Compact Bench Scales

SJ-HSSeries

SJ-1000HS SJ-2000HS SJ-5000HS SJ-5001HS SJ-12KHS SJ-20KHS SJ-30KHS

ISO
9000 CERTIFIED

http://www.aandd.jp

Features■ Large, removable stainless steel (SUS304) weighingpan, which is easy to wash and keep clean■ High-speed weighing in less than 1 second toincrease your work efficiency■ Resolution up to 1/5000 (SJ-5001HS) for highly
precise measurements■ Clearly visible LCD (with 25 mm character height)
ensuring accurate reading of results■ Low-profile construction for storage in narrow shelves,
which also helps when lifting heavy objects to the weighing pan■ Tare function to exclude the weight of the container from the weighing result■ Long battery life of approx. 2000 hours (optional AC adapter is also available)●●

Auto power off function so as not to waste battery power
Battery indicator that tells you the remaining battery level■ Four weighing units (g, lb, oz, and lb-oz) that can be switchedwith a single key■ Two side handles allowing easy carrying of the scale■ Zero tracking function to maintain the display at zero

The battery box makes it quick
and easy to replace batteries.

before measurement
Specifications
SJ-1000HS

SJ-2000HS

SJ-5000HS

SJ-5001HS

SJ-12KHS

SJ-20KHS

g1000 g x 0.5 g

2000 g x 1 g

5000 g x 2 g

5000 g x 1 g

12 kg x 5 g

20 kg x 10 g

30 kg x 20 glb

2.2 lb x 0.001 lb

4.4 lb x 0.002 lb

11 lb x 0.005 lb

11 lb x 0.002 lb

26 lb x 0.01 lb

44 lb x 0.02 lb

66 lb x 0.05 lboz

35 oz x 0.02 oz

70 oz x 0.05 oz

176 oz x 0.1 oz

176 oz x 0.05 oz

423 oz x 0.2 oz

705 oz x 0.5 oz

1058 oz x 1 oz

lb-oz

2.2 lb x 0.1 oz

4.4 lb x 0.1 oz

11 lb x 0.1 oz

11 lb x 0.1 oz

26 lb x 1 oz

44 lb x 1 oz

66 lb x 1 oz

Repeatability (Std. deviation)

1g

2g

5g

5g

10 g

20 g

30 g

Linearity

±1 g

±2 g

±5 g

±5 g

±10 g

±20 g

±30 g

Capacity x Resolution±0.015% / °C (5 °C to 35 °C / 41 °F to 95 °F)

Sensitivity drift

7 segment LCD (character height: 25 mm / 0.98 in.)

Display

10 times per second

Display update–10 °C to 40 °C / 14 °F to 104 °F, 85%R.H. or less (no condensation)

Operating conditions
Power supply

Four D (R20P / LR20) batteries or AC adapter (optional)

Battery life

Approximately 2,000 hours using alkaline batteries at 20 °C / 68 °F
230 (W) x 190 (D) mm / 9.05 (W) x 7.48 (D) in.

Weighing pan size

265 (W) x 250 (D) x 106 (H) mm / 10.4 (W) x 9.8 (D) x 4.2 (H) in.

Dimensions

2.2 kg / 4.85 lb

Net weight1000 gCalibration massOption

SJ-30KHS

2000 g5000 g

2.5 kg / 5.51 lb
5000 g

10 kg

20 kg

30 kg

AC adapter
LCD
Display

External
Calibration

Auto PowerOFF

NormalBattery...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www.aandd.jp
1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

A&D Australasia
ia Ptyy Ltd.
d.
32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66<German Sales Office>509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599

Hamburger Straße 30 D-22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

A&D
D Instruments
Inst uments India Private
Pre Limited
mited

Manhattan Bldg. 8F, 36-2 Yoido-dong, Youngdeungpo-gu, Seoul, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280*SJ-HS-ADCC-03-PR3-12503