User Manual

PROFESSIONAL

Model No.

Dial Indicator / Comparator

DIAL INDICATOR

WTI-＊＊＊＊＊

Tight places where plunger type Dial Gauge can not be used.
Use on milling machines and laths for centering, measure runout, parallelism,
flatness, etc.
●
Mount on height gauge for various measurements.
●Always follow the procedures specified below in order to prevent harm to yourself or others, and toprevent damage to property.

WARNING Indicates risk of personal injury or property damage if not followed.
■These symbols mark content that must be observed.
Denotes a prohibition – You MUST NOT do．

●●For safe and proper use of this product, please read thisinstruction manual before use and follow the procedures
described. Please keep manual where it is accessible touser for future reference.
●Keep this manual with the instrument if transferred orleased to a third party.
●For inquiries about this product, please contact distributor
or place of purchase.

Please Observe■Content marked as follows indicates risk of injury or damage if not followed.

Thank you for purchasing the Niigata Seiki Dial Indicator.
Used with a Magnetic Base or Indicator Stand, this gauge will show the difference in height or position
relative to a zero point set at a reference position.

■APPLICATIONSSAFETY PRECAUTIONSSAFETY NOTIFICATIONS

Throughout this manual, “ ” symbol indicates
RISK OF PERSONAL INJURY OR PROPERTYDAMAGE if not followed.
The “” symbol indicates something which is
PROHIBITED, and the “” symbol Indicates
REQUIRED step or necessary condition.

Denotes a requirement – You MUST do．

CAUTION
Read the manual and follow all instructions.

Do not disassemble or modify.

・Use of product other than as described in themanual may cause accident.

・It may damage Gauge and cause pooraccuracy.
・If Bezel Clamp Screw is removed, internal
components may come loose and become
misaligned causing product failure.

Use only as indicator Gauge.

・Use for any purpose other than measuring
may damage or wear the instrument.
Improper use may also cause accident.

Do not shock Probe.

・Rapid motion, or lateral force may damage
Gauge and cause poor accuracy.

Use in an environment which meets thefollowing conditions:● Temperature within range of 0~40°
C、■PART IDENTIFICATION, FUNCTION

humidity 30~70% (non-condensing.)● Location with minimal dust, oil, oil mist,※Model WTI-8038S shownand protected from direct sunlight.

● Location protected from use by children

and unauthorized people.
・Use in location contrary to the above may
cause poor accuracy, damage to the product,
or may result in accident or injury.

⑤ Dovetails① Pointer(Needle)

Handle With Care.

・Do not drop or subject to shock, do not place
under heavy objects. Damage may cause
failure or poor accuracy.

② Scale plate④ Bezel[Accessories]⑥ Stem(φ6･φ8mm)

Gauge Holder

Make sure Gauge holder is rigid.

⑧ Dovetail groove①Pointer ……………… Shows measurement on the Scale Plate②Scale plate …………… Graduation. Turn Bezel to rotate③Probe ………………… Feeler which contacts workpiece④Bezel ………………… Rotate to turn Scale Plate⑤Dovetails ……………… For attaching Stem
⑥Stem ………………… For mounting to Holder⑦Mounting bracket …… Connects Stem to gauge⑧Dovetail groove ……… Matches Dovetail in gauge body for mounting Stem
⑨Spanner ……………… Tool to change Probe

PREPARATION - Mounting
Dial Indicator must be securely mounted such as on a comparator stand or magnetic base.
Please follow these guidelines.

⑦ Mounting bracket③ Probe (Stylus)⑨ Spanner (for Probe)

・Mounting of gauge by other than Stem or Dovetail will cause
inaccuracy and product damage.
※Mounting Example- 2-

TROUBLESHOOTING⑤ Dovetails⑤ Dovetail

Insert⑧ Dovetailgroove②Confirm that the Probe and stem are secure
without any play.

⑦ Mountingbracket

If they are loose, tighten to secure.

③Attach the Indicator Gauge to the Holder.

Slightly Loosen⑤Make sure the Probe is parallel to themeasured surface, and perpendicular to thedirection of the measurement

Direction ofMeasurement※Rotate to loosen

Direction ofMeasurement90°

If probe is not positioned properly, the measurement
will be inaccurate and operation may be faulty.

Workpiece

Direction ofMeasurement

Workpiece90°

Workpiece

Direction ofMeasurement

Workpiece

HOW TO USE - Comparison Measurements①Set Up Reference Part.

Ex.) Reference Part 30mm Gauge Block : 30mm
Workpiece height : 30.08mm

Position the reference Part with the measured
surface parallel to the probe and perpendicular to thedirection of the measurement. Bring into contact withthe probe, but use care that the part does not bumpthe probe from the side.

・Temperature changes during measurement can cause repeatability error. Please try the following solutions:●Use in location with constant temperature.
●When taking measurements, periodically adjust zero point using a Master reference to correct for temperatureinduced drift.
・Make sure the Stem is not loose, and tighten if necessary.

■Measurement is not stable, or measurement accuracy is poor.

・Probe may be worn; a worn Probe will affect accuracy. Periodically check for wear, and replace if wear is affecting
measurement accuracy.

(φ6･φ8mm)

InsertBring the Probe into contact with the workpiece, andgently raise and lower the gauge while checking themovement. Confirm that the pointer returns to initial
position when raised.

■Origin position shifts during measurement.

SPECIFICATIONS

InsertPlease attach by the Stem only.

④Confirm that the pointer moves freely.

Minimize Length

Dial Gauge must only be attached by Dovetail or Stem.

HOW TO USE - Preparation
Loosen the Stem mounting bracket and slide thegroove onto the Dovetail in the desired position.
Tighten the Stem to secure. (Choose a stem withdiameter to match the Gauge Holder you plan touse.)

Support Arm⑥ Stem・Holder must be sufficiently secure to prevent deflection
from the weight of the Gauge.
・Holder must be rigid enough to hold Gauge and not tolift from measurement force.
・Holder support arm should be as short as possible toprevent deflection.
※Deflection or lifting will cause measurement error, suchas origin position error and inaccuracies in measured
reading.

- 1-①Attach Mounting Stem.

Measurement Direction〈Set origin with Reference Part〉〈Measure workpiece〉① Pointer②Set the Origin.

Model No.

WTI-2030R
WTI-2030W
WTI-2038R
WTI-2038W
WTI-8030S
WTI-8030R
WTI-8030W
WTI-8038S
WTI-8038R
WTI-8038W

Graduation Measurement
Range
(mm)
(mm)

Scale

Error of indication Hysteresis
over a range ofError
Measuring range
(μm)
(μm)

Measuring Repeatability
Force
(μm)
(N)

0.002

0.2

0-100-062≦0.5

10.0020.20-100-062

≦0.5

10.01

0.8

0-40-0133

≦0.5

30.01

0.80-40-0133≦0.5

3Weight(g)

Probe
Ruby
Carbide Ball
Ruby
Carbide Ball
Steel Ball
Ruby
Carbide Ball
Steel Ball
Ruby
Carbide Ball

70807080●Replacement Probe Part No. : WTI-2030R･WTI-2038R…WTI-20R、WTI-2030W･WTI-2038W…WTI-20W
WTI-8030S･WTI-8038S…WTI-80S、WTI-8030R･WTI-8038R…WTI-80R、WTI-8030W･WTI-8038W…WTI-80W

AFTER USE CARE, STORAGE
Remove any dust or dirt after use. ※Do not lubricate.
・Wipe any contamination from Probe and moving surfaces using
a dry cloth, or cloth moistened with alcohol.
・To clean other surfaces, wipe with a soft dry cloth, or a clothmoistened with a mild cleaner.

Replacing Probe
Please use Spanner supplied with gauge.
⑨ Spanner
Slot

Check for wear of Probe tip.

・Measurement accuracy will be affected by worn Probe.
Regularly check for wear and replace Probe if worn.

TightenLoosen

Store in provided case in a cool, dark, and dry location.

Adjust the gauge mount or rotate the Bezel to set theGauge to “0”.

③Remove the Reference Part and Measure theWorkpiece.

Remove the master with care not to shock the gauge.
Replace with the workpiece and read the scale.

② Scale plate④ BezelHOW TO USE - Parallelism, Flatness, Runout, etc.
①Position the probe on the workpiece.

Position the Part with the measured surface parallel to the probe and perpendicular to the direction of themeasurement. Bring into contact with the probe, but use care that the part does not bump the probe from the side.

②Set the Origin.

Adjust the gauge mount or rotate the Bezel to set the Gauge to “0”.

③Move the workpiece and read the scale.

・During storage, make sure there is no force on the Probe (suchas pushed in, or lateral force.)
・Keep away from moisture and direct sunlight, and secure fromunauthorized personnel.

Use only factory approved Probe. Using alternate probe
may cause error.
・Always perform calibration after replacing Probe.

Screw

CALIBRATION
To maintain measurement accuracy, periodic calibration is recommended
Wear of measuring surfaces from repeated use may affect accuracy andperiodic accuracy checks should performd.
Outside Japan: Please contact distributor or place of purchase to inquire
about calibrations services.

Move the workpiece slowly while reading the position of the pointer on the scale.

- 3-Adjusting Length After Changing Probe※WTI-2030R・WTI-2038R
WTI-2030W・WTI-2038W only
Rotate Probe to expose screw as shown.
Using a screwdriver, turn screw andadjust Probe length.

- 4-Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

業務用

取扱説明書

てこ式精密比較測定器

ダイヤルインジケータ

WTI -＊＊＊＊＊

品番

必ずお守りください。

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。
■表示内容を無視して、誤った使い方をした時に生じる危害や損害の程度を次の表示で区分しています。

この度は「ダイヤルインジケータ」をお買上げいただきありがとうございます。この商品はマグネットベースやコンパレータスタンドなどの保持具に固定して、任意の基準地点を０とした時の測定値の増減を読み取る比較測定器です。

■用途の一例

安全上のご注意

注意 『傷害を負う、または物的損害が発生するおそれがある内容』です。

■お守りいただく内容の種類を次の図記号で区分しています。
してはいけない内容『禁止事項』です。

標準形のダイヤルゲージが使用できない狭い場所での測定旋盤・フライス盤に取り付けて、内外側・ミゾ幅・芯出し・平行度・平面度などの測定● ハイトゲージに取り付けての各種測定必ず実行していただく内容『強制事項』です。

注意●●●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。

取扱説明書をよく読み、指示に従う。

安全上のお知らせこの説明書には拡大損害が予想される事項には、禁止事項には、必ずお守りいただきたい強制事項にはマークを付けて表示しています。

●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。

分解、改造を行わない。

・取扱説明書に記載された内容以外での使用は、事故の原因となります。

・精度不具合や製品の破損の原因となります。
・止めねじを外すと、内部部品の歯車の噛み合いがズレて製品不具合の原因となる場合があります。

精密比較測定作業にのみ使用する。

・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。

測定子に衝撃を与えない。

・測定子に測定方向以外から必要以上の力を加えないでください。
精度不具合や故障、破損の原因となります。

下記の条件を満たす環境で使用する。

●温度０〜４０℃、湿度３０〜７０％の結露しない場所●埃、油、オイルミストが少なく、直射日光が当たらない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は、精度不良や製品の破損、事故やけがの原因となります。

●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称と機能 ※イラストはWTI-8038Sです。

本器は大切に取り扱う。

⑤ アリ板

・落下などの衝撃を与えたり、重量物を乗せたりしないでください。精度不具合や故障、破損の原因となります。

① 指針

② 目盛板

ご使用の前に[付属品]
④ 外枠

⑥ ステム

⑦ 取付金具（φ6･φ8mm）

ダイヤルインジケータはマグネットベースやコンパレータスタンドなどの保持具で保持して使用します。
取り付けの際は下記の内容をお守りください。

保持具のたわみ・浮きに注意する。

⑧ アリ溝

・保持具は十分に剛性のあるものを使用し、ダイヤルインジケータの重みによる保持具のたわみを防いでください。
・保持具はダイヤルインジケータの測定力に対して、十分に固定可能なものを使用し、測定時の保持具の浮きを防いでください。
・保持具の支柱はなるべく短く設定し、支柱のたわみを防いでください。
※たわみ、浮きは原点位置ズレや指示値のバラつきなど測定誤差の原因となります。

⑨ スパナ（測定子交換用）③ 測定子①指 針 ……… 目量を表す針です。
②目盛板 ……… 目量を表す目盛です。外枠を回すと回転します。
③測定子 ……… マスターやワークに当てる測定部です。
④外 枠 ……… 目盛板を回転させます。

測定方向⑤アリ板 ……… ステムを取り付ける部分です。
⑥ステム ……… 保持具取付部です。
⑦取付金具 …… 本体取付部です。
⑧アリ溝 ……… ステムを本体に取り付けるための溝です。
⑨スパナ ……… 測定子交換用です。

支柱

⑥ ステム

保持具

なるべく短く保持具への取り付けは、ステムもしくはアリ板で行う。
・ステム、アリ板以外での取り付けは、精度不具合や製品の破損の原因となります。

※保持例- 1-- 2-

使用方法（準備）

トラブルシューティング①ステムを取り付ける⑤ アリ板

ステムの取付金具を少しゆるめ、アリ板にアリ溝を通して差し込み、任意の位置でステムを締めて固定してください。
（ステム径は保持具の仕様に合わせて選んでください。）

⑤ アリ板

差し込む

⑧ アリ溝

②測定子とステムにゆるみが無いことを確認する

ゆるみがある場合は、全てのねじを締め直してください。

⑦ 取付金具③保持具にダイヤルインジケータを取り付ける

差し込むステムで保持具に取り付けてください。

④指針の動きが滑らかであることを確認する

差し込む

ワークに測定子を接触させ、ダイヤルインジケータ本体を軽く上下させ、動きを確認してください。また、測定子を上下させた後、指針が最初の位置に戻ることを確認してください。

⑥ ステム

少しゆるめておく
少しゆるめておく（φ6･φ8mm）※取付金具は回すとゆるみます。

■比較測定時、使用前と使用後で原点の位置がズレる

・測定室の温度変化や、ダイヤルインジケータの繰返し誤差が原因として考えられます。下記の解決策をお試しください。
●測定室の温度が一定となるよう、環境を整える●測定時に任意の一定周期毎にマスタゲージを使用して、温度変化や繰返し誤差から来る原点ズレを調整する・ステムの締め付けを確認してください。ゆるい場合は確実に締めてください。

■測定値が安定しない、測定精度外の数値を示す・測定子の磨耗が考えられます。測定子が磨耗すると測定値に影響が出ますので、定期的に磨耗確認を行い、測定値に影響が出る場合は交換を行ってください。

本体仕様

品

番

WTI-2030R

WTI-2030W
測定方向⑤測定子がワークの被測定面と平行に、測定方向と直角に接するよう設置する

測定方向

90°

正しく接しないと、測定値や動作に不具合が出ます。

ワーク

90°

WTI-2038R

WTI-2038W
WTI-8030S

WTI-8030R

ワーク

WTI-8030W
WTI-8038S
測定方向

測定方向

WTI-8038R

WTI-8038W

ワーク測定範囲
目量（mm） （mm）

目盛仕様

全測定範囲行き 戻り誤差
指示誤差（μm） （μm）

測定力（N）

繰返し精密度測定子（μm）

0.002

0.2

0-100-0620.5 以下

10.0020.20-100-0620.5 以下

10.01

0.8

0-40-01330.5 以下

30.01

0.8

0-40-01330.5 以下

3

質量

ルビー

70g

超硬球
ルビー80g

超硬球
鋼球

ルビー

70g

超硬球
鋼球

80g

ルビー
超硬球

●交換用測定子品番：WTI-2030R･WTI-2038R…WTI-20R、WTI-2030W･WTI-2038W…WTI-20W
WTI-8030S･WTI-8038S…WTI-80S、WTI-8030R･WTI-8038R…WTI-80R、WTI-8030W･WTI-8038W…WTI-80W

ワーク

使用方法（比較測定）

使用後のお手入れ・保管方法

誤差の影響を少なくするために、測定物と同じ高さのマスターを使い、比較測定をします。

〔例〕マスター（ブロックゲージ）：30mm
ワークの高さ：30.08mm①マスターを設置する〈マスターによる原点の設定〉

測定子をマスターの被測定面と平行に、測定方向と直角に接するように当てます。測定子の側面から衝突させないように注意してください。

① 指針〈ワークの測定〉

使用中に付着した埃、汚れを清掃する。※注油は行わないこと。
・測定子摺動面の汚れは、乾いた布かアルコールを少量含ませた布で拭き取ってください。
・その他外観の汚れは、柔らかく乾いた布か、中性洗剤を少量含ませた布で拭き取ってください。

測定子の磨耗を確認する。

測定子の交換方法
付属のスパナを使用して測定子を交換してください。
スパナ

スパナ差込ミゾ・測定子が磨耗すると測定値に影響が出ます。定期的に磨耗確認を行い、測定値に影響が出る場合は交換を行ってください。

②原点を設定する

付属のケースに収納し、乾燥した冷暗所に保管する。

保持位置を上下させるか、外枠を回して目盛板を回転させ、指針と「０」を合わせます。

③マスターを外して、測定する

測定子に衝撃を与えないようマスターを静かに外し、ワークを当てて目盛を読み取ってください。

② 目盛板

④ 外枠

使用方法（平行度・平面度・偏心測定等）①測定子をワークの測定箇所に当てる・保管時、測定子に負荷をかけないよう、力（押し込み力・曲げ荷重など）をかけないでください。
・直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

必ず専用の測定子を使用してください。他の測定子を使用すると、精度不具合の原因となります。
・測定子交換後は必ず校正を行ってください。

保持位置を上下させるか、外枠を回して目盛板を回転させ、指針を「０」に合わせます。

当社にて校正を受け付けております。
全国にある当社代理店、取扱店を経由してお申し込みください。

性能の劣化は、使用の頻度・環境によって大きく異なります。
お客様の社内規格に準じた周期で、定期点検を行うことをおすすめ致します。

ワークをゆっくりと動かし、針の動きを読み取ってください。

- 3-

測定子交換後の長さ調整方法※WTI-2030R・WTI-2038R・
WTI-2030W・WTI-2038Wのみ

測定子を動かすと下図のようにねじが現れます。お手持ちのドライバー等で、ねじを回して、測定子の長さを調節してください。

校正について測定精度を保つため、定期校正を実施することをおすすめします③測定対象を動かし、目盛を読み取る

ゆるめる

ねじ

測定子をワークの被測定面と平行に、測定方向と直角に接するように当てます。測定子の側面から衝突させないように注意してください。

②原点を設定する

締める

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

E345-T2- 4-2109500