PicoScope 6000E Series®

A smarter scope for faster debug

Deep-memory, high-performance oscilloscopes and MSOs
8-bit to 12-bit FlexRes® ADC
A choice of 4 or 8 analog channels
Add 8 or 16 digital MSO channels
Up to 500 MHz bandwidth
200 ms capture time at 5 GS/s
Up to 4 GS capture memory
50 MHz 200 MS/s 14-bit AWG
300 000 waveforms per second update rate
PicoScope 6 software and PicoSDK included
21 serial protocol decoder / analyzers included
Mask limit testing and user-definable alarms
High-resolution time-stamping of waveforms
Over ten million DeepMeasure™ results per acquisition
Advanced triggers: edge, window, pulse width, window pulse width,level dropout, window dropout, interval, runt and logic

www.picotech.com

Product overview
The PicoScope 6000E Series fixed-resolution and FlexRes oscilloscopes provide 8 to 12 bits of vertical resolution, with up to 500 MHz bandwidth and 5 GS/s sampling rate. Models
with four or eight analog channels have the timing and amplitude resolution you need to reveal signal integrity issues such as timing errors, glitches, dropouts, crosstalk andmetastability issues.

Typical applications
These oscilloscopes are ideal for design engineers working with high-performance embedded systems, signal processing, power electronics, mechatronics and automotive designs,
and for researchers and scientists working on multi-channel high-performance experiments in physics labs, particle accelerators and similar facilities.

Best-in-class bandwidth, sampling rate and memory depth
Capture time at maximum sampling rate: 200 ms at 5 GS/s
With up to 500 MHz analog bandwidthcomplemented by a real-time sampling rate of
5 GS/s, the PicoScope 6000E Series scopes
can display single-shot pulses with 200 ps time
resolution.
The PicoScope 6000E Series gives you thedeepest capture memory available as standard on
any oscilloscope at any price – up to 4 GS in total.
This ultra-deep memory allows the oscilloscopeto capture 200 ms waveforms at its maximum
sampling rate of 5 GS/s.
The SuperSpeed USB 3.0 interface and hardware
acceleration ensure that the display is smoothand responsive even with long captures.
The PicoScope 6000E Series gives you the
waveform memory, resolution and analysis toolsthat you need to perform stringent testing of
today’s high‑performance embedded computersand next-generation embedded system designs.

PicoScope® 6000E Series

Power, portability and performance
Traditional benchtop mixed-signal oscilloscopes take up a lot of bench space, and models with eight analog channels are prohibitively expensive for many engineers working on
next-generation designs. PicoScope 6000E Series oscilloscopes are small and portable while offering the high-performance specifications required by engineers in the lab or on themove, and deliver lowest cost of ownership for this class of instrument.
The PicoScope 6000E Series offers up to 8 analog channels, plus an optional 8 or 16 digital channels with the plug-in 8-channel TA369 MSO (mixed-signal oscilloscope) pods. The
flexible high‑resolution display options enable you to view and analyze each signal in detail.
Supported by PicoScope 6 software, these devices offer an ideal, cost-effective package for many applications, including design, research, test, education, service, and repair.
PicoScope 6 is included in the price of your scope, available for free download, with free updates, and can be installed on as many PCs as you want, allowing you to view/analyze
data off-line without the scope.

PicoScope® 6000E Series

What is FlexRes?
Pico FlexRes flexible-resolution oscilloscopes allow you to reconfigure the scope
hardware to optimize either the sampling rate or the resolution.

8 bitsThis means you can reconfigure the hardware to be either a fast (5 GS/s) 8-bit
oscilloscope for looking at digital signals, a 10-bit oscilloscope for general-purpose use
or a high-resolution 12-bit oscilloscope for audio work and other analog applications.

10 bits

Whether you’re capturing and decoding fast digital signals or looking for distortion insensitive analog signals, FlexRes oscilloscopes are the answer.

12 bits

FlexRes is available on the 8-channel PicoScope 6824E and the 4-channel PicoScope
6424E.

The FlexRes architecture employs multiple high‑resolution
ADCs at the input channels in different time-interleaved andparallel combinations to optimize, for example, the sampling
rate to 5 GS/s at 8 bits or the resolution to 12 bits at 1.25 GS/s.
For simplicity, the diagram shows one bank of four channels;
the 8-channel PicoScope 6824E has two banks. The PicoScope
6424E uses one quad-ADC chip for each pair of analog
channels.
Coupled with high signal-to-noise ratio amplifiers and a lownoise system architecture, FlexRes technology can capture anddisplay signals up to 500 MHz with a high sampling rate, or
lower-speed signals with 16 times more resolution than typical
8-bit oscilloscopes.
PicoScope 6 software lets you choose between setting theresolution manually and leaving the scope in auto resolutionmode, where the optimal resolution is used for the chosensettings.

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

ADC
1.25 GS/s

MULTI-CHANNEL*

TIME-INTERLEAVED*

Independent sampling on all
channels at 8-bit or 10-bit
resolution.

TIMING

Most digital oscilloscopes gain their high sampling rates byinterleaving multiple 8-bit ADCs. This interleaving processintroduces errors that always make the dynamic performance
worse than that of the individual ADC cores.

TIMING

FlexRes – how we do it

Maximum 5 GS/s sampling rate
in 8 or 10-bit mode.

TIMING

Resolution enhancement—a digital signal processing technique built into PicoScope 6—
can further increase the effective vertical resolution of the scope to 16 bits.

ADC
1.25 GS/s

PARALLEL*

Simultaneous sampling in 12-bit
mode at up to 1.25 GS/s on two
channels.

* See technical specifications for channel and sampling rate combinations.

PicoScope® 6000E SeriesMixed signal option
When fitted with optional 8-channel TA369 MSO pods, the PicoScope 6000E Series adds up to 16 high-performance digital channels to up to eight analog
channels, enabling you to accurately time-correlate analog and digital channels. Digital channel bandwidth is 500 MHz, equivalent to 1 Gb/s, and the input
capacitance of only 3.5 pF minimizes loading on the device under test.
Digital channels, captured from either parallel or multiple serial buses, may be grouped and displayed as a bus, with each bus value displayed in hex, binary
or decimal, or as a level (for DAC testing). You can set advanced triggers across the analog and digital channels.
The digital inputs also bring extra power to the serial decoding feature. You can decode serial data on all analog and digital channels
simultaneously, giving you up to 24 channels of data – for example, decoding multiple SPI, I²C, CAN bus, LIN bus and FlexRay signals all at thesame time!

TA369 MSO pod with 8 digital inputs

Analog waveforms (top) and digital waveforms (bottom) shown on PicoScope 6 display

Digital channels connected to a device under test

PicoScope® 6000E SeriesPicoScope 6000E Series inputs, outputs and indicators

8-channel front panel

Rear-panel

Input channels A to H

Intelligent probe interfaces
Power LED
Status/trigger LED

AUX TRIGGER I/O – trigger
from an external logic levelsource and integrate the scope
into a larger system

Digital 1 and Digital 2
MSO pod interfaces –accept TA369 MSO pods

Probe compensation output
Probe compensation groundUSB 3.0 port

10 MHz clock reference input
The scope will automaticallyswitch to the external reference
when a clock signal is detected.

Ground – accepts barewire or 4 mm (banana)plug.

Intelligent probe interface (probes coming soon!)

4-channel front-panel
Analog input channels A to D, with intelligent probe interfaces

Power LED
Status/trigger LED

AWG output
50 MHz 14 bits
200 MS/s

12 V DC input – use only themains power adaptor suppliedwith the oscilloscopeWith an intelligent probe interface on channels C to F on 8-channel models and allchannels on 4-channel models, the PicoScope 6000E Series will support innovative
active probes with a low-profile mechanical design for ease of connectivity and low
loading of the device under test.

Digital 1 and Digital 2
MSO pod interfaces –accept TA369 MSO pods

Probe compensation output
Probe compensation ground

PicoScope® 6000E SeriesPicoScope 6 software
The display can be as simple or as
advanced as you need. Begin witha single view of one channel, andthen expand the display to include
any number of live channels, mathchannels and reference waveforms.

Tools: Including
serial decoding,referencechannels, macro
recorder, alarms,
mask limit
testing and mathchannels.

Waveform replay tools:
PicoScope 6 automaticallyrecords up to 10 000 of themost recent waveforms. You
can quickly scan through tolook for intermittent events,
or use the Buffer Navigator tosearch visually.

Zoom and pan tools:
PicoScope 6 allows azoom factor of several
million, which is necessary
when working with theultra-deep memory of the6000E Series scopes.

Signal generator:
Generates standard
signals or arbitrary
waveforms.
Includes frequency
sweep mode.

Auto setup button:
Configures the collection
time and voltage range forclear display of signals.

Rulers: Eachaxis has tworulers that can be
dragged acrossthe screen to
make quick
measurementsof amplitude,
time andfrequency.

Channel options:
Filtering, offset, resolution
enhancement, custom
probes and more.
Oscilloscope controls:
Controls such as voltage
range, scope resolution,
channel enable, timebase
and memory depth.

Properties
sheet: Showsa summary of
the settings that
PicoScope isusing.

Movable axes: The
vertical axes can bescaled and dragged up
or down. This feature
is particularly usefulwhen one waveformis obscuring another.
There’s also an
Auto Arrange Axes
command.

Zoom overview: Clickand drag for quick
navigation in zoomed
views.

Ruler legend:
Absolute anddifferential ruler
measurements
are listed here.

Trigger toolbar: Quick
access to main controls,with advanced triggers
in a pop-up window.

Automatic measurements: Display calculated measurements
for troubleshooting and analysis. You can add as manymeasurements as you need on each view. Each measurement
includes statistical parameters showing its variability.

Trigger marker: Drag
the yellow diamond
to adjust trigger leveland pre-trigger time.

Views:
PicoScope 6
is carefully
designed tomake the bestuse of thedisplay area. You
can add newscope, spectrum
and XY views
with automaticor customlayouts.

PicoScope® 6000E Series

Advanced display
PicoScope 6 software dedicates the majority of the display area to the waveform,
ensuring that the maximum amount of data is visible at all times. The size ofthe display is only limited by the size of your computer’s monitor, so even with alaptop, the viewing area is much bigger, with much higher resolution, than that of abenchtop scope.
With such a large display area available, you can create a customizable split-screen
display and view multiple channels or different views of the same signal at thesame time – the software can even show multiple oscilloscope and spectrum
analyzer views at once. Each view has separate zoom, pan and filter settings, forultimate flexibility.
You can control the PicoScope software using a mouse, a touchscreen or
customizable keyboard shortcuts.

SuperSpeed USB 3.0 connection
PicoScope 6000E Series oscilloscopes feature a
USB 3.0 connection, providing lightning-fast savingof waveforms while retaining compatibility witholder USB standards.
PicoSDK® supports continuous streaming to thehost computer at rates of over 300 MS/s.
The USB connection not only allows high-speed
data acquisition and transfer, but also makes
printing, copying, saving and emailing your datafrom the field quick and easy.

PicoScope 6 custom colors
In PicoScope 6, you can customize thecolor scheme and line thicknesses. Display
elements you can adjust in this way include
the channel traces, background color andgrid lines.

PicoScope® 6000E Series

Signal fidelity
Careful front-end design and shielding reduces noise, crosstalk and harmonic distortion.
PicoScope 6000E Series oscilloscopes exhibit a dynamic performance of up to
60 dB SFDR.
With PicoScope 6, when you probe a circuit, you can trust in the waveform you see on
the screen.

High resolution for low-level signals
With their 12-bit resolution, the PicoScope 6824E and 6424E can display low-level signals athigh zoom factors. This allows you to view and measure features such as noise and ripple
superimposed on larger DC or low-frequency voltages.
Additionally, you can use the lowpass filtering controls on each channel independently, to hide
noise and reveal the underlying signal.

PicoScope® 6000E Series

High-end features as standard
Buying a PicoScope is not like making a purchase from other oscilloscope companies,
where optional extras considerably increase the price. With our scopes, high-end
features such as serial decoding, mask limit testing, advanced math channels,
segmented memory, hardware‑based time-stamping and a signal generator are all
included in the price.
To protect your investment, both the PC software and firmware inside the scope can be
updated. Pico Technology has a long history of providing new features for free throughsoftware downloads. We deliver on our promises of future enhancements year afteryear. Users of our products reward us by becoming lifelong customers and frequently
recommending us to their colleagues.

Total cost of ownership (TCO) and environmental benefits
Total cost of ownership of a PicoScope 6000E is lower than traditional benchtop instruments for several reasons:
1. Low power consumption - just 60W - saves hundreds of dollars throughout the lifetime of the product compared
to benchtop instruments. It’s kinder to the environment too, with lower CO2 emissions.
2. Everything is included in the purchase price: serial protocol decoders / math channels and mask limit testing. No
expensive optional upgrades or annual license fees.
3. Free updates: new features and capabilities are provided throughout the lifetime of the product as we develop andrelease them.

PicoScope® 6000E Series

Ultra-deep memory
PicoScope 6000E Series oscilloscopes have waveform capture memories of up to 4 gigasamples – many times larger than competing scopes. Deep memory enables the captureof long-duration waveforms at maximum sampling speed. In fact, the PicoScope 6000E Series can capture waveforms 200 ms long with 200 ps resolution. In contrast, the same
200 ms waveform captured by an oscilloscope with a 10 megasample memory would have just 20 ns resolution. The scope automatically shares the capture memory between theanalog channels and MSO ports you have enabled.
Deep memory is invaluable when you need to capture fast serial data with long gaps between packets, or nanosecond laser pulses spaced milliseconds apart, for example. It canbe useful in other ways too: PicoScope lets you divide the capture memory into a number of segments, up to 10 000. You can set up a trigger condition to store a separate capturein each segment, with as little as 300 ns dead time between captures. Once you have acquired the data, you can step through the memory one segment at a time until you find theevent you are looking for.
Powerful tools are included to allow you to manage and examine all of this data. As well as functions such as mask limit testing and color persistence mode, PicoScope 6 software
enables you to zoom into your waveform up to 100 million times. The Zoom Overview window allows you to easily control the size and location of the zoom area. Other tools, suchas the waveform buffer, serial decoding and hardware acceleration work with the deep memory, making the PicoScope 6000E Series some of the most powerful oscilloscopes on
the market.

PicoScope® 6000E Series

Persistence mode
PicoScope’s persistence mode options allow you to see old and new data superimposed, making it easy to spot glitches and dropouts and estimate their relative frequency – usefulfor displaying and interpreting complex analog signals such as video waveforms and amplitude modulated signals. Color-coding and intensity-grading show which areas are stable
and which are intermittent. Choose between Digital Color, Analog Intensity, Fast and Advanced display modes or create your own custom setup.
An important specification to understand when evaluating oscilloscope performance, especially in persistence mode, is the waveform update rate, which is expressed as
waveforms per second. While the sampling rate indicates how frequently the oscilloscope samples the input signal within one waveform or cycle, the waveform capture rate refers
to how quickly an oscilloscope acquires waveforms.
Oscilloscopes with high waveform capture rates provide better visual insight into signal behavior and dramatically increase the probability that the oscilloscope will quickly capture
transient anomalies such as jitter, runt pulses and glitches – that you may not even know exist.
The PicoScope 6000E Series’ HAL4 hardware acceleration can achieve update rates of 300 000 waveforms per second in fast persistence mode.

PicoScope® 6000E Series

Serial bus decoding and protocol analysis
PicoScope can decode 1-Wire, ARINC 429, BroadR-Reach, CAN & CAN FD, DALI, DCC, DMX512, Ethernet 10Base-T and 100Base-TX, FlexRay, I²C, I²S, LIN, PS/2, Manchester, Modbus,
SENT, SPI, UART (RS-232 / RS-422 / RS-485), and USB 1.1 protocol data as standard, with more protocols in development and available in the future with free-of-charge software
upgrades.
Graph format shows the decoded data (in hex, binary, decimal or ASCII) in a data-bus timing format beneath the waveform on a common time axis, with error frames marked in red.
These frames can be zoomed to investigate noise or signal integrity issues.
Table format shows a list of the decoded frames, including the data and all flags and identifiers. You can set up filtering conditions to display only the frames you are interested in or
search for frames with specified properties. The statistics option reveals more detail about the physical layer such as frame times and voltage levels. PicoScope can also import a
spreadsheet to decode the data into user-defined text strings.
Click on a frame in the table to zoom the oscilloscope display and show the waveform for that frame.

PicoScope® 6000E Series

DeepMeasure
One waveform, millions of measurements.
Measurement of waveform pulses and cycles is key to verification of the performance of electrical and electronic devices.
DeepMeasure delivers automatic measurements of important waveform parameters, such as pulse width, rise time and voltage. Up to a million cycles can be displayed with eachtriggered acquisition. Results can be easily sorted, analyzed and correlated with the waveform display.

PicoScope® 6000E SeriesMask limit testing
Mask limit testing allows you to compare live signals against known good signals, and is designed for production and debugging environments. Simply capture a known goodsignal, draw (or have PicoScope auto-generate) a mask and then measure the system under test. PicoScope will check for mask violations and perform pass/fail testing, captureintermittent glitches, and can show a failure count and other statistics in the Measurements window.

Waveform buffer and navigator
Ever spotted a glitch on a waveform, butby the time you’ve stopped the scope it
has gone? With PicoScope you don’t need
to worry about missing glitches or othertransient events. PicoScope can store thelast ten thousand oscilloscope or spectrum
waveforms in its circular waveform buffer.
The buffer navigator provides an efficient
way of navigating and searching throughwaveforms, effectively letting you turnback time. Tools such as mask limit testing
can also be used to scan through eachwaveform in the buffer looking for mask
violations.

PicoScope® 6000E Series

FFT spectrum analyzer
The spectrum view plots amplitude against frequency and is ideal for finding noise,crosstalk or distortion in signals. The spectrum analyzer in PicoScope is of the Fast
Fourier Transform (FFT) type that, unlike a traditional swept spectrum analyzer, candisplay the spectrum of a single, non-repeating waveform. With up to a million points,
PicoScope’s FFT has excellent frequency resolution and a low noise floor.
With a click of a button, you can display a spectrum plot of the active channels, with amaximum frequency of up to 500 MHz. A full range of settings gives you control overthe number of spectrum bands (FFT bins), scaling (including log/log) and display modes(instantaneous, average, or peak-hold). A selection of window functions allow you to
optimize for selectivity, accuracy or dynamic range.
You can display multiple spectrum views alongside oscilloscope views of the samedata. A comprehensive set of automatic frequency-domain measurements can be added
to the display, including THD, THD+N, SNR, SINAD and IMD. A mask limit test can beapplied to a spectrum and you can even use the AWG and spectrum mode together toperform swept scalar network analysis.
FM radio broadcasts

10 MHz sine wave showing 60 dB SFDR

Harmonics of a square-wave signal

PicoScope® 6000E Series

Powerful tools provide endless options
Your PicoScope is provided with many powerful tools to help you acquire and analyze waveforms. While these tools can be used on their own, the real power of PicoScope lies in
the way they have been designed to work together.
As an example, the rapid trigger mode allows you to collect 10 000 waveforms in a few milliseconds with minimal dead time between them. Manually searching through thesewaveforms would be time-consuming, so just pick a waveform you are happy with and let the mask tools scan through for you. When done, the measurements will tell you how
many have failed and the buffer navigator allows you to hide the good waveforms and just display the problem ones.
The screenshot below shows changing frequency versus time as a graph. Perhaps instead you want to plot changing duty cycle as a graph? How about outputting a waveform fromthe AWG and also automatically saving the waveform to disk when a trigger condition is met? With the power of PicoScope the possibilities are almost endless. To find out evenmore about the capabilities of PicoScope software, visit our online A to Z of PC Oscilloscopes.

PicoScope® 6000E Series

Arbitrary waveform and function generator
The PicoScope 6000E scopes have a built-in 50 MHz function (sine and square wave) generator, with triangle, DC level, white noise, PRBS and other waveforms possible at lower
frequencies. As well as basic controls to set level, offset and frequency, more advanced controls allow you to sweep over a range of frequencies. Combined with the spectrum peakhold option, this makes a powerful tool for testing amplifier and filter responses.
Trigger tools allow one or more cycles of a waveform to be output when various conditions are met, such as the scope triggering or a mask limit test failing.
All models include a 14-bit 200 MS/s arbitrary waveform generator (AWG). This has a variable sample clock, which avoids the jitter on waveform edges seen with fixed-clock
generators and allows generation of accurate frequencies down to 100 µHz. AWG waveforms can be created or edited using the built-in editor, imported from oscilloscope traces,
loaded from a spreadsheet or exported to a .csv file.

PicoScope® 6000E Series

Digital triggering architecture
Many digital oscilloscopes still use an analog trigger
architecture based on comparators. This causes time andamplitude errors that cannot always be calibrated out andoften limits the trigger sensitivity at high bandwidths.
In 1991 Pico pioneered the use of fully digital triggering
using the actual digitized data. This technique reduces
trigger errors and allows our oscilloscopes to trigger on thesmallest signals, even at the full bandwidth. Trigger levels
and hysteresis can be set with high precision and resolution.

Advanced triggers
The PicoScope 6000E Series offers an industry-leading setof advanced trigger types including pulse width, runt pulse,
windowed, logic and dropout.
The digital trigger available during MSO operation allows you
to trigger the scope when any or all of the 16 digital inputs
match a user-defined pattern. You can specify a condition foreach channel individually, or set up a pattern for all channels
at once using a hexadecimal or binary value.
You can also use the logic trigger to combine the digitaltrigger with an edge or window trigger on any of the analog
inputs, for example to trigger on data values in a clocked
parallel bus.

AlarmsPicoScope can be programmed to execute actions when certain events occur.
The events that can trigger an alarm include mask limit fails, trigger events and buffers
full.
The actions that PicoScope can execute include saving a file, playing a sound, executing
a program and triggering the signal generator or the AWG.
Alarms, coupled with mask limit testing, help create a powerful and time-saving
waveform monitoring tool. Capture a known good signal, auto-generate a mask around it
and then use the alarms to automatically save any waveform (complete with a time/datestamp) that does not meet specification.

PicoScope® 6000E Series

Hardware acceleration engine (HAL4)
Some oscilloscopes struggle when you enable deep memory; the screen update rate
slows and the controls become unresponsive. The PicoScope 6000E Series avoids thislimitation with the use of a dedicated fourth-generation hardware acceleration (HAL4)
engine inside the oscilloscope.
Its massively parallel design effectively creates the waveform image to be displayed on
the PC screen and allows the continuous capture and display to the screen of 2.5 billion
samples every second.
The hardware acceleration engine eliminates any concerns about the USB connection or
PC processor performance being a bottleneck.

Time-stamping
The PicoScope 6000E Series features
hardware-based trigger time-stamping.
Each waveform can be time-stamped with thetime in sample intervals from the previous
waveform.
Fast trigger rearm times are possible down to
300 ns (typical).

Time from first trigger incircular buffer to current trigger

Time from previous trigger tocurrent trigger

PicoScope® 6000E Series

Ultra-high-definition display
PicoScope PC-based instruments use the host computer’s display, which is typically larger and of higher resolution than the dedicated displays installed in traditional benchtop
oscilloscopes. This allows room for simultaneous display of time- and frequency-domain waveforms, decoded serial bus tables, measurement results with statistics and more.
PicoScope 6 software scales automatically to take full advantage of the improved resolution of larger display sizes, including 4K ultra-high definition models. At 3840 x 2160
resolution—over eight million pixels—PicoScope allows engineers to get more done in less time through split-screen views of multiple channels (or different views of the samechannel) from the device under test. As the example shows, the software can even show multiple oscilloscope and spectrum analyzer traces at once.
Large, high-resolution displays really come into their own when viewing high-resolution signals with the PicoScope 6824E and 6424E 8- to 12-bit FlexRes models. With a 4K monitor,
PicoScope can display more than ten times the information of some of our competitors’ scopes, solving the problem of how to match a big display and features with a smallfootprint portable oscilloscope.
PicoScope also supports dual monitors: instrument control and waveforms displayed on the first, and large data sets from serial protocol decoders or DeepMeasure results on thesecond. The software can be controlled by mouse, touchscreen or keyboard shortcuts.

PicoScope® 6000E Series

Math channels and filters
With PicoScope 6 you can select simple functions such as addition and inversion,or open the equation editor to create complex functions involving filters (lowpass,
highpass, bandpass and bandstop filters), trigonometry, exponentials, logarithms,
statistics, integrals and derivatives.
Display up to eight real or calculated channels in each scope view. If you run out of
space, just open another scope view and add more. You can also use math channels toreveal new details in complex signals, for example graphing the changing duty cycle orfrequency of your signal over time.

Custom probes in PicoScope oscilloscope software
The custom probes feature allows you to correct for gain, attenuation, offsets andnonlinearities in probes, sensors or transducers that you connect to the oscilloscope.
This could be used to scale the output of a current probe so that it correctly displays
amperes. A more advanced use would be to scale the output of a nonlinear temperature
sensor using the table lookup function.
Definitions for standard Pico-supplied oscilloscope probes and current clamps areincluded. User-created probes may be saved for later use.

PicoScope® 6000E SeriesPicoSDK® - write your own apps
Our free software development kit, PicoSDK, allows you to write your own software and includes drivers for Windows, macOS and Linux. Example code supplied on our GitHub
organization page shows how to interface to third-party software packages such as National Instruments LabVIEW and MathWorks MATLAB.
Among other features, the drivers support data streaming, a mode that captures continuous gap-free data directly to your PC or host computer at rates of over 300 MS/s, so you are
not limited by the size of your scope’s capture memory. Sampling rates in streaming mode are subject to PC specifications and application loading.
There is also an active community of PicoScope users who share both code and whole applications on our Test and Measurement Forum and the PicoApps section of the website.
The Frequency Response Analyzer shown here is a popular application on the forum.

Copyright © 2014-2017 Aaron Hexamer. Distributed under GNU GPL3.

PicoScope® 6000E Series

Optional accessories
TA369 MSO pod
The PicoScope 6000E Series can be upgraded to MSO capability. This consists of adetachable active MSO pod, powered by the scope, with eight permanently attachedflying leads terminating in MSO probes for connection to the circuit under test.
The active MSO pods bring the MSO input circuitry closer to the device under test
minimizing loading and giving the best possible performance.
The MSO pod connects to either of two digital interface ports on the scope front panel
using a 0.5 m digital interface cable. All PicoScope 6000E Series models support the
MSO pod.
The innovative single and multi-way ground clips allow fast and flexible connection to all
signal and ground pins in a double row header, regardless of where the layout engineer
has placed them.
Features:• 8 digital inputs per pod• 500 MHz bandwidth, 1 Gb/s
• 5 GS/s sampling on 16 digital channels• 1 ns minimum pulse width• Minimal load on the device under test: 101 kΩ ∥ 3.5 pF• Innovative ground clips for easy connection to 2-row, 2.54 mm-pitch headers• 8 ground leads and 12 mini test hooks included
An MSO pod spares kit (PQ221) is also available which contains extra 1-way, 4-way and8-way MSO ground clips and MSO ground leads.

When a header has signal
pins on one row andgrounds on the other row.

When you find a header
with signals situated
together and not enoughgrounds. A ground leadcan be used to connect toa remote ground pin on thedevice under test.

For a header with a mixof non-adjacent andadjacent signal pins. Thank
goodness my MSO is so
versatile!

PicoScope® 6000E Series

Probe positioning system
The Pico oscilloscope probe positioning system holds your circuit board firmly during soldering, inspection and test.
The kits include flexible probe holders which secure magnetically to the steel base plate. When the probes are installed in the holders they can be positioned to make contact withpoints of interest on the circuit board and will remain in contact while you make measurements in the PicoScope software.
The large steel base plate is mirror-finished allowing you to see any items such as status LEDs underneath the PCB.

Probe positioning system kit contents:
Item
PCB holder
Base plate, 210 x 297 mm
Set of insulation washers for PCB holders
Pico probe holder, 2.5 mm
Set of cable holders channels A-D
Set of cable holders channels E-H
P2056 500 MHz 10:1 passive BNC probe

PQ215 kit
4
1
1
4
1
1
If you already own a
4- or 8-channel scopewith four probes, thiskit is the ideal add-on.

PQ219 kit
PQ218 kit
4
1
1
8
4
1
1
1
1
4
Upgrade your 8-channel Four extra probe
scope from four toholders.
eight probes, and add
eight probe holders.

Passive analog high-impedance probes:
P2056 500 MHz and P2036 300 MHz passive probes are supplied with your scope and are also available separately insingle or dual packs. These probes feature a probe-detect readout BNC connector allowing automatic recognition as a
10:1 attenuator by the scope.
Probe connection is confirmed by a notification in PicoScope 6.
Features:••••

Up to 500 MHz bandwidth
10:1 attenuation
Trimmed to match the oscilloscope
Probe-detect readout pin for automatic range scaling

A comprehensive selection of accessories is supplied in the single probe packs and a basic selection in the dual packs.
Further accessories are available as listed in the P2056 and P2036 user’s guide.

PicoScope® 6000E SeriesPicoScope 6000E Series specifications
PicoScope 6824E

PicoScope 6424E

PicoScope 6804E

PicoScope 6404E

PicoScope 6403E

Vertical (analog channels)
Input channels
Bandwidth (–3dB)
Rise time
Bandwidth limit
Vertical resolution
Enhanced vertical resolution (software)
Input connector
Input characteristics
Input coupling
Input sensitivity

8
4
8
4
4
500 MHz
300 MHz< 850 ps
< 1.3 ns
20 MHz, selectable
8, 10 or 12 bits FlexRes
8 bits fixed
Up to 4 extra bits beyond ADC resolution
BNC(f)
1 MΩ ±0.5% ∥ 12 pF ±1 pF, or 50 Ω ±2%. Single-ended.
1 MΩ AC/DC or 50 Ω DC
2 mV/div to 4 V/div (10 vertical divisions)
1 MΩ ranges: ±10 mV, ±20 mV, ±50 mV, ±100 mV, ±200 mV, ±500 mV, ±1 V, ±2 V, ±5 V, ±10V, ±20 V
Input ranges (full scale)
50 Ω ranges: ±10 mV, then as above up to ±5 V
DC gain accuracy±(0.5% of signal + 1 LSB)±(1.5% of signal + 1 LSB)±(1% of full scale + 250 µV)
DC offset accuracy
Offset accuracy can be improved by using the “zero offset” function in PicoScope 6.
8 bits: < 0.4 % of input range
LSB size (quantization step size)
10 bits: < 0.1 % of input range
8 bits: < 0.4 % of input range
12 bits: < 0.025 % of input range
Analog offset range (vertical position±1.25 V (10 mV to 1 V ranges)
adjustment)±20 V (2 V to 20 V ranges)
Analog offset control accuracy±0.5% of offset setting, additional to DC accuracy above
1 MΩ ranges: ±100 V (DC + AC peak) up to 10 kHz
Overvoltage protection
50 Ω ranges: 5.5 V RMS
Vertical (digital channels with optional TA369 8-channel MSO pods)
Input channels
8 channels per MSO pod. Supports up to 2 pods / 16 channels.
Maximum detectable input frequency
500 MHz (1 Gb/s)
Minimum detectable pulse width
1 ns
Input connector (probe tip)
Staggered signal and ground sockets for each channel, to accept 0.64 - 0.89 mm round or 0.64 mm square pin, 2.54 mm pitch
Input characteristics
101 kΩ ±1% ∥ 3.5 pF ±0.5 pF
Threshold range and resolution±8 V in 5 mV steps
Threshold accuracy±(100 mV + 3% of threshold setting)
PicoScope 6: Two independent threshold controls, one per 8-channel port
Threshold grouping
PicoSDK: Individual threshold for each channel
Threshold selection
TTL, CMOS, ECL, PECL, user-defined
Maximum input voltage at probe tip±40 V up to 10 MHz, derated linearly to ±5 V at 500 MHz
Minimum input voltage swing (at 500 MHz)
400 mV peak to peak

PicoScope® 6000E Series

Hysteresis (at DC)
Minimum input slew rate
Horizontal
Maximum sampling rate (real time, 8-bit mode)

PicoScope 6824E
PicoScope 6424E
PicoScope 6804E
PicoScope 6: Fixed hysteresis approx. 100 mV.
PicoSDK: selectable per port; approx. 50 mV, 100 mV, 200 mV or 400 mV
No minimum slew rate requirement

PicoScope 6404E

Up to 2 total analog channels and MSO pods

5 GS/s[¹]5 GS/s[³]5 GS/s[¹]5 GS/s[³]Up to 4 total analog channels and MSO pods

2.5 GS/s[²]2.5 GS/s

2.5 GS/s[²]2.5 GS/s

1.25 GS/s
N/A

1.25 GS/s
625 MS/s

1.25 GS/s
N/A

5 GS/s
2.5 GS/s
1.25 GS/s
625 MS/s
N/A

N/A

1.25 GS/s[³]N/AUp to 8 total analog channels and MSO pods
1.25 GS/s
Over 8 channels and digital ports
625 MS/s
Maximum sampling rate (real time, 10-bit mode)
1 analog channel or digital port
5 GS/sUp to 2 total analog channels and MSO pods
2.5 GS/s[²]Up to 4 total analog channels and MSO pods
1.25 GS/s
Up to 8 total analog channels and MSO pods
625 MS/s
Over 8 channels and digital ports
312.5 MS/s
Maximum sampling rate (real time, 12-bit mode)
Up to 2 analog channels plus any MSO pods
1.25 GS/s[¹][ ]¹ No more than one channel from each of ABCD and EFGH
[ ]² No more than one channel from each of AB, CD, EF and GH[ ]³ No more than one channel from each of AB and CD

PicoScope 6403E

2.5 GS/s [³](2 analog channels)
5 GS/s (1 or no analog
channels)
1.25 GS/s (3-4 analog
channels)
2.5 GS/s [³](2 analog channels)
1.25 GS/s

Max. sampling rate, USB 3.0 streaming mode,
PicoScope 6 (split between active channels,~20 MS/s
PC dependent)
Max. sampling rate, USB 3.0 streaming mode,
PicoSDK (split between active channels, PC
dependent)
Max. sampling rate to on-device buffer(continuous USB streaming of downsampled
data, PicoSDK only, split between enabledchannels)
Capture memory(shared between active channels)~312 MS/s (8-bit mode)~156 MS/s (10/12-bit modes)~312 MS/s

1.25 GS/s (8-bit mode)
625 MS/s (10/12-bit modes)

N/A

4 GS (2 GS at 10/12-bit)

2 GS

1 GS

PicoScope® 6000E SeriesPicoScope 6824E
Maximum single capture duration at
maximum sampling rate (PicoScope 6)
Maximum single capture duration at
maximum sampling rate (PicoSDK)Capture memory (continuous streaming)
Waveform buffer (number of segments)
Timebase ranges
Initial timebase accuracy
Timebase drift
ADC samplingExternal reference clock
Input characteristics
Input frequency range
Input connector
Input level
Overvoltage protection
Dynamic performance (typical)
CrosstalkHarmonic distortion, 8-bit modeHarmonic distortion, 10/12-bit mode
SFDR, 8-bit mode
SFDR, 10/12-bit mode
Noise, 8-bit mode
Noise, 10/12-bit mode
Linearity, 8-bit mode
Linearity, 10-bit mode
Bandwidth flatness
Low frequency flatness
Triggering
SourceTrigger modes
Advanced trigger types (analog channels)
Trigger sensitivity (analog channels)
Advanced trigger types (digital inputs)

PicoScope 6424E

PicoScope 6804E

PicoScope 6404E

PicoScope 6403E

200 ms
800 ms (8-bit)
400 ms (10-bit)
400 ms
400 ms200 ms1600 ms (12-bit)
100 MS in PicoScope software. Buffering using full device memory when using PicoSDK, no limit on total duration of capture.
PicoScope 6: 10 000
PicoSDK: 2 000 000
PicoSDK: 1 000 000
1 ns/div to 5000 s/div±2 ppm±1 ppm/year
Simultaneous sampling on all enabled analog and digital channels
Hi-Z, AC coupled (> 1 kΩ at 10 MHz)
10 MHz ±50 ppm
Rear-panel BNC, dedicated
200 mV to 3.3 V peak to peak
±5 V peak max±10 mV to ±1 V ranges: better than 1200:1 up to full bandwidth (equal voltage ranges)±2 V to ±20 V ranges: better than 300:1 up to full bandwidth (equal voltage ranges)–50 dB at 1 MHz full scale–60 dB at 1 MHz full scale
N/A> 50 dB on ±50 mV to ±20 V ranges> 60 dB on ±50 mV to ±20 V ranges
N/A< 200 μV RMS on most sensitive range< 150 μV RMS on most sensitive range
N/A< 2 LSB< 4 LSB
N/A(+0.3 dB, –3 dB) from DC to full bandwidth< ±3% (or ±0.3 dB) from DC to 1 MHz
Any analog channel, AUX trigger, plus digital ports with optional TA369 MSO pods
None, auto, repeat, single, rapid (segmented memory)
Edge, window, pulse width, window pulse width, level dropout, window dropout, interval, runt, logic
Logic allows arbitrary combinations of up to 4 analog channels or MSO ports
Digital triggering provides 1 LSB accuracy up to full bandwidth of scopeWith optional MSO pods: Edge, pulse width, dropout, interval, pattern, logic (mixed signal)

PicoScope® 6000E SeriesPicoScope 6824E
Maximum pre-trigger capture
Maximum post-trigger delay
Rapid trigger mode rearm time
Maximum trigger rate
Trigger time-stamping
Auxiliary trigger I/O
Connector type
Trigger types (triggering scope)
Input bandwidth
Input characteristics
Threshold range
Hysteresis
Overvoltage protection
Function generator
Standard output signals
Standard signal frequency
Output frequency accuracy
Output frequency resolution
Sweep modes
Sweep frequency range
Sweep frequency resolution
Triggering
GatingPseudorandom output signals
Output voltage range
Output voltage adjustment
DC accuracy
Amplitude flatness
Analog filters
SFDR

PicoScope 6424E

PicoScope 6804E

PicoScope 6404E

PicoScope 6403E

100% of capture size
PicoScope 6: up to 0.8 s at fastest timebase
PicoSDK: > 1012 sample intervals, settable in 1 sample steps (delay range at fastest sample rate of > 200 s in 200 ps steps)
700 ns max, 300 ns typical (single channel, 5 GS/s)
PicoScope 6: 10 000 waveforms in 3 ms; PicoSDK: 1 million waveforms in 0.3 s
Each waveform is timestamped with a resolution of one sample interval relative to the previous waveform. Time resets when any
settings are changed.
Rear-panel BNC
Edge, pulse width, dropout, interval, logic> 10 MHz
2.5 V CMOS Hi-Z input, DC coupled
Fixed threshold, 1.25 V nominal
1 V max (VIH < 1.75V, VIL > 0.75V)±20 V peak max
Sine, square, triangle, DC voltage, ramp up, ramp down, sinc, Gaussian, half-sineSine: 100 μHz to 50 MHz; Square: 100 μHz to 50 MHz; Other waves: 100 μHz to 1 MHz
Oscilloscope timebase accuracy ± output frequency resolution
0.002 ppm
Up, down, dual with selectable start/stop frequencies and increments
Sine / square waves: 0.075 Hz to 50 MHz
Other waves: 0.075 Hz to 1 MHz
Swept frequencies down to 100 μHz are possible using PicoSDK with some restrictions
In PicoScope 6 software: 0.075 Hz. Sweep frequency resolution down to 100 μHz is possible via PicoSDK with some restrictions
Free-run, or from 1 to 1 billion counted waveform cycles or frequency sweeps. Triggered from scope trigger or manually.
Software controlled gating of waveform output
White noise, selectable amplitude and offset within output voltage range
Pseudorandom binary sequence (PRBS), selectable high and low levels within output voltage range, selectable bit rate up to
50 Mb/s±5 V into open circuit; ±2.5 V into 50 Ω
Signal amplitude and offset adjustable in < 1 mV steps within overall range±(0.5% of output voltage + 20 mV)< 2.0 dB to 50 MHz (sine wave into 50 Ω)< 0.5 dB to 50 MHz (square)< 1.0 dB to 1 MHz (other waveforms)
50 MHz selectable filter (5-pole, 30 dB/octave)
70 dB (10 kHz 1 V peak to peak sine into 50 Ω)

PicoScope® 6000E SeriesPicoScope 6824E
Output noise
Output resistanceConnector type
Overvoltage protection
Arbitrary waveform generator
Update rate
Buffer size
Vertical resolution

PicoScope 6424E

PicoScope 6804E

PicoScope 6404E

PicoScope 6403E< 700 μV RMS (DC output, filter enabled, into 50 Ω)
50 Ω ±3%
Rear-panel BNC±20 V peak max

Variable from < 1 S/s to 200 MS/s with < 0.02 ppm resolution
40 kS
14 bits (output step size < 1 mV)
No filter: 100 MHz
Bandwidth
Filtered: 50 MHz
No filter: 3.5 ns
Rise time (10% to 90%)
Filtered: 6 ns
Sweep modes, triggering, frequency accuracy and resolution, voltage range and accuracy and output characteristics as for function generator.
Probe support
Active probe interface on four channels supporting A6000 Series active probes. Probe interface supplies power and controls theActive probe interface
active probe.
Probe detection
Automatic detection of P2036 (300 MHz) and P2056 (500 MHz) 10:1 oscilloscope probes
Probe compensation pin
1 kHz, 2 V peak to peak square wave, 600 Ω
Probe compensation pin rise time
< 50 ns
Spectrum analyzer
DC to 500 MHz
DC to 300 MHz
Frequency range
Magnitude, average, peak hold
Display modes
Logarithmic (dbV, dBu, dBm, arbitrary dB) or linear (volts)
Y axisLinear or logarithmic
X axis
Rectangular, Gaussian, triangular, Blackman, Blackman−Harris, Hamming, Hann, flat-top
Windowing functions
Selectable from 128 to 1 million in powers of 2
Number of FFT points
Math channels−x, x+y, x−y, x*y, x/y, x^y, sqrt, exp, ln, log, abs, norm, sign, sin, cos, tan, arcsin, arccos, arctan, sinh, cosh, tanh, delay, average,
Functionsfrequency, derivative, integral, min, max, peak, duty, highpass, lowpass, bandpass, bandstop, coupler
A to H (input channels), T (time), reference waveforms, pi, 1D0 to 2D7 (digital channels), constants
Operands
Automatic measurements
AC RMS, true RMS, frequency, cycle time, duty cycle, DC average, falling rate, rising rate, low pulse width, high pulse width, fall time,
Scope moderise time, minimum, maximum, peak to peak
Frequency at peak, amplitude at peak, average amplitude at peak, total power, THD %, THD dB, THD+N, SFDR, SINAD, SNR, IMD
Spectrum mode
Minimum, maximum, average, standard deviation
Statistics
DeepMeasure™
Cycle number, cycle time, frequency, low pulse width, high pulse width, duty cycle (high), duty cycle (low), rise time, fall time,
Parameters
undershoot, overshoot, max. voltage, min. voltage, voltage peak to peak, start time, end time

PicoScope® 6000E SeriesPicoScope 6824E

PicoScope 6424E

PicoScope 6804E

PicoScope 6404E

PicoScope 6403E

Serial decoding
Protocols
Mask limit testingStatistics
Mask creation
DisplayInterpolation
Persistence modes
Languages

1-Wire, ARINC 429, BroadR-Reach, CAN & CAN FD, DALI, DCC, DMX512, Ethernet 10Base-T and 100Base-TX, FlexRay, I²C, I²S, LIN,
PS/2, Manchester, Modbus, SENT, SPI, UART (RS-232 / RS-422 / RS-485), and USB 1.1
Pass/fail, failure count, total countUser-drawn, table entry, auto-generated from waveform or imported from file
Linear or sin(x)/x
Digital color, analog intensity, custom, fast
Chinese (simplified), Chinese (traditional), Czech, Danish, Dutch, English, Finnish, French, German, Greek, Hungarian, Italian,Japanese, Korean, Norwegian, Polish, Portuguese, Romanian, Russian, Spanish, Swedish, Turkish

PicoScope® 6000E SeriesPicoScope 6824E
General specifications
PC connectivity
PC connector type
Power requirement
Ground terminal
Dimensions
WeightTemperature range
Humidity range
Altitude range
Pollution degree
IP rating
Safety compliance
EMC compliance
Environmental compliance
Software
PC requirements
Warranty
MSO pod dimensions
MSO digital interface cable length
MSO probe flying lead length
MSO pod size
MSO probe size

PicoScope 6424E

PicoScope 6804E

PicoScope 6404E

PicoScope 6403E

USB 3.0 SuperSpeed (USB 2.0 compatible)
USB Type B
12 V DC from supplied PS016 PSU. Up to 7 A including powered accessories.
Functional ground terminal accepting wire or 4 mm plug, rear-panel
245 x 192 x 61.5 mm
2.2 kg (scope only)
5.6 kg (in carry case with PSU and cables)
Operating: 0 to 40 °C
For quoted accuracy after 20 minutes warm-up: 15 to 30 °C
Storage: –20 to +60 °C
Operating: 5% to 80% RH non-condensing
Storage: 5% to 95% RH non-condensing
Up to 2000 m
EN 61010 pollution degree 2: “only nonconductive pollution occurs except that occasionally a temporary conductivity caused bycondensation is expected”
IP20
Designed to EN 61010-1:2010
Tested to EN 61326-1:2013 and FCC Part 15 Subpart B
RoHS & WEEE compliance
PicoScope 6: All supported Windows operating systems. Beta software also available for 64-bit Linux and macOS.
PicoSDK: All supported Windows operating systems. Drivers also available for 64-bit Linux and macOS.
Users writing their own apps can find example programs for all platforms on the Pico Technology organization page on GitHub.
Processor, memory and disk space: as required by the operating system
Ports: USB 3.0 (recommended) or 2.0 (compatible)
5 years
500 mm (scope to pod)
225 mm (pod to probe tip)
75 x 55 x 18.2 mm
34.5 x 2.5 x 6.7 mm (including ground clip)

PicoScope® 6000E Series

Optional accessories
Order code
Description
MSO pods
TA369
8-channel MSO Pod for PicoScope 6000E Series
MSO pod replacement accessories
PQ221
MSO pod spares kit
TA139
MSO grabbers, pack of 12
TA365
MSO digital interface cable
Probe positioning system
TA102
Two-footed probe holder
PQ215
Probe holder and PCB holder kit, no probes
PQ219
8-channel probe holder upgrade kit with 4 probes for PicoScope 6000E Series
PQ218
4 additional probe holders
Passive probes
TA437
P2056 500 MHz 10:1 passive probe
TA480
P2056 500 MHz 10:1 passive probe dual pack
TA436
P2036 300 MHz 10:1 passive probe
TA479
P2036 300 MHz 10:1 passive probe dual pack
High-impedance active probe
TA112
TETRIS 1000 1 GHz high-impedance active probe, standard BNC connector
High-voltage differential probes
TA042
100 MHz 1400 V differential oscilloscope probe 100:1/1000:1, standard BNC connector
TA043
100 MHz 700 V differential oscilloscope probe 10:1/100:1, standard BNC connector
Power supply
PQ247
12 V 7 A PS016 PSU, IEC input, DIN output and supplied with 4 IEC mains cables (UK, EU, US and Australia/China)

USD*

EUR*

GBP*

709

599

499

41
30
35

35
25
20

29
21
25

25
369
1295
245

21
309
1095
205

18
259
909
169

235
389
199
309

199
329
169
259

165
269
139
219

989

839

679

609
659

519
559

419
459

93

79

65

PicoScope® 6000E SeriesPicoScope 6000E Series oscilloscope pack contents

TA369 MSO pod pack contents

MSO pod spares kit contents

•
•

•
•
•
•
•
•
•
•

A spares kit is available containing the following
items:• MSO ground clip 1-way (8)• MSO ground clip 4-way• MSO ground clip 8-way• MSO ground lead (8)

•
•
•
•
•
•

PicoScope 6000E Series PC oscilloscope
P2056 500 MHz 10:1 passive probes (4)(supplied with the PicoScope 6824E, 6424E, 6804E and 6404E)
P2036 300 MHz 10:1 passive probes (4)(supplied with the PicoScope 6403E)
Quick start guide
12 V power supply, universal input
Localized IEC mains lead
USB cable, 1.8 m
Storage / carry caseTA369 8-channel MSO pod
MSO test hooks (pack of 12)
MSO ground lead (8)MSO ground clip 1-way (8)MSO ground clip 4-way
MSO ground clip 8-way
MSO digital interface cable
Storage / carry case

PicoScope® 6000E SeriesPicoScope 6000E Series ordering information
Order code
PQ198
PQ201
PQ197
PQ200
PQ199

Description
PicoScope 6824E
PicoScope 6424E
PicoScope 6804E
PicoScope 6404E
PicoScope 6403E

Bandwidth (MHz)
500
500
500
500
300

Channels
84844Resolution (bits)
8 to 128 to 128
8
8

Memory (GS)
4
4
2
2
1

USD*
14 2458 725
11 395
7 305
4 865

EUR*
12 095
7 415
9 675
6 205
4 135

GBP*
9 995
6 125
7 995
5 125
3 415

USD*
179

EUR*
159

GBP*
139

Calibration service
Order code
CC051

Description
Calibration certificate for PicoScope 6000E Series oscilloscopes* Prices correct at time of publication. Sales taxes not included. Please contact Pico Technology for the latest prices before ordering.

More devices from Pico Technology...
PicoLog ADC-20/24
High-resolution andhigh-accuracy voltage
input data loggers

PicoVNA
Low-cost, professionalgrade 6 GHz vector
network analyzer forboth lab and field use

PicoScope
9400 SXRTO
Sampler-extended realtime oscilloscopes
5 to 16 GHz

United Kingdom global headquarters

North America regional office

Asia-Pacific regional office

Pico Technology
James House
Colmworth Business Park
St. Neots
Cambridgeshire
PE19 8YP
United Kingdom

Pico Technology
320 N Glenwood Blvd
Tyler
TX 75702
United States

Pico Technology
Room 2252, 22/F, Centro
568 Hengfeng Road
Zhabei District
Shanghai 200070
PR China

www.picotech.com
 +44 (0) 1480 396 395
 sales@picotech.com

www.picotech.com
 +1 800 591 2796
 sales@picotech.com

www.picotech.com
 +86 21 2226-5152
 pico.asia-pacific@picotech.com

Errors and omissions excepted.
Pico Technology, PicoScope, PicoSDK, and FlexRes are internationally registered trademarks of Pico Technology Ltd.
LabVIEW is a trademark of National Instruments Corporation. Linux is the registered trademark of Linus Torvalds, registered in the U.S. andother countries. macOS is a trademark of Apple Inc., registered in the U.S. and other countries. MATLAB is a registered trademark of The
MathWorks, Inc. Windows is a registered trademark of Microsoft Corporation in the United States and other countries.
MM105.en-4 Copyright © 2020 Pico Technology Ltd. All rights reserved.

Pico Technology@LifeAtPico

PicoSource AS108
8 GHz agile USB
controlled vector
modulating signal
synthesizer

www.picotech.com@picotechnologyltd

Pico Technology@picotech