Surgical Solutions

Brightness.
Clarity. Efficiency.
G-Series Surgical Lighting System

One Integrated Approach to Healthcare

PURE, WHITE BRILLIANCE
Experience the clear difference in coolness, comfort and natural color rendition
inherent in pure, white LED light. The G-Series Surgical Lighting Systemdelivers it to you affordably without compromising quality.

Why White Light is Ideal in a Surgical Environment
The G-Series uses a single-color, phosphor-coated LED chip for crisp,
white light, without the color-fringing or color-shifting effects of multiplecolor LEDs. Natural color rendition allows surgeons to accurately andconsistently assess and interpret tissue and vessel appearance, helping
provide the best possible patient outcomes.

Bright, Wide Illuminationof the Surgical Spot
The G-Series delivers up to 160,000 lux over agreater portion of the spot or pattern size, making
the spot visibly brighter, with crisp, defined edges.
This more uniform beam and larger depth offield means no hot or cold spots, and less visual
distraction or eye fatigue.

Cool Under Pressure
The new LED array consumes less than one-third of theenergy of previous LED technology, providing a more
comfortable OR experience for the surgical staff. Less
heat also reduces the risk of drying out exposed tissue.

FLEXIBLE SUSPENSIONS
Simplified Controls
G-Series suspension systems are made to the highest quality standards. Their flexible nature
integrates with your workflow and case load, providing you with the ideal solution for yourunique challenges. The system’s open architecture and non-wired suspension arms give you
the flexibility to use any major integration or monitor provider.

Suspensions to Fit Any Room Configuration
Experience full-range maneuverability, single- or dual- cardanic arms, and fixed oradjustable spot size light heads, including camera-ready ones for recording cases.

Easy-to-Position and Stow Monitors
Specially engineered monitor yokes are future-ready,accommodating up to 32-inch monitors. Simply push up
and away to a stow position when not in use, avoiding heatdamage to the screen face.

Centralized, Integrated Controls
The sterile surgeon controls enable you to turn the lights
on and off, and adjust the light intensity, spot size andposition from within the sterile field. In addition, an intuitive
wall control enables complete command of lights and thein-light camera from a distance.

LIGHTS.
CAMERA.
OPTIONS.
See every image with stunning clarity andenhanced contrast with our HD in-light camera. Its
small size fits easily into the sterile handle for an unobstructed
view of the surgical site. Controlled from the wall-mounted control
panel or through the Harmony iQ® Integration System, the HD camera
provides superior visualization for reference, broadcasting and teaching.

The Harmony® ConnectPoint™ Video Connectivity System
improves safety and efficiency for your staff and patients. It is an ideal solution toquickly integrate your room and increase surgeon visualization.

Your Partner in Green
We’re committed to providing you with environmentally responsible solutions that
help you meet your goals, so that you can focus on what you do best – provide
the highest quality patient care.
Our Environmental Stewardship Initiative encompasses our products andsolutions, company operations, and social responsibility.

Less is More
G- Series lights use up to 70 percent
less energy per light head thanprevious LED surgical lights, all while
delivering more light to the surgicalfield.

Long Live LEDs
Longer-life LED bulbs last in excess of 50,000
hours. Further, LEDs are mercury-free,and do not contain CFCs, POPs, VOCs,halogens or other harmful chemicals.
They’re also RoHS-compliant.

Technical Specifications

FeatureG5 Series

Diameter

585 mm# of LED

60

Maximum Central Illuminance160 klx

Maximum Central Irradiance< 500 W/m1

Heat to Light Ratio x

3.1 mW/m2-lx

Depth of Field38” (97 cm) at 20%
21” (53 cm) at 60%

Pattern Size (D10)7” - 11” (18-28 cm)
7” (18cm) fixed

D50 (small pattern)

4.4” (11cm)

D50/D10 (small pattern)

62%

CCT (Color Temp)

4400 +/- 300 K

CRIup to 96

R9up to 98

Chromaticity Coordinates(0.369, 0.371)

UV emission (300-400 nm)

2.69 W/m2

Power Consumption (LEDs only)

40 watts

Life (B50, L70 definition)

50,000 hrs™

Surgical Lights
For more information, contact your STERIS representative
or visit www.steris.com.

Document # M4083EN.2016-04, Rev C.
© 2016 STERIS Corporation.
All rights reserved.

STERIS Corporation
5960 Heisley Road
Mentor, OH 44060-1834 USA
800-548-4873
www.steris.com•