Please keep manual where it is accessible to user for future reference.

● Keep this manual with the instrument if transferred or leased to a third party.
● For inquiries about this product, please contact dealer or Niigata Seiki at the address listed on the following page.

■PARTIDENTIFICATION

Meas. Surfaces（Carbide Tips）

SpindleAnvil（Fixed Side）

50253035
400-Point
Adj. Hole

ON/OFFFrame【REAR】

0SET

2530

345

Ratchet StopABS

EOM133-125
EOM133-150【ACCS.】

Calibration Std.

Meas. Range（mm）

Accuracy※（ µm ）

Weight（g）

0− 25EOM133-50

25− 50

EOM133-7550− 75460

EOM133-10075−100600

EOM133-125

100−125

EOM133-150

125−15046265
380515560● Resolution： 0.001mm● Power： SR44 (Silver Oxide Battery)

TROUBLESHOOTINGCORRECTIVE ACTION● Clean measuring surfaces
with cloth.
● Confirm 0-Point setting andre-zero with long press of[SET] Button (> 2 sec.)● LCD does not display. ● Cycle power by removing※If problem persists, or if you have any questions, please contact
distributor or place of purchase.
※Please note, manufacturer is unable to respond to inquires or provide
service directly. Please contact distributor or place of purchase.

※For TestFEATURES● Carbide measuring surfaces.

● EOM133-25･････ 0.000● Ratchet stop.

● EOM133-75････ 50.000● EOM133-125･･ 100.000● Spherical Anvil Attachment (SR2.5) included.
● Auto-display-off function (approx. 6 minutes.)● Insulating cover・Thermal shield

SAFETY

PLEASE OBSERVE

Always follow the procedures specified below in order to prevent harm
to yourself or others, and to prevent damage to property.
Indicates risk of personal injury or propertyCAUTION damage if not followed.

● Always clean measuring surfaces with soft cloth before use.
● Do not turn spindle more than 3mm above uppermeasurement range.
● This is a precision instrument, handle with care. Do notdrop or subject to excessive forces.
● Keep away form high voltage and strong magnetic fields.
Instrument damage may result.
● Keep instrument away from direct sunlight or hightemperatures such as in a car, or near a stove or heat source.
● Do not disassemble or modify.
● Please do not use organic solvents to clean the body ofthe instrument.
● When not being used for long period of time, please
remove battery.
● Only use as directed. Improper use may cause accident or injury.

Make sure the Battery Cover is closed
securely. Please make sure the gasket isin the proper place on the Battery Cover.

"Click Click"

50ON/OFF

45

SET

40

Ratchet
Stop

Fig.1

Match origin lines
50ON/OFF

45

SET

40

Thimble

ABS

Fig.2

Rotate until origins match

Spanner

Fig.3

Press for > 2 sec.

● EOM133-50････ 25.000● EOM133-100･･･ 75.000● EOM133-150･･ 125.000① Remove any grease or dust from the measuring surfaces
using a clean cloth, such as lens cleaning cloth. Make sureno foreign objects are adhered to the measuring surfaces.
② Place object to be measured between the Anvil and Spindle.
③ With the Thimble Clamp loose, turn the Thimble to capture
the measured object between the measuring surfaces.
Stop after 1~3 “Clicks” from the Ratchet Stop. (Fig. 5)④ The measured value is displayed on the LCD display.
※ Measurement can also be read off Thimble and Sleeve aswith a standard micrometer.

Clamp

Meas. surfaces touch

50ON/OFF45

SET40

ABS

The gauge is now ready to take measurements.

● SR44 (silver oxide battery) included for testing.

[SET] Button
Workpiece

Fig. 4"Click Click"
5ON/OFFSET

Fig.5

00５

１4540

RatchetStop

Spherical Attachment
5CAREand replacing battery.
● Replace battery with newunstable.
one.
● Display is not clear.
● Displayed value isSR44（Silver Oxide Cell）

BALL ANVIL

ERROR● Measured value iswrong.

ＳＲ４４OPERATION

EOM133-25

Setting the origin used for measurements.
Once set, even when power is off the origin is saved anddoes not have to be reset with each use.
When battery is replaced origin will go to default setting,please reset.
① Remove any grease or dust from the measuring surfaces
using a clean cloth, such as a lens cleaning cloth. Make sureno foreign matter is adhered to the measuring surfaces.
② With Spindle Clamp loose, turn until measuring surfaces
touch. Turn until Ratchet Stop clicks 1~3 times. (Fig. 1)※ For models other than EOM133-25, please insert thesupplied Calibration Standard between measuring surfaces.
③ To align the Thimble Scale 0-Point reference lines. (Fig. 2)
1. Tighten Spindle Clamp to lock Spindle.
2. Insert the spanner into 0-point Adjustment Hole on rearof Thimble.
3. Use spanner to rotate Sleeve to align the 0-Point lines. (Fig. 3)④ Press the [SET] Button for more than two seconds. (Fig. 4)
LCD will display “SET” for 0.5 seconds, and then thedisplay will read as shown below:

Spherical (Ball)
Attachment（Not w/ EOM133-25）

Spanner※Not incl. quantization error (±1 count)Lock

Battery
Cover

When [ON/OFF] Button is pressed, power is turned ON, andmeasurement is displayed on the LCD.
Press [ON/OFF] Button a second time to turn power OFF.

Loosen40※ No Insulating CoverModel No.

40

Clamp

Insulating
Cover

50
ON/OFF

SPECIFICATIONS

45

SET

【FRONT】[ON/OFF] Button,[SET] Button

Battery
Cover

ThermalShieldSleeveThimbleLCD Display

Open

SETTING THE ZERO-POINT● For safe and proper use of this product, please read this instruction manual before use and follow the procedures described.

① Rotate the Battery Cover on rearusing coin or similar tool.
② Insert battery with “+” side up.
③ Replace Battery Cover.

POWER

Thank you for purchasing the Niigata Seiki DIGITAL OUTSIDE MICROMETER.
This precision gauge measures outside dimension of object placed between the Anvil and Spindle.

35

DIGITAL OUTSIDE MICROMETERbattery.

30

PRECISION OUTSIDE MEASURING INSTRUMENT

CAUTION Only use SR44 type40

EOM133-***BATTERY

ModelNo.

INSTRUCTION MANUALThe Spherical (Ball) Anvil Attachment can be attachedas required.
After mounting, please set 0-Point following the procedure“Setting the Zero-Point” above.
※ When the Spherical Attachment is attached, the measurement
indicated on the Sleeve scale will read 5mm larger thanactual size. (Fig. 6)● If any cutting fluid or moisture gets on the instrument body, wipe

using a soft dry cloth to prevent corrosion.
● After use, remove any cutting chips or contamination using a softcloth and apply rust preventative oil. Keep out of reach of children.

0ON/OFF0SETFig.6

4540

Sleeve reading = Meas. + 5mm

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

スピンドル

測定面

スリーブ

LCD表示部

シンブル

アンビル（超硬チップ）

（固定側）

2530

フレーム

3540

【裏面】

０点調整穴

50
ON/OFF

045

SET

【表面】

40

ラチェットストップ
ON/OFFボタン、SETボタン

電池蓋
防熱カバー

クランプ

締める

5

ゆるめる

0ON/OFF
SET

2530

34540

ABS

防熱板

EOM133-125
EOM133-150※防熱カバーなし

基準棒

【付属品】

キースパナ

製品仕様品番※量子化誤差（±1カウント）含まず

測定範囲（mm）

最大許容誤差※
（µm）

質量（g）

0− 25

EOM133-50

25− 50

EOM133-75

50− 75460

EOM133-100

75−100600

EOM133-125

100−125

EOM133-150

125−150●最小読取値：0.001mm●電源：SR44（酸化銀電池）

46265
380515560

トラブルと対応※症状が改善されない場合や不明な点がございましたらお買い上げの販売店、または発売元までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

特長●測定面は超硬チップ付●防熱カバー・防熱板付●球面アタッチメント付（SR2.5）●定圧装置付●オートディスプレイオフ機能付（約6分）●SR44（酸化銀電池）テスト用１個付属

安全上のご注意

必ずお守りください

お使いになる人や他の人への危害、財産への損害を未然に防止するた
め、必ずお守りいただくことを下記のように説明しています。

注 意 するおそれがある内容』です。

お守りいただかないと『傷害を負う、または物的損害が発生●ご使用の前に、必ずグラスペーパなどで測定面の汚れを除去してください。
●測定範囲の上限を3mm以上上回って、スピンドルを回さないでください。
●精密機器ですので、落としたりぶつけたり、強い力を加えたりしないでください。
●強い磁気や高電圧に近付けないでください。故障の恐れがあります。
●直射日光の当たる場所や炎天下の車の中、火のそばなどでの使用、保管はしないでください。
●分解・改造をしないでください。
●本体の手入れに有機溶剤を使用しないでください。
●長期間使用しない場合は、本体から電池を外して保管してください。
●用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。

電池蓋

クランプ

測定面を合わせる

カチカチ

5①アンビル・スピンドルの測定面を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。
②アンビルとスピンドルの間に被測定物を入れます。
③クランプを回してゆるめ、ラチェットストップを持ってシンブルを回し、測定面で被測定物を挟みます。ラチェットが 『カチカチ』と鳴ったところから1〜3回転させたところで停止してください。 図 5④LCD表示部に表示された数値が､測定値となります。
※通常のマイクロメータ同様にスリーブとシンブルの目盛を読み取ることもできます。
必要に応じてアンビルまたはスピンドルの測定面に、球面アタッチメントを取り付けてください。
球面アタッチメント取付後、上記『原点の設定』を参照して0点を合わせてください。
※球面アタッチメント取付時にスリーブとシンブルの目盛を読む場合は、測定値はスリーブの表示値より5mmマイナスとしてください。 図 60ON/OFF45

SET40

ラチェット
ストップ
図1

通常測定で使用する原点を設定します。
この原点は、電源をOFFにしても記憶されますので、毎回設定し直す必要はありません。
電池交換後はリセットされますので、再度設定し直してください。
①アンビル・スピンドルの測定面を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。
②クランプを回してゆるめ、ラチェットストップを持って回し、測定面を合わせます。
測定面が合ってから、ラチェットが「カチカチ」と1〜3回鳴った所で停止してください。 図 1※EOM133-25以外は、測定面の間に付属の基準棒をはさんで合わせてください。
③基準線目盛の『0』にシンブルの『0』目盛が一致していれば、スリーブ・シンブルの0点が合っています。 図 2
一致していない場合は、下記の手順で0点を合わせてください。
1. クランプを回してシンブルを固定する。
2. キースパナをスリーブ裏面の0点調整穴に入れる。
3. 0点が一致するまで、キースパナでスリーブを回す。 図 3④SETボタンを2秒以上長押ししてください。
LCD表示部に『Set』が0.5秒ほど点灯後、数字の表示が下記のように表示されます。 図 4●EOM133-25･････ 0.000 ●EOM133-50････ 25.000●EOM133-75････50.000 ●EOM133-100･･･75.000●EOM133-125･･100.000 ●EOM133-150･･125.000
これで測定の準備ができました。

使用後のお手入れ●LCD表示部に何も表 ●電池を入れ直してください。
示されない●新しい電池と交換してください。
●表示が安定しない●表示が不明確である

SR44（酸化銀電池）※テスト用

球面アタッチメントトラブルの内容
対応策●測定値が間違ってい ●測定面の汚れをグラスペーる
パなどで拭き取ってください。
●0点が合っているか確認し、SETボタンを２秒以上長押してください。

ＳＲ４４

使用方法

EOM133-25

球面アタッチメント（EOM133-25を除く）

35■各部の名称①本体裏面の電池蓋を、矢印の向きにコインなどで回して開けます。
②電池を必ず（＋）側を表にしてセットしてください。
③電池蓋を元通りに装着してください。

電池蓋は確実に閉めてください。
パッキンが電池蓋から外れていないことを確認してください。

ON/OFFボタンを押すと電源がONになり、LCD表示部に数値が表示されます。もう一度押すと電源がOFFになります。

原点の設定●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

開ける

電源

この度は「デジタル外側マイクロメータ」をお買上げいただきありがとうございます。
この商品はアンビルとスピンドルで測定対象物を挟み、外側長さを測定する精密測定器です。

電池は必ずSR44をお使いください。
30

デジタル外側マイクロメータ

ご注意

40

精密外側測定器

品番 EOM133-***

電池の入れ方

取扱説明書

原点が合っていることを確認
50ON/OFF

45

SET

40

シンブル

ABS

図2

原点が合うまで回す

キースパナ

図3

2秒以上長押し50ON/OFF

45

SET

40

図4

ABS

SETボタン
被測定物を挟む

カチカチ
5ON/OFF
SET

図500５１

4540

ラチェット
ストップ

球面アタッチメント
50ON/OFF0SET

図6●切削液等が本体にかかった場合は、柔らかい乾いた布で水分を拭き取ってください。そのまま放置すると錆の原因となります。
●使用後は、柔らかい布などでゴミや切粉などを取り除き、防錆油を塗布してお子様の手の届かない場所へ保管してください。

45
40

スリーブの表示値より５マイナス
発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256) 33-5502(代) FAX (0256) 33-5528

URL http://www.niigataseiki.co.jp

E1141-Q

20062000