PicoVNA®

Vector Network Analyzer

Quick Start Guide빠른 시작 가이드クイックスタートガイドКраткое руководство пользователя
快速入门指南PicoVNA Vector Network Analyzer Quick Start Guide

CONTENTS
English..................................................................................................................... 1
1.
Introduction.................................................................................................. 1
2.
Safety information....................................................................................... 1
3.
System requirements.................................................................................. 54.
Contents of your PicoVNA kit..................................................................... 55.
External connections................................................................................... 6
6.
Installing the PicoVNA vector network analyzer....................................... 7
7.
Further information................................................................................... 12
한국어............................................................................................................................. 13
1.
소개.................................................................................................................... 132.
안전 정보............................................................................................................ 133.
시스템 요구사항.................................................................................................. 17
4.
PicoVNA 키트의 내용물....................................................................................... 175.
외부 연결............................................................................................................ 18
6.
PicoVNA 벡터 네트워크 분석기 설치.................................................................... 197.
추가 정보............................................................................................................ 24
日本語............................................................................................................................ 25
1.
はじめに............................................................................................................. 25
2.
安全に関する情報............................................................................................... 25
3.
システム要件...................................................................................................... 29
4.
PicoVNAキットの同梱物..................................................................................... 29
5.
外部接続............................................................................................................ 30
6.
PicoVNAベクトルネットワークアナライザーの設置.............................................. 31
7.
詳細な情報......................................................................................................... 36Русский................................................................................................................. 371.
Введение................................................................................................... 372.
Информация по технике безопасности............................................... 373.
Системные требования.......................................................................... 424.
Комплект поставки PicoVNA................................................................. 425.
Внешние соединители............................................................................ 436.
Установка векторного анализатора цепей PicoVNA......................... 447.
Подробная информация......................................................................... 49
简体中文........................................................................................................................ 51
1.
简介................................................................................................................... 51
2.
安全信息............................................................................................................ 51
3.
系统要求............................................................................................................ 55
4.
PicoVNA 套件内容............................................................................................. 55
5.
外部连接............................................................................................................ 56
6.
安装 PicoVNA 矢量网络分析仪........................................................................... 57
7.
更多信息............................................................................................................ 62
DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

iPicoVNA Vector Network Analyzer Quick Start Guideii

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA Vector Network Analyzer Quick Start Guide

1. Introduction

EnglishThank you for purchasing a PicoVNA vector network analyzer.
The following models are available:•

PicoVNA 106: a 6 GHz two-port VNA with built-in bias-T networks•

PicoVNA 108: an 8.5 GHz two-port VNA with built-in bias-T networks, mixer
measurements and save on trigger

For further setup information following basic installation, see the User’s Guide suppliedwith the PicoVNA 2 [108: PicoVNA 3]* software.
For detailed specifications see the PicoVNA Data Sheet.
* Unless otherwise stated, information in this guide applies to all models in the
PicoVNA series. Information that applies exclusively to the PicoVNA 108 is written like
this: “[108: <information>]”.

2. Safety information
To prevent possible electrical shock, fire, personal injury, or damage to the product,
carefully read this safety information before attempting to install or use the product. In
addition, follow all generally accepted safety practices and procedures for working withand near electricity.
This instrument and its accessories have been designed to meet the requirements of
EN 61010-1 (Safety Requirements for Electrical Equipment for Measurement, Control
and Laboratory Use).
The following safety descriptions are found throughout this guide:
A WARNING identifies conditions or practices that could result in injury or death.
A CAUTION identifies conditions or practices that could result in damage to the
product or equipment to which it is connected.
WARNING
This product is for professional use by trained and qualified technicians only.
To prevent injury or death, use the product only as instructed and use only
accessories that have been supplied or recommended by Pico Technology.
Protection provided by the product may be impaired if used in a manner not
specified by the manufacturer.

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

1

PicoVNA Vector Network Analyzer Quick Start Guide

2.1 Symbols
These safety and electrical symbols may appear on the product or in this guide.
Symbols

Description

This terminal can be used tomake a measurement groundconnection. It is NOT a safety orprotective earth.

Earth (ground) terminal

Chassis terminal
Possibility of electric shockAppearance on the productindicates a need to read this safety
and operation manual.

Caution
Do not dispose of this product as
unsorted municipal waste

2.2 Maximum input and output ranges
WARNINGTo prevent electric shock, do not attempt to measure or apply signal levels
outside the specified maxima below.
The table below indicates the maximum voltage of the outputs and the overvoltage
protection range for input on the VNA and its calibration accessories. The overvoltage
protection ranges are the maximum voltages that can be applied without damaging the
instrument or accessory.
Instrument connectors
Ports 1 and 2
Bias-Ts 1 and 2
Trigger and reference in
Trigger and reference out
Calibration accessory connectors
SOLT-PREM and SOLT-STD ports
SOLT-AUTO ports A/B

Maximum operating voltage(output or input)+10 dBm(about 710 mV RMS)
±15 V DC
0 V to +5 V

Overvoltage or overcurrent
protection+20 dBm(about 2.2 V RMS)
250 mA±6 V pk
Do not apply a voltage
Overvoltage protection+20 dBm+20 dBmWARNING

Signals exceeding the voltage limits in the table below are defined as“hazardous live” by EN 61010.
±60 V DC
2

Signal voltage limits of EN 61010-1:2010+A1:2019
30 V AC RMS± 42.4 V pk max.

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA Vector Network Analyzer Quick Start Guide
WARNINGTo prevent electric shock, take all necessary safety precautions whenworking on equipment where hazardous live voltages may be present.
WARNINGTo avoid equipment damage and possible injury, do not operate
the instrument or an accessory outside its rated supply voltages orenvironmental range.
CAUTION
Exceeding the overvoltage protection range on any connector can cause
permanent damage to the instrument and other connected equipment.
To prevent permanent damage, do not apply an input voltage to the trigger
or reference output of the VNA.

2.3 Grounding
WARNING

The instrument’s or SOLT-AUTO E-Cal module’s ground connection
through the USB cable is for functional purposes only. The instrument andaccessories do not have protective safety grounds.
To prevent injury or death, or permanent damage to the instrument, neverconnect the ground of an input or output (chassis) to any electrical powersource. To prevent personal injury or death, use a voltmeter to check that
there is no significant AC or DC voltage between the instrument’s groundand the point to which you intend to connect it.
CAUTIONTo prevent signal degradation caused by poor grounding, always use thehigh-quality USB cable supplied with the instrument.

2.4 External connections
WARNINGTo prevent injury or death, use only the power adaptor supplied with the
instrument. This is approved for the voltage and plug configuration in yourcountry.
PicoVNA model

USB connection

PicoVNA 106
PicoVNA 108
SOLT-AUTO E-Cal module

DO296-6

USB 2.0
Compatible withUSB 3.0

Voltage

External power supply
Current
Total power

12 to 15 V DC

1.85 A pk

22 W
25 W

N.A.

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

3

PicoVNA Vector Network Analyzer Quick Start Guide

WARNING
Containment of radio frequencies
The instrument contains a swept or CW radio frequency signal source(300 kHz to 6.02 GHz at +6 dBm max. [108: 300 kHz to 8.50 GHz at +6 dBmmax.]) The instrument and supplied accessories are designed to contain andnot radiate (or be susceptible to) radio frequencies that could interfere withthe operation of other equipment or radio control and communications. To
prevent injury or death, connect only to appropriately specified connectors,
cables, accessories and test devices, and do not connect to an antenna
except within approved test facilities or under otherwise controlled
conditions.

2.5 Environment
WARNING
This product is suitable for indoor or outdoor use, in dry locations only. The
product’s external mains power supply is for indoor use only.
WARNINGTo prevent injury or death, do not use the VNA or accessory in wet or damp
conditions, or near explosive gas or vapor.
CAUTIONTo prevent damage, always use and store your VNA or accessory inappropriate environments.
Temperature
Humidity
Altitude
Pollution degree

Storage
Operating–20 °C to +50 °C+5 °C to +40 °C
Up to 80% RH (non-condensing)2000 m2

CAUTION
Do not block the air vents at the back of the instrument as overheating willcause damage.
Do not insert any objects through the air vents as internal interference willcause damage.

2.6 Care of the product
The product and accessories contain no user-serviceable parts. Repair, servicing andcalibration require specialized test equipment and must only be performed by Pico
Technology or an approved service provider. There may be a charge for these services
unless covered by the Pico three-year warranty.
WARNINGTo prevent injury or death, do not use the VNA or accessory if it appears to
be damaged in any way, and stop use immediately if you are concerned byany abnormal behavior.
4

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA Vector Network Analyzer Quick Start Guide

CAUTION
Regularly inspect the instrument and all probes, connectors, cables andaccessories before use for signs of damage or contamination.
To prevent damage to the device or connected equipment, do not tamper
with or disassemble the instrument, case parts, connectors, or accessories.
When cleaning the product, use a soft cloth and a solution of mild soap
or detergent in water, and do not allow liquids to enter the casing of the
instrument or accessory.
Take care to avoid mechanical stress or tight bend radii for all connectedleads, including all coaxial leads and connectors. Mishandling will cause
deformation of sidewalls, and will degrade performance. In particular, notethat test port leads should not be formed to tighter than 5 cm (2”) bend
radius.
To prevent measurement errors and extend the useful life of test leads andaccessory connectors, ensure that liquid and particulate contaminantscannot enter. Always fit the dust caps provided and use the correct torquewhen tightening. Pico recommends: 1 Nm (8.85 inch-lb) for supplied and allstainless steel connectors, or 0.452 Nm (4.0 inch-lb) when a brass or goldplated connector is interfaced.

3. System requirements
To ensure that the PicoVNA vector network analyzer and PicoVNA software operatecorrectly, you must use a computer with the system requirements and one of theoperating systems shown in the table below.
Processor, memory, free disk space
Operating system
Ports
Display resolution

As required by the operating system
Microsoft Windows 7, 8 or 10. 32-bit or 64-bit
USB 2.0 (also compatible with USB 3.0)
1280 x 720 minimum

4. Contents of your PicoVNA kit
The PicoVNA vector network analyzer kit contains the following items:
•
•
•
•
•
•
•

PicoVNA 106 [108: PicoVNA 108] vector network analyzer
PicoVNA 2 [108: PicoVNA 3] USB flash drive
DC power supply with four international plug adaptors
USB 2.0 cable, 1.8 m
SMA/PC3.5/2.92 and N-type fixed multi-wrench
Quick Start Guide
Storage and carry caseSome product packs may contain additional items. For a full list of all available kits andadditional accessories, see the User’s Guide or visit www.picotech.com.
DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

5

PicoVNA Vector Network Analyzer Quick Start Guide

5. External connections
5.1 PicoVNA 106 and 108 vector network analyzers

Port 1

Test
I/O

Bias-T
input

Sweep
indicator

Sweep
indicator

USB connection

DC input

10 MHz In

Bias-T
input

Test
I/O

Port 2

On/off switch
Trig In

S/N
OFF

10 MHz Out

12–15 V DC

Trig OutON

10 MHz reference in/out

Chassis ground

Trigger in/out

5.2 SOLT-STD-M/F and SOLT-PREM-M/F manual calibration standards
Load

Open

Through

Short

5.3 SOLT-AUTO-M/F automatic (E-Cal) calibration standards
Indicator

USB port

Port APort B
6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA Vector Network Analyzer Quick Start Guide

6. Installing the PicoVNA vector network analyzer
6.1 Installing the software
1.

Obtain the PicoVNA 2 [108: PicoVNA 3] software installer from the USB flash drive
supplied with your VNA or from:			
www.picotech.com/downloads.

2.

Run the installer (right-click and Run as administrator) and ensure that the
installation was successful.

3.

Connect the PicoVNA unit to the computer and wait while Windows automaticallyinstalls the driver.

The installer creates a support directory at:
C:\Users\<user name>\Documents\Pico Technology\PICOVNA2[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3]
This directory contains the following files (PicoVNA 106 example shown):•••

xxxx-log.txt
CalKits\DefUnc.dat
DefCal.cal•

FactoryDefaults.cal

This is the status log file. ‘xxxx’ is the serial number
Default measurement uncertainty values
Default calibration data (last used calibration)– will appear after first connection to a VNA device
Backup calibration data from factory

On Windows 7 machines it is common to see the following error message:

It is safe to click
Ignore to continue

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

7

PicoVNA Vector Network Analyzer Quick Start Guide

6.2 The PicoVNA 2 [108: PicoVNA 3] main window

You can adjust the vertical scale of any of the rectangular plots in the above window, as
follows:
Drag to adjust

8

Drag to adjust or
right-click and type

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA Vector Network Analyzer Quick Start Guide

6.3 Loading the calibration kit(s)PicoVNA 106 or 108

PicoVNA 106 or 108

PicoVNA 106 or 108

Port 1

Port 1

Port 1Maleconnector

Port 2InsertabledeviceNon-insertable device(test ports will connect directly together)FemaleconnectorMaleconnectorMaleconnector

Male cable.
Use female cal kit.

Port 2

Non-insertable device(test ports will not connect directly together)Femaleconnector

Femaleconnector

DUT

DUT

Female cables.
Use same male kit
for both ports.

Male cables. Usesame female kit
for both ports.

DUT
Female cable.
Use male cal kit.

Port 2

Calibrate using fixed manual
calibration standards:

Calibrate using fixed manual
calibration standards:

Calibrate using fixed manual
calibration standards:

SOLT-STD-F and SOLT-STD-M
or
SOLT-PREM-F and SOLT-PREM-M

SOLT-STD-M
or
SOLT-PREM-M

SOLT-STD-For
SOLT-PREM-F

Or calibrate with PicoVNA 108 andautomatic E-Cal Standard:

Or calibrate with PicoVNA 108 andautomatic E-Cal Standard:

Or calibrate with PicoVNA 108 andautomatic E-Cal Standard:

SOLT-AUTO-For
SOLT-AUTO-M

SOLT-AUTO-M

SOLT-AUTO-F

For a manual SOLT calibration:••••

Run the PicoVNA 2 [108: PicoVNA 3] software
In the main menu, select Tools > Calibration kit
Click Load P1 kit, locate the data file for your Port 1 cal kit (on the USB flash drive
supplied with the cal kit) and then click Apply
If a different cal kit is to be used on Port 2, click Load P2 kit, select the data foryour Port 2 kit and then click Apply

Select the calibration kit(s) required depending on the device to be tested. For quicker
access in future, you may wish to copy the cal kit data to your
C:\Users\<user name>\Documents\Pico Technology\PICOVNA2\CalKits\[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3\CalKits\]
folder. You can do this using the Save Kit button in the Cal Kit Editor.
For the PicoVNA 108 and an automated calibration:
•
•
•
•
•
•
•
•
•

Run the PicoVNA 3 software.
Connect the E-Cal module to a spare USB port on the controlling PC.
In the main menu, select Tools > Calibration kit.
Click Electronic Cal kit.
Select the E-Cal module and Port Adapter arrangement to suit the Test Ports.
Connect the E-Cal module, port adapter and test ports to exactly match theselected configuration.
Click Find and load EasyCal data.
Upon completion select Exit.
Click Apply in the Calibration Kit Parameters menu.

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

9

PicoVNA Vector Network Analyzer Quick Start Guide

6.4 Setting the resolution
Before proceeding with a calibration, click Enhance in the main window to set theminimum resolution bandwidth that you need the calibration to support. Lower values
will result in a slower calibration.
Measurement
Fastest speed
Best accuracy and~100 dB dynamic range
General use, fast speed,~90 dB dynamic range
Best dynamic range

Calibration Calibration Calibration
bandwidth averaging
power
10 kHzNone+0 dBm100 HzNone–3 dBm

1 kHz

None

+0 dBm

10 Hz

None

+6 dBmComments
Set bandwidth to 140 kHz
during measurement
Leave bandwidth set to 100 Hz
during measurement
Leave bandwidth set to 1 kHz
during measurement
Leave bandwidth set to 10 Hz
during measurement

6.5 Setting up calibration parameters
Click Calibration to open the Calibration window:
The cal kitsthat you haveloaded

1. Set sweep
parameters

3. Select
measurement
required

2. After setting therequired sweep
bandwidth, apply
values
5. IMPORTANT!
On completion, apply calibration

10

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

4. Perform required
calibration steps.
Any sequence ispossible, but
down-then-acrossorder usually
minimizes thenumber ofconnection steps.

DO296-6

PicoVNA Vector Network Analyzer Quick Start Guide

6.6 Setting up display parameters
Click Display in the main window to open the Display Set Up window:
3. Select channel
to set up
1. Select
one-channel,
two-channel orfour-channelview
4. Check to show
vertical scalevalues on graphs

2. Selectchannel toshow in theMarkers panel
5. Apply settings before
selecting next channel

When finished, click Start in the main window to begin measurements.

6.7 Running in demo mode
Demo mode allows you to explore the PicoVNA 2 [108: PicoVNA 3] software without
the need to have an instrument running.
••

To enter demo mode, run the PicoVNA 2 [108: PicoVNA 3] software with no
instrument connected.
Click Ignore in the dialog that appears.

PicoVNA 2 [108: PicoVNA 3] will then offer you a selection of demonstration
measurements.

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

11

PicoVNA Vector Network Analyzer Quick Start Guide

7. Further information
7.1 Free newsletter
Pico Technology sends out a free monthly newsletter by email. This gives news of
product launches, software upgrades and application notes, as well as hints andadvice. We treat your e-mail address and other details as confidential and will not passthem to any third party. To subscribe, visit:www.picotech.com/signup

7.2 Software updates and replacement
The latest versions of all Pico software and manuals can be downloaded free of charge
at:www.picotech.com/downloads
If you require a new software disk, please contact Pico Technology or your distributor.
There may be a small charge for this service.

7.3 Writing your own software
Code examples in MATLAB and MATLAB RF toolbox, LabVIEW, C, C# and Python areavailable on GitHub (github.com/picotech). Examples include multiple instrument
addressing and control.

7.4 User’s and programmer’s guides
The PicoVNA User’s Guide and PicoVNA Programmer’s Guide are available on the USB
flash drive supplied and from:www.picotech.com/downloads

7.5 Specifications
The User’s Guide and Data Sheet contain the latest specifications for your PicoVNA
vector network analyzer.

7.6 Technical support
Regularly updated technical support information can be found in the Pico Technology
support forum:
www.picotech.com/tech-support

7.7 Warranty and returns
These PicoVNA vector network analyzers are supplied with a three-year
return-to-manufacturer warranty. For full terms and conditions, see the
PicoVNA User’s Guide.
12

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

YEAR W A R R A N T Y

DO296-6

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서

한국어

1. 소개

PicoVNA 벡터 네트워크 분석기를 구매해 주셔서 감사합니다.
제공되는 모델은 다음과 같습니다.
••PicoVNA 106: 바이어스-T 네트워크가 내장된 6 GHz 2포트 VNA

PicoVNA 108: 바이어스-T 네트워크, 믹서 측정, 트리거 시 저장 기능이 내장된 8.5 GHz
2포트 VNA기본 설치 후 추가 설정 정보는 PicoVNA 2 [108: PicoVNA 3]* 소프트웨어와 함께 제공되는사용 설명서를 참조하십시오.
자세한 사양은 PicoVNA 데이터 시트를 참조하십시오.

* 달리 명시되지 않는 한 이 설명서의 정보는 PicoVNA 시리즈의 모든 모델에 적용됩니다.
PicoVNA 108에만 적용되는 정보는 다음과 같이 표시됩니다: “[108: <정보>]”.

2. 안전 정보감전, 화재, 부상 사고 또는 제품 손상을 방지할 수 있도록 제품을 설치하거나 사용하기 전에 이안전 정보를 주의 깊게 읽으십시오. 또한 전기 근처에서 진행하거나 전기를 사용하는 작업에일반적으로 적용되는 모든 안전 방침과 절차를 따르십시오.
이 계측기와 부속품은 EN 61010-1 요건(측정, 제어 및 실험실용 전자 장비의 안전 요건)의요구 사항을 충족하도록 설계되었습니다.
다음은 이 가이드 전반에서 사용된 안전성에 대한 설명입니다.

경고는 부상 또는 사망을 초래할 수 있는 조건이나 사용 방법을 나타냅니다.

주의는 연결된 제품 또는 장비의 손상을 초래할 수 있는 조건이나 사용 방법을 나타냅니다.

경고본 제품은 전문적인 교육을 받고 해당 자격을 갖춘 전문 기술자만이 사용할 수
있습니다. 부상 또는 사망을 방지하기 위해 지침에 따라 제품을 사용하고 Pico
Technology에서 제공되거나 권장된 부속품만을 사용하십시오.  제조업체에서
지정하지 않은 방법으로 사용할 경우, 제품에서 제공하는 보호 기능이 훼손될 수있습니다.

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

13

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서

2.1 기호제품 또는 이 가이드에 표시된 안전 및 전기 기호는 다음과 같습니다.
기호설명측정 접지 연결을 위한 단자입니다.
안전 또는 보호 접지용으로는 사용할수 없습니다.

접지 단자
섀시 단자
감전 위험
주의
본 제품을 분류되지 않은생활폐기물로 폐기하지 마십시오제품에 표시된 경우, 이 안전 및 작동설명서를 읽어 보아야 합니다.

2.2 최대 입력/출력 범위경고감전 방지를 위해 아래 명시된 최대값을 벗어난 신호 레벨을 측정하거나 인가하지마십시오.

아래 표에는 VNA와 교정 부속품 입력에 대한 과전압 보호 범위와 최대 출력 전압이 나와
있습니다. 과전압 보호 범위는 기기나 부속품에 손상을 주지 않고 인가할 수 있는 최대 전압을말합니다.
기기 커넥터최대 작동 전압(출력/입력)포트 1 및 2

바이어스-T 1 및 2트리거 및 기준 입력트리거 및 기준 출력+10 dBm(약 710 mV RMS)
±15 V DC교정 부속품 커넥터
SOLT-PREM 및 SOLT-STD 포트
SOLT-AUTO 포트 A/B경고

0 V ~ +5 V

과전압/과전류 보호+20 dBm.
(약 2.2 V RMS)
250 mA±6 V pk가압 금지과전압 보호
+20 dBm+20 dBm아래 표의 전압 한계를 초과한 신호는 EN 61010에 따라 “위험한 통전”으로정의됩니다.
±60 V DC

14

EN 61010-1:2010+A1:2019의 신호 전압 한계
30 V AC RMS± 42.4 V 피크 최대

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서경고위험한 통전 조건 전압이 흐를 수 있는 장비에서 작업할 때, 감전 사고를 방지하기위해 필요한 모든 안전 예방조치를 취하십시오.
경고장비 손상과 부상 위험을 방지하기 위해 정격 공급 전압 또는 환경 범위를 벗어나기기나 부속품을 작동하지 마십시오.

주의커넥터의 과전압 보호 범위를 초과할 경우 기기 및 기타 연결된 장비를 영구적으로
손상시킬 수 있습니다.

영구적인 손상 방지를 위해 VNA의 트리거 또는 기준 출력에 입력 전압을 인가하지마십시오.

2.3 접지 연결경고USB 케이블을 통한 기기 또는 SOLT-AUTO E-Cal 모듈의 접지 연결은기능용으로만 제공됩니다. 본 기기와 부속품에는 보호 안전 접지가 없습니다.
부상 또는 사망 또는 기기의 영구적인 손상을 방지하기 위해 입력/출력(섀시)의
접지를 전원에 절대 연결하지 마십시오. 부상 또는 사망 사고를 방지하기 위해,전압계를 사용하여 계측기 접지와 연결하려는 지점 사이에 유의미한 AC 또는 DC가흐르지 않는지 확인하십시오.
주의접지 불량으로 인한 신호 저하를 방지하기 위해 기기와 함께 제공된 고품질 USB케이블을 항상 사용하십시오.

2.4 외부 연결경고부상 또는 사망 사고를 방지하려면 기기와 함께 제공된 전압 어댑터만사용하십시오. 해당 국가의 전압 및 플러그 구성에 맞도록 승인된 어댑터입니다.

PicoVNA 모델

USB 연결

PicoVNA 108

USB 2.0 .
USB 3.0.
과 호환

PicoVNA 106

SOLT-AUTO E-Cal 모듈

DO296-6

전압

12 ~ 15 V DC외부 전원 공급 장치전류
1.85 A pk

총 전력

해당 없음

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

22 W
25 W

15

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서경고무선주파수 억제본 기기에는 스윕 또는 CW 무선주파수 신호 공급원(최대 +6 dBm에서 300 kHz ~6.02 GHz [108: 최대 +6 dBm에서 300 kHz ~ 8.50 GHz])이 포함되어 있습니다.
본 기기 및 제공된 부속품은 다른 장비 또는 전파 제어 및 통신의 작동을 방해할수 있는 무선주파수를 억제하고 방사하지 않도록(또는 이러한 주파수에 취약하지않도록) 설계되어 있습니다. 부상 또는 사망을 방지하기 위해 올바르게 지정된커넥터, 케이블, 부속품 및 테스트 장치만을 연결하고, 승인된 테스트 시설이나 달리통제된 조건인 경우 외에는 안테나에 연결하지 마십시오.

2.5 작업 환경경고이 제품은 실내용 또는 실외용으로 적합하며 반드시 건조한 곳에서 사용해야합니다. 제품의 외부 주 전원 공급장치는 실내에서만 사용할 수 있습니다.

경고부상 또는 사망을 방지하기 위해 습기나 물기가 있는 환경 또는 폭발성 가스나증기가 있는 곳에서 VNA나 부속품을 사용하지 마십시오.
주의손상을 방지하기 위해 VNA나 부속품을 항상 적절한 환경에서 사용하고보관하십시오.
온도
습도
고도
오염도

보관
작동
–20 °C ~ +50 °C+5 °C ~ +40 °C
최대 80% RH(비응축)2000 m2주의과열로 인해 기기가 손상되므로 기기 후면의 통기구를 막지 마십시오.

내부 간섭으로 인해 기기가 손상되므로 통기구에 이물질을 삽입하지 마십시오.

2.6 제품 관리본 제품과 부속품에는 사용자가 수리할 수 있는 부품이 없습니다. 수리, 서비스 및 교정 작업을위해서는 특수한 테스트 장비가 필요하며, Pico Technology 또는 공인 서비스 공급자만이러한 작업을 수행할 수 있습니다. Pico 3년 보증이 적용되는 경우 외에 이러한 서비스에는
수수료가 부과될 수 있습니다.
경고부상 또는 사망 사고를 방지하기 위해 손상된 것으로 보이는 VNA나 부속품은사용하지 말고, 비정상적인 작동이 우려되면 사용을 즉시 중단하십시오.

16

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서주의사용하기 전에 계측기와 모든 프로브, 커넥터, 케이블 및 액세서리를 주기적으로
점검하여 손상이나 오염의 징후를 확인하십시오.
장치 또는 연결된 장비의 손상을 방지하기 위해, 기기, 케이스 부품, 커넥터 또는부속품을 개조하거나 분해하지 마십시오.

제품 청소 시에는 부드러운 천을 연성 비누 또는 세제 용액에 적셔 사용하되 기기케이스나 부속품 내에 액체가 유입되지 않도록 주의하십시오.
모든 동축 선과 커넥터를 비롯한 모든 연결된 선에 기계적인 압박이 가해지지않도록 하고 선을 너무 많이 구부리지 않도록 주의하십시오. 잘못 취급하면 측면벽이 변형되어 성능이 저하됩니다. 특히, 테스트 포트 선의 굴곡 반경이 5 cm(2인치)보다 작게 구성해서는 안 됩니다.

측정 오류를 방지하고 테스트 선과 부속품 커넥터의 내구연한을 늘리기 위해,오염물질 액체 및 입자가 유입되지 않도록 주의하십시오. 제공된 먼지 캡을 항상장착하고 올바른 토크로 조이십시오. Pico는 황동 또는 금 도금 커넥터 체결 시
0.452 Nm(4.0 inch-lb)를, 그리고 제공하는 모든 스테인리스강 커넥터에 대해 1
Nm(8.85 inch-lb)를 권장합니다.

3. 시스템 요구사항PicoVNA 벡터 네트워크 분석기와 PicoVNA 소프트웨어가 올바르게 작동하려면 아래 표에나와 있는 운영 체제 중 하나를 실행하고 시스템 요구 사항을 충족하는 컴퓨터를 사용해야합니다.
프로세서, 메모리, 여유 디스크 공간
운영 체제

포트
디스플레이 해상도운영 체제 요구사항을 따름
Microsoft Windows 7, 8 또는 10. 32비트 또는 64
비트USB 2.0(USB 3.0 호환)
1280 x 720 이상

4. PicoVNA 키트의 내용물

PicoVNA 벡터 네트워크 분석기 키트 구성은 다음과 같습니다.
•
•
•
•
•
•
•

PicoVNA 106 [108: PicoVNA 108] 벡터 네트워크 분석기
PicoVNA 2 [108: PicoVNA 3] USB 플래시 드라이브DC 전원 공급 장치(해외 플러그 어댑터 4개 포함)
USB 2.0 케이블, 1.8 m
SMA/PC3.5/2.92 및 N-형식 고정 멀티 렌치빠른 시작 가이드보관 및 운반 케이스일부 제품 팩에는 추가 품목이 포함될 수 있습니다. 모든 사용 가능한 키트와 추가 부속품의전체 목록은 사용 설명서 또는 www.picotech.com에서 확인할 수 있습니다.
DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

17

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서

5. 외부 연결5.1 PicoVNA 106 및 108 벡터 네트워크 분석기

포트 1테스트 바이어스-T 스윕표시
I/O
입력
등

USB 연결

스윕표시등바이어스-T 테스트
입력
I/O

DC 입력

10 MHz In포트 2온/오프 스위치
Trig In

S/N
OFF
12–15 V DC

10 MHz Out

Trig OutON

10 MHz 레퍼런스 입/출력섀시 접지 트리거 입/출력5.2 SOLT-STD-M/F 및 SOLT-PREM-M/F 수동 교정 표준
로드

열기

통과

짧음

5.3 SOLT-AUTO-M/F 자동 (E-Cal) 교정 표준
표시기

포트 A

18

USB 포트포트 B

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서

6. PicoVNA 벡터 네트워크 분석기 설치
6.1 소프트웨어 설치
1.
2.
3.

VNA와 함께 제공된 USB 플래시 드라이브나 .
			
www.picotech.com/downloads에서 PicoVNA 2 [108:
PicoVNA 3] 소프트웨어 설치 프로그램을 가져옵니다.
설치 프로그램을 실행(마우스 오른쪽 버튼을 클릭하고 관리자로 실행)하고 설치가성공적으로 수행되었는지 확인합니다.

PicoVNA 장치를 컴퓨터와 연결하고 Windows가 드라이버를 자동 설치할 때까지기다립니다.

설치 프로그램은 다음 위치에 지원 디렉터리를 만듭니다.

C:\Users\<user name>\Documents\Pico Technology\PICOVNA2[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3]이 디렉터리에는 다음과 같은 파일이 포함됩니다(PicoVNA 106 예 표시).
••••xxxx-log.txt
CalKits\DefUnc.dat
DefCal.cal

FactoryDefaults.cal상태 로그 파일. ‘xxxx’는 일련 번호기본 측정 불확도 값기본 교정 데이터(마지막으로 사용한 교정) .
- VNA 장치에 처음 연결한 후 나타남공장에서 제공하는 백업 교정 데이터

Windows 7 시스템에서는 대개 다음 오류 메시지가 표시됩니다.

Ignore (무시)를
클릭하고 계속해도
안전합니다.

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

19

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서

6.2 The PicoVNA 2 [108: PicoVNA 3] 기본 창
디스플레이
채널 1

기기 일련
번호

교정 상태

USB 포트
상태

디스플레이
채널 3

스윕 상태

테스트 신호
출력

디스플레이
채널 2

측정점 수

사용 중인
주파수

측정 대역폭

평균
산출스윕
횟수

디스플레이
채널 4

선택한 활성채널의 마커 값트리거컨트롤에서저장(PicoVNA 108
전용)

마커 값

내부 온도(PicoVNA
108 전용)위 창에서 직사각형 플롯의 세로 크기를 다음과 같이 조절할 수 있습니다.
마우스를 끌어서
조절합니다.

20마우스를 끌어서 조절하거나 마우스 오른쪽 버튼을클릭하고 입력합니다.

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서

6.3 교정 키트 로드
PicoVNA 106 또는108

포트 1

PicoVNA 106 또는108

포트 2

삽입형
장치

수
커넥터

포트 1

암
커넥터

수
커넥터

비 삽입형 장치(테스트 포트가 직접 연결됨)수커넥터수 케이블.
암 교정 키트 사용.

PicoVNA 106 또는108

포트 1

포트 2

비 삽입형 장치(테스트 포트가 직접 연결되지 않음)

암
커넥터

암
커넥터

DUT

DUT

암 케이블.
두 포트에 모두같은 수 키트 사용.

수 케이블.
두 포트에 모두같은 암 키트 사용.

DUT암 케이블.
수 교정 키트 사용.

포트 2고정된 수동 교정 표준을 사용하여 교정: 고정된 수동 교정 표준을 사용하여 교정: 고정된 수동 교정 표준을 사용하여 교정:
SOLT-STD-F, SOLT-STD-M.
또는.
SOLT-PREM-F, SOLT-PREM-M

SOLT-STD-M.
또는.
SOLT-PREM-M

SOLT-STD-F.
또는.
SOLT-PREM-F

SOLT-AUTO-F.
또는.
SOLT-AUTO-M

SOLT-AUTO-M

SOLT-AUTO-F

또는 PicoVNA 108 및 자동 E-Cal표준으로 교정:또는 PicoVNA 108 및 자동 E-Cal표준으로 교정:또는 PicoVNA 108 및 자동 E-Cal표준으로 교정:수동 SOLT 교정은 다음과 같습니다.
••••PicoVNA 2 [108: PicoVNA 3] 소프트웨어 실행기본 메뉴에서 Tools > Calibration kit(도구 > 교정 키트)를 선택합니다.
Load P1 kit(P1 키트 로드)를 클릭하고 교정 키트와 함께 제공된 USB 플래시드라이브에서 포트 1 교정 키트의 데이터 파일을 찾은 후 Apply(적용)를 클릭합니다.
포트 2에 다른 키트를 사용하려면 Load P2 kit(P2 키트 로드)를 클릭하고 포트 2 키트의데이터를 선택한 후 Apply(적용)를 클릭합니다.

테스트 대상 장치에 따라 필요한 교정 키트를 선택합니다. 나중에 빠르게 액세스하기 위해 교정키트 데이터를 .
.
C:\Users\<user name>\Documents\Pico Technology\PICOVNA2\CalKits\[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3\CalKits\]폴더에 복사해도 됩니다. 이 작업은 Cal Kit Editor(교정 키트 편집기)에서 Save Kit(키트저장) 버튼을 사용하여 수행할 수 있습니다.
PicoVNA 108과 자동 교정은 다음과 같습니다.
•
•
•
•
•
•
•
•

PicoVNA 3 소프트웨어를 실행합니다.
E-Cal 모듈을 제어하는 PC의 예비 USB 포트에 연결합니다.
기본 메뉴에서 Tools > Calibration kit(도구 > 교정 키트)를 선택합니다.
Electronic Cal kit(전자 교정 키트)를 클릭합니다.
테스트 포트에 맞게 E-Cal 모듈 및 포트 어댑터 배열을 선택합니다.
선택한 구성과 정확히 일치하도록 E-Cal 모듈, 포트 어댑터 및 테스트 포트를 연결합니다.
Find(찾기)를 클릭하고 EasyCal 데이터를 로드합니다.
완료되면 Exit(끝내기)를 선택합니다.

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

21

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서•

Calibration Kit Parameters(교정 키트 매개변수) 메뉴의Apply(적용)을 클릭합니다.

6.4 해상도 설정교정을 진행하기 전에 기본 창에서 Enhance(강화)를 클릭하여 교정에서 지원할 최소 분해능대역폭을 설정합니다. 값을 낮추면 교정 속도가 느려집니다.
측정

최대 속도
최대 정확도와 ~100 dB
다이나믹 레인지일반 용도, 빠른 속도, ~90
dB 다이나믹 레인지최대 다이나믹 레인지교정대역폭
10 kHz

교정
평균 산출
없음

교정
출력
+0 dBm1 kHz

없음+0 dBm100 Hz
10 Hz

없음
없음

–3 dBm

+6 dBm

6.5 교정 파라미터 설정주석측정 중 대역폭을 140 kHz로 설정측정 중 대역폭 설정을 100 Hz로 유지측정 중 대역폭 설정을 1 kHz로유지측정 중 대역폭 설정을 10 Hz로유지Calibration(교정)을 클릭하여 Calibration(교정) 창을 엽니다.

로드된 교정
키트

1. 스윕
파라미터를
설정합니다.

3. 필요한
측정을선택합니다.

2. 필요한 스윕대역폭 설정이끝나면 값을
적용합니다.
5. 중요!
작업을 마쳤으면 교정을 적용합니다.

22

4. 필요한 교정 단계
수행. 어떤 순서를따르든 상관 없지만,나와 있는 순서를따르면 대개 연결단계의 수가
최소화됩니다.

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

DO296-6

6.6 디스플레이 파라미터 설정

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서기본 창에서 Display(디스플레이)를 클릭하여 Display Set Up(디스플레이 설정) 창을엽니다.
3. 설정할 채널을선택합니다.
1.
1-채널,2-채널 또는
4-채널보기를선택합니다.

4. 그래프에 세로
눈금 값을
표시하려면
선택합니다.

2. 마커 패널에표시할 채널을
선택합니다.
5. 다음 채널을 선택하기 전에설정을 적용합니다.

작업을 마쳤으면 기본 창에서 Start(시작)를 클릭하여 측정을 시작합니다.

6.7 데모 모드로 실행데모 모드에서는 기기를 작동하지 않고도 PicoVNA 2 [108: PicoVNA 3] 소프트웨어를 살펴볼수 있습니다.
••데모 모드에 들어가려면 기기를 연결하지 않은 상태에서 PicoVNA 2 [108: PicoVNA 3]소프트웨어를 실행합니다.
나타난 대화 상자에서 Ignore(무시)를 클릭합니다.

그러면 PicoVNA 2 [108: PicoVNA 3]에 데모 측정 선택 항목이 표시됩니다.

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

23

PicoVNA 벡터 네트워크 분석기 빠른 시작 설명서

7. 추가 정보7.1 무료 뉴스레터Pico Technology에서는 무료 월간 뉴스레터를 이메일로 발송해드립니다. 이 뉴스레터에는제품 출시, 소프트웨어 업그레이드, 애플리케이션 정보, 팁 및 권장 정보가 담겨 있습니다.
고객의 이메일 주소 및 기타 정보는 기밀로 유지되며 타사에 제공되지 않습니다. 구독하려면다음 웹 사이트를 방문하십시오.
www.picotech.com/signup

7.2 소프트웨어 업데이트 및 교체모든 Pico 소프트웨어 및 설명서의 최신 버전은 다음 웹 사이트에서 무료로 다운로드할 수있습니다.
www.picotech.com/downloads

새로운 소프트웨어 디스크가 필요한 경우에는 Pico Technology나 현지 대리점에문의하십시오. 이 서비스에는 소정의 수수료가 부과될 수 있습니다.

7.3 자체 소프트웨어 작성

MATLAB 및 MATLAB RF 툴박스, LabVIEW, C, C# 및 Python 코드 예제는 GitHub(github.
com/picotech)에서 제공됩니다. 예제에는 다수의 기기 처리 및 제어가 포함됩니다.

7.4 사용 설명서와 프로그래머 가이드

PicoVNA 사용 설명서와 PicoVNA 프로그래머 가이드는 제공되는 USB 플래시 드라이브에들어 있으며 다음 위치에서 다운로드할 수 있습니다.
www.picotech.com/downloads

7.5 사양사용 설명서와 데이터 시트에는 PicoVNA 벡터 네트워크 분석기의 최신 사양이 포함되어있습니다.

7.6 기술 지원다음 Pico Technology 지원 포럼에서는 정기적으로 업데이트되는 기술 지원 정보를 확인할수 있습니다.

7.7 보증 및 반품

www.picotech.com/tech-support

본 PicoVNA 벡터 네트워크 분석기에는 3년 반품 보증이 제공됩니다. 전체 보증약관은 PicoVNA 사용 설명서를 참조하십시오.

24

Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.

YEAR W A R R A N T Y

DO296-6

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

1. はじめに日本語

PicoVNAベクトルネットワークアナライザーをお買い上げいただき誠にありがとうございます。
以下のモデルをご利用いただくことができます：••

PicoVNA 106：6 GHzの2ポートVNA、内蔵バイアスTネットワーク

PicoVNA 108: 8.5 GHzの2ポートVNA（内蔵バイアスTネットワーク、
ミキサー測定、
トリ
ガー保存機能付き）

基本的な設置後の詳細な設定情報については、PicoVNA 2[108: <情報PicoVNA 3]*ソフトウ
ェアに同梱されているユーザーガイドを参照してください。
詳細な仕様については、PicoVNAデータシートを参照してください。

*特に指示がない限り、本ガイドの情報はPicoVNAシリーズの全モデルに適用されます。
PicoVNA 108にのみ適用される情報には、[108: <情報>]と記載されます。

2. 安全に関する情報

感電、火災、負傷、また製品の損傷などを避けるため、本製品を設置、使用する前に本安全に関する情報をしっかりとお読みください。
また、電気を使う作業や電気の近くで行う作業に関して、一般的に認められている安全な作業方法や手順すべてに従ってください。
本製品は欧州規格EN 61010-1（計測、制御および試験用電気機器の安全要件）に従って設計されています。
本書には、以下の安全に関する説明が記載されています。

警告：けがや死亡につながる恐れのある状況や慣行を示します。

注意：本製品や本製品に接続する機器の破損につながる恐れのある状況や慣行を示します。

警告
本製品は、訓練され、認証された技術者のみによる専門的な使用を目的としています。
けがや死亡を避けるため、説明書の指示通りに本製品を使用してください。
また、同梱されたアクセサリ、または Pico Technology の推奨アクセサリのみを使用してください。
メーカーによって指定されていない方法で使用した場合、本製品による保護が十分に機能しない場合があります。

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

25

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

2.1 記号

本製品、および本書では、以下の安全および電気に関する記号が使用されています。
記号

説明

この端子は、計測機器とアースを接続するために使用できます。
安全な保護アースではありません。

アース（接地）端子
筐体端子
感電の危険
注意
分別せずに都市廃棄物として本製品を廃棄しないでください。

本製品にこの記号が付いている場合は、本書を参照する必要があります。

2.2 最大入力・出力範囲警告

感電を防ぐため、以下で指定された最大値を超える信号レベルを測定または適用しないでください。

下表は、VNAおよびキャリブレーションアクセサリの出力の最大電圧および入力の過電圧保護範囲を示しています。
過電圧保護範囲は、機器やアクセサリを損傷することなく適用できる最大電圧です。
最大動作電圧（出力または入力）

機器のコネクタ
ポート1および2

バイアスT 1および2
トリガーおよび基準入力
トリガーおよび基準出力

キャリブレーションアクセサリのコ
ネクタ
SOLT-PREM および SOLT-STD ポート
SOLT-AUTO ポート A/B
警告+10 dBm（約710 mV RMS）
±15 V DC

0 V ～ +5 V

過電圧または過電流保護+20 dBm .
（約2.2 V RMS）
250 mA±6 V pk
電圧を適用しないでください
過電圧保護+20 dBm+20 dBm

以下の表の電圧最大値を超える信号は、EN 61010により「危険電位」
として指定されています。

26±60 V DCEN 61010-1: 2010+A1:2019の信号電圧限度
30 V AC RMS
最大± 42.4 V pk

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

DO296-6

警告

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

感電を防ぐため、危険電位電圧が存在する可能性がある場所で本機器の作業を行う場合、必要な安全対策をすべて取ってください。
警告

機器の損傷とけがの可能性を避けるため、本機器またはアクセサリは定格供給
電圧または環境範囲外で操作しないでください。

注意
いずれかのコネクタで過電圧保護範囲を超えると、本機器や接続されている他の機器が永久に破損する原因となる場合があります。

恒久的な損傷を防ぐため、VNAのトリガーまたは基準出力に入力電圧を適用しないでください。

2.3 接地警告

USBケーブル経由での本機器またはSOLT-AUTO E-Calモジュールのアース接続
は、機能のみを目的としています。
本機器およびアクセサリには、安全用または保護用アースはついていません。
けがや死亡、
または機器の恒久的な損傷を避けるため、
アース入力または出力（筐体）は、種類を問わず絶対に電源に接続しないでください。
けがや死亡を避けるため、電圧計を使って本機器のアースとアースを接続しようとしている部分との間に重大なACまたはDC電圧が存在していないことを確認してください。
注意

接地の不備による信号劣化を防ぐため、本機器に同梱されている高品質のUSB
ケーブルを常に使用してください。

2.4 外部接続

警告
負傷または死亡を避けるため、本機器本機器に同梱されている電源アダプターのみを使用してください。
これはお住まいの国の電圧・プラグ設定向けに承認されています。

PicoVNA モデル
PicoVNA 106

PicoVNA 108

SOLT-AUTO E-Cal モジュールDO296-6USB接続

電圧

USB 2.0 .
12 ～ 15 V DC（.
USB 3.0とも互換性
あり）

外部電源
電流

1.85 A pk
該当なし

総電力
22 W
25 W

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

27

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

警告
無線周波数の抑制
本機器には、
スイープまたはCW無線周波数信号源（300 kHz～6.02 GHz、最大+6 dBm[108: 300 kHz～8.50 GHz、最大6 dBm]）が含まれています。
本機器と同梱のアクセサリは、他の機器の操作や無線制御・通信を妨害する可能性のある無線周波数を放出する（またはこれに敏感な）のではなく、無線周波数を含むよう設計されています。
けがや死亡を避けるため、適切に指定されたコネクタ、
ケー
ブル、
アクセサリ、
テスト機器にのみ接続し、承認された試験施設内やその他の制
御された状態以外ではアンテナに接続しないでください。

2.5 環境警告
本製品は屋内・屋外で使用することができますが、乾燥した場所でのみ使用する必要があります。
本製品の外部主電源は、屋内でのみ使用することができます。

警告
けがや死亡を避けるため、VNAまたはアクセサリは濡れている場所や湿っている
場所、
または爆発性ガスや蒸気の付近では使用しないでください。
注意
損傷を防ぐため、必ず適切な環境でVNAまたはアクセサリを使用・保管してくだ
さい。
温度
湿度
高度
汚染度

保管
動作
–20 °C ～ +50 °C+5 °C ～ +40 °C
最大80 %（相対湿度、結露なきこと）
2000 m2

注意
本機器の背面にある通気口をブロックしないでください。
加熱によって損傷する原因になります。

通気口には何も挿入しないでください。
内部干渉によって機器に損傷が生じる可能性があります。

2.6 本製品のお手入れ

本製品とアクセサリには、ユーザーが修理・点検できる部品はありません。
修理、点検、
キャリ
ブレーションには特別な試験装置が必要であり、実行できるのはPico Technology、
または承認されたサービスプロバイダーのみです。
Picoの3年間保証の対象でない場合、これらのサービスには料金がかかる場合があります。
警告
けがや死亡を避けるため、多少なりとも破損しているように見える場合、本製品
またはアクセサリを使用しないでください。
また、異常な動作に不安がある場合は、直ちに使用を停止してください。

28

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

DO296-6

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

注意
本機器およびすべてのプローブ、
コネクタ、
ケーブル、
アクセサリなどは、使用する前に破損したり汚れたりしていないか必ず点検してください。

本機器または接続されている機器の損傷を避けるため、本機器、ケース部品、コ
ネクタ、
またはアクセサリを改ざんあるいは分解しないでください。

本機器のお手入れを行う際は、柔らかい布と液体中性洗剤もしくは水に溶かした洗剤を使用してください。
液体が機器やアクセサリの筐体に入らないようにしてください。

あらゆる同軸リードやコネクタを含むすべての接続リードで機械的負荷やきつい
曲げ半径を避けられるよう配慮してください。
取り扱いを誤ると、側壁がゆがむ
原因になり、性能や精度が劣化します。特に、
テストポートリードは曲げ半径が5cm以下にならないよう注意してください。

測定エラーを防ぎ、
テストリードとアクセサリコネクタの使用期間を延ばすため、液体や粒子状の汚染物質が入り込まないようにしてください。
必ず同梱のダスト
キャップをはめ、適正なトルクで締めてください。
Picoでは、同梱のあらゆるステ
ンレススチール製コネクタで1 Nm（8.85 inch-lb）を使用するか、真ちゅう製または金メッキのコネクタをインターフェイスで接続する際は0.452 Nm（4.0 inchlb）を使用するようお勧めしています。

3. システム要件

PicoVNAベクトルネットワークアナライザおよびPicoVNAソフトウェアが正常に動作するようにするため、以下の表に記載されているシステム要件を満たすコンピューターを使用してください。
プロセッサ、
メモリ、ディスク空き容量
オペレーティングシステム
ポート
ディスプレイ解像度

OSの要件に応じてMicrosoft Windows 7/8/10（32ビット＆64ビット）
USB 2.0（USB 3.0とも互換性あり）
1280 x 720以上

4. PicoVNAキットの同梱物

PicoVNAベクトルネットワークアナライザーキットには以下のアイテムが含まれています。

•
•
•
•
•
•
•

PicoVNA 106 [108: PicoVNA 108]ベクトルネットワークアナライザー
PicoVNA 2 [108: PicoVNA 3] USBフラッシュドライブ
4つの国際的なプラグアダプタを含むDC電源
USB 2.0ケーブル（1.8m）
SMA/PC3.5/2.92、およびNタイプ固定マルチレンチ
クイックスタートガイド
保管・持ち運び用ケース

一部の製品には追加アイテムが含まれています。
すべての利用可能なキットと追加アクセサ
リの詳細なリストについては、
ユーザーガイドを参照するか、www.picotech.comにアクセスしてください。
DO296-6

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

29

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

5. 外部接続5.1 PicoVNA 106/108ベクトルネットワークアナライザー

ポート1

テスト
I/O

バイアスT
入力

スイープ

インジケータUSB接続
10 MHz In

スイープ

インジケータ

バイアスT
入力

テスト
I/O

ポート2

オン/オフスイッチ

DC入力

Trig In

S/N
OFF

10 MHz Out

12–15 V DC

Trig OutON

10 MHz参照入力/出力

筐体接地 トリガ入力/出力

5.2 SOLT-STD-M/FおよびSOLT-PREM-M/F手動キャリブレーション基準
ロード

開く

経由

ショート

5.3 SOLT-AUTO-M/F自動（E-Cal）キャリブレーション基準
インジケータ

ポートA

30

USBポート

ポートB

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

DO296-6

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

6. PicoVNAベクトルネットワークアナライザーの設置

6.1 ソフトウェアのインストール
1.

2.
3.

PicoVNA 2 [108: PicoVNA 3] ソフトウェアインストーラーは、VNAに同梱されている
USBフラッシュドライブ、
または .
			
www.picotech.com/downloads から入手します。

インストーラーを実行し（[Run as administrator（管理者として実行）]を右クリックします）、確実にインストールします。
PicoVNAユニットをコンピューターに接続し、Windowsが自動的にドライバをインストールするまで待ちます。

インストーラーが以下の場所にサポートディレクトリを作成します。

C:\Users\<user name>\Documents\Pico Technology\PICOVNA2[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3]

このディレクトリには以下のファイルが含まれています（PicoVNA 106の例を示しています）
：
•

•
•
•

xxxx-log.txt
番号です
CalKits\DefUnc.dat
DefCal.cal
キャリブレーション）.

FactoryDefaults.cal

これはステータスログファイルです。
「xxxx」はシリアル

デフォルトの測定不確実性の値
デフォルトのキャリブレーションデータ
（最後に使用した- VNA装置に最初に接続した後に表示されます
工場からのバックアップキャリブレーションデータ

Windows 7を搭載したコンピューターでは、以下のエラーメッセージが表示される場合があります。

[Ignore（無視）]をクリックして続行できますDO296-6

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

31

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

6.2 PicoVNA 2 [108: PicoVNA 3] メインウィンドウ
チャンネル
1を表示

機器のシリ
アル番号

キャリブレ
ーションの
ステータス

USBポートのステータス

チャンネル3
を表示

スイープの
ステータス

テスト信号
電源

使用中の
周波数

測定ポイント数選択されたアクティ
ブなチャンネル向けのマーカー値

チャンネル
2を表示

チャンネル
4を表示

平均スイー
プ数

トリガコントロールに マーカ
保存（PicoVNA 108 ー値
のみ）

測定帯域幅

内部温度（PicoVNA 108のみ）

上のウィンドウではどの長方形プロットの垂直スケールでも、以下のように調整できます。
ドラッグして調整

32

ドラッグして調整するか、
右クリックして入力してください

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

DO296-6

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

6.3 キャリブレーションキットの読み込み
PicoVNA 106 または 108
ポート1

オス
コネクター

ポート2

挿入可能な
デバイス

メス
コネクター

PicoVNA 106 または 108

PicoVNA 106 または 108

挿入不可能なデバイス（テストポートを直接接続することはできます）

挿入不可能なデバイス（テストポートを直接接続することはできません）

DUT

DUT

メスケーブル。
両方のポートで同じオスのキットを使用してください。

オスケーブル。
両方のポートで同じメスのキットを使用してください。

ポート1

オス
コネクター

DUT
メスケーブル。
オスのキャリブレーションキ
ットを使用してください。

オスケーブル。
メスのキャリブレーションキ
ットを使用してください。

固定手動キャリブレーション基準を使用したキャリブレーション：

SOLT-STD-FおよびSOLT-STD-M.
または.
SOLT-PREM-FおよびSOLT-PREM-M

またはPicoVNA 108および自動E-Cal基準を使用したキャリブレーション：
SOLT-AUTO-F.
または.
SOLT-AUTO-M

ポート2

オス
コネクター

固定手動キャリブレーション基準を使用したキャリブレーション：

SOLT-STD-M.
または.
SOLT-PREM-M

またはPicoVNA 108および自動E-Cal基準を使用したキャリブレーション：
SOLT-AUTO-M

ポート1

ポート2

メス
コネクター

メス
コネクター

固定手動キャリブレーション基準を使用したキャリブレーション：

SOLT-STD-F.
または.
SOLT-PREM-F

またはPicoVNA 108および自動E-Cal基準を使用したキャリブレーション：
SOLT-AUTO-F

手動SOLTキャリブレーション：

•
•
•
•

PicoVNA 2 [108: PicoVNA 3] ソフトウェアを実行します
メインメニューで[Tools（ツール）] > [Calibration kit（キャリブレーションキット）]を選択します[Load P1 kit（P1キットを読み込む）]をクリックし、ポート1キャリブレーションキット向けのデータファイル（キャリブレーションキットに同梱されているUSBフラッシュドライブ
にあります）を特定してから[Apply（適用）]をクリックします
異なるキャリブレーションキットをポート2で使用する場合は、[Load P2 kit（P2キットを読み込む）]をクリックし、ポート2キット向けのデータを選択してから[Apply（適用）]をクリックしてください

テストする機器に応じて、必要なキャリブレーションキットを選択します。
今後、より素早くアクセスするには、キャリブレーションキットデータを.
.
C:\Users\<user name>\Documents\Pico Technology\PICOVNA2\CalKits\[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3\CalKits\]

フォルダにコピーしておくようお勧めします。
これは、[Cal Kit Editor（キャリブレーションキッ
トエディタ）]で[Save Kit（キットの保存）]ボタンを使用すると実行できます。
PicoVNA 108および自動キャリブレーション：

•
•
•

•

PicoVNA 3ソフトウェアを実行します。
E-CalモジュールをPCの空いているUSBポートに接続します。
メインメニューで[Tools（ツール）] > [Calibration kit（キャリブレーションキット）]を選択します。
[Electronic Cal kit（E-Calキット）]をクリックします。

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

33

PicoVNAベクトルネットワークアナライザー クイックスタートガイド•

テストポートにあったE-Calモジュールおよびポートアダプタの組み合わせを選択します。
選択した構成に合うように、E-Calモジュール、ポートアダプタ、
テストポートを接続します。
[Find（検索）]をクリックして、EasyCalデータを読み込みます。
完了したら、[Exit（終了）]を選択します。
[Calibration Kit Parameters（キャリブレーションキットパラメータ）]メニューで[Apply（適用）]をクリックします。

•

•
•
•

6.4 解像度の設定

キャリブレーションを進める前に、
メインウィンドウで[Enhance（拡張）]をクリックし、
キャリ
ブレーションでサポートする必要のある最低解像度帯域幅を設定します。
値が低ければ、キャリブレーションが鈍化します。
測定
最高速度

最良の精度と最高100 dBのダイナミックレンジ
汎用、高速、最高90 dBのダイナミックレンジ
最適なダイナミックレンジ

キャリブレ
ーション
帯域幅

キャリブレ
ーション
平均

キャリブレ
ーション
電力

100 Hzなし–3 dBm

10 kHz
1 kHz

10 Hz

なし
なし

なし

+0 dBm
+0 dBm
+6 dBm

6.5 キャリブレーションパラメーターの設定

備考

測定中、帯域幅を140 kHzに設定します
測定中、帯域幅は100 Hzから変更しないでください
測定中、帯域幅は1 kHzから変更しないでください
測定中、帯域幅は10 Hzから変更しないでください[Calibration（キャリブレーション）]をクリックして[Calibration（キャリブレーション）]ウィ
ンドウを開きます：

読み込まれたキャリブレーションキット

1. スイープパ
ラメーターを設定します

3. 必要な測定を選択します2. 必要なスイープ
帯域幅の設定後、値を適用します34

5. 重要！
完了後、キャリブレーションを適用します

4. 必要なキャリブレーションステップを実施します。
どのようなシーケンスでも構いませんが、通常は下向きの後に横向きの順序にすると、接続ステップの数を最低限に抑えられます。

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

DO296-6

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

6.6 表示パラメーターの設定

メインウィンドウで[Display（ディスプレイ）]をクリックし、[Display Set Up（ディスプレイの設定）]ウィンドウを開きます：
3.チャンネルを選択して設定します

4.チェックして、グ
ラフ上に垂直ス
ケール値を表示します

1 .1チャンネ
ル、2チャンネル、または4チャンネルの表示を選択します2.マーカーパネ
ルで表示するチ
ャンネルを選択します
5.次のチャンネルを選択する前に設定を適用してください

終了したら、
メインウィンドウで[Start（スタート）]をクリックして測定を開始します。

6.7 デモモードでの実行

デモモードでは、本機器を実効しなくてもPicoVNA 2 [108: PicoVNA 3] ソフトウェアをお試しいただけます。
••

デモモードに入るには、本機器を接続せず、PicoVNA 2 [108: PicoVNA 3] ソフトウェア
を実行してください。
表示されたダイアログで[Ignore（無視）]をクリックします。

その後、PicoVNA 2 [108: PicoVNA 3] で特定のデモ測定をご覧いただけます。

DO296-6

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

35

PicoVNAベクトルネットワークアナライザー クイックスタートガイド

7. 詳細な情報

7.1 無料のニュースレター

Pico Technologyは、メールで無料のニュースレターを毎月送信させていただいております。
ニュースレターでは、新製品の発表、
ソフトウェアの更新、用途に関する情報はもちろん、お役立ち情報やアドバイスも提供しています。
お客様の電子メールアドレスとその他の詳細は機密として扱われ、第三者には提供されません。
購読するには以下のサイトよりニュースレターの配信登録を行うことができます。
www.picotech.com/signup

7.2 ソフトウェアの更新と交換

Picoソフトウェアの最新版とマニュアルは、
すべて無料で以下から入手できます：
www.picotech.com/downloads

新しいソフトウェアディスクが必要な場合は、Pico Technologyまたは最寄りの代理店まで
お問い合わせください。
このサービスを利用する場合、少額の料金が発生する場合があります。

7.3 オリジナルソフトウェアの作成

MATLABおよびMATLAB RFツールボックス、LabVIEW、C、C#、Pythonのコード例
は、GitHub（github.com/picotech）からご利用いただけます。
コード例には、装置の様々なアドレッシングや制御が含まれています。

7.4 ユーザーガイドおよびプログラマー向けガイド

PicoVNAユーザーガイドおよびPicoVNAプログラマーガイドは、同梱されているUSBフラッシュドライブまたは以下から入手できます：

7.5 仕様

www.picotech.com/downloads

ユーザーガイドおよびデータシートには、PicoVNAベクトルネットワークアナライザーの最新の仕様が含まれています。

7.6 テクニカルサポート

Pico Technologyテクニカルサポートウェブサイトおよびユーザーフォーラムでは、
テクニカ
ルサポートに関する情報が定期的に更新されています。
以下のサイトにアクセスしてご確認ください。

7.7 保証および返品

www.picotech.com/tech-support

PicoVNAベクトルネットワークアナライザーには、メーカー返品3年間保証がつ Y E A R W A R R A N T Y
いています。
詳細な条件については、PicoVNAユーザーガイドを参照してください。
36

Copyright © 2017–2022 Pico Technology Ltd. 無断複写・複製・転載禁止。

DO296-6Краткое руководство пользователя векторного анализатора цепей PicoVNA

1. Введение

Русский

Благодарим вас за приобретение векторного анализатора цепей PicoVNA.
Доступны следующие модели:•PicoVNA 106: двухпортовый ВАЦ на 6 ГГц со встроенными цепями стройниками смещения

•PicoVNA 108: двухпортовый ВАЦ на 8,5 ГГц со встроенными цепями стройниками смещения, измерения смесителей и сохранение по триггеру

Дополнительную информацию о настройках после базовой установки см.
в Руководстве пользователя, поставляемом с программным обеспечением
PicoVNA 2 [108: PicoVNA 3]*.
Подробные характеристики см. в паспорте PicoVNA.
* Если не указано иное, информация, приведенная в данном руководстве,относится ко всем моделям серии PicoVNA. Информация, которая относится
исключительно к модели PicoVNA 108, приводится в следующем формате: «[108:<информация>]».

2. Информация по технике безопасностиВо избежание возможного удара электрическим током, пожара, травмы илиповреждения изделия внимательно прочтите эту информацию по технике
безопасности, прежде чем приступать к установке или использованию продукта.
Кроме того, следуйте всем общепринятым мерам и процедурам техники
безопасности при работе с электричеством и вблизи него.
Данный прибор с его аксессуарами был разработан специально длясоответствия требованиям стандарта EN 61010-1 (Требования к безопасностидля электрического оборудования для измерений, управления и лабораторного
использования).
В данном руководстве правила техники безопасности сопровождаются
следующими сигнальными словами:ВНИМАНИЕ — информация об условиях или действиях, которые могут привестик травмам или смерти.
ОСТОРОЖНО — информация об условиях или действиях, которые могут
привести к повреждению изделия или оборудования, к которому оноподключено.
ВНИМАНИЕ
Данное изделие предназначено исключительно дляпрофессионального использования специалистами, которые
обладают соответствующими знаниями и квалификацией. Воизбежание травм или смерти используйте продукт исключительнопо инструкциям и только с аксессуарами из комплекта поставкиили теми, которые были рекомендованы Pico Technology. Защита,обеспечиваемая продуктом, может быть нарушена, если оборудование
используется способом, не указанным производителем.
DO296-6© © 2017–2022 Pico Technology Ltd. Все права защищены.

37Краткое руководство пользователя векторного анализатора цепей PicoVNA

2.1 Символы
Следующие символы безопасности и электрооборудования могут находиться напродукте или в данном руководстве.
Символы

Описание
Клемма заземления (земля)Данная клемма может
использоваться для измерения
заземляющего соединения. Данная
клемма НЕ является защитным илибезопасным заземлением.

Клемма шасси
Возможность поражения
электрическим током
Осторожно
Данное изделие нельзя
утилизировать вместе сбытовыми отходами

Данный символ на продукте говорито том, что необходимо прочитать
руководство по эксплуатации имерам предосторожности.

2.2 Максимальные входные/выходные диапазоны
ВНИМАНИЕ

Чтобы избежать поражения электрическим током, не пытайтесь
измерять или использовать уровни сигнала, которые выходят запределы указанных ниже максимальных значений.
В приведенной ниже таблице указано максимальное напряжение на выходеи диапазон защиты от перенапряжения на входе ВАЦ и его калибровочных
аксессуаров. Диапазоны защиты от перенапряжения — это максимальное
напряжение, которое можно использовать без повреждения прибора илиаксессуара.
Соединители прибора
Порты 1 и 2Тройники смещения 1 и 2Триггер и опорный сигнал (вход)Триггер и опорный сигнал (выход)Соединения калибровочных
аксессуаров
Порты SOLT-PREM и SOLT-STD
Порты A/B SOLT-AUTO
38

Максимальное рабочее
напряжение (на входеили выходе)+10 дБм
(прибл. 710 мВ RMS)
±15 В пост. токаОт 0 В до +5 ВЗащита отперенапряжения илисверхтока
+20 дБм
(прибл. 2,2 В RMS)
250 мА
±6 В (пиковое значение)Запрещено подавать
напряжение
Защита отперенапряжения
+20 дБм
+20 дБм

© © 2017–2022 Pico Technology Ltd. Все права защищены.

DO296-6Краткое руководство пользователя векторного анализатора цепей PicoVNAВНИМАНИЕ

Сигналы, превышающие пределы напряжения в приведенной нижетаблице, обозначаются как «опасное токонесущее напряжение»согласно стандарту EN 61010.
Пределы напряжения сигнала согласно EN 61010-1:2010+A1:201930 В перем. тока±42,4 В (макс. пиковое
±60 В пост. тока(среднеквадр.)значение)ВНИМАНИЕВо избежание поражения электрическим током примите всенеобходимые меры предосторожности при работах на оборудовании сопасным для жизни напряжением.
ВНИМАНИЕВо избежание повреждения оборудования и возможных травм неиспользуйте прибор или аксессуары за пределом их номинального
напряжения питания или диапазона изменения окружающих условий.
ОСТОРОЖНО
Превышение диапазона защиты от перенапряжения на любом
соединителе может привести к необратимому повреждению прибора идругого подключенного к нему оборудования.
Во избежание неисправимого повреждения не подключайте входное
напряжение к выходному сигналу триггера или опорного напряжения
ВАЦ.

2.3 Заземление
ВНИМАНИЕ

Заземляющее соединение через кабель USB предназначено
исключительно для работы прибора или модуля SOLT-AUTO E-Cal.
Прибор и аксессуары не оборудованы защитными заземляющими
соединениями.
Во избежание травм, смерти или необратимого повреждения прибора
никогда не подключайте к входному или выходному заземлению
(шасси) источник электроэнергии. Во избежание травм или смерти
используйте вольтметр для проверки на наличие напряжения
постоянного или переменного тока между заземлением прибора иточкой, к которой его необходимо подключить.
ОСТОРОЖНОВо избежание снижения уровня сигнала по причине плохого
заземления всегда используйте высококачественный кабель USB изкомплекта поставки прибора.

DO296-6© © 2017–2022 Pico Technology Ltd. Все права защищены.

39

Краткое руководство пользователя векторного анализатора цепей PicoVNA

2.4 Внешние соединители
ВНИМАНИЕВо избежание травм или смерти используйте только адаптер питанияиз комплекта поставки прибора. Он одобрен для напряжения испособа подключения в вашей стране.
Модель PicoVNA
PicoVNA 106
PicoVNA 108
Модуль SOLT-AUTO E-Cal

USB-подключение
USB 2.0Совместимо с
USB 3.0Внешний источник питания
ОбщаяНапряжение
Токмощность
1,85 А
22 Вт
12-15 В пост.
(пиковоетока
25 Втзначение)Н/П

ВНИМАНИЕ
Изоляция радиочастотВ данном приборе используется радиосигнал качающейся илинемодулированной частоты (от 300 кГц до 6,02 ГГц при макс.
+6 дБм [108: от 300 кГц до 8,50 ГГц при макс. +6 дБм]). Прибор ипоставляемые с ним аксессуары предназначены для блокировки,а не излучения радиоволн (или восприимчивости к ним), которые
являются потенциальной помехой при работе другого оборудованияили передаче сигналов радиоуправления или радиосвязи. Воизбежание травм или смерти подключайте к прибору только
одобренные соединители, кабели, аксессуары и тестовые устройства,а также избегайте подключения к антеннам, которые не находятсяв одобренной среде для тестирования или других контролируемых
условиях.

2.5 Окружающая среда
ВНИМАНИЕ
Данный продукт предназначен для использования в помещении ина улице, только в сухих местах. Сетевой источник питания продукта
предназначен только для использования в помещении.
ВНИМАНИЕВо избежание травм или смерти не используйте ВАЦ или аксессуарыв условиях повышенной влажности или сырости, а также вблизи
взрывоопасных газов или паров.
ОСТОРОЖНО
Всегда используйте и храните ВАЦ или аксессуары в соответствующих
условиях во избежание их повреждения.
Температура
Влажность
Высота над уровнем моряУровень загрязнения
40Хранение
Эксплуатация
От –20 °C до +50 °C
От +5 °C до +40 °C
До 80% относительной влажности (безконденсации)2000 м
2© © 2017–2022 Pico Technology Ltd. Все права защищены.

DO296-6Краткое руководство пользователя векторного анализатора цепей PicoVNAОСТОРОЖНОНе заслоняйте вентиляционные отверстия в задней части прибора,поскольку перегрев прибора приводит к его повреждению.
Не вставляйте какие-либо предметы в вентиляционные отверстия,поскольку вмешательство в работу внутренних компонентов прибора
приводит к его повреждению.

2.6 Уход за продуктом
Данное изделие и его аксессуары не содержат деталей, обслуживаемых
пользователями. Ремонт, обслуживание и калибровка прибора требуют
специализированного тестового оборудования и должны проводиться
исключительно со стороны Pico Technology или другого одобренного
поставщика услуг. За эти услуги может взиматься отдельная плата, если она непокрывается трехлетней гарантией Pico.
ВНИМАНИЕВо избежание травм или смерти не используйте ВАЦ или аксессуары,если они каким-либо образом повреждены, и прекратите
эксплуатацию в случае выявления каких-либо аномалий в их работе.
ОСТОРОЖНО
Перед использованием регулярно осматривайте прибор, все его щупы,соединители, кабели и принадлежности на предмет повреждений илизагрязнения.
Во избежание повреждения устройства или подключенного к немуоборудования не нарушайте целостность и не разбирайте прибор,детали корпуса, соединители или аксессуары.
При чистке продукта используйте мягкую ткань и слабый мыльный
раствор или разбавленное в воде моющее средство, и не допускайте
попадания жидкостей внутрь корпуса прибора или аксессуаров.
Соблюдайте осторожность, чтобы не допускать механических нагрузокили сильного изгиба любых подключенных кабелей и проводов,в том числе коаксиальных. Ненадлежащее обращение приведетк деформации боковых стенок и ухудшит производительность.
В частности, обратите внимание, что радиус изгиба кабелей
измерительного порта не должен быть меньше 5 см.
Во избежание ошибок при измерении, а также для продления срока
службы измерительных проводов и соединителей аксессуаров недопускайте попадания внутрь никаких жидкостей или загрязняющих
примесей. Всегда надевайте пылезащитные крышки из комплекта
поставки и используйте рекомендуемый момент затяжки.
Рекомендуемый Pico момент затяжки: 1 Нм для соединителей изкомплекта поставки и всех соединителей из нержавеющей стали или
0,452 Нм для соединений в контакте с латунными или позолоченными
соединителями.
DO296-6© © 2017–2022 Pico Technology Ltd. Все права защищены.

41

Краткое руководство пользователя векторного анализатора цепей PicoVNA

3. Системные требования
Чтобы обеспечить правильную работу векторного анализатора цепей PicoVNAи программного обеспечения PicoVNA, требуется компьютер, соответствующий
системным требованиям и работающий под управлением одной изоперационных систем, указанных в таблице ниже.
Процессор, память, свободное
пространство на диске
Операционная система
ПортыРазрешение экрана

Согласно требованиям операционной
системы
Microsoft Windows 7, 8 или 10 (32- или
64-разрядная)USB 2.0 (также совместимо с USB 3.0)
1280 x 720 (минимум)4. Комплект поставки PicoVNAВ комплект векторного анализатора цепей PicoVNA входят следующие
компоненты:
•
•
•
•
•
•
•

Векторный анализатор цепей PicoVNA 106 [108: PicoVNA 108]Флэш-накопитель USB PicoVNA 2 [108: PicoVNA 3]Источник питания постоянного тока с четырьмя штепсельными адаптерами
международного формата
Кабель USB 2.0 (1,8 м)
SMA/PC3.5/2.92 и многофункциональный гаечный ключ N-типаКраткое руководство пользователя
Чехол для хранения и переноскиВ некоторых комплектациях продукт может содержать дополнительные
компоненты. Полный список всех доступных комплектаций и дополнительных
аксессуаров см. в Руководстве пользователя или по адресу www.picotech.com.

42© © 2017–2022 Pico Technology Ltd. Все права защищены.

DO296-6Краткое руководство пользователя векторного анализатора цепей PicoVNA

5. Внешние соединители
5.1 Векторные анализаторы цепей PicoVNA 106 и 108Порт 1Тестовый ВходИндикатор
вход/тройника колебания
выходсмещения

Тестовый
Индикатор ВходПорт 2колебания тройника вход/смещения выход

USB-подключение Подача пост. тока Выключатель питания
10 MHz In

Trig In

S/N
OFF

10 MHz Out

12–15 V DC

Trig OutONВход/выход опорного сигнала 10 МГцЗаземление
шассиВход/выход
триггера

5.2 Стандарты ручной калибровки SOLT-STD-M/F и SOLT-PREM-M/FСогласованная нагрузка
Холостой ходПеремычка

Короткое замыкание

5.3 Стандарты автоматической (E-Cal) калибровки SOLT-AUTO-M/FИндикатор

Порт USBПорт AПорт B

DO296-6© © 2017–2022 Pico Technology Ltd. Все права защищены.

43Краткое руководство пользователя векторного анализатора цепей PicoVNA

6. Установка векторного анализатора цепей
PicoVNA
6.1 Установка программного обеспечения
1.

Подготовьте установщик программного обеспечения PicoVNA 2 [108:
PicoVNA 3], который находится на флэш-накопителе USB (входит в комплект
поставки) или по адресу
			www.picotech.com/downloads.

2.

Запустите установщик (щелкните правой кнопкой мыши и выберите Запускот имени администратора). Проследите за тем, чтобы процесс установки
завершился успешно.

3.

Подключите прибор PicoVNA к компьютеру и подождите, пока Windowsавтоматически устанавливает для него драйвер.

Установщик создаст каталог поддержки по адресу:C:\Users\<имя пользователя>\Documents\Pico Technology\
PICOVNA2[108: C:\Users\<имя пользователя>\Documents\Pico Technology\
PICOVNA3]
Этот каталог содержит следующие файлы (в примере приводится модель
PicoVNA 106):
•
•
•

•

xxxx-log.txt
номер)
CalKits\DefUnc.datумолчанию
DefCal.calпоследней калибровки)прибору ВАЦ
FactoryDefaults.calзаводскими значениями

Файл журнала статусов («xxxx» — серийный
Значения погрешностей измерения поКалибровочные данные по умолчанию (данные
— появляются после первого подключения кРезервная копия калибровочных данных сСледующее сообщение об ошибке является характерным для компьютеров с
Windows 7:Можно нажать
Пропустить дляпродолжения
44© © 2017–2022 Pico Technology Ltd. Все права защищены.

DO296-6Краткое руководство пользователя векторного анализатора цепей PicoVNA

6.2 Главное окно PicoVNA 2 [108: PicoVNA 3]Каналотображения 1Серийный
номеринструмента

Состояние
калибровки

Состояние
порта USBКаналотображения 3Состояние
колебания

Количество
точекизмерения

Используемая
Мощность
тестового сигнала частота

Канал
Канал
отображения 2 отображения 4Количество
усредненных
колебаний

Полоса пропускания
измерения

Значения маркеровдля выбранного
активного канала

Контроль сохраненияпо триггеру (только
PicoVNA 108)Значения
маркеров

Внутренняя температура (только
PicoVNA 108)Вы можете настроить вертикальную шкалу любого из прямоугольных графиков
(см. изображение выше) следующим образом:Для настройки
перетащитеДля настройки перетащите или щелкните
правой кнопкой мыши и введите

DO296-6© © 2017–2022 Pico Technology Ltd. Все права защищены.

45

Краткое руководство пользователя векторного анализатора цепей PicoVNA

6.3 Загрузка калибровочных комплектов
PicoVNA 106 или 108

PicoVNA 106 или 108

PicoVNA 106 или 108Порт 1Порт 1Порт 1Штекер

Порт 2Вставное
устройство

Порт 2Невставное устройство
(тестовые порты соединяются напрямую)Гнездо

Штекер

Штекер

Порт 2Невставное устройство
(тестовые порты не соединяются напрямую)Гнездо

Гнездо

DUT

DUT(Тестируемое устройство)(Тестируемое устройство)(Тестируемое устройство)Кабели с гнездами.
Используйте те же штекеры калибровочного
набора для обоих портов.

Кабели со штекерами. Используйтете же гнезда калибровочного
набора для обоих портов.

Кабель с гнездом.
Используйте штекер
калибровочного комплекта.

Кабель со штекером.
Используйте гнездо
калибровочного комплекта.

DUTОткалибруйте с использованием
фиксированных стандартов
ручной калибровки:Откалибруйте с использованием
фиксированных стандартов
ручной калибровки:Откалибруйте с использованием
фиксированных стандартов
ручной калибровки:SOLT-STD-F и SOLT-STD-M
или
SOLT-PREM-F и SOLT-PREM-M

SOLT-STD-M
или
SOLT-PREM-M

SOLT-STD-Fили
SOLT-PREM-FИли откалибруйте с помощьюИли откалибруйте с помощьюИли откалибруйте с помощью
PicoVNA 108 или стандарта
PicoVNA 108 или стандарта
PicoVNA 108 или стандарта
автоматической калибровки E-Cal: автоматической калибровки E-Cal: автоматической калибровки E-Cal:
SOLT-AUTO-Fили
SOLT-AUTO-M

SOLT-AUTO-M

SOLT-AUTO-FДля ручной калибровки SOLT:
•
•
•

•

Запустите программное обеспечение PicoVNA 2 [108: PicoVNA 3]
В главном меню выберите Tools (Инструменты) > Calibration kit(Калибровочный комплект)Щелкните Load P1 kit (Загрузить комплект P1), укажите путь к файлу сданными для калибровочного комплекта порта 1 (на флэш-накопителе USBиз комплекта поставки калибровочного комплекта), а затем щелкните Apply(Применить)Если необходимо использовать другой калибровочный комплект,подключенный к порту 2, щелкните Load P2 kit (Загрузить комплект P2),укажите данные для порта 2 для подключенного к нему комплекта, а затем
щелкните Apply (Применить)Выберите требуемый калибровочный комплект (один или несколько) взависимости от типа тестируемого устройства. Вы можете скопировать
калибровочные данные в каталог
C:\Users\<user name>\Documents\Pico Technology\PICOVNA2\CalKits\[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3\
CalKits\]для упрощения доступа к ним в будущем. Для этого нажмите кнопку Save
Kit (Сохранить комплект) в разделе Cal Kit Editor (Редактор калибровочных
комплектов).
46© © 2017–2022 Pico Technology Ltd. Все права защищены.

DO296-6Краткое руководство пользователя векторного анализатора цепей PicoVNA

Для PicoVNA 108 и автоматической калибровки:
•
•
•
•
•
•
•
•
•

Запустите программное обеспечение PicoVNA 3.
Подключите модуль E-Cal к свободному порту USB управляющего ПК.
В главном меню выберите Tools (Инструменты) > Calibration kit(Калибровочный комплект).
Щелкните Electronic Cal kit (Электронный калибровочный комплект).
Выберите модуль E-Cal и расположение адаптера порта, которое отвечает
тестовым портам.
Подключите модуль E-Cal, адаптер порта и тестовые порты в точном
соответствии с выбранной конфигурацией.
Щелкните Find (Найти) и загрузите данные EasyCal.
По завершении нажмите Exit (Выйти).
Щелкните Apply (Применить) в меню Параметры калибровочного
комплекта.

6.4 Установка разрешения
Прежде чем приступать к калибровке, щелкните Enhance (Повысить качество)в главном меню, чтобы установить минимальное разрешение по полосе
пропускания, необходимое для выполнения калибровки. Более низкие значения
приведут к более медленной калибровке.

Измерение

Максимальная
скорость
Самая высокая
точность,динамический
диапазон ~100 дБОбщее использование,высокая скорость,динамический
диапазон ~90 дБЛучший динамический
диапазон

DO296-6Полоса
пропусканияпри калибровке

Калибровочное
усреднение

Калибровочная
мощность

10 кГц

Нет+0 дБмВо время измерения
установите для полосы
пропускания значение 140кГц

100 Гц

Нет-3 дБмВо время измерения
оставьте для полосы
пропускания значение 100 Гц

1 кГц

Нет

+0 дБмВо время измерения
оставьте для полосы
пропускания значение 1 кГц

10 Гц

Нет

+6 дБмВо время измерения
оставьте для полосы
пропускания значение 10 ГцПримечание

© © 2017–2022 Pico Technology Ltd. Все права защищены.

47

Краткое руководство пользователя векторного анализатора цепей PicoVNA

6.5 Настройка параметров калибровки
Щелкните Calibration (Калибровка), чтобы открыть окно Calibration (Калибровка):Загружаются
вашикалибровочные
комплекты

1. Установите
параметры
колебания

3. Нужно
выбрать
измерение

2. После установки
необходимой полосы
пропускания
колебаний, примените
значения

5. ВАЖНО!
По завершении примените калибровку

4. Выполните
необходимые
процедуры
калибровки.
Возможна любая
последовательность,однако порядок
вниз-затем-насквозь
обычно сводит кминимуму
количество
операций
подключения.

6.6 Настройка параметров отображения
Щелкните Display (Отображение) в главном окне, чтобы открыть окно Display Set
Up (Настройка отображения):3. Выберите
настраиваемый канал

1. Выберите
одноканальное,двухканальное иличетырехканальное
отображение
4. Выберите дляотображения
значений навертикальной шкале
графиков

2. Выберите
канал, который
будетотображаться напанели маркеров

5. Примените настройки перед выбором
следующего канала

После завершения настроек щелкните Start (Начать) в главном окне, чтобы
начать измерения.

48© © 2017–2022 Pico Technology Ltd. Все права защищены.

DO296-6Краткое руководство пользователя векторного анализатора цепей PicoVNA

6.7 Работа в демонстрационном режиме
Демонстрационный режим позволяет ознакомиться с функциями программного
обеспечения PicoVNA 2 [108: PicoVNA 3] без необходимости в подключении
измерительного прибора.
••Чтобы открыть демонстрационный режим, запустите PicoVNA 2 [108:
PicoVNA 3], не подключая прибор к компьютеру.
Щелкните Ignore (Игнорировать) в появившемся диалоговом окне.

После этого PicoVNA 2 [108: PicoVNA 3] предложит на выбор несколько
вариантов демонстрационных измерений.

7. Подробная информация
7.1 Бесплатная информационная рассылка
Pico Technology ежемесячно отправляет по электронной почте бесплатную
информационную рассылку. Подписка включает: новости о запуске
продуктов, обновления программного обеспечения, характерные примеры
использования продуктов, а также различные советы и рекомендации. Мыобращаемся с вашим адресом электронной почты и другими данными как сконфиденциальной информацией, которую запрещено передавать третьим
лицам. Чтобы подписаться, перейдите по адресу:www.picotech.com/signup

7.2 Обновление программного обеспечения и замена
Актуальные версии программного обеспечения и руководств Pico можно
бесплатно загрузить по адресу:www.picotech.com/downloads
Если вам требуется новый диск с программным обеспечением, обратитесь в
Pico Technology или к своему поставщику оборудования. За данную услугу может
взиматься небольшая плата.

7.3 Написание собственного программного обеспечения
Примеры программ для MATLAB и MATLAB RF toolbox, LabVIEW, C, C# и Pythonдоступны на GitHub (github.com/picotech). Примеры включают несколько
функций использования прибора и управления им.

7.4 Руководство пользователя и руководство программиста
Руководство пользователя PicoVNA и Руководство программиста PicoVNAдоступны на флэш-накопителе USB из комплекта поставки или по адресу:www.picotech.com/downloads

DO296-6© © 2017–2022 Pico Technology Ltd. Все права защищены.

49

Краткое руководство пользователя векторного анализатора цепей PicoVNA

7.5 Характеристики
Руководство пользователя и Паспорт содержат актуальные характеристики
вашего векторного анализатора цепей PicoVNA.

7.6 Техническая поддержка
Постоянно обновляемую информацию о технической поддержке можно найтина нашем форуме Pico Technology по адресу:www.picotech.com/tech-support

7.7 Гарантийные обязательства и возврат
YEAR W A R R A N T Y
На векторные анализаторы цепей PicoVNA распространяется
трехлетняя гарантия с возвратом к производителю. Полную
информацию о гарантийных условиях см. в Руководстве пользователя
PicoVNA.

50© © 2017–2022 Pico Technology Ltd. Все права защищены.

DO296-6

PicoVNA 矢量网络分析仪快速入门指南

简体中文

1. 简介

感谢您购买 PicoVNA 矢量网络分析仪。
可提供以下型号：••

PicoVNA 106：带内置 Bias-T 网络的 6 GHz 两个端口 VNA。

PicoVNA 108：
8.5 GHz 两个端口 VNA，带内置 Bias-T 网络、混频器测量值以及触发
保存功能。

有关进行基本安装后的更多设置信息，请参见随 PicoVNA 2 [108:PicoVNA 3]* 软件一起提供的《用户指南》。
有关详尽规格，请参见 PicoVNA 数据表。

* 除非另有说明，否则本指南中的信息可应用于 PicoVNA 产品系列的所有型号。
专门应用
于 PicoVNA 108 的信息书写为以下类似情况：“[108: <信息>]”。

2. 安全信息

为防止可能发生的电击、火灾、人身伤害或产品损坏，请仔细阅读这些安全信息，然后再尝试安装或使用本产品。
此外，在使用和靠近电时，遵循所有普遍接受的安全措施和程序。
本仪器及其配件的设计符合 EN 61010-1 标准（测量、控制和实验室使用的测量电气设备安全要求）。
本指南包含下列安全说明：

警告 表示存在可能造成人身伤害或死亡的条件或做法。

小心 表示存在可能造成相连产品或设备损坏的条件或做法。

警告
本产品仅供受过培训和具备相应资质的专业技术人员使用。
为防止造成人身伤亡，必须按指示使用产品，且仅使用 Pico Technology 提供或建议的附件。
若以
制造商未指定的方式使用，则可能削弱产品提供的保护功能。

DO296-6

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

51

PicoVNA 矢量网络分析仪快速入门指南

2.1 符号

这些安全和电气符号可能出现在产品上或本指南中。
符号描述

此接线柱可用于进行接地测量。
它不
是安全或保护性接地。

接地接线柱
机箱接线柱
可能存在电击风险
小心请勿将此产品当作未分类的城市垃圾处理产品上出现此符号表示需要阅读本
安全和操作手册。

2.2 最大输入和输出范围警告为防止电击，切勿尝试应用以下指定的最大最以外的信号电平：

以下表格指出了 VNA 及其校准上的最大输出电压和输入过压保护范围。
过压保护范围是
可施加的不会损坏该分析仪或配件的最高电压。
最大值工作电压（输出或输入）分析仪连接器端口 1 和 2

Bias-T 1 和 2
触发器和参考输入触发器和参考输出+10 dBm（约 710 mV RMS）
±15 V DC

校准配件连接器
SOLT-PREM 和 SOLT-STD 端口
SOLT-AUTO 端口 A/B
警告

0 V 至 +5 V过压或过流保护+20 dBm.
（约 2.2 V RMS）
250 mA±6 V pk
不要应用电压
过压保护
+20 dBm+20 dBm

下表中超过电压限值的信号被 EN 61010 定义为“危险带电”。
±60 V DC
警告52

EN 61010-1:2010+A1:2019 的信号电压限值30 V AC RMS± 42.4 V pk 最大值为防止电击，操作可能存在危险电压的设备时，请采取所有必需的安全预防措施。
版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

DO296-6

PicoVNA 矢量网络分析仪快速入门指南警告为了避免设备损坏和可能受伤，不要在额定电压或环境范围外操作该分析仪或配件。
小心
超出任何连接器上的过压防护范围，都有可能导致仪器和其它连接设备永久性损坏。
为了防止永久性损坏，不要对 VNA 的触发器和参考输出应用电压。

2.3 接地警告分析仪或 SOLT-AUTO E-Cal 模块通过 USB 缆线的接地连接仅用于功能目的。
分析仪及配件无防护性安全接地装置。
为了防止人身伤害或死亡或对分析仪造成永久性损坏，切勿将输入或输出接地（机箱）连接至任何电源。
为防止人身伤害或死亡，请使用电压表检查分析仪的
接地端与要连接的点之间是否存在明显交流或直流电压。
小心为防止接地不良引起信号衰减，请始终使用分析仪随附的高质量 USB 线缆。

2.4 外部连接

警告为了避免伤亡，请仅使用随分析仪一起提供的电源适配器。
这是您所在的国家
批准的电压和插头配置。

PicoVNA 型号

USB 连接PicoVNA 108

USB 2.0 .
兼容 .
USB 3.0

PicoVNA 106

SOLT-AUTO E-Cal 模块电压

12 至 15 V DC

外部电源
电流

1.85 A pk
不适用

总电源
22 W
25 W

警告
无线电频率封隔
该仪器可隔离扫频或 CW 射频信号源（最大 +6 dBm 时为 300 kHz 至 6.02 GHz[108：最大 +6 dBm 时为 300 kHz 至 8.50 GHz]）。
该仪器及所提供的附件的设计目的是隔离而不是辐射（或易受）射频，因为射频可能会干扰其他设备或无线电控制和通信的运行。
为了防止人身伤害或死亡，请仅连接到适当的指定连接
器、电缆、配件和测试设备，除了经过批准的测试设施内部的天线或其他可控条
件下的天线，不要连接到其他天线。

2.5 环境警告本产品仅适用于室内或室外的干燥位置。
产品的外部主电源仅供室内使用。

DO296-6

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

53

PicoVNA 矢量网络分析仪快速入门指南警告为防止人身伤害或死亡，请勿在潮湿的条件下或者附近有爆炸性气体或蒸汽时
使用 VNA 或配件。
小心为防止损坏，请始终在适当的环境中使用和存储 VNA 或配件。

温度
湿度
海拔
污染度

存储
工作
–20 °C 至 +50 °C+5 °C 至 +40 °C
最大为 80% 相对湿度（非冷凝）2000 m2

小心请勿堵塞仪器后面的通风孔，因为过热会造成损坏。

请勿通过通风孔插入任何物体，因为内部干扰将会造成损坏。

2.6 产品照管本产品及其配件不含可由用户维修的部件。
维修、保养和校准需要专用的测试设备且必须仅由 Pico Technology 或获批的服务提供商执行。
如果不在 Pico 三年质保范围内，这些服
务可能需要收费。
警告为避免人身伤害或死亡，请勿使用出现任何损坏的产品；如果您担心出现任何异常行为，请立即停止使用 VNA 或配件。
小心使用之前，请定期检查设备和所有探针、连接器、线缆和配件，查找是否存在损坏或污染迹象。

为了对设备或所连接的装备造成损坏，请勿乱动或拆卸该分析仪、机箱部件、连接器或配件。
清洁产品时，请使用软布以及温和肥皂溶液或洗涤剂水，不要让液体进入仪器
或配件外壳。
小心避免所有连接导线（包括所有同轴导线和连接器）出现机械应力或过小的弯曲半径。
处理不当可能会造成侧壁变形，从而会降低性能。
特别是，注意测试端口导线不能形成小于 5 厘米（2 英寸）的弯曲半径。

为了防止测量错误并延长测试导线和配件连接器的有效寿命，请确保不要让液
体和颗粒污染物进入。
始终装上提供的防尘盖，拧紧时请使用正确的扭矩。
Pico
推荐：对于提供的所有不锈钢连接器，推荐 1 Nm（8.85 英寸-磅），或当接口使用铜质或镀金连接器时，推荐 0.452 Nm（4.0 英寸-磅）。

54

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

DO296-6

PicoVNA 矢量网络分析仪快速入门指南

3. 系统要求

为了确保 PicoVNA 矢量网络分析仪和 PicoVNA 软件正确运行，您必须使用一台符合系统
要求且能够运行下表中所示的其中一种操作系统的计算机。
处理器、内存、可用磁盘空间
操作系统
端口
显示分辨率

如操作系统所需Microsoft Windows 7、8 或 10，32 或 64 位。
USB 2.0（同时兼容 USB 3.0）
最小 1280 x 720

4. PicoVNA 套件内容

PicoVNA 矢量网络分析仪套件包含以下物品：
•
•
•
•
•
•
•

PicoVNA 106 [108：
PicoVNA 108] 矢量网络分析仪
PicoVNA 2 [108：
PicoVNA 3] USB 闪存驱动器带四个国际插头转换器的 DC 电源
USB 2.0 线缆，1.8 米
SMA/PC3.5/2.92 和 N 型固定扳手
快速入门指南存放和携带箱某些产品包可能还含其它一些物品。
有关所有可用套件和其他配件的完整列表，请参阅《用户指南》或访问www.picotech.com。

DO296-6

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

55

PicoVNA 矢量网络分析仪快速入门指南

5. 外部连接

5.1 PicoVNA 106 和 108 矢量网络分析仪

端口 1

测试
I/O

Bias-T输入扫频
指示器USB 连接10 MHz In

扫频
指示器

DC 输入

测试
I/O

Bias-T输入

端口 2开启/关闭开关
Trig In

S/N
OFF

10 MHz Out

12–15 V DC

Trig OutON

10 MHz 参考输入/输出

机箱接地

触发输入/输出

5.2 SOLT-STD-M/F 和 SOLT-PREM-M/F 手动校准标准
加载

打开

通过

短

5.3 SOLT-AUTO-M/F 自动 (E-Cal) 校准标准
指示器

端口 A

56

USB 端口端口 B

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

DO296-6

6. 安装 PicoVNA 矢量网络分析仪

PicoVNA 矢量网络分析仪快速入门指南

6.1 安装软件1.
2.
3.

从随您的 VNA 提供的 USB 闪存驱动器或从.
			
www.picotech.com/downloads 上获取 PicoVNA 2 [108:
PicoVNA 3] 软件安装程序。
运行该安装程序（右键单击然后选择以管理员身份运行）并确保安装成功。
将 PicoVNA 设备连接到计算机并等待 Windows 自动安装驱动程序。

安装程序将创建以下支持目录：C:\Users\<user name>\Documents\Pico Technology\PICOVNA2[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3]

此目录包含以下文件（PicoVNA 106 示例）：
• xxxx-log.txt• CalKits\DefUnc.dat• DefCal.cal•

FactoryDefaults.cal这是状态日志文件。
“xxxx”指序列号默认的测量不确定值默认的校准数据（最后使用的校准）.
- 将在首次连接到 VNA 设备后显示
来自厂家的备份校准数据

在 Windows 7 机器上，出现以下错误消息很常见：可以安全地单击
忽略继续

DO296-6

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

57

PicoVNA 矢量网络分析仪快速入门指南

6.2 PicoVNA 2 [108: PicoVNA 3] 主窗口显示通道 1

仪器序
列号

校准状态

USB 端口状态

显示通道 3

扫频状态

测量点数量

测试信号电源 使用中的频率测量带宽显示通道 2

平均扫频数量显示通道 4

选定活动通
道的标记值

触发控制时保存（仅
限 PicoVNA 108）标记值

内部温度（仅限
PicoVNA 108）您可以调整以上窗口中任何矩形绘图中的垂直缩放比例，如下所示：
拖动进行调整58

拖动进行调整或右键单击并输入

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

DO296-6

PicoVNA 矢量网络分析仪快速入门指南

6.3 加载校准套件
PicoVNA 106 或 108

端口 1

阳接头
连接器

PicoVNA 106 或 108

端口 2

可插入
设备

端口 1

非可插入设备
（测试端口将直接连接在一起）

阴接头
连接器

阳接头
连接器

阳接头
连接器

阳接头电缆。
使用阴接头校准套件。

PicoVNA 106 或 108

端口 1

端口 2

非可插入设备
（测试端口将不会直接连接在一起）

阴接头
连接器

阴接头
连接器

DUT

DUT

阴接头线缆。
两个端口都
使用相同阳接头套件。

阳接头电缆。
两个端口都
使用相同阴接头套件。

DUT
阴接头电缆
使用阳接头校准套件。

端口 2使用固定的手动校准标准进行校准：

使用固定的手动校准标准进行校准：

使用固定的手动校准标准进行校准：或使用 PicoVNA 108 和自动 E-Cal 标
准进行校准：

或使用 PicoVNA 108 和自动 E-Cal 标
准进行校准：

或使用 PicoVNA 108 和自动 E-Cal 标
准进行校准：

SOLT-STD-F 和 SOLT-STD-M.
或.
SOLT-PREM-F 和 SOLT-PREM-M
SOLT-AUTO-F.
或.
SOLT-AUTO-M

SOLT-STD-M.
或.
SOLT-PREM-M
SOLT-AUTO-M

SOLT-STD-F.
或.
SOLT-PREM-F
SOLT-AUTO-F对于手动 SOLT 校准：•••运行 PicoVNA 2 [108: PicoVNA 3] 软件
在主菜单中，选择工具 > 校准套件单击加载 P1 套件，找到端口 1 校准套件（在随校准套件提供的 USB 闪存驱动器上）的数据文件，然后单击应用• 如果要在端口 2 上使用不同的校准套件，请单击加载 P2 套件，为端口 2 套件选择数据，然后单击应用根据要测试的设备选择所需的校准套件。
为了以后能更快访问，您可能希望将校准套件数据复制到 .
.
C:\Users\<user name>\Documents\Pico Technology\PICOVNA2\CalKits\[108: C:\Users\<user name>\Documents\Pico Technology\PICOVNA3\CalKits\]
文件夹。
可以使用保存套件按钮（在校准套件编辑器中）来执行此操作。

对于 PicoVNA 108 和自动校准：
•
•
•
•
•
•
•
•
•

运行 PicoVNA 3 软件。
将 E-Cal 模块连接到控制 PC 上的闲置 USB 端口。
在主菜单中，选择工具 >校准套件。
单击电子校准套件。
选择 E-Cal 模块和端口适配器排列以适合测试端口。
连接 E-Cal 模块、端口连接器和测试端口，使其与选定的配置完全匹配。
单击查找并加载 EasyCal 数据。
完成后请选择退出。
单击校准套件参数菜单中的应用。

DO296-6

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

59

PicoVNA 矢量网络分析仪快速入门指南

6.4 设置分辨率继续进行校准之前，请单击主窗口中的增强来设置您想要让校准支持的最小分辨率带宽。
值越低将导致校准速度越慢。
测量

最快速度

最佳精度和 ~100 dB 动态范围常规应用、快速、~90 dB动态范围
最佳动态范围

校准
带宽

校准
平均

100 Hz无

10 kHz
1 kHz

10 Hz

校准
电源

无

+0 dBm无+0 dBm无

6.5 设置校准参数–3 dBm

+6 dBm备注

测量过程中将带宽设置为 140
kHz
测量过程中保持带宽设置为
100 Hz测量过程中保持带宽设置为 1
kHz
测量过程中保持带宽设置为
10 Hz

单击校准可打开校准窗口：
1.设置扫频参数

您已加载的
校准套件
3.选择所需的测量

4.执行所需的校
准步骤。
可以使用
任何顺序，但使用
先向下再横向顺序通常可以最小化连接步骤的数量。

2.设置所需的扫频带宽后，请应用值

5.重要说明！
完成时，请应用校准60版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

DO296-6

6.6 设置显示参数

PicoVNA 矢量网络分析仪快速入门指南单击主窗口中的显示可打开显示设置窗口：3.选择要设置的通道

1.选择一个通
道、两个通道或四个通道视
图
4.选中可在图形上显示垂直缩放
比例2.选择要在标记面板中显示的通道完成后，请单击主窗口中的开始来开始测量。

5.选择下一个通道之前请
应用设置

6.7 在演示模式中运行

演示模式允许您在无需分析仪运行的情况下探索 PicoVNA 2 [108: PicoVNA 3] 软件的功能。
••

要进入演示模式，请在不连接分析仪的情况下运行 PicoVNA 2 [108: PicoVNA 3] 软件。
单击所出现对话框中的忽略。

PicoVNA 2 [108: PicoVNA 3] 将为您提供演示测量的选择。

DO296-6

版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

61

PicoVNA 矢量网络分析仪快速入门指南

7. 更多信息

7.1 免费通讯期刊

Pico Technology 通过电子邮件发送免费月刊。
为您提供有关产品首发、软件升级、应用说
明、提示及建议等方面的信息。
我们将会对您的电子邮件地址及其他详细信息进行保密，不会将其透露给任何第三方。
若要订阅，请访问:7.2 软件升级和更新www.picotech.com/signup

所有 Pico 软件和手册的最新版本均可从以下网址免费下载：www.picotech.com/downloads

如果需要新的软件光盘，请联系 Pico Technology 或您的经销商。
此项服务可能需收取少量费用。

7.3 编写您自己的软件

MATLAB 和 MATLAB RF 工具箱、LabVIEW、C、C# 和 Python 的代码示例可从 GitHub(github.com/picotech) 上获取。
这些示例包括多种仪器寻址和控制方式。

7.4 用户和编程人员指南

PicoVNA 用户指南 和 PicoVNA 编程人员指南在所提供的 USB 闪存驱动器上，也可从以下地址获取：
www.picotech.com/downloads

7.5 规格《用户指南》和数据表包含有关 PicoVNA 矢量网络分析仪的最新规格信息。

7.6 技术支持

Pico Technology 支持论坛提供定期更新的技术支持信息：

7.7 质保与退回

www.picotech.com/tech-support

这些 PicoVNA 矢量网络分析仪提供有三年的返厂质保。
有关完整条款和条件，Y E A R W A R R A N T Y
请参阅PicoVNA 用户指南。

62版权所有 © 2017–2022 Pico Technology Ltd。
保留所有权利。

DO296-6

UK headquarters

Asia-Pacific regional office

英国本社

アジア太平洋地域管轄支社

英国总部

亚太地区办公室

영국 본사Центральный офис в Соединенном
Королевстве
Pico Technology
James House
Colmworth Business Park
St. Neots
Cambridgeshire
PE19 8YP
United Kingdom아시아 태평양 지사Региональный офис в Азиатскотихоокеанском регионе
Pico Technology
Room 2252, 22/F, Centro
568 Hengfeng Road
Zhabei District
Shanghai 200070
PR China

 +44 (0) 1480 396 395
 sales@picotech.com

 +86 21 2226-5152
 pico.asia-pacific@picotech.com

North America regional office

EU authorised representative

Pico Technology
320 N Glenwood Blvd
Tyler
TX 75702
United States

Pico Technology GmbHIm Rehwinkel 630827 Garbsen
Germany +1 800 591 2796
 sales@picotech.com

 +49 (0) 5131 907 62 90
 info.de@picotech.com

Pico Technology is an internationally registered trademark of Pico Technology Ltd. PicoVNA isa registered trademark of Pico Technology Ltd. Windows is a registered trademark of Microsoft
Corporation in the United States and other countries.
Copyright © 2017–2022 Pico Technology Ltd. All rights reserved.
DO296-6

www.picotech.com