INSTRUCTION MANUAL

S-LINE DIGITAL MICROMETER

Model No. : MCD-30IP67S / MCD-66IP67S / MCD-102IP67S <combined>

Thank you for purchasing the S-Line Digital Micrometer.
Please read this manual thoroughly before use for proper operation.

FEATURES

SPECIFICATIONS● Waterproof protection grade: IP67
● Adjustable measuring force (5N / 10N) [MCD-30IP67 only]● Spindle moves 12mm with one rotation allowing
quick measurement of objects of different dimensions.
● Wide measuring range. Only three sizes needed to cover
102mm measuring range.
● Constant Force Mechanism: Thimble Ratchet● Measuring Surfaces: Flat Carbide Tip● Spindle : Straight Feed, φ6.5mm Cylindrical Spindle● Preset Function● Sleep Mode Function (approx. 20 min.)● Locking Function● Made In Switzerland

Part Number

PART IDENTIFICATION

Measuring Range

MCD-30IP67S

MCD-66IP67S

MCD-102IP67S

0-30mm30-66mm66-102mm

Resolution

0.001mm

Accuracy

0.003mm

0.004mm5N/10N(switchable)

Measuring Force
Flatness

0.6μm

Parallelism

2μm

3μm

Repeatability

0.002mm

Waterproof Protection GradeIP67

Operating Temperature Range

5〜40℃

WeightMeasuring SurfacesAnvil(Carbide tip)

(Fixed side)

0.005mm
10N

280g

440g

Spindle560g

LCD Display

ThimbleFramemm

MODE

Battery Cover

MCD-30IP67SSETInsulating Cover[SET] Button
0 - 30mm
0.001mm【BACK】IP67
SWISS MADE

[MODE] Button【FRONT】

IP CODE SUMMARY【ACCESSORIES】

IP rating for electrical equipment defined by IEC standard for water and solid
body protection.
MICROMETER MCD-***IP67S

ＩＰ６７
2nd Digit : Water proof rating
1st Digit : Direct contact-protection rating for ingress of solid foreign object

CALIPER D-***IP67S

Battery cover tool
1st Digit

６
2nd Digit７Description
Solid IngressProtection

Description
Liquid IngressProtection

Degree of Protection
No ingress of dust allowed.
Dust tight.

Test Method
Complete protection from dust
intrusion

Degree of protection

Test Method

No harmful effect when immersed
in water for specified period andpressure.

Submersion for 30min.
at depth of 1m(for devices of height less than 850mm)

TROUBLE SHOOTING

ＣＲ２０３２Calibration Standard(Excluding MCD-30IP67S)
MCD-66IP67S:50mm
MCD-102IP67S:75mm

CR2032 (Lithium Cell)※ for testing purposes

BATTERY REPLACEMENTERROR CONDITION

CORRECTIVE ACTIONLCD display reads “ERR0”

Reading Error. Please reset zero-point.

LCD display reads “ERR3”

Internal Calculation Error. Please reset zero-point.

LCD display reads “ERR8”

Memory error during storage of preset data.
Please retry saving preset data.
※ If error persists, the memory may be damaged please contact distributor or place of purchasefor service.

LCD display reads “ERR9”

Memory error. Please contact distributor or place ofpurchase for service.

Measured value is wrong● Clean measuring surfaces with cloth.
● Check to insure zero-point is correct andre-set zero-point.

● LCD does not display● Displayed value is unstable● Display is not clear● Cycle power by removing and replacing
battery.
● Replace battery with new one.

※ If issue does not resolve, or if you have any questions, please contact distributor or place of purchase.
※ Please note, manufacturer is unable to respond to inquires or provide service directly. Please contact
distributor or place of purchase.

① With the battery cover tool, unscrew the battery cover using
the side marked as “MICROMETER MCD-***IP67S”.
② Insert the new battery with the (+) side outward.
③ Replace the cover using the Battery Cover Tool.

CautionRotateUse CR2032 battery.

Battery Cover Tool.

Battery CoverReverse Side -- How to Use →POWER

OPERATION【POWER ON】
Power is turned on by pressing either the [MODE] or [SET] buttons one time.

【POWER OFF - Sleep Mode】
Press and Hold the [SET] button to enter sleep mode. Continuing to hold thebutton will power off the unit.

①Remove any grease and dust from the measuring surfaces with a clean
cloth, such as lens cleaning cloth. Make sure no foreign objects are adhered
to the surfaces.
②Place object to be measured between the anvil and spindle.
③Turn the thimble to capture the measured-object between the measuringsurfaces. Please stop after 2~3 “Clicks” from the spindle ratchet.
④The measured value is displayed on the LCD display.

mmmmmm[SET] Button Press andHold[SET] Button Press andHold

Measured-object between measuring surfaces

Sleep Mode

“Click, Click”

Power Off

※Sleep mode is activated automatically after approximately 20 min. of no activity.
※In sleep mode, pressing any button or turning the spindle returns unit to normal operation.
※When battery power is low, 『 B 』indicator appears. Please replace themm

MODE

SETThimblebattery.

IP67

0 - 30mm
0.001mm

SWISS MADESETTING FUNCTIONS
Pressing the [MODE] button switches the display in the following sequence.
Pressing the [SET] button allows modifying the displayed function.

【Measurement Display】※Please wear dust free gloves for this operation.

Used during measurements.

①With power off, open thimble cap with coin or similar tool.

mm[MODE] Button

MEASUREMENT FORCE ADJUSTMENT
For MCD-30IP67 only (Measurement force can be set to 5N or 10N.)②Turn thimble as shown to remove spindle.

【PRESET FUNCTION】
A preset value can be input and then recalled using the[SET] button.

mm

To Set:[MODE] Button②

Press [SET] button to modify, then use [MODE] button
to change position of cursor and [SET] button tochange value at the cursor. When cursor is moved past
final digit instrument returns to measurement mode.

①Remove●In measurement mode, press and hold the [SET] button
until “SEt” is displayed. When released the preset value
will display and instrument reading will be set to thatvalue.

【LOCK FUNCTION】
Locks buttons while allowing instrument to take measurements.

ThimbleSpindle③While squeezing spindle end from both sides, rotate outer plate
under square plate as shown in diagram to the desired setting.
Reinstall spindle to thimble and replace thimble cap.

mm

To Set:Press [MODE] button for this option; press[SET] button to enter lock mode.

●When lock function is on,[MODE] Button『』 will be displayed.

●To unlock the instrument buttons, press and holdthe [SET] button until “Unloc” is displayed.
Squeeze

Turn While Squeezing【HOLD FUNCTION】
When enabled, [SET] button will hold measured value on thedisplay.

mm

To Set:[MODE] Button

Measuring force=10N

Measuring force=5N

Press [MODE] button for this option; press[SET] button to turn On/Off.

●Momentarily pressing [SET] during measurement will holdthe value and display『H』on LCD. Press [SET] a second
time to release the held value.
※If hold function is off, the [SET] button can not be usedduring measurement to hold the value.
※When changing functions, after five seconds of inactivity the instrument will return tomeasurement mode.
※While inputting preset value, there will be an error if there is no operation for 15 seconds.

CAUTION● Always remove dirt from measuring surfaces with lens cloth before

use.
● Do not turn spindle more than 3mm above upper measurement

range.

ZERO POINT SETTING● This is a precision instrument. Handle with care. Do not drop orsubject to excessive forces.

①Wipe measuring surfaces of anvil and spindle with clean cloth toremove any grease or contamination.

● Keep away form high voltage and strong magnetic fields. Instrument②Turn spindle to bring measuring surfaces together until thimble“clicks” 2~3 times.

● Keep instrument away from direct sunlight or high temperatures such

※Excepting MCD-30IP67S, please use
supplied calibration standard between
measuring surfaces.

● When not in use keep spindle 1-2mm from closed position.
● Please do not disassemble or modify.
● Remove dust and cutting chips after use and apply rust preventative

Measuring Faces In Contact③Press the [SET] button until “SEt”
is displayed. When released,the following values will display.

damage may result.
as in a car, or near a stove or heat source.

“Click, Click”

oil to metallic surfaces. Please keep out of reach of children.
● Please do not use organic solvents to clean the body of themmMODE

0 - 30mm
0.001mm

IP67

SETThimble※If appropriate value does not
display, please refer to “PRESET FUNCTION” above. Enter the proper
value from the following list, and repeat steps ①〜③.
●MCD-30IP67Săăă0.000 ●MCD-66IP67Săăă50.000●MCD-102IP67Săăă75.000
Instrument is now ready for measurement.
SWISS MADE

instrument.
● When not being used for long period of time, please remove battery.
● Only use as directed. Improper use may cause accident or injury.

Niigata seiki Co., Ltd.
6-15-22, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-31-5660 Fax. : +81-256-39-7730

MAIL:intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I214-K1411