Intrinsically Safe Platform Scales

HV-CEP/ HW-CEP
SeriesMinimum Investment,
Maximum Safety!

www.aandd.jp

Danger Comes When You Least Expect It
Boom! Sparks in potentially explosive atmospheres (like those that arise wherepetrol and other ﬂammable chemicals are processed) can turn a plant ﬂoor into araging inferno in an instant. If you perform weighing in these types ofenvironments, your equipment must have the proper features to prevent explosions
and loss of life.
The HV-CEP/HW-CEP series of internationally certiﬁed, intrinsically safe*1 platform
scales will always keep you and your colleagues safe from the risk of these tragedies!
the energy available to a level insuﬃcient to cause*1 Limitssparks or heat that can ignite ﬂammable gases or vapors

Areas in which the HV-CEP/HW-CEP series is certiﬁed for safe use
Flammable gases/
vapors present
continuously

Flammable gases/
vapors presentintermittently

Flammable gases/
vapors present
only accidentally✔

✔

✔
✔

IECEx
ATEX
FM/FMc (NEC505)

Zone 0FM/FMc (NEC500)

IECEx:
ATEX:

Zone 1✔

Division 1

Ex ia IIB T4 Ga
II 1G Ex ia IIB T4 Ga

FM/FMc (NEC505): Class I, Zone 0, AEx/Ex ia IIB T4
FM/FMc (NEC500): IS / I / 1 / C,D / T4
Tamb -25 to +40℃

Zone 2Division 2
You can comfortably bring the scale tolocations where ignitable concentrationsof ﬂammable gases, vapors or liquids arepresent continuously or for long periods oftime under normal operating conditions(i.e. Zone 0 or Division 1).

✔

IP65 dust & waterproof base unit withstainless steel (SUS430) panIn addition to being intrinsically safe, the base unit of theHV-CEP/HW-CEP series can be washed with water and canweigh wet objects without problems. Further, the surface ofthe pan is resistant to chemicals, scratches and rust, andeasy to keep clean and hygienic.
IP65 base unit with a SUS430 pan

✔

Multi-colored 5-step LED comparator lights
With the clearly-discernible LED signals, check weighing (judging whether the measured
weight is within the designated limits) becomes far more accurate and eﬃcient than with anumeric display alone.

■ Auto-tare and negative comparison functions

LED comparator lights

・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・・

The scale can perform tare automatically when theweight reaches the set target range. Additionally,
negative values can be used as comparator limits. Withthese functions combined, you can perform "take-away"
check weighing, that is, removing a target amountrepeatedly from its container with no key press.

LCD display
with white backlight✔

LCD display (26 mm character height)
with white backlight・・・・・・・・・・

Do not underestimate the eﬀect of display visibilityon operator work eﬃciency, fatigue and tendency tomake errors, especially in areas where illumination
is not suﬃcient.

Alphanumeric keypad✔

Quick and easy settings using the alphanumeric keypad
Direct alphanumeric input makes it simple and straightforward to enter/re-enter preset tarevalues, comparator limits, calibration values and various internal settings.

■ Key lock function

You can inhibit the use of certain keys to prevent operators from making unwanted changes to thescale settings.

✔

Long battery life of approx. 1,500 hours *2
The HV-CEP/HW-CEP series operates on 4 D batteries*3, making it unnecessary to secure powerfrom a safe area via a safety barrier.

■ Auto power OFF function

To avoid wasting power, the scale powers oﬀ automatically when it is inactive for a set period of time (5to 60 minutes).
LED and backlight OFF (or approx. 1,000 hours with LED and backlight ON)*23 With
The batteries to be used are speciﬁed: DURACELL MN1300, ENERGIZER E95, or* Panasonic LR20 (XJ).

Other Useful Features✔ High-speed measurement—response in 1 second or less (typical for the HV-CEP series) * 4✔ Adjustable response characteristics and display hold function to be used depending on the weighing/environmental conditions✔ Counting mode to count a number of small pieces of the same mass value instantly and precisely✔ Accumulation (M+) function to display the sum total of separate weighing results✔ Display angle adjustable to ﬁve levels to suit the convenience of individual operators
required for the last digit to settle (the stabilization mark will be lit one second after).
*4 TimeWith the HW-CEP series, response typically takes not more than 1.5 seconds.

Speciﬁcations
kglbozlb-oz
Counting mode

HV-15KCEP HV-60KCEP HV-200KCEP HW-10KCEP HW-60KCEP HW-100KCEP HW-200KCEP
3 / 6 / 15*i

Capacity

15 / 30 / 60*i 60 / 150 / 220*i

60

100

220

0.001

0.005

0.010.02

20

150

200

500

i

0.002

0.01

0.020.05

320

2400

3200

8000

0.05

0.2

0.5

10.001 / 0.002 / 0.005* 0.005 / 0.01 / 0.02*

Capacity

6 / 15 / 30*i

Readability

0.002 / 0.005 / 0.01* 0.01 / 0.02 / 0.05* 0.05 / 0.1 / 0.2*

Capacity

96 / 240 / 480* 480 / 960 / 2400* 2400 / 4800 / 8000*

Readability

0.05 / 0.1 / 0.2*

Capacity

30 lb

Readability

0.1 ozMax. countable number

150,000i

0.02 / 0.05 / 0.1*

10Readability

ii

30 / 60 / 150*i 150 / 300 / 500*i
i

i

i

i

i

0.2 / 0.5 / 1*

1 / 2 / 5*

ii

i

20 lb

N/A
120,000

100,000

110,000

Number of samples

N/A

0.1 oz120,000

100,000

110,000

5, 10, 20, 50 or 100

Stabilization time* ii

Within 1.5 seconds

Within 1 second

Repeatability (std. deviation) [kg]0.001 / 0.002 / 0.005 0.005 / 0.01 / 0.02 0.02 / 0.05 / 0.1

0.002

0.010.02

0.04

Linearity [kg]±0.001 / ±0.002 / ±0.005 ±0.005 / ±0.01 / ±0.02 ±0.02 / ±0.05 / ±0.1

±0.002

±0.01

±0.02

±0.04

Sensitivity drift (typical)±20 ppm/˚C (5 to 35 ˚C / 41 to 95 ˚F)Display

7 segment LCD (Character height: 26 mm) with backlight
3-color, 5-step LED comparator lights

Display refresh rate

10 times / second

Operating environment– 10 to 40 ˚C / 14 to 104 ˚F, 85% RH or less (no condensation)
Four D-size, alkaline, 1.5 V batteries *iii

Power supply
Battery life (approx.)

1,500 hours with LED and backlight OFF or 1,000 hours with LED and backlight ON

Weighing pan size
Dimensions (W × D × H)
Net weight (approx.)

250 × 250 mm 330 × 424 mm 390 × 530 mm 250 × 250 mm 330 × 424 mm

390 × 530 mm

9.8 × 9.8 in

15.4 × 20.9 in

13.0 × 16.7 in 15.4 × 20.9 in

9.8 × 9.8 in

13.0 × 16.7 in

250 × 439 × 397 mm 330 × 585 × 776 mm 390 × 691 × 776 mm 250 × 439 × 397 mm 330 × 585 × 776 mm

390 × 691 × 776 mm

9.8 × 17.3 × 15.6 in 13.0 × 23.0 × 30.6 in 15.4 × 27.2 × 30.6 in 9.8 × 17.3 × 15.6 in 13.0 × 23.0 × 30.6 in

15.4 × 27.2 × 30.6 in

8 kg / 17.6 lb 12 kg / 26.5 lb 18 kg / 39.7 lb 8 kg / 17.6 lb 12 kg / 26.5 lb

18 kg / 39.7 lb

The lighter the target object, the ﬁner the readability.
*iii Multi-interval:
Time required for the last digit to settle (the stabilization mark will be lit one second after), which is subject* to the weighing environment, response characteristic setting and how the target object is placed.
*iii Only DURACELL MN1300, ENERGIZER E95, or Panasonic LR20 (XJ) can be used (but not mixed).

IS
Counting
Function

Intrinsic
SafetyComparator
Function

IP65

Platform
only

A&D Company, Ltd.

3-23-14 Higashi-Ikebukuro, Toshima-Ku, Tokyo, 170-0013, Japan Tel: +81 3-5391-6132 Fax: +81 3-5391-1566 www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131, U.S.A. Tel: +1 408-263-5333 Fax: +1 408-263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031, Australia Tel: +61 8-8301-8100 Fax: +61 8-8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire, OX14 1DY, United Kingdom Tel: +44 1235-550420 Fax: +44 1235-550485<German Sales Office>Hamburger Straße 30, D-22926, Ahrensburg, Germany Tel: +49 4102-459230 Fax: +49 4102-459231

A&D Korea Ltd.

8F Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331, Korea Tel: +82 2-780-4101 Fax: +82 2-782-4280

A&D Rus Co., Ltd.

Vereyskaya Str. 17, 121357, Moscow, Russia Tel: +7 495-937-33-44 Fax: +7 495-937-55-66

A&D Instruments India (P) Ltd.

509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India Tel: +91 (124) 471-5555 Fax: +91 (124) 471-5599* HVW-CEP-ADCC-02-PR2-19501GP