Dust & Water-Proof Precision Balances

FZ-iWPiWP
FX-Series

ISO9000 CERTIFIED...Clearly a Better Value
http://www.aandd.jp

The World’s First IP65 Compliant,
Dust & Water-ProofW
Water-Pro
oof Precision
Precisi ion (1 mg)
Balancese
Yet another breakthrough from A&D enabling the measurement ofpowder or liquid at 1 mg resolution, eliminating any worries aboutdamage to the balance!

What does “IP65” mean?
An International Protection (IP) Code, as definedby the international standard IEC 60529,
specifies the degrees of protection provided byenclosures. The two digits next to IP indicate thelevels of protection against solid foreign objects(first digit) and water (second digit). For example,a device rated as IP65 should permit no ingressof dust (level 6) and withstand water jettingfrom any direction (level 5).

Newly Developed, Air-Pump Controlled Internal Calibrationi(FZ-iWP models only) * i Patent Pending
For the internal calibration of the FZ-iWP Series, A&D adopted a direct-driving methodthat utilizes air pressure for its power source. This new method features a simple
structure, high reliability and a fail-safe mechanism – the internal mass returns to theoriginal position even if the calibration is interrupted by power failure, etc.

One Touch Automatic Calibration (FZ-iWP models only)
With just one press of a key, you can ensure the accuracy of your balance.

Super Hybrid
Sensor

VFD Display

Quick USB

LAN-Ethernet
Interface

RS-232C
Interface

GLP
CompliantCounting
Function

Percentage
Function

AnimalWeighing

Statistical
Calculation

Auto PowerOFF

Underhook

Rechargeable
Battery

IP65

Software

Internal
Calibration

EC TypeApproval

External
Calibration

ON
Comparator
Function

Auto PowerON(FZ-iWP)(FX-iWP)

Compact, B5 Size Footprint
A&D’s latest, Compact Super Hybrid Sensor (C-SHS) technology ii has significantly reduced
the size of the FZ-iWP/FX-iWP Series – a 25% smaller footprint than the GX/GF Series –making it ideal for installation and use in narrow spaces.
* ii Patent Pending

Fast, 1 Second Stabilization
While maintaining extremely high span stability, the C-SHS also realizes an ultra-fast
response speed – just 1 second – to maximize your productivity and efficiency.

RS-232C Interface (standard)
Bi-directional communication is possible with a PC,
printer or other peripheral device.
Dust & Water-Proof RS-232C Cable (optional)

Quick USB Interface (FXi-02)

The easy-to-use optional USB interface requires no driver or software installation foriii
transfer of weighing data to a PC.

LAN-Ethernet Interface (FXi-08) with WinCT-Plus Software
The user can send commands for controlling and acquiring data from multiple balances.

WinCT-PlusLANUSB

Ethernet Hub
Provided by end-user

FX i -08

FX i -08

FX i -02

FX i -08

communication from a balance to a
* iii Unidirectional
PC only. Please use the standard RS-232C interface tosend commands from a PC to the balance.

Shock Absorber Functioniv
The internal shock absorber protects the C-SHS from the adverse effects
of vertical and transverse overloading. * iv Patent Pending

Multiple Weighing Units
Programmable or standard weighing units of measurement can beselected: g, pcs, %, oz, lb, L oz, ozt, ct, mom, dwt, GN, tl, tol, mes, and MLT.

Statistical Calculation Function
The balance displays and/or outputs statistical calculation data as No.
(number of data), SUM (total), MAX (maximum), MIN (minimum),R [range (maximum – minimum)], AVE (average), SD (standard deviation)
and CV (coefficient of variation).

Built-in Rechargeable Battery (FXi-09)
An optional Ni-MH rechargeable battery pack can be fitted into thebalance for use without AC power and for mobile applications (10hours charging time for 8 hours operation).

Large Breeze BreakNote:(FXi-WP-11)

All FZ-iWP models, the FX-120iWP,
the FX-200iWP and the FX-300iWP
come with a small breeze break asstandard.

Specifications

FZ-120iWP
FX-120iWP
122g

Weighing capacity

FZ-200iWP
FX-200iWP

FZ-300iWP
FX-300iWP

220g

320g

Minimum weighing value

0.001gCounting mode

0.001gMinimum unit mass

1220g

2200g

3200g

0.01g
0.01g
5, 10, 25, 50 or 100 pieces

Number of samples
Percent mode

FZ-1200iWP FZ-2000iWP FZ-3000iWP
FX-1200iWP FX-2000iWP FX-3000iWP

0.100gMinimum 100% reference mass

1.00g

0.01%, 0.1%, 1% (depends on the reference mass stored)

Minimum 100% display
Repeatability (Standard Deviation)

0.001g0.01g

Linearity±0.002g±0.02gApprox. 1 second

Stabilization time (typically set at FAST)±2ppm / °C

Sensitivity drift (10°C to 30°C)

5°C to 40°C (41°F to 104°F), 85%RH or less (no condensation)

Operating temperature

Vacuum Fluorescent Display (VFD)

Display type

5 times /second, 10 times /second or 20 times /second

Display refresh rate

RS-232C Interface

Standard interface

130mm

Weighing pan (diameter)

150mm

Air-pump Controlled Internal Calibration (FZ-i WP models only)Internal calibration
External calibration mass

100g

200g

300g

1000g

2000g

3000g

50g

100g

200g

500g

1000g2000g

100g

1000g

Confirm that the adaptor type is correct for the local voltage and power receptacle

AC adaptor

Approx. 11VA (supplied to the AC adaptor)

Power consumption

193 (W) X 262.5 (D) X 85.5 (H) mm

External dimensions
Net weight

FZ-i WP Approx. 3.2kg / FX-i WP Approx. 2.7kg

Options

External Dimensions (mm/inches)

168.5 / 6.63

Accessories*

61.5 / 2.42

194.5 / 7.66
192*

3*

*Underhook

Dimensions with Small Breeze Break (FX i-WP-10)

193 / 7.6

(Comes equipped with FZ,FX-120iWP/200iWP/300iWP, FZ-1200iWP/2000iWP/3000iWP)

184 / 7.24

184 / 7.24
168.5 / 6.63

115 / 4.53

171.5 / 6.75
175 / 6.89

ø130 / 5.12

81 / 3.19 83.5 / 3.29

115 / 4.53
85.5 / 3.33 85 / 3.35

Static eliminator
Electrostatic fieldmeter
Weighing data logger
Tweezers for calibration weight
Remote display
Remote controller
Compact printer
Universal printerSerial/Ethernet converter
Quick USB adapter
Serial/USB converter
Waterproof RS-232C cable (5m)
Carat pan (silver, 2pcs)
Carat pan (black, 2pcs)
Main unit cover (5pcs)

262.5 / 10.33

A162 / 6.38

AD-1683
AD-1684
AD-1688*
AD-1689
AD-8920A*
AD-8922*
AD-8121B*
AD-8118C*
AD-8526*
AD-8527*
AX-USB-9P*
AX-KO2737-500-EX
AX-CARATPAN-W
AX-CARATPAN-B
AX-FXi-31

A
130 / 5.12
150 / 5.91

Model
FZ,FX-120iWP / 200iWP / 300iWP
FZ,FX-1200iWP / 2000iWP / 3000iWP

262.5 / 10.33

FXi-02, FXi-08 and FXi-09 cannot be used at the same time
* (Not
IP65 compliant when used with FZ,FX-i WP)

Pan size

70.5 / 2.78

Quick USB interface with cable
Ethernet interface with WinCT-Plus software
Built-in rechargeable battery
Small breeze break (gray)
Large breeze break (gray)
Animal weighing pan
Carrying case85.5 / 3.37

FXi-02*
FXi-08*FXi-09*FXi-WP-10
FXi-WP-11
FXi-12FXi-15

193 / 7.6

Dimensions with Large Breeze Break (FX i-WP-11)(Not IP65 compliant when used with FZ,FX-i WP)

105/4.13

112/4.41

294/11.57

315/12.4

199.5/7.85

193/7.6

Ø156/6.14

FXi-12

107/4.21...Clearly a Better Value
3-23-14 Higashi-lkebukuro, Toshima-ku,Tokyo 170-0013 JAPAN
Telephone:[81](3) 5391-6132 Fax:[81](3) 5391-6148
http://www.aandd.jp

Unit 24/26 Blacklands Way Abingdon Business Park,Abingdon, Oxon OX14 1DY UNITED KINGDOM
Telephone:[44](1235) 550420 Fax:[44](1235) 550485<German Sales Office>

Vereyskaya str.17, Moscow, 121357 RUSSIA
Telephone: [7] (495) 937-33-44 Fax: [7] (495) 937-55-66

A&D
D InstInstrumentsuments India Pr
Privatee Limited
mited
509 Udyog Vihar Phase V
Gurgaon-122 016, Haryana, INDIA
Telephone: [91](124) 471-5555 Fax: [91](124) 471-5599

1756 Automation Parkway, San Jose, CA 95131 U.S.A.
Telephone:[1](408) 263-5333 Fax:[1](408) 263-0119

Große Straße 13 b 22926 Ahrensburg GERMANY
Telephone:[49](0) 4102 459230 Fax:[49](0) 4102 459231

32 Dew Street, Thebarton, South Australia 5031 AUSTRALIA
Telephone:[61](8) 8301-8100 Fax:[61](8) 8352-7409

Manhattan Bldg. 8F, 36-2 Yoido-dong, Youngdeungpo-gu, Seoul, KOREA
Telephone:[82](2) 780-4101 Fax:[82](2) 782-4280*FZi/FXiWP-ADCC-01-PR1-09703