Washdown Scales

SeriesELBISR
EUBMSISO
9000 CERTIFIED...Clearly a Better Value
http://www.aandd.jp

IP68 Washdown – A New Standard ofPlatform Scales for Harsh Environments
The SC/SE Series, equipped with a completely sealed indicator, achieves a submersiblelevel of watertightness *, while providing high accessibility and excellent weighingperformance, including stabilization in just 1 second!

* 1 m under water for 24 hours

SC
SC

SeriesPremium Resistance toHostile Environments!
Stainless steel(SUS304) frame

SC-30KAM
60KAM
150KAM

SC-60KAL
150KALSESeriesHigh Resistance toHostile Environments!

Steel frame

SE-30KAM
60KAM
150KAM

SE-60KAL
150KALFeatures
IP68 dust and water-proof (submersible)
Washdown construction for easy cleaning
4500 hours of continuous operation with 6 D batteries iSupersized LCD of 39 mm character height

Easily removable battery unit

Easily switchable, multiple weighing units (kg, lb, oz, and pcs)
High speed stabilization of 1 second
Battery life indicator

Stainless steel weighing pan (SC: SUS304, SE: SUS430)
Waterproof and corrosion-resistant load cellii

Quick USB interface iii and RS-232C + comparator relay
output available as optionsi) Batteries are not included.
ii) Aluminum anodized and coated with epoxy resin with protective silicon over wiring.
iii) Cables are not included.

LCD
Display

External
Calibration

Quick USB

RS-232C
Interface

Counting
Function

Comparator
Function

Auto PowerOFF

NormalBattery

IP68

SoftwareSpecifications
SC-30KAM
SE-30KAM
kg

lb

oz

SC-60KAM
SE-60KAM

SC-150KAM
SE-150KAM

SC-60KAL
SE-60KAL

SC-150KAL
SE-150KAL

Capacity

30

60

150

60

150

Min. display

0.01

0.02

0.05

0.02

0.05

0.005 *

0.01 *

0.02 *

0.01 *0.02 *Capacity

66

130

330

130330Min. display

0.02

0.05

0.1

0.05

0.1

0.01 *

0.02 *

0.05 *

0.02 *

0.05 *

Capacity

1050

2100

5200

2100

5200

Min. display

0.5

12

12

0.2 *

0.5 *

1*

0.5 *

1*

5 pieces (Can be changed to 10, 20, 50 or 100)

Number of samples
Maximum count60,000 pcs

60,000 pcs

75,000 pcs

60,000 pcs

75,000 pcs

Minimum unit weight

0.0005 kg

0.001 kg

0.002 kg

0.001 kg

0.002 kg

0.005 kg

0.01 kg

0.02 kg

0.01 kg

0.02 kg±0.01 kg

±0.02 kg

±0.05 kg±0.02 kg±0.05 kg

Repeatability(Std. deviation)
Linearity
Sensitivity drift±20 ppm / °C (5 °C to 35 °C / 41 °F to 95 °F)

Display

7 segment LCD (Character height: 39 mm)
10 times per second

Display update–10 °C to 40 °C / 14 °F to 104 °F, 85%R.H. or less (no condensation)

Operating conditions

Size D (R20P / LR20) batteries x 6

Power supply

Approximately 4,500 hours using alkaline batteries

Battery operation

390 x 530 mm / 15.4 x 20.9 in.

300 x 380 mm / 11.8 x 15.0 in.

Weighing pan size
SCDimensionsSE300(W) x 543(D) x 717(H) mm

390(W) x 688(D) x 717(H) mm

11.8(W) x 21.4(D) x 28.2(H) in.

15.4(W) x 27.1(D) x 28.2(H) in.

300(W) x 582(D) x 722(H) mm

390(W) x 732(D) x 722(H) mm

11.8(W) x 22.9(D) x 28.4(H) in.

15.4(W) x 28.8(D) x 28.4(H) in.
16 kg / 35.3 lb

12 kg / 26.5 lb

Mass (approximate)Calibration weight

30 kg

60 kg

150 kg

60 kg

150 kg(factory setting)

60 lb

120 lb300 lb120 lb300 lb* Factory settingOptions SCE-02 Quick USB interface

and SCE-03 cannot be used at the same time.
* SCE-02* The SC / SE is not IP68 compliant with SCE-02 or SCE-03.

SCE-03 RS-232C + comparator relay output

External Dimensions (mm/inches)

SC series

60°

SE series

60°
D

φ38.1/1.5

722/28.4

717/28.2

D

40/1.6 × 40/1.6
BB

173/6.8

C

E

E

Note

202/8.0

FFC

A

A

Model

A

B

C

D

ModelEF

E

F

SC-30KAM / 60KAM / 150KAM 300/11.8 380/15.0 300/11.8 543/21.4

117/4.6

220/8.7

SE-30KAM / 60KAM / 150KAM300/11.8 380/15.0 300/11.8 582/22.9

118/4.7

220/8.7

SC-60KAL / 150KAL

123/4.8300/11.8

SE-60KAL / 150KAL

390/15.4 530/20.9 440/17.3 732/28.8

128/5.0300/11.8

390/15.4 530/20.9 440/17.3 688/27.1

ABCDNote) The pole support foot is for the SE-30KAM/60KAM/150KAM only.

...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www.aandd.jp1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

A&D Australasiaia Ptyy Ltd.
d.
32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66<German Sales Office>509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599

Große Straße 13 b 22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

A&D
D InstInstrumentsuments India PrPrivatee Limited
mited

Manhattan Bldg. 8F, 36-2 Yoido-dong, Youngdeungpo-gu, Seoul, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280*SC/SE-ADCC-01-PR1-10403