SJ-H series Digital Scale
Operating Instructions

Consignes d’exploitation
Balance compacte numérique de série SJ-H
1WMPD4001461E

Cautions

Avoid installing the scale in direct sunlight.
 Avoid over / shock load that could cause damage to the scale.
 Use the scale under conditions of -10 °C to 40 °C, less than 85% R.H.
 Do not mix the type of batteries. Replace the batteries all together.
 If the scale is not to be used for a long period of time, remove all batteries.
 Use only specified AC adapter for the scale (AC adapter is optional).

Compliance with FCC Rules

Please note that this equipment generates, uses and can radiate radio frequency
energy. This equipment has been tested and has been found to comply with the limitsof Class A digital devices pursuant to Part 15 of FCC rules. These rules are designed toprovide reasonable protection against interference when equipment is operated in acommercial environment. If this unit is operated in a residential area, it may causesome interference and under these circumstances the user would be required to take,at his own expense, whatever measures are necessary to eliminate the interference.
(FCC = Federal Communications Commission in the U.S.A.)

Specifications

SJ-1000H
1000 x 0.5 g / 2.2 x 0.001 lb / 35 x 0.02 oz / 2.2 lb x 0.1 oz
SJ-2000H
2000 x 1 g / 4.4 x 0.002 lb / 70 x 0.05 oz / 4.4 lb x 0.1 oz
SJ-5000H
5000 x 2 g / 11 x 0.005 lb / 176 x 0.1 oz / 11 lb x 0.1 oz
SJ-5001H
5000 x 1 g / 11 x 0.002 lb / 176 x 0.05 oz / 11 lb x 0.1 oz
SJ-12KH
12 kg x 5 g / 26 x 0.01 lb / 423 x 0.2 oz / 26 lb x 1 oz
SJ-20KH
20 kg x 10 g / 44 x 0.02 lb / 705 x 0.5 oz / 44 lb x 1 oz
SJ-30KH
30 kg x 20 g / 66 x 0.05 lb / 1058 x 1 oz / 66 lb x 1 oz
Power
4 x “D” size batteries (not included) or AC adapter (option)
Battery life
Approximately 2000 hours with alkaline cells at 20 °C / 68 °F
Pan size
230 (W) x 190 (D) mm / 9.1 (W) x 7.5 (D) in.
Dimensions
265 (W) x 250 (D) x 106 (H)mm / 10.4 (W) x 9.8 (D) x 4.2 (H) in.
Keys and Symbols
ON/OFF key
Turns the scale power on/off.
RE-ZERO key
Sets the display to zero.
UNITS key
Changes weighing units, (k) g, lb, oz or lb-oz.
SymbolIndicates when the reading is stable.
Symbol ZERO
Indicates when the scale zero is correct.
Symbol NET
Indicates when net weight is displayed.
SymbolBattery indicator.

Installing batteries / Connecting the AC adapter
Installing / Replacing Batteries
Slide the battery cover downward toopen, and push the battery box tounhook and pull it out. Insert 4
batteries (D size) into the battery
box, taking extreme care of thepolarities.

AC adapter (optional)
Verify that the AC adapter is correct.
Open the rubber cap and plug theAC adapter.

Battery box

Vorsichtsmaßnahmen

Precauciones







Spécifications

SJ-1000H
1000 x 0,5 g / 2,2 x 0,001 lb / 35 x 0,02 oz / 2,2 lb x 0,1 oz
SJ-2000H
2000 x 1 g / 4,4 x 0,002 lb / 70 x 0,05 oz / 4,4 lb x 0,1 oz
SJ-5000H
5000 x 2 g / 11 x 0,005 lb / 176 x 0,1 oz / 11 lb x 0,1 oz
SJ-5001H
5000 x 1 g / 11 x 0,002 lb / 176 x 0,05 oz / 11 lb x 0,1 oz
SJ-12KH
12 kg x 5 g / 26 x 0,01 lb / 423 x 0,2 oz / 26 lb x 1 oz
SJ-20KH
20 kg x 10 g / 44 x 0.02 lb / 705 x 0.5 oz / 44 lb x 1 oz
SJ-30KH
30 kg x 20 g / 66 x 0.05 lb / 1058 x 1 oz / 66 lb x 1 oz
Puissance
4 piles de format “D” (non inclues) ou adapteur secteur (option)
Durée de vie des piles Approximativement 2000 heures avec des piles alkalines à 20 °C/68 °F
Dimensions du plateau de pesage 230 (Largeur)x 190 (Profondeur) mm / 9,05 (L) x 7,48 (P) pouces
Dimensions
265 (Largeur) x 250 (Profondeur) x 106 (Hauteur)mm / 10.4 (L) x 9.8 (P) x 4.2 (H) pouces
Cléf d’interrupteur et Symboles d’affichage
Touche ON/OFF
Mise en marche et arrêt de la balance.
Touche RE-ZERO
Met l’affichage à zéro.
Touche UNITS
Change les unités de pesage, (k)g, lb, oz ou lb-oz.
SymboleIndique quand la lecture est stable.
Symbole ZERO
Indique quand le zéro de la balance est correct.
Symbole NET
Indique quand le pesage net est affiché.
Symboleindicateur de batterie.

Mise en place des piles /Mise en marche de l’adapteur
secteur

Mise en place/Remplacement des piles
Logement des piles
Glissez le couvercle de logement despiles en bas pour ouvrir, et poussez lelogement des piles pour le décrocher etle tirer. Insérez 4piles(D) dans lelogement des piles, prenez soin
particulier aux polarités.
Couvercle des piles
Adapteur secteur(optionnel)
4 piles nouvelles
Vérifiez que l’adapteur secteur est
correct. Ouvrez le couvercle en
Ouvrir le couvercle en
caouchouc et branchez l’adapteur
caouchouc
secteur.

Prise de l’adapteur continu

Procédures de pesage
Battery cover
New four batteries(R20P / LR20 / D size)
Open rubber cap.
AC adapter plug

FunctionsOperations for parameters and items
Turn off the scale using the ON/OFF key. While pressing
Entering to the preset mode
the RE-ZERO key, press the ON/OFF key.
Selecting an item
Press the UNITS key.
Changing the parameter Press the RE-ZERO key.
Storing them and exiting Press the UNITS key until displaying.
the mode
Turn off the scale using the ON/OFF key.
List of parameters and items
Items
Parameters
Auto power off function: Disabled.
: Enabled.
or,(Pound),(Ounce),(lb-oz,),Units(Taiwan,),(HK jewelry,)
Zero tracking: Disabled.
: Enabled.
Decimal point: Dot ( ).
: Comma ( ).
The factory settings are stored into parameters.
Initialization: Not used.
: Restored them.
Error message
Overload. Remove the objects from the pan.
The scale is out of zero range or unstable when powered on.
Low battery. Replace the batteries or use the AC adapter.
High voltage. Use the specified AC adapter.
Calibration
Please visit the A&D website (https://www.aandd.jp).

.

,

EnglishInstrucciones de Operación parala balanza digital, serie SJ-H

Remarques

Évitez d'installer la balance en plein soleil.
 Évitez une charge excessive et /ou un choc qui pourrait causer des dommages à la balance.
 Utilisez la balance dans les conditions de -10 °C à 40 °C, au-dessous de 85% H.R.
 Ne mélangez pas de types de piles. Remplacez toutes les piles en même temps.
 Si la balance ne doit pas être utilisée pendant une longue période, enlevez toutes les piles.
 N’utilisez que l’adapteur secteur pour la balance. (L’adapteur secteur est optionnel).



SJ-H-Serie Digitalwaage
Bedienungsanleitung

1. Mettre en marche et arrêter la balanceAppuyez sur la touche ON/OFF pour mettre en marche la balance. La balance affiche
tous les segments pendant quelques secondes et indique alors le zéro. Appuyez denouveau sur la touche ON/OFF de nouveau pour arrêter la balance.
2. Choisir l’ unité de pesage
Appuyez sur la touche UNITS pour choisir l’unité de pesage. La balance semettra en marche avec l’unité de pesage qui avait été utilisée quand elle a étéarrêtée la dernière fois.
3. Pesage
Vérifiez que la lecture est bien zéro. Sinon, appuyez sur la touche RE-ZERO pourmettre l’affichage à zéro. Si vous utilisez un récipient pour pesage, placez le récipientvide sur le plateau de pesage et appuyez sur la touche RE-ZERO pour mettrel’affichage à zéro. Placez les objets devant être pesés sur le plateau de pesage ou dansle récipient. Quand la lecture devient stable, l’indicateur STABLE
est affiché.

FonctionsOperations for parameters and items
Entrée en mode préréglé Éteignez la balance à l’aide de la touche ON/OFF . Tout en
appuyant sur la touche RE-ZERO , appuyez sur la touche ON/OFF .
Sélection d’un élément Appuyez sur la touche UNITS .
Modification des paramètres Appuyez sur la touche RE-ZERO .
Stockage des éléments Appuyez sur la touche UNITS jusqu’à l’affichage de. Étéignez la balance à l’aide de la touche ON/OFF .
et sortie du mode
Liste des paramètres et des élémentsÉléments
Paramètres
Fonction d’extinction: désactivée.
: activée.
automatique
Unitésou,(livre),(once),(lb-oz,),(Taïwan,),(joaillerie HK,)
Suivi du zéro
: désactivé.
: activé.
Virgule décimale: point ( ).
: virgule ( ).
Initialisation
Les réglages d’usine sont stockés dans les paramètres.
: non utilisé.
: restauration des réglages.
Message d’erreur
Surcharge. Enlevez les objets excessifs du plateau de pesage.
Balance est hors le gamme de zéro ou instable en démarrage.
Piles faibles. Changez les piles ou utilisez un adapteur secteur.
Haute tension. Utilisez l’adaptateur secteur spécifié.
Etalonnage
Veuillez visiter le website A&D (https://www.aandd.jp).
French.

,Die Waage nicht im direkten Sonnenlicht aufstellen.
 Starke Erschütterungen vermeiden, da diese die Waage beschädigen können.
 Die Waage bei Temperaturen von -10 °C bis 40 °C und weniger als 85% relativer
Luftfeuchtigkeit betreiben.
 Nicht verschiedene Sorten von Batterien mischen. Die Batterien immer gemeinsam
ersetzen.
 Wenn die Waage längere Zeit über nicht verwendet werden soll, alle Batterien
entnehmen.
 Nur das vorgeschriebene Netzteil für die Waage verwenden (das Netzteil ist
Sonderzubehör).

Technische Daten

SJ-1000H
1000 x 0,5 g / 2,2 x 0,001 lb / 35 x 0,02 oz / 2,2 lb x 0,1 oz
SJ-2000H
2000 x 1 g / 4,4 x 0,002 lb / 70 x 0,05 oz / 4,4 lb x 0,1 oz
SJ-5000H
5000 x 2 g / 11 x 0,005 lb / 176 x 0,1 oz / 11 lb x 0,1 oz
SJ-5001H
5000 x 1 g / 11 x 0,002 lb / 176 x 0,05 oz / 11 lb x 0,1 oz
SJ-12KH
12 kg x 5 g / 26 x 0,01 lb / 423 x 0,2 oz / 26 lb x 1 oz
SJ-20KH
20 kg x 10 g / 44 x 0.02 lb / 705 x 0.5 oz / 44 lb x 1 oz
SJ-30KH
30 kg x 20 g / 66 x 0.05 lb / 1058 x 1 oz / 66 lb x 1 oz
Strom
4 x "D"-Batterien (nicht mitgeliefert) oder Netzteil (Option)
Batterielebensdauer Etwa 2000 Stunden mit Alkalibatterien bei 20 °C
Schalengröße
230 (B) x 190 (T) mm
Abmessungen265 (B) x 250 (T) x 106 (H) mm
Tastenschalter und Displaysymbole
ON/OFF -Taste
Schaltet den Betriebsstrom für die Waage ein/aus.
RE-ZERO -Taste
Stellt die Anzeige auf Null und tariert.
UNITS -Taste
Schaltet die Wiegeeinheiten zwischen (k) g, lb, oz oder lb-oz um.
SymbolZeigt an, wenn das Wägeergebnis stabil ist.
Symbol ZERO
Zeigt an, wenn die Nullstellung der Waage korrekt ist.
Symbol NET
Erscheint, wenn das NETTO-Gewicht angezeigt wird.
SymbolBatterieanzeige.

Einsetzen von Batterien / Anschließen des Netzteils
Einsetzen/Ersetzen von Batterien
Schieben Sie denBatteriekastendeckel zum Öffnen
nach unten und drücken Sie denBatteriekasten zum Lösen undHerausziehen. Setzen Sie 4 Batterien(Format D) in den Batteriekasten undachten Sie auf die richtige Polung.
Netzteil (optional)
Prüfen Sie, ob das Netzteil richtig ist.
Öffnen Sie die Gummikappe undstecken das Netzteil ein.

Batteriefach

Vier neue Batterien(Format R20P / LR20 / D)
Gummikappe öffnen.

Netzteilstecker
1. Ein-/Ausschalten der Waage
Drücken Sie die ON/OFF -Taste zum Einschalten. Die Waage zeigt einige Sekundenlang alle Segmente an und zeigt dann Null. Drücken Sie die ON/OFF -Taste erneut
zum Ausschalten.
2. Die Wiege-Einheit wählen
Drücken Sie die UNITS -Taste zum Wählen der Wiege-Einheit. Die Waage schaltet
mit der Wiege-Einheit ein, die bei der letzten Benutzung verwendet wurde.
3. WiegenPrüfen Sie, ob die Ablesung Null ist. Wenn nicht, drücken Sie die RE-ZERO -Taste, um das
Display auf Null zu schalten. Wenn Sie einen Behälter zum Wiegen verwenden, setzen Sieeinen leeren Behälter auf die Wiegewanne und drücken dann die RE-ZERO -Taste, um das
Display auf Null zu schalten. Legen Sie die zu wiegenden Objekte auf die Wiegewanne oderin den Behälter. Wenn die Ablesung stabil wird, erscheint die Anzeige.

Funktionen

Bedienungsvorgänge für Parameter und Elemente
Den Voreinstellungs-Modus Die Waage mit der ON/OFF -Taste ausschalten.
Die RE-ZERO -Taste gedrückt halten und gleichzeitig
aufrufen.
die ON/OFF -Taste drücken.
Ein Element wählen.
Die UNITS -Taste drücken.
Den Parameter ändern.
Die RE-ZERO -Taste drücken.
Speichern und den
Die UNITS -Taste drücken, bisangezeigt wird.
Die Waage mit der ON/OFF -Taste ausschalten.
Modus verlassen.
Liste der Parameter und Elemente
Elemente
Parameter
Automatische Ausschaltfunktion: Deaktiviert.
: Aktiviert.
Einheitenoder,(Pfund),(Unze),(lb-oz,),(Taiwan,),(HK Schmuck,)
Null-Verfolgung: Deaktiviert.
: Aktiviert.
Dezimalstelle: Punkt ( ).
: Komma ( ).
Initialisierung
Die Werkseinstellungen werden in den Parametern gespeichert.
: Nicht verwenden.
: Wiederherstellen.
FehlermeldungÜberlast. Entfernen Sie Überlast von der Wiegewanne.
Die Waage ist beim Einschalten außerhalb des Nullbereichs oder instabil.
LSchwache Batterie. Ersetzen Sie die Batterien oder verwenden Sie das Netzteil.
Hochspannung. Das angegebene Netzteil verwenden.
Kalibrierung
Bitte besuchen Sie die Website von A&D (https://www.aandd.jp).
German.

Especificaciones

SJ-1000H
1000 x 0.5 g / 2.2 x 0.001 lb / 35 x 0.02 oz / 2.2 lb x 0.1 oz
SJ-2000H
2000 x 1 g / 4.4 x 0.002 lb / 70 x 0.05 oz / 4.4 lb x 0.1 oz
SJ-5000H
5000 x 2 g / 11 x 0.005 lb / 176 x 0.1 oz / 11 lb x 0.1 oz
SJ-5001H
5000 x 1 g / 11 x 0.002 lb / 176 x 0.05 oz / 11 lb x 0.1 oz
SJ-12KH
12 kg x 5 g / 26 x 0.01 lb / 423 x 0.2 oz / 26 lb x 1 oz
SJ-20KH
20 kg x 10 g / 44 x 0.02 lb / 705 x 0.5 oz / 44 lb x 1 oz
SJ-30KH
30 kg x 20 g / 66 x 0.05 lb / 1058 x 1 oz / 66 lb x 1 oz
Alimentación
4 pilas de tamaño “D” (éstas no están incluidas) o adaptador de CA (opción)
Vida útil de pilas Aproximadamente 2000 horas con las pilas alcalinas a 20 °C / 68 °F
Tamaño del platillo de pesaje
230 (An) x 190 (Pr) mm / 9.05 (An) x 7.48 (Pr) pulgadas.
Dimensiones
265 (An) x 250 (Pr) x 106 (Al)mm / 10.4 (An) x 9.8 (Pr) x 4.2 (Al) pulgadas.
Interruptores y símbolos que aparecen en el visualizador
Interruptor ON/OFF
Interruptor destinado a encender o apagar la balanza.
Interruptor RE-ZERO
Ajusta la lectura del visualizador a cero.
Interruptor UNITS
Cambia las unidades de pesaje a (k)g, oz o lb-oz.
Indica cuando la lectura está estable.
Símbolo
Indica cuando la escala cero es correcta.
Símbolo ZERO
Indica cuando se visualiza el peso neto.
Símbolo NET
Indicador de la batería.
Símbolo

Instalación de pilas / Conexión del adaptador de CA
Instalación / Reemplazo de las pilas
Deslice la cubierta del compartimiento de Compartimiento de pilas
pilas hacia abajo para abrir elcompartimiento, y empuje la caja de pilaspara extraerla y sáquela. Inserte 4 pilas(tamaño D) en la caja de pilas, respetando
Cubierta dellas polaridades de las pilas.
compartimiento
Adaptador de CA (optativo)de pilasVerifique que el adaptador de CA es
Cuatro pilas nuevas(Tamaño R20P / LR20 / D)correcto. Abra el tapón de caucho yconecte el adaptador de CA.

Batteriefachdeckel

Wiegeverfahren

Evite instalar la balanza en un lugar expuesto a la luz solar directa.
Evite la sobrecarga o impactos que puedan causar daño a la balanza.
 Utilice la balanza a una temperatura ambiente de -10 °C a 40 °C, y humedad relativa
menos de 85%.
 No mezcle diferentes tipos de pilas. Reemplace todas las pilas con las nuevas.
 Si la balanza no va a utilizarse por un período prolongado, retire todas las pilas.
 Utilice sólo el adaptador de CA destinado a esta balanza (el adaptador de CA es optativo).
,Abra el tapón de caucho.
Tapón del adaptador de CA

Procedimientos de pesaje

1. Encendido/apagado de la balanza
Presione el interruptor ON/OFF para encenderla. El visualizador de la balanza
enciende todos los segmentos por unos segundos y luego muestra cero. Vuelva apresionar el interruptor ON/OFF para apagarla.
2. Seleccione la unidad de pesaje.
Presione el interruptor UNITS para seleccionar la unidad de pesaje. El visualizador
indica la unidad de pesaje que ha sido utilizada cuando ésta fue apagada la última vez.
3. Pesaje
Asegúrese de que la lectura sea cero. Si no, presione el interruptor RE-ZERO para
ajustarla a cero. Si desea utilizar un contenedor para pesaje, coloque el contenedor
vacío sobre el platillo de pesaje y presione el interruptor RE-ZERO para ajustar elvisualizador a cero. Coloque los objetos a ser pesados sobre el platillo o en elcontenedor. Cuando la lectura se quede estable, se visualiza STABLE.

Funciones

Operaciones con parámetros y elementos
Acceder al modo
Apague la balanza utilizando el interruptor ON/OFF .
predeterminado.
Mientras mantiene presionado el interruptor RE-ZERO ,presione el interruptor ON/OFF .
Seleccionar un elemento. Presione el interruptor UNITS .
Cambiar el parámetro.
Presione el interruptor RE-ZERO .
Almacenarlo y
Presione el interruptor UNITS hasta que aparezca.
Apague la balanza utilizando el interruptor ON/OFF .
salir del modo.
Lista de parámetros y elementos
Elementos
Parámetros
Función de apagado: Desactivado.
: Activado.
automático
Unidadeso,(libra),(onza),(libra-onza,),(tael Taiwán,),(tael Hong Kong,)
Seguimiento de cero
: Desactivado.
: Activado.
Separador decimal: Punto ( ).
: Coma ( ).
Inicialización
Los valores de fábrica se almacenan en los parámetros.
: No usados.
: Restaurados.
Mensajes de error
Sobrecarga. Retire la carga excesiva del platillo de pesaje.
La balanza está fuera de la gama cero o está inestable cuando seenciende la balanza.
Pilas débiles. Reemplace todas las pilas con las nuevas o utilice eladaptador de CA.
Alto voltaje. Utilice el adaptador de CA especificado.
Calibración
Visite por favor el sitio WEB de A&D (https://www.aandd.jp).
Spanish.

,Bilancia digitale della serie SJ-H
Istruzioni per l'uso

AvvertenzePrecauções

Evitate di esporre la bilancia alla luce solare diretta.
 Evitate di sottoporre la bilancia a sovraccarico e a forti urti, poiché si potrebbe danneggiare.
 Usate la bilancia tra -10 °C e 40 °C di temperatura e a un'umidità relativa inferiore all' 85%.
 Non usate contemporaneamente batterie di tipo diverso. Le batterie devono essere sempre
sostituite tutte insieme.
 Rimuovete le batterie qualora non intendiate usare la bilancia per lungo tempo.
 Usate esclusivamente l'adattatore di rete CA specificato (non fornito in dotazione alla bilancia).

Caratteristiche tecniche
SJ-1000H
SJ-2000H
SJ-5000H
SJ-5001H
SJ-12KH
SJ-20KH
SJ-30KH
Alimentazione:

Durata delle batterie:
Dimensioni del piatto:
Dimensioni della bilancia:
Tasti e indicazioni
Tasto ON/OFF
Tasto RE-ZERO
Tasto UNITS
Simbolo
Simbolo ZERO
Simbolo NET
Simbolo1000 x 0,5 g / 2,2 x 0,001 lb / 35 x 0,02 oz / 2,2 lb x 0,1 oz
2000 x 1 g / 4,4 x 0,002 lb / 70 x 0,05 oz / 4,4 lb x 0,1 oz
5000 x 2 g / 11 x 0,005 lb / 176 x 0,1 oz / 11 lb x 0,1 oz
5000 x 1 g / 11 x 0,002 lb / 176 x 0,05 oz / 11 lb x 0,1 oz
12 kg x 5 g / 26 x 0,01 lb / 423 x 0,2 oz / 26 lb x 1 oz
20 kg x 10 g / 44 x 0.02 lb / 705 x 0.5 oz / 44 lb x 1 oz
30 kg x 20 g / 66 x 0.05 lb / 1058 x 1 oz / 66 lb x 1 oz
4 batterie di formato “D” (non fornite in dotazione) oadattatore di rete CA (opzione)
circa 2000 ore con batterie alcaline alla temperatura di 20 °C.
230 (L) x 190 (P) mm / 9.1 (L) x 7.5 (P) in.
265 (L) x 250 (P) x 106 (A) mm / 10.4 (L) x 9.8 (P) x 4.2 (A) in.

accende e spegne la bilancia.
azzera il valore di pesatura visualizzato.
seleziona l'unità di pesatura, (k)g, lb, oz o lb-oz.
indica la stabilità della lettura.
indica che il punto di zero della bilancia è correttamente impostato.
indica che è visualizzato il peso netto.
carica residua della batteria; appare esclusivamente dopo
l'accensione della bilancia.

Inserimento delle batterie/Collegamento dell'adattatore di rete CA
Inserimento/Sostituzione delle batterie
Aprite il coperchio delle batterie facendolo
Porta-batteriescorrere verso il basso e quindi premete ilporta-batterie per sbloccarlo ed estrarlo.
Inserite nel porta-batterie 4 batterie diformato D prestando molta attenzione al
corretto allineamento delle polarità.
Adattatore di rete CA (opzione)
Verificate innanzi tutto che l'adattatore di Coperchio delle batterie
Quattro nuove batterierete CA sia del tipo corretto. Aprite il(R20P/LR20/ Formato D)
cappuccio di gomma e collegate l'adattatore
alla presa.
Aprire il cappuccio di gomma

Procedura di pesatura

Spinotto dell'adattatore CA

1. Per accendere/spegnere la bilancia
Per accenderla premete il tasto ON/OFF . La bilancia mostrerà per alcuni secondi tutte
le indicazioni e quindi il valore zero. Per spegnere la bilancia premete nuovamente iltasto ON/OFF .
2. Selezione dell'unità di pesatura
Per selezionare l'unità di pesatura desiderata premete il tasto UNITS . Alla riaccensione
la bilancia s'imposta automaticamente sull'unità di pesatura selezionata prima dell'ultimo
spegnimento.
3. Pesatura
Verificate innanzi tutto che la bilancia visualizzi il valore zero. Se non visualizza il valore
zero premete il tasto RE-ZERO . Se per la pesatura si fa uso di un contenitore, èinnanzi RE-ZERO per azzerarne la lettura. Ponete sul piatto o nel container ilmateriale da pesare. Quando la lettura diviene stabile appare l'indicatore STABLE.

Funzione

Operazioni relative a parametri e vociSpegnere la bilancia mediante il tasto ON/OFF . Mentre si
Accesso alla modalitàpreimpostata.
preme il tasto RE-ZERO , premere il tasto ON/OFF .
Selezione di una voce. Premere il tasto UNITS .
Modifica del parametro. Premere il tasto RE-ZERO .
Memorizzazione deiPremere il tasto UNITS fino a visualizzare.
parametri e chiusura
Spegnere la bilancia mediante il tasto ON/OFF .
della modalità.
Elenco di parametri e voci
Voci
Parametri
Funzione di spegnimento: disattivata.
: attivata.
automatico
Unità
o
,
(libbra),(oncia),(lb-oz,),(Taiwan,),(HK Jewelry,)
Regolazione dello zero: attivato.
: disattivato.
Separatore decimale: punto ( . ).
: virgola( , ).
Inizializzazione
Le impostazioni di fabbrica vengono memorizzate nei parametri.
: non utilizzata.
: ripristinate.
Messaggio d'errore
Sovraccarico. Rimuovete dal piatto il peso in eccesso.
La bilancia non è correttamente azzerata oppure all'accensione è instabile.
Batterie scariche. Sostituitele oppure usate l'adattatore di rete CA.
Alta tensione. Utilizzare l'adattatore di rete CA specificato.
Calibrazione
Vi preghiamo di visitare il sito A&D all'indirizzo (https://www.aandd.jp).
ItalianBalança Digital da Série SJ-H
Instruções de Operação

Evite instalar a balança sob a luz direta do sol.
 Evite sobrecargas e impactos que poderiam danificar a escala.
 Utilize a balança nas condições entre -10 °C e 40 °C, com menos de 85% de umidade relativa.
 Não misture pilhas de tipos diferentes. Troque todas as pilhas ao mesmo tempo.
 Se a balança não for ser utilizada durante um longo período de tempo, retire as pilhas.
 Utilize somente o adaptador de CA especificado para a balança (o adaptador de CA é
opcional).

Especificações
SJ-1000H
SJ-2000H
SJ-5000H
SJ-5001H
SJ-12KH
SJ-20KH
SJ-30KH
Energia
Vida útil das pilhas
Tamanho do prato
Dimensões

1000 x 0,5 g / 2,2 x 0,001 lb / 35 x 0,02 oz / 2,2 lb x 0,1 oz
2000 x 1 g / 4,4 x 0,002 lb / 70 x 0,05 oz / 4,4 lb x 0,1 oz
5000 x 2 g / 11 x 0,005 lb / 176 x 0,1 oz / 11 lb x 0,1 oz
5000 x 1 g / 11 x 0,002 lb / 176 x 0,05 oz / 11 lb x 0,1 oz
12 kg x 5 g / 26 x 0,01 lb / 423 x 0,2 oz / 26 lb x 1 oz
20 kg x 10 g / 44 x 0.02 lb / 705 x 0.5 oz / 44 lb x 1 oz
30 kg x 20 g / 66 x 0.05 lb / 1058 x 1 oz / 66 lb x 1 oz
Pilha de tamanho “D” x 4 (não incluídas) ou adaptador de CA (opção)Aproximadamente 2000 horas com pilhas alcalinas a 20 °C
230 (L) x 190 (P) mm / 9,05 (L) x 7,48 (P) polegadas
265 (L) x 250 (P) x 106 (A) mm / 10,4 (L) x 9,8 (P) x 4,2 (A)polegadas
Teclas e símbolos do mostrador
Tecla ON/OFF
Liga/desliga a balança.
Tecla RE-ZERO
Reposiciona o mostrador a zero.
Tecla UNITS
Muda as unidades de pesagem, (k) g, lb, oz ou lb-oz.
Símbolo
Indica quando a leitura está estável.
Símbolo ZERO
Indica quando a reposição a zero da balança está orreta.
Símbolo NET
Indica quando o peso líquido está exibido.
SímboloNível restante de energia das pilhas, mostrado somente
quando a balança é ligada.

Instalação das pilhas / Conexão do adaptador de CA

Instalação e troca das pilhas
Deslize a tampa do compartimento das
Caixa das pilhas
pilhas para baixo para abri-la, e empurrea caixa das pilhas para desbloquear eretirá-la. Coloque 4 pilhas (tamanho D)
na caixa de pilhas, tomando muito
cuidado com as polaridades.
Tampa do compart
Adaptador de CA (opcional)
das pilhas
Quatro pilhas novas
Certifique-se de que o adaptador de CA(R20P / LR20 / Tamanho D)
seja do tipo correto. Abra a cobertura deborracha e conecte o adaptador de CA.
Abra a cobertura de borracha
Plugue do adaptador de CA

Procedimentos de pesagem

1. Ligue / desligue a escala.
Pressione a tecla ON/OFF para ligar a balança. A balança exibe todos ossegmentos durante alguns segundos e, em seguida, mostra zero. Pressione a tecla
ON/OFF de novo para desligar a balança.
2. Selecione a unidade de pesagem
Pressione a teclas UNITS para selecionar a unidade de pesagem A balança éligada com a unidade de pesagem que foi utilizada na última vez que foi desligada.
3. Pesagem
Certifique-se de que a leitura seja zero. Se não for, pressione a tecla RE-ZEROpara reposicionar o mostrador a zero. Se você utilizar um recipiente para pesagem,
coloque o recipiente vazio no prato de pesagem e pressione a tecla RE-ZERO para
reposicionar o mostrador a zero. Coloque os objetos a serem pesados no prato ou
no recipiente. Quando a leitura se estabilizar, o indicador STABLE
será exibido.

Funções

Operações para parâmetros e itens
Entrar no modo
Desligue a balança utilizando a tecla ON/OFF . Mantendo
premida a tecla RE-ZERO , prima a tecla ON/OFF .
predefinido.
Selecionar um item.
Prima a tecla UNITS .
Mudar o parâmetro.
Prima a tecla RE-ZERO .
Guardá-los e
Prima a tecla UNITS até visualizar.
Desligue a balança utilizando a tecla ON/OFF .
sair do modo.
Lista dos parâmetros e itens
Itens
ParâmetrosFunção de desactivação: Desactivada.
: Activada.
automática
Unidadesou,(Libra),(Onça),(lb-oz,),(Taiwan,),(HK jewelry,)
Rastreio zero
: Desactivado.
: Activado.
Ponto decimal: Ponto ( ).
: Vírgula ( ).
Inicialização
As definições de fábrica são guardadas nos parâmetros.
: Não utilizadas.
: Restauradas.
Mensagem de erro
Sobrecarga. Retire a carga excedente do prato de pesagem.
A balança está fora do intervalo zero ou instável após ser ligada.
Pilhas fracas. Troque as pilhas ou utilize o adaptador de CA.
Alta tensão. Utilize o adaptador de CA especificado.
Calibração
Visite o site da A&D na Web (https://www.aandd.jp).
Portuguese.

,Цифровые весы серии SJ-HИнструкция по использованию

SJ-H 系列電子秤使用說明書Внимание

注意事項





Избегайте располагать весы в месте воздействия прямых солнечных лучей.
Избегайте перезагружать или резко загружать весы, так как это может привестик повреждению весов. ( Используйте весы в условиях-10 °C до 40 °C, меньше
85% относительной влажности.) Не смешивайте разные типы батареек. Производите замену сразу всех батареек.
 В случае длительного перерыва в использовании весов достаньте все батарейки из весов.
 Используйте только специальный адаптер питания для весов (адаптер питания
продается отдельно).
Спецификации
SJ-1000H
SJ-2000H
SJ-5000H
SJ-5001H
SJ-12KH
SJ-20KH
SJ-30KHПитание

1000 x 0.5 гр / 2.2 x 0.001 фунт / 35 x 0.02 oz / 2.2 фунт x 0.1 унция
2000 x 1 гр / 4.4 x 0.002 фунт/ 70 x 0.05 унция/ 4.4 фунт x 0.1унция
5000 x 2 гр / 11 x 0.005 фунт/ 176 x 0.1 унция/ 11 фунт x 0.1унция
5000 x 1 гр / 11 x 0.002 фунт/ 176 x 0.05 унция/ 11 фунт x 0.1унция
12 кг x 5 гр / 26 x 0.01фунт/ 423 x 0.2 унция/ 26 фунт x 1 унция
20 кг x 10 гр / 44 x 0.02 фунт / 705 x 0.5 унция / 44 фунт x 1 унция
30 кг x 20 гр / 66 x 0.05 фунт / 1058 x 1 унция / 66 фунт x 1 унция
4 батарейки размера “D” (не прилагаются ) или адаптер питания
(продается отдельно)Срок службы батареек Приблизительно 2000 часов для щелочных батареек при 20 °C / 68 °FРазмер чашки весов 230 (Ширина) x 190 (Длина) мм / 9.05 (Ширина) x 7.48 (Длина) дюйм.
Измерения
265 (Ширина) x 250 (Длина) x 106 (Высота) миллиметр /10.4 (Ширина) x 9.8 (Длина) x 4.2 (Высота) дюймКнопки и символы Дисплея
Кнопка ON/OFFВключает/выключает питание весов.
Кнопка RE-ZEROУстанавливает дисплей на ноль.
Кнопка UNITSМеняет единицы веса (к)г, фунты, унции или фунты-унции.
Символ
Загорается при стабильном показателе.
Символ ZEROЗагорается при правильности нулевой точки.
Символ NET
Загорается при отображении веса нетто.
Символ
Уровень зарядки батареек, отображается только после
включения питания.

Установка батареек/ Подсоединение адаптера питания.
Установка / Замена батареек.
Сместите крышку батареек вниз, надавитена отделение батареек и снимите его. Отсек для батареек
Вставьте 4 батарейки (размер D) вотделение батареек. Обратите особое
внимание на совпадение полюсов.
Адаптер питания
(продается отдельно)Крышка отсек дляУбедитесь в соответствии адаптера
батареек
питания параметрам вашей сети.
Четыре новые батарейки
Откройте резиновую крышку и(R20P / LR20 / размер D)подсоедините адаптер питания.
Откройте резиновую крышку

Процесс взвешивания

Подсоединение адаптера
питания
1. Turn on / off the scaleНажмите кнопку ON/OFF для включения. Весы отобразят все сегменты нанесколько секунд и покажут ноль. Нажмите кнопку ON/OFF снова длявыключения.
2. Выберите единицу взвешивания
Нажмите кнопку UNITS для выбора единицы взвешивания. Весы начнут
взвешивать в той единице измерения, которая была выбрана перед выключением впоследний раз.
3. Взвешивание
Убедитесь в показателе ноль. В противном случае нажмите кнопку RE-ZEROдля отображения нуля на дисплее. В случае использования упаковки длявзвешивания установите пустую упаковку на чашу весов и нажмите кнопку
RE-ZERO для отображения нуля на дисплее. Поместите объекты длявзвешивания на чашу весов или в упаковку. При стабилизации показателя
отобразится индикатор STABLE.

Функции

Операции для параметров и пунктов
Режим предустановки Выключите весы, используя кнопку ON/OFF .
Удерживая нажатой кнопку RE-ZERO , нажмите кнопку
ON/OFF .
Выбор пункта
Нажмите кнопку UNITS .
Смена параметра
Нажмите кнопку RE-ZERO .
Сохранение настроек и Нажимайте на кнопку UNITS , пока не появится
.
Выключите весы, используя кнопку ON/OFF .
выход из режима
Список параметров и пунктов
Пункты
Параметры
Автоматическое
: Отключено.
: Включено.
отключение питания
Единицы измерения
or,(Фунт),(Унция), (фунт-унция,),(Бас (Тайвань),),(Таел (Гонконг),)Отслеживание нуля: Отключено.
: Включено.
Десятичная точка
: Точка ( ).
: Запятая ( ).
Настройки по умолчанию сохранены в параметрах.
Инициализация
: Не используются.
: Восстановлены.
Сообщения об ошибках
Перегруз. Удалите лишний груз с чаши весов.
Нулевая точка смещена или нестабильны при включенном питании.
Слабые батарейки. Замените батарейки или используйте адаптер
питания.
Высокое напряжение. Используйте указанный адаптер питания.
Проверка
Посетите сайт (https://www.aandd.jp).

.

,

Russian

請避免將磅秤放置在陽光直射的地方。
請避免超重或負荷衝擊致使磅秤本身的損壞。
 請在攝氏-10 °C ~ 40 °C，相對濕度不滿 85%的條件下，使用本磅秤。
 請勿混用不同類型的電池，須同時全部一起更換。
 若長期不使用，請將所有的電池卸下來。
 僅指定使用 AC 變壓器 (AC 變壓器為選購配件)。
規格
SJ-1000H
SJ-2000H
SJ-5000H
SJ-5001H
SJ-12KH
SJ-20KH
SJ-30KH
電源
電池壽命
秤台尺寸
體積
按鈕和顯示符號ON/OFF 鍵
RE-ZERO 鍵
UNITS 鍵
符號
ZERO
符號
NET
符號
符號

1000 x 0.5 公克/ 2.2 x 0.001 磅/ 35 x 0.02 盎司/ 2.2 磅 x 0.1 盎司2000 x 1 公克/ 4.4 x 0.002 磅/ 70 x 0.05 盎司/ 4.4 磅 x 0.1 盎司
5000 x 2 公克/ 11 x 0.005 磅/ 176 x 0.1 盎司/ 11 磅 x 0.1 盎司
5000 x 1 公克/ 11 x 0.002 磅/ 176 x 0.05 盎司/ 11 磅 x 0.1 盎司
12 公斤 x 5 公克/ 26 x 0.01 磅/ 423 x 0.2 盎司/ 26 磅 x 1 盎司
20 公斤 x 10 公克/ 44 x 0.02 磅/ 705 x 0.5 盎司/ 44 磅 x 1 盎司
30 公斤 x 20 公克/ 66 x 0.05 磅/ 1058 x 1 盎司/ 66 磅 x 1 盎司
4 x “D” 尺寸電池 (不包含) 或 AC 變壓器 (選購配件)
大約 2000 小時鹼性電池，在攝氏 20 °C /華氏 68 °F 條件下230 (寬) x 190 (長) 毫米 / 9.05 (寬) x 7.48 (長) 英吋
265 (寬) x 250 (長) x 106 (高)毫米 / 10.4 (寬) x 9.8 (長) x 4.2 (高) 英吋
開啟磅秤的電源 開 / 關
設定顯示歸零更換重量單位(公斤)公克, 磅, 盎司或磅-盎司當讀數穩定時顯示顯示磅秤歸零顯示淨重顯示剩餘電量，僅在打開電源後顯示

安裝電池 / 連接 AC 變壓器

電池盒

更換電池
滑動打開電池盒蓋並按下電池盒，使固定鉤拖開。插進 4 個電池(D 尺寸) 於電
池盒， 並請注意電池的極性。
AC 變壓器 (選購配件)
請確認 AC 變壓器是否正確， 打開橡膠蓋並插入 AC 變壓器。

電池蓋

4 個新電池(R20P / LR20 / D 尺寸)
打開橡膠蓋

測量順序

AC 變壓器1. 開啟磅秤的開關
按下 ON/OFF 按鍵到開的位置。 數秒鐘之間顯示磅秤的全區域，然後顯示歸零。
再按一下 ON/OFF 按鍵關閉電源。
2. 選擇測量重量單位
按下 UNITS 按鍵用以選擇重量單位。 磅秤將顯示上次所使用的重量單位。
3. 測量
確認歸零。 如果不是零,按一下 RE-ZERO 按鍵顯示歸零。 如果使用容器秤重，請將空容器放在秤台上並按下 RE-ZERO 按鍵顯示歸零。
請將秤重的東西放在秤台上或容器中，並確認出數字變化穩定，而顯示出 STABLE
指示器。

機能
Operations for parameters and items
輸入至預先設定模式。
使用 ON/OFF 按鍵，關閉磅秤。按下 RE-ZERO 按鍵時，請同時按下 ON/OFF 按鍵。
選擇項目。
按下 UNITS 按鍵。
變更參數。
按下 RE-ZERO 按鍵。
進行存儲並離開該模式。 請按下 UNITS 按鍵，直到顯示。
使用 ON/OFF 按鍵，關閉磅秤。
參數與項目清單項目
參數自動關閉電源功能
：已停用。
：已啟用。
單位
或
，
(磅)，
(盎司)，
(磅-盎司，)，(台灣，)，(香港珠寶，
)
零位追蹤
：已停用。
：已啟用。
小數點
：點 ( )。
：逗號 ( )。
初始化
出廠設定已存儲至參數。
：未使用。
：回復至出廠設定。
錯誤訊息
超重 請取下秤台上的超重物品。
打開電源時磅秤超出了零的範圍或不穩定。
電池電量不足，請更換電池或使用 AC 變壓器。
高電壓。請使用指定的 AC 變壓器。
校正請瀏覽 A&D 網站 (https://www.aandd.jp)。

.

,

Chinese