Circumference / Diameter
Precision Meas. Instrument

ModelNo.

INSTRUCTION MANUALCJU-****BKD

CIRCUMFERENCE MEASURING TAPE
(CIRCO METER) KAIDANREADING THE SCALE

HOW TO USE① Place the Tape around the surface to be measured.
② Pass the end of the Tape through the Reader as shown below.
Workpiece

Thank you for purchasing CIRCUMFERENCE MEASURING TAPE (CIRCO METER) KAIDAN. This product is designed formeasuring the circumference and diameter of large cylindrical parts.
● For safe and proper use, please read this instruction manual before use and follow the procedures described.

Measurement = Main Scale reading + Vernier Scale reading

2hThroug

1 In s ertTape● Please keep manual where it is accessible to user for future reference.

The measurement is determined by reading the Main Scale,and then adding the Vernier Scale reading determined by theposition where the Vernier and Main Scale graduations align.

●Diameter Measurement Example① Diam. Main Scale ＝271.5mm② Diam. Vernier Scale aligns ＝0.35mm

Tape

Keep this manual with the instrument if transferred or leased to a third party.

③ Measurement ＝271.5＋0.35＝ 271.85mm②Reader● For inquiries about this product, please contact the dealer where purchased, or the address listed on the back page.

■ PARTS ID & FUNCTION ※ Model CJU-950BKD shown.

Vernier Scale（0.05mm）<Reader Scales>③ Diameter Main

④ Diameter Vernier

0270
840 850① Tape② Reader

0.5

860
00.5

870

890

900

Pull

⑥ Circum. Vernier

ReaderVernier Scale 840

Keep Tape straight

Denotes a requirement –You MUST do.

WARNING: Indicates risk of personal injury or property damage if not followed.
Read the manual and follow the directions.
・Use of product other than as described in thismanual may cause accident.

Only use within specified measurement range.
・Use outside of specified range may cause deformation and damage resulting in poor accuracy.

Use only for measuring.
・Use for any purpose other than measuring may
damage or wear the instrument. Improper use may
also cause accident.

Handle with Care.
・Do not subject gauge to excessive shock. Do notplace under heavy objects. Improper handling may
cause damage or poor accuracy.
・Do not scratch instrument surfaces.

Make sure power is turned off to equipment before using.
・There is risk of electrical shock.
Wear gloves to protect hands during use.
・Tape edges may cause injury to unprotected hands.
Protect eyes from tape tip during use.
・Tape is spring steel and may whip off surface.
Use care to prevent injury.

850（1mm）

Tape

Main Scale（0.1mm）④ Read the diameter or circumference measurement.

0Use and store in location meeting the following conditions:●Dry and protected from rain, water, and oils.
●Protected from direct sunlight.
●Protected from children and unauthorized users.
・Use in places contrary to the above may cause
damage to the product or result in accident or injury.
Do not disassemble or modify.
・ Disassembly may damage product or cause accident.

860

870

0.5

880

890

1.0②

Reading The Scale

NOTE: When using model CJU-950BKD

Please Note
● When taking a reading, make sure the Reader is tight against

When measuring workpieces with a diameter of φ180mm or less,there will be gap between the Reader and the Tape, causing thereading to be slightly larger than the actual measurement.

the Tape, and read from directly in front of the Scale. Readingat an angle may cause measurement error.

● When reading diameter and circumference, for each reading

hold the Tape in place and take a reading. If attempting to read
both simultaneously, the measurement may not match acalculated value due to how the Tape is held.

Workpiece diameter≦ φ180mm
Gap

Denotes a prohibition –You MUST NOT do.

280

①①1.0

⑤ Circum. Main

1.0① Circum. Main Scale＝853mm② Circum. Vernier Scale aligns＝0.6mm③ Measurement = 853＋0.6＝ 853.6mm

Please Observe

To prevent harm to yourself or others, and to prevent damage to property,
always follow the procedures marked by the following symbols.

270

0.5

●Circumference Measurement Example

Workpiece290910① Tape …………………………… Goes around the object to be measured. Contains diameter and circumference MainScales.
② Reader ………………………… Contains diameter and circumference Vernier Scales. For reading instructions,please refer to section below.
③ Diameter Main Scale ………… 0.5mm Diameter Scale.
④ Diameter Vernier Scale ……… 0.05mm Diameter Scale.
⑤ Circumference Main Scale …… 1mm Circumference Scale.
⑥ Circumference Vernier Scale … 0.1mm Circumference Scale.

SAFETY NOTICESMain Scale（0.5mm）③ Hold the Reader secure and pull the Tape tight. Makesure the Tape is not at an angle, and is in full contact
with the workpiece.

1.02808800

AFTER USE CARE ・STORAGE● Using a dry cloth, wipe off any dirt or oils.
● Store in a safe place, out of harm's way.
If stored under heavy object, or subject to excessive
forces, the tape may be bent and become inaccurate.
● Store in a cool, dry, and dark location.
Avoid direct sunlight or high humidity, and access fromunauthorized users.

TapeReader【Diameter Measurements】●φ20〜60mm ：~ 0.5mm●φ60〜80mm ：~ 0.2mm●φ80〜180mm：~ 0.15mm
Reading oversize amount【Circumference Measurements】● 60〜188mm ：~ 1.6mm● 188〜251mm：~ 0.6mm● 251〜565mm：~ 0.5mm
Reading oversize amount

For small diameter workpieces, account for the error
when taking a measurement.

SPECIFICATIONS
Model No.
D ia m- Meas. Range（mm）
m e te r Resolution（mm）
Circum- Meas. Range（mm）
ference Resolution（mm）
Accuracy of Length（mm）
Mass
Material（g）

CJU-950BKD

CJU-2200BKD

60 〜 950

940 〜 2200

φ20 〜 300

± 0.2050

φ300 〜 700

82

CJU-3460BKD

φ700 〜 1100

φ1100 〜 1500

2190 〜 3460

3450 〜 4720

0.05
0.1

CJU-4720BKD± 0.25114

Stainless Steel± 0.30150

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書
円周・直径測定用
精密測定器

使用方法

CJU-****BKD

品番サークルメジャー 快段目盛

目盛の読み方

①対象ワーク（被測定物）にテープを巻き付けます。
②読取部へテープを図のように通してください。

この度は「サークルメジャー」をお買上げいただきありがとうございます。この商品は、大径部品の円周、直径測定が可能な精密測定工具です。

ワーク●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社（裏面記載）にご連絡ください。

1

本尺目盛に、本尺目盛と副尺（バーニヤ）目盛が合致したところの副尺目盛をプラスして、測定値を読みます。

測定値＝本尺目盛＋副尺（バーニヤ）目盛●直径測定例

2通す

通す

テープ

①直径本尺目盛＝271.5mm②直径副尺目盛＝0.35mmが合致③271.5＋0.35＝ 271.85mm が測定値となります。

テープ読取部②■各部の名称と機能 ※イラストはCJU-950BKDです。

直径副尺目盛（0.05mm）<読取部詳細>③直径本尺目盛

830

①テープ
②読取部

840

270
850

0.58600③読取部を固定し、テープを引きます。
テープが斜めに傾かないよう注意しながら、ワークへと密着させてください。

④直径副尺目盛

0⑤円周本尺目盛

280880890

900290910920③直径本尺目盛 …… 直径の０.５ｍｍ単位を測定する目盛です。

④直径副尺目盛 …… 直径の０.０５ｍｍ単位を測定する目盛です。
⑤円周本尺目盛 …… 円周の１ｍｍ単位を測定する目盛です。

⑥円周副尺目盛 …… 円周の０.１ｍｍ単位を測定する目盛です。
必ずお守りください注意

してはいけない内容『禁止事項』です。

必ず実行していただく内容『強制事項』です。

お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

取扱説明書をよく読み、指示に従う。
・取扱説明書に記載された内容以外での使用は、事故の原因となります。

指定された測定範囲内で使用する。
・測定範囲外の対象物への使用は、商品の変形や破損、精度不良の原因となります。

測定作業にのみ使用する。
・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。

大切に取り扱う。
・ぶつける、落とすなどの衝撃を与えたり、重量物を上に乗せたりしないでください。精度不良や破損の原因となります。
・商品に傷をつけないでください。

通電している箇所へ使用しない。
・感電の恐れがあり大変危険です。
作業用手袋を着用して使用する。
・テープで手を切る可能性があります。作業用手袋などで手を保護し、使用してください。
テープの跳ね返りに注意する。
・金属製テープのため弾力性が高く、使用中に跳ね返る恐れがあります。先端が目などに当たらないよう注意してください。

下記の条件を満たす環境で使用・保管する。
●雨や水、油などがかからない、乾燥した場所●高温にならない場所●子どもや、使用者以外が近づかない場所
・上記に反する場所での使用は製品の破損、事故やけがの原因となります。
分解・改造を行わない。
・製品の破損や事故の原因となります。

01.08700.5280880890

1.0①円周本尺目盛＝853mm
00.5
1.0②円周副尺目盛＝0.6mmが合致③853＋0.6＝ 853.6mm が測定値となります。

引く

読取部

0.5

270
①840 850 860●円周測定例

ワーク

300
940

⑥円周副尺目盛②読取部 …………… 直径・円周それぞれの副尺目盛が表示されています。目盛の読み方は裏面を参照してください。

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを右記のように説明しています。

930

1.0

①テープ …………… 直径・円周それぞれの本尺目盛が表示されています。対象ワーク（被測定物）に巻きつけて測定を行います。

安全上のご注意

830

1.08700.5

直径本尺目盛（0.5mm）

0270
①830
840
850860
円周本尺目盛テープを傾けないこと

テープ

870（1mm）

円周副尺目盛（0.1mm）

00.5

280880

890

1.0

②

④直径または円周、いずれかの目盛を読み取ってください。
目盛の読み方CJU-950BKD使用時のご注意

ご注意●測定時は必ず目盛をしっかりと合わせ、使用する目盛の正面から読み取ってください。
斜めから読み取ると、測定誤差の原因となります。
●直径、円周の同時測定は固定方法により、計算値と異なる値が出る場合があります。
直径、円周の測定はそれぞれ目盛を固定して、読み取りを行ってください。

φ180mm以下の小径ワークを測定する場合、読取部とテープに隙間が生じるため、誤差が発生いたします。
φ180mm以下の小径ワーク
隙間ができる

使用後のお手入れ・保管方法

テープ●汚れや油などを乾いたウエスで拭き取る●無理な力がかからないよう、大切に保管する
重量物を上に乗せたり、無理な力がかかった状態で保管すると本体が曲がり、精度不良の原因となります。
●乾燥した冷暗所に保管する
直射日光や湿気は避け、管理者以外が触れない状態で保管してください。

読取部【直径測定時】●φ20〜60mm ：約 0.5mm●φ60〜80mm ：約 0.2mm●φ80〜180mm：約 0.15mm
大きく表示されます。
【円周測定時】● 60〜188mm ：約 1.6mm● 188〜251mm：約 0.6mm● 251〜565mm：約 0.5mm
大きく表示されます。

小径ワーク測定の際は、誤差を考慮して測定を行ってください。

製品仕様
品

直径
円周番

測 定 範 囲（mm）

最小読取値（mm）

測 定 範 囲（mm）
最小読取値（mm）

長さの許容量（mm）
質

材量 （g）質

CJU-950BKD
φ20 〜 30060 〜 950± 0.2050

CJU-2200BKD
φ300 〜 700
940 〜 2200

82

CJU-3460BKD

φ1100 〜 1500

2190 〜 3460

3450 〜 4720

0.05
0.1

CJU-4720BKD

φ700 〜 1100

± 0.25ステンレス114± 0.30
150

発売元
〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I391-K22110