UV Sensor "UV-Cure-HT"
UV sensor for high UV irradiance and high operating temperature (170°C)
1/2General Features

The “UV-Cure-HT” is a UV sensor for high UV irradiance and high operating temperature
up to 170°C. Typically this sensor is used for measurement of high UV radiation at hightemperature (up to 170°C / 338°F) e.g. for curing and drying processes. The sensor works
with a diffuser made of radiation hard and temperature resistant microporous quartz
glass and is configured with a heat resistant cable. The signal output is photocurrent(nA ... μA). The UV-Cure-HT needs an external amplifier (such as sglux RADIKON-simple).
The spectral responsivity will be configured upon individual customer’s requirements
which are clarified within the order process.
Figure 1 shows the different options regarding the spectral responsivity. Our sales team is happy to assist ourcustomers selecting the best suitable responsivity for the specific application. Alternatively, technical reports andselection guides are available on our website providing further assistance. A PTB traceable calibration can beordered.

spectral responsivity selection options
Figure 1 shows the available spectral responsivites. Table 1 shows the position of the peak and the 10% of maximun
margins. For UV measurement, by default, unfiltered broadband SiC is applied. If a UV source also emits radiationthat must not contribute to the sensor’s signal a filtered SiC sensor (UVC, UVB or UVA only) is to be selected. For
measurement of radiation above 390nm GaP based detectors are used.

Table 1: position of peak responsivity and
10% of maximum margins, values in nml_Slowl_Shigh280

221

358

331

309

367

280

231

309

275

225

287

UV+VIS

445

240

560

BLUE

445

390

515

SR

PeakBroadBUVAUVBUVC

Figure 1: available spectral responsivities

sglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.de
Rev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.

UV Sensor "UV-Cure-HT"
UV sensor for high UV irradiance and high operating temperature (170°C)
2/2general Specifications
Fixed Specifications Parameter
Dimensions and Field of view
WeightTemperature coefficient (30 to 65°C)

Value
Please refer to drawing and graph at the bottom of this page
100 g
< 0.1%/K

Operating temperature-55 to +170˚C

Storage temperature-55 to +170˚C

Humidity
Temperature Sensor< 80%, non condensing
electrical resistance PT100 Type K, class B

Signal output

photodiode current approx. 1 nA ... 1 μA, needs external amplifier

Connections

high temperature two wire cable with open wires or BNC plug

Configurable Specifications Parameter

Value

Spectral sensitivity

Broadband UV, UVA, UVB, UVC, Bluelight or UV+VIS (see Fig. 1 at page 1)

Measurement range

The measurement range depends on the used amplifier.

Field of  View and Drawing (drawing values in mm)

54.5
3118

Ø10

Ø22M 22x 1.5

Ø22

wrenchwidth 190.8
cable glandwindow viewsglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.deRev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.

Sensor Probes Overview
laboratory & experiments
UV-Surface
Universal radiometric UV sensor
for calibration and reference
measurements, cosine correction.
Often used with radiometer SXL55.
UV-Cosine
Waterproof dirt repellent UV sensor
for outdoor measurement, cosine
field of view. Also available as UVIsensor (ERYCA), M20x1.5 thread
UV-AirAxial measuring screw-in UV sensor
very good EMC properties, M22x1.5thread
TOCON-Probe
Miniature UV sensor with0 to 5 V voltage output,
M12x1 thread

special applications
UV-ArcWaterproof UV sensor for measurement ofelectric arcs between overhead contact wires
and pantograph, complies with EN 50317,
G3/4" threadsglux ERYCA
highaccuracy
UV-Indexsensor,
measurement uncertainty is <5%. The sensor
complies with ISO 17166, M20x1.5 thread

UVI-Solo
like sglux ERYCA but configured as aready-to-mount system (avaliable for pole orrailings assembly)

UV-Wirelesswireless UV sensor with a display unit forintensity and dose measurement

DUty Sensors monitoring
UV disinfection of Air,surfaces and water
UV-Sanitize
UV sensor for monitoring of air and surface UV
disinfection systems, configurable formonitoring of Hg low pressure lamps, excimer
lamps or xenon flash lamps, M20x1.5 thread
UV-Water-G3/4
UV sensor for operation in pressurized water(10 bar), for Hg medium and low pressure
lamps. Also available as UV-Water-G1/4,G3/4" thread
UV-Water-PTFE
PTFE UV sensor for operation in pressurizedwater (10 bar), only for Hg low pressure
lamps or LEDs, G1/4" thread

UV-ÖNORM / UV-DVGW
UV sensor for DVGW(160°) and ÖNORM
certified water purifiers, also avaliable as
UV-DVGW (40°). The sensors comply withÖNORM M5873, DVGW W294(06), DIN19294
UV-Radial
Waterproof side looking UV sensor formonitoring of lamp bundles, for operation ina cladding tube or directly in water, M20x1.5
threadHIGH UV radiation
UV-CureUV sensor for high irradiance(>100mW/cm2) for LED curing or cooled
medium pressure lamps, M22x1.5
thread (temperature sensor available).
UV-Cure_HTLike UV-Cure but for temperatures up to170°C, e.g. for uncooled medium pressure
systems, M22x1.5 threadsglux GmbH | Richard-Willstätter-Str. 8 | D–12489 Berlin | Tel. +49 30 5301 5211 | welcome@sglux.de | www.sglux.de
Rev. 5.0 Due to our strive for continuous improvement, specifications are subject to change within our PCN policy according to JESD46C.