Thank you for purchasing the Niigata Seiki Digital Gap Caliper.
This is a precision measuring tool for step height measurements.
●For safe and proper use, please read this instruction manual before use and follow the procedures described. Please keep manual
where it is accessible to user for future reference.
●Keep this manual with the instrument if transferred or leased to a third party.
●For inquiries about this product, please contact dealer or Niigata Seiki.

■PART IDENTIFICATION AND FUNCTION※Model GDG-4F-J1 shown

DIGITAL GAP CALIPER⑭ [HOLD] Button⑪Battery Cover② Lock Screw⑩ LCD Display⑫ [ ZERO ] Button③Arm Lock Screw(GDG-4F-J1 only)④Clamp Screw⑬ [ ON/OFF ] Button⑤Probe Holder⑥ Jaw⑦Lock Nut

⑧ Probe※Stainless Tip SR0.5mm※1/8" Nylon Ball Tip(GDG-P-S2 only)⑨Reference Surface

SAFETY PRECAUTIONS
To prevent harm to yourself or others, and to prevent damage toproperty, always follow the procedures marked by the following
symbols.

WARNING

Denotes a requirement –You MUST do.

Indicates risk of personal injury or property damage if not followed.

Read the manual and follow the directions.
・Use of product other than as described in this manual
may cause accident.
Use only for measuring.
・Use for any purpose other than measuring may
damage or wear the instrument. Improper use may
also cause accident.
Handle with Care.
・Do not drop or subject instrument to excessive shock.
Do not place under heavy objects. Improper handling
may cause damage or poor accuracy.
・Do not scratch or mark the body of the gauge.
Avoid using in high electric fields, such as nearfluorescent lights or switching power supplies.
・Incorrect value may be displayed on LCD.
Probe Tip is Sharp-Handle With Care.
・Tip can cause injury.

Battery Cover●Use care to avoid damaging terminals when inserting
battery.
●Terminals may be broken if battery is pressed straight
down.
●Insert battery at angle, underneath the small hook, asshown.
Battery

HookSlide In

Terminals

Remove any dirt or rustproofing from the measuring
surfaces and moving parts before use. Make sure thehardware is tight and secure.
・Improper preparation may cause measurement errors.
Use and store only in environment meeting thefollowing conditions:●Dry location protected from rain, water, and oils.
●Location that is not excessively hot.
●Location protected from use by children andunauthorized people.
・Use in places contrary to the above may cause damage
to the product resulting in accident or injury.
Do not disassemble or modify.
・Disassembly may damage product or cause accident.
For repair, please contact distributor or place of purchase.
Do not write on instrument with electric engraver.
・Do not engrave instrument body or jaw arm, such as an
ID number with engraver or electric pen.

■Absolute Measurement Mode

①Turn on the Gauge, and loosen the LockScrew.
②Position the probe tip for the desired
measurement, and tighten the Clamp
Screw, and the Arm Lock Screw(Model GDG-4F-J1). (Figure 1.)③Place the gauge on a flat reference surface,and slide the tip down for good contact.
Press the [Zero] Button.
(Figure 2.)
Gauge is ready for measurement when
the display reads "0.00 mm"④Slide the gauge head up and down tomake contact with the surface you wantto measure. (Figure 3.)※ Do not change the position of theProbe Holder as set in step ②, or themeasurement will not be correct.

BatteryTROUBLESHOOTING
Q. Nothing is displayed on LCD
A. The battery may be running low, please replacebattery with a new one.
Q. With power on, the number on the LCD does notchange as Display Unit is moved.
A. Please reset by removing battery for about
30sec., and then replacing.
Q. Nothing happens when buttons are pressed.
A. Make sure battery is properly inserted.
Q. The characters “MIN”, “MAX”, or “△” aredisplayed on LCD.
A. Cycle power, or apply several long presses to[HOLD] Button. Use as normal once character isoff.
Q. The correct value is not displayed.
A. High humidity may cause dampness. Please dryoff the instrument and use only in dry location.

SERVICE

Please Observe
Denotes a prohibition –You MUST NOT do.

Notice Use only CR2032 typebattery.

Pressed

⑮ Output Connector① Main Beam

①Main Beam ・・・・・・・・・・・Supports LCD Display Unit.
②Lock Screw ・・・・・・・・・・・Tighten to Prevent the Slide and Display frommoving.
③Arm Lock Screw ・・・・・Fixes the joint between the Arm and Jaw.
(GDG-4F-J1 only)④Clamp Screw・・・・・・・・・Secure the sliding Probe Holder in place.
⑤Probe Holder ・・・・・・・・・Holds the Probe.
⑥Jaw ・・・・・・・・・・・・・・・・・・・・Supports the sliding Probe Holder.
Standard straight arm, jointed version for(GDG-4F-J1 only).
⑦Lock Nut ・・・・・・・・・・・・・・Tighten to secure the Probe in place.
⑧Probe・・・・・・・・・・・・・・・・・・Tip is applied to workpiece for measurement.
Tip available in stainless steel, or plastic(GDG-P-S2 only).
⑨Reference Surface・・・Measurement is difference between Tip position,and this surface.
4-point (flat bottom) (GDG-4F-S1・GDG-4F-J1)
3-point (round) (GDG-3R-S1)
2-point (round) (GDG-2R-S1)
Flat (GDG-F-S1)
Plastic version available (GDG-P-S2)⑩LCD Display ・・・・・・・・・・Displays digital measurement.
⑪Battery Cover・・・・・・・・・Open to access battery.
⑫[ZERO] Button・・・・・・・・・・When pressed, Display will read "0.00"⑬[ON/OFF] Button・・・・・For turning power ON / OFF.
⑭[HOLD] Button ・・・・・・・When pressed, "HOLD"Icon appears on Display
and measurement will be fixed.
⑮Output Connector・・・・Measurement value can be outputted to the PC
by connecting BluetoothBOX of the option.

●If Instrument is not working properly, or if you have anyquestions, please contact distributor or place ofpurchase.
●Please note, manufacturer is unable to respond toinquires or provide service directly. Please contact
distributor or place of purchase.

Lock Screw

SecureClamp Screw
Probe Holder

Figure 1.

②ZERO① Contact

ReferenceSurface

Flat Surface■Comparisons Measure Mode

①Turn on the Gauge, and loosen the LockScrew.
②Position the Gauge to measure theReference surface, and press the [Zero]
Button.
③Next, position the Gauge to measure thesurface for comparison, and the differencein height with the reference will bedisplayed.

DIGITAL GAP CALIPER①Slide cover in direction of arrow.
②Insert battery with (＋) sideoutward.
③Replace battery cover.

DIGITAL GAP CALIPER

GDG-＊＊-＊＊vemoRe

DIGITAL GAP CALIPER

Model No.

●[ON/OFF] Button turns power ON / OFF.
●[ZERO] Button sets the display to read "0.00" when pressed.
●[HOLD] Button will fix the display value when pressed.

DIGITAL GAP CALIPER

PRECISION MEASURING TOOLSHOW TO USE

INSERTING・REPLACING BATTERY

Figure 2.

DIGITAL GAP CALIPER

INSTRUCTION MANUAL

Contact
Figure 3.

■Measurement Precaution●To prevent error, make sure the Base is nottilted when making measurements.
(Figure 4.)

Figure 4.

AFTER USE CARE・STORAGE●After use, wipe off any dirt or oil from measuring surfaces, sliding
surfaces, and exterior using a dry cloth and perform rustprevention treatment.
For rust prevention, lightly apply anti-rust oil, or place in rustpreventive bag.
●Store with Lock Screw slightly loose
If Lock Screw is tightened, expansion due to temperature changes
may cause excessive force which will cause inaccuracy.
●Store in a cool, dry, and dark location in the provided case.
Keep out of direct sunlight and moisture, and please keep securefrom unauthorized personnel.
●Remove battery if Gauge will not be used for a long period of time.

SPECIFICATIONS●Resolution ･･････････････････････ 0.01mm●Instrument Error････････････････ ±0.03mm

Model※Not including quantization error(±1count)●Measurement Range ･･････････ ー13 to ＋16mm●Display ･･････････････････････････ 4 Digit LCD●Auto Off ･････････････････････････ approx. 20 min.
●Maximum response speed ････ Upward... 1.5m/s
Downward ... 0.75 m/s※Direction as shown in diagram at topof manual.

●Operating Temperature Range ･･･ 0°
C~40°
C●Storage Temperature Rang ･･･ ー10°
C~60°
C
●Power ･･･････････････････････････ (1x ) CR 2032 Lithium Battery●Battery Life ･････････････････････ approx. 2.5 years in normaluse●Accessory ･･････････････････････ CR 2032 Lithium Battery(for test)

GDG-4F-S1 GDG-3R-S1 GDG-2R-S1 GDG-F-S1 GDG-P-S2 GDG-4F-J1
4-Point(Flat)

3-Point(Round)2-Point(Round)

Flat

Plastic

4-Point(Flat)

1/8" Nylon
Ball Tip

Stainless Tip
SR 0.5mm

145

150

BaseStraight

Arm
Probe
Weight(g)JointedStainless Tip SR 0.5mm
150155

145

150

Niigata seiki Co., Ltd.
5-3-14, Tsukanome, Sanjo, Niigata, Japan, 955-0055
Tel. : +81-256-33-5522 Fax. : +81-256-33-5518

MAIL intl.sales@niigataseiki.co.jp
URL http://www.niigataseiki.co.jp

●正しく安全にお使いいただくため、ご使用の前に本取扱説明書を必ず読み、記載の手順に従ってご使用ください。
お読みになった後は、お使いになる方がいつでも見ることができる場所に、大切に保管してください。
●第三者に譲渡・貸与される場合も、この説明書を必ず添付してください。
●当商品に関するお問い合わせは、お買い求めの販売店もしくは当社にご連絡ください。

■各部の名称と機能⑮出力コネクタ①本尺
DIGITAL GAP CALIPER※図は GDG-4F-J1です。

⑪電池蓋

②止めねじ

⑩LCD表示部③アーム止めねじ（GDG-4F-J1のみ）④ホルダ止めねじ⑫ZEROスイッチ⑬ ON/OFFスイッチ⑤測定子ホルダ⑥ジョウ⑦ナット⑧測定子※ステン先端SR0.5mm※ナイロン先端1/8インチ球（GDG-P-S2のみ）

安全上のご注意⑨基準面

必ずお守りください

お使いになる人や他の人への危害、財産への損害を未然に防止するた
め、必ずお守りいただくことを右記のように説明しています。

してはいけない内容『禁止事項』です。

必ず実行していただく
内容『強制事項』です。

注 意 お守りいただかないと『傷害を負う、または物的損害が発生するおそれがある内容』です。
取扱説明書をよく読み、指示に従う。
・取扱説明書に記載された内容以外での使用は、事故の原因となります。
測定作業にのみ使用する。
・指定用途以外へのご使用は、製品の破損や磨耗、予測できない事故の原因となります。
大切に取り扱う。
・ぶつける、落とすなどの衝撃を与えたり、重量物を上に乗せたりしないでください。精度不良や破損の原因となります。
・本体に傷をつけないでください。

インバータ式蛍光灯などの電界が発生する場所での使用は避ける。
・ＬＣＤ表示部の数値が誤作動を起こす原因となります。
測定子先端は尖っているので、扱いに注意する。
・けがの原因となります。

使用前に測定面・摺動面・外観などの汚れや防錆油などを拭き取り、測定子ホルダのナットがゆるんでいる場合は締め直す。
・そのまま測定を行うと誤差の原因となります。

電池は必ずCR2032をお使いください。

DIGITAL GAP CALIPER●電池をセットする際は、端子の折れにご注意ください。
●電池をセットする際に上から強く押し込むと、端子が折れて破損する場合があります。
●電池をセットする際は、右斜め方向から端子を外側に押すように、本体のツメの下へ入れ込んでください。
ツメ電池

端子入れ込む①止めねじをゆるめ、電源を入れてください。
②測定したい位置に測定子ホルダを合わせます。ホ ル ダ 止 め ね じ・ア ー ム 止 め ね じ（GDG-4F-J1）を締めて固定してください。
（図１）③平らな面に基準面をあてます。測定子を押し下げ、先端を平らな面に合わせます。合わせた所で”ZERO”スイッチを押して（図２）ください。
LCD 表示 部に"0.0 0 mm"が表示されると計測開始ＯＫです。
④測定子を上下にスライドさせ、測定したい面に合わせて計測してください。
（図３）※この時、測定子ホルダは②で固定した位置で計測してください。誤差の原因となります。

電池

トラブルと対応

止めねじ

合わせて固定

ホルダ
止めねじ
測定子
ホルダ

（図１）

②ZERO

①合わせる

Ｑ．電源ONの状態でスライドさせてもLCD表示の数字が変わらない。
Ａ．一度電池を取り外し、30秒後に電池をセットし直してください。
Ｑ．スイッチを押しても何も反応しない。
Ａ．電池が正しくセットされているか確認してください。

Ｑ．LCD表示部に『MAX』『MIN』『△』などの文字が表示されている。
Ａ．一度電源を切るか、HOLDスイッチを何回か長押しして文字を消してからご使用ください。
Ｑ．正しい数値が表示されない。
Ａ．使用環境の湿度が高い可能性があります。本器を乾燥させ、湿度が低い環境下でご使用ください。

修理の際は●正常に作動しない場合や、不明な点がございましたらお買い上げの販売店または、発売元までご連絡ください。
●お問い合わせや、ご連絡が無いまま直接発売元に修理品などを送付されても処理、対応ができない場合がありますのでご了承ください。

基準面

平らな面

■比較測定モード

Ｑ．LCD表示部に何も表示されない。
Ａ．電池の残量が少なくなっている可能性があります。新しい電池と交換してください。

（図２）①止め ねじをゆるめ 、電 源を入れてください。
②基 準 と な る 被 測 定 物 を 測 定 し た 状 態 で”ZERO”を押してください。
③次に比較する被測定物を測定すると、基準との差が表示されます。

合わせる
（図３）

■測定上の注意●ベース基準面が傾いたまま測定すると正しく測定されません。
（図４）
（図４）

使用後のお手入れ・保管方法●測定面・摺動面・外観などの汚れや油などを乾いたウエスで拭き取り、防錆処理を行う
防錆処理は防錆油を薄く塗るか、防錆袋などに入れてください。
●止めねじをゆるめた状態で保管する止めねじを締めて固定すると、温度変化による膨張などで無理な力がかかり、精度不良の原因となります。

●付属のケースに入れ、乾燥した冷暗所に保管する
直射日光や湿気は避け、管理者以外が触れない状態で保管してください。
●長期間使用しない場合は電池を製品から取り外した状態で保管する製品仕様●最 小 読 取 値：0.01mm●器差：±0.03mm

品番※量子化誤差（±1カウント）を含まず下記の条件を満たす環境で使用・保管する。
●雨や水、油などがかからない、乾燥した場所●高温にならない場所●子どもや、使用者以外が近付かない場所
・上記に反する場所での使用は製品の破損、事故やけがの原因となります。

●測 定 範 囲：−13 〜 ＋16mm●表示：LCD 4桁表示
●オ ー ト オ フ：約20分●最大応答速度：上方向へのスライド…1.5m/s
下方向へのスライド…0.75m/s

電気ペンなどで書き込まない。
・電気ペンなどで本尺やジョウなどに、番号などの書き込みをしないでください。

●使用温度範囲：0℃〜40℃●保存温度範囲：−10℃〜60℃●電源：リチウム電池 CR2032×1●電 池 寿 命：通常使用状態で約2.5年●付 属 品：リチウム電池 CR2032（テスト用）

分解・改造を行わない。
・製品の破損や事故の原因となります。修理はお買い上げの販売店、または発売元までご連絡ください。

■絶対測定モード

電池蓋押し込む⑭ HOLDスイッチ①本尺 ………………… デジタル液晶部のスライドを保持します。
②止めねじ …………… デジタル液晶部のスライドを固定します。
③アーム止めねじ …… ジョウのアーム関節を固定します。
（GDG-4F-J1のみ）④ホルダ止めねじ …… 測定子ホルダのスライドを固定します。
⑤測定子ホルダ ……… 測定子を支えます。
⑥ジョウ ……………… 測定子ホルダがスライドします。
アーム形状はストレートと関節（GDG-4F-J1のみ）があります。
⑦ナット ……………… ここを締めて測定子を固定します。
⑧測定子 ……………… 被測定物にこの先端を当てて測定します。
ステン先端とナイロン先端（GDG-P-S2のみ）があります。
⑨基準面 ……………… この面を基準として測定子との差を測定します。
4点支持〈フラット〉（GDG-4F-S1・GDG-4F-J1）
3点支持〈球面〉（GDG-3R-S1）
2点支持〈球面〉（GDG-2R-S1）
フラット（GDG-F-S1）
樹脂（GDG-P-S2）があります。
⑩LCD表示部 ………… 測定数値をデジタル表示します。
⑪電池蓋 ……………… 開けて電池をセットします。
⑫ZEROスイッチ …… 押すと画面数値が"0.00"と表示されます。
⑬ON/OFFスイッチ … 電源のON/OFFを行います。
⑭HOLDスイッチ …… 押すと画面上に"HOLD"と表示され、表示している数値を固定できます。
⑮出力コネクタ ……… 別売の Bluetooth BOX を接続することで、測定値をパソコンへ出力することができます。

ご注意●”ON/OFF”スイッチを押すと電源のON/OFFができます。
●”ZERO”スイッチを押すと画面数値が "0.00"と表示されます。
●”HOLD”スイッチを押すと数値の表示を固定することができます。

す外

この度は「デジタルギャップキャリパ」をお買上げいただきありがとうございます。
この商品は、測定物の段差測定が可能な精密測定工具です。

①電池蓋を矢印の向きに開けます。
②電池を、必ず（＋）側を表にしてセットしてください。
③電池蓋を元通りに装着してください。

DIGITAL GAP CALIPER

デジタルギャップキャリパ

使用方法

電池のセット・交換

DIGITAL GAP CALIPER

精密測定工具品番 ＧDG-＊＊-＊＊DIGITAL GAP CALIPER

取扱説明書※上方向、下左方向は各部の名称の図を正面にした場合

ベース

GDG-4F-S1 GDG-3R-S1 GDG-2R-S1 GDG-F-S1
2点支持
４点支持
３点支持
フラット〈フラット〉〈球面〉

ストレ ート

アーム
測定子
質量（g）

〈球面〉

155

145

〈フラット〉

関 節

ナイロン先端 ステン先端
1/8 インチ球 SR0.5mm

ステン先端 SR 0.5mm
150

GDG-P-S2 GDG-4F-J1
４点支持
樹脂

150

145

150

発売元
〒955-0055 新潟県三条市塚野目5丁目3番14号(0256)33-5502(代) FAX(0256)33-5528

URL http://www.niigataseiki.co.jp

E341-T1 2010200