M200D

M-SERIES

M200D

DUAL-ENERGYCOMPACTUP TO 5 MW RF PEAK
POWER
FULLY COMPUTERIZEDMINIMIZEDCONSUMABLESMAGNETRON MODULATOR UP TO 5 MW
ScandiNova's M200D is a dual-energy pulse Modulator
in a very compact enclosure. Our dual-energy models
are for applications where there is a need to switch
between two different energy levels. The Modulator is
optimized for a wide range of different magnetrons withan RF peak power ranging from 2.5 MW to 5 MW, andcan also power klystrons. The M200D is also available asa single-energy, stand-alone version (see M200).
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers a user-friendly interface. The pulse length,
amplitude and repetition rate can all be changed.
The M200D can be delivered as a pure high-power
pulse modulator or as part of an RF Unit including
magnetron and related components. Different levels ofservice and support programmes are available.

THE M-SERIES
ScandiNova's M-series
contains a range of magnetronpulse modulators with highreliability and performance,
a compact design and low
energy consumption.
Our magnetron pulse
modulators require about
one third of the space ofalternative solutions yet still
offer around 30% higher
efficiency, two operating
advantages that especiallyappeal to cancer treatment
providers.
Their high reliability andperformance also attract manyindustry applications.

Excellence in pulsed powerwww.scandinovasystems.com

M200DUNIT

VALUE

NOTES

Magnetron RF peak power, highMW

2.5-5

Depends on choice of magnetron

Magnetron RF peak power, lowMW

1–3

Depends on choice of magnetron

Magnetron RF average powerkW

7Maximum, depends on choice of magnetron

Modulator peak power, highMW

14Maximum

Modulator peak power, lowMW

6.2

Maximum

Modulator average power

kW

16Maximum

Pulse voltage

kV40-75

Pulse current

A30-250

RF pulse length, top

μs

0.5-5

Depends on max average power

Pulse repetition rate

Hz0-500

2x250 Hz, depends on max average power

Modulator Voltage Stability, RMS%

0.4

Verified on resistive load (see options)

Water coolingl/min818-40 °C, non-condensing

INTERFACECONNECTOR

DEFAULTOPTION

Mains power, three-phase

Terminal block (4-wire)

400±10 % VAC, 50/60 Hz

208/380/480 VAC

Mains power, single-phaseC20

230±10 % VAC, 50/60 Hz

115 VAC

Control interface

RJ45

Modbus TCP

Water cooling interface

12 mm

Legris Push-in

SwagelockTrig input

BNC

Electrical

OpticalDiagnostics

BNC

Pulse voltage & current signals

SIZE AND WEIGHT
Weight (incl. oil)

UNIT
kg

VALUE
200

The Standard Modulator Includes

Services

Filament power supply

Training in handling, operationand maintenanceControl system
For more information, visit www.scandinovasystems.com/M200

Control via Modbus TCP
Pulse sensors
HV pulse cable

544 mm

DOC-004509-02 / 2024-01-19

SPECIFICATIONS

1267 mm

647 mmService contract

Typical Magnetron Loads

Factory acceptance test certificate(M200D can also power klystrons)

Manuals

BVERIOptions

VE2116

Enhanced PRF Range (1000–2000 Hz)
Enhanced Stability (down to 0.1 %)Digitizer

Information contained in this document is subject to change
without notice.

Installation and start-up

Internal trig generator

Pulse/RF diagnostics

Contact ScandiNova for any requirements not covered bythis specification.

Shipping

Additional System Components
Circulator & RF loadsE2VM5028
MG8076

GLVAC
GLM8028

Directional coupler
Magnet power supply
Waveguide windows
Magnetron

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power