1WMPD4001896

Calibration Instructions for theStandard Version of the SJ-H/HS series
When is to Calibration Required?

Calibration may be required when the SJ-H/HS series initially installed or if the scale is
moved a substantial distance.

Calibration switch

Calibration using a Calibration Weight
1. Entering calibration mode.

Press the [ON/OFF] key to turn the power off.

While pressing

both the [RE-ZERO] and [UNITS] keys, press the [ON/OFF] key.
ThenCalCal

will be displayed.

can also be displayed by removing the calibration switch

cover at the bottom of the device and pressing the calibration
switch while the device is in the Weighing Mode.

2. Zero calibration
Press the [RE-ZERO] key to calibrate zero.

Calibration switch coverCal0 will be displayed.

Wait for the Stable Indicator to be displayed, then press the [RE-ZERO] key.

Calf will be displayed after a few seconds. Press the [UNITS] key to perform only a
Zero Calibration.

The scale will then automatically return to the Weighing Mode.

3. Span calibration
When Calf is displayed, place the weight at the center of the platform. Wait for theStable Indicator to be displayed, then press the [RE-ZERO] key. The display will show

end and the scale will automatically return to the Weighing Mode.
See “SPECIFICATIONS” about the calibration weight.

Calibration by Gravity Compensation

If the acceleration of gravity at your location is not 9.798 m/s2 and you do not havecalibration weights, the scale can be calibrated by compensating for the acceleration ofgravity.

(Refer to “The Value of Gravity at Various Locations”.)

1. Setting a new acceleration value.
When Calis displayed, press the [UNITS] key and 9.798 will be displayed. Pressthe [RE-ZERO] key to increment the blinking digit and press the [UNITS] key to move theblinking digit.

2. Storing the value in the memory.
While pressing the [UNITS] key, press and hold the [RE-ZERO] key and release the[UNITS] key.

Then the display will show

end and return to Cal

off to finish the procedure.

1/2.

Turn the scale

SPECIFICATIONS
MODEL

SJ-1000H/HS

SJ-2000H/HS

SJ-5000H/HS

SJ-5001H/HS

SJ-12KH/HS(k)g1000 g x 0.5 g

2000 g x 1 g

5000 g x 2 g

5000 g x 1g

12 kg x 5glb

2.2 lb x 0.001 lb 4.4 lb x 0.002 lb 11 lb x 0.005 lb 11 lb x 0.005 lb

26 lb x 0.01 lboz

35 oz x 0.02 oz

70 oz x 0.05 oz

176 oz x 0.1 oz

176 oz x 0.1 oz

423 oz x 0.2 oz

lb-oz

2.2 lb x 0.1 oz

4.4 lb x 0.1 oz

11 lb x 0.1 oz

11 lb x 0.1 oz

26 lb x 1 oz

Calibration weight

1000g ± 0.1g

2000g ± 0.2g

5000g ± 0.5g

5000g ± 0.5g

10 kg ± 1g

CapacityxResolution

The Value of Gravity at Various Locations
Amsterdam
AthensAuckland NZ
BangkokBirmingham
Brussels
Buenos Aires
Calcutta
Cape Town
ChicagoCopenhagen
CyprusDjakarta
Frankfurt
Glasgow

9.813 m/s2 Havana
9.788 m/s2 Rio de Janeiro 9.788 m/s22
9.807 m/s Helsinki
9.819 m/s2 Rome
9.803 m/s2
9.799 m/s2 Kuwait
9.793 m/s2 San Francisco 9.800 m/s2
9.783 m/s2 Lisbon
9.801 m/s2 Singapore
9.781 m/s222
9.813 m/s London (Greenwich) 9.812 m/s Stockholm
9.818 m/s2
9.811 m/s2 Los Angeles
9.796 m/s2 Sydney
9.797 m/s222
9.797 m/s Madrid
9.800 m/s Taichung9.789 m/s2
9.788 m/s2 Manila
9.784 m/s2 Vancouver, BC 9.809 m/s2
9.796 m/s2 Melbourne
9.800 m/s2 Washington DC 9.801 m/s229.803 m/s Mexico City9.779 m/s2 Wellington NZ 9.803 m/s2
9.815 m/s2 Milan
9.806 m/s2 Zurich
9.807 m/s222
9.797 m/s New York9.802 m/s Taiwan
9.788 m/s2
9.781 m/s2 Oslo
9.819 m/s2 Taipei
9.790 m/s29.810 m/s2 Ottawa
9.806 m/s2 Tokyo
9.798 m/s222
9.816 m/s Paris9.809 m/s

2/2