Moisture Analyzers

MS-70/ MX-50
MF-50/ ML-50

Test with The BEST –Quick, Safe and Accuratewww.aandd.jp

A&D's Moisture Analyzers

MS-70
MX-50
MF-50
ML-50

Fast and uniform heating with halogen lamp and innovative SRA technology
Straight halogen lamp and uniquely designed SRA (Secondary Radiation Assist) ﬁlter gives shortermeasurement time, thanks to fast and uniform heatingHigh repeatability

With Super Hybrid Sensor (SHS) featured as the weight sensor, ultra accurate moisture content
determination is possible based on high precision weighing of even a small sample

Low moisture content measurement

MS-70 measures the moisture content at 0.001% resolution suitable for low moisture content samples aswell as Karl Fischer method, yet requires no special knowledge or training and produces no harmful waste

WinCT-Moisture (for MS&MX) for real-time graph displaying

WinCT-Moisture is an original software application designed to display a graph of moisture content
rate change while measuring with a connected PC

Sodium Tartrate Dihydrate comes as standard for accuracy checking

Sodium Tartrate Dihydrate is a chemical material that has stable moisture content of 15.66% +0.3
–0.1 ,and thus is best to use for accuracy check to maintain the reference value of the analyzer

Calibration of the heater temperature (for MS&MX)

With the temperature calibrator (optional), calibration result can be output in the format that
conforms to GLP, GMP, ISO

Memory function

According to sample up to 20 suitable measurement conditions can be stored and recalled, which saves time and prevents
the user from making a mistake when setting (10 for MF & 5 for ML)
For measurement result, up to 100 data can be stored and output at once (50 for MF & 30 for ML)

Five measurement programs

Five choices of measurement programs, Standard, Automatic, Quick, Timer, and Manual Mode are provided
Standard Mode : Just measurement accuracy, HI, MID or LO needs to be set
Automatic Mode : Ends measurement when moisture content changes at a rate less than the set rate
Quick Mode : Begins heating samples at 200°C for 3 minutes, then is the same as Automatic Mode
Timer Mode : Continues measurement for a set duration of time (1~60mins.: by 1min, 60~480mins.: by 5mins.)
Manual Mode : STOP button should be pressed to end measurement (Max. heating time: 480mins.)

Selectable Heating Mode

Choose the heating mode from standard, quick, step and ramp heating modes for the most suitable measurement(ML has Standard and Quick heating modes only)

Clear and easy-to-see, large VFD display

Measurement, setting value, change in moisture content, action status, data number and other important
information are clearly displayed

Easy handling of the unit

Ergonomically designed pan handle eliminates mishaps such as burns when moving a hot sample pan into or out of theunit from either side. Wing handle for easy opening and closing of the heater coverLow maintenance cost assured

The halogen lamp is user replaceable without unit downtime with protective chamber for easy cleaning(Lamp life 5000 hours)

Select the best moisture analyzer
for your application–Test with The BestMSMXMF

ML

Progress window for heating check

Heating process can be checked through translucent window

Self Check function

Defect check function is available along with temperature control

Quick Reference Card

A convenient operating guide is installed at the bottom of the analyzer

RS-232C Interface

Bi-directional communication with a PC or connects directly to a printerConformity to GLP, GMP, GCP and ISO with date/time, ID, calibration
data and check record outputs
Data output for daily record management

WinCT-MoistureWith our WinCT–Moisture software, data measured by the Moisture Analyzers
can be easily displayed on your computer.
Eﬀective for determining measurement conditions such as heating temperature,and useful in reducing the time needed for measurement and improving accuracy.

Moisture rate

Change in moisture rate

Displays moisture rate changes over time in a graph (RsFig)
Displays changes in moisture rate in real time

Measures moisture rate in a minimum time with excellent accuracy
Heats at the highest temperature without changing the physical properties of thesample and provides measurements with good repeatability

Automatically determines the most suitable heating conditions
in a short time (RsTemp)

Automatically changes the heat applied by the set increments and interval time withina range of 30°C-200°C. From the moisture rate change over that time can determine
the most suitable heating temperature in one time measurement (*Patent pending)Shows sample data summaryProvides a data summary for the sample with the results of moisture rate change forthe representative material’s measurement

Calculates measurement data statistics
Saves the recorded data as a CSV ﬁleDetermines other changes to the sample material in addition tomoisture rate

Continuously measures changes to the mass in response to heating temperature and detects other materialchanges besides the moisture rate

WinCT-Moisture
Measurement example

WinCT–Moisture consists of RsTemp software to determine the heating temperature and RsFig software for graphics.

1. Example of measurement using RsTemp software to determine the heating temperature
Automatically changes the heating temperature (by your setting increments and interval time) within a range of 30°C - 200°C. From the rateof change in moisture over time, in one time measurement it can determine the optimum heating temperature for the sample.

Soybean ﬂourSodium tartrate dihydrate

2. Example of measurement using RsFig graphic software

Can show moisture rate against time when the heating temperature is changed as well as showing the results of repeated measurements at acertain temperature. From the graph, the highest possible heating range for the sample and the quickest measuring time can be determined.

Corn starch

Sodium tartrate dihydrate

Comparison between MS-70 and the KF (Karl Fischer) method
Example of PET plastic pellet measurement
Moisture rate
Averagemeasurement
Average
Repeatability
timevalue(standard deviation)

Measurement conditions

MS-70

0.298%

0.0045%

6.8 mins

Heating temperature 180°C
Test sample
10g
Measurements
5 times

KF method

0.307%

0.0065%19.1 mins

Heating temperature 180°C
Test sample
0.3g
Measurements
5 times

KF method: a way of measuring moisture content through chemical determination.

With PET plastic and other materials, the MS-70 can measure a drop of moisture content of less than 1%.
Specialist knowledge is not necessary to operate the MS-70, and since measurement occurs quickly noharmful waste is produced.

Speciﬁcations

MS-70

Measurement method

MX-50

ML-50

400W straight halogen lamp heating system with SRA ﬁlterand SHS weighing technology

Max sample weight capacity

71g

Weight readability
Moisture content readability
Moisture content accuracy over 1g(standard deviation)over 5g

51g

0.0001g0.001g0.002g

0.005g0.001%/0.01%/0.1%

0.01%/0.1%

0.05%/0.1%/1%

0.1%/1%

0.05%

0.10%

0.20%

0.5%

0.02%

0.05%

0.1%

0.01%

Heating technology

Halogen lamp (Straight type, 400 Watt max, 5000 hours)

Drying temperature (1°C increment)

30-200°C

Memory of measurement programs

50-200°C
10 sets

20 sets

Measurement programs

5 setsStandard mode/Automatic mode/Quick mode/Timer mode/Manual mode

Measurement mode

Moisture content(Wet or Dry base) /Dry content/Ratio/Weight

Heating modeStandard/Quick/Step/Ramp

Display type

Standard/QuickLarge VFDStandard interface

RS-232C

Data memory capacity

50 data

100 data

Operating temperature

30 data

5-40ºC (41-104ºF) less than 85%RH

GLP/GMP/ISO compliant output

Available

Self check function

Standard

Standard communication software

WinCT

WinCT-MoistureSample pan size

Ø85mm

Power

AC 100V to 120V (3A) or AC 200V to 240V (1.5A), 50/60 Hz, Approx. 400W

External dimensions / Net weightStandard accessories

MF-50215(W) × 320(D) × 173(H) mm / Approx. 6kg
Sample pans (20 for MS/MX/MF 10 for ML), Pan handles (2 for MS/MX/MF, 1 for ML),
Tweezers (for MS/MX/MF), Spoon (for MS/MX/MF), Test sample (30g of sodium tartrate
dihydrate for MS/MX/MF), CD-ROM (WinCT-Moisture for MS/MX),
Glass ﬁber sheet (for MS/MX/MF), RS-232C cable (for MS/MX), Display cover,
Dust cover (for MS/MX/MF), Instruction manual, Quick reference card, Power cable, Fuse
Speciﬁcations are subject to change for improvement without notice.

Accessories
AD-8127
AX-MX-30
AX-MX-31
AX-MX-32-1
AX-MX-32-2
AX-MX-33

Compact printer
Disposable aluminum pan (100 pcs)
Sample pan (ø85mm × 100 pcs)
Glass ﬁber sheet (ø70mm × 100 sheets)Glass ﬁber sheet (ø78mm × 100 sheets)
Test sample (Sodium tartrate dihydrate,30g × 12 pcs)
AX-MX-34-120V Halogen lamp for AC100V to 120V
AX-MX-34-240V Halogen lamp for AC200V to 240V
AX-MX-35
Pan handle (2 pcs)

AX-MX-36
AX-MX-37
AX-MX-38
AX-MX-39
AX-MX-40
AX-MX-41
AX-MX-42

Tweezers (2 pcs)
Spoon (2 pcs)
AD-8127
Display cover (5 pcs)
Compact printerDust cover
RS-232C cable (2m, 25-9pins)
Calibration mass (20g, equivalent to OIML class F1)
WinCT-Moisture (CD-ROM, application softwarefor Windows)
AX-MX-43
Certiﬁed temperature calibrator (only for MS/MX)
AX-USB-25P Serial/USB converter with cable

A&D Company, Ltd.

3-23-14 Higashi-Ikebukuro, Toshima-Ku, Tokyo, 170-0013, Japan Tel: +81 3-5391-6132 Fax: +81 3-5391-1566 www.aandd.jp

A&D Engineering, Inc.

1756 Automation Parkway, San Jose, CA 95131, U.S.A. Tel: +1 408-263-5333 Fax: +1 408-263-0119

A&D Australasia Pty Ltd.

32 Dew Street, Thebarton, South Australia 5031, Australia Tel: +61 8-8301-8100 Fax: +61 8-8352-7409

A&D Instruments Ltd.

Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire, OX14 1DY, United Kingdom Tel: +44 1235-550420 Fax: +44 1235-550485<German Sales Office>Hamburger Straße 30, D-22926, Ahrensburg, Germany Tel: +49 4102-459230 Fax: +49 4102-459231

A&D Korea Ltd.

8F Manhattan Bldg., 33, Gukjegeumyung-ro 6-gil, Yeongdeungpo-gu, Seoul, 07331, Korea Tel: +82 2-780-4101 Fax: +82 2-782-4280

A&D Rus Co., Ltd.

Vereyskaya Str. 17, 121357, Moscow, Russia Tel: +7 495-937-33-44 Fax: +7 495-937-55-66A&D Instruments India (P) Ltd.

509 Udyog Vihar Phase V Gurgaon-122 016, Haryana, India Tel: +91 (124) 471-5555 Fax: +91 (124) 471-5599* MS/ML-ADCC-17-PR11-19501GP