Digital Platform Scale
HV-15KCEP
HV-60KCEP
HV-200KCEP
HW-10KCEP
HW-60KCEP
HW-100KCEP
HW-200KCEP

1WMPD4003607B

2024.04.17

© 2019 A&D Company Ltd. All rights reserved.
No part of this publication may be reproduced, transmitted, transcribed, or translated into any language inany form by any means without the written permission of A&D Company Ltd.
The contents of this manual and the specifications of the instrument covered by this manual are subject
to change for improvement without notice.
Windows, Word and Excel are registered trademarks of the Microsoft Corporation.

Contents
1. COMPLIANCE ............................................................................................................... 5
1.1. Compliance with FCC rules ..................................................................................... 5
2. OUTLINE AND FEATURES .......................................................................................... 6
3. SAFETY INSTRUCTIONS AND PRECAUTIONS ......................................................... 7
3.1. Safety Instructions on Intrinsically safe Structure ................................................ 7
3.2. Precautions on Installing the Scale ........................................................................ 7
3.3. Precautions on operating the Scale ....................................................................... 8
3.4. Precautions on storing the Scale............................................................................ 8
4. INSTALLING THE SCALE ............................................................................................ 94.1. Setting up the Scale.................................................................................................. 9
5. UNPACKING ............................................................................................................... 10
5.1. Accessories and Options List................................................................................ 11
5.2. Installing the Batteries ........................................................................................... 125.3. Grounding the scale ............................................................................................... 13
6. CONFORMING STANDARDS ................................................................................. 14
6.1. IECEx ....................................................................................................................... 14
6.2. ATEX ........................................................................................................................ 14
6.3. FM,FMc (Zone)......................................................................................................... 14
6.4. FM,FMc (Division) ................................................................................................... 14
7. DESCRIPTION OF EACH PART ................................................................................ 16
7.1. Display and Symbols .............................................................................................. 17
7.2. Keys ......................................................................................................................... 19
8. BASIC OPERATION ................................................................................................... 21
8.1. Turning the Scale on/off and Weighing ................................................................ 21
8.1.1. When Using Batteries ....................................................................................... 21
8.2. Tare (And Net Display) ........................................................................................... 22
8.2.1. Tare Input by Weighing ..................................................................................... 22
8.2.2. Digital Input (Preset Tare) ................................................................................. 228.3. Switching the mode ................................................................................................ 23
HV/HW-CEP Series

Page 3

9. COUNTING MODE ...................................................................................................... 24
9.1. Storing a Unit Mass ................................................................................................ 24
9.2. Counting the Number of Articles ........................................................................... 25
10. ACCUMULATION FUNCTION .................................................................................... 26
11. COMPARATOR ........................................................................................................... 28
11.1. The formula to compare....................................................................................... 29
11.2. Entering the comparator values .......................................................................... 30
12. AUTO-TARE ................................................................................................................ 31
13. CALIBRATION ............................................................................................................ 32
13.1. Gravity Acceleration Table .................................................................................. 33
13.2. Complete Calibration Procedure ......................................................................... 34
13.2.1. Gravity Acceleration Correction ..................................................................... 34
13.2.2. Preparation ....................................................................................................... 34
13.2.3. Calibration of the Zero Point........................................................................... 35
13.2.4. Span Calibration .............................................................................................. 35
14. FUNCTION TABLE ..................................................................................................... 36
14.1. Parameter Setting Procedure .............................................................................. 36
14.2. Parameter List ...................................................................................................... 37
15. SPECIFICATIONS ....................................................................................................... 39
16. MAINTENANCE .......................................................................................................... 43
16.1. Check Points Before Calling Maintenance Service ..................................................... 43
16.1.1. Repair ............................................................................................................... 43

Page 4

HV/HW-CEP Series1. Compliance1.1. Compliance with FCC rules
 Please note that this equipment generates, uses and can radiate radio frequency
energy. This equipment has been tested and has been found to comply with thelimits of Class A digital devices pursuant to Part 15 of FCC rules. These rules aredesigned to provide reasonable protection against interference when equipment is
operated in a commercial environment. If this unit is operated in a residential area, it
may cause some interference and under these circumstances the user would be
required to take, at his own expense, whatever measures are necessary to eliminate
the interference. (FCC = Federal Communications Commission in the U.S.A.)

HV/HW-CEP Series

Page 5

2. Outline and Features
 The HV-CEP/HW-CEP series are intrinsically safe electronic scales complying withinternational standards such as IECEx and ATEX or FM requirements.
 The HV-CEP/HW-CEP series can be used in areas where explosive atmospheres arepresent. (For details on the explosion-proof structure, refer to “6. CONFORMING
STANDARDS.”) The scale automatically switches to small scale interval when a light sample isweighed or large scale interval when a heavy sample is weighed, depending on thesample weight (multi-interval).
 The HW-CEP series are platform scales with 1/10000 resolution.
 The HV-CEP/HW-CEP series use batteries. As a power source, only the four “D”,
“LR20” or “AM1” alkaline 1.5 V batteries listed below can be used.
DURACELL MN1300, ENERGIZER E95, Panasonic LR20(XJ) The scales use a back lit liquid crystal display to enable viewing in dim light.
 The counting mode converts the total mass value (total weight) of articles to be
counted, to a count, when each article has the same mass value.
 The comparator function compares the displayed weight value with the upper limit
value (HI) and the lower limit value (LO) and displays the result.
 The scale has an accumulation function with a maximum of 6 digits, which has amaximum accumulations of 999 times. (The number of times weighed and the total
mass of that can be stored in the scale.) The following parameters are stored in the scale even if the power is removed.
Display mode (unit)
Unit mass of counting mode
Total count and total mass of accumulation function
Upper limit value and lower limit value of upper / lower comparator function
Calibration data
Parameters of the function table Multi-interval: This is what the readability is automatically switched to dependingon the sample weight.
Example) With the HV-60KCEP, weighing capacity is 60kg andreadability is 0.005kg, 0.01kg or 0.02kg.
Multi-interval: When exceeding the range for small, medium or large, the readability
is automatically switched. Weighing capacity
Point:
When a light sample is
60kg
Large
weighed, the readability isrange
small. When a heavy
30kg
Mediumsample is weighed, therangereadability is large.
15kg
Small
range
0.005kgPage 6

0.01kgMinimum
display
Readability

0.02kg

HV/HW-CEP Series3. Safety Instructions and Precautions
3.1. Safety Instructions on Intrinsically safe Structure Do not disassemble or modify the scale.
 When determining the installation site, take the following into consideration.
Will explosive gases be generated? How often will they be generated?
Refer to “6. CONFORMING STANDARDS.” A portion of the enclosure is non-conducting and, under certain extreme conditions,
may generate an ignition-capable level of electrostatic charges. The user shallensure that the equipment is not installed in a location where it may be subjected toexternal conditions (such as high-pressure steam) which might cause a build-up ofelectrostatic charges on non-conducting surfaces. Additionally, cleaning of theequipment should be done only with a damp cloth.
 Replace the batteries in non-hazardous areas.
 As a power source, only the four “D”, “LR20” or “AM1” alkaline 1.5 V batteries listed
below can be used.
DURACELL MN1300, ENERGIZER E95, Panasonic LR20(XJ) All 4 cells in single platform scale must be by the same manufacturer.
 When replacing the batteries, be sure to prevent foreign materials from entering thebattery compartment.
 The enclosure contains aluminum and is considered a potential risk of ignition by
impact or friction. Care must be taken during installation to prevent impact orfriction.

3.2.

Precautions on Installing the Scale

Consider the following conditions to get the most from your scale.
Install the scale where the temperature and relative humidity is stable, there is no draft
and a stable power source is available.
Install the scale on a solid and level surface.
Do not install the scale in direct sunlight.
Do not install the scale near heaters or air conditioners.
Do not install the scale where there is corrosive gas present.
Do not install the scale near equipment which produces magnetic fields.
Do not install the scale where there is apt to be static electricity, in a place where therelative humidity is lower than 45% RH. Plastic and isolators are apt to be charged withstatic electricity.
The display unit is not water resistant. Use the display unit cover to avoid damage.
Do not use an unstable power source.
Remove the protective film from the weighing pan before use.
HV/HW-CEP Series

Page 7

3.3.

Precautions on operating the Scale

Periodically ensure that the weight value is correct.
Calibrate the scale before using and after moving it to another location.
Do not place anything on the pan which exceeds the weighing capacity.
Do not drop anything upon the pan.
Do not use a sharp instrument such as a pencil to press the keys. Press the keys
gently using your finger.
Pressing the ZERO key before each weighing is recommended to prevent possible
error.
Replace the used batteries with four new ones when the “lb” mark is displayed.
As a battery, only the four “D”, “LR20” or “AM1” alkaline 1.5 V batteries listed below
can be used.
DURACELL MN1300, ENERGIZER E95, Panasonic LR20(XJ)

3.4.

Precautions on storing the Scale

Do not disassemble the scale.
Do not use solvents to clean the scale. Wipe it with a dry lint free cloth or a lint freecloth which is moistened with warm water and a mild detergent.
The base unit can be cleaned with gentle running tap water. Do not scratch the baseunit with a brash. Allow the unit to dry before using.
Protect the display unit from dust and water.
Remove the batteries from the display unit when the scale is not to be used for a longtime. If you leave the batteries installed, they may leak and damage the scale.

Page 8HV/HW-CEP Series

4. Installing the Scale
4.1. Setting up the Scale
This procedure includes all of the steps for installingthe scale. Therefore, there may be someunnecessary steps for some models.
Step 1 Take the base unit and pole out, taking care
not to pull on the load cell cable.

Pan
Pole
Step 2
Step 1

Step 2 Place the pan on the base unit. Remove theprotective film from the pan before use.
Step 3 Attach the pole to the bracket of the baseunit, while using care not to damage the loadcell cable.
Insert the remainder of the load cell cable
into the pole.
Affix the pole to the bracket using two hex
screws.
* With the HW-10KCEP and HV-15KCEP,
this procedure is not required because thepole and bracket is a combined unit.

Load cell cableBase unit
Pole
Step 33mm Allen wrench
Allen screws

BracketBubble spirit level

Step 5

Leveling feetOK

Step 4 Select a place for installing the scale. Referto "3.2. Precautions on Installing the Scale".

10°
Display unit

Step 5 Adjust the level of the base unit using the
bubble spirit level and the leveling feet.

NG

30°
50°
70°

Groundterminal

90°

Step 6 Adjust the angle of the display unit and fix bydisplay knobs.
Step 7 Check the weighing accuracy. If the scaleneeds calibration, refer to "13. Calibration".

Step 6

Display knobs

The display unit can be adjusted in five steps in the up-and-down direction. Setting
the display sideways is also possible. (Make sure that the pole is secured at thelower part of the pole using the hex screws. Do not turn the display unit at a joint forthe pole.)
HV/HW-CEP Series

Page 9

5. Unpacking
HV-15KCEP
HW-10KCEP

Display unit

HV-200KCEP
HW-100KCEP
HW-200KCEP

Display unit

Pan

Pan
Base unit

HV-60KCEP
HW-60KCEP

Display unit

Base unit
Caution
Do not pull the load cell cable.

Pan2.5mm Allen wrench

Base unit

3mm Allen wrench

Caution
Do not pull the load cell cable.

Page 10

HV/HW-CEP Series

5.1.

Accessories and Options List

Accessories List
Models
HV-15KCEP
HW-10KCEP

HV-60KCEP
HV-200KCEP
HW-60KCEP
HW-100KCEP
HW-200KCEP

HV/HW-CEP Series

Accessories (Quantity)- 2.5mm hex wrench (1)- Instruction manual (1)- 2.5mm hex wrench (1)- 3mm hex wrench (1)- Instruction manual (1)Page 11

5.2.

Installing the Batteries

Step 1

Turn off the display.
Remove the right side display knob.
Tilt the Display unit to the front.

Step 2

Open the rear cover.

Display unit

Display knobs

Step 1

Step 3

Disconnect the power connector.
Power connector

Step 4

Carry the battery box to non-hazardous
areas.

Step 5Using the hex wrench provided, remove thebattery box securing screws.

Rear cover
Step 2

Step 3

Battery box
Rear knobs

Step 6

Insert four new batteries with proper polarity(+,-). Batteries are only the four “D”, “LR20”
or “AM1” alkaline 1.5 V batteries listed below
can be used.
DURACELL MN1300, ENERGIZER E95,
Panasonic LR20(XJ)

Step 7 Attach the battery cover and secure it using
the battery box securing screws.

Step 4

Battery box

2.5mm Allen wrench

Step 5

Step 8 Close the covers in reverse order of step 3 tostep 1.

CautionReplace used batteries with four new ones, when “lb” is displayed.

All 4 cells in single platform scale must be by the same manufacturer.
Do not mix used and new batteries. It may cause damage to the batteries or
the scale, if used.
Check the battery direction. If the batteries are installed in the wrong
direction, it may cause battery leakage. If the direction of a single battery iswrong, the scale may work only temporarily.
Page 12HV/HW-CEP Series

The battery life depends on the ambient temperature.
Remove the batteries from the display unit, when the scale is not to be used
for a long time. They may leak and cause damage.
Damage which is due to battery leakage is not covered under warranty.

5.3.

Grounding the scale

When using where there may be static electricity, ground the scale.
The grounding procedure depends on the scale model. Refer to the table below.
These procedures are only for grounding part of the scale.
ModelsRefer toHV-15KCEP/HW-10KCEP
HV-60KCEP/HV-200KCEP
HW-60KCEP/HW-100KCEP/HW-200KCEP

Procedure A
Procedure B

Procedure A(HV-15KCEP/HW-10KCEP)
Base unit
bottom side

Secure the grounding cable using a M4 screw in thescrew hole between the two hexagon bolts on the baseunit bottom side. (Part of “ ”)
Levering feet
Under cover

Procedure B

HV-60KCEP/HV-200KCEP
HW-60KCEP/HW-100KCEP/HW-200KCEP

Secure the grounding cable using the screw that
secures the under cover. (Part of “ ”)

HV/HW-CEP Series

Page 13

Base unit
bottom side

6.

CONFORMING

6.1.

IECEx

STANDARDS

Ex ia ΙΙB T4 Ga Tamb -25 ºC to +40 ºC (IECEx FMG 17.0018X)
Ex:
ia:
ΙΙB:
T4:
Ga:

Ex Component
Type of Protection
Gas Classification
Temperature Class
Equipment Protection Level

6.2. ATEX
ΙΙ 1G Ex ia ΙΙB T4 Ga Tamb -25 ºC to +40 ºC (FM17ATEX0038X)
Ex:
ia:
ΙΙB:
T4:
Ga:

Ex Component
Type of Protection
Gas Classification
Temperature Class
Equipment Protection Level

6.3. FM,FMc (Zone)
Class Ι, Zone 0, AEx / Ex ia ΙΙB T4 Tamb -25 ºC to +40 ºC(FM17US0171X/FM17CA0089X)
Class:
Zone:
Ex:
(AEx:
ia:
ΙΙB:
T4:

Permitted Class
Permitted Zone
Ex Component
American National Standard)Type of Protection
Gas Classification
Temperature Class

6.4. FM,FMc (Division)
Intrinsically Safe for Class Ι, Division 1, Groups C, D T4
Tamb -25 ºC to +40 ºC (FM17US0171X/FM17CA0089X)
Intrinsically Safe: Type of Protection
Class:
Permitted Class
Division:
Permitted Division
Groups:Permitted Groups
T4:
Temperature Class

Page 14

HV/HW-CEP Series

The label on the scale

HV/HW-CEP Series

Page 15

7. Description of Each Part
Models
HV-15KCEP
HW-10KCEP

Display unit

Models
HV-60KCEP
HV-200KCEP
HW-60KCEP
HW-100KCEP
HW-200KCEP

Display knobs
Pole

Display unit

Display knobs
Pole

Pan (Weighing Pan)
Base unit
Leveling footDisplay

Leveling footBubble spirit level
Leveling footCautionThe certified mass is required.

CAL switch
Calibrating the scale
to weigh correctly.

Battery box

Rear cover

Ground terminal

Rear knobs

Display knobs

Page 16

HV/HW-CEP Series7.1.

Display and Symbols

Display and Symbols

PT

HOLD

M+

A
A
Stability markWhen the current weight value is stable, this mark is A
displayed. It means a proper condition that this value is A
readable.
A
A
Zero point markWhen the ZERO key is pressed with nothing on the pan, A
this mark is displayed. The zero point is the fundamental A
starting point to weigh anything.
A
Net markA
When the TARE key is pressed, this mark is displayed. A
Used to indicate that the mass of the container placedA
Description

on the pan has been subtracted from the gross value.
Preset tare markA
While a tare with digital input is displayed, this markblinks.
Hold markWhile the display is held, this mark is displayed.
Accumulation markWhile the accumulation function is used, this mark is
displayed.
Low battery markWhen the battery is depleted (becoming low voltage),
this mark is displayed. Replace with four new batteries.
Comparator indicator
When using the comparator function, the result is
indicated after the weight value is compared with theupper and lower limits.

Weighed mass value
STABLEZERO

HV/HW-CEP Series

unit

A
A
A
A
A
A
A

A
A
Zero point (Example)
When the ZERO key is pressed with nothing on the pan, A
the zero point mark and the stability mark are displayed. A
APage 17

Display and Symbols

Description
Counting mode (Example)
This mode uses the stored unit mass and counts thenumber of articles on the pan. The unit is.
Storing the unit mass for the counting mode (Example)
The unit mass is stored, using 20 pieces of samples.
The zero value means that no articles are on the pan.
Storing the unit mass for the counting mode (Example)
The unit mass is stored, using 10 pieces of samples.
Sign "-" means that something is placed on the pan.
Function settings (Example)- Press the MODE key to select the item, and then press
the ENTER key to finalize the selection.
- Enter a parameter using the numerical keypad.
- Press the ENTER key to store the new parameters.

Item Parameter

ExamplePT

While preset tare setting (Example)- Enter a setting value using the numerical keypad.
- Press the ENTER key to store the new tare.
Hold display (Example)
The hold display is set using Hold of the function table.
When the weight value is "Near zero" (within the zeroband) or changes more than 25% +30 digits, the hold iscanceled.
Weighing error
Check the base unit and the weighing pan.

Overload display
Remove anything that is on the pan.
Calibration error
The calibration mass is too light.
Check the base unit and the weighing pan.
The "digit" is a unit of display, and is equivalent to the minimum measurable mass.
The "Near zero" or zero band is within ±4 digits from zero point in the unit of kg.

Page 18

HV/HW-CEP Series

A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A

A
A
Calibration errorA
The calibration mass is too heavy.
A
Check the base unit and the weighing pan.
A
Does not display zero when the scale is turned on.
A
Remove anything that is on the weighing pan.
A
Perform zero point calibration.
AOr
The weight value is unstable due to drift or vibration A

Display and Symbols

Description

when the scale is turned on.
A breeze or vibration may be affecting the measurement.
Check around the weighing pan.
Blinking

M+

Blinking

M+

Accumulated data countand lighting up

Total mass value of the accumulated data

The weighing range and measurable minimum mass.
Example: Displays the weight value by 5 g up to 15 kg.
e.g. CAP. MAX. 3/6/15kg d=1/2/5g
Displays the weight value by 2 g up to 6 kg.
Displays the weight value by 1 g up to 3 kg.

7.2.

A
A
A
A
A
A

Keys

Display and Symbols

A
A
ON/OFF key
The scale is in standby status when power is connected using A
the AC adapter.
A
ZERO keyA
The ZERO key will zero the scale if the weight value is within A±2% of the weighing capacity (kg) around the power-on zero A
point. The zero point indicator turns on (Zero operation).
A
If the scale is in tare in this time, the tare value is cleared.
Description

When accumulation is displayed, the accumulation is cleared.
TARE keyA
The TARE key will tare the scale and subtract the weight to A
zero as a tare (container) weight when the weight is a plus A
stable value. In this case the zero point indicator and net
indicator turn on. (TARE operation)
When the tare is removed while the tare operation is inprogress and the scale returns to the zero point, the zero point
indicator and net indicator turn on.
In this case the displayed tare value will be negative.
Note The tare reduces the weighing range.
Adds to the accumulated data.

HV/HW-CEP Series

Page 19

Display and Symbols

Description
SET key
When setting the upper/lower limit, switch between + and -.

Press
and
hold

and
press

Enters preset tare setting mode

Press
and
hold

and
press

Displays the accumulated results

Press
and
hold

and
press

Sets upper/lower limit values for comparator

Press
and
hold

and
press

Proceeds to unit mass storing when using counting mode
MODE key- Switches the mode (unit) to be displayed- The mode (unit) is maintained in non-volatile memory, so
the scale displays using the most recently used mode (unit)
when turning on the power next time
- Used as key to select the items at each setting.
HOLD key
Holds the display. Refer to function settings for details.

Not used in the HV/HW-CEP series
Display off
Press
and
hold

and
press

Enters function settings

Page 20

HV/HW-CEP Series

8. Basic Operation
8.1.

Turning the Scale on/off and Weighing

8.1.1. When Using Batteries
Step 1 Install four new batteries. Refer to "5.2. Installing the Batteries".
Step 2 Confirm that nothing is placed on the pan.
Step 3 The scale turns on/off using the ON/OFF key alternately.
Step 4 Check the accuracy of weighing. Allow 30-minute warm up before calibration.
Step 5 With nothing on the pan, press the ZERO key to display zero.
Step 6 Place an article to be weighed on the pan gently.
Step 7 Wait for the stability mark to be displayed. Read the weight value.
Step 8 Remove the article from the pan.
Step 9 Turn the scale off using the ON/OFF key.

CautionWhenis displayed, this means the batteries will run out soon.
Replace used batteries with four new ones when “lb” is displayed.
Battery life depends on the ambient temperature.
Remove the batteries from the display unit when the scale is not to be used
for a long time. The batteries may leak and cause damage.

HV/HW-CEP Series

Page 21

8.2.

Tare (And Net Display)

The "tare" is used to cancel the mass of a container, which is placed on the pan tocontain the article to be weighed.

Caution

The tare reduces the weighing range.
The current tare value is reset by pressing the ZERO key or turning the scale
off. (Reset value is zero.)
The storable preset tare value must be within the minimum weighing range.

8.2.1. Tare Input by Weighing
Step 1 Place the container on the pan.
Step 2 Wait for the stability mark to be displayed. Press the TARE key. The display
becomes zero and the net mark is displayed.
Step 3 Place an article to be weighed into the container. Wait for the stability mark to be
displayed and read its net display.
Step 4 Remove the article and the container from the pan.

8.2.2. Digital Input (Preset Tare)
Step 1 Press and hold the SET key and press the TARE key.
Then the blank or stored tare value is displayed. The blank display means that thetare value is zero (reset value). And
PT blinks.
Step 2 Enter the preset tare value using the numerical keypad.
Step 3 Press the ENTER key to store the new preset tare value.
Then the scale displays the net value (the gross weight value minus the tare value).
Step 4 Place an article to be weighed into the container. Wait for the stability mark to be
displayed and read its net display.
Step 5 Remove the article and the container from the pan.

Page 22

HV/HW-CEP Series

8.3.

Switching the mode+

kg weighing

Unit mass storing mode

Counting mode

Ib weighing

OZ weighing* Ib-OZ display is available only with
HV-15KCEP and HW-10KCEP

Ib-OZ weighing+
Number of

Accumulation
value *1

accumulations *1+
Upper/lower limit values
setting mode+
Preset tare setting mode

* The scale displays the number of accumulations and an accumulation value only whenaccumulation data exists.

HV/HW-CEP Series

Page 23

9. Counting Mode
The counting mode is the function to convert the total mass value (total weight) ofarticles to a count, when each article has the same mass value.
To use this function, store a unit mass in advance.
Even if the batteries are removed, the unit mass is maintained in non-volatilememory.

9.1.

Storing a Unit Mass

Step 1 Press the MODE key to display the unit.

Counting mode
Step 2 Press the SET and ENTER key to enter the modethat stores a unit mass.
STABLE

Step 3 Press the SET key to select the number of samples.
The greater the quantity of samples, the greater theaccuracy of the count.
5 pieces, 10 pieces, 20 pieces, 50 pieces,
100 pieces

STABLE

Number of samples
A container(bowl)

Step 4 Place the container on the pan.
Press the TARE key.

Pan

Step 5 Place the number of samples selected at step 3.
Wait for the stability mark to be displayed. Pressthe ENTER key to store. The count is displayed.

STABLE

Zeroing value

CautionWhen the sample total mass value is too small andit is not possible to calculate a unit mass, the scale
displays lo ut and returns to step 3. Increase thenumber of samples. More than 10 digits of the total
sample mass, in the unit of kg, are required.
When the unit mass is too light to store, the scale
displays lo ut . In this case, the unit mass will not be
stored even if the number of samples is increased.
Pressing the MODE key, after lo ut is displayed,displays the next unit.

Step 6 Remove the samples and the container from the pan.
Page 24

STABLECount

Note
The pan shape depends
on the scale model.

HV/HW-CEP Series

9.2.

Counting the Number of Articles

Step 1 Press the MODE key to display the unit.

Counting mode
Step 2 Store the unit mass of the article.
Refer to "9.1. Storing a Unit Mass."

Storing a unit mass
A container(bowl)

Step 3 Place the container on the pan.
Press the TARE key.

Pan

STABLE

Step 4 Place articles in the container. Wait for the stability
mark to be displayed and read the count.

Zeroing value

STABLE

Step 5 Remove the articles and the container from the pan.
Note
The pan shape depends
on the scale model.

HV/HW-CEP Series

Page 25

10.

Accumulation Function

This function counts the number of times articles are weighed, calculates the total mass
value and can display the number (accumulation count) and accumulated mass value.
The accumulation function is displayed with up to 6 digits. The balance can not display 7 ormore digits, therefore the leading digits are not displayed.
Example: With 60K type, when importing the data of 17 accumulations of up to 60kg (60.000
X 17=1020.000), the balance displays this as “020.000”.
To use this function, set the parameters of the "Accumulation function ( 5um )" in thefunction table in advance.
The accumulation count and accumulated mass value are stored in the scale even if
the power is removed.

Operation and Keys
The display of the accumulation count has a blinking

M+ without a unit.

The display of the accumulated mass value has a unit and a blinking

M+.

Press the SET and M+ key to display the accumulation count and accumulated mass
value.
Press the ZERO key in the accumulation function (with a blinkingM+) to reset thecurrent function (The count and accumulated mass value become zero.)

Caution

The accumulation function is available only when weighing is performed in thesame unit.

Parameter List and Word Definition

The "Near zero" is within ±4 digits from the zero point in the unit of kg.
The "digit" is a unit of display, and is equivalent to the minimum measurable mass.
The "zero point" is the fundamental starting point to weigh anything.

A
A
Accumulation function not used.
5um 0
The scale accumulates the data, if the M+ key is pressed, when A
the display is a positive stable value and not near zero. The next A
5um 1
accumulation can be performed after the display becomes near A
zero or a negative value.
A
The scale accumulates the data, if the M+ key is pressed, when the A
display is a stable value and not near zero. The next accumulation A
5um 2
can be performed after the display becomes near zero.
A
Function table
Description
A
Function table

5um 3

Description

When the display is a positive stable value and not near zero, thescale accumulates the data automatically. The next accumulation
can be performed after the display becomes near zero or anegative value.
Page 26

HV/HW-CEP Series

5um 4

HV/HW-CEP Series

When the display is a stable value and not near zero, the scale
accumulates the data automatically. The next accumulation can be
performed after the display becomes near zero.
Use
Recording the number and mass of articles removed fromthe pan. (Place the articles on the pan. Press the TARE key
at each removal.)

Page 27

A
A
A
A
A

11.

Comparator

Five-level, three-level and seven-level (portion weighing mode) comparators areavailable.
Each comparator mode compares the weight value against the preset limit values anddisplays the results using LEDs (yellow / green / red).
• Five-level comparator mode:
Uses four comparator values tocompare the weight value and displays
results in five levels of LOLO, LO, OK,
HI and HIHI.
• Three-level comparator mode:
Uses two comparator values (upperand lower limit values) to compare theweight value and displays results inthree levels of LO, OK and HI.
• Seven-level comparator mode (portion weighing mode):
Uses six comparator values to compare
the weight value and displays results inseven levels of over in the negativevalue, level 1 (LOLO), level 2 (LO),Level 3 (OK), level 4 (HI), level 5 (HIHI)
and over in the positive value.
 To use the comparator modes, the function settings “Cp-l” and “Cp” must be
specified and the comparator values must be set.
 Using the function setting “Cp-l”, select a comparator mode.
0: five-level comparator mode
1: three-level comparator mode
2: Seven-level comparator mode (portion weighing mode) Using the function setting “Cp”, select comparison conditions.
0: No comparison (comparator mode disabled).
1: To compare all data.
2: To compare all stable data.
3: To compare all data which are more than or equal to +5d, or less thanor equal to -5d.
4: To compare stable data which are more than or equal to +5d, or lessthan or equal to -5d.
5: To compare all data which are more than or equal to +5d.
6: To compare stable data which are more than or equal to +5d.
d = readability in kg (Refer to “15. Specifications”.)
Also in the counting mode, “d” is equal to the readability of kg mode.

Page 28

HV/HW-CEP Series11.1. The formula to compareComparison is performed using the formula listed below and the results are displayed.
Five-level comparator mode
Results
Comparison formula
LED display
Displayed
value
<
LOLO
limit
value
LOLO
(Or over in the negative value)( Red LED on)
LO

Displayed value < LO limit valueOKLO limit value ≤ Displayed value ≤ HI limit valueHIHI limit value < Displayed value

HIHI

HIHI limit value < Displayed value(Or over in the positive value)

Three-level comparator mode
Results
Comparison formula
Displayed value < LO limit valueLO(Or over in the negative value)OK

LO limit value ≤ Displayed value ≤ HI limit valueHIHI limit value < Displayed value(Or over in the positive value)

Seven-level comparator mode (portion weighing mode)
Results
Comparison formula
Displayed value < Level 1 lower limit valueNone(Or over in the negative value)
LOLO(Level 1) Displayed value < Level 2 lower limit value
LO(Level 2) Displayed value < Level 3 lower limit value
OK
Level 3 lower limit value ≤ Displayed value ≤ Level(Level 3) 3 upper limit value
HI(Level 4) Level 3 upper limit value < Displayed value
HIHI(Level 5) Level 4 upper limit value < Displayed value
Level 5 upper limit value < Displayed valueNone(Or over in the positive value)( Yellow LED on)( Green LED on)）( Yellow LED on)( Red LED on)
LED display( Red LED on)( Green LED on)）
( Red LED on)
LED display( No LEDs on)( Red LED on)( Yellow LED on)( Green LED on)）( Yellow LED on)( Red LED on)( No LEDs on)

 The comparator values are common to the weighing and counting mode.
 Ignore the decimal point when setting the comparator values.
 The comparator values are maintained even if the power is turned OFF.
 Judgment order of comparison is from the top row to the bottom in the comparator
mode tables
 The entered comparator values are not judged. Even if the upper limit value is lessthan the lower limit value, no error will be display.

HV/HW-CEP Series

Page 29

11.2. Entering the comparator values
1. In the weighing mode, press and hold the SET key and press the COMP key to enter
the comparator value setting mode.
2. Enter the comparator values using the following keys.
0 to 9 To enter numerical valueC
To cancel settings
SET
To switch between + and ENTER To store setting values* Each time the SET key is pressed, “－” switches between being lit and off at thefirst digit. “－”being lit means a minus setting.
3. When the setting is complete, “end” is displayed and the scale returns to the weighing
mode. (At this time, power-on-zero is not performed.)
Example of five-level
Example of three-level
Example of seven-levelcomparator modecomparator modecomparator mode (portion(upper/lower limit mode)
weighing mode)
With the HW-60KCEP, set as
follows.
LOLO 8.500 kg
LO 10.000 kg
HI 10.500 kg
HIHI 12.000 kg

With the HV-200KCEP, set as
follows.
LO 148.85 kg
HI 152.5 kg

Press

Press

+

To set level 1
lower limit vaue
Press

Press

To set level 2
lower limit vaue

To set the HI
limit vaue

To set the LO
limit vaue

Press

Press

Press

Press

+

To set the LO
limit vaue

To set the LOLO
limit vaue
Press

Weighing
mode

Weighing
mode

Weighing
mode+With the HV-15KCEP, set as follows.
Level 1 (LOLO) 0.500 kg or more
Level 2 (LO) 1.000 kg or more
Level 3 (OK) 1.500 kg to 2.000 kg
Level 4 (HI)up to 2.500 kg
Level 5 (HIHI)
up to 3.000 kg

To set level 3
lower limit vaue

To set the HI
limit vaue
Weighing
mode

Press

Press
To set level 3
upper limit vaue

To set the HIHI
limit vaue
Press

Press

* With the HV-C models, thescale changes thereadability digit depending
Weighing
on the display range.
modeEnter 0 for the final digit.

To set level 4
upper limit vaue
Press
To set level 5
upper limit vaue
Press

Weighing
mode

Page 30

HV/HW-CEP Series

12.

Auto-tare

The HV/HW-CEP series has an auto-tare function to be used with the comparator mode
enabled. Using this function in check weighing, the scale automatically tares, then displays “OK”
for a certain amount of sample and repeats this process for the next weighing.
Start with display zero after tare operation. Place or take away objects until thecomparison result will show OK. When the stable display is maintained for the duration
specified in the function setting “at-t”, the scale will automatically tare the weight, show
zero and be ready for next weighing.
 In some countries or areas, the auto-tare function can not be used on the Legal for Trademodels and the selection in the function settings “at”, “at-t” and “at-f” is not available.
 To use the auto-tare function, set the function settings below.

Cp 1:

Compare all weighing data (other settings may be used depending on theapplication).
at 1:
Auto-tare function enabled.
at-t 0 ~ 9: Select the timing to tare automatically to avoid the wrong tare operation, forexample; too early to tare, to take a longer time to go to the next weighing.
 Take-away check weighing “Cp-p 1” (Example with “Cp-l 0” setting)Take-away check weighing (negative comparison) is the way to compare thenegativeweightwhile
takingaway
objects
fromacontainer.
Set the function “Cp-p 1” together with the auto-tare function enabled “at 1”. In thisoperation mode, the scale operates as “take-away the objects”  “OK and stable” “auto-tare”  “take-away the objects”  ······.
In this setting, the polarity of LOLO, LO, HI, and HIHI limit values are ignored and thescale shows the comparator results as below.

Note: To start the take-away check weighing, be sure to use the TARE key to tare theweight of the container filled with objects. The ZERO key may zero the display,
and the scale goes below the zero point by taking out the objects. Then, theauto-tare function or TARE key does not work.
 When the function “at-f 1 Tares the initial (container) weight.” is selected:
To start the auto-tare function, usually the container (filled with objects) will be placed on theweighing pan and its weight must be tared using the TARE key. When the function “at-f1” is selected, the scale will tare the initial (container) weight automatically.
When all load on the weighing pan is removed, the scale will return to the zero point andthe tare weight will be automatically cleared. If the scale does not return to the zero point,press the ZERO key to clear the tare weight.

HV/HW-CEP Series

Page 31

13.

Calibration

The scale is an instrument which weighs the "weight" and displays its "mass".
Calibration is the adjustment function so that the scale can weigh correctly.
Three steps of calibration are available
Gravity Acceleration Correction ... The function to correct the scale’s local gravity
acceleration, so that the scale functions
correctly when the calibrated scale has been
moved to a distant place.
Refer to the "Gravity Acceleration Table" on
the next page.
Calibration of the Zero Point ......... The function to adjust the zero point, so thatthe zero point mark is displayed when there isnothing on the pan.
Comment The zero point is the fundamental starting point
to weigh anything and influences theperformance of scale.
Span Calibration .............................. The function to adjust the span with acalibrated mass, so that the scale canaccurately weigh anything within the weighing
capacity.
Comment Span means the range of weighing capacity.
Use a calibration mass heavier than two thirdsof the weighing capacity.

CautionCheck the accuracy of weighing periodically. Calibrate the scale, if it has
been moved to another location or the environment has changed.
Gravity acceleration correction is not required, when the scale is calibrated
with the calibration mass at the place where the scale is used.

Page 32

HV/HW-CEP Series13.1. Gravity Acceleration Table
Amsterdam
AthensAuckland, NZ
BangkokBirmingham
Brussels
Buenos Aires
Calcutta
ChicagoCopenhagen
CyprusDjakarta
Frankfurt
Glasgow
HavanaHelsinki
KuwaitLisbonLondon (Greenwich)
Los Angeles
Madrid

HV/HW-CEP Series

9.813
9.800
9.799
9.783
9.813
9.811
9.797
9.788
9.803
9.815
9.797
9.781
9.810
9.816
9.788
9.819
9.793
9.801
9.812
9.796
9.800

m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2

Page 33

ManilaMelbourne
MexicoMilan
New York
Oslo
OttawaParis
Rio de Janeiro
Rome
San Francisco
Singapore
Stockholm
SydneyTainanTaipeiTokyo
Vancouver, BC
Washington, DC
Wellington, NZ
Zurich

9.784
9.800
9.779
9.806
9.802
9.819
9.806
9.809
9.788
9.803
9.800
9.781
9.818
9.797
9.788
9.790
9.798
9.809
9.801
9.803
9.807

m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2
m/s2

13.2. Complete Calibration Procedure
13.2.1. Gravity Acceleration Correction
Step 1 Turn on the display.
Open the rear cover of the display unit. Locate the
CAL switch inside.
Press the CAL switch to enter the calibration mode.
Then Calis displayed.

CAL switch

Step 2 Press the MODE key to enter the gravity acceleration
correction mode. Press the ENTER key to make thenumerical value blink and proceed to input mode.

Step 3 Set your local gravity acceleration using the numerical
keypad.

Enter using numerical
keypadStep 4 Press the ENTER key to store the new value.
The display returns to gravity acceleration display
after displaying end .

Step 5 Press the CAL switch again to finish gravity
acceleration correction.

CAL switch
The power is automaticallyturned off.

13.2.2. Preparation

Step 6 Confirm the environmental conditions as follows:
Maintain a constant temperature and stable power.
Install the scale on a solid floor where there is no draft, vibration, strong magetic
fields or direct sunlight.
Refer to "3.2. Precautions on Installing the Scale".
Step 7 Keep the display turned on for at least 30 minutes to warm up the scale.
Page 34

HV/HW-CEP Series13.2.3. Calibration of the Zero
Point
Step 8 After 30-minute warm up, press theNote
The pan shape depends
on the scale model.

Weighing mode

CAL switch

CAL

switch to display Cal .
Press the ENTER switch to display Cal 0 .
Step 9 Confirm that nothing is placed on the pan.
Wait for the stability mark to be displayed.
Press the ENTER key.

STABLEThe scale stores the current condition as
the zero point.

Nothing on
the pan
STABLE

Step 10 Calibration of the zero point is finished. To
exit the calibration mode, proceed to step
14.

Stability mark13.2.4. Span Calibration
Step 11 Set the value of the calibration massusing the numerical keypad. (The initialvalue depends on the scale model.)Enter using numerical
keypadWeignt value
15kg
STABLE

Step 12 Place the mass on the pan which was set
at step 11. Wait for the stability mark to be
displayed. Press the ENTER key.

Stability markThe scale then calculates the span andstores it.
Step 13 The scale displays Cal after displaying
end
to finish setting.
Remove the mass from the pan.
Step 14 Press the CAL switch to finish calibraiton.

CAL switch
The power is automaticallyturned off.

HV/HW-CEP Series

Page 35

14.

Function Table

The function table is used to store and refer items that determine the performance ofthe scale. Each item has a parameter.
The parameters are stored in the scale even if the power is removed.

Item

Parameter14.1. Parameter Setting Procedure
Display off

Step 1 Turn off the display.
Press
Press and hold the TARE key and press the ON/OFF key and holdto display software version p-XXX .
Press the MODE key to enter function setting mode.
Class

ba5fncand press

.

Class

Step 2Press the MODE key to select the desired class, and thenpress the ENTER key.

Items

Step 3

The first item of class is displayed. Press the MODE
key to select the desired item, and then enter a value
using numerical keypad.

Step 4Press the ENTER key to store. The display returns toclass after displaying

Step 5Enter using
numerical
key padend.

Press the ZERO key to return to normal weighing mode.

Weighing value
Page 36

HV/HW-CEP Series14.2. Parameter List
Class

Items
Key locklock
Automatic
off
poff

Automaticon
p-on
Zero tracking
trc

Parameter
Details and usage■0
Functions with all the keys
1
Functions with the ON/OFF, ZERO, TARE, HOLD and M+ key
2
Functions with the ON/OFF, ZERO, TARE key
■ OFF
0
power
1
After 5 minutes
2
After 10 minutes
3
After 15 minutes
4
After 30 minutes
5
After 60 minutes
■
power
0
OFF

ba5fncStability band width
5t-b
Stability band time
5t-t
Weighing
stabilization level
cond

Backlight control
l-it

Decimal point
pnt
Automatic tare
at

1

ON

0
1 ■
0 ■
1
2
0
1 ■
2
0
1
2 ■
3
4
0
1
2 ■
3
4
5
0 ■
1
0 ■
1

OFF
ON
±0.5 digit±1 digit±2 digits
0.5 second
1.0 second
1.5 seconds
Fast

Slow
Always turned off
Always lit
Turns off 5 seconds after stabilizing
Turns off 10 seconds after stabilizing
Turns off 15 seconds after stabilizing
Turns off 30 seconds after stabilizing
Dot (.)
Comma (,)
OFFON

: factory settings"Near zero" is within ±4 digits (four times the minimum mass that can be weighed) fromzero point in the unit kg.
■

HV/HW-CEP Series

Page 37

Bba5fnc

Class

Items
Parameter
Details and usage
Interval until making
00 second
automatic tare1
0.5 second■
at-t
2
1.0 second
3
1.5 seconds
4
2.0 seconds
5
2.5 seconds
6
3.0 seconds
7
4.0 seconds
8
5.0 seconds
9
10 seconds■
0OFF
Tare on initial load
at-f
1
ON
0
OFF
Accumulation mode
1 ■ Accumulates by M+ key when the value is +, excluding near zero
5um
2
Accumulates by M+ key when the value is + or -, excluding near zero3
Accumulates automatically when the value is +, excluding near zero
4
Accumulates automatically when the value is + or -, excluding near zero■0OFF
Hold condition
1
Holds or releases by the HOLD key
Hold
2
Automatically holds when detecting a stabilization■: factory settings"Near zero" is within ±4 digits (four times the minimum mass that can be weighed) fromzero point in the unit kg.

Cp fnc

Class

Items

Parameter
Details and usage■ Five-level comparator
0Comparator
1
Three-level comparator (upper and lower limits)
judgment value
Cp-l
2
Seven-level comparator (portion weighing)0OFF■ Compares all
1
Comparator
2
Compares all the stabilized data
judgment condition
3
Compares all the data, excluding from –4 digits to +4 digitsCp
4Compares the stabilized data, excluding from –4 digits to +4 digits
5Compares all the data more than +4 digits
6
Compares the stabilized data more than +4 digits■
0OFF
Comparator reversal
Cp-p
1
ON
All LEDs upon
the power on
Cp-d■

0 ■ ON
1

OFF: factory settings

Page 38

HV/HW-CEP Series

15.

Specifications

HV-CEP Series (Weighing capacity of 15 kg to 220 kg)
Models
Weighing Capacity

HV-15KCEP

HV-60KCEP

HV-200KCEP

[kg]

3

6

15

15

30

60

60

150

220

Min. weighing (1 digit) [g]

1

2

5

5

10

20

20

50

100

6

15

30

30

60

150

150

300

500

Weighing Capacity

[lb]

#

Min. weighing (1 digit) [lb]#0.002 0.005

0.01

0.01

0.02

0.05

0.05

0.1

0.2

Weighing Capacity

[oz]

#

96

240

480

480

960

2400

2400

4800

8000

Min. weighing (1 digit) [oz]

#

0.05

0.1

0.2

0.2

0.5

1

1

2

5

Weighing Capacity[lb_oz] #30 lbMin. weighing (1 digit) [lb_oz] #Number of samples in counting
mode [pieces]Max. count number [pieces]

0.1 oz5（can be changed to 10､20､50､100）
150,000

Linearity [g]Span drift
Power source

120,000

Ａ

110,000

Ａ

7 segment LCD, Character height 25 mm,
3 color 5 level comparator LED,
Display refreshment rate: 10 times per second

Display
Repeatability [g]

---

Ａ

Ａ
±1
±2
±5
±5
±10 ±20 ±20 ±50 ±100 Ａ
20ppm/°C typ. (5°C to 35°C)
Ａ
Ａ
Battery
TYPE D (LR20) x 4
Ａ
DURACELL MN1300, ENERGIZER E95, Panasonic LR20(XJ)
Ａ
±1

±2

±5

±5

±10

±20

±20

±50

±100

Battery life (Approximately)

1500 hours (LED and Backlight off)
1000 hours (LED and Backlight on)

Ambient temperature and humidity-10°
C to 40°
C, Less than 85%R.H. (Do not allow condensation)Ａ

Weighing pan size [mm]250 x 250

330 x 424

390 x 530

Dimensions [mm]Width x Depth x Height

250 x 439 x 397

330 x 585 x 776

390 x 691 x 776

Weight[ kg ]

8

12

18# : If the law in your area permits, you can use these units.

HV/HW-CEP Series

Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ

Page 39ＡＡＡＡＡHW-CEP Series (Weighing capacity of 10 kg to 220 kg)
Models

HW-10KCEP

HW-60KCEP

HW-100KCEP

HW-200KCEP

[kg]

10

60

100

220

Min. weighing (1 digit) [g]

1

5

10

20

Weighing Capacity
Weighing Capacity

[lb]

#

20

150

200

500

Min. weighing (1 digit) [lb]

#

0.002

0.01

0.02

0.05

Weighing Capacity

[oz]

#

320

2400

3200

8000

Min. weighing (1 digit) [oz]

#

0.05

0.2

0.5

1

[lb_oz] #20 lbMin. weighing (1 digit) [lb_oz] #Number of samples in counting
mode [pieces]Max. count number [pieces]

0.1 ozWeighing Capacity--5（can be changed to 10､20､50､100）

100,000

120,000

100,000

Ａ
110,000

7 segment LCD, Character height 25 mm,
3 color 5 level comparator LED,
Display refreshment rate: 10 times per second

Display
±2

±10

±20

±40

Linearity [g]

±2

±10

±20

±40

Power source

20ppm/°C typ. (5°C to 35°
C)
Battery
TYPE D (LR20) x 4
DURACELL MN1300, ENERGIZER E95, Panasonic LR20(XJ)

Battery life (Approximately)

1500 hours (LED and Backlight off)
1000 hours (LED and Backlight on)

Ambient temperature and humidity-10°
C to 40°
C, Less than 85%R.H. (Do not allow condensation)

Weighing pan size [mm]Dimensions [mm]Width x Depth x Height
Weight[ kg ]Ａ
Ａ

Repeatability [g]Span drift

Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ

Ａ
Ａ
Ａ
Ａ
Ａ
Ａ
Ａ

Ａ
Ａ
Ａ
250 x 439 x 397 330 x 585 x 776 390 x 691 x 776 390 x 691 x 776
Ａ
250 x 250

330 x 424

390 x 530

390 x 530

8

12

18

18

# : If the law in your area permits, you can use these units.

Page 40

HV/HW-CEP SeriesＡDimensions
Models250

HV-15KCEP
HW-10KCEP

250

214

108
60

397

439

180

35

180

201

Models330

HV-60KCEP
HW-60KCEP

424
214

105
55

776

585

240

45

334

187

Unit: mm
HV/HW-CEP Series

Page 41Models

HV-200KCEP
HW-100KCEP

390

HW-200KCEP

530

214

129
70

776

691

300

45

440

187

Unit: mm

Page 42

HV/HW-CEP Series

16.

Maintenance

Refer to "3. Precautions" for use.
Refer to "7.1. Display and Symbols " for displayed error code.
Refer to "13. Calibration (Adjusting the Scale)" for precision weighing.
Periodically check the accuracy of the weighing. Calibrate the scale, if it has been
moved to another location or the environment has changed.

16.1. Check Points Before Calling Maintenance Service
In this situation

Confirm these items

A
A
A
A
A

Nothing is displayed.
Scale does not turn on.

Are the batteries dead ?
Is the battery direction correct ?

The scale does not display
zero upon power-on.

Check around the weighing pan.
Is there anything on the pan?
Perform zero point calibration.

A
A------ is displayed anddoes not proceed.

The weight value is unstable due to drift, vibrationor other. A breeze or vibration may be affecting themeasurement. Check around the weighing pan.
Check the connection of load cell cable.
No zero display when the display is turned on.
Remove anything that is on the pan.
Perform zero point calibration.

A
A
A
A
A
A
A
A
A
A
A
A
A
A
A
A

Cal e is displayed

Calibration error that means "Too heavy".

-Cal e is displayed

Calibration error that means "Too light".

eis displayed

Weighing error that means "Overload".

-eis displayed

Weighing error that means "Underload".

Fixed display

Do you use the hold function?
Turn off the scale and turn it on.

16.1.1. Repair
Do not disassemble or assemble the scale without an authorized service engineer. It
may cause an electrical shock or damage to the scale. In this case, repair is notcovered under warranty. Contact your local A&D dealer if your scale needs service orrepair.
HV/HW-CEP Series

Page 43

MEMO

Page 44

HV/HW-CEP Series

MEMO

HV/HW-CEP Series

Page 45

MEMO

Page 46

HV/HW-CEP Series

THIS PAGE INTENTIONALLY LEFT BLANKBack cover: A&D Global

A&D Company, Limited

3-23-14 Higashi-Ikebukuro, Toshima-ku, Tokyo 170-0013, JAPAN
Telephone: [81] (3) 5391-6132
Fax: [81] (3) 5391-1566

A&D ENGINEERING, INC.

47747 Warm Springs Blvd, Fremont, California 94539, U.S.A.
Tel: [1] (800) 726-3364
Weighing Support:[1] (888) 726-5931

Inspection Support:[1] (855) 332-8815

A&D INSTRUMENTS LIMITED

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY United Kingdom
Telephone: [44] (1235) 550420
Fax: [44] (1235) 550485

A&D AUSTRALASIA PTY LTD

32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100
Fax: [61] (8) 8352-7409

A&D KOREA Limited

한국에이.엔.디(주)OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"서울특별시 영등포구 국제금융로6길33 (여의도동) 맨하탄빌딩 817 우편 번호 07331( 817, Manhattan Bldg., 33. Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331 Korea )전화: [82] (2) 780-4101
팩스: [82] (2) 782-4264

Почтовый адрес:121357, Российская Федерация, г.Москва, ул. Верейская, дом 17Юридический адрес: 117545, Российская Федерация, г. Москва, ул. Дорожная, д.3, корп.6, комн. 8б
( 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Instruments India Private Limited

D-48( D-48, Udyog Vihar, Phase- , Gurgaon - 122016, Haryana, India ): [91] (124) 4715555: [91] (124) 4715599

A&D SCIENTECH TAIWAN LIMITED. A&D台灣分公司 艾安得股份有限公司
台湾台北市中正區青島東路5 號4 樓
( 4F No.5 Ching Tao East Road, Taipei Taiwan R.O.C. )Tel : [886](02) 2322-4722
Fax : [886](02) 2392-1794

A&D INSTRUMENTS (THAILAND) LIMITEDบริษัท เอ แอนด์ ดี อินสทรูเม้นท์ (ไทยแลนด์) จำกัด
168/16 หมู่ที่ 1 ตำบลรังสิต อำเภอธัญบุรี จังหวัดปทุมธานี 12110 ประเทศไทย
( 168/16 Moo 1, Rangsit, Thanyaburi, Pathumthani 12110 Thailand )Tel : [66] 20038911