INJECTORSDISINFECTINGGUIDELINES

E21000433

Diagnostic imaging

READY BOX®
ILLUMENA® NEO
OPTIVANTAGE®

OPTISTAR® ELITEThe Chemical agents listed
are determined using
the EPA list for disinfectants
against COVID-19

OPTIONE®
2

OPTIVANTAGE® / OPTIONE®
DISINFECTING GUIDELINES

3OPTIVANTAGE® / OPTIONE® DISINFECTING GUIDELINES%CHEMICAL AGENTPERCENTAGE
CONCENTRATION

Ethyl Alcohol(Ethanol)

AGENT PRODUCT EFFECTRECOMMENDATION OF USE

PRODUCT DETAILS

62%

3 No components were affected
3 Did not cause any damage
or discoloration to any of the testedcomponents or surfaces

3 The most benign chemical for use

Hydrogen
Peroxide

2.2%

Isopropyl Alcohol(Isopropanol)

0.5%

3 The powerhead screw coating
had a minor affect
3 No plastic or rubber components
were affected
3 There was a slight effecton hardware: power head screws

3 Only a small effect on some platingof metals. Should recommendthat the percentage levels
are kept this low for either one
or both chemicals

3 Powerhead Screws:
Material: 904142 - 10 mm Screw;
Steel with Black Zinc Coating:
3 in the pre vs. post disinfectant,		 there was some slight removal		 of the black zinc coating

Citric Acid

0.6%

3 Two components were affected
3 Moderate effect on some platingof metals
3 The powerhead screw coating
was slightly affected
3 The console hinge screws
was moderately affected as far
as appearance with respect
to the oxide coating

3 For use on plastics only

3 Power head screws:
904142 - 10 mm Screw; Steel With
Black Zinc Coating
3 Console Hinge Screws:
904159 – #10 Screw, Steel WithZinc Coating

BACK TO CONTENTS

4ILLUMENA® NEO
DISINFECTING GUIDELINES

5ILLUMENA® NEO DISINFECTING GUIDELINES%CHEMICAL AGENTPERCENTAGE
CONCENTRATION

Ethyl Alcohol(Ethanol)

AGENT PRODUCT EFFECTRECOMMENDATION OF USE

PRODUCT DETAILS

62%

3 The product did not cause any damage
or discoloration to any of the tested 		
components or surfaces
3 No components were affected
3 No damage on any hardware with zincor black oxide plating

3 Most benign chemical of the test
disinfectants and should be the firstto be recommended

Hydrogen
Peroxide

2.2%

3 A minor effect on the Pedestal Hubblue paint (dots created)
3 A medium effect on the lag screws
zinc plating

3Should recommend that thepercentage levels are kept this low(or lower) for either one or bothchemicals

Isopropyl Alcohol(Isopropanol)

0.5%

3 PEDESTAL HUB: Material: 905838 - 		Lexan FL9OO Polycarbonate StructuralFoam. Only the portion painted PMS
296 (Guerbet Blue) was affected.
3 LAG SCREWS: Material: 904162 - 		STEEL; ZINC PLATING

Citric Acid

0.6%

3 The Pedestal Hub blue paint had
minor staining
3 A medium effect on the lag screws
zinc plating
3 Seven additional hardware
components had medium
discoloration of the plating
3 Six of the hardware components had
minor to medium plating removal

3 For use on plastics only

Only the portion painted PMS 296 (Guerbet Blue)was affected.
3 904119 - STEEL; ZINC PLATING
3 904101 - STEEL; ZINC PLATING
3 904162 - STEEL; ZINC PLATING
3 904145 - STEEL; ZINC PLATING
3 904134 - SPRING STEEL; BLACK OXIDE
3 311S0012 - STEEL; ZINC PLATING
3 311S0041 - STEEL; ZINC PLATING
3 311S0032 – STAINLESS STEEL; ZINC PLATINGBACK TO CONTENTS6OPTISTAR® ELITEDISINFECTING GUIDELINES

7OPTISTAR® ELITE DISINFECTING GUIDELINES%CHEMICAL AGENTPERCENTAGE
CONCENTRATION

Ethyl Alcohol(Ethanol)

62%

3 No tested components / materials
were affected by the disinfectants
3 Most benign chemical

Hydrogen
Peroxide

2.2%

3 Isopropyl Alcohol (Isopropanol)
3Recommend that the percentage
had only some minor effects on metallevels are kept this low (or lower)
hardware coating
for either one or both chemicals

Isopropyl Alcohol(Isopropanol)

0.5%

3 SCREW,M4X.7,10MM,PR,TH,BLK:            Steel: Black Zinc Finish
3 SCREW, M4X.7,10MM,PH W/LCK& FLT WSHR: Brass with nickel plating

Citric Acid

0.6%

3 It had some minor to medium issues
with metal hardware components
and their coatings

3 SCREW,M4X.7,10MM,PR,TH,BLK:            Steel; Black Zinc Finish
3 SCREW, M4X.7,10MM,PH W/LCK& FLT WSHR: Brass with nickel plating

AGENT PRODUCT EFFECTRECOMMENDATION OF USE

PRODUCT DETAILS

3 Primary recommended disinfectant

3 For use on plastics only

BACK TO CONTENTS

8READY BOX®
DISINFECTING GUIDELINES

9READY BOX® DISINFECTING GUIDELINES%CHEMICAL AGENTPERCENTAGE
CONCENTRATION

Ethyl Alcohol(Ethanol)

62%

Hydrogen
Peroxide

2.2%

Isopropyl Alcohol(Isopropanol)

0.5%

Citric Acid

0.6%

AGENT PRODUCT EFFECTRECOMMENDATION OF USE

PRODUCT DETAILS

3 Most benign chemical
3 No tested components/materials
were affected by the disinfectants.

3 The primary recommended
disinfectant

3 Minor effects on metal hardware
coating

3Recommend that the percentage
levels are kept this low (or lower)
for either one or both chemicals

3 SCREW,SEM,6-32X1/4” SLOT HEX,WASHER: Zinc plated steel
3 SCREW-THREAD FORMING 6-32 X1/2
HEX WASHER HEAD: Zinc plated steel
3 WASHER - #8 INT. TOOTH: steel
3 NUT-8-32X1/4 HEX: steel
3 SCREW-8X1/2 PR TH: steel

3 Minor to medium issues with metal
hardware components
and their coatings

3 For use on plastics only

3 SCREW,SEM,6-32X1/4” SLOT HEX,WASHER: Zinc plated steel
3 SCREW-THREAD FORMING 6-32 X1/2
HEX WASHER HEAD: Zinc plated steel
3 WASHER - #8 INT. TOOTH: steel
3 NUT-8-32X1/4 HEX: steel
3 SCREW-8X1/2 PR TH: steel

BACK TO CONTENTS

10LEGAL MENTIONS
Optivantage® DH Contrast Media Injectors is a medical device intended for use by qualified healthcare professionals, it is a contrast delivery system and is designed to inject
radiopaque contrast media into a patient’s vascular system to obtain diagnostic images when used with computed tomography (i.e. “CT”) equipment. Ensure that injection frommulti-dose containers and to more than one patient respects the valid rules of good practice and the marketing authorization (MA) of the contrast agent and local regulatoryrequirements.For complete information about precautions and optimal usage conditions, we recommend consulting the instruction for use supplied with the device or by yourlocal Guerbet representative(s). For use only in countries with applicable health authority registrations
Class IIb/CE TÜV SÜD 0123 Manufacturer: Liebel-Flarsheim Company LLC EC Rep: Guerbet
The Optione® Single-Head Contrast Media Injector is a class IIb (CE) medical device intended for use by qualified healthcare professionals, is intended to inject radiopaque
contrast media into a patient’s vascular system to obtain diagnostic images when used in conjunction with computed tomography (CT) equipmentFor complete information about
precautions and optimal usage conditions, we recommend consulting the instruction for use supplied with the device or by your local Guerbet representative(s). For use only incountries with applicable health authority registrations
Class IIb/CE TÜV SÜD 0123 Manufacturer: Liebel-Flarsheim Company LLC EC Rep: Guerbet
Illumena® Neo Contrast Media Injectors is a medical device intended for use by qualified healthcare professionals, it is designed to inject a radiopaque contrast mediuminto the vascular system for Angiographic or CT procedures as prescribed by qualified health care professionals. Contraindications for the use of this device are determinedby the prescribing physician at the time of use based upon the contrast media package inserts. For complete information about precautions and optimal usage conditions,
we recommend consulting the instruction for use supplied with the device or by your local Guerbet representative(s). For use only in countries with applicable health authority
registrations
Class IIb/CE TÜV SÜD 0123 Manufacturer: Liebel-Flarsheim Company LLC EC Rep: Guerbet
OptiStar® Elite Contrast Delivery System is a medical device intended for use by qualified healthcare professionals. It is intended to inject MR contrast media and flushingsolutions into a patient’s vascular system to obtain diagnostic images when used in conjunction with Magnetic Resonance (MR) Imaging equipment. For complete information
about precautions and optimal usage conditions, we recommend consulting the instruction for use supplied with the device or by your local Guerbet representative(s). For use
only in countries with applicable health authority registrations.
Class IIb/CE. TÜV SÜD 0123 Manufacturer: Liebel-Flarsheim Company LLC EC Rep: Guerbet
Ready Box® Contrast Warmer is a medical device intended for use by qualified healthcare professionals to warm contrast media to approximately body temperature. For
complete information about precautions and optimal usage conditions, we recommend consulting the instruction for use supplied with the device or by your local Guerbet
representative(s). For use only in countries with applicable health authority registrations
Class IIa/CE TÜV SÜD 0123 Manufacturer: Liebel-Flarsheim Company LLC
OptiVantage®, OptiOne®, Illumena® Néo, OptiStar® Elite and Ready-Box® are trademarks of Guerbet.
Issued patents. © Copyright 2015 Guerbet - All rights reserved.

BACK TO CONTENTS

11

www.guerbet.com