Electrical Thermometer
Model UT-201BLE-A
Oral type

Instruction manual
Original
Type oral

Manuel d’instructions
Traduction
Tipo oral

Manual de instrucciones
Traducción
Tipo orale

Manuale di istruzioni
Traduzione
Oraler typ

BedienungsanleitungÜbersetzung
口溫計

使用手冊
翻譯
1WMPD4003378D

Contents
Dear Customers ....................................................................... 2
Preliminary Remarks ................................................................. 2
Precautions .............................................................................. 3
Contraindications ................................................................. 4
Parts Identification ................................................................... 5
Symbols ................................................................................... 6
Using the Thermometer ............................................................ 7
Installing / Changing the Battery .......................................... 7
Wireless Function ................................................................. 8
Bluetooth Transmission .................................................... 9
Pairing ............................................................................... 10
Measurement and Transmitting Data ..................................11
Changing Units .................................................................. 11
Applying the Thermometer ................................................ 12
Measurement Time ............................................................ 12
After Measurement............................................................. 12
Measurements ........................................................................ 13
Adjusted Mode Measurement ............................................. 13
Direct Mode Measurement.................................................. 14
Notes for Accurate Measurement........................................14
Features ................................................................................. 15
Adjusted Mode Measurement ............................................. 15
Direct Mode Measurement.................................................. 15
Switch with Flash Action ..................................................... 15
Last Reading Display and Memory ...................................... 15
Temperature Unit ............................................................... 15
Bluetooth ........................................................................ 15
Troubleshooting ..................................................................... 16
Maintenance ........................................................................... 17
Maintenance ...................................................................... 17
Cleaning ............................................................................ 17
Storage .............................................................................. 17
Cautions ............................................................................ 17
Regular Inspection ............................................................. 17
Disposal ............................................................................. 17
Technical Data........................................................................ 18

English 1

Dear Customers
Congratulations on purchasing a state-of-the-art A&D thermometer, oneof the most advanced thermometers available today. Designed for easeof use and accuracy, this thermometer will facilitate your thermometer
regimen.
We recommend that you read through this manual carefully before
using the device for the first time.
The Bluetoothword mark and logos are registered trademarks owned
by Bluetooth SIG, Inc. and any use of such marks by A&D is underlicense. Other trademarks and trade names are those of their respective
owners.

Preliminary Remarks
This device conforms to the European Directive 93/42 EEC for Medical
Products. This is made evident by themark of conformity.
( 0123 : The reference number to the involved notified body. )This device fulfills the provisions of BS EN 12470 Clinical thermometers Part 3: Performance of compact electrical thermometers (non-predictive
and predictive) with maximum device.
Hereby, A&D Company, Limited declares that the radio equipment type
UT-201BLE-A is in compliance with Directive 2014/53/EU. The full text of
the EU declaration is available at the following internet address:
http://www.aandd.jp/products/manual/manual_medical.html
The device is a Continua certified, Bluetooth
enabled medical device.

wireless technology

The device is designed to be used in the medical facilities.
This device is designed to measure body temperature.
This device is designed to be operated by an adult (18 years old orolder).
This device intends to measure the body temperature of the patient(5 years old or older).

English 2

Precautions
Precision components are used in the construction of this device.
Extremes in temperature, humidity, direct sunlight, shock or dust should
be avoided. It may be cause of losing performances of sensor, battery,electrical terminals and this device.
This device is the thermometer to measure an oral temperature ofbottom side of tongue. Do not measure a temperature of other position
so it is incorrect.
Clean the device with a dry, soft cloth or a cloth dampened with waterand a neutral detergent. Never use alcohol, benzene, thinner or otherharsh chemicals to clean the device.
Clean the device before and after use. Keep cleanly to be able to insert
into mouth. It may be the cause of occurring a cross-infection if not clean.
Avoid excessive shock. It may be the cause of a malfunction.
Do not put the device in the neighborhood of heater. Prevent the device
from splashing of a hot water. It may be the cause of a malfunction.
The device is not water resistant. Prevent rain, sweat and water fromsoiling the device.
Measurements may be distorted if the device is used close to televisions,microwave ovens, X-ray or other devices with strong electrical fields.
Wireless communication devices, such as networking devices, mobile
phones, cordless phones and their base stations, walkie-talkies canaffect this thermometer. Therefore, a minimum distance of 30 cm should
be kept from such devices.
When reusing the device, confirm that the device is clean.
Used equipment, parts and battery are not treated as ordinary household
waste, and must be disposed of according to the applicable local
regulations.
Do not modify the device. It may cause accidents or damage to thedevice.
Do not let children use the device by themselves and do not use thedevice in a place within the reach of infants.
There are small parts that may cause a choking hazard if swallowed by
mistake by infants.
When the liquid inside of the battery invades into an eye, wash eye withlarge quantities of water as quickly as possible, consult the doctor for
diagnose and treatment. It may be the cause of blindness and injury, if
not perform.
English 3

When your skin and cloth are touched to the liquid inside of the battery,wash them with large quantities of water.
Replacement of battery by inadequately trained personnel could result ina HAZARD (such as excessive temperatures, fire or explosion).
Use the battery, removable parts and materials that are described in thismanual. It may be the cause of malfunction and injury, if not use.
Insert the battery with proper polarities (+) and (-) into the compartment.
It may be the cause of malfunction and injury, if not insert correctly.
Do not short-circuit the battery. Failure to do so may lead to fluid leakage,heat generation or bursting, and resulting in injury.
Do not heat the battery. Failure to do so may lead to fluid leakage,
bursting, and resulting in injury.
Prevent the device from chewing and bending. It may be the cause of
malfunction and injury, if it is chewed and bended.
We recommend that you read through this manual carefully before using
the device for the first time.
Do not use this device with equipment such as a sensor cover. This may
affect the accuracy of measurements.

Contraindications
The following are precautions for proper use of the device.
Do not use the device where flammable gases such as anesthetic gases
are present. It may cause an explosion.
Do not use the device in highly concentrated oxygen environments, suchas a high-pressure oxygen chamber or an oxygen tent. It may cause afire or explosion.

English 4

Parts Identification
ON / Standby switch with LED.
Battery : CR2032

Battery cover on
the battery compartment

Case
Temperature sensor

This manual

Display
Communication mark
Memory markBattery indicator
Temperature valueCount down indicatorfor measurements

Units of temperature
Adjusted Mode measurement markDirect Mode measurement markEnglish 5

Symbols
Symbols that are displayed on the device
SymbolsSN
BT

Function / MeaningStandby and turn on the thermometer.
Last reading is stored in memory when the mark is displayed.
Full battery mark.
Low battery mark: A half of the battery capacity was consumed.
Battery is low when it blinks. Replace a battery with new one.
Temperature is above 42 °C during measurement.
Temperature is below 32 °C during measurement.
Thermometer or room temperature is above 40 °C.
Thermometer or room temperature is below 10 °C
Measurement is not correct. Check the way of use.
Malfunction of thermometer. Contact your dealer.
Time out of Bluetooth communication.
Bluetooth communication error.
Bluetooth communication mark.
Pair mark to construct Bluetooth communication pair.
Adjusted Mode measurement mark.
Direct Mode measurement mark.
Temperature units of Celsius and Fahrenheit.
Count down indicator for measurements means a waiting time until
displaying temperature. This indicator may include few timing error
in process.
Direct current.
Type BF: Device is designed to provide special protection againstelectrical shocks.
EC directive medical device label
EU-representative
Manufacturer
Date of manufacture
Not waterproof
Class II device
WEEE label
Serial number
Bluetooth address
Refer to instruction manual/booklet
Negative electrode
English 6

Symbols

Function / Meaning
To indicate generally elevated, potentially hazardous, levels of
non-ionizing radiation, or to indicate equipment or systems e.g. inthe medical electrical area that include RF transmitters or that
intentionally apply RF electromagnetic energy for diagnosis ortreatment.

Using the Thermometer
Installing / Changing the Battery
1.

Remove the battery cover.

2.

Remove the used battery with astick.

3.

Insert a new battery into thebattery compartment as shown,taking care that the polarities (+)
and (-) are correct.

4.

Step 2

Step 1
Step 4

Step 3Replace the battery cover.
Use only CR2032 battery.

CautionInsert the battery as shown in the battery compartment. If installed
incorrectly, the device will not work.
When(Low battery mark) blinks in the display, replace the battery
with a new one. Replace the battery after the device turns off and wait for
two seconds or more.
(Low battery mark) does not appear when the battery is drained.
Use the specified battery only. The battery provided with the device is fortesting thermometer performance and may have a limited life.
Remove the battery if the device is not to be used for a long time.
The battery may leak and cause a malfunction.
Keep the thermometer out of the reach of children. A child may swallow
the battery while playing with it. If a child should swallow the battery,seek medical treatment immediately.

English 7

Using the Thermometer
Wireless Function
CautionIn the unlikely event that this thermometer causes radio wave
interference to a different wireless station, change the location where thisthermometer is used or stop use immediately.
Be sure to use in a location where visibility between the two devices thatyou want to connect is good. The connection distance is reduced by the
structure of buildings or other obstructions. In particular, connection may
be impossible when devices are used on either side of reinforced
concrete.
Do not use Bluetoothconnection in the range of a wireless LAN orother wireless devices, near devices that emit radio waves such as
microwaves, in locations where there are many obstructions, or in otherlocations where signal strength is weak. Doing so may result in frequentloss of connection, very slow communication speeds and errors.
Using close to an IEEE802.11g/b/n wireless LAN device may cause
mutual interference to occur, which may result in reduced communication
speeds or which may prevent connection. In this case, switch off thepower supply to the device that is not being used, or use thethermometer in a different location.
If the thermometer does not connect normally when used near a wirelessstation or broadcast station, use the thermometer in a different location.
A&D Company, Limited cannot accept liability for any damages incurred
due to impaired operation or data loss, etc. that occur through the use ofthis device.
This device is not guaranteed to connect to all Bluetooth
devices.

English 8

compatible

Using the Thermometer
BluetoothTransmission

This device is equipped with a Bluetooth
wireless function and canconnect to the following Bluetooth
devices.
Continua certified devices
iPhone, iPad, iPod (iPhone 4S or later)
Applications and devices that are compatible with Bluetooth 4.0.
Each device needs an application to receive data.
For connection methods, refer to the manual for each device.
Bluetoothmark.

devices carry the Bluetoothlogo

Continua certified devices carry the Continua logo
mark.

English 9

Using the Thermometer
Pairing

A Bluetooth
device needs to be paired with a different specific device inorder to communicate with that device. If this thermometer is paired with areceiver device from the start, measurement data is transmitted
automatically to the receiver device each time a measurement is made.

Cautions for Pairing
Only one device can be paired with this thermometer at one time. If thereceiver device cannot receive measurement data, try pairing again
If another receiver device is paired, the first device will be unpaired toenable the new device to be paired.
Follow the steps below to pair the thermometer with a Bluetoothcompatible receiver device. Also refer to the manual of the receiver device.
Please use a pairing wizard if one is provided.
Pairing Procedure
1.

Follow the instructions in the manual of the receiverdevice to switch it to the pairable status. When pairing
this thermometer, place it as close as possible to thereceiver device to be paired with.

2.

Install the battery as described on page 7.
Press theswitch to turn the thermometer on.
Press theswitch while " " is displayed.
The thermometer can be found by the receiver device
while " " is displayed for approx. one minute.

3.

Find, select and build a pair with the receiver device inaccordance with its manual. When the pairing of thereceiver device is built, "" of the decision of thepairing is displayed.

4.

If "
" is displayed or pairing is failed, remove thebattery and try steps 1 to 3 again.

5.

Follow the manual of the pairing receiver device tosearch for, select and pair with this thermometer.

Communication DistanceThe communication distance between this thermometer and the receiverdevice is approximately 5 m.
This distance is reduced by the conditions in the surrounding environment,so be sure to check that the distance is short enough for a connection to bemade after measurement is complete.
English 10

Using the Thermometer
Measurement and Transmitting Data
1.
2.

The communication performs the following steps after building the paring.
Keep the condition of the receiver device so as to communicate.
Turn on the thermometer. Data is measured automatically.
Data is transmitted after finishing the measurement.

Transmitting Temporarily Stored Data
In cases when the receiver device cannot receive measurement data, themeasurement data is temporarily stored in the thermometer memory. The
data stored in the memory is transmitted the next time a connection is
successfully made to the receiver device
A total of 90 sets of measurement data can be stored. When the amount ofdata exceeds 90 sets, the oldest data is deleted and the new data is stored.
The amount of data that can be stored temporarily may vary with theapplication.
Time
This thermometer has a built-in clock. The date and time that ameasurement was taken is included in the measurement data.
The built-in clock is designed to be automatically adjusted by syncing withthe clock of a receiver device. Refer to the specifications of the receiverdevice. This thermometer has no clock adjustment function.

Changing Units
1.

2.

Press and hold theswitch above 6 seconds when turning off thethermometer. The new unit is displayed after blinking the last unit.
The thermometer turns off automatically.
Turn off the device.
When the same operation is
performed again, an effective unit
Press and hold theis exchanged.
switch above six seconds.
The unit is stored in the memory.
The unit of the factory setting
is °C (centigrade).
Last unit

New unit

The device is turned off automatically.
English 11

Using the Thermometer
Applying the Thermometer

Put the thermometer sensor on the mouth floor under the tongue, at theroot of the tongue and at the side of the lingual frenulum.
Keep the position of the sensor during measurement
Tongue

Lingual frenulum
TongueMouth floor

Sensor position

Measurement Time

When the adjusted mode measurement is used, keep the sensor
position for approximately 30 seconds with placing the sensor at acorrect position in the mouth.

When the direct mode measurement is used, keep the sensor position
for approximately 5 minutes with placing the sensor at a correct positionin the mouth.
We recommend to use the direct mode measurement for a precision
thermometry.

After Measurement

After measurement, press and hold theseconds to turn the device off.

switch for one or more

Note: The device has an automatic power shut-off function, which turns thepower off approximately one minute after measurement.

English 12

Measurements
Adjusted Mode Measurement
1. Press theswitch.
When the last measurement is stored,
it is displayed for approximately two
seconds.

2.

Turn on the device.
Last data

Wait until " " is displayed.

Ready
Sensor position

3.

Put the temperature sensor on themouth floor (under the tongue, at theroot of the tongue and at the side ofthe lingual frenulum).
Close the mouth gently.

4.

keep the sensor position during
measurement for approximately
30 seconds.

Measurement startCount down indicator
5.

6.

7.

The count down indicator isdisplayed.
The adjusted mode measurement
mark blinks and rotates.

The result is displayed forapproximately 15 seconds, theadjusted mode measurement marklights, the LED of theswitchblinks and buzzer sounds when theadjusted mode measurement finishes.

The result
Mark lights
To next page if the directmode measurement is used.

Select an operation.
Press theAdjisted modemeasurement markswitch to turn the thermometer off.

Keep the sensor position to use the direct modemeasurement. Proceed to next page.
English 13

Measurements
From last page
Direct Mode Measurement
8. When the direct modemeasurement starts, the markblinks. Keep the sensor position
for approximately five minutes.
9.

The result is displayed forapproximately one minute, thedirect mode measurement marklights, the LED of theswitchblinks and buzzer sounds when
the direct mode measurement
finishes.

10. Press and hold theturn the device off.

Measurement startCount down indicator

switch to

Direct modemeasurement markThe result
Mark lights
Turn off the device.

Notes for Accurate Measurement
Themark lights when the data is stored in memory.

The direct mode measurement is performed after the adjusted modemeasurement is finished.
The device is provided with an automatic power shut-off function with thebuzzer that the device is turned off at approximately one minute later fromremoving it or displaying data.
The device can be turned off when pressing and holding theswitch.
In measurement, breathe with using nose and closing mouth.
Should the device detect a condition that is abnormal, it will stop themeasurement and display an error symbol. See page 6 for the description of
the symbols.
This thermometer is intended for use by adults only. Consult with yourphysician before using this device on a child. A child should not use thisdevice unattended.
To ensure accurate measurement, wait at least 30 seconds between each
use to allow the thermometer to return to room temperature.

English 14

Features
Direct mode reading
Adjusted modereading

Adjusted Mode Measurement
The adjusted modemeasurement calculates the
direct mode temperature
after five minutes when the
body temperature ismeasured for 30 seconds. It
is based on an ascending
thermal curve.
If you need a precision data,
we recommend to use thedirect mode measurement fora precision thermometry.

Body temperature
Adjusted mode
calculation

Thermal curvein ascending.

Room temperature
30 seconds
5 minutes

Results of Clinical Evaluation
Clinical Bias
-0.06 °C
Clinical Repeatability
0.12 °C
Limit of Agreement
0.22 °C
Standard Deviation
0.08 °C

Direct Mode Measurement
The body temperature can be measured using for five minutes.

Switch with Flash Action
When pressing theswitch , this switch flashes.
When finishing the measurement, this switch flashes.

Last Reading Display and Memory
The previous reading stored in memory is automatically displayed whenturning on the thermometer. The new reading is stored in memory whenmeasuring the temperature.

Temperature Unit
Temperature unit of Celsius or Fahrenheit can be selected .

Bluetooth
Temperature data can be transmitted to the receiver that is paired with thethermometer.

English 15

Troubleshooting
ProblemPossible Reason

Recommended Action
Replace the battery with a newBattery is drained.
one.
Nothing appears in
the display, even
Reinstall the battery with negative
when the power is Battery terminals are not in and positive terminals matchingturned on.
those indicated on the battery
the correct position.
compartment.
Battery power is low.
If the battery is drained
Replace the battery with a new
No measurement
completely, the mark does one.
not appear.
Temperature will change at
Normal body
Measure the temperature underawaking, in activity, after
temperature
the same condition.
eating.
includes error.
Body temperature
Incorrect sensor position Check the sensor position.
is displayed low.
The device calculates an
Measure after several minutes
Body temperature direct mode temperature
again or use the direct mode
is displayed high. after five minutes.
measurement.
Therefore, it includes error.
The paring is not
Place the device in proximity of
the receiver. Make a paring.
Data transmission established.
error
Replace the battery with a newBattery is not enough.
one.
Note: If the actions described above do not solve the problem, contact the
dealer. Do not attempt to open or repair this device, as any attempt to do
so will make your warranty invalid.

English 16

A&D Company, Limited
1-243 Asahi , Kitamoto-shi, Saitama 364-8585, JAPAN
Telephone: [81] (48) 593-1111 Fax: [81] (48) 593-1119
Emergo Europe B.V.
Westervoortsedijk 60, 6827 AT Arnhem, The Netherlands

A&D INSTRUMENTS LIMITED
Unit 24/26 Blacklands Way, Abingdon Business Park, Abingdon, Oxfordshire
United Kingdom
Telephone: [44] (1235) 550420 Fax: [44] (1235) 550485
A&D ENGINEERING, INC.
1756 Automation Parkway, San Jose, California 95131,
Telephone: [1] (408) 263-5333 Fax: [1] (408)263-0119

OX14 1DY

U.S.A.

A&D AUSTRALASIA PTY LTD
32 Dew Street, Thebarton, South Australia 5031, AUSTRALIA
Telephone: [61] (8) 8301-8100 Fax: [61] (8) 8352-7409
OOO A&D RUS

OOO "ЭЙ энд ДИ РУС"121357, Российская Федерация, г.Москва, ул. Верейская, дом 17( Business-Center "Vereyskaya Plaza-2" 121357, Russian Federation, Moscow, Vereyskaya Street 17 )тел.: [7] (495) 937-33-44
факс: [7] (495) 937-55-66

A&D Technology Trading(Shanghai) Co. Ltd 爱安德技研贸易（上海）有限公司
中国 上海市浦东新区浦东大道138号永华大厦21楼A室 邮编200120( 21F Room A, Majesty Building, No.138 Pudong Avenue, Pudong New Area, Shanghai, 200120, China )电话: [86] (21) 3393-2340
传真: [86] (21) 3393-2347

A&D INSTRUMENTS INDIA PRIVATE LIMITED( 509, Udyog Vihar, Phase- , Gurgaon - 122 016, Haryana, India ): 91-124-4715555: 91-124-4715599