Specification List of RL Series
LIST OF SPECIFICATION
Consumption VA &Product

DC AmmeterMax. Input

Model& Rated

Voltage

Current

Response time (sec)
L-110C

L-80C

Attached
Transducer

0.15

MRL-110C□

20mA－below 500Ω

MRL-80C□

50mA

－500Ω

MTRL-110□, 80C□

100µA~10A

－

60mV

0.2

0.15,0.2－

MRL-110C□, 80C□

15V~300V

20mA

－

0.2

0.15,0.2－－

0.2

0.15,0.2－

DC Voltmeter
MTRL-110C□,80C□

AC Ammeter

Internal Resistance1kΩ

500mV~14V

10ｋΩ

－0.2

*1

CRL-110C□
CRL-80C□

60mV~200mV

0.2*11A~10A－

below 6VA

0.1, 0.15, 0.2－

0.5 ~30*2

AC Voltmeter

DCReceiving
Indicator

CRL-110C□
CRL-80C□*2－

below 500Ω

XRL-80C□

50mA－500Ω

20mA－

YRL-110C□
YRL-80C□

YRL-110C□

Singlephase

3 phasePowerFactor
Meter

－

0.1, 0.15, 0.2
0.5~300.2

0.15,0.2

Singlephase
3 phase(balance)
Singlephase

3 phaseMRL-80CHL response
time: 0.2 sec. only
Specify aux. power
AC100V or AC200V
MRL-80CHL response
time: 0.2 sec. only
Specify aux. power
AC100V or AC200V*1) CRL-80 response
time: 0.1, external
AT-62M, response time:
1~15, external MR-CTN
80 degree: below 15 sec*2) CRL-80 response
time: 0.1,1~15, external
VT-62M,
80 angle: below 15 sec.
0.15~0.5 sec. external
DH-41 (Except Vo Meter)－－

XRL-80CHL response
time: 0.2 sec only.

15V~300V

0.2－

0.15, 0.2
*3

1A~10A－－

below 6VA
0.1, 0.15, 0.2

YRL-80C□

VarMeter－20mA

XRL-80C□

WattMeter

below 6VA

XRL-110C□

XRL-110C□

AC
Receiving
Indicator

100~300V

Notes*40.5~30
100~300V

below 6VA－2VA

1VA

WRL-110C□-12110V, 5A

WRL-80C□-12220V, 5A

WRL-110C□-33110V, 5A

Each phase

Each phase

WRL-80C□-33220V, 5A

2VA

1VA

WVRL-110C□-12110V, 5A

WVRL-80C□-12220V, 5A

2VA

1VA

WVBRL-110C□-33110V, 5A

Each phase

Each phase

WVBRL-80C□-33220V, 5A

2VA

1VA

PRL-110C□-12110V, 5A

PRL-80C□-12220V, 5A

3VA

1VA

PRL-110C□-33110V, 5A

Each phase

Each phase

PRL-80C□-33220V, 5A

3VA

1VA－*3) YRL-80 response time:0.1, external AT-62M,response time: 1~15,external MR-CTN,
80 degree: below 15 sec.
*4) YRL-80 response time:
0.1,1~15, external
VT-62M,
80 degree: below 15 sec.
0.15~0.5 sec. external
DH-41 (except Vo Meter)

0.5~30

WT-83M-12－

0.5~30

WT-83M-33－0.5~30

WVT-83M-12

Specify Frequency

0.5~30WVBT-83M-33

Specify Frequency

2~30

PT-83M-12

Specify Frequency

2~30

PT-83M-33

Specify Frequency

6

Specification List of RL Series
LIST OF RL SERIESMODELTypeRL – 110CPrinciple

DC Ammeter

Moving coil

DC Voltmeter
AC Ammeter

Rectifier

AC Voltmeter

RL – 80C

ModelClass

Record PointerAccuracy

ModelClass

Record PointerAccuracy

MRL-110C□

1.5± 2.0%

MRL-80C

1.5± 2.0%

MRL-110C□

1.5± 2.0%

MRL-80C

1.5± 2.0%

CRL-110C□

1.5± 2.0%

CRL-80C

1.5± 2.0%

CRL-110C□

1.5± 2.0%

CRL-80C

1.5± 2.0%DC Receiving Indicator

Moving coil

XRL-110C□

1.5± 2.0%XRL-80C

1.5± 2.0%AC Receiving Indicator

Rectifier

YRL-110C□

1.5± 2.0%

YRL-80C

1.5± 2.0%Singlephase

Transducer

WRL-110C□-121.5± 2.0%

WRL-80C-121.5± 2.0%

WRL-110C□-33

1.5± 2.0%

WRL-80C-33

1.5± 2.0%

WVRL-110C□-121.5± 2.0%

WVRL-80C-12

1.5± 2.0%WVRL-110C□-33

1.5± 2.0%WVRL-80C-33

1.5± 2.0%

PRL-110C□-125.0± 6.0%

PRL-80C-12

1.5± 6.0%

PRL-110C□-335.0± 6.0%

PRL-80C-33

1.5± 6.0%

Watt Meter

3 phaseVar Meter
PowerFactorMeter

Singlephase
3 phaseSinglephase

Transducer

Transducer

3 phaseConnection Diagram
DC Ammeter

DC Ammeter external with Shunt(Type inner with amplifier)

DC Voltmeter

DC Voltmeter(Type inner with amplifier)AC Ammeter

AC Ammeter(External with AT-62M)

7

Specification List of RL Series
Connection Diagram
AC Voltmeter(External with VT-62M)

AC Voltmeter3 phase Wattmeter3 phase Var Meter
3 phase Power Factor Meter

Single phase Wattmeter
Single phase Var Meter
Single phase Power Factor Meter

Dimensions

Dimensions For Accessory Transducer

AT-62M, VT-62M

WT-83M, WVT-83M, PT-83M

TypeA

BC

DE

F1

F2

G1G2weight (kg)RL-110C

90110

30

167 (MTRL: 169)1599Φ101Φhole

M5 screw

7Φhole1.1

RL-80C

6480

22.5

168 (MTRL: 169)1065Φ67ΦholeM4 screw

5.5Φhole

1.08