Thimble③ Replace Battery Cover.

Ratchet StopFrame0­Point Adj. Hole

[ON/OFF] Button● SSR44 (Silver Oxide【Rear】Batt.) ※For test

● Calibration Standard

Battery Cover※For 25­50mm Gauge only

APPLICATIONS・FEATURES・SPECIFICATIONS※ Accuracy specification does not include quantization error (±1 count)■DIGITAL DEEP THROAT MICROMETER

Model No.

Range
（mm）

Resolution（mm）Accuracy（µm）

MCD233-100U

0-250.001

6

MCD233-150U

0-250.001

6Flatφ6.5mm

MCD233-300U

0-250.001

8Flat / Roundφ6.5mm

Model No.

Range（mm）

Resolution（mm）

Accuracy（µm）

MeasuringSurface Profile

MCD230-25SA0-250.001

4φ2× 5mm

MCD230-25SB0-250.001

4φ3×10mm● For measuring thickness ofProfile

steel plate.

Insulating cover

ProfileSR5100U・150U300U■DIGITAL SMALL MEASURING FACE MICROMETER● Narrow measuring surfaces for measuring

diameter of splined shafts.

● Carbide tipped measuring surfaces.

Profile● Ratchet stop・Insulating cover

CORRECTIVE ACTION● Clean measuring surfaces withcloth.

● Confirm 0­Point and re­zero with

● LCD does not display.
● Displayedunstable.

value is● Display is not clear.

● Cycle power by removing andreplacing battery.

● Replace battery with new one.

※ If problem persists, or if you have any questions, please contact distributor
or place of purchase.
※ Please note, manufacturer is unable to respond to inquires or provide
service directly. Please contact distributor or place of purchase.

Press [ON/OFF] Button to turn power ON; measurement will display on the LCD. Press a second time to turn power OFF.

● Spanner

30[SET] Button

Insulating
Cover

25

[ABS/INC] Button【Accessories】

20

LCD Displaywrong.

value isPOWER

15ABS/INCSETON/OFF● Carbide tip・Ratchet stop・BatteryCover40Scale

ABS INC Set● Measured

long press of [SET] Button (> 2 s.)

Only use SR44 type battery.

045

5ERROR
OpenCAUTION

5030

Anvil (Stationary)SleeveClamp① Rotate the Battery Coveron rear using coin orsimilar tool.

② Insert battery with（＋）
side up.

※Model MCD230-25SA shown

MeasuringSurfaces Spindle

TROUBLESHOOTING25

PARTS IDENTIFICATIONINSERTING BATTERY

20

Thank you for purchasing the Niigata Seiki Digital Micrometer.
For proper operation, please read this manual thoroughly and follow the procedures described.

Model：MCD**-** ＜combined＞

15INSTRUCTION MANUALDIGITAL MICROMETER

MeasuringSurface Profile
Flatφ6.5mm

SETTING THE ZERO-POINT

Setting the origin used for normal measurements.
Once set, even when power is off the origin is saved and does not have to be reset with each use.
When battery is replaced, origin will go to default setting. Please reset.
① Remove any grease or dust from the measuring surfaces using a clean cloth, such as a lens cleaning cloth. Make sure noforeign matter is adhered to the measuring surfaces.
② With Spindle Clamp loose, turn until measuring surfaces touch. Turn until Ratchet Stop clicks 1 ­ 3 times.
※For 20­50mm range models, please insert the supplied Calibration Standard between measuring surfaces.
③ Check to Scale to see if it reads "0" and follow the procedure below if necessary.
To align the Thimble Scale 0­Point reference lines.
1. Tighten Spindle Clamp to lock Spindle.
2. Insert the spanner into 0­point Adjustment Hole on rear of Thimble.
3. Use spanner to rotate Sleeve to align the 0­Point lines.
④ Press the [SET] Button for more than two seconds.
LCD will display "SET" for 0.5 seconds, and then the display will read as shown below:● Measurement Range of 0­25mm･････ 0.000● Measurement Range of 25­50mm･････ 25.000
The gauge is now ready to take measurements.

OPERATION■DIGITAL BLADE MICROMETER● Blade shaped measuring surfaces formeasuring diameter of grooves.

● Ratchet Stop・Insulating cover■DIGITAL GEAR TOOTH MICROMETER

Profile● For measuring tooth span for spur andhelical gears.

● Measures gears of module 0.5 to 6.
● Ratchet stop・Insulating cover■DIGITAL TUBE MICROMETER● For measuring pipe wall thickness.
● Carbide tip ・Ratchet stop・Insulating cover■DIGITAL POINT MICROMETER● For measuring narrow groove OD, such asdrill web thickness.
● Carbide tip ・Ratchet stop・InsulatingcoverRange（mm）

Resolution（mm）Accuracy（µm）

MCD235-25BA

0-250.001

46×0.75mm

MCD235-50BA

25-50

0.001

46×0.75mm

MCD235-25BAU

0-250.001

46×0.4mm

MCD235-50BAU

25-500.001

46×0.4mm

Model No.

Range（mm）

Resolution（mm）Accuracy（µm）

MeasuringSurface Profile

MCD230-25D

0-250.001

4φ20mmMCD230-50D25-500.001

4φ20mmModel No.
ProfileMeasuringSurface Profile① Remove any grease or dust from the measuring surfaces using a clean cloth, such as lens cleaning cloth.
Make sure no foreign objects are adhered to the measuring surfaces.
② Place object to be measured between the Anvil and Spindle.
③ With the Thimble Clamp loose, turn the Ratchet Stop to capture the measured object between the measuring surfaces.
Stop after 1 ­ 3 Clicks from the Ratchet Stop.
④ The measured value is displayed on the LCD display.
※ Measurement can also be read off Thimble and Sleeve as with a standard micrometer.

INC MODEFor comparative measurements, the display reading can be zeroed at any position by pressing the [ABS/INC] Button.
In this mode, the zero point is retained in memory. Press the [ABS/INC] Button again to return to absolute mode.

CAUTION● Always clean measuring surfaces with soft cloth before use.

ProfileSR5

Model No.

Range（mm）

Resolution（mm）

Accuracy（µm）

MCD236-25T

0-250.001

4MeasuringSurface ProfileFlat / Roundφ6.5mm● This is a precision instrument, handle with care. Do not drop or subject to excessive forces.

● Keep away form high voltage and strong magnetic fields. Instrument damage may result.
● Do not turn spindle more than 3mm above upper measurement range.

● Keep instrument away from direct sunlight or high temperatures such as in a car, or near a heat source.
● Do not expose to water or oils.
● Do not disassemble or modify.

ProfileModel No.

MCD232-25P

Range
（mm）

Resolution（mm）Accuracy（µm）

0-250.001

4Measuring
Surface ProfileTaper angle~15°● After use, wipe with a soft cloth to remove any dirt or grime, and apply a rust preventative

oil before storage. Keep out of reach of children.

● Please do not use organic solvents to clean the body of the instrument.
● When not being used for long period of time, please remove battery.
● Only use as directed. Improper use may cause accident or injury.

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

アンビル（固定側）

測定面

スピンドル

スリーブ シンブル③電池蓋を元通りに装着してください。

ラチェットストップ

ご注意

5フレーム

45

540

基準線目盛

ABS INC Set

０点調整穴

ON/OFFボタン● 基準棒※測定範囲25-50タイプのみ※最大許容誤差は全て量子化誤差（±１カウント）を含みません。

■デジタルＵ字形鋼板マイクロメータ● 測定面超硬チップ・定圧装置・

※テスト用

電池蓋

用途・特長・仕様● 鋼板の厚み測定に。

● SR44（酸化銀電池）

【裏面】

測定面形状

防熱カバー付

品
測定面形状
SR5

100U・150U■デジタルスプラインマイクロメータ● スプラインシャフトの溝の径を測定するのに便利なように、両測定面を細くしてあります。

300U

測定面形状

● 測定面は超硬チップ付● 定圧装置・防熱カバー付

どを簡単に測定できます。
● 定圧装置・防熱カバー付■デジタル直進式歯厚マイクロメータ

測定面形状● 測定可能モジュール：0.5〜6● 定圧装置・防熱カバー付● 管の厚さ測定に。

● 測定面超硬チップ・定圧装置・防熱カバー付■デジタルポイントマイクロメータ●ドリルのウェブ直径などのような谷径の測定に。
● 測定面超硬チップ・定圧装置・防熱カバー付

最小読取値 最大許容誤差
（mm）
（µm）

測定面形状
平面φ6.5mm

MCD233-100U

0-250.001

6

MCD233-150U

0-250.001

6

平面φ6.5mm

MCD233-300U

0-250.001

8

片球面φ6.5mm品

番

測定範囲（mm）

最小読取値 最大許容誤差（mm）
（µm）

測定面形状

MCD230-25SA

0-250.001

4φ2× 5mm

MCD230-25SB

0-250.001

4φ3×10mm品
測定面形状● 平歯車、はすば歯車のまたぎ歯厚の測定に。

■デジタル片球面マイクロメータ

測定範囲（mm）●電池を入れ直してください。
●新しい電池と交換してください。

※症状が改善されない場合や不明な点がございましたらお買い上げの販売店、または発売元までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

原点の設定

通常測定で使用する原点を設定します。
この原点は、電源をOFFにしても記憶されますので、毎回設定し直す必要はありません。
電池交換後はリセットされますので、再度設定し直してください。
①アンビル・スピンドルの測定面を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。
②クランプを回してゆるめ、ラチェットストップを持って回し、測定面を合わせます。
測定面が合ってから、ラチェットが『カチカチ』と1〜3回鳴ったところで停止してください。
※測定範囲25-50mmタイプは、測定面の間に付属の基準棒をはさめて合わせてください。
③基準線目盛の『０』にシンブルの『０』目盛が一致していれば、スリーブ・シンブルの０点が合っています。
一致していない場合は、下記の手順で０点を合わせてください。
１.クランプを回してシンブルを固定する。
２.キースパナをスリーブ裏面の０点調整穴に入れる。
３.０点が一致するまで、キースパナでスリーブを回す。
④SETボタンを2秒以上長押ししてください。
LCD表示部に『Set』が0.5秒ほど点灯後、数字の表示が下記のように表示されます。
●測定範囲0-25mmタイプ･･････0.000●測定範囲25-50mmタイプ･････25.000
これで測定の準備ができました。

使用方法■デジタル直進式ブレードマイクロメータ● 測定面がブレードタイプで、丸物の細溝径な番●LCD表示部に何も表示されない●表示が安定しない●表示が不明確である

対応策●測定面の汚れをグラスペーパなどで拭き取ってください。
●0点が合っているか確認し、SETボ
タンを2秒以上長押ししてください。

ON/OFFボタンを押すと電源がONになり、LCD表示部に数値が表示されます。もう一度押すと電源がOFFになります。

● キースパナ

30

SETボタン25

ABS/INCボタン
防熱カバー

【付属品】

20

LCD表示部

トラブルの内容

測定値が間違っている

電源

15ABS/INC
SET
ON/OFF

電池蓋

電池は必ずSR44をお使いください。

00

開ける

②電池を必ず（＋）側を表にしてセットしてください。

※イラストはMCD230-25SAです。

クランプ①本体裏面の電池蓋を、矢印の向きにコインなどで回して開けます。
30

各部の名称

トラブルと対応25

この度はデジタルマイクロメータシリーズをお買上げ頂きありがとうございます。
ご使用前に、本書をよくお読み頂き正しくお使いくださいますようお願い申し上げます。

電池の入れ方20

取扱説明書

品番：MCD**-** ＜兼用＞15

デジタルマイクロメータ

測定面形状
SR5

測定面形状

番

測定範囲（mm）

最小読取値 最大許容誤差（mm）
（µm）

測定面形状

MCD235-25BA

0-250.001

46×厚 0.75mm

MCD235-50BA

25-500.001

46×厚 0.75mm

MCD235-25BAU

0-250.001

46×厚 0.4mm

MCD235-50BAU

25-500.001

46×厚 0.4mm

品

番

測定範囲
（mm）

最小読取値 最大許容誤差（mm）
（µm）

測定面形状

MCD230-25D

0-250.001

4φ20mmMCD230-50D

25-500.001

4φ20mm

品番

MCD236-25T

品

番

MCD232-25P

測定範囲
（mm）

0-25

測定範囲
（mm）

0-25

最小読取値 最大許容誤差（mm）
（µm）

0.001

4

最小読取値 最大許容誤差（mm）
（µm）

0.001

4

測定面形状
片球面φ6.5mm

測定面形状
先端テーパ角度約15°①アンビル・スピンドルの測定面を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。
②アンビルとスピンドルの間に被測定物を入れます。
③クランプを回してゆるめ、ラチェットストップを持って回し、測定面で被測定物を挟みます。
ラチェットが『カチカチ』と1〜3回鳴ったところで停止してください。
④LCD表示部に表示された数値が､測定値となります。
※通常のマイクロメータ同様にスリーブとシンブルの目盛を読み取ることもできます。

INCモードABS/INCボタンを押すことで､原点を維持したまま表示をゼロリセットすることができます。比較測定などに便利です。
原点からの測定値に戻りたい場合は、再度ABS/INCボタンを押してください。

ご注意●ご使用の前に、必ずグラスペーパなどで測定面の汚れを除去してください。
●測定範囲の上限を3mm以上上回って、スピンドルを回さないでください。
●精密機器ですので、落としたりぶつけたり、強い力を加えたりしないでください。
●強い磁気や高電圧に近付けないでください。故障の恐れがあります。
●直射日光の当たる場所や炎天下の車の中、火のそばなどでの使用、保管はしないでください。
●水や油のかかるところで使用しないでください。
●分解・改造をしないでください。
●使用後は柔らかい布などでゴミや切粉などを取り除き、防錆油を塗布してお子様の手の届かない場所へ保管してください。
●本体の手入れに有機溶剤を使用しないでください。
●長期間使用しない場合は、本体から電池を外して保管してください。
●用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。

発売元〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I96-K32101