Mass Comparators

MC-1000

MCseries

1100 g × 0.0001 g

MC-6100

6100 g × 0.001 g

MC-10K

10.1 kg × 0.001 g

MC-30K

31 kg × 0.01 g

MC-100KS
101 kg × 0.1 g

MC-6100

MC-10K

MC-100KS

ISO9000 CERTIFIED...Clearly a Better Value
http://www.aandd.jp

Revealed — Here Are What A&D Use
To Manage Their Masses!
It’s not easy to find high performance, quality mass comparators that meet yourbudget, is it? Discover how A&D, an ISO17025 accredited calibration laboratory
as well as being a weighing equipment manufacturer, can finally solve suchproblems for you —.

1. Enhanced stabilizing filter

High resolution balances like the MC series can respond to even the slightest environmental
disturbances. For smooth and accurate mass comparison, the MC series is equipped with a special
stabilizing function that copes with the effect of drafts or vibrations.

2. Simplified calibration of the balance

Using the internal mass, you can ensure the balance accuracy with one key press when changes tothe environment (such as temperature or location) take place. The internal mass value can becorrected as necessary.

3. Various options to minimize possible measurement errors
Even under a perfectly controlled weighing environment, there still remain errors that can arise
from operation. The following tools are effective in managing such errors:

Auto-centering pan (available for MC-1000/6100/10K/30K)

Corrects eccentric loading automatically by bringing the mass to the center.

Please also visit our video library at www.aandd.jp to see how it operates!

MC Series
with auto-centering pans

Remote controller(AD-8922A)

Static eliminator(AD-1683)

Enables operations without directly
touching and consequently shakingthe balance.

1293

Instantly neutralizes charged objects
such as the operator’s clothes to ensurestable weighing.

ONMemory

6Internal
Calibration

Super HybridSensor

VFD Display

RS-232C
Interface

GLP
Compliant

Clock & Calendar

Data Memory
Function

Counting with
ACAI

Percentage
Function

AnimalWeighing

Comparator
Function

Auto PowerON

Auto PowerOFF

Recommended Measurement Ranges

Nominal value

Model
BM-20
BM-252
BM-500MC-1000
MC-6100
MC-10K
MC-30K
MC-100KS
Class E2 F1 F2 M1 M2 F1 F2 M1 M2 F1 F2 M1 M2 F1 F2 M1 M2 F1 F2 M1 M2 F1 F2 M1 M2 F1 F2 M1 M2 M1 M2
1 mg

1 mg

2 mg

2 mg

5 mg

5 mg

10 mg

10 mg

20 mg

20 mg

50 mg

50 mg

100 mg

100 mg

200 mg

200 mg

500 mg

500 mg

1g

1g

2g

2g

5g

5g

10 g

10 g

20 g

20 g

50 g

50 g

100 g

100 g

200 g

200 g

500 g

500 g

1 kg

1 kg

2 kg

2 kg

5 kg

5 kg

10 kg

10 kg

20 kg

20 kg

50 kg

50 kg

100 kg100 kg* Please refer to the BM series of micro analytical balances for the BM-20, BM-252 and BM-500.

Wait, there is more! If you are in the industrial sector, how about using the MC series as

Precision Balances With Extended Resolution

The MC series displays the weighing results with an extra decimal place not normally foundin balances of the same capacity. Such extended resolution enables measurements of minuteweight changes with heavy objects or containers. Example applications include:

Management of Liquids

Management of Powders

Management of Gases

Production Line Weighing

Resist inks on flat panels (glass)
Coating materials on silicon wafers
Moisture contents of catalysts
Solder pastes and sealing materials onelectronic parts
Resins used for LED lenses
Surface materials on optical lenses
Grease in miniature bearings for hard disks

Medicines in pharmaceutical manufacturing –from full flow to minute flow
Discharges (losses in weight) from feeders
Wear volumes of metallic materials
Cutting dust (contaminants) left on mechanicalparts
Particles caught by filters

As precision check-weighing sensors withwider ranges

Fill ration of hydrogen gas
Gasified components

OptionalUnderhook(Optional forMC-100KS)

SoftwareAccumulation
Function

IP65

MC-10K/30K/100KS

Quick USB

MC-1000/6100

Specifications
MC-1000

MC-6100

MC-10K

MC-30K

MC-100KS

Weighing capacity

1100 g6100 g10.1 kg

31 kg

101 kgMinimum weighing value

0.0001 g

0.01 g

0.1 g

ModelsRepeatability (standard deviation) * 1

0.001 g0.0005 g (for 1 kg)

0.004 g (for 5 kg)

0.005 g (for 10 kg)

0.0004 g (for 500 g)

0.0015 g (for 2 kg)

0.004 g (for 5 kg)±0.003 g

Linearity

Accuracy immediately after internal calibration * 2±2 ppm / °C
±0.010 g±0.2 g±2 g±3 ppm / °C

±6 ppm / °C

±1.5 g±10.0 g±0.15 gOperating environment

5°C to 40°C (41 F to 104 F), 85%RH or less (no condensation)

Display refresh rate
Display mode * 3

0.1 g (for 60 kg)±0.03 gSensitivity drift (10 to 30°C / 50 to 86°F)

0.2 g (for 100 kg)

0.015 g (for 20 kg)

5 times/second or 10 times/second
Gram, kilogram (for MC-10K/30K/100KS), ounce, pound, pound-ounce, troy ounce, metric carat, momme, pennyweight, grain,
counting mode, percent mode, density mode, and a user-programmable unit

Standard interface
Weighing pan size

RS-232C
128 x 128 mm

165 x 165 mm

386 x 346 mm

270 x 210 mm

External dimensions

Display unit: 260 (W) x 164 (D) x 202 (H) mm
210 (W) x 317 (D) x 86 (H) mm

300 (W) x 355 (D) x 111 (H) mm

Platform: 346 (W) x 443 (D) x 130 (H) mm
Cable length: 3 m

Net weight

Approx. 4.6 kg

Approx. 5.1 kg

Power consumption

Approx. 9.3 kg

Approx. 18 kg
Approx. 15 VA (supplied to the AC adapter)

Approx. 11 VA (supplied to the AC adapter)* 1 Repeatability when the auto-centering pan or an automatic weight loader is used under good environmental conditions.
* 2 Accuracy immediately after internal calibration performed under good environmental conditions. The values are for the weighing capacities.
* 3 Either tael (Singapore/HK jewelry/Taiwan) or tola can be added upon request.

Options

Accessories

For MC-1000/6100

AD-1603-E2 * 9
AD-1603-F1 * 9
AD-1682
AD-1683
AD-1684
AD-1687
AD-1688
AD-1689
AD-8127
AD-8526
AD-8527
AD-8920A
AD-8922A
AX-MC1000PAN
AX-MC6100PAN
AX-MC10K/30KPAN
AX-SW128
AX-USB-25P

GX-02 * 4 Quick USB interface with cable
GX-04 * 4 * 5 Comparator output with a buzzer/RS-232C/Current loop output
GX-06 * 4 * 5 Analog output/Current loop output
Animal weighing pan
GX-12
Density determination kit (for MC-1000 only)
GX-13* 4 Only one of GX-02, GX-04, GX-06 or the standard RS-232C interface can be used.
* 5 Factory-installed options

For MC-10K/30K
GX-04K * 6
GX-06K * 6
GX-07K
GXK-012
GXK-015Comparator output with a buzzer/RS-232C/Current loop output
Analog output/Current loop output
Waterproof RS-232C cable (25P-9P, 5 m)
Animal weighing pan
Carrying case* 6 Only one of GX-04K, GX-06K or the standard RS-232C can be used.
The MC-10K/30K are not IP65 compliant with these options.

* 9 Selectable from 1 mg to 20 kg.

For MC-100KS
GP-04 * 7
GP-06 * 7
GP-07 * 8
GP-12
GP-21
GP-22

Calibration masses, OIML class E2
Calibration masses, OIML class F1
Rechargeable battery unit
Static eliminator
Electrostatic fieldmeter
Weighing environment logger
Weighing data logger
Tweezers for calibration mass
Compact printerRS-232C/Ethernet (TCP/IP) converter
Quick USB adapter
Remote display
Remote controller
Auto-centering pan (MC-1000)
Auto-centering pan (MC-6100)
Auto-centering pan (MC-10K/30K)
Foot switch
RS-232C/USB converter

Comparator output with a buzzer/RS-232C/Current loop output
Analog output/Current loop output
5-meter extension cable
Animal weighing panUnderhook
AD-8121B printer support* 7 Only one of GP-04, GP-06 or the standard RS-232C can be used.
* 8 Factory-installed option

Auto-centering pan...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www.aandd.jp
1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66

Hamburger Straße 30 D-22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599<German Sales Office>817, Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil,Yeongdeungpo-gu, Seoul, 07331, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280

A&D Instruments India Private Limited*MC Series-ADCC-01-PR4-22300