PROFESSIONALCALIBRATIONMASTER

INSTRUCTION MANUALCALIPER CALIBRATION SETModelNo. CC-S-*** <comb.>

Thank you for purchasing the Caliper Calibration Set. This product is for calibrating outside calipers and checking accuracy.
● For safe and proper use of this product please read this instruction manual before use and follow the procedures

described. Please keep manual where it is accessible to user for future reference.

●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact dealer or Niigata Seiki at the address listed below.

■ PART IDENTIFICATIONPRODUCT SPECIFICATIONS
Set Contents

Model No.
Measurement
Surface50

CC-S-100

Measurement
Surface

CC-S-150
CC-S-200

The Measurement Surfaces can be identified
as the sides when facing the size marking.
Use these surfaces between Caliper Jaws
during measurement.

CC-S-300

JIS Grade 2 Gauge Blocks
JIS Grade 2 Gauge Blocks

JIS Grade 2 Gauge Blocks
75mm（Dimensional tolerance：±1.00µm）
JIS Grade 2 Gauge Blocks 150mm（Dimensional tolerance：±1.60µm）

JIS Grade 2 Gauge Blocks 100mm（Dimensional tolerance：±1.20µm）
JIS Grade 2 Gauge Blocks 200mm（Dimensional tolerance：±2.00µm）

JIS Grade 2 Gauge Blocks 150mm（Dimensional tolerance：±1.60µm）
JIS Grade 2 Gauge Blocks 300mm（Dimensional tolerance：±2.80µm）● Material： High Carbon High Chromium Steel

PREPERATION FOR USE

SAFETY PRECAUTIONS Please Observe
Always follow the procedures specified in order to prevent harm
to yourself or others, and to prevent damage to property.

Denotes a requirement – You MUST do

CAUTION

Items marked where there is potentialof injury or property damage.

Read the manual and follow all instructions.
・Use of product other than as described in the manual
may cause accident.
Use only for measuring.
・Use for any purpose other than measuring may cause damage orwear to the gauge. Improper use may also cause accident.
Handle with Care.
・Do not drop or subject gauge to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage.

・Do not engrave or scratch gauge.

Before using, be sure to do the following.

●When first purchased:・Confirm that package includes correct Gauge Blocks.
・Confirm that Inspection Report is included.

●Before each use:

・Check that there is no corrosion or scratches on the Gauge Blocks
・ Check for burrs.
If necessary, use burnishing stone to remove any burrs.

HOW TO USE① Clean the Caliper and Gauge Blocks.

Using cleaning solution, wipe off any oils or dirt frommeasurement surfaces.

② Make sure Gauge Blocks and Caliper are at same
temperature.
③ Measure

Use and store only in the following conditions:● Dry location protected from rain, water, and oils.
● Location that does not become hot.
● Location protected from use by children andunauthorized people.
・Use in conditions contrary to the above may cause
damage to the product resulting in accident or injury.
Use corrosion protection when not in use.
・After use, wipe surfaces with a cloth to remove any dirtor oils, and place in a rust preventive bag. Store ina cool, dry location protected from direct sunlight.

Place the measurement surfaces of the Gauge Block between
the Outside Caliper Jaws and perform the measurement.

mm

75

CARE AND STORAGE AFTER USE

MAINTANENCE① Clean Gauge before storage.

● Wear on the measurement surfaces willaffect the dimension so please verify thedimensions regularly.
● Rate of wear greatly depends of frequency
and conditions of use.
Periodic inspection frequency should bedetermined based on your companies internal
standards in order to insure accuracy.

Use a solvent moistened cloth to wipe off any
dirt or oils from the measurement surfaces.

② Storage

Place in the provided rust preventive bagand store in a cool, dry place.

50mm（Dimensional tolerance：±0.80µm）
100mm（Dimensional tolerance：±1.20µm）

Niigata Seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055+81-256-33-5522

FAX : +81-256-33-5518

MAIL: intl.sales@niigataseiki.co.jp
URL: http://www.niigataseiki.co.jp

I402­K1911