Digital Torque Wrench for Tightening

CPT-G (Global model)
PAT. PEND

The tightening process is simplified with visualconfirmation of the applied torque

Torque Tightening with Digital Precision
PRO TORK

Highly responsive torque sensor for precise application oftightening torque10 different target torque values can be programmed and easily
recalled by alphabetic registration ID
LED indicator allows the user to follow the changing torque values
Multiple units of measure (SI, Metric, American) setup capabilities
50 data memory capacity with RS232C output
Accuracy ±3% of indicated value, CW & CCW
measurement, Power source: 2 x AA batteries

Features and Functionsz Several different tightening modes available to cater to a variety ofapplications. Quick and accurate tightening while preventing errors.
Modes include:
Preset Tightening Mode, Judgment Tightening Mode, Peak/Run Modes* Retightening/loosening torque is performed in the Peak Mode.

Display example 1: Preset Tightening Mode
(Red LED shows the level of the applied torque)

Preset Tightening Mode: Allows user to set the target torque with specific % oftorque allowable beyond target, then the red LED moves towards the right toindicate the level of the applied torque. When it reaches the target torque range,
the blue LED blinks and the buzzer signals tightening completion.

Display example 2: Judgment Tightening Mode
(As torque is being applied prior to completion)

Judgment Tightening Mode: Allows user to set judgment ranges for lower/upper
limit in the tightening operation. Upon tightening completion a judgment is made as
torque value is stored in the memory.

z Equipped with bright LED lamp
Bright LED lamp indicates the current torque level against the target torque,which enables a precise application in the tightening operation.

Display example 3: Judgment Tightening Mode
(The case of exceeding target torque range)
Direction

Common Specificationsz Multiple unit of measure selection
5 changeable units of measure through keypad setup, including
N·m, kgf·cm, kgf·m, lbf·in, and lbf·ft.

z Data Memory, Torque Set Registration and Output functions
Up to 50 torque measurements can be saved as well as up to 10 different
target torque or judgment range registrations. By selecting the registeredtorque setting (indicated alphabetically as A-J), you have the flexibility towork on several different tightening applications with ease. The stored
data can be transferred to PC with RS232C cable (optional).

z TQH Ratcheting Square Drive Head ideal for tight spaces
72 teeth ratchet with 5 degrees of operation,
Push release socket lock function* For CPT20x10D-G-SET to CPT200x19D-GSET (4 Models).
Note:
TQH22D for CPT280x22D-G-SET
has 24 teeth without push releasesocket lock function.
TQH heads are contained only when 72 teeth ratchet headpurchasing the “Set” model version. (TQH15D)

Accuracy±3% of indicated valueTightening Direction
Clockwise/Counter clockwise
Display/Character Height 14 segment LCD 6 digits/7mm
7 segment LCD 4 digits/3mm
Battery Life Indicator
4 stepsNumber of Data Memory 50
Torque Setting Memory Preset Tightening mode: 10 torque values to register
Judgment Tightening mode: Up to 10 values of each
Upper/Lower/Tightening direction
Basic Function
Auto power off ( 3 minutes)
Auto memory/Reset
Auto zeroOver torque alarm
PowerAA battery x 2pcs
Continuous Use
Approx. 40 hours
Temperature in Use
0-40 Celsius below 85%RH (no condensation)“Torque Wrench Only” ModelsAccuracy ±3%

Torque Range
OverallWeight[N·m][kgf·cm][kgf·m][lbf·in][bf·ft]LengthModel
Min.-Max. Grad. Min.-Max. Grad. Min.-Max. Grad. Min.-Max. Grad. Min.-Max. Grad. [mm][kg]
CPT20x10D-G
4-20
0.02 40-200 0.2 0.4-2 0.002 36-180 0.2 3-14.5 0.02 280.5 0.63
CPT50x12D-G
10-50 0.05 100-500 0.5
1-50.005 100-440 0.5 7.5-36 0.05 282.5 0.65
CPT100x15D-G 20-100 0.1 200-1000 12-100.01 200-880 1
15-73
0.1 384.5 0.85
CPT200x19D-G 40-200400-20004-20
360-1700
30-150475.5 1.37
0.2
20.02
20.2
CPT280x22D-G 56-280
560-2800
5.6-28500-2400
42-200591.5 1.76

Note 1. "Torque Wrench Only" version is provided in basic carton product box and does notinclude TQH Head, Batteries, Storage Case, or Product Box.
2. "Overall Length" does not include the length of interchangeable head TQH.
3. "Weight" does not include the weight of interchangeable head TQH and batteries.
4. Approx. 230 different types of interchangeable heads are available and sold separately.
Refer to Tohnichi's Torque Handbook or Product Catalog for additional information.

“Set” Models (including Accessories)
Accuracy ±3%
Standard Accessories
Model
CPT20x10D-G-SET
CPT50x12D-G-SET
CPT100x15D-G-SET
CPT200x19D-G-SET
CPT280x22D-G-SET

Ratchet Head
Sq. Drive
Model
[mm]
TQH10D
9.5
TQH12D
TQH15D
12.7
TQH19D
19.0
TQH22D

Battery

AA Alkaline
Battery (2pcs)

Storage
Case

ProductBox

Small

Small

Large

Large

Storage Case
(For “Set” model only)

CPT-G Optional Accessories
PC Connecting Cable
Part #585

Applicable Model
CPT-G → PC (D-Sub 9 Pin Female)Note. Recommendation: Use 2xAA Ni-MH batteries for longer continuous use.

How to Order[Ex. 1] CPT100x15D-G-SET* “Set” model version (with standard accessories)[EX. 2] CPT200x19D-G* “Torque Wrench Only” version (without standard accessories)

Data Processing Software
ModelEXCEL RECEIVERMediaCD-ROM

10.07D