FT Technologies Test Equipment makes it easy to test, evaluate and change the settings on the FT range of analogue
output wind sensors. The Acu-Test (Analogue) pack is made-up of the FT055 cable and our Acu-Vis software.

INCLUDED IN THE PACK

toSensor1

USB

toComputer4Speed Loop
(4-20mA)

toPower2

Speed Loop
(4-20mA)

to
Meter3+24 VDC

toPower20 VDC

toPower2

Direction Loop
(4-20mA)

toPower2

Direction Loop
(4-20mA)

to
Meter31x FT055 cable
Each cable is labelled with a unique serial number and build date to ensure quality andtraceability. Cable accessories include a protection cap for sensor connector and an adjustablerubber sleeve to keep the wires tidy. For full connection details please visit our website.

Acu-Test (Analogue)

Software andDatasheet

FT Technologies Ltd
Church Lane
Teddington
Middlesex
TW11 8PA UK
sales@fttech.co.uk
www.fttech.co.uk

1x Acu-Vis CD

YOU WILL ALSO NEED+Sensor1

1.
2.
3.
4.

+
OR+
Power Source2

Meter3

Computer4

This equipment can only be used to test the FT range of analogue output wind sensors.
Use a bench power supply unit, 20-30 VDC (6A), Rechargeable Battery or +24 VDC Laptop charger.
The sensor current loops can be measured using an ammeter, voltmeter or oscilloscope.
Acu-Vis software will run on Window XP, Vista or Windows 7 computers.

FT TECHNOLOGIES LTD
The FT and Acu-Vis logos are registered trademarks of FT technologies Ltd
2A4204 May 2012

ACU-VIS SOFTWARE
The Acu-Vis software can be installed from the CD provided and is available in English, French and Chinese.
The software automatically finds and identifies if a FT055 cable is connected to the computer.

1. Sensor Serial Number including VersionNumber

2. Software Version on Sensor
3. Datum Current Configuration
4. Error Current
5. Wind Speed Range
6. Heater Setpoint
7. Heater Current Limit
8. Wind Velocity Filter
9. Datum Offset
Above - Screen shot with sensor 7000-696 connected

CHANGING SETTINGS
Settings can be saved in a log file along with a time and date stamp. Making it easy to keep a record of which settings
are applied to every sensor.
Depending on the sensor version, different settings can also be changed.
 The options that are configurable can be selected and the “Update Settings” button used to apply thesesettings to the sensor.
 It is also possible to return the sensor to its original settings using the “Factory Defaults” button.

SEEING THE SENSOR WORKBy clicking on the “Wind Display” tab, you can visualise the real time air flow that the sensor is calculating

By clicking on the “Loop Test” tab, you can step the analogue electronics and current loop outputs through a testroutine. This sets the outputs to fixed values that can then be measured allowing you to check their accuracy.

The Acu-Test (Analogue) pack is available from sales@fttech.co.uk+24 VDC Laptop charger (available on request).

Information furnished by FT Technologies Ltd is believed to be accurate
and reliable. However no responsibility is assumed by FT Technologies Ltdfor its use; nor for any infringements of patents or other rights of third
parties which may result from its use. No licence is granted by implicationor otherwise under any patent rights of FT Technologies Ltd.

FT Technologies Ltd Church Lane, Teddington, Middlesex, TW118PA, England. Tel:+44(0)20 8943 0801 Fax:+44(0)20 8943 3283
E-mail: sales@fttech.co.uk Web: http://www.fttech.co.uk