespeichert zu werden. Die Probe kann nicht verwendet
werden.

Einheitsmasse-Fehler
Die Probeneinheitsmasse ist für den Zählmodus zu leicht. Diese zu speichern
und im Zählmodus zu benutzen, könnte zu einem Zählfehler führen. Fügen Sie
weitere Proben hinzu, um die angegebene Zahl zu erreichen, und drücken Sie
die PRINT -Taste. Wenn Sie die PRINT -Taste drücken, ohne weitere
Proben hinzuzufügen, schaltet die Waage in den Zählmodus. Für eine genaue
Zählung sollten Sie weitere Proben hinzufügen.

Uhr-Batteriefehler
Die Backup-Batterie der Uhr ist leer. Drücken Sie eine beliebige Taste undstellen Sie die Uhrzeit und das Datum ein. Selbst wenn die Ersatzbatterieder Uhr leer ist, funktioniert die Uhr- und Kalenderfunktion normal, solange
die Waage mit dem Wechselstromadapter mit Strom versorgt wird. Wenndieser Fehler häufig auftritt, wenden Sie sich an Ihren lokalen Händler.

Spannungsfehler in der Stromversorgung
Die vom Wechselstromadapter gelieferte Spannung ist nicht normal.
Bitte prüfen Sie, ob das Problem bei dem mit der Waage mitgelieferten
Wechselstromadapter liegt.

47

DE
Anzeige

Code

Beschreibung

Wiederholbarkeitsfehler
Die Standardabweichung (SD) der Wiederholbarkeit überschreitet 50
Ziffern. Überprüfen Sie die Installationsumgebung der Waage.
In der Wiederholbarkeitsanzeige.
In der Mindestwägewertanzeige (Referenzwert).
Weitere Einzelheiten finden Sie in dem separaten Bedienungshandbuch,aufgeführt unter „1.1. Detaillierte Handbücher“.

Voller Speicher
Blinkt (abwechselnd)

Die Anzahl der gespeicherten Wägewerte hat die Obergrenze erreicht. Umneue Wägewerte zu speichern, müssen Daten gelöscht werden. Weitere
Einzelheiten finden Sie in dem separaten Bedienungshandbuch „1.1.
Detaillierte Handbücher“.

Voller Speicher
Blinkt (abwechselnd)

Die gespeicherte Empfindlichkeitsanpassungs- und Kalibriertest-Historie hat
eine Anzahl von 50 erreicht. Um ein neues Ergebnis zu speichern, wird dieälteste Historie gelöscht. Weitere Einzelheiten finden Sie in dem separaten
Bedienungshandbuch „1.1. Detaillierte Handbücher“.

Kommunikationsfehler
EC,E00 Bei der Übertragung trat ein Protokollfehler auf.
Prüfen Sie Format, Baudrate und Parität.

Fehler Undefinierter Befehl
EC,E01 Ein nicht definierter Befehl wurde gefunden.
Prüfen Sie den übertragenen Befehl.

Nicht bereit
Der empfangene Befehl kann nicht ausgeführt werden:
EC,E02 (z. B.) Q-Befehl wurde außerhalb des Wägemodus empfangen.
(z. B.) Q-Befehl wurde beim erneuten Nullstellen empfangen.
Passen Sie die Verzögerungszeit für die Befehlsübertragung an.

Timeout-Fehler
EC,E03Ist der Timeout-Parameter auf
eingestellt, gab es eine
Wartezeit von etwa 1 Sekunde oder mehr beim Erhalten derBefehlszeichen. Überprüfen Sie die Kommunikation.

Zeichenlängenfehler
EC,E04 Die Anzahl der Zeichen in dem erhaltenen Befehl hat die Grenzeüberschritten. Überprüfen Sie den zu übertragenden Befehl.

Formatfehler
Die Beschreibung des erhaltenen Befehls ist falsch:
EC,E06 (z. B.) Die Anzahl der Ziffern der Zahlenwerte ist inkorrekt.
(z. B.) Unter den Zahlenwerten befinden sich Buchstabenzeichen.
Prüfen Sie den übertragenen Befehl.

Parameter-Einstellungsfehler
EC,E07 Die Daten des empfangenen Wertes übersteigen den zulässigen Bereich.
Überprüfen Sie den Einstellbereich des Zahlenwerts des Befehls.

48

DE

7.1. Weitere Anzeigen
Dies ist der Hinweis zur automatischen Empfindlichkeitsanpassung (dieMarkierung blinkt).
Wenn die Waage mehrere Minuten lang nicht benutzt wird und diese Markierung
blinkt, führt die Waage automatisch eine Empfindlichkeitsanpassung mithilfe derinternen Gewichts durch. (Die Blinkzeit hängt von der Betriebsumgebung ab.)
Tipp

Obwohl die Waage verwendet werden kann, wenn diese Markierung blinkt,
empfehlenwireineVerwendungnachdemAbschließen
derEmpfindlichkeitsanpassung jedoch nur, um die Wägegenauigkeit aufrechtzuerhalten.

49

DE

8.

Ionisator (AX-ION-25)

Der Ionisator AX-ION-25 ist standardmäßig bei BA-6E/BA-6DE enthalten. Der Ionisator beseitigt statische
Elektrizität durch Bestrahlung des Zielobjekts mit positiven oder negativen Ionen, die von vier
Entladungselektroden durch Gleichstrom-Koronaentladung erzeugt werden. Normalerweise neigen
Isolatoren wie Pulver, Filter und Wägepapier dazu, sich bei einer Luftfeuchtigkeit von 45 % RH oder weniger
aufzuladen, und beim Wägen kann ein Fehler von einigen Milligramm auftreten. Durch Entfernen derstatischen Aufladung des Wägeobjekts mithilfe des Ionisators ist es möglich, den Fehler des Wägewerts
aufgrund der Aufladung zu beseitigen, und es kann eine korrekte Wägung durchgeführt werden. Weitere
Einzelheiten finden Sie in dem separaten Bedienungshandbuch unter „1.1. Detaillierte Handbücher“.

8.1. Verwenden des Ionisators
Schritt 1. Stellen Sie sicher, dass der Wechselstromadapter der Waage nicht verbunden ist, undverbinden Sie dann die Waage und den Ionisator (wie beschrieben unter „2. Aufbau undInstallation“).
Schritt 2. Verbinden Sie den Wechselstromadapter, stecken Sie ihn ein und schalten Sie den Strom ein.
Die Stromleuchte des Ionisators leuchtet auf.
Schritt 3. Platzieren Sie das Zielobjekt innerhalb des Bereichs der statischen Eliminierung.
Schritt 4. Halten Sie Ihre Hand über den IR-Sensor (berührungsloser Infrarot-Näherungssensor) an der
Vorderseite des Ionisators, um die statische Eliminierung zu starten. Die ION-Leuchte, dieanzeigt, dass die statische Eliminierung ausgeführt wird, leuchtet auf.
Schritt 5. Beachten Sie die nachfolgende Abbildung für den Wirkungsbereich der statischen
Eliminierung und führen Sie die statische Eliminierung durch. In den Werkseinstellungen,wenn 3 Sekunden nach dem Beginn der statischen Eliminierung vergangen sind, erlischt die
ION-Leuchte, die anzeigt, dass die statische Eliminierung ausgeführt wird, und die statische
Eliminierung hält an.

IR-SENSOR (berührungsloser Infrarot-Näherungssensor)
Stromleuchte (leuchtet auf, wenn der Stromeingeschaltet ist)
ION-Leuchte (leuchtet während der statischen
Eliminierung auf)

50

Ionisator(AX-ION-25)

DE
Wirkungsbereich der statischen Eliminierung
Draufsicht

Wirkungsbereich
0 cm
Seitenansicht

5 cm
Entfernung

7 cm

10 cm

Wirkungsbereich

1 Sek. 1,5 Sek.
Zeit der statischen Eliminierung

51

3 Sek.

DE

9.

Spezifikationen
BA-6E

Wägekapazität
Maximale Anzeige

BA-6DE
6,2 g

6,2 g

2,1 g
6,20008 g

6,200084 g

2,100009 g
0,01 mg

Lesbarkeit

0,001 mg

Wiederholbarkeit(Standardabweichung)

0,0030 mg (6 g)

0,01 mg (6 g)

0,0010 mg (1 g)

0,0025 mg (1 g)

Linearität±0,010 mg

0,001 mg

±0,02 mg
±0,010 mg

Anzeige

Flüssigkristallanzeige (monochrom)

Bedienung

Tastenschalter

Kommunikation

RS-232C (Drucker, PLC, usw.), USB Typ Mini-B (PC)
Klinkenstecker (externer Schalter)

BA-225
Wägekapazität

Maximale Anzeige

Lesbarkeit
Wiederholbarkeit(Standardabweichung)Linearität

220 g

220,00084 g

0,01 mg
0,03 mg (200 g)
0,015 mg

BA-225D

BA-125D

220 g

120 g

51 g

51 g

220,0008 g

120,0008 g

51,00009 g

51,00009 g

0,1 mg0,1 mg0,01 mg

0,01 mg

0,1 mg (200 g)(50 g)

0,025 mg

±0,15 mg

(50 g)±0,2 mg

0,1 mg (100 g)
0,030 mg

±0,2 mgAnzeige

Flüssigkristallanzeige (monochrom)

Bedienung

Tastenschalter

Kommunikation

RS-232C (Drucker, PLC, usw.), USB Typ Mini-B (PC)
Klinkenstecker (externer Schalter)

52(50 g)

ES

1.

Guía de inicio rápido

Gracias por adquirir una báscula analítica de la serie BA de A&D. En esta guía de inicio rápido se describe
el proceso de instalación de la báscula de la serie BA, así como las funciones básicas y las operaciones
principales de la báscula.
Para obtener más información sobre la báscula de la serie BA, consulte los manuales de instrucciones
por separado mencionados en “1.1. Manuales detallados”.

1.1. Manuales detallados
Tiene a su disposición manuales detallados que puede descargar en el sitio web de A&D( https://www.aandd.jp ):BA Series Instruction Manual (Manual de instrucciones de la serie BA)
Este manual le ayudará a entender las funciones y operaciones de la serie BA con más detalle y aaprovecharlas al máximo.

Ionizer (static eliminator) Instruction Manual (Manual de instrucciones del ionizador(eliminador estático))
En este manual se describen las funciones del ionizador AX-ION-25 y sus instrucciones de uso.

2.

Montaje e instalación

Precaución
La báscula analítica de la serie BA es un instrumento de precisión. Desembálela con cuidado.
Se recomienda conservar el material de embalaje con el fin de utilizarlo para transportar la básculaen caso necesario.
El contenido del paquete varía dependiendo del producto. Consulte las ilustraciones del contenidodel paquete para confirmar que contiene todos los artículos necesarios.
Conecte el adaptador de CA a la báscula únicamente después de haberla montado e instalado.
Utilice el adaptador de CA específico indicado para la báscula.
No conecte el adaptador de CA incluido a otros dispositivos.
Si utiliza un adaptador de CA incorrecto, la báscula y el resto del equipo podrían no funcionar bien.
Asegúrese de que el adaptador de CA esté desenchufado antes de conectar la unidad de cabina
cortavientos y el ionizador.
La báscula analítica de la serie BA se compone de una unidad de pesaje y una unidad devisualización. No es posible sustituir las unidades por separado.

53

ES

2.1. Desembalado
Contenido del paquete común para toda la serie BA
Tapa protectora de launidad de visualización

Unidad de pesaje

Placa inferior de la Puerta superior de lacabina cortavientos cabina cortavientos

PinzasUnidad devisualización
Cable A (aprox. 0,2 m)

Cable B
(aprox. 1,0 m)

Cable USB
(aprox. 1,8 m)

Adaptador de CA(aprox. 1,5 m)

Este documento

Contenido del paquete para los modelos BA-6E / BA-6DE

Cristal delantero de la Unidad de cabina
cortavientos
cabina cortavientos

Placa antipolvo A

Panel superior Soporte de base

Plato grande,Φ 15, 0,8 ml

Cable C
(aprox. 0,6 m)

Ionizador

Contenido del paquete para los modelos BA-225 / BA-225D / BA-125D

Cristal delantero de lacabina cortavientos

Puertas de lacabina cortavientos

54

Unidad de cabina cortavientos

ES

2.2. Montaje
Se muestra el montaje del modelo BA-6E a modo de ejemplo. El resto de modelos se montan de lamisma forma.
Unidad de cabina cortavientos

Paso 1 Retire los mecanismos debloqueodelacabinacortavientos.

Paso 1

Paso 2

Paso 2 Coloque la unidad de cabina
cortavientos en la unidad depesaje.

Paso 3 Acople los mecanismos debloqueodelacabinacortavientos para fijar ambas
unidades.

Paso 4 Monte el plato de pesaje.
Para los modelos BA-6E / BA6DE, consulte la vista en
sección transversal para montarlas cinco piezas del plato depesaje en la posición correcta.
Además del plato de pesaje
estándar (Plato A), también
puede utilizar el plato de pesajepara filtros. Para obtener más
información, consulte el manualdeinstrucciones
detalladomencionado en “1.1. Manuales
detallados”.
Para los modelos BA-225 / BA225D / BA-125D, consulte lavista en sección transversal
para montar las cuatro piezasdel plato de pesaje en la
posición correcta.

Paso 3

Unidad depesajePara BA-225 / BA-225D / BA-125D
Para BA-6E / BA-6DE
Plato A

Placa inferior
de la cabina
cortavientos

Vista en sección
transversal de
BA-6E / BA-6DE

Vista en sección transversal deBA-225 / BA-225D
BA-125D

55

Plato BES
Paso 5 Monte la cabina cortavientos de Panel superior
cristal.
Paneles laterales
Monte la cabina cortavientos de (3 piezas)
cristal de los modelos BA-6E /BA-6DE en la cámara de pesaje.
Alinee los paneles laterales conla muesca del soporte de base.
Paso 5

Soporte de base

Ejemplo de cabina cortavientos
pequeña con el lado derecho
abierto.

Panel lateral

MuescaSoporte de base

Paso 6 Inserte la puerta superior de lacabina cortavientos por losmecanismos de bloqueo.
Paso 7 Monte las puertas de la cabina
cortavientos de los lados
derecho e izquierdo.
7a7bInserte las puertas de la cabina
cortavientos de los lados
derecho e izquierdo.

Paso 7aPaso 6
Paso 8b
Paso 7b
Paso 7aPaso 8a

Fije las puertas de la cabina
cortavientos con los cierres.

Paso 8b

Paso 8 Monte el cristal delantero de lacabina cortavientos.
8a

Inserte el cristal delantero de lacabina cortavientos.

8b

Fije el cristal delantero de lacabina cortavientos con loscierres.

Paso 7b56

ES
Paso 9

Utilice los cables A y B incluidos para conectar la “unidad de pesaje” y la “unidad de cabina
cortavientos”, y la “unidad de visualización” y la “unidad de pesaje” respectivamente; para ello,preste atención a la dirección de la flecha de cada cable.
Precaución
Asegúrese de desenchufar el adaptador de CA antes de llevar a cabo la conexión.
Paso 10 Conecte la unidad de cabina cortavientos y el ionizador con el cable C incluido.
Precaución
El ionizador funciona con el adaptador de CA conectado a la báscula.
Paso 11 Conecte el adaptador de CA incluido a la báscula.
Terminal de
Unidad de visualización
Unidad de cabina
cortavientos

conexión a tierra

Número de serie
Ionizador.

Cable A
Paso 9Cable B
Paso 11Adaptador de CA
Unidad de pesaje
Paso 10

Cable C

BA-225 / BA-225D / BA-125D

BA-6E / BA-6DE

57

ES

3. Consideraciones durante la instalación, preparativos yprecauciones
A continuación se describen las condiciones de instalación óptimas para obtener el máximo rendimiento de la báscula.
Coloque la báscula en un entorno cuya temperatura y humedad no sean excesivos. La temperatura de funcionamientoóptima es de aproximadamente 20 °C ± 2 °C con una humedad relativa de alrededor del 45 al 60 ％.
Coloque la báscula en un lugar sin polvo.
La mesa de pesaje debería ser sólida. (las mesas antivibración o las mesas de piedra son ideales).
Coloque la báscula sobre una mesa horizontal y asegúrese de que no esté inclinada.
Coloque la báscula en un lugar estable; evite las vibraciones y los golpes. El lugar idóneo son las esquinas
de las salas situadas en la primera planta, ya que son zonas menos propensas a las vibraciones.
Coloque la báscula en un lugar en el que no se vea afectada por calefactores ni aires acondicionados.
Coloque la báscula en un lugar en el que no esté expuesta ala luz solar directa.
Coloque la báscula alejada de equipos que produzcan
campos magnéticos.
Nivel de burbuja
Nivele la báscula con las ruedas niveladoras y el nivel deburbuja. Consulte “3.1. Cómo ajustar el nivel de la báscula”.
Antes de usar la báscula, asegúrese de calentarla durante al menos una hora (o un mínimo de cuatrohoras para los modelos BA-6E / BA-6DE) con el adaptador de CA conectado a la alimentación.
Ajuste la sensibilidad de la báscula antes de usarla por primera vez o tras moverla de ubicación paraobtener resultados precisos. Para obtener más información, consulte el manual de instrucciones
detallado mencionado en “1.1. Manuales detallados”.
Precaución
No coloque la báscula en lugares donde haya gases inflamables o corrosivos.

3.1. Cómo ajustar el nivel de la báscula
Subir

Bajar

Bajar

Círculo rojo

Burbuja

Subir

Nivele la báscula ajustando las ruedas niveladoras de forma que la burbuja del nivel de burbuja
quede centrada en el círculo rojo.

Si la burbuja está a la izquierda.

Si la burbuja está a la derecha.

Si la burbuja está detrás.

Si la burbuja está delante.

Gire la rueda niveladora delantera derecha en elsentido de las agujas del reloj.

Gire la rueda niveladora delantera izquierda enel sentido de las agujas del reloj.

Gire las dos ruedas niveladoras delanteras a lavez en el sentido de las agujas del reloj.

Gire las dos ruedas niveladoras delanteras a lavez en el sentido contrario a las agujas del reloj.

58

ES

4.

Pantalla y panel de teclas (funcionamiento básico)

Visualización de elementos iluminados
Número de memoria de datos
Indicador de respuesta (se enciende durante 30
segundos cuando se inicia el pesaje)
Indicador de impactos

Indicador de sensor de IR
Marca de tara predefinida
Marca bruta
Marca neta

Marca de cero bruto

Indicador de espera

Visualización de unidades
Visualización del valor de pesaje, losdatos almacenados y otros elementos

Visualización intermitente
Datos del número que se muestranIndicador de procesamiento
Aviso de ajuste de sensibilidad
automático
Indicador de activación del modo de salida de intervalos

TeclasSensor de IR
derechoSensor de IR
izquierdo

Tecla

SAMPLE

Tecla

MODE

Tecla

PRINT

Tecla

CAL

Tecla

TARE

Tecla

ON/OFF

Tecla

RE-ZERO

59

ES

Funcionamiento de las teclas

El funcionamiento de las teclas afecta al de la báscula. El
funcionamiento normal de las teclas durante la medición es“Pulse la tecla y suéltela inmediatamente” o “Pulse la tecla”.
No “Mantenga pulsada la tecla (durante 2 segundos)” amenos que así se requiera.
Tecla

Cuando se pulsa y se suelta

Pulsar la tecla(pulse la tecla y suéltela
inmediatamente).

Mantenerpulsada la tecla
2 segundos.

Cuando se mantiene pulsada (durante 2 segundos)

Enciende/apaga la pantalla.
Cuando se apaga la pantalla, solo se
muestra el indicador de espera.
Cuando se enciende la pantalla, se puede
realizar el pesaje.
Con la función de contraseña activada, se
mostrará “Introduzca la contraseña al iniciodel pesaje” cuando se encienda la pantalla.
Para obtener más información, consulte el
manual de instrucciones por separado
mencionado en “1.1. Manuales detallados”.
La tecla ON:OFF puede activarse en
cualquier momento. Si se pulsa con labáscula en funcionamiento, se apagará lapantalla.

En el modo de pesaje, pulse la tecla paraencender y apagar la legibilidad de losdígitos.
En el modo de recuento o de porcentaje,
pulse la tecla para entrar en el modo dealmacenamiento de muestras.

Enciende y apaga el sensor de IR.

Entra en el modo de tabla de funciones.
Consulte el manual de instrucciones por
separado mencionado en “1.1. Manuales
detallados”.
Ejecuta la función de comprobación derepetibilidad cuando se mantiene pulsada
durante otros 2 segundos después de que
se muestre el menú de la tabla defunciones. Para obtener más información,
consulte el manual de instrucciones por
separado mencionado en “1.1. Manuales
detallados”.

Cambia las unidades de pesaje
almacenadas en la tabla de funciones.

Entra en el modo “Cambiar velocidad depesaje”.

Lleva a cabo el ajuste de sensibilidad de labáscula con el peso interno.

Muestra otros elementos del menú de ajustede sensibilidad.

Almacena y envía el valor de pesaje cuando
es estable conforme a la configuración de latabla de funciones.
(Los datos se envían conforme a los ajustes
de fábrica.)

Entra en el modo para cambiar el número deregistro de la masa de la unidad, el modo derecuento.
Es posible ajustar la siguiente configuración
con la tabla de funciones:
Envía el “Title block (Bloque de título)” y el“End block (Fin del bloque)” para losinformes de GLP y GMP.
Se muestra el menú de memoria de datos.

60

ES
Lleva a cabo la operación de tara.

-

Restablece el valor mostrado a cero.

-Sensor de IR (sensor sin contacto).
Reacciona al acercar la mano.
Tiene asignadas la apertura y el cierre de lapuerta de la cabina cortavientos.
Para obtener más información, consulte “5.
Sensores de IR y puertas automáticas”.

-

61

ES

5.

Sensores de IR y puertas automáticas

5.1. Sensores de IR
Las básculas analíticas de la serie BA están equipadas con sensores de IR que permiten su
funcionamiento sin tocar directamente la pantalla de la báscula. En los ajustes de fábrica, los sensoresde IR izquierdo y derecho de la pantalla están asignados a la apertura y el cierre de las puertas de lacabina cortavientos. Manteniendo pulsada la tecla ON:OFF (durante unos 2 segundos), puede
encender o apagar los sensores de IR con un toque.

5.2. Puertas automáticas
Las básculas analíticas de la serie BA están equipadas con puertas automáticas que permiten abrir/cerrar
la cabina cortavientos sin tocar directamente las puertas.
La configuración de los sensores de IR izquierdo y derecho de la pantalla está asignada a la apertura yel cierre de las puertas derecha e izquierda de la cabina cortavientos respectivamente.
Conforme a los ajustes de fábrica, cada puerta de la cabina cortavientos se abre hasta la posición en laque estaba abierta previamente.
También se puede cambiar a qué sensores se asignan las puertas de la cabina cortavientos y configurar
que las puertas se queden total o parcialmente abiertas en la tabla de funciones de la báscula.
Para obtener más información, consulte el manual de instrucciones por separado mencionado en “1.1.
Manuales detallados”.

Apertura de las puertas de la cabina cortavientos

Paso 1. Si desea abrir la puerta de la cabina cortavientos del lado izquierdo (o del lado derecho) cuando
esté cerrada, ponga la mano sobre el sensor de IR derecho (o el sensor de IR izquierdo).
Paso 2. Se escuchará un sonido de confirmación de detección y la puerta de la cabina cortavientos dellado izquierdo (o derecho) se abrirá.

Cierre de las puertas de la cabina cortavientos

Paso 1. Si desea cerrar la puerta de la cabina cortavientos del lado izquierdo (o del lado derecho)
cuando esté abierta, ponga la mano sobre el sensor de IR derecho (o el sensor de IR izquierdo).
Paso 2. Se escuchará un sonido de confirmación de detección y la puerta de la cabina cortavientos dellado izquierdo (o derecho) se cerrará.

62

ES

Resumen del funcionamiento
El sensor de IR izquierdo abre/cierra la puertadel lado derecho.

Se muestra el modelo BA-6E /BA-6DE a modo de ejemplo

El sensor de IR derecho abre/cierra lapuerta del lado izquierdo.

63

ES

6.

Ajuste de sensibilidad

6.1. Ajuste de sensibilidad automático
La sensibilidad de la báscula se ajusta de forma automática con el peso interno en función de los cambios
de temperatura ambiente, el tiempo de ajuste o el tiempo de intervalo. Para obtener más información,
consulte el manual de instrucciones por separado mencionado en “1.1. Manuales detallados”.

6.2. Ajuste de sensibilidad con el peso interno
El ajuste de sensibilidad con el peso interno puede llevarse a cabo con un toque.
Paso 1. Con el plato de pesaje vacío, enchufe la báscula con el adaptador de CA para calentarla almenos 1 hora (o 4 horas en el caso de los modelos BA-6E / BA-6DE).
Paso 2. Pulse la tecla

CAL hasta que se visualice.

Paso 3. La báscula lleva a cabo el ajuste de sensibilidad con el peso interno. No permita que la báscula
se vea afectada por vibraciones o corrientes de aire.
Paso 4. Tras el ajuste de sensibilidad, se enviará un “registro del ajuste de sensibilidad” que se guardará
como configuración en la tabla de funciones.
Para obtener más información, consulte el manual de instrucciones por separado mencionado
en “1.1. Manuales detallados”.
Paso 5. Finalmente, la báscula volverá automáticamente al modo de pesaje.

6.3. Ajuste de sensibilidad con un peso externo
Ajuste la sensibilidad de la báscula con un peso externo.
Para obtener más información, consulte el manual de instrucciones por separado mencionado en “1.1.
Manuales detallados”.

64

ES

7.

Visualización de errores (códigos de error)

Visualización

CódigoDescripción

Error de estabilidad
El valor de pesaje es inestable y no pueden ejecutarse la “visualización acero”, el “ajuste de sensibilidad”, etc. Compruebe el entorno del plato.
EC,E11 Consulte el manual de instrucciones por separado mencionado en “1.1.
Manuales detallados”. Mejore las condiciones del entorno de instalación(vibraciones, corrientes de aire, electricidad estática, etc.). Para volver almodo de pesaje, pulse la tecla CAL .

Fuera del intervalo de valores de ajuste
El valor configurado está fuera del intervalo de ajustes.
Introduzca un valor que esté dentro del intervalo de ajustes.

Mal funcionamiento del elemento de la memoria interna de la báscula
Si el error se sigue visualizando, será necesario realizar una reparación.
Póngase en contacto con su proveedor local para solicitar la reparación.

Error del peso interno
La aplicación del peso interno no produce un cambio en el valor del peso
EC,E16 tal como se especifica.
Confirme que no haya nada en el plato y vuelva a realizar la operación
desde el principio.

Error del peso interno

EC,E17 El mecanismo de aplicación del peso interno no funciona correctamente.
Vuelva a realizar la operación desde el principio.

Anomalía en los datos de la memoria interna de la báscula
Si el error se sigue visualizando, será necesario realizar una reparación.
Póngase en contacto con su proveedor local para solicitar la reparación.

Error de la unidad de pesaje
No se han recibido datos de la unidad de pesaje. Compruebe la conexión
entre la unidad de pesaje y la unidad de visualización y luego apague yvuelva a encender la báscula.

Error de la unidad de cabina cortavientos
No se establece comunicación con la cabina cortavientos.
Desenchufe la alimentación y vuelva a conectar el cable entre la unidad decabina cortavientos y la unidad de pesaje.

Error de número de serie
Los números de serie de la unidad de pesaje y la unidad de visualización no coinciden.
Vuelva a conectar la combinación de unidades correcta.

Error del peso de ajuste de sensibilidad (valor positivo)
EC,E20El peso de ajuste de sensibilidad es demasiado alto. Compruebe el entornodel plato. Compruebe el valor del peso del ajuste de sensibilidad.
Para volver al modo de pesaje, pulse la tecla CAL .

65

ESVisualización

CódigoDescripción

Error del peso de ajuste de sensibilidad (valor negativo)
EC,E21El peso de ajuste de sensibilidad es demasiado bajo. Compruebe el entornodel plato. Compruebe el valor del peso del ajuste de sensibilidad.
Para volver al modo de pesaje, pulse la tecla CAL .

Error de sobrecarga
Se ha colocado en el plato una muestra que supera la capacidad de pesajede la báscula. Retire el objeto del plato.

Error del plato de pesaje
El valor de la masa es demasiado bajo. Compruebe que el plato de pesaje
esté colocado correctamente. Configure el plato de pesaje correctamente.
Calibre la báscula.

Error de masa de la muestra
La muestra es demasiado ligera para almacenarse como masa de la muestra en elmodo de recuento o de porcentaje. No puede utilizarse esa muestra.

Error de masa de la unidad
La masa de la unidad de muestra para el modo de recuento es demasiado
ligera. Puede provocar un error de recuento si se almacena y se usa pararealizar recuentos. Añada muestras para alcanzar el número especificado
y pulse la tecla PRINT . Si pulsa la tecla PRINT sin añadir ninguna
muestra, la báscula pasará al modo de recuento. No obstante, asegúrese
de añadir muestras para obtener un recuento preciso.

Error de la batería del reloj
La batería de reserva del reloj se ha agotado. Pulse cualquier tecla y ajustela fecha y la hora. Incluso si se ha agotado la batería de reserva del reloj,
las funciones de reloj y calendario funcionarán con normalidad siempre quela báscula esté conectada al adaptador de CA. Si este error aparece con
frecuencia, póngase en contacto con su proveedor local.

Fallo de tensión de alimentación
La tensión suministrada por el adaptador de CA es anómala.
Compruebe si el problema está en el adaptador de CA suministrado con la báscula.

Error de repetibilidad
La desviación estándar (SD) de la repetibilidad excede de 50 dígitos. Revise
el entorno de instalación de la báscula.
en la visualización de repetibilidad.
en la visualización del valor de pesaje mínimo (valor de referencia).
Para obtener más información, consulte el manual de instrucciones por separado
mencionado en “1.1. Manuales detallados”.

Memoria llena
Parpadeo (alternado)

El número de valores de pesaje almacenados ha llegado al límite superior.
Para almacenar nuevos valores de pesaje es necesario eliminar datos. Paraobtener más información, consulte el manual de instrucciones por separado
mencionado en “1.1. Manuales detallados”.

66

ES
Visualización

CódigoDescripción

Memoria llena

Parpadeo (alternado)

El historial de pruebas de calibración/ajuste de sensibilidad almacenado ha
llegado a los 50 resultados. Se eliminará el historial más antiguo para poderalmacenar nuevos resultados. Para obtener más información, consulte el
manual de instrucciones por separado mencionado en “1.1. Manuales
detallados”.

Error de comunicación
EC,E00 Ha habido un error del protocolo de comunicaciones.
Compruebe el formato, la velocidad en baudios y la paridad.

Error de comando no definido
EC,E01 Se ha detectado un comando no definido.
Compruebe el comando transmitido.

La báscula no está lista
No se puede ejecutar el comando recibido:
EC,E02 (Ejemplo) Se ha recibido un comando Q cuando la báscula no estaba en el modo de pesaje.
(Ejemplo) Se ha recibido un comando Q cuando se estaba procesando la puesta a cero.
Ajuste la sincronización de la transmisión del comando.

Error de tiempo límite
EC,E03Con el parámetro de tiempo límite ajustado en, se ha producido untiempo de espera de aproximadamente 1 segundo o más mientras se recibían
caracteres de comando. Compruebe la comunicación.

Error de longitud de caracteres
EC,E04 El número de caracteres del comando recibido excede el límite.
Compruebe el comando que debe transmitirse.

Error de formato
La descripción del comando recibido es incorrecta:
EC,E06 (Ejemplo) El número de dígitos de valores numéricos es incorrecto.
(Ejemplo) Hay caracteres alfabéticos entre los valores numéricos.
Compruebe el comando transmitido.

Error del valor de ajuste
EC,E07 El valor del comando recibido ha excedido el límite permitido.
Compruebe el intervalo de ajustes del valor numérico del comando.

7.1. Otras visualizaciones
Este es el aviso de ajuste de sensibilidad automático (marcaintermitente).
Si no se utiliza la báscula durante varios minutos con esta marca parpadeando, esta
lleva a cabo el ajuste de sensibilidad automáticamente con el peso interno. (Elperiodo de parpadeo depende del entorno de funcionamiento.)
Consejo

Aunque es posible seguir utilizando la báscula cuando esta marca parpadea, serecomienda utilizarla una vez que haya finalizado el ajuste de sensibilidad paramantener la precisión de pesaje.

67

ES

8.

Ionizador (AX-ION-25)

El ionizador AX-ION-25 viene de serie con los modelos BA-6E / BA-6DE. El ionizador elimina laelectricidad estática irradiando iones positivos o negativos al objeto generados por cuatro electrodos dedescarga mediante descarga de corona de CC. Normalmente, ciertos aislantes como los polvos, losfiltros y el papel de pesaje tienden a cargarse con una HR del 45 % o menos, por lo que puede producirse
un error de unos cuantos miligramos durante el pesaje. Suprimiendo la estática del objeto que debe
pesarse con el ionizador, es posible eliminar el error del valor de pesaje gracias a la carga para llevar acabo un pesaje correcto. Para obtener más información, consulte el manual de instrucciones por
separado mencionado en “1.1. Manuales detallados”.

8.1. Uso del ionizador
Paso 1. Asegúrese de que el adaptador de CA de la báscula no esté conectado y, a continuación,
conecte la báscula al ionizador (tal y como se describe en “2. Montaje e instalación”).
Paso 2. Conecte el adaptador de CA, enchúfelo y encienda la alimentación. El piloto de encendido delionizador se iluminará.
Paso 3. Coloque el objeto en un rango efectivo de eliminación de estática.
Paso 4. Ponga la mano sobre el sensor de IR (sensor de proximidad de infrarrojos sin contacto) en laparte delantera del ionizador para iniciar la eliminación de estática. El piloto ION que indica quela eliminación de estática está en curso se iluminará.
Paso 5. Consulte la siguiente figura para confirmar el rango efectivo de eliminación de estática y llévela
a cabo. Conforme a los ajustes de fábrica, 3 segundos después del inicio de la eliminación deestática, el piloto ION que indica que la eliminación de estática está en curso se apaga y laeliminación de estática se detiene.
SENSOR DE IR (sensor de proximidad deinfrarrojos sin contacto)

Ionizador(AX-ION-25)

Piloto de encendido (se ilumina cuando seenciende la alimentación)
Piloto ION (se ilumina durante la eliminación deestática)

Rango efectivo de eliminación de estática
Vista superior

Rango efectivo
0 cm
Vista lateral

5 cm
Distancia

7 cm

10 cm

Rango efectivo

1s
1,5 s
Tiempo de eliminación de estática

68

3s

ES

9.

Especificaciones
BA-6E

Capacidad de pesaje

6,2 g

Valor de visualización
máximo6,200084 g

BA-6DE
6,2 g
2,1 g
6,20008 g
2,100009 g
0,01 mg

Legibilidad

0,001 mg

Repetibilidad(desviación estándar)

0,0030 mg (6 g)

0,01 mg (6 g)

0,0010 mg (1 g)

0,0025 mg (1 g)

Linealidad±0,010 mg

0,001 mg

±0,02 mg
±0,010 mg

Pantalla

Pantalla de cristal líquido (monocromática)

Funcionamiento

TeclasComunicación

RS-232C (impresora, controlador lógico programable, etc.),
USB tipo B mini (ordenador)
Conector estéreo (interruptor externo)

BA-225
Capacidad de pesaje

220 gValor de visualización
máximo220,00084 g

Legibilidad
Repetibilidad(desviación estándar)
Linealidad

0,01 mg
0,03 mg (200 g)
0,015 mg

BA-225D

BA-125D

220 g

120 g

51 g

51 g

220,0008 g

120,0008 g

51,00009 g

51,00009 g

0,1 mg0,1 mg0,01 mg

0,01 mg

0,1 mg (200 g)(50 g)

0,025 mg

±0,15 mg

(50 g)

0,1 mg (100 g)
0,030 mg

±0,2 mg(50 g)±0,2 mg

Pantalla

Pantalla de cristal líquido (monocromática)

Funcionamiento

TeclasComunicación

RS-232C (impresora, controlador lógico programable, etc.),
USB tipo B mini (ordenador)
Conector estéreo (interruptor externo)

69

IT

1.

Guida introduttiva

Grazie per aver acquistato la bilancia analitica della serie BA di A&D. Questa guida introduttiva descrive
la procedura di installazione della bilancia della serie BA e include le funzioni di base e le operazioni dellabilancia.
Per ulteriori informazioni sulla bilancia della serie BA, consultare i manuali di istruzioni separati elencati
in “1.1. Manuali dettagliati”.

1.1. Manuali dettagliati
I manuali dettagliati possono essere scaricati dal sito web di A&D ( https://www.aandd.jp ):BA Series Instruction Manual (Manuale di istruzioni della serie BA)
Questo è il manuale che aiuta a comprendere nel dettaglio e a sfruttare pienamente le funzioni e leoperazioni della serie BA.

Ionizer (static eliminator) Instruction Manual (Manuale di istruzioni dello Ionizzatore(eliminatore di elettricità statica))
Questo è il manuale che descrive le funzioni dello ionizzatore AX-ION-25 e le istruzioni per l’uso.

2.

Assemblaggio e installazione

Attenzione
Una bilancia analitica della serie BA è uno strumento di precisione e deve essere disimballata concura.
Si consiglia di conservare i materiali di imballaggio, in modo da poterli utilizzare per trasportare labilancia.
Il contenuto della confezione varia a seconda del prodotto. Fare riferimento all’illustrazione con ilcontenuto dell’imballaggio e assicurarsi che tutto sia incluso.
Non collegare l’adattatore CA alla bilancia prima di assemblarla e installarla.
Utilizzare l’apposito adattatore CA specificato per la bilancia.
Non collegare l’adattatore CA incluso ad altri dispositivi.
Se si utilizza l’adattatore CA sbagliato, la bilancia o altri apparecchi potrebbero non funzionare
correttamente.
Accertarsi che l’adattatore CA sia scollegato prima di collegare l’unità frangivento e lo ionizzatore.
La bilancia analitica della serie BA è costituita dall’unità di pesatura e dall’unità display. Non èpossibile sostituire solo una delle unità.

70

IT

2.1. Disimballaggio
Contenuto dell’imballaggio comune della serie BA
Copertura di protezione
dell’unità display

Unità di pesatura

Placca inferiore
frangiventoSportello superiore
frangivento

Pinze
Unità display
Cavo A (circa 0,2 m)

Cavo B(circa 1,0 m)

Cavo USB(circa 1,8 m)

Adattatore CA(circa 1,5 m)

Attaccare le etichette dell’adattatore CA sull’adattatore.

Contenuto dell’imballaggio per BA-6E / BA-6DE

Unità frangivento
Vetro anteriore
frangivento

Set frangivento in vetro piccolo

Supporto diPannelli laterali (3 pezzi) Pannello
superiore
base
Piatti rotondi in alluminio per l’analisi (10 pezzi ciascuno)Sportelli frangivento
Piatto grande,Φ15, 0,8 mL

Piatto medio,
Φ12, 0,3 mL

Piatto piccolo,
Φ8, 0,05 mL

Cavo C(circa 0,6 m)

Ionizzatore

Contenuto dell’imballaggio per BA-225 / BA-225D / BA-125D
Piatto A, Φ25 Anello frangivento A

Placca anti-polvere B

Vetro anteriore frangiventoSportelli frangivento

71

Unità frangivento

IT

2.2. Assemblaggio
Qui viene utilizzata la BA-6E come esempio per l’assemblaggio. Gli altri modelli possono essere
assemblati allo stesso modo.
Unità frangivento

Passaggio 1 Estrarre le maniglie dibloccaggio del frangivento.

Passaggio 1

Passaggio 2 Posizionare
l’unitàfrangivento sull’unità dipesatura.

Passaggio 2

Passaggio 3 Spingere le maniglie dibloccaggio del frangivento
verso l’interno per fissare leunità tra loro.

Passaggio 4 Assemblare il piatto per lapesatura.
Per BA-6E / BA-6DE,
assemblare le cinque partidel piatto per la pesatura
nella posizione corretta
facendo riferimento alla vista
in sezione.
Oltre al piatto per la pesatura
standard (Piatto A), è
possibile utilizzare anche ilpiatto per la pesatura per filtri.
Permaggioridettagli,
consultare il manuale diistruzioni dettagliato elencato
in “1.1. Manuali dettagliati”.
Per BA-225 / BA-225D /BA-125D, assemblare lequattro parti del piatto per lapesatura nella posizione
corretta facendo riferimento
alla vista in sezione.

Passaggio 3

Unità di pesatura

Per BA-225 / BA-225D / BA-125D
Per BA-6E / BA-6DE

Piatto per lapesatura per
filtriPiatto A

Anellofrangivento A
Placca antipolvere A
Placca inferiore
frangivento

Vista in sezione
BA-6E / BA-6DE

Vista in sezione
BA-225 / BA-225D
BA-125D

72

Piatto B

Anellofrangivento B

Placca antipolvere B

IT
Passaggio 5 Assemblare il frangivento in
vetro.
Assemblare il frangivento in
vetro BA-6E / BA-6DE nella
cameradipesatura.
Allineare i pannelli laterali
con l’incavo del supporto dibase.

Pannello superiore
Pannelli laterali(3 pezzi)

Passaggio 5

Supporto di base

Esempio di frangivento piccolo
con il lato destro aperto.

Pannello
lateraleIncavoSupporto di base

Passaggio 6 Inserirelosportello
superiore del frangivento
dal lato della maniglia.

Passaggio 7a
Passaggio 6
Passaggio 8b

Passaggio 7 Assemblare gli sportelli
frangivento sui lati sinistro e Passaggio 7b
destro.
7aInseriregli
sportellifrangivento sui lati sinistro e
destro.

7b

Fissaregli
sportellifrangivento con i fermi.

Passaggio 8b

Passaggio 8 Assemblareilvetro
anteriore del frangivento.
8a

Inserire il vetro anteriore delfrangivento.

8b

Fissare il vetro anteriore delfrangivento con i fermi.

Passaggio 7a

Passaggio 7b73

IT
Passaggio 9 Utilizzando il cavo A e il cavo B inclusi, collegare rispettivamente l’“unità di pesatura” el’“unità frangivento” e l’“unità display” e l’“unità di pesatura”, prestando attenzione alla
direzione della freccia su ciascun cavo.
Attenzione Accertarsi di scollegare l’adattatore CA prima di effettuare il collegamento.
Passaggio 10 Collegare l’unità frangivento e lo ionizzatore con il cavo C incluso.
Attenzione Lo ionizzatore funziona quando l’adattatore CA è collegato alla bilancia.
Passaggio 11 Collegare l’adattatore CA incluso alla bilancia.
Terminale dimessa a terra
Unità display
Unità frangivento

Numero di serie
Ionizzatore.

Cavo A
Passaggio 9

Cavo B
Passaggio 11

Adattatore CA
Unità di pesatura
Passaggio 10

Cavo C

BA-225 / BA-225D / BA-125D

BA-6E / BA-6DE

74

IT

3.

Considerazioni sull’installazione, precauzioni e preparazione

Predisporre le seguenti condizioni di installazione al fine di ottenere le massime prestazioni dalla bilancia.
Installare la bilancia in un ambiente in cui la temperatura e l’umidità non siano eccessive. La temperatura
migliore per il funzionamento è di circa 20°C ±2°C a circa il 45%-60% di umidità relativa (UR).
Installare la bilancia in un luogo privo di polvere.
Il tavolo di pesatura deve essere solido. (Sono particolarmente indicati un tavolo antivibrazione o un tavolo di pietra)
Posizionare la bilancia su un tavolo orizzontale e accertarsi che non sia inclinata.
Installare la bilancia in una posizione stabile ed evitare vibrazioni e scosse. I luoghi migliori sono gli
angoli delle stanze al primo piano, poiché sono meno soggetti a vibrazioni.
Installare la bilancia in un luogo che non sia in prossimità di termosifoni o impianti di aria condizionata.
Installare la bilancia in un luogo che non sia direttamente Piedino dilivellamento
esposto alla luce solare.
Installare la bilancia lontano da apparecchi che producono
campi magnetici.
Livella a bolla d'aria
Livellare la bilancia con i piedini di livellamento e la livella a bollad’aria. Consultare “3.1. Come regolare il livello della bilancia”.
Accertarsi di far riscaldare la bilancia prima dell’uso per almeno un’ora, o per almeno quattro ore peri modelli BA-6E / BA-6DE, con l’adattatore CA collegato all’alimentazione.
Al fine di eseguire una pesatura accurata, regolare la sensibilità della bilancia prima di utilizzarla perla prima volta o dopo averla spostata in un’altra posizione. Per maggiori dettagli, consultare il manuale
di istruzioni dettagliato elencato in “1.1. Manuali dettagliati”.
Attenzione
Non installare la bilancia in presenza di gas infiammabili o corrosivi.

3.1. Come regolare il livello della bilancia
Livella abolla d'aria
La bilancia è livellata(Pesatura corretta)

Su

Giù

Piedino dilivellamento
La bilancia è inclinataGiù(Pesatura inaccurata).

Piedino dilivellamento
Cerchio rosso

Bolla

Su

Livellare la bilancia regolando i piedini di livellamento in modo che la bolla della livella a bolla d’aria
sia centrata nel cerchio rosso.

Quando la bolla è decentrata verso sinistra.

Quando la bolla è decentrata verso destra.

Quando la bolla è decentrata verso la parte posteriore.

Quando la bolla è decentrata verso la parte anteriore.

Ruotare il piedino di livellamento anteriore
destro in senso orario.

Ruotare il piedino di livellamento anterioresinistro in senso orario.

Ruotare contemporaneamente entrambi i piedini
di livellamento anteriori in senso orario.

Ruotare contemporaneamente entrambi i piedini
di livellamento anteriori in senso antiorario.

75

IT

4.

Display e pannello dei tasti (operazioni di base)

Display acceso
Numero di memoria dati
Indicatore di risposta (si accende per 30 secondi
dopo l’inizio della pesatura)
Indicatore d’urto

Indicatore del sensore IR
Segnale tara preimpostata
Segnale lordo
Segnale netto
Indicatore di processo

Segnale zero lordo

Indicatore di standby

Visualizzazione dell’unità
Mostra il valore di pesatura, i dati memorizzati e le voci

Display lampeggiante
Numero di dati attualmente visualizzati
Indicatore di processo
Avviso di regolazione automaticadella sensibilità
Indicatore della modalità intervallo di output attivo

Tasti
Sensore IR destro

Sensore IR sinistro

Tasto

SAMPLE

Tasto

MODE

Tasto

PRINT

Tasto

CAL

Tasto

TARE

Tasto

ON/OFF

Tasto

RE-ZERO76

IT

Operazioni dei tasti

Le operazioni dei tasti influiscono sul funzionamento dellabilancia. La normale operazione dei tasti durante la Premere il tasto
misurazione è “Premere e rilasciare immediatamente il tasto” (Premere e rilasciareo “Premere il tasto”. Non “Tenere premuto il tasto (per 2 immediatamente il tasto).
secondi)” a meno che non sia necessario.
Tasto

Se premuto e rilasciato

Tenere premuto
il tasto per 2
secondi.

Se mantenuto premuto (per 2 secondi)

Accende e spegne il display.
Quando il display è spento, viene
visualizzato solo l’indicatore di standby.
Quando il display è acceso, è possibile
eseguire la pesatura.
Con la funzione password, all’accensionedel display verrà visualizzato il messaggio“Inserire la password all’inizio dellapesatura”. Per maggiori dettagli, consultare ilmanuale di istruzioni separato elencato in“1.1. Manuali dettagliati”.
Il tasto ON:OFF è attivo in qualsiasi
momento e premendo questo tasto durante
l’operazione, il display verrà sempre spento.

Attiva e disattiva il sensore IR.

In modalità di pesatura, premere il tasto per
accendere e spegnere le cifre per laleggibilità.
In modalità di conteggio o percentuale,premere il tasto per entrare in modalità dimemorizzazione del campione.

Entra in modalità tabella delle funzioni.
Consultare il manuale di istruzioni
separato elencato in 1.1. Manuali
dettagliati”.
Se si mantiene premuto per altri 2 secondi
dopo la visualizzazione del menu dellatabella delle funzioni, esegue la funzione
di verifica della ripetibilità. Per maggioridettagli, consultare il manuale di istruzioni
separato elencato in “1.1. Manuali
dettagliati”.

Commuta le unità di pesatura memorizzate
nella tabella delle funzioni.

Entra in modalità di “Modifica della velocità dipesatura”.

Esegue la regolazione della sensibilità dellabilancia utilizzando la peso interna.

Mostra altre voci del menu di regolazionedella sensibilità.

Memorizza o emette il valore di pesatura,una volta stabile, secondo le impostazionidella tabella delle funzioni.
(Per impostazione di fabbrica, i dati vengono
emessi).

Entra in modalità di modifica del numero diregistrazione della massa dell’unità nella
modalità di conteggio.
Sono possibili le seguenti impostazioni con latabella delle funzioni:
Invia il “Title block (Blocco del titolo)” el’“End block (Blocco finale)” per il rapporto
GLP/GMP.
Mostra il menu di memoria dati.

Esegue l’operazione di tara.

-

Imposta il valore visualizzato a zero.

-

77

IT
Sensore IR (sensore touchless).
Reagisce quando la mano viene avvicinata
ad esso.
Sono assegnate l’apertura e la chiusura
degli sportelli frangivento.
Per maggiori dettagli, consultare “5. Sensori
IR e sportelli automatici”.

78

-

IT

5.

Sensori IR e sportelli automatici

5.1. Sensori IR
Le bilance analitiche della serie BA sono dotate di sensori IR che consentono di eseguire le operazioni senza
toccare direttamente il display della bilancia. Per impostazione di fabbrica, ai sensori IR a sinistra e a destradel display sono assegnate l’apertura e la chiusura degli sportelli frangivento. Tenendo premuto il tasto
ON:OFF (per circa 2 secondi), è possibile attivare e disattivare i sensori IR con un singolo tocco.

5.2. Sportelli automatici
Le bilance analitiche della serie BA sono dotate di sportelli automatici che consentono di aprire/chiudere
il frangivento senza toccare gli sportelli.
Ai sensori IR a sinistra e a destra del display sono assegnate le impostazioni per aprire/chiudere
rispettivamente gli sportelli frangivento a destra e sinistra.
Per impostazione di fabbrica, ogni sportello frangivento si apre nella posizione in cui era stato aperto in precedenza.
Nella tabella delle funzioni della bilancia, è inoltre possibile modificare i sensori ai quali sono assegnati
gli sportelli frangivento e scegliere se aprire gli sportelli completamente o parzialmente.
Per maggiori dettagli, consultare il manuale di istruzioni separato elencato in “1.1. Manuali dettagliati”.

Apertura dello sportello frangivento

Passaggio 1. Se si desidera aprire lo sportello frangivento sul lato sinistro (o sul lato destro) quando èchiuso, tenere la mano sul sensore IR destro (o sul sensore IR sinistro).
Passaggio 2. Viene emesso un segnale acustico di rilevamento e lo sportello frangivento sul lato sinistro (o destro) si apre.

Chiusura dello sportello frangivento

Passaggio 1. Se si desidera chiudere lo sportello frangivento sul lato sinistro (o sul lato destro) quando èaperto, tenere la mano sul sensore IR destro (o sul sensore IR sinistro).
Passaggio 2. Viene emesso un segnale acustico di rilevamento e lo sportello frangivento sul lato sinistro(o destro) si chiude.

Panoramica dell’operazione
Il sensore IR sinistro apre/chiude lo sportello di destra.

Esempio del modello BA-6E / BA-6DE

Il sensore IR destro apre/chiude lo
sportello di sinistra.

79

IT

6.

Regolazione della sensibilità

6.1. Regolazione automatica della sensibilità
La sensibilità della bilancia viene regolata automaticamente utilizzando la peso interna in base al cambio
di temperatura dell’ambiente, all’ora impostata o all’intervallo di tempo. Per maggiori dettagli, consultare
il manuale di istruzioni separato elencato in “1.1. Manuali dettagliati”.

6.2. Regolazione della sensibilità mediante peso interna
È possibile eseguire la regolazione della sensibilità mediante peso interna con un singolo tocco.
Passaggio 1. Senza posizionare niente sul piatto per la pesatura, collegare la bilancia con l’adattatore CA
per far riscaldare la bilancia per almeno 1 ora (o 4 ore per i modelli BA-6E / BA-6DE).
Passaggio 2. Premere il tasto

CAL per visualizzare.

Passaggio 3. La bilancia esegue la regolazione della sensibilità utilizzando la peso interna. Non esporrea vibrazioni o correnti d’aria che influirebbero sulla bilancia.
Passaggio 4. Al termine della regolazione della sensibilità, il “registro della regolazione della sensibilità”
viene emesso e memorizzato a seconda dell’impostazione nella tabella delle funzioni.
Per maggiori dettagli, consultare il manuale di istruzioni separato elencato in “1.1. Manuali
dettagliati”.
Passaggio 5. Una volta terminato, la bilancia ritorna automaticamente alla modalità di pesatura.

6.3. Regolazione della sensibilità mediante peso esterno
Utilizzando il proprio peso esterno, regolare la sensibilità della bilancia.
Per maggiori dettagli, consultare il manuale di istruzioni separato elencato in “1.1. Manuali dettagliati”.

80

IT

7.
Display

Display di errore (Codici di errore)
CodiceDescrizione

Errore di stabilità
Il valore di pesature è instabile e non è possibile eseguire l’“azzeramentodel display”, la “regolazione della sensibilità”, ecc. Controllare intorno al
EC,E11 piatto. Consultare il manuale di istruzioni separato elencato in “1.1. Manuali
dettagliati”. Migliorare l’ambiente del luogo di installazione (vibrazioni,
correnti d’aria, elettricità statica, ecc.). Per ritornare alla modalità dipesatura, premere il tasto CAL .

Non compreso nell’intervallo di impostazione
Il valore da impostare non rientra nell’intervallo di impostazione.
Impostare nuovamente entro l’intervallo di impostazione.

Malfunzionamento del componente della memoria interna dellabilancia
Se si continua a visualizzare questo errore, è necessaria la riparazione. Si
prega di contattare il proprio rivenditore locale per la riparazione.

Errore della peso interna
EC,E16

Applicando la peso interna non si verifica nessun cambiamento nel valoredella massa come specificato.
Verificare che non ci sia niente sul piatto ed eseguire l’operazione dall’inizio.

Errore della peso interna

EC,E17 Il meccanismo di applicazione della peso interna non funziona
correttamente. Eseguire l’operazione dall’inizio.

Anomalia nei dati della memoria interna della bilancia
Se si continua a visualizzare questo errore, è necessaria la riparazione.
Si prega di contattare il proprio rivenditore locale per la riparazione.

Errore nell’unità di pesatura
Non sono stati ricevuti i dati dall’unità di pesatura. Controllare ilcollegamento tra l’unità di pesatura e l’unità display, quindi spegnere eriaccendere.

Errore dell’unità frangivento
La comunicazione con il frangivento è scollegata.
Scollegare l’alimentazione e ricollegare il cavo tra l’unità frangivento el’unità di pesatura.

Errore del numero di serie
I numeri di serie dell’unità di pesatura e dell’unità display non
corrispondono.
Ricollegare con la combinazione corretta.

Errore nel peso di regolazione della sensibilità (valore positivo)
EC,E20Il peso di regolazione della sensibilità è troppo elevato. Controllare intorno
al piatto. Controllare il valore della massa di regolazione della sensibilità.
Per ritornare alla modalità di pesatura, premere il tasto CAL .

81

IT
Display

CodiceDescrizione

Errore nel peso di regolazione della sensibilità (valore negativo)
EC,E21

Il peso di regolazione della sensibilità è troppo basso. Controllare intorno alpiatto. Controllare il valore della massa di regolazione della sensibilità.
Per ritornare alla modalità di pesatura, premere il tasto CAL .

Errore di sovraccarico
È stato posizionato sul piatto un campione che supera la capacità dipesatura della bilancia. Rimuovere l’oggetto dal piatto.

Errore nel piatto per la pesatura
Il valore della massa è troppo basso. Controllare che il piatto per la pesatura
sia installato correttamente. Posizionare il piatto per la pesatura
correttamente. Calibrare la bilancia.

Errore della massa del campione
Il campione è troppo leggero per essere memorizzato come massa
campione per la modalità di conteggio o per la modalità percentuale. Non èpossibile usare il campione.

Errore della massa dell’unità
La massa dell’unità di campionamento per la modalità di conteggio è troppobassa. La sua memorizzazione e il suo utilizzo per il conteggio può causare unerrore di conteggio. Aggiungere campioni per raggiungere il numero specificato
e premere il tasto PRINT . Se si preme il tasto PRINT senza aver aggiunto
i campioni, la bilancia passerà alla modalità di conteggio. Tuttavia, per unconteggio accurato, accertarsi di aggiungere dei campioni.

Errore nella batteria dell’orologio
La batteria di backup dell’orologio si è scaricata. Premere qualsiasi tasto e
impostare l’ora e la data. Anche se la batteria di backup dell’orologio è
scarica, la funzione orologio e calendario funziona normalmente se labilancia viene alimentata con l’adattatore CA. Se questo errore appare difrequente, contattare il proprio rivenditore locale.

Guasto di tensione dell’alimentazione
Il voltaggio fornito dall’adattatore CA è anomalo.
Controllare se il problema è l’adattatore CA fornito in dotazione con la bilancia.

Errore di ripetibilità
La deviazione standard (SD) della ripetibilità supera le 50 cifre. Controllare
l’ambiente di installazione della bilancia.
nel display della ripetibilità.
nel display del valore minimo di pesatura (valore di riferimento).
Per maggiori dettagli, consultare il manuale di istruzioni separato elencato
in “1.1. Manuali dettagliati”.

Memoria piena
Lampeggiante(in modo alterno)

Il numero di valori di pesatura memorizzati ha raggiunto il limite massimo.
Per memorizzare nuovi valori di pesatura, è necessario cancellare dei dati.
Per maggiori dettagli, consultare il manuale di istruzioni separato “1.1.
Manuali dettagliati”.

82

IT
Display

CodiceDescrizione

Memoria piena
Lampeggiante(in modo alterno)

La cronologia dei test di calibrazione / regolazione della sensibilità memorizzata
ha raggiunto i 50 risultati. Per memorizzare un nuovo risultato, la cronologia
meno recente verrà eliminata. Per maggiori dettagli, consultare il manuale diistruzioni separato “1.1. Manuali dettagliati”.

Errore nelle comunicazioni
EC,E00 Si è verificato un errore di protocollo nelle comunicazioni.
Controllare il formato, la velocità in baud e la parità.

Errore di comando non definito
EC,E01 È stato rilevato un comando non definito.
Controllare il comando trasmesso.

Non pronto
Non è possibile eseguire il comando ricevuto:
EC,E02 (es.) Il comando Q è stato ricevuto quando non si era in modalità di pesatura.
(es.) Il comando Q è stato ricevuto durante l’azzeramento.
Regolare il tempo di ritardo per trasmettere un comando.

Errore di timeout
EC,E03Con il parametro di timeout impostato su, si è verificato un
tempo di attesa di circa 1 secondo o più durante la ricezione dei caratteridel comando. Controllare la comunicazione.

Errore nella lunghezza dei caratteri
EC,E04 Il numero di caratteri nel comando ricevuto supera il limite consentito.
Controllare il comando da trasmettere.

Errore di formato
La descrizione del comando ricevuto non è corretta:
EC,E06 (es.) Il numero di cifre dei valori numerici non è corretto.
(es.) Sono presenti caratteri alfabetici tra i valori numerici.
Controllare il comando trasmesso.

Errore nell’impostazione del parametro
EC,E07 Il valore del comando ricevuto supera il valore consentito.
Controllare l’intervallo di impostazione del valore numerico del comando.

7.1. Altri avvisi
Questo è l’avviso di regolazione automatica della sensibilità (il segnale
lampeggiante).
Se la bilancia non viene utilizzata per alcuni minuti mentre questo segnale lampeggia,
la bilancia esegue automaticamente la regolazione della sensibilità utilizzando lapeso interna. (Il periodo di lampeggiamento dipende dall’ambiente operativo).
Suggerimento Sebbene sia possibile continuare a utilizzare la bilancia anche mentre questo
segnale lampeggia, si raccomanda di utilizzarla dopo aver completato la regolazionedella sensibilità al fine di mantenere la precisione di pesatura.

83

IT

8.

Ionizzatore (AX-ION-25)

Lo ionizzatore AX-ION-25 viene fornito di serie con i modelli BA-6E / BA-6DE. Lo ionizzatore rimuove l’elettricità
statica irradiando l’oggetto target con ioni positivi o negativi, generati da quattro elettrodi di scarica mediantescarica a corona CC. Normalmente, isolanti come polveri, filtri e carta per pesatura tendono a caricarsi quando
l’umidità relativa è pari o inferiore al 45% (UR) e può verificarsi un errore di pochi milligrammi durante lapesatura. Rimuovendo l’elettricità statica dall’oggetto da pesare con lo ionizzatore, è possibile eliminare l’errore
nel valore di pesatura dovuto alla carica ed è possibile eseguire una pesatura corretta. Per maggiori dettagli,
consultare il manuale di istruzioni separato in “1.1. Manuali dettagliati”.

8.1. Utilizzo dello ionizzatore
Passaggio 1. Accertarsi che l’adattatore CA della bilancia sia scollegato, quindi collegare la bilancia e lo
ionizzatore (come indicato in “2. Assemblaggio e installazione”).
Passaggio 2. Collegare l’adattatore CA, inserirlo nella presa di corrente, quindi accendere l’alimentazione.
La spia di accensione dello ionizzatore si accende.
Passaggio 3. Posizionare l’oggetto target entro il campo effettivo di eliminazione dell’elettricità statica.
Passaggio 4. Per avviare l’eliminazione dell’elettricità statica, tenere la mano sul sensore IR (sensore diprossimità a infrarossi touchless) nella parte anteriore dello ionizzatore. La spia IONI si
accende ad indicare che l’eliminazione dell’elettricità statica è in corso.
Passaggio 5. Per quanto riguarda il campo effettivo di eliminazione statica, fare riferimento alla figura
seguente ed eseguire l’eliminazione statica. Per impostazione di fabbrica, una volta trascorsi
3 secondi dall’avvio dell’eliminazione statica, la spia IONI, che indica che l’eliminazione è incorso, si spegne e l’eliminazione statica si arresta.

Ionizzatore(AX-ION-25)

Spia di accensione (si accende quando
l’alimentazione è accesa)

Campo effettivo di eliminazione statica
Vista dall’altoCampo d’azione
0 cm
Vista laterale

5 cm
Distanza

7 cm

10 cm

1 sec 1,5 sec
Tempo di eliminazione statica

3 sec

Campo d’azione

84

IT

9.

Specifiche tecniche
BA-6E

Capacità di pesatura
Display massimo

BA-6DE
6,2 g

6,2 g

2,1 g
6,20008 g

6,200084 g

2,100009 g
0,01 mg

Leggibilità

0,001 mg

Ripetibilità(Deviazione
standard)

0,0030 mg (6 g)

0,01 mg (6 g)

0,0010 mg (1 g)

0,0025 mg (1 g)

Linearità±0,010 mg

0,001 mg

±0,02 mg
±0,010 mg

Display

Display a cristalli liquidi (monocromatico)

Funzionamento

Tasti di commutazione

Comunicazione

RS-232C (stampante, PLC, ecc.), USB tipo Mini-B (PC)Jack stereo (commutatore esterno)

BA-225
Capacità di pesatura

Display massimo

Leggibilità
Ripetibilità(Deviazione
standard)
Linearità

BA-225D

BA-125D

220 g

120 g

51 g

51 g

220,0008 g

120,0008 g

51,00009 g

51,00009 g

0,1 mg0,1 mg0,01 mg

0,01 mg

220 g

220,00084 g

0,01 mg
0,03 mg (200 g)
0,015 mg

0,1 mg (200 g)(50 g)

0,025 mg

±0,15 mg

(50 g)±0,2 mg

0,1 mg (100 g)
0,030 mg

(50 g)±0,2 mg

Display

Display a cristalli liquidi (monocromatico)

Funzionamento

Tasti di commutazione

Comunicazione

RS-232C (stampante, PLC, ecc.), USB tipo Mini-B (PC)Jack stereo (commutatore esterno)

85

CT

1.

快速入門指南感謝您購買 A&D BA 系列分析天平。本快速入門指南描述如何安裝 BA 系列天平，以及天平的基本功能
和主要操作。
如需 BA 系列天平的進一步資訊，請參閱「1.1.詳細使用說明書」中額外列出的使用說明。

1.1. 詳細使用說明書可從 A&D 網站 ( https://www.aandd.jp ) 下載以下的詳細使用說明書：BA Series Instruction Manual（BA 系列使用說明書）本說明書可協助您理解 BA 系列的詳細功能和操作，讓您可以完整使用。

Ionizer (static eliminator) Instruction Manual（電離器（靜電消除器）使用說明書）本說明書描述 AX-ION-25 電離器的功能和使用說明。

2.

組裝和安裝

警告
BA 系列分析天平是精密儀器，拆開包裝時請小心。
建議您保留包裝材料，以便將來搬運天平時使用。
包裝內容物視天平型號而定。請參閱插圖，確認沒有任何遺漏。
天平組裝完成並安裝前，請勿連接變壓器到天平。
使用天平時請搭配專屬變壓器。
請勿將隨附的變壓器連接到其它裝置。
如果使用的變壓器有誤，天平和其他設備可能無法正常使用。
連接防風單位和電離器之前，請務必拔下變壓器。
BA 系列分析天平是由稱重單位和顯示單位組成。無法僅取代任一個部分。

86

CT

2.1. 拆開包裝BA 系列共用包裝內容物
稱重單位

防風底板

顯示部分保護蓋

防風頂門

鑷子
顯示單位
纜線 A（約 0.2 m）

纜線 B（約 1.0 m）

USB 纜線（約 1.8 m）

變壓器（約 1.5 m）

變壓器 ID 標示

本文件

將變壓器標示貼到變壓器上。

BA-6E/BA-6DE 的包裝內容物
秤盤 A，Φ25

秤盤支撐 A

濾紙秤盤

防風單位
防風罩前玻璃

防塵板 A

防風環 A

小型玻璃防風組

頂板

側板（3 塊）基座支架

分析用鋁圓盤（每個 10 件）
防風門
大型秤盤，
Φ15，0.8 mL

中型秤盤，
Φ12，0.3 mL

小型秤盤，
Φ8，0.05 mL

纜線 C（約 0.6 m）

電離器

BA-225/BA-225D/BA-125D 的包裝內容物
秤盤 A，Φ25

防風環 A

防塵板 B

防風罩前玻璃

防風門

87

防風單位

CT

2.2. 配件
此處使用 BA-6E 作為配件範例展示。其它型號可以相同方式組裝。

防風單位

步驟 1 拿出防風鎖定把手。

步驟 1

步驟 2 將防風單位置於稱重單位上。
步驟 2

步驟 3 將防風鎖定把手內推，讓單位相
互固定。

步驟 3

稱重單位

針對 BA-225/BA-225D/BA-125D

步驟 4 組裝秤盤。
針對 BA-6E/BA-6DE，請參閱剖面圖，在正確位置組裝秤盤的五個零件。
除了標準秤盤（秤盤 A）之外，
您也可以使用濾紙秤盤。如需詳
細資訊，請參閱「1.1.詳細使用說
明書」列出的詳細使用說明。

針對 BA-6E/BA-6DE

濾紙秤盤
秤盤支撐 A防風環 A

針 對
BA-225/BA-225D/BA125D，請參閱剖面圖，在正確位
置組裝秤盤的四個零件。

防塵板 A
防風底板

剖面圖
BA-6E/BA-6DE
剖面圖
BA-225/BA-225D
BA-125D

88

秤盤 A

秤盤 B

防風環 B

防塵板 B

CT
步驟 5 組裝玻璃防風罩。
在 稱 重 艙 中 組 裝 BA-6E/BA6DE 的玻璃防風罩。將側板對齊
基座支架的凹槽。

頂板
側板（3 塊）

步驟 5

基座支架

側板
右側開口的小型防風罩範例。
凹槽
基座支架步驟 6 從把手側插入防風頂門。
步驟 7 在左和右側組裝防風門。
7a在左和右側插入防風門。

7b

使用鎖栓固定防風門。

步驟 7a
步驟 6
步驟 8b
步驟 7b
步驟 7a

步驟 8 組裝防風罩前玻璃。
8a插入防風罩前玻璃。

8b

使用鎖栓固定防風前門。

步驟 8a

步驟 8b
步驟 7b

89

CT
步驟 9

使用內附的纜線 A 和纜線 B，分別連接「稱重單位」和「防風罩單位」以及「顯示單位」和「稱重單位」，請注意每條纜線上的箭頭方向。
警告
連接前，請務必拔下變壓器。
步驟 10 使用內附的纜線 C，連接防風單位和電離器。
警告
電離器會在變壓器連接到天平時開始運作。
步驟 11 連接內附的變壓器到天平。
接地端子

顯示單位
防風單位

序號
電離器

。

纜線 A

步驟 9

纜線 B
步驟 11

變壓器
稱重單位
步驟 10

纜線 C

BA-225 / BA-225D / BA-125D

BA-6E / BA-6DE

90

CT

3.

安裝考量事項、準備和注意事項

請注意以下安裝環境，以充分發揮天平的效能。
將天平放置在溫度和濕度適中的環境。最佳運作溫度約為 20°C ±2°C，相對濕度約為 45 到 60％ RH。
將天平放置在沒有灰塵的地方。
稱量台應牢固。(最好使用防震桌或石桌)
將天平放在水平桌上並儘可能保持水平。
將天平放置在平穩的位置，避免震動和撞擊。最好放在一樓房間的角落，比較不會受到震動。
將天平放置在不受暖爐或冷氣空調影響的地方。
將天平放置在不受陽光直射的地方。
水平調整腳座切勿將天平放置在會產生磁場的設備附近。
調整水平腳座和氣泡式水平儀，讓天平處於水平狀態。請參閱
氣泡式水平儀。
「3.1.如何調整天平水平」
請確定使用前至少熱機一小時，針對 BA-6E/BA-6DE，請在連
接變壓器到電源的情況下，至少熱機四小時。
第一次使用前請調整天平敏感度，或在使用後移動到其他位置，以進行準確的稱重。如需詳細資訊，請參閱「1.1.詳細使用說明書」列出的詳細使用說明。
警告
請勿在有易燃氣體或腐蝕性氣體的地方放置天平。

3.1. 如何調整天平水平上氣泡式水平儀天平處於水平狀態（正確稱重）

天平處於傾斜狀態（不正確稱重）
。

下

水平調整
腳座

水平調整
腳座

下

紅色圓圈

氣泡

上

調整水平腳座，讓天平處於水平狀態，讓氣泡式水平儀的氣泡處於紅圈的正中央。

氣泡向左偏移時。

氣泡向右偏移時。

將右前方的水平調整腳座依順時針方向旋轉。

將左前方的水平調整腳座依順時針方向旋轉。

氣泡向後偏移時。

氣泡向前偏移時。

同時將前方的兩個水平調整腳座依順時針方向旋
轉。

同時將前方的兩個水平調整腳座依逆時針方向旋
轉。

91

CT

4.

顯示和按鍵面板（基本操作）

照明顯示
IR 感應器指示燈
預設皮重標記資料記憶數字
反應指示燈（開始稱重後，亮燈 30 秒）

毛重標記
淨重標記

震盪感應顯示器處理指示燈毛重歸零標記穩定性指示燈USB 連接標記

間隔輸出模式待機指示燈
待機指示燈

單位顯示
顯示稱重值、儲存的資料和物品

閃爍顯示顯示資料數字處理指示燈
自動敏感度調整通知
間隔輸出模式使用中指示燈

按鍵
右 IR 感應器

左 IR 感應器

SAMPLE 按鍵
MODE 按鍵

PRINT 按鍵CAL 按鍵TARE 按鍵

ON/OFF 按鍵

RE-ZERO 按鍵

92

CT

按鍵操作
按鍵操作會影響天平運作。測量過程中正常的按鍵操作是「按按鍵並立即放開」或「按按鍵」。除非必要，否則請勿「長按按鍵（2 秒）」。
按鍵

按下並放開時

按下按鍵（按下按鍵並立即放開。）

長按按鍵 2秒。

長按（2 秒）時

開啟和關閉顯示。
關閉顯示時，僅顯示待機指示燈。
開啟顯示時，可以稱重。
使用密碼功能時，開啟顯示將顯示「開始稱
重時輸入密碼」。如需詳細資訊，請參閱
「1.1.詳細使用說明書」額外列出的使用說
明。
ON:OFF 按鍵可隨時使用，操作時按下此按鍵將關閉顯示。
在稱重模式下，按下按鍵可開啟和關閉可讀數的數字。
在計數或百分比模式下，按下按鍵進入樣品
儲存模式。

切換 IR 感應器為開啟和關閉。

進入功能表模式。請參閱「1.1.詳細使用說
明書」額外列出的使用說明。
出現功能表選單後再長按 2 秒，執行重複性檢查功能。如需詳細資訊，請參閱「1.1.
詳細使用說明書」額外列出的使用說明。

切換功能表中儲存的稱重單位。

進入「變更稱重速度」模式。

使用內部砝碼，執行天平的敏感度調整。

顯示敏感度調整功能表的其他項目。

依據功能表設定，儲存或輸出穩定的稱重值。
（出廠設定為輸出資料。）

進入模式以變更計數模式下的單位質量註冊編號。
使用以下功能表可進行以下設定：
輸出 GLP/GMP 報告的「Title block（標題區塊）」和「End block（結束區塊）」。
顯示資料記憶體功能表。

執行皮重操作。

-

將顯示值設為零。

-

IR 感應器（無接觸式感應器）。
您將手靠近時會做出反應。
指定開啟和關閉防風門。
如需詳細資訊，請參閱「5.IR 感應器和自動
門」
。

-

93

CT

5.

IR 感應器和自動門

5.1. IR 感應器
BA 系列分析天平配備 IR 感應器，讓您不須直接觸碰天平顯示螢幕，就可進行操作。出廠設定中，位於顯示螢幕左和右側的 IR 感應器會被指定開啟和關閉防風門。長按 ON:OFF 按鍵（2 秒），您只需觸碰一
次就可切換 IR 感應器為開啟和關閉。

5.2. 自動門
BA 系列分析天平配備自動門，讓您不需觸碰門，就可開啟/關閉防風罩。
顯示螢幕左右側的 IR 感應器會分別在右側和左側指定開啟/關閉防風門的設定。
出廠設定中，每個防風門的開啟位置為之前曾開啟的位置。
同時也可變更防風門被指定的感應器，以及在天平功能表中變更要完全開啟或半開啟防風門。
如需詳細資訊，請參閱「1.1.詳細使用說明書」列出的詳細使用說明。

開啟防風門
步驟 1. 如果您想要開啟關閉的左側（或右側）的防風門，將手舉到右側的 IR 感應器（或左側的 IR 感應器）。
步驟 2. 偵測蜂鳴聲響起且左側（或右側）防風門開啟。

關閉防風門
步驟 1. 如果您想要關閉開啟的左側（或右側）的防風門，將手舉到右側的 IR 感應器（或左側的 IR 感應器）。
步驟 2. 偵測蜂鳴聲響起且左側（或右側）防風門關閉。

操作概覽
左 IR 感應器開啟/關閉右側的門。

BA-6E/BA-6DE 範例

右 IR 感應器開啟/關閉左側的門。

94

CT

6.

敏感度調整

6.1. 自動敏感度調整
將依據環境溫度變化、設定時間或間隔時間，使用內部砝碼，自動調整天平敏感度。如需詳細資訊，請參
閱「1.1.詳細使用說明書」額外列出的使用說明。

6.2. 使用內部砝碼進行敏感度調整
只需觸碰一次，就可執行使用內部砝碼進行敏感度調整。
步驟 1. 秤盤上無任何物品時，將變壓器插入天平，為天平至少熱機 1 小時（或 BA-6E/BA-6DE 4 小時）。
步驟 2. 按下

CAL 按鍵顯示。

步驟 3. 天平會使用內部砝碼執行敏感度調整。請勿讓震動或氣流影響天平。
步驟 4. 調整敏感度後，會如功能表設定，輸出並儲存「敏感度調整記錄」。
如需詳細資訊，請參閱「1.1.詳細使用說明書」額外列出的使用說明。
步驟 5. 完成後，天平將自動回到稱重模式。

6.3. 使用外部砝碼進行敏感度調整
使用外部砝碼，調整天平敏感度。
如需詳細資訊，請參閱「1.1.詳細使用說明書」額外列出的使用說明。

95

CT

7.
顯示

顯示螢幕（錯誤代碼）
代碼

描述

穩定性錯誤
稱重值不穩定且「顯示為零」、「敏感度調整」等無法執行。請檢查秤盤。請EC,E11 參閱「1.1.詳細使用說明書」列出的詳細使用說明。改善安裝位置的環境（震動、氣流、靜電等）。若要返回稱重模式，按下CAL 按鍵。

超出設定範圍
要設定的值超出設定範圍。
在設定範圍內再設定一次。

天平內部記憶體元素故障
如果持續顯示此錯誤，表示需要維修。請聯絡當地的經銷商維修。

內部砝碼錯誤EC,E16 內部砝碼並未使質量的值按指示改變。
請確認秤盤上沒有任何物件，然後從頭開始執行操作。
EC,E17

內部砝碼錯誤內部砝碼應用機制無法正常運作。請從頭開始執行操作。

天平內部記憶體資料異常如果持續顯示此錯誤，表示需要維修。
請聯絡當地的經銷商維修。

稱重單位錯誤無法接收來自稱重單位的資料。請檢查稱重單位和顯示單位間的聯線並關閉
電源然後再次開啟電源。

防風單位錯誤
未連接與防風罩的通訊。
拔下電源插頭，並重新連接防風單位與秤重單位間的纜線。

序號錯誤
稱重單位和顯示單位的序號不相符。
與正確的組和重新連接。

敏感度調整砝碼錯誤（正值）EC,E20 敏感度砝碼太重。請檢查秤盤。請檢查敏感度調整質量值。
若要返回稱重模式，按下 CAL 按鍵。

敏感度調整砝碼錯誤（負值）EC,E21 敏感度砝碼太輕。請檢查秤盤。請檢查敏感度調整質量值。
若要返回稱重模式，按下 CAL 按鍵。

超載錯誤
在秤盤上放置了超出天平稱重上限的樣品。從秤盤取下物品。

秤盤錯誤
質量值太輕。請檢查秤盤是否正確安裝。正確設定秤盤。校正天平。

樣品質量錯誤樣本太輕，無法儲存為計數模式或百分比模式的樣品質量。無法使用樣品。

96

CT
代碼

顯示

描述

單位質量錯誤計數模式的樣品單位質量太輕。儲存並用於計數可能導致計數錯誤。將樣品
增加到指定數量，然後按下 PRINT 按鍵。未增加樣品即按下 PRINT 按鍵將使天平變為計數模式。為了正確計數，請務必增加樣品。

時鐘電池錯誤時鐘備用電池已耗盡。按下任意鍵設定時間和日期。即使時鐘備用電池電量
耗盡，只要使用變壓器為天平供電，時鐘和日曆功能就可正常運作。如果頻繁出現此錯誤，請聯絡當地經銷商。

電源電壓故障變壓器電壓異常。
請檢查問題是否出在天平隨附的變壓器上。

重複性錯誤
重複性的標準偏差 (SD) 超過 50 個數字。檢查天平的安裝環境。
在重複性顯示螢幕。
在最小稱重值（參考值）顯示螢幕。
如需詳細資訊，請參閱「1.1.詳細使用說明書」列出的詳細使用說明。

記憶體已滿閃爍（交替）

儲存的稱重值數量已達到上限。為了儲存新的稱重值，您必須刪除資料。如需詳細資訊，請參閱「1.1.詳細使用說明書」列出的詳細使用說明。

記憶體已滿閃爍（交替）

儲存的敏感度調整/校正測試記錄已達 50 個。為了儲存新的結果，最舊的記錄將刪除。如需詳細資訊，請參閱「1.1.詳細使用說明書」列出的詳細使用說
明。

通訊錯誤
EC,E00 通訊時發生通訊協定錯誤。
請檢查格式、鮑率和奇偶。

未定義的命令錯誤
EC,E01 發現未定義的命令。
請檢查傳輸的命令。

尚未準備就緒無法執行收到的命令：
EC,E02 （例如）不在稱重模式時，收到 Q 命令。
（例如）歸零時，收到 Q 命令。
調整傳輸命令的延遲時間。

逾時錯誤EC,E03 逾時參數設定為時，接收命令字元時，約有 1 秒鐘或以上的等待時間。請檢查通訊狀態。
EC,E04

字元長度錯誤接收命令的字元數量超過限制。請檢查要傳輸的命令。

97

CT
代碼

顯示

描述

格式錯誤
接收命令的描述不正確：
EC,E06 （例如）述職的數字數量不正確。
（例如）數值中有字母字元。
請檢查傳輸的命令。

參數設定錯誤EC,E07 接收命令的值超過容許值。
請檢查命令數值的設定範圍。

7.1. 其它顯示
此為自動敏感動調整通知（
標記閃爍）。
此指示燈閃爍時，如果未在數分鐘內使用天平，天平會使用內部砝碼自動執行敏感度調整。
（閃爍期間視操作環境而定。）
建議雖然此標記閃爍時也可以繼續使用天平，但是為了維持稱重精準度，建議在敏感度調
整完成後再使用。

98

CT

8.

電離器 (AX-ION-25)

AX-ION-25 電離器是 BA-6E/BA-6DE 的標準配備。電離器使用直流電暈放電，從四個放電電極產生的正離子或負離子射出目標物，去除靜電。濕度為 45% RH 或更低時，通常粉末、濾紙和稱重紙之類的絕緣體往往會帶電，並且在稱重時可能會出現幾毫克的誤差。使用電離器去除稱重物品的靜電，可以減少因充
電導致的稱重值誤差，並進行正確的稱重。如需詳細資訊，請參閱「1.1.詳細使用說明書」額外列出的使用
說明。

8.1. 使用電離器
步驟 1. 請確定天平的變壓器未連接，然後連接天平和電離器（如「2.組裝和安裝」所述）。
步驟 2. 連接變壓器、插入插頭，然後開啟電源。電離器的電源指示燈亮起。
步驟 3. 將目標物品放置在靜電消除的有效範圍內。
步驟 4. 將手舉到電離器前的 IR 感應器（無接觸式紅外線距離感應器），開始消除靜電。指示正在進行靜
電消除的 ION 指示燈會亮起。
步驟 5. 請參閱下圖，瞭解靜電消除的有效範圍並執行靜電消除。出廠設定中，開始靜電消除過了 3 秒
鐘時，指示正在進行靜電消除的 ION 指示燈會熄滅且會停止靜電消除。

IR 感應器（無接觸式距離感應器）

電離器(AX-ION-25)

電源指示燈（開啟電源時亮起）
ION 指示燈（進行靜電消除時亮起）靜電消除的有效範圍
俯視圖
有效範圍
0 cm

5 cm
距離

7 cm

10 cm

1 秒
1.5 秒
靜電消除時間

3 秒

側視圖

有效範圍

99

CT

9.

規格
BA-6E

可讀數
可重複性
（標準偏差）
線性

6.2 g

6.2 g

稱重上限

最大顯示數值

BA-6DE
2.1 g
6.20008 g

6.200084 g

2.100009 g
0.01 mg0.001 mg0.001 mg

0.0030 mg (6 g)

0.01 mg (6 g)

0.0010 mg (1 g)

0.0025 mg (1 g)±0.02 mg

±0.010 mg±0.010 mg

顯示液晶顯示螢幕（單色）

操作

按鍵切換

通訊

RS-232C（印表機、PLC 等）、USB type Mini-B (PC)立體聲插孔（外部切換）

BA-225
稱重上限最大顯示數值

可讀數
可重複性（標準偏差）
線性BA-225D

BA-125D

220 g

120 g

51 g

51 g

220.0008 g

120.0008 g

51.00009 g

51.00009 g

0.1 mg

0.1 mg0.01 mg

0.01 mg

220 g

220.00084 g

0.01 mg
0.03 mg (200 g)
0.015 mg

0.1 mg (200 g)(50 g)

0.025 mg

±0.15 mg

(50 g)±0.2 mg

0.1 mg (100 g)
0.030 mg±0.2 mg顯示

液晶顯示螢幕（單色）

操作

按鍵切換

通訊

RS-232C（印表機、PLC 等）、USB type Mini-B (PC)立體聲插孔（外部切換）

100(50 g)

MEMO

MEMO

A&D Company, Limited
3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-1566

A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131, U.S.A.
Telephone: [1] (408) 263-5333
Fax: [1] (408)263-0119

A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 07331( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4264

Почтовый адрес:121357, Российская Федерация, г.Москва, ул. Верейская, дом 17Юридический адрес: 117545, Российская Федерация, г. Москва, ул. Дорожная, д.3, корп.6, комн. 8б
( 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Instruments India Private Limited( 509, Udyog Vihar, Phase- , Gurgaon - 122016, Haryana, India ): [91] (124) 4715555: [91] (124) 4715599

A&D SCIENTECH TAIWAN LIMITED.

A&D台灣分公司 艾安得股份有限公司

台湾台北市中正區青島東路 5 號 4 樓
( 4F No.5 Ching Tao East Road, Taipei Taiwan R.O.C. )Tel : [886](02) 2322-4722
Fax : [886](02) 2392-1794

A&D INSTRUMENTS (THAILAND) LIMITEDบริษัท เอ แอนด ดี อินสทรูเมนท (ไทยแลนด) จํากัด
168/16 หมูที่ 1 ตําบลรังสิต อําเภอธัญบุรี จังหวัดปทุมธานี 12110 ประเทศไทย
( 168/16 Moo 1, Rangsit, Thanyaburi, Pathumthani 12110 Thailand )Tel : [66] 20038911