Central Sterile Services

DEVICE

REFERENCE GUIDEOne Integrated Approach to Healthcare

Lumen Cycle

Non Lumen Cycle• Single or Dual Stainless Steel Lumen Devices- ≥ 0.77 mm internal diameter (ID) and ≤ 410 mm in length (L)• Triple Stainless Steel Lumen Devices- ≥ 1.2 mm ID and ≤ 275 mm L
- ≥ 1.8 mm ID and ≤ 310 mm L
- ≥ 2.8 mm ID and ≤ 317 mm L

• Non-lumened rigid andsemi-rigid endoscopes• Non-lumened flexible endoscopes• Stainless steel or Titanium diffusionrestricted spaces

Flexible Cycle• Single or Dual Lumen Devices- ≥ 1 mm ID and ≤ 990 mm L
• Polyethylene (PE)/PTFE (Teflon®) Tubing*- ≥ 1 mm ID and ≤ 4,000 mm L

Examples of Representative Devices

Refer to the complete Indications for Use and validation loads on the reverse side.

IMAGE
KEY LUMEN
LENS

Single Lumen
Endoscope

CameraLIGHT

Non-Lumened Flexible
EndoscopeSingle Lumen
Endoscope

LIMITSPER CYCLELIMITSPER CYCLE

12 POUNDS(5.4 kg)

11 POUNDS(5.0 kg)
12 LUMENS

Non-LumenedTelescope

LIMITSPER CYCLE
1 ENDOSCOPE

Dual Lumen
Endoscopeda Vinci®2Endoscopes

Dual Lumen
Endoscope

BatteryNon Lumen
Cycle

Triple Lumen
Endoscope

Lumen
Cycle

Flexible
Cycle
LIMITSPER CYCLE

OrthopedicDrill

Light Cord

14 piecesno additionalload

Scissors /Forceps(Teflon®) PTFE+PE
Tubing*See device matrix to identify the correct cycle for your device.

INSIDE LUMEN DIAMETERS.77
Millimeters0.030 Inches
2.3
French1.0Millimeters0.039 Inches
3.0
French2.00.078
6.0* Tubing claims have not been reviewed by the Food and Drug Administration

Millimeters
InchesFrench

3.0
0.118
9.0

MillimetersInches
French

4.00.157
12.0Millimeters
InchesFrenchPictorial representation of device measurements areapproximate and for reference only.

LUMEN LENGTHS

Lumen Length (410 mm)

0 mm

50 mm

100 mm

150 mm

275 mm
310 mm

317 mm

410 mmINDICATIONS FOR USE
The V-PRO 60 Low Temperature Sterilization System, with VAPROX® HC Sterilant, is a vaporized hydrogen peroxide
sterilizer intended for use in the terminal sterilization of cleaned, rinsed and dried reusable metal and nonmetal
medical devices used in Healthcare facilities. The pre-programmed sterilization cycles (Lumen Cycle, Non Lumen
Cycle, and Flexible Cycle) operate at low pressure and low temperature and are thus suitable for processing medicaldevices sensitive to heat and moisture.

V-PRO 60
Sterilizer Cycle

Indications for Use•

Instruments with diffusion-restricted spaces such as thehinged portion of forceps and scissors•

Non-lumened devices including non-lumened rigid andsemi-rigid endoscopes•

Medical devices, including single, dual and triple
channeled rigid and semi-rigid endoscopes, with thefollowing configurationsa:

1. Single or dual lumen devices with stainless steel
lumens that are ≥ 0.77 mm (~1/32”) internal diameter(ID) and ≤ 410 mm (16-9/64”) in length

Lumen Cycle
Can sterilizea:

Total Load

11 lbs (5.0 kgs)
and12 Lumens

2. Triple lumen devices with stainless steel lumens that are:≥ 1.2 mm (~3/64”) ID and ≤ 275 mm (~10-55/64”) in length≥ 1.8 mm (~5/64”) ID and ≤ 310 mm (~12-13/64”) in lengthor≥ 2.8 mm (~7/64”) ID and ≤ 317 mm (12-31/64”) in length•Non Lumen Cycle

Non-lumened devices including non-lumened rigid, semirigid and flexible endoscopes and non-lumened devices
with stainless steel or titanium diffusion-restricted spaces
such as the hinged portion of forceps and scissors.

12 lbs (5.4 kgs)

Can sterilizeb:•Flexible Cycle
Can sterilizec:

One flexible surgical endoscope or bronchoscope witha light cord (if not integral to endoscope) and mat without
any additional load. The flexible endoscope may be a:NASingle or dual lumen device with lumens that are > 1 mm(~3/64”) ID and < 990 mm (38-63/64”) in lengtha The validation studies for all lumen configurations were conducted using a maximum of twelve (12) lumens per load. Hospital loads should not exceed the maximum numberof lumens validated by this testing. The validation studies were performed using a validation load consisting of one instrument tray with instrument organizers and mat and two
pouches for a total weight of 11 lbs (5.0 kg).
b The validation studies were performed using a validation load consisting of one instrument tray with instrument organizers and mat and one pouch for a total weight of 12 lbs(5.4 kg).
c The validation studies were conducted with one flexible endoscope, packaged into a tray with silicone mat, instrument organizers and light cord (if not integral to scope) andno additional load.

Teflon is a registered trademark of the E.I. du Pont de Nemours and Company.
da Vinci® is a registered trademark of Intuitive Surgical, Inc.

1

2

Document # M4070EN.2016-03, Rev. B©2016 STERIS Corporation.
All rights reserved.