INSTRUCTION MANUALDIGITAL S-LINE INDICATOR

Model No. DEI-WKS1/DEI-WKS2 <combined>

Thank you for purchasing the Digital S-Line Indicator.
For proper operation, please read this manual thoroughly before use.

PARTS IDENTIFICATIONFEATURES● Tolerance Setting capability

LCD Display

Display Rotates 270°
Easy to read from any position.

● Calculation capability● Display rotates 270°

● Programmable “Favorite” Button

SWISS MADE

(Can be set for one touch access to frequently used functions.)● MIN、MAX、MINMAX (DELTA, or TIR) display functions● Direction of measurement (＋ / −) switchable● Preset function● Sleep mode (after approx. 20 min.)[MODE] Button

MODE[SET] Button

SETBattery CoverCenter (“Favorite”)
ButtonAccessories

Stem (Mounting Shaft)

ＣＲ２０３２Spindle

CR2032 (Lithium Cell)※Included for test

Contact Point①Remove the (2x) battery screws using (+) PH#0 screwdriver.
Place a small standard screwdriver in the notch to gently pry
open the cover and slide out the battery.
②Insert the new battery following the polarity markings on thecover.
③Replace Battery Cover and tighten screws to secure.

Measurement RangeResolution0.01mm

Accuracy

0.02mm
0.003mm
Less than 1.5N
0.002mm
φ8mm
5〜40℃
CR2032 (Lithium cell) ×1
115g

Repeatability
Mounting Shaft Diameter
Operating Temp. Range
Power SupplyWeight

TROUBLESHOOTINGCorrective Action●＋Measured value is wrong

Clean measuring surfaces withlens tissue or similar.
● Confirm and re-set 0-Point setting.

●●

LCD does not display
Displayed value isunstable● Display is hard to readNotch●

NOTICE Use only CR2032 type battery.

POWER【Power ON】
Press [MODE], [SET], or Center Button to turn on power.
【Power OFF, Sleep Mode】
Press and Hold the [SET] button to enter Sleep Mode; continuepressing to turn power OFF.
1mm

0.01mm/0.001mm(switchable)

Displayed Resolution

Battery CoverＣＲ２０３２[SET] Button (Press and Hold)

DEI-WKS1
DEI-WKS2
0-12.5mm
0.01mm0.001mm

Model No.

Error Condition

Battery[SET] Button (Press and Hold)

SPECIFICATIONS

Measurement ForceBATTERY REPLACEMENT

Battery Cover● Made in Switzerland

mmSleep Mode
Power OFF※Sleep Mode is entered automatically if gauge is idles for about
20 min.
※To exit Sleep Mode, press any button or move the Spindle.
※When remaining battery level is low, the『 B 』icon is displayed;
please replace battery.
※Press and hold [SET] and [MODE] at same time for about 4
sec. to display “CLEAr” and reset all settings (except units, andresolution).

Cycle power by removing andreplacing battery.
● Replace battery with new one.

※If problem persists, or if you have any questions, please contact distributor
or place of purchase.
※Please note, manufacturer is unable to respond to inquires or provide
service directly. Please contact distributor or place of purchase.

HOW TO USE①Mount indicator securely, such as on optional Magnetic Base,and make sure that the Spindle is perpendicular to themeasurement surface.
②Press [MODE], [SET], or Center Button to turn on power.

③Check that the probe is in contact with surface, and refer to thefollowing to set the 0-Point.
Refer toRefer toFunction Setup: “Preset Value”

Programmable “Favorite” Button: “Zero Set”

LCD will display “0.00mm”④With Gauge body held fixed, and the probe in contact with theworkpiece, the LCD will display the difference between
workpiece and reference plane.
See Reverse for “FUNCTION SETUP”→

FUNCTION SETUP【Measurement Display】
Press the [MODE] Button, or Press-and-hold, to access and cycle through the functions
as show in the chart below. When the function is displayed, press Center Button to access.
※If no operation is performed after 5 seconds, display will return to measurement screen.

1mm[MODE] Button[MODE] Button【REF1 / REF2 Mode】
Two different Zero points, REF1 and REF2 canbe used (also Tolerance settings.) This feature ishelpful when there are multiple measurementcriteria in use at the same time.

1mm[MODE] Button【Display Mode】(MIN),
Set measurement mode:(MINMAX),or(NORMAL)Long Pressmm※“deltA” (MINMAX)→Display maximum displacement without regard to direction (TIR).

To Set[MODE] Button

Press Center Button to display options, then press[SET] to select.

To SetPress Center Button to display REF1 / REF2, thenpress [SET] to select.

1mm

1mm

1mm

Center Button【Preset Value】
Any number can be preset for use in setting display
during measurement by using this function.
PRESET INPUT1mmTOL

Return toMeasurement Display

1mm

Set Lower
Limit
TOL INPUT

Press Center Button to move the cursor, and [SET]
to change the value of that character. Press [Mode]to enter.
●Press [SET] in measurement screen to set display
the preset value.

[MODE] Button
1mm[MODE] Button●Center Button can be programmed to call up a frequently usedfunction with a single press while taking measurements.
※Programmed setting is reset when power is turned off.

※Tolerance values can be set for MIN,
【Tolerance Function】
MAX, and DELTA modes.
With tolerance values set, gauge will show if
measurement is within range.
Colored lights will show tolerance determination
for intuitive judgment.
Measurement: 2.25mm Measurement: 4.00mm Measurement: 7.25mm

Press and hold the Center “Favorite” Button to enter the button
programming selection mode.Press Center Button to cycle through theoptions and then press [MODE] or [SET] to select.
※This option
not used

Long Pressmm

Center Button

Center Button

Red light

Hold functionmmCenter ButtonmmCenter ButtonmmCenter Button

Reset MIN,MAX, DELTA
display
Zero setting

Center Button

Measurement: 2.25mm Measurement: 4.00mm Measurement: 7.25mm
Orange Light

Tolerance
Setting function
ON / OFF

Press Center Button to display 0.001 / 0. 01, thenpress [SET] to select.

【Measurement Direction】

mmTo Set※DEI-WKS2 only

Press Center Button to display
press [SET] to select.

[MODE] Button

Center Buttonmm

Calculation
functionmm

Center Button

Presetmm(−), thenTo Set※Set to 1.000
to disable

Press Center Button to move the cursor, and [SET]
to change the value of that character. Press [Mode]to enter.

[MODE] Button

Ex: If computation factor is set to “2.00", and measured value is “4.25”(actual amount of Spindle motion,) the displayed reading will be “8.50”(repeat cycle)【Auto Off】

Center Buttonmm

To SetCAUTION● This is a precision instrument, handle with care. Do not drop or subject

to excessive forces.
● Keep away form high voltage and strong magnetic fields; instrument
damage may result.
● Do not use or store in direct sunlight or locations with hightemperatures such as in a car, or near a stove or heat source.
● When taking measurements, hold gauge securely, such as inmagnetic base. (sold separately)● Please do not disassemble or modify.
● Remove dust and cutting chips after use and apply rust preventative
oil to metallic surfaces. Please keep out of reach of children.
● Do not use organic solvents to clean the body of the instrument.
● When not being used for a long period of time, please remove battery.
● Only use as directed. Improper use may cause accident or injury.

(＋) /【Calculation Function】
You can enter a number to multiply by theamount of Spindle movement.

Center Button

SwitchREF1 / REF2mm

Red light

To Setmm

Center ButtonOK Green Light【Resolution】※DEI-WKS2 only

[MODE] Button

Measurement
Direction ＋ / −mm

Orange LightTo reverse the display indication, reverse the entry of the Lower andUpper limits.

NOR, MIN,MAX, DELTA
mode

Resolution setting
0.01 / 0.001

mm

OK Green Light

Ex: For measurement Lower limit set to “+6”, Upper limit set to “+3”mmmm

Set UpperLimit

Press Center Button
to move the cursor,and [SET] to change
the value of thatcharacter. Press[Mode] to enter.

Ex: For measurement Lower limit set to “+3”, Upper limit set to “+6”

To SetmmTOLTOL INPUT

Programmable “Favorite” Button

Center Button

Center Button[MODE] Button

To Set[MODE] Button(MAX),[MODE] Button

Press Center Button to display “MAn” / “Auto”, andpress [SET] to select.
『MAn』…disabled『Auto』…Auto off if idle for 20 min.

【Button Lock】
For locking buttons while taking measurements.

mm

To Set[MODE] Button

Return toMeasurement Display

Press Center Button to display ON / OFF, thenpress [SET] to select.
●When lock is ON,『』icon will display on LCD.
●To release the lock mode, press and hold [SET]
until “UnLoc” is displayed.

Niigata seiki Co., Ltd.
6-15-22, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-31-5670 Fax. : +81-256-39-7730

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I300-K1604