INSTRUCTION MANUALID Measuring
Instrument

ModelNo.

SCDI-***SS-LINE BORE GAUGE

Thank you for purchasing the S-LINE BORE GAUGE.
This product is a two-point mechanical instrument for measuring the ID of bores and holes relative to a reference
standard. Instrument can be used with a Dial Gauge, Digital Indicator Gauge, Electronic Sensor, etc.
● For safe and proper use of this product, please read this instruction manual before use and follow the procedures described.
Please keep manual where it is accessible to user for future reference.
● Keep this manual with the instrument if transferred or leased to a third party.
● For inquiries about this product, please contact dealer or Niigata Seiki at the address listed on the following page.

■ PART IDGauge

1. Gauge Mount
2. Shaft
3. Measurement Head10(soldseparately)

0SAFETY

1020

30

40

40

50

50

CAUTION:

Indicates risk of personal injury orproperty damage if not followed.

●Use in accordance to the measuring range specified
in the manual for the measuring instrument model.

4. Centering Plate(Guide)●This is a precision instrument, handle with care.

5. Washers
6. Extension Rod
7. Extension RodClamp Nut●Do not disassemble or modify.

●Instrument is not waterproof or corrosion resistant;
Keep away from water and oils.

2●Use only with supplied Extension Rods and Washers.

●Keep instrument away from direct sunlight and hightemperatures such as in a car, or near a stove or heatsource.

34

Please Observe

Always follow the procedures specified below in order to prevent
harm to yourself or others, and to prevent damage to property.

156●After use, remove dust and cutting chips, and apply
rust preventative oil to Extension Rods and Washers.
When not in use store in supplied storage case.
●This product is for measuring inside diameter; use
only as directed. Improper use may cause accident orinjury.

7SPECIFICATIONS● Accuracy： 2 µm

● Retrace Error： 2.5µm● Repeatability： 0.5µm※ Accuracy specifications are for Instrument Body and do not include the Gauge. Total accuracy specification with Gauge
attached will be the sum of Instrument Body accuracy plus the Gauge accuracy.

● Gauge Mount Specifications： Clamp Diameter …φ8mm h6Gauge Insertion Length (distance from probe tip to top of clamp surface … ≥ 40mm
Meas. Range (mm)SCDI−8S
6-8

SCDI−12S
8-12

SCDI−20S
12-20

SCDI−35S
18-35

SCDI−60S
35-60

Stroke Length

0.5

0.7

0.9

1.3

1.3

7pcs.

9pcs.

9pcs.

Model No.

(mm)

9pcs.
18・20・22
24・26・28
18・19・20 30・32・34
1pcs.
2pcs.

8・8.5・9
12・13・14
Ex. Rod Quant.／ 6・6.4・6.8
7・7.2・7.6 9.5・10・10.5 15・16・17
Sizes (mm)Washer Quant. ／Sizes (mm)Weight (g)Notes

811・11.5・12−−0.5656595
Probe replacement wrench included

6pcs.
35・39
44・4954・59
2pcs.

0.5・1.0

2.0・3.085

175

SCDI−100S SCDI−160S
50-100100-160
1.4① Select a combination of Extension Rod and Washers as close as possible to the ID of the part to be measured.
※ Must be withing measuring range of attached Gauge.
※ For SCDI-8S、12S、20S the Extension Rod and Washer attach directly to the Measuring Head.
For SCDI-35S、60S、100S、160S、attach the Extension Rod and Washer using the Clamp Nut.
② Attach a measuring instrument such as a dial gauge (sold separately) to the Instrument Body.

OPERATION

2030

PREPARATION

1.6

11pcs.
7pcs.
50・55・60・65 100・110・120
70・75・80・85 130・140・150
90・95・100
160
3pcs.
4pcs.
1.0・2.0
1.0・2.0
3.0
3.0・6.0
190530※SSCDI Series Bore Gauge is a comparative gauge and requires a Reference Master such as a ring gauge to setup for comparative measurements.
※Prepare a Reference Master appropriate for the dimensions of the part to be measured.
※Before use, make sure the components on the Measurement Head are all securely attached and not loose.

SETTING THE 0-POINT①Prepare a Ring Gauge to use as a reference standard for setting
the zero-point. Insert the Measurement Head into the ring gauge.

Fig.１
10②In the direction perpendicular to the Ring Gauge axis, position theBore Gauge to measure diameter ⓐ shown in the cross section(Fig. 1). This is the maximum distance, and the Gauge will show
the minimum reading. The Guide moves to automatically position
the Gauge Head at this diameter when inserted.
③In the cross section shown in Figure 2, parallel the Ring Gauge
axis though diameter ⓐ, position the Bore Gauge to minimize thedistance ⓑ. In this position the Gauge will read the maximumvalue.
This position is determined by manually moving the gauge andnoting the position where the Pointer shows the maximum value.
(Fig. 3)

2030
40
50

Fig.２

Fig.３

ⓑMEASUREMENT
Insert the Measurement Head of the Bore Gauge into theobject to be measured. As in step ③ of “Setting the 0Point” above, pivot the gauge while checking the pointer todetermine the diameter ⓑ where the gauge will show themaximum value to determine the measurement reading.
This measurement will be the difference in diameter between
the part and the reference Ring Gauge.

Set to Zero1020

● Clean outside surfaces regularly using a cloth to remove any
dirt or contamination.
● Use a cloth dampened with a cleaning solution if required forexcess dirt.
● After use, place in a rust preventive bag or use a corrosion
inhibiting oil for protection.
● Store in supplied case, protected from direct sunlight or
humid environments, and accessible only to authorized
personnel.

1050

�

④The reading obtained at step ③ is the zero-point. Set the Gaugeto zero and the zero point is now set. (Fig. 4)

AFTER USE CARE, STORAGE

020304030010102020303001020304040404050505050REPAIRS AND SERVICE●If Gauge does not operate properly, or if you have any questions,please contact distributor or place of purchase.
●Please note, manufacturer is unable to respond to inquires or provide
service directly. Please contact distributor or place of purchase.

Fig.４

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書

内径測定器

Sラインシリンダゲージ

品番SCDI-***S

この度は「Sラインシリンダゲージ」をお買上げいただきありがとうございます。
この商品は内径基準とその寸法差を測定するための機械式2点比較測定器です。ダイヤルゲージ、精密指示計、電子プローブなどを取り付けて使用できます。
●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称測定器（別売）

1. 測定器ホルダ
2. 外筒
3. 測定ヘッド4. センタリングプレート（案内板）
5. ワッシャ6. 換えロッド
7. 換えロッド取付ねじ100安全上のご注意

10202030404050501●Sラインシリンダゲージシリーズの測定器は、指定された測定範囲および取扱説明書に基づいてご使用ください。
●精密機器ですので、落としたりぶつけたりしないでください。
●分解・改造はしないでください。
●防水・防錆仕様ではありません。水や油のかかるところでは使用しないでください。
●換えロッド・ワッシャなどは付属のものを使用し、他との併用は避けてください。
●直射日光の当たる場所や炎天下の車の中、火のそばなどに放置しないでください。
●使用後は柔らかい布などでゴミや切粉などを取り除き、換えロッド・ワッシャなどは防錆処理をして、収納ケースの元の位置に保管してください。
●本製品は内径測定用機器です。
用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。

345

製品仕様

●行き精度：2µm

67●戻り誤差：2.5µm

必ずお守りください

注意 お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

2●繰返し精度：0.5µm※精度はすべて、本体のみの数値です。測定器を取り付けた時の総合精度は、上記の精度に測定器の精度が付加されます。

●取付可能な測定器の仕様：クランプ部径…φ8mm h6
差込部長さ（把持可能な部分から測定子先端までの距離）…40mm以上品

番測 定 範 囲（ m m ）

測定子の有効ストローク（mm）

SCDI−8S
6-8

SCDI−12S
8-12

SCDI−20S
12-20

SCDI−35S
18-35

SCDI−60S
35-60

0.5

0.7

0.9

1.3

1.3

7本

9本

9本

811・11.5・12−−9本
18・20・22
24・26・28
18・19・20 30・32・34
1枚
2枚

換えロッド数 ／ 6・6.4・6.8
8・8.5・912・13・14
付属サイズ（mm） 7・7.2・7.6 9.5・10・10.5 15・16・17
換えワッシャ数／
付属サイズ（mm）
質

備量（g）
考

6565

測定子交換用レンチ付属6本
35・3944・4954・59
2枚

0.50.5・1.0

2.0・3.09585

175

SCDI−100S SCDI−160S
50-100100-1601.4①測定対象物の内径にできるだけ近くなるよう、換えロッドとワッシャを組み合わせて選択し、測定ヘッドに取り付けてください。
※測定器の測定範囲でご使用ください。
※SCDI-8S、12S、20Sは、直接測定ヘッドにねじで換えロッドとワッシャを固定してください。
SCDI-35S、60S、100S、160Sは、換えロッド取付ねじで換えロッドとワッシャを固定してください。
②ダイヤルゲージなどの測定器（別売）を、測定器ホルダに取り付けてください。

測定方法

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。

30

ご使用前に

1.611本
7本
50・55・60・65 100・110・120
70・75・80・85 130・140・150
90・95・100
160
3枚4枚
1.0・2.0
1.0・2.0
3.0
3.0・6.0
190530※SCDIシリーズは、比較測定器であり、測定する寸法を決定するためにリングゲージ等の内径用マスタが必要です。
用意する基準マスタは、測定する内径の寸法公差に合わせてください。
※ご使用の前に、測定ヘッドと外筒の間がゆるんでいないか確認し、ゆるんでいたら締め付けてください。

０点の調整①最初に０点を調整します。
基準とするリングゲージを用意し、測定ヘッドを差し込んでください。

（図１）
10

④ ③で求められた点が、０点となります。
測定器を調整し、０に設定してください。
（図４）102030405050

�②円筒の軸に垂直な断面において、直径ⓐは最大値となり、測定器は最小値を指します。
（図１）
この直径ⓐは、可動する案内板によって自動的に求められます。
③直径ⓐを通る円筒の軸に平行な断面において、直径ⓑは最小値となり、測定器は最大値を指します。
（図２）
この直径ⓑは、手動で求めます。
シリンダゲージを（図３）のように振り、測定器が最大値を示す点を探してください。

0203040（図２）

測定

（図３）

ⓑ

測定物にシリンダゲージの測定ヘッドを差し込み、測定します。
上記「０点の調整」③と同様に、目盛を確認しながらシリンダゲージを振り、直径位置ⓑ（測定器が最大値を示す点）で測定を行ってください。
基準リングゲージからの差の値が表示されます。

使用後のお手入れ・保管方法●外部から触れることのできる研磨部品は、定期的に清潔な布で拭いて汚れを除去してください。
●汚れがひどい場合は測定器用の洗浄液等を使用して清潔な布でふき取ってください。
●防錆油を薄く塗るか、防錆袋に入れるなどの防錆処理を行ってください。
●付属のケースに収納し、直射日光や湿気を避け、管理者以外が触れない状態で保管してください。

修理の際は０に設定する
102030010102020303001020304040404050505050●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、発売元までご連絡ください。
●お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

（図４）

発売元
〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I407-K2003