Digital Blood Pressure Monitor
Model UA-651BLE
Instruction Manual
Original

Manuel d’instructions
Traduction

Manual de Instrucciones
Traducción

Manuale di Istruzioni
Traduzione

BedienungsanleitungÜbersetzung

使用手冊
翻譯

1WMPD4003018C

Contents
Dear Customers ....................................................................... 2
Preliminary Remarks ................................................................. 2
Precautions .............................................................................. 2
Parts Identification ................................................................... 5
Symbols ................................................................................... 6
Using the Monitor..................................................................... 8
Installing / Changing the Batteries .......................................8
Connecting the Air Hose ...................................................... 8Connecting the AC Adapter .................................................. 8
Wireless Function ................................................................. 9Bluetooth® Transmission ..................................................... 9
Pairing ............................................................................... 10
Selecting the Correct Cuff Size ...........................................12
Applying the Arm Cuff ....................................................... 13
How to Take Accurate Measurements ................................... 13
Measurement ..................................................................... 13
After Measurement ............................................................ 13
Measurements ........................................................................ 14
Normal Measurement ......................................................... 14
Measurement with the Desired Systolic Pressure ................15
Notes for Accurate Measurement ....................................... 15
What Is The IHB/AFib Indicator? .............................................. 16
What Is The AFib? ................................................................... 16
Pressure Bar Indicator ............................................................. 16
About Blood Pressure ............................................................. 17
What is Blood Pressure? ..................................................... 17
What is Hypertension and How is it Controlled? ..................17
Why Measure Blood Pressure at Home?...............................17
WHO Blood Pressure Classification .....................................17
Blood Pressure Variations................................................... 17
Troubleshooting ..................................................................... 18
Maintenance ........................................................................... 19
Technical Data........................................................................ 19

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D blood pressure
monitor, one of the most advanced monitors available today. Designed
for ease of use and accuracy, this monitor will facilitate your daily bloodpressure regimen.
We recommend that you read through this manual carefully before
using the device for the first time.
The Bluetooth® word mark and logos are registered trademarks owned byBluetooth SIG, Inc. and any use of such marks by A&D is under license.
Other trademarks and trade names are those of their respective owners.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
Products. This is made evident by themark of conformity.
(0123: The reference number to the involved notified body)
Hereby, A&D Company, Limited declares that the radio equipment typeUA-651BLE is in compliance with Directive 2014/53/EU. The full text of
the EU declaration is available at the following internet address:
http://www.aandd.jp/products/manual/manual_medical.html
The device is a Continua certified, Bluetooth® wireless technology
enabled medical device.
The device is designed for use on adults, not newborns or infants.
Environment for use. The device is for use to operate by yourself in thehome healthcare environment.
This device is designed to measure blood pressure and pulse rate of people
for diagnosis.

Precautions
Precision components are used in the construction of this device.
Extremes in temperature, humidity, direct sunlight, shock or dust should
be avoided.
Clean the device and cuff with a dry, soft cloth or a cloth dampened withwater and a neutral detergent. Never use alcohol, benzene, thinner orother harsh chemicals to clean the device or cuff.
Avoid tightly folding the cuff or storing the hose tightly twisted for long
periods, as such treatment may shorten the life of the components.
Take care to avoid accidental strangulation of babies or infants with the hose.
Do not twist the air hose during measurement. This may cause injury
due to continuous cuff pressure.
The device and cuff are not water resistant. Prevent rain, sweat andwater from soiling the device and cuff.
Measurements may be distorted if the device is used close to televisions,microwave ovens, cellular telephones, X-ray or other devices with strong
electrical fields.
English 2

Wireless communication devices, such as home networking devices,mobile phones, cordless phones and their base stations, walkie-talkies
can affect this blood pressure monitor. Therefore, a minimum distance of
30 cm should be kept from such devices.
When reusing the device, confirm that the device is clean.
Used equipment, parts and batteries are not treated as ordinary
household waste, and must be disposed of according to the applicable
local regulations.
When the AC adapter is used, make sure that the AC adapter can bereadily removed from the electrical outlet when necessary.
Do not modify the device. It may cause accidents or damage to the device.
To measure blood pressure, the arm must be squeezed by the cuff hardenough to temporarily stop blood flow through the artery. This may cause
pain, numbness or a temporary red mark to the arm. This condition willappear especially when measurement is repeated successively.
Any pain, numbness, or red marks will disappear with time.
Measuring blood pressure too frequently may cause harm due to bloodflow interference. Check that the operation of the device does not resultin prolonged impairment of blood circulation, when using the device
repeatedly.
If you have had a mastectomy, please consult a doctor before using thedevice.
Do not let children use the device by themselves and do not use the device
in a place within the reach of infants. It may cause accidents or damage.
There are small parts that may cause a choking hazard if swallowed by
mistake by infants.
Unplug the AC adapter when not in use during the measurement.
Use of accessories not detailed in this manual may compromise safety.
Should the battery short-circuit, it may become hot and potentially causeburns.
Allow the device to adapt to the surrounding environment before use (about
one hour).
Clinical testing has not been conducted on newborn infants and pregnant
woman. Do not use on newborn infants or pregnant woman.
Do not touch the batteries, the DC jack, and the patient at the same time.
That may result in electrical shock.
Do not inflate without wrapping the cuff around the upper arm.

Contraindications
The following are precautions for proper use of the device.
Do not apply the cuff on an arm with another medical electricalequipment attached. The equipment may not function properly.
People who have a severe circulatory deficit in the arm must consult adoctor before using the device, to avoid medical problems.

English 3

Do not self-diagnose the measurement results and start treatment by yourself.
Always consult your doctor for evaluation of the results and treatment.
Do not apply the cuff on an arm with an unhealed wound.
Do not apply the cuff on an arm receiving an intravenous drip or bloodtransfusion. It may cause injury or accidents.
Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, such as ahigh-pressure oxygen chamber or an oxygen tent. It may cause a fire or explosion.

English 4

Parts Identification
Proper Fit Range
Index
Artery Position MarkAir Hose

DC Jack

Arm Cuff
AC Adapter Plug
Display
START Button
Air Socket
Air Connector Plug
Battery Compartment
Battery Cover

1.5V Batteries(LR6, R6P or AA)

Display

Systolic Pressure

Diastolic Pressure

Pressure Bar Indicator

Pulse Rate

Heart Mark
Communication Mark

Battery Indicator
IHB/AFib symbol
English 5

Symbols
Symbols that are printed on the device caseSymbols

Function / MeaningStandby and Turn the device on.

SYS

Systolic blood pressure in mmHg

DIA

Diastolic blood pressure in mmHg

PUL

Pulse per minuteBattery installation guide
Direct current

IP
SN
BT

Type BF: Device, cuff and tubing are designed to provide
special protection against electrical shocks.
EC directive medical device label
EU-representative
Manufacturer
Date of manufactureInternational protection symbol
WEEE label
Serial number
Bluetooth address
Refer to instruction manual/booklet
Polarity of DC jackTo indicate generally elevated, potentially hazardous, levels of
non-ionizing radiation, or to indicate equipment or systems e.g.
in the medical electrical area that include RF transmitters orthat intentionally apply RF electromagnetic energy for diagnosis
or treatment.
Keep drySymbols that appear on the display
Symbols

Function / Meaning
Appears while measurement is inprogress. It blinks when the pulse isdetected.

English 6

Recommended Action
Measurement is inprogress. Remain as
still as possible.

Symbols

Function / Meaning
Recommended Action
IHB/AFib symbol
Appears when an irregular heartbeat
is detected.
It may light when a very slight vibration
like shivering or shaking is detected.
andare illuminatedalternately during communication.
FULL BATTERY
The battery power indicator during
measurement.
Replace all batteries
LOW BATTERY
with new ones when
The battery is low when it blinks.
the mark blinks.
Take anothermeasurement.
Unstable blood pressure due tomovement during measurement.
Remain very still during
measurement.
The systolic and diastolic values arewithin 10 mmHg of each other.
The pressure value did not increase
Apply the cuff correctly,
during the inflation.
and take anothermeasurement.
The cuff is not applied correctly.
PUL DISPLAY ERROR
The pulse is not detected correctly.

Blood pressure monitor internal error

Pairing has not been performed
correctly.

Pairng in progress.

Pairng complete.

English 7

Remove the batteries
and press the START
button, and then install
the batteries again. If
the error still appears,
contact the dealer.
Remove and reinstall thebatteries. Try pairing again.

Using the Monitor
Step 2

Installing / Changing the Batteries
1.
2.

3.

Remove the battery cover.
Remove the used batteries and insert
new batteries into the battery
compartment as shown, taking care thatthe polarities (+) and (-) are correct.
Step 2
Replace the battery cover.
Use only LR6, R6P or AA batteries.

Step 1
Step 3

CAUTION
Insert the batteries as shown in the battery compartment. If installed
incorrectly, the device will not work.
When(LOW BATTERY mark) blinks in the display, replace all batteries
with new ones. Do not mix old and new batteries. It may shorten thebattery life, or cause the device to malfunction.
Replace the batteries two seconds or more after the device turns off.
If
(LOW BATTERY mark) appears even after the batteries are replaced,make a blood pressure measurement. The device may then recognize
the new batteries.
(LOW BATTERY mark) does not appear when the batteries are drained.
The battery life varies with the ambient temperature and may be shorterat low temperatures. Generally, four new LR6 batteries will lastapproximately for one year when used twice for measurement each day.
Use the specified batteries only. The batteries provided with the device
are for testing monitor performance and may have a limited life.
Remove the batteries if the device is not to be used for a long time.
The batteries may leak and cause a malfunction.

Connecting the Air HoseAir Hose

Insert the air connector plug
into the air socket firmly.

Air Connector Plug
Air Socket

Connecting the AC Adapter

Insert the AC adapter plug into the
DC jack. Next, connect the AC
adapter to an electrical outlet.

AC adapter plugDC jack

Use the specified AC adapter.
(Refer to page 20.)
When disconnecting the AC adapter
from the electrical outlet, grasp and pull
the AC adapter body out of the outlet.
When disconnecting the AC adapter plug from the blood pressure
monitor, grasp and pull the AC adapter plug out of the monitor.
English 8

Using the Monitor
Wireless Function
CAUTION
In the unlikely event that this monitor causes radio wave interference to adifferent wireless station, change the location where this monitor is used orstop use immediately.
Be sure to use in a location where visibility between the two devices that you
want to connect is good. The connection distance is reduced by the structureof buildings or other obstructions. In particular, connection may be
impossible when devices are used on either side of reinforced concrete.
Do not use Bluetooth® connection in the range of a wireless LAN or otherwireless devices, near devices that emit radio waves such as microwaves, inlocations where there are many obstructions, or in other locations where
signal strength is weak. Doing so may result in frequent loss of connection,very slow communication speeds and errors.
Using close to an IEEE802.11g/b/n wireless LAN device may cause mutualinterference to occur, which may result in reduced communication speeds or
which may prevent connection.
In this case, switch off the power supply to the device that is not being used,
or use the monitor in a different location.
If the monitor does not connect normally when used near a wireless stationor broadcast station, use the monitor in a different location.
A&D Company, Limited cannot accept liability for any damages incurred due toimpaired operation or data loss, etc. that occur through the use of this product.
This product is not guaranteed to connect to all Bluetooth® compatible devices.

Bluetooth® Transmission

This product is equipped with a Bluetooth® wireless function and canconnect to the following Bluetooth® devices.
 Continua certified devices
 iPhone, iPad, iPod (iPhone 4S or later)
 Applications and devices that are compatible with Bluetooth 4.0.

Each device needs an application to receive data.
For connection methods, refer to the manual for each device.
Bluetooth® devices carry the Bluetooth® logo
mark.
Continua certified devices carry the Continua logo
mark.
To connect with your mobile device – download and install the “A&D
Connect” app:
Follow the instructions in the app to connect.
English 9

Using the Monitor
Pairing

A Bluetooth® device needs to be paired with a different specific device in order
to communicate with that device. If this monitor is paired with a receiver device
from the start, measurement data is transmitted automatically to the receiverdevice each time a measurement is made.

Cautions for pairing
Only one device can be paired with this monitor at one time. If the receiverdevice cannot receive measurement data, try pairing again
If another receiver device is paired, the first device will be unpaired toenable the new device to be paired.
Follow the steps below to pair the monitor with a Bluetooth® compatible
receiver device. Also refer to the manual of the receiver device. Please use
a pairing wizard if one is provided.
Pairing procedure
1.

Follow the instructions in the manual of the receiver device to switch it to thepairable status. When pairing this monitor, place it as close as possible to
the receiver device to be paired with.

2.

Install the batteries or connect the AC adapter as
described on page 8.
Press and hold the START button until “pr” isdisplayed, and then release the button.
The monitor will be in a state that can be found by thereceiver device for about one minute.

3.

The monitor displays “end” to indicate that pairing iscomplete.

4.

If “err10” is displayed or pairing is failed, remove thebatteries or disconnect the AC adapter and try steps
1-3 again.

5.

Follow the manual of the pairing receiver device to search for, select andpair with this monitor.

English 10

Using the Monitor
Communication distanceThe communication distance between this monitor and the receiver device isabout 10 m.
This distance is reduced by the conditions in the surrounding environment,so be sure to check that the distance is short enough for a connection to bemade after measurement is complete.
Transmitting temporarily stored data
In cases when the receiver device cannot receive measurement data, themeasurement data is temporarily stored in the monitor memory. The datastored in the memory is transmitted the next time a connection is
successfully made to the receiver device
A total of 30 sets of measurement data can be stored. When the amount ofdata exceeds 30 sets, the oldest data is deleted and the new data is stored.
The amount of data that can be stored temporarily may vary with theapplication.
Time
This monitor has a built-in clock. The date and time that a measurement was
taken is included in the measurement data.
The built-in clock is designed to be automatically adjusted by syncing withthe clock of a receiver device. Refer to the specifications of the receiverdevice.
This monitor has no clock adjustment function.

English 11

Using the Monitor
Selecting the Correct Cuff Size

Using the correct cuff size is important for an accurate reading. If the cuff is not
the proper size, the reading may yield an incorrect blood pressure value.
The arm size is printed on each cuff.
The index
and proper fit range, on the cuff, tell you if you are applying thecorrect cuff. (Refer to "Symbols that are printed on the cuff".)
If the index
points outside of the range, contact your local dealer to purchase a
replacement cuff.
The arm cuff is a consumable. If it becomes worn, purchase a new one.
Arm Size
Recommended Cuff Size
31 cm to 45 cm
Large adult cuff
22 cm to 32 cm
Adult cuff
Arm size: The circumference of the biceps.

Catalog Number
CUF-F-LA
CUF-F-A

Note: Model UA-651BLE is not designed for using a small adult cuff.
Symbols that are printed on the cuff
Symbols
Function/Meaning

Recommended Action
mark on the artery of
Set thethe upper arm or in line with thering finger on the inside of the arm.

Artery position mark
▲
REF

Index
Catalog number
Proper fit range for the adult cuff.
It's printed on the adult cuff.
Range to use the large adult cuff.
Over range printed on the adult cuff.
Proper fit range for the large adult cuff.
It's printed on the large adult cuff.
Under range printed on the adult cuff.
Range to use the adult cuff.
It's printed on the large adult cuff.

Use the large adult cuff instead of
the adult cuff.

Use the adult cuff instead of thelarge adult cuff.

Lot number
Proper fit range

Large adult cuff
Proper fit range

Adult cuff

English 12

Using the Monitor
Applying the Arm Cuff
1. Wrap the cuff around the upper arm,
about 1-2 cm above the inside ofthe elbow, as shown.
Place the cuff directly against theskin, as clothing may cause a faintpulse, and result in a measurement
error.

Do not roll up shirtsleeve tightly
Cuff
1 to 2 cm

Artery position mark2. Constriction of the upper arm,
caused by tightly rolling up a
shirtsleeve, may prevent accurate
readings.
3. Confirm that the index
within the proper fit range.

pointsSecure the fabricfastener firmly
Air hose
Index

and Proper fit range

How to Take Accurate Measurements
For the most accurate blood pressure measurement:
Sit comfortably on a chair. Rest your arm on the table. Do not cross yourlegs. Keep your feet on the floor and straighten your back.
Relax for about five to ten minutes before measurement.
Place the center of the cuff at the same height as your heart.
Remain still and keep quiet during measurement.
Do not measure immediately after physical exercise or a bath. Rest fortwenty or thirty minutes before taking the measurement.
Try to measure your blood pressure at the same time every day.

Measurement
During measurement, it is normal for the cuff to feel very tight. (Do not be
alarmed)After Measurement
After measurement, press the START
Remove the cuff and record your data.

button to turn off the power.

Note: The device has an automatic power shut-off function, which turns thepower off approximately one minute after measurement.
Allow at least three minutes between measurements on the same person.

English 13

Measurements
Before measurement, read “Notes for Accurate Measurement” on the next page.

Normal Measurement
1.

Place the cuff on the arm (preferably the left arm).
Sit quietly during measurement.

2.

Press the START button.
All of the display segments are displayed.
Zero is displayed blinking briefly. Then thedisplay changes, as indicated in the figure
at the right, as the measurement begins.
The cuff starts to inflate. It is normal for the
cuff to feel very tight. A pressure bar
indicator is displayed, as in the figure at
the right, during inflation.
Note: If you wish to stop inflation at any time,press the START button again.

3.

When inflation is complete, deflation starts
automatically and the(heart mark)blinks, indicating that the measurement isin progress. Once the pulse is detected,
the mark blinks with each pulse beat.
Note: If an appropriate pressure is not
obtained, the device starts to inflateagain automatically.

4.

5.

When the measurement is complete, thesystolic and diastolic pressure readings andpulse rate are displayed. The cuff exhausts
the remaining air and deflates completely.
Press the START
off the power.

button again to turnAt heart level
START button

All of the display
segments

Zero display

Starts inflation

Pressurizing

Measurementin progress
Systolic pressure
Diastolic pressure
Pulse rate
Exhausts remaining
air automaticallyNote: Model UA-651BLE is provided with
an automatic power shut-off function.
Allow at least three minutes between
measurements on the same person.
English 14

Measurements
Measurement with the Desired Systolic Pressure
Model UA-651BLE is designed to detect the pulse and to inflate the cuff toa systolic pressure level automatically.
If re-inflation occurs repeatedly or your systolic blood pressure is
expected to exceed 230 mmHg, use the following method.
1.

Place the cuff on the arm (preferably the left arm).

2.

Press the START button.

3.

While the zero blinks, press and holdthe START button until a number
about 30 to 40 mmHg higher than yourexpected systolic pressure appears.

4.

When the desired number is reached,release the START button to startmeasurement. Then continue tomeasure your blood pressure as
described on the previous page.

At heart level
Press

Press and hold
Inflation in progress
ReleaseDesired pressure
Refer to the previous pagefor measurement

Notes for Accurate Measurement
Sit down in a comfortable position. Place your arm on a table with yourpalm facing upward and the cuff at the same level as your heart.
Relax for about five or ten minutes before taking a measurement. If you
are excited or depressed by emotional stress, the measurement willreflect this stress as a higher (or lower) than normal blood pressure
reading and the pulse reading will usually be faster than normal.
An individual's blood pressure varies constantly, depending on what you
are doing and what you have eaten. What you drink can have a verystrong and rapid effect on your blood pressure.
This device bases its measurements on the heartbeat. If you have a very
weak or irregular heartbeat, the device may have difficulty determining
your blood pressure.
Should the device detect a condition that is abnormal, it will stop themeasurement and display an error symbol. See page 7 for thedescription of the symbols.
This blood pressure monitor is intended for use by adults. Consult withyour physician before using this device on a child. A child should not use
this device unattended.
English 15

The automatic blood pressure monitor’s performance may be affected byexcessive temperature or humidity, or altitude.

What Is The IHB/AFib Indicator?
When the monitor detects an irregular rhythm during the measurements,
the IHB/AFib indicator will appear on the display with the measurement
values.
Note: We recommend contacting your physician if you see thisIHB/AFib indicator frequently.

What Is The AFib?
The heart contracts due to electrical signals occurring in heart and sends
blood through the body. Arterial fibrillation (AFib) occurs when the electricalsignal in the atrium becomes confused and leads to disturbances in thepulse interval. AFib can cause blood to stagnate in the heart, which caneasily create clots of blood, a cause of stroke and heart attack.

Pressure Bar Indicator
The indicator monitors the progress of pressure during measurement.
Pressurizing

Inflation in progress

Releasing air

Measurement in progress
Inflation complete

English 16

About Blood Pressure
What is Blood Pressure?
Blood pressure is the force exerted by blood against the walls of the arteries.
Systolic pressure occurs when the heart contracts. Diastolic pressure occurs
when the heart expands. Blood pressure is measured in millimeters of
mercury (mmHg). One's natural blood pressure is represented by thefundamental pressure, which is measured first thing in the morning while one
is still at rest and before eating.

What is Hypertension and How is it Controlled?
Hypertension, an abnormally high arterial blood pressure, if left unattended,
can cause many health problems including stroke and heart attack.
Hypertension can be controlled by altering lifestyle, avoiding stress, and withmedication under a doctor’s supervision.
To prevent hypertension or keep it under control:
Do not smoke
Exercise regularly
Reduce salt and fat intake
Have regular physical checkups
Maintain proper weight

Why Measure Blood Pressure at Home?
Blood pressure measured at a clinic or doctor's office may cause
apprehension and can produce an elevated reading, 25 to 30 mmHg higher
than that measured at home. Home measurement reduces the effects ofoutside influences on blood pressure readings, supplements the doctor's
readings and provides a more accurate, complete blood pressure history.

WHO Blood Pressure Classification

Blood Pressure Variations

Diastolic blood pressureStandards to assess high Reference Material: Journal of Hypertension 1999, Vol 17 No.2
bloodpressure,
without mmHg
Grade 3 hypertension (severe)
110
regard to age, have been
Grade 2 hypertension (moderate)105
established by the World
100
Health Organization (WHO),
Grade 1 hypertension (mild)95
as shown in the chart.
90
85
80

High-normal
NormalOptimalAn individual’s blood pressure
120 130 140 150 160 170 180
varies greatly on a daily andSystolic blood pressure
mmHg
seasonal basis. It may vary by
30 to 50 mmHg due to various conditions during the day. In hypertensive
individuals’ variations are even more pronounced. Normally, the bloodpressure rises while at work or play and falls to its lowest levels during sleep.
So, do not be overly concerned by the results of one measurement.
Take measurements at the same time every day using the procedure

English 17

Typical fluctuation within a day(Measured every five minutes)
Systolic
Blood pressure

described in this manual to get to know yournormal blood pressure. Regular readings givea more comprehensive blood pressure history.
Be sure to note date and time when recording
your blood pressure. Consult your doctor tointerpret your blood pressure data.

Sleep

Diastolic
PM

AM

Time

Troubleshooting
ProblemPossible Reason
Recommended Action
Replace all batteries with new ones.
Nothing appears Batteries are drained.
in the display,
Reinstall the batteries witheven when theBattery terminals are not negative and positive terminals
power is turned in the correct position.
matching those indicated on theon.
battery compartment.
Battery power is low.
(LOW BATTERYmark) blinks.
The cuff does
Replace all batteries with new
not inflate.
If the batteries areones.
drained completely, themark does not appear.
The cuff is not applied
Apply the cuff correctly.
properly.
You moved your arm
Make sure you remain very stillor body duringand quiet during measurement.
The device does measurement.
not measure.
Sit comfortably and still.
Readings are
The cuff position is not
Place your arm on a table with yourtoo high or too
correct.
palm facing upward and the cuff atlow.
the same level as your heart.
If you have a very weak or
irregular heat beat, the device may
have difficulty in determining yourblood pressure.
The value is different
See “Why measure bloodfrom that measured at apressure at home?”
clinic or doctor’s office.
Other
Remove the batteries. Place themback properly and try themeasurement again.
Note: If the actions described above do not solve the problem, contact the
dealer. Do not attempt to open or repair this product, as any attempt to do
so will make your warranty invalid.
English 18

MaintenanceDo not open the device. It uses delicate electrical components and an
intricate air unit that could be damaged. If you cannot fix the problem
using the troubleshooting instructions, request service from your dealer
or from the A&D service group. The A&D service group will provide
technical information, spare parts and units to authorized dealers.
The device was designed and manufactured for a long service life.
However it is generally recommended to have the device inspected
every 2 years, to ensure proper functioning and accuracy. Please contact
the authorized dealer in your area or A&D for maintenance.

Technical DataType
Measurement method
Measurement range

UA-651BLE
Oscillometric measurement
Pressure: 0 - 299 mmHg
Systolic pressure: 60 - 279 mmHg
Diastolic pressure: 40 - 200 mmHg
Pulse:
40 - 180 beats / minute
Measurement accuracy
Pressure: ±3 mmHg
Pulse:±5%
Power supply
4 x 1.5V batteries (LR6, R6P or AA) or
AC adapter (TB-233C) (Not included)
Number of measurements
Approx. 700 times LR6 (alkaline batteries)
Approx. 200 times R6P (manganese batteries)
With pressure value of 180 mmHg at room
temperature of 23 °C.
Classification
Internally powered ME equipment (Supplied bybatteries) /Class II (Supplied by adapter)
Continuous operation mode
Clinical test
According to ISO81060-2：2013
EMD
IEC 60601-1-2: 2014
Wireless communication VZ (MURATA Manufacturing Co. Ltd.)
Bluetooth Ver.4.0LE BLP
Frequency band: 2402 MHz to 2480 MHz
Maximum RF output power: 1.6 dBm
Modulation:GFSK
Operating conditions+10 to +40 °C / 15 to 85 %RH / 800 to 1060 hPa
Transport / Storage conditions -20 to +60 °C / 10 to 95 %RH / 700 to 1060 hPa
Dimensions
Approx. 96 [W] x 68 [H] x 130 [D] mm
WeightApprox. 250 g, excluding batteries
English 19

Applied part
Useful life
Ingress protection
Accessory AC adapter
TB-233C

Cuff Type BF
Device: 5 years (when used six times a day)Cuff: 2 years (when used six times a day)AC adapter: 5 years (when used six times a day)Device: IP20
The adapter is to connect the blood pressure
monitor to a power source at home.
Please contact your local A&D dealer forpurchasing.
The AC adapter is required to be inspected or
replaced periodically.
Symbols that are printed on the AC adapter
Symbols
Function / Meaning
For indoor use only
Class II device
Thermal fuse
Fuse
EC directive device label
EAC certification device label
Polarity of AC adapter plug

Accessories sold separately
Cuff
Arm Size
Catalog Number
Cuff Size
CUF-F-LA
Large adult cuff 31 cm to 45 cm
CUF-F-A
Adult cuff
22 cm to 32 cm
AC adapter
Catalog Number
Plug
TB-233C
Type C
Note: Specifications are subject to change for improvement without prior notice.
IP classification is the degrees of protection provided by enclosures inaccordance with IEC 60529. This device is protected against solid
foreign objects of 12 mm diameter and greater such as a fingers. This
device is not protected against water.

English 20

A&D Company, Ltd.
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111 Fax: [81] (48) 593-1119
Emergo Europe B.V.
Prinsessegracht 20, 2514 AP The Hague, The Netherlands
Tel: [31] (70) 345-8570
Fax: [31] (70) 346-7299
A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire OX14 1DY
United Kingdom
Telephone: [44] (1235) 550420 Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131,
Telephone: [1] (408) 263-5333 Fax: [1] (408)263-0119

U.S.A.

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409
OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd 爱安德技研贸易（上海）有限公司
中国 上海市浦东新区浦东大道138号永华大厦21楼A室 邮编200120( 21F Room A, Majesty Building, No.138 Pudong Avenue, Pudong New Area, Shanghai, 200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599