Model

INSTRUCTION MANUALPRECISION OUTSIDEMEASURING INSTRUMENT

MC122-25C/50C＜comb＞

No.

COUNT MICROMETER

Thank you for purchasing the Niigata Seiki COUNT MICROMETER.
This precision gauge measures outside dimension of object placed between the Anvil and Spindle.
●For safe and proper use of this product, please read this instruction manual before use and follow the procedures
described. Please keep manual where it is accessible to user for future reference.
●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact dealer or Niigata Seiki.

■ PART IDENTIFICATION
0-Point
Adj. HoleMeas.Surfaces（Carbide Tips）

Anvil（Fixed Side）

SpindleClamp

SleeveThimbleFrame205250300【REAR】【FRONT】

30 5 0 2
mm145

Ratchet Stop
Scale
Counter Cover Plate
Counter

Thermal Protection Cover【ACCS.】

SPECIFICATIONS
Model No.
MC122-25C
MC122-50C

Meas.Range Resolution Maximum permissible Weight
（g）
（mm）
（mm）
error（µm）
0〜 250.01
4220
25 〜 500.01

4300

FEATURESOPERATION●Mechanical digit counter for quickand error-free reading●Carbide measuring faces●With ratchet stop
●With thermal protection cover①Use a lens cleaning cloth or similar to clean the Anvil and Spindle ofany grease or debris.
②Loosen the Clamp and place the workpiece between the Measuring
Surfaces.
③Rotate the Thimble using the Ratchet Stop until the workpiece contacts
the Measurement Surfaces and the Ratchet clicks 1〜3 times. （Fig.7）④A rough reading is displayed on the Counter display.
※For a precise measurement please read the value shown on theSleeve/Thimble Scale.（Fig.8）

HOW TO USE
ADJUSTING SLEEVE/SPINDLE ZERO SETTING①Remove any grease or dust from the measuring surfaces using aclean cloth, such as a lens cleaning cloth.
Make sure no foreign matter is adhered to the measuring surfaces.
②With Spindle Clamp loose, turn until measuring surfaces touch.
Turn until Ratchet Stop clicks 1-3 times.（Fig.1）※For MC122-50C, place the provided Calibration Standard between thevq

30 0 0 2mm1

Fig.１vquv

ThimbleFig.２

Rotate until origins match
５０１００ ０ ００４５

Remove４０４５４０Spanner

Fig.３

CAUTION

３５１１１ ００００ ９９９１Counter
Module Screws

９Fig.４
Rotate the gear to set thereference value:（MC122-25C：0000）（MC122-50C：2500）
Set counter to 0002〜0003
above ref. value.

Gear

３１２０ １００ ９９ ０
９ ８８１１８Fig.５

Counter Module

Hold by outer frame.
５１１０１ ３０ ０ ０２

９ ９ ９１４５４０Fig.６
Workpiece“Click Click”
uqq81 2 8 7mm6Indicates risk of personal injury or property damage
if not followed.

●Always clean measuring surfaces with soft cloth before use.
●Do not turn spindle more than 3mm above upper measurement range.
●When turning Thimble, be gentle and do not apply force. High force on Thimble willcause inaccuracy.
●This is a precision instrument, handle with care. Do not drop or subject toexcessive forces.
●Keep instrument away from direct sunlight or high temperatures such as in a car,or near a stove or heat source.
●Do not expose to water or oils.
●Do not disassemble or modify.
●After use, remove any chips or debris using a soft cloth and take appropriate
rust-preventive measures. Keep out of reach of children.
●Please do not use organic solvents to clean the body of the instrument.
●Only use as directed. Improper use may cause accident or injury.
※If problem persists, or if you have any questions, please contact distributor.
※Please note, manufacturer is unable to respond to inquires or provide service directly.
Please contact distributor.

v

rqtvtqRatchet StopFig.７⫰⫱PLEASE OBSERVE

Always follow the procedures specified below in order to prevent harm toyourself or others, and to prevent damage to property.

Match origin lines

30 0 0 2mm1

SAFETY

５

Cover Plate

uquv

Ratchet Stop

Measuring Surfaces.

③If the zero line on the Thimble lines up with the line on the Sleeve, thenthe origin is set.（Fig.2）
If they do not line up, adjust the zero-point according to the procedure
below.
1. Tighten Spindle Clamp to lock Spindle.
2. Insert the spanner into 0-point Adjustment Hole on rear of Sleeve.
3. Use spanner to rotate Sleeve to align the 0-Point lines.（Fig.3）“Click Click”

Meas. surfaces touch Clamp

Cover Plate
ScrewsInsert※Model MC122-25C shown

Calibration Std.

（※For MC122-50C only）

Spanner

COUNTER DISPLAY SETTING
Gauge includes a Mechanical Counter for use as a quick reference
display in addition to the measurement shown conventionally on theSleeve Scale. Please note the following points about the measurement
displayed on the Mechanical Counter:
1. Use the reading from the Mechanical Counter only as a guide.
Since the Counter is a gear driven mechanism, there is somebacklash error and it will not necessarily match the reading on theSleeve/Thimble. When there is a discrepancy, please treat themeasurement shown on the Sleeve Scale as the correct value.
※The specifications listed for this product are for the measurement displayedon the Sleeve Scale. The Counter reading is only to be used as a guide.
2. In the unlikely event that the Counter and Scale readings deviatesignificantly, use the following procedure to adjust the Counter display.
Please note, before adjustment check that the Sleeve/Thimble zerosetting is correct.
※For MC122-50C, use the provided Calibration Standard between theMeasuring Surfaces.
＋ type screwdriver, remove the two screws①Using a precision ○securing the Counter Cover Plate, and then the two additionalscrews holding the Counter Module in place.（Fig.4）②Remove the Counter for access, and rotate the gear to adjust. Set thecounter to 0.02〜0.03 above the reference value.（Fig.5）③Hold the Counter Module by the outer frame without touching thecounter dials or gears, and replace the Module into the Micrometer
body.（Fig.6）※If you touch the counter dials or gears the value may change whenbeing inserted.
④Replace the Counter Module screws to secure, and then replace theCounter Cover Plate and screws.

qv

rqtvtqSleeve Reading：１２．
５
Thimble Reading：．
３７
Measurement：１２．
８７(mm)

Fig.８

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

取扱説明書精密外側測定器

品番 MC122-25C/50C ＜兼用＞

カウントマイクロメータ

この度はカウントマイクロメータをお買上げいただきありがとうございます。
この商品はアンビルとスピンドルで測定対象物を挟み、外側長さを測定する精密測定器です。

測定面

０点調整穴

スピンドル クランプ

アンビル（固定側）（超硬チップ）フレーム

20250300

【裏 面】

【表 面】

4530 5 0 2mm1

ラチェットストップ

基準線目盛

カウンタ保護パネルカウンタ
防熱カバー

【付属品】
キースパナ

※イラストはMC122-25Cです。

基準棒（MC122-50Cのみ）

製品仕様品番MC122-25C
MC122-50C

特長

測定範囲（mm）

0 〜2525 〜50

最小読取値（mm）

最大許容誤差（μm）

質量（g）

0.0142200.014300●機械式カウンタ表示により読み間違いがありません。
●測定面は超硬チップ付●定圧装置付●防熱カバー付

使用方法
スリーブ・シンブルの０点の合わせ方①アンビル・スピンドルの測定面を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。
②クランプを回してゆるめ、ラチェットストップを持ってシンブルを回し、測定面を合わせます。測定面が合ってから、ラチェットが『カチカチ』と1 〜 3回鳴ったところで停止してください。（図１）※MC122-50Cは、測定面の間に付属の基準棒を挟んで合わせてください。
③基準線目盛の『０』にシンブルの『０』目盛が一致していれば、原点が合っています。（図２）
一致していない場合は、下記の手順で原点を合わせてください。
１. クランプを回してシンブルを固定する。
２. キースパナをスリーブ裏面の０点調整穴に入れる。
３. 原点が一致するまで、キースパナでスリーブを回す。（図３）

クランプ

測定面を合わせるカチカチv

q

30 0 0 2mm1uv

ラチェットストップ

図１

原点が合っていることを確認v

quv

30 0 0 2mm1

シンブル

図２

原点が合うまで回す
５０

１００ ０ ０４５４０

キースパナ

図３

カウンタ
保護パネル
固定ねじ

カウンタ
保護パネル

５０４５外す

４０

３５１

カウンタ
モジュール
固定ねじ

１
１ ００００ ９９９１９

図４歯車を回して、基準となる値（MC122-25C：0000）（MC122-50C：2500）に0002〜0003をプラスした
数値にカウンタを合わせてください。

歯車３１２０ １００ ９９ ０
９ ８８１１８

図５カウンタモジュール

カウンタに触れないように
５１１０１ ３０ ０ ０２

９ ９ ９１

４５４０

戻す

2. 万一カウンタ表示が大きくズレた場合、以下の手順によって数値表示の調節を行なってください｡
また、本作業の前にスリーブ・シンブルの０点が正しいことを確認し、正しくない場合は左記の『スリーブ・シンブルの０点の合わせ方』に従って調整してください｡※MC122-50Cについては、必ず付属の基準棒を使用してご確認ください｡＋の精密ドライバで外し、さらにカウンタモ①カウンタ保護パネルの固定ねじを○＋の精密ドライバで外してください。
ジュールを固定しているねじも同様に○（図４）②カウンタモジュールを取り出し、歯車を回して基準となる値に0.02〜0.03をプラスしたところに合わせてください。（図５）③カウンタや歯車に触れないようにカウンタモジュールの外側フレームを持ち、マイクロメータ本体に戻してください。（図６）※この時にカウンタの歯車に触れた場合、調整したカウンタ表示が変化する場合があります｡④カウンタモジュール、カウンタ保護パネルを、それぞれの固定ねじで固定し元に戻してください。

スリーブ シンブル
5

本商品には機械式カウンタが付いており、測定結果については、従来の方式による目盛の読み取りに加えて、カウンタの数値表示で確認することができます｡
カウンタの数値表示については以下の点についてご注意いただきますよう、お願い致します｡

1. 機械式カウンタは、目安としてお使いください｡
本商品のカウンタは歯車駆動の機械式のカウンタを使用しているため、バックラッシュによってカウンタの表示とスリーブ･シンブルの目盛の値が必ずしも一致しない場合があります｡
この場合はスリーブ・シンブルの表示値を、正しい値として扱ってください｡※当社での最大許容誤差保証は目盛の値となっており、カウンタは目安的な位置付けとなっております｡●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称

カウンタの調整

使用方法

図６①アンビル・スピンドルの測定面を脱脂し、グラスペーパなどで拭き取って異物が付着していない状態にしてください。
②クランプを回してゆるめ、アンビルとスピンドルの間に被測定物を入れます。
③ラチェットストップを持ってシンブルを回し、測定面で被測定物を挟みます。ラ
チェットが『カチカチ』と1〜3回鳴ったところで停止してください。
（図７）④カウンタに大まかな測定値が表示されます。
※精密な測定値が必要な場合は、スリーブとシンブルの目盛を読み取ってください。（図８）

安全上のご注意

カチカチuqq8
1 2 8 7mm6v

お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。

●ご使用の前に、必ずグラスペーパなどで測定面の汚れを除去してください。
●測定範囲の上限を3mm以上上回って、スピンドルを回さないでください。
●シンブルは強い力を加えずに、ゆっくりと回転させてください。精度不良の原因となります。
●精密機器ですので、落としたりぶつけたり、強い力を加えたりしないでください。
●直射日光の当たる場所や炎天下の車の中、火のそばなどでの使用、保管はしないでください。
●水や油のかかるところで使用しないでください。
●分解・改造をしないでください。
●使用後は柔らかい布などでゴミや切粉などを取り除き、防錆処理をしてお子様の手の届かない場所へ保管してください。
●本体の手入れに有機溶剤を使用しないでください。
●用途以外のご使用は、事故やけがの原因となりますので絶対におやめください。
※症状が改善されない場合や不明な点がございましたらお買い上げの販売店、または当社までご連絡ください。
※お問い合わせや、ご連絡が無いまま直接当社に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

rqtvtqラチェットストップ

図７

必ずお守りください

お使いになる人や他の人への危害、財産への損害を未然に防止するため、必ずお守りいただくことを下記のように説明しています。

注意

被測定物を挟む⫰⫱uqqv

rq

tvtq

スリーブの読み：１２．
５
シンブルの読み：．
３７
読み：１２．
８７(mm)

図８〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

I82-K1 2308