DIGITAL S-LINE TEST INDICATOR（IP65）

INSTRUCTION MANUAL
Model No. DET-234S1/DET-234S2 <combined>

Thank you for purchasing the S-Line Digital Test Indicator.
For proper operation, please read this manual thoroughly before use.

PARTS IDENTIFICATIONFEATURES
Battery CoverEasy to read from any position.

LCD Display

Zero Point can be set
by rotating display.

● IP65 class waterproof rating.
● Display rotates 360°

● Digital LCD with analog bar display
mmNORx1

MODE[MODE] Button● Small size and wide range probe angle for use in tightspaces.

Display Rotates 360°

● MIN、MAX、MIN MAX (DELTA, or TIR) functions● Direction of measurement (＋ / −) switchable

SETSET● Preset function● Auto-off feature (approx. 2 hour)

SWISS MADE

[SET] Button

Accessories※when enabled● Made in Switzerland

Stem
ＣＲ２０３２

SPECIFICATIONSCR2032(Lithium Cell)※Included for test

Tool for changing
Contact Point

DET-234S1
DET-234S2
0-0.8mm0-0.5mm0.001mm0.01mm/0.001mm (switchable)

Model No.
Measurement Range

Contact Point

Resolution
Displayed Resolution

φ8mm Stem(Dovetail mount)

0.01mm
0.004mm0.005mm
Less than 0.2N0.003mm
YES
Stem: φ8mm
12.5mm
36mm
5〜40℃
70g

Accuracy
Backlash (Retrace) Error
Measurement Force
Repeatability

BATTERY REPLACEMENT①Remove the (2x) battery screws using (+) PH#0 screwdriver.
Place a small standard screwdriver in the notch to gently pry
open the cover and slide out the battery.
②Insert the new battery following the polarity markings on thecover.
③Replace Battery Cover and tighten screws to secure.

Measurement Dir. Switchable
Mounting Shaft DiameterContact Point Length
Operating Temp. Range
WeightTROUBLESHOOTING
Error Condition

Corrective Action“Err 0” is Displayedon LCD

Read Error.
Reset the sensor by pressing the[MODE] or [SET] button.
※If message is displayed again,
there is possibility that thememory is corrupted; Pleasecontact distributor or place ofpurchase.

Notch

Battery CoverBattery CoverBattery
ＣＲ２０３２＋NOTICE Use only CR2032 type battery.

POWER【Power ON】
Press [MODE], or [SET] button to turn on power.
【Power OFF】
Press and hold the [SET] button to turn off power.
※Power will switch off automatically if idle after approximately 2hours.
(※If Auto Off is enabled.)※When battery level is low, the『 B 』icon is displayed; please
replace battery.

Measured valueis wrong●LCD does notdisplay● Displayed value isunstable● Display is hard toread●

Clean measuring surfaces withlens tissue or similar.
● Confirm and re-set 0-Point setting.
●

Cycle power by removing andreplacing battery.
● Replace battery with new one.

※If problem persists, or if you have any questions, please contact distributor
or place of purchase.
※Please note, manufacturer is unable to respond to inquires or provide
service directly. Please contact distributor or place of purchase.

See Reverse for “How to Use”→

FUNCTION SETUPHOW TO USE

Press the [MODE] button to cycle through the functions in thesequence shown below. When desired function is displayed,press [SET] button to access the various settings.

mm

NOR

x 1 SETNOR

SET【Preset Function】
Preset Value set can be called up fordisplay.
※Analog display will also be set.

[MODE] Button

To SetNOR

SET[MODE] Button

mm

NOR

SET

Press [MODE] to display “NOR”, “MIN”,“MAX”, or “MIN MAX”, then press [SET]to select.

【Preset Value】
Numerical value can be preset, andlater called up to measurement screen.
To SetPress [MODE] to move the cursor,
and press [SET] to change value.

[MODE] Button【Analog Display Scale】
Scale of analog display can be set.
mm

NOR

Direction of Measurement

90°

90°

Direction of Measurement

Direction of Measurement②Press [MODE], or [SET] to turn on power.
【Display Mode】

mm

Direction of Measurement※MIN MAX(DELTA)
Displays total displacement, without
regard to direction (TIR).

[MODE] Buttonmm【Measurement Screen】
Four display modes: NOR(NORMAL),MIN, MAX, or MIN MAX(DELTA).

①Mount securely, such as on optional Magnetic Base, and makesure that the Contact Point is parallel to the surface being
measured, and perpendicular to the Measurement direction.

SET※If “0.00mm” or “0.000mm” is not displayed, please refer to“Function Setup - Preset Value” to set value to 0.00mm, andthen perform steps ① to ③.

④With Gauge body held fixed, and the probe in contact with theworkpiece, the LCD will display the difference between
workpiece and reference plane.
※If gauge can not be set up with contact point parallel tosurface, the measurement will have a “cosine error.”
You can correct for this error using the following table.

To SetPress [MODE] to display “F1”, “F2”,or “F5”, then press [SET] to select.

『F1』…Analog display
1 div.→0.001mm『F2』…1 div.→0.002mm『F5』…1 div.→0.005mm[MODE] Button③Setting 0-Point
With gauge in contact with measurement surface, press the[MODE] button. When display reads “SEt”, press the [SET]button.
(ref. “Function Setup - Preset Function”)
Display will read “0.00mm” or “0.000mm.”

displacement

displacement＝measurement reading × cosααα【Measurement Direction】
To SetmmNORSET[MODE] Button

Press [MODE] to display “Inc” (+) /“dEc” (-), then press [SET] to select.

To SetNOR

SET[MODE] Button

Press [MODE] to display 0.01 / 0.001,
then press [SET] to select.

To SetNOR[MODE] Button

CAUTION● This is a precision instrument, handle with care. Do not drop or● Keep away form high voltage and strong magnetic fields;

instrument damage may result.

● Do not use or store in direct sunlight or locations with hightemperatures such as in a car, or near a stove or heat source.

● When taking measurements, hold gauge securely, such as inmagnetic base (sold separately.)● Please do not disassemble or modify.

【Auto Off】
mm

cos α
0.996
0.985
0.966
0.866
0.707
0.500

subject to excessive forces.

【Resolution】
mmangle (α)
5°
10°
15°
30°
45°
60°

SET

Press [MODE] to display “nor” / “Auto”,
then press [SET] to select .
『nor』…Auto Off disabled『Auto』…Will turn off after approximately
2 hours idle

● Remove dust and cutting chips after use and apply rustpreventative oil to metallic surfaces. Please keep out of reach ofchildren.
● Do not use organic solvents to clean the body of the instrument.
● When not being used for a long period of time, please remove
battery.
● Only use as directed. Improper use may cause accident or injury.

Niigata seiki Co., Ltd.
※If no operation is performed after 5 seconds, display will return
to measurement screen.

6-15-22, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-31-5670 Fax. : +81-256-39-7730

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

I298-K1604