Instruction Manual

ZEISS Axiocam 208 color / 202 mono
Microscope Camera

ZEISS Axiocam 208 color / 202 mono
Original Manual
Carl Zeiss Microscopy GmbH
Carl-Zeiss-Promenade 10
07745 Jena
Germany
microscopy@zeiss.com
www.zeiss.com/microscopy

Carl Zeiss Suzhou Co., Ltd.
Modern Industrial Square 3-B, No.333 XingPu Road SIP
215126 Suzhou
China

Document Name: ZEISS Axiocam 208 color / 202 mono Instruction Manual
Order Number: 426570-7011-001
Revision: 5
Language: en-US
Effective from: 11/2021© 2021 Without the prior written consent of ZEISS, this document or any part of it must neither be translated nor
reproduced or transmitted in any form or by any means - including electronic or mechanic methods, by photocopying, recording or by any information or filing system. The right to make backup-copies for archiving purposes shallremain unaffected thereby. Any violations may be prosecuted as copyright infringements.
The use of general descriptive names, registered names, trademarks, etc. in this document does not imply thatsuch names are exempt from the relevant intellectual property laws and regulations and therefore free for generaluse. This shall also apply if this is not specifically referred to. Software programs shall entirely remain the propertyof ZEISS. No program or subsequent upgrade thereof may be disclosed to any third party, copied or reproduced inany other form without the prior written consent of ZEISS, even if these copies or reproductions are destined forinternal use at the customer's only, the only exception being one single back-up copy for archiving purposes.

Content​

ZEISSContent
1 General Information ......................................................................

5

1.1

Introduction ...............................................................................................

5

1.2

Text Conventions and Link Types................................................................

5

1.3

Explanation warning messages and additional information.........................

6

2 Safety ............................................................................................

7

2.1

Intended use ..............................................................................................

7

2.2

Prevention of hazards.................................................................................

7

2.3

Limitation of liability ...................................................................................

8

2.4

Warranty....................................................................................................

8

2.5

Warning labels ...........................................................................................

8

2.5.1
2.5.2

9
9

Position of the warning labels.....................................................
Meaning of the warning labels ...................................................

3 Technical Data and Conformity ..................................................... 10
3.1

Axiocam 208 color ..................................................................................... 10
3.1.1
3.1.2

3.2

Axiocam 202 mono.................................................................................... 12
3.2.1
3.2.2

3.3

Specifications ............................................................................. 10
Spectral Sensitivity ...................................................................... 12
Specifications ............................................................................. 12
Spectral Sensitivity ...................................................................... 14

Applicable Standards and Regulations ........................................................ 15
3.3.1

Symbols on the camera .............................................................. 15

4 Shipment ....................................................................................... 16
4.1

Axiocam 208 color ..................................................................................... 16

4.2

Axiocam 202 mono.................................................................................... 16

5 Connecting the Camera ................................................................. 17
5.1

Camera layout and accessories ................................................................... 17
5.1.1
5.1.2
5.1.3

Camera connections ................................................................... 17
Camera controls ......................................................................... 17
Accessories................................................................................. 18

5.2

Mounting the camera to the microscope.................................................... 20

5.3

Connecting the camera to the power supply .............................................. 21
5.3.1
5.3.2

5.4

Powering the camera via microscope.......................................... 21
Powering the camera via mains supply ....................................... 21

Connecting the camera to a display (without PC)........................................ 22

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

3

Content​

ZEISS

5.5

Connecting the camera to a network ......................................................... 22
5.5.1
5.5.2

Connecting camera via Ethernet ................................................. 23
Connecting camera via Wi-Fi adapter ......................................... 23

5.6

Connecting the camera to a certified PC..................................................... 24

5.7

Function indicator signals ........................................................................... 25

6 On Screen Display (OSD) Menu ...................................................... 26
6.1

Opening the OSD menu ............................................................................. 26
6.1.1
6.1.2
6.1.3

Opening menu via camera's operator panel................................ 26
Opening menu via microscope stand buttons ............................. 26
Opening menu via network browser........................................... 26

6.2

Home menu ............................................................................................... 27

6.3

Settings menu ............................................................................................ 30
6.3.1
6.3.2
6.3.3

Image Settings menu.................................................................. 31
Microscope System menu ........................................................... 32
Operating System menu ............................................................. 34

7 Installing Software and Drivers ..................................................... 39
7.1

Installing software on PC............................................................................ 39

7.2

Installing software on iPad ......................................................................... 39

7.3

Installing TWAIN plugin on PC.................................................................... 39

7.4

Installing TWACKER DEMO Application ...................................................... 39

8 Acquiring Images and Videos ........................................................ 41
8.1

Introduction ............................................................................................... 41

8.2

Basic procedure using operator panel......................................................... 41

8.3

Basic procedure using OSD menu ............................................................... 41

8.4

Image acquisition with Labscope................................................................ 41

9 Care and Maintenance................................................................... 43
9.1

Optical system............................................................................................ 43

9.2

Cleaning the infrared filter or protective glass............................................. 43

9.3

Updating the firmware ............................................................................... 43

10 Troubleshooting ............................................................................ 45
10.1 ZEN software.............................................................................................. 45
10.2 Camera ...................................................................................................... 46
10.3 Labscope.................................................................................................... 47

11 Disposal and Recycling .................................................................. 48

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

41 General Information | 1.1 Introduction

ZEISS

1 General Information
1.1 Introduction
Welcome to the Axiocam 208 color / Axiocam 202 mono user guide.
These cameras are multi-functional digital CMOS cameras for use in light microscopy applications.
To make it easier for you to set up the camera, please follow the instructions in these chapters
step by step.
Content

ChapterContent

About this guideIntroduction and overview of this manual.

SafetyImportant information on the safe handling of the camera.
Read this chapter before unpacking and operating thecamera.

Technical data

Here you will find your camera's technical data.

Shipment

The contents of delivery and optional attachments will be
described here.

Connecting the camera

In this chapter, you will find detailed instructions on connecting and using the camera.

OSD menu

This chapter lists the functions of the On Screen Display(OSD) menu.

Installing software and drivers

Here you will learn how to install the software and camera
drivers.

Acquiring images

This chapter provides the basics of image acquisition.

Troubleshooting

In this chapter, we have listed some solutions to various
problems. If you are still unable to solve your problem, contact ZEISS support.

MaintenanceThis chapter describes some measures for the maintenanceand care of your camera. In case of greater damage, always
contact ZEISS support.

Disposal and Recycling

Important instructions for disposal and recycling.

1.2 Text Conventions and Link Types
The following text conventions and link types are used:

Text convention

Meaning

Click Start.

The names of controls and important information are shown in bold letters.

Press the STANDBY button.
Press [Enter] on the keyboard.
Press <Ctrl+Alt+Del>

Press several keys on the keyboard simultaneously.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

51 General Information | 1.3 Explanation warning messages and additional information

ZEISS

Text convention

Meaning

Select Tools > Goto Control Panel > Airlock.

Follow a path in the software.

Text input

Text to be entered by the user

Programming and Macros

Anything typed in literally during programming, including, for example, macro codes,
keywords, data types, method names, variables, class names, and interface names.

Tab. 1: Text convention

Link type

Meaning

See: Text Conventions and Link Types [} 5].

Link to further information for this topic.

https://www.zeiss.com/corporate/int/
home.html

Link to a website on the internet.

Tab. 2: Link types

1.3 Explanation warning messages and additional information
The safety notes in this document follow a system of risk levels, defined in the following manner:

CAUTION
Risk of personal injury
CAUTION indicates a potentially hazardous situation which, if not avoided, may result in minor
or moderate personal injury.

NOTICE
Risk of property damage
NOTICE indicates a property damage message. In addition, NOTICE is used for data loss or corrupt data.

Info
Indicates useful additional information. It helps you to make your daily work easier, but it is optional. There is no risk for personal injury or property damage involved.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

6

2 Safety | 2.1 Intended use

ZEISS

2 Safety
2.1 Intended use
The Axiocam 208 color and the Axiocam 202 mono are high definition cameras for color andmonochromatic imaging, respectively. They are suitable for use as accessories for educational androutine microscopy in laboratory environments and for use by trained laboratory personnel. The
cameras have been designed to be used in the field of light microscopy for general observation,
routine work, and simple applications in which a sufficient amount of light is available.
These cameras should only be used for training and research. The images / videos from thesecameras must not be used for the direct generation of diagnostic results.

2.2 Prevention of hazards
Axiocam 208 color and Axiocam 202 mono have been manufactured and tested by ZEISS according to the regulations specified in CE and has left the manufacturer’s premises in perfect working
order. The cameras are intended to be used in a basic electromagnetic environment while thenoise emission conforms to CISPR 11 Class B level. The EMC performance might be impaired while
integrated into a system. In order to ensure that this condition is maintained and to avoid any
risks when operating the system, the user must comply with any notes and warnings contained inthis manual. The manufacturer shall be exempt from statutory liability for accidents should the operator fail to observe the safety regulations.

CAUTION
Personal Injury
To avoid personal injury, read and adhere to the safety notes below.
4 To avoid the risk of fire or explosion, do not use the camera near inflammable liquids orgases.
4 Setup, expansions, re-adjustments, alterations, and repairs must be carried out only by
persons who have been authorized by ZEISS.
4 Do not allow any cables, particularly power cords, to trail across the floor, where they canbe snagged by people walking past.
4 Protect the cables from excessive heat (e.g. halogen lamps, microscope fluorescence illumination).
4 To avoid injuries due to potentially high surface temperatures, do not touch the camera's
surface for a prolonged time.
4 Do not position the equipment in a way that makes it difficult to operate or disconnect thedevice.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

7

2 Safety | 2.3 Limitation of liability

ZEISS

NOTICETo avoid equipment damage, data loss, or corrupted data, read and adhere to the safety notes
below.
4 Protect the camera against mechanical impact. External damage may affect the operationof inner components.
4 Keep chemicals and fluids away from the camera.
4 Make sure there is sufficient ventilation of the camera head. Avoid direct exposure to sunlight and locations near heat sources (radiators, stoves). Overheating can cause noisy images.
4 Use the camera in a clean and dry location.
4 Attach all connectors firmly and securely.
4 Use only the accessories supplied by ZEISS, when applicable.
4 Use only normal microscope cleaning material to clean the camera housing.
4 Contact your local ZEISS service organization if a repair is necessary. Do not disassemble
the camera by yourself, otherwise the warranty will be lost.
4 Save all your data, such as images, measurement data, archives, reports, forms and documents, at regular intervals on an external storage medium. Otherwise it cannot be avoided
that access to this data may be lost as a result of operational errors or hardware defects.
ZEISS accepts no liability for consequential damage resulting from insufficient data protection.

2.3 Limitation of liability
No warranty shall be assumed by ZEISS during the warranty period if the equipment is operated
without observing the safety regulations. In any such case, ZEISS shall be exempt from statutory liability for accidents resulting from such operation.

2.4 Warranty
ZEISS shall be exempt from any warranty obligations should the user fail to observe the safety regulations. ZEISS only guarantees the safety, reliability, and performance of the system if the safety
notes are closely observed.

2.5 Warning labels
All points that may pose special risks are additionally marked by warning labels (pictograms) on
the camera. These warning labels indicate possible dangers. They are part of this instruction manual. They are to be kept in a clean and legible state. Warning labels that are damaged or nolonger clearly legible must be replaced immediately. Always observe all warning labels on thecamera.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

8

2 Safety | 2.5 Warning labels

ZEISS

2.5.1 Position of the warning labels

1Fig. 1: Warning labels on the camera

2.5.2 Meaning of the warning labels
The meaning of each warning label is explained below.

No.

Symbol

1

Description
Follow the notes in the instruction manualand the supplied documents. For more information, see Prevention of hazards [} 7].

Tab. 3: List of attached Warning Labels

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

9

3 Technical Data and Conformity | 3.1 Axiocam 208 color

ZEISS

3 Technical Data and Conformity
3.1 Axiocam 208 color
3.1.1 Specifications
Features

ValuesSensor type

CMOS sensor with rolling shutter

Sensor Size

Diagonal 8.1 mm (1/2.1")

Effective Sensor Pixel Count

8.3 Megapixels: 3840 (H) x 2160 (V)Pixel size

1.85 μm

Spectral Sensitivity

Approx. 400 nm – 700 nm, IR filter RGB
Bayer color mask

Selectable Resolution

3840 x 2160 (Ultra HD, 4K)
1920 x 1080 (Full HD, 1080p)

Gain (Signal Amplification)

1x – 22x adjustable

Digitization

3 x 8 bit / pixel

Exposure Time Range (Integration time)

0.06 ms - 1 s

Image enhancement functions

Active denoising, active sharpening, auto
white balance, HDR

Automatic features

Automatic exposure and gain regulation at
Ultra HD resolution (4K), fast live image under low light conditions

Status-LED for camera

Color coded operation status

Interfaces

HDMI for monitor
USB 3.0 Type-C for USB hub, flash drive, WiFi adapter or PC connection
Ethernet (RJ45) for LAN connection
Micro-D for power and communication withdedicated stands

Wi-Fi compatibility

Via USB Wi-Fi adapter and router

Optical Interface

C-mount

Camera control buttons for stand-alone

1x Image capture
1x OSD (On Screen Display menu)
1x Camera factory reset

Stand-alone operation:§ Image storage format§ tiff or jpg§ Video stream format§ mp4§ Live frame rate via HDMI

§ 30 fps @ Ultra HD (4K)

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

10

3 Technical Data and Conformity | 3.1 Axiocam 208 color

ZEISS

Features

ValuesMaximum live frame rate at configuration:@ 4K (3840 x 2160)§ HDMI§ 30fps§ Ethernet§ -§ USB 3.0

§ 15 fps

Maximum live frame rate at configuration:@ 1080p (1920 x 1080)§ HDMI§ 30 fps§ Ethernet§ 30 fps§ USB 3.0

§ 27 fps

Max. file size per image.tiff up to: 24 MB (not content dependent).jpg up to: approx. 0.5 MB – 3.3 MB (content
dependent)

Size/ Weight

Approx. 135 x 97 x 54mm / 590 g

Housing

Blue painted aluminum and cooling fins on
the top plate

Registration

CE, RoHS, CSA

Power supply

via Micro-D interface

Power consumption

Max. 9 W (24 V DC, 0.375 A)

Environmental conditions for storage and operation+10°C to +40°C, max. 75% relative air humidity at 35°C, no condensation, free air circulation required, CAT II, pollution degree 2,
altitude <2000m, indoor use

Environmental conditions for transport inpackaging-40 °C to +70 °C, max. 75 % relative air humidity at 35 °C

IP code

IP20

Operating systems:§ for ZEN

§ Windows 10 x64 Prof./Ultimate and higher§ for Labscope§ Windows 10 x64 Prof./Ultimate and iOS
v12 and higher

Supported Application Software

ZEN blue v3.0 and higher (includes ZEN lite/
pro/system)
ZEN core v2.7 and higher (includes ZEN
starter/core)
Labscope v2.9 (win) v2.8.3 (iOS) and higher

TWAIN plugin

Software interface to control camera by 3rdparty application software

Order number

426570-9001-000

Info

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

11

3 Technical Data and Conformity | 3.2 Axiocam 202 mono

ZEISSComputer hardware, operating system, and software may decrease the frame rates. All specifications are subject to change without notice.

Relative Spectral Sensitivity

3.1.2 Spectral Sensitivity

Wavelenght (nm)Fig. 2: Spectral Sensitivity of Axiocam 208 color (incl. IR Filter)

3.2 Axiocam 202 mono
3.2.1 Specifications
Features

ValuesSensor TypeCMOS sensor with global shutter

Sensor Size

Diagonal 13 mm (1/1.23")

Effective Sensor Pixel Count

2.1 Megapixels: 1920 (H) x 1080 (V)Pixel size

5.86 μm

Spectral Sensitivity

Approx. 350 nm – 850 nm, protection glass(coated)

Selectable Resolution

1920 x 1080 (Full HD, 1080p)

Gain (Signal Amplification)

1x – 16x adjustable

Digitization

12 or 8 bit / pixel

Cooling

Passive cooling

Exposure Time Range (Integration time)

0.3 ms - 2 s

Image enhancement functions

Active denoising, active sharpening

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

12

3 Technical Data and Conformity | 3.2 Axiocam 202 mono

ZEISS

Features

ValuesAutomatic features

Automatic exposure and gain regulation at
Full HD resolution (1080p), fast live image
under low light conditions

Status-LED for camera

color coded operation status

Interfaces

HDMI for monitor
USB 3.0 Type-C for USB hub, flash drive, WiFi adapter or PC connection
Ethernet (RJ45) for LAN connection
Micro-D for power and communication withdedicated stands

Wi-Fi compatibility

Via USB Wi-Fi adapter and router

Optical Interface

C-mount

Camera control buttons for stand-alone

1x Image capture
1x OSD (On Screen Display menu)
1x Camera factory reset

Stand-alone operation:§ Image storage format§ tiff or jpg§ Video stream format§ mp4§ Live frame rate via HDMI

§ 30 fps @ Full HD (1080p)

Maximum live frame rate at configuration:@ 1080p (1920 x 1080)§ HDMI§ 30 fps§ Ethernet§ 30 fps§ USB 3.0

§ 27 fps

Max. file size per image

12bit .tiff: 4 MB (one channel, not content
dependent)
8bit .tiff: 2 MB (one channel, not content dependent)
8 bit .jpg: approx. 0.1 MB – 1.3 MB (content
dependent)

Size/Weight

Approx. 135 x 97 x 54mm / 590 g

Housing

Blue painted aluminum and cooling fins on
the top plate

Registration

CE, RoHS, CSA

Power supply

via Micro-D interface

Power consumption

Max. 9 W (24 V DC, 0.375 A)

Environmental conditions for storage and operation+10°C to +40°C, max. 75% relative air humidity at 35°C, no condensation, free air circulation required, CAT II, pollution degree 2,
altitude <2000m, indoor use

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

13

3 Technical Data and Conformity | 3.2 Axiocam 202 mono

ZEISS

Features

ValuesEnvironmental conditions for transport inpackaging-40 °C to +70 °C, max. 75 % relative air humidity at 35 °C

IP code

IP20

Operating systems:§ for ZEN

§ Windows 10 x64 Prof./Ultimate and higher§ for Labscope§ Windows 10 x64 Prof./Ultimate and iOS
v12 and higher

Supported Application Software

ZEN blue v3.0 and higher (includes ZEN lite/
pro/system)
ZEN core v2.7 and higher (includes ZEN
starter/core)
Labscope v2.9 (win) v2.8.3 (iOS) and higher

TWAIN plugin

Software interface to control camera by 3rdparty application software

Order number

426570-9011-000

Info
Computer hardware, operating system, and software may decrease the frame rates. All specifications are subject to change without notice.

Relative Spectral Sensitivity

3.2.2 Spectral Sensitivity

Wavelength (nm)Fig. 3: Relative Spectral Sensitivity of Axiocam 202 mono

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

14

3 Technical Data and Conformity | 3.3 Applicable Standards and Regulations

ZEISS

3.3 Applicable Standards and Regulations
Observe the generally applicable local and national safety and accident prevention regulations, aswell as the applicable laws and regulations in your country. The camera and its accessories havebeen designed, manufactured and tested to comply with the guidelines and specifications as
named in the following. The accordance to the relevant specification is indicated by a respective
symbol on the unit.

3.3.1 Symbols on the camera
SymbolDescription
Conforms to:§ EU Directive 2014/35/EU (LVD)§ 2014/30/EU (EMC)§ 2015/863/EU (RoHS)
Complies with EU Directive 2012/19/EU(WEEE)

CSA certificate mark, conforms to:§ CAN/CSA-C22.2 No. 61010-1-12§ UL Std. No. 61010-1 (3rd edition)
Tab. 4: List of attached labels concerning standards and regulations

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

15

4 Shipment | 4.1 Axiocam 208 color

ZEISS

4 Shipment
4.1 Axiocam 208 color§

1x Axiocam 208 color§

1x Power & signal cable, camera to microscope stand (compatible with Aixolab 5 and Axioscope 5/7/Vario)§

1x Power adapter with country-specific plug§

1x USB 3.0 cable, Type-C to Type-A§

1x USB 3.0 flash drive, Type-C and Type-A§

1x USB hub, Type-C

Accessories for Stand-Alone and for Usage with Labscope

Order Number

Accessory

000000-0626-248

High-Speed-HDMI-Cable, Premium, Resolution 4K, 2m

000000-0626-246

Optical USB-Scroll Mouse

000000-0626-245

Keyboard, USB, Language DE

000000-0626-267

Keyboard, USB, Language US

426570-9110-000

Wi-Fi dongle package containing Wi-Fi Dongle and USB adaptor Type-C to Type-A

4.2 Axiocam 202 mono§

1x Axiocam 202 mono§

1x Power & signal cable, camera to microscope stand (compatible with Aixolab 5 and Axioscope 5/7/Vario)§

1x Power adapter with country-specific plug§

1x USB 3.0 cable, Type-C to Type-A§

1x USB 3.0 flash drive, Type-C and Type-A§

1x USB hub, Type-C

Accessories for Stand-Alone and for Usage with Labscope

Order Number

Accessory

000000-0626-248

High-Speed-HDMI-Cable, Premium, Resolution 4K, 2m

000000-0626-246

Optical USB-Scroll Mouse

000000-0626-245

Keyboard, USB, Language DE

000000-0626-267

Keyboard, USB, Language US

426570-9110-000

Wi-Fi dongle package containing Wi-Fi Dongle and USB adaptor Type-C to Type-A

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

16

5 Connecting the Camera | 5.1 Camera layout and accessories

ZEISS

5 Connecting the Camera
5.1 Camera layout and accessories
5.1.1 Camera connections

1

2

3

4

Fig. 4: Camera connector panel

No.

Name

Function1

Micro-D port

Power supply and communication

2

USB 3.0 Type-C

Camera control and image data transfer

3

Ethernet port (RJ45)

Communication and image data transfer

4

HDMI port

Image data transfer to a certified monitor, TV or projector

5.1.2 Camera controls

1

2

3

4

Fig. 5: Camera operator panel

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

17

5 Connecting the Camera | 5.1 Camera layout and accessories

No.

ZEISS

Name

Function1

LED function indicator

Please see section Function indicator signals [} 25]
for detail.

2

Camera factory reset
buttonPress to reset all parameters to default.

3

Image / video capture
buttonPress shortly for capturing an image.

OSD menu button

Press to open the OSD menu. Press again to close the
OSD menu.

4

Press for five seconds to start recording a video. Pressagain shortly to stop the video recording.

See also
2 Acquiring Images and Videos [} 41]

5.1.3 Accessories
The following items are required for power supply and use of the ports:

Name

FigureRemarks

Power & signal cable, camera to microscope standConnection between camera andcompatible microscopes (e.g.: Axioscope 5/7 or Axiolab 5, power is supplied via microscope)Power adapter with country-specific plug

Power supply from external powersocket if the camera is used with microscopes other than Axioscope 5/7
or Axiolab 5.

USB 3.0 cable, Type-C to
Type-A

Connection between camera and PC

USB flash drive, Type C andType A

Connection to camera (Type-C) or to
USB hub (Type-A) for immediate image and video storage

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

18

5 Connecting the Camera | 5.1 Camera layout and accessories

Name

FigureZEISSRemarks

USB hub, Type-C

Connection between camera andmultiple USB Type-A devices, such as
keyboard, mouse and flash drive

Ethernet cable(not supplied in package)Connection between camera and network or WLAN router

HDMI cable(not supplied in package,please order seperately:

Connection between camera andmonitor, TV, or projector

Order No.
000000-0626-248)
Mouse(not supplied in package,please order seperately:

For control and navigation in the OSD
menu

Order No.
000000-0626-246)
Keyboard(not supplied in package,please order seperately:

For typing in the OSD menu

Order No.
000000-0626-267 for US
layout,Order No.
000000-0626-245 for DE
layout)
Wi-Fi Adapter package(not supplied in package,please order separately:
Order No.
426570-9110-000)
Monitor TFT 32" 4K(not supplied in package,please order seperately:

Wi-Fi Adapter package consisting of
Wi-Fi adapter and USB adapter TypeC to Type-A for wireless transmissionof camera images to PC or iPad with
Labscope
For display of camera image and operating the OSD menu

Order No.
410350-3201-000)

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

19

5 Connecting the Camera | 5.2 Mounting the camera to the microscope

ZEISS

5.2 Mounting the camera to the microscope
To mount the camera to your microscope’s camera port, use a C-mount camera adapter. The
adapter is not included in the scope of delivery. You will find some suitable examples for adaptersin the list below:

CameraPort

Adapter

Order number

Axiocam 208
color

60N

Camera Adapter 60N-C 2/3" 0.5x426112-0000-000

Axiocam 202
mono

60N

Camera Adapter 60N-C 2/3" 0.63x

426113-0000-000

NOTICE
Loss of warranty
The Axiocam 208 color is delivered with an integrated IR filter (infrared cut filter). The Axiocam
202 mono is delivered with a protective glass to shield the camera against dust and to reduce
optical interferences.
4 Do not remove the filter or the protective glass. Otherwise the warranty will be lost.

1

2
3

4

Fig. 6: Mounting the camera to the microscope

1

Camera2

C-mount camera adapter

3

Ring nut

4

60N port at microscope stand

1.
2.
3.
4.

Remove the dust cap from the camera's C-mount port.
Mount the C-mount camera adapter to the camera.
Attach the camera with the adapter to the microscope's 60N port.
Orient the camera to the stand and fix its position by tightening the ring nut.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

20

5 Connecting the Camera | 5.3 Connecting the camera to the power supply

ZEISS

5.3 Connecting the camera to the power supply
When connected to Axioscope 5/7 or Axiolab 5, the Axiocam 208 color or the Axiocam 202
mono are powered by the microscope. Otherwise the cameras have to be connected to the supply mains via the plug-in power supply.

5.3.1 Powering the camera via microscope

NOTICE
Risk of damage to the microscope or the camera
4 Always switch off the microscope before you disconnect the power cable from the camera!
Prerequisite ü The camera is mounted to Axioscope 5/7 or Axiolab 5.
1. Insert the Micro-D plug of the power cable into
the Micro-D port of the camera.

2. Insert the power cable's opposite connector
into the corresponding socket on your microscope.

Info
Refer to the instruction manual of your microscope for further information.

2

5.3.2 Powering the camera via mains supply
Prerequisite ü The power supply has been adapted with the appropriate country-specific adapter.
ü The camera is not already powered via a compatible microscope.
1. Insert the Micro-D plug of the plug-in power
supply into the Micro-D port of the camera.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

21

5 Connecting the Camera | 5.4 Connecting the camera to a display (without PC)

ZEISS

2. Insert the power plug of the plug-in power supply into a power outlet.

3. Turn on the camera via cable switch.

5.4 Connecting the camera to a display (without PC)
The camera can be connected to a certified monitor, TV, or projector for visualization of the live
image data and for operating the OSD menu functions. Certain HDMI functions (e.g. audio, commands from monitor to camera) are not supported.

Info
If you connect the Axiocam 208 color / 202 mono to a monitor, these are the minimal monitor
requirements:§

HDMI 1.4 or HDMI 2.0 input port§

1920 x 1080 or higher resolution§

16:9 or 16:10 aspect ratio§

Progressive scanning§

HDMI cable with less than 3 m length (shorter cable enables better signal integrity)

Note that the maximum resolution of the Axiocam 202 mono is Full HD (1920 x 1080), while
the Axiocam 208 color supports resolutions up to Ultra HD (4K).
1. Insert the HDMI cable into the HDMI port ofthe camera.

2. Insert the HDMI cable's opposite connector into the corresponding socket on your display
device.
3. Set the display device’s aspect ratio to 16:9 or Aspect.
For further camera settings using the OSD a mouse is required (e.g. Optical USB-Scroll Mouse, Order No. 000000-0626-246) . An additional keyboard is optional (Keyboard, USB, Design "Business
Slim", Language English US, Order No. 000000-0626-267 or Keyboard, USB, Design "Business
Slim", Language German, Order No. 000000-0626-245).

5.5 Connecting the camera to a network
If you want to connect the camera to a network, you can choose between several options, ofwhich all require network access and the ZEISS imaging software Labscope (available as windows
or iOS version). The camera identifies itself automatically to the network (DHCP) and is automatically recognized by Labscope, provided the device is on the same network.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

22

5 Connecting the Camera | 5.5 Connecting the camera to a networkZEISSNOTICE
Display errors
In the event of an overloaded or slow WLAN, the live image of the camera may be delayed orincorrectly displayed on the iPad.
4 If possible, use a high-performance 802.11n WLAN.
4 Provide a sufficient contingent of free bandwith for communication.
For an overview of all ZEISS Microscopy apps and further information on individual apps, visit
https://www.zeiss.com/microscopy/int/products/microscope-software/microscopy-apps.html?
vaURL=www.zeiss.com/micro-apps

5.5.1 Connecting camera via Ethernet
Prerequisite ü The camera is powered via mains supply or microscope.
1. Insert the Ethernet cable into the camera’s Ethernet port.

2. Insert the Ethernet cable's opposite connector into the corresponding socket on your
WLAN router.

5.5.2 Connecting camera via Wi-Fi adapter
Prerequisite ü The camera is powered via mains supply or microscope.
ü An USB Wi-Fi adapter is available.
ü An USB hub is connected to the camera.
ü An USB mouse is connected to the USB hub to operate the OSD.
ü The camera is connected to a display via HDMI.
1. Insert the USB Wi-Fi adapter into the camera's
USB Type-C port or into the USB hub.

2. Open On Screen Display menu (see chapter Opening the OSD menu [} 26]).
3. Select Settings icon in Home Menu (Home menu [} 27]) to open the Settings menu.
4. Select Operating System Settings (Settings menu [} 30]) to open the Operating Systemmenu.
5. Select Wi-Fi Settings.
The Choose WLAN mode menu shows up. It offers two ways to connect the camera to a Wi-Fi
device (e.g. iPad or Laptop).
5.5.2.1 Using the camera as access point
Follow the subsequent instructions to directly connect your camera to the Wi-Fi device:

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

23

5 Connecting the Camera | 5.6 Connecting the camera to a certified PC

ZEISS

1. Select As an access point from the Choose
WLAN mode menu.

2. Click the Next button.
à The subsequent screen appears.
3. Type in the network name (or keep default name) and password ZEISS1846 into the respective input fields.
4. Click the Save button.
à The camera can be found by other devices as access point: Other devices can connect tothe camera using the specified network name and password.
5.5.2.2 Connecting to existing WLAN
Follow the subsequent instructions to connect your camera to an existing WLAN:
1. Select Connect to existing WLAN from theChoose WLAN mode menu.

2. Click the Next button.
à The subsequent screen appears.
3. Select the network name from the respective selection field.
4. Type in the password into the respective input field.
5. Click the Save button.
à The camera is connected to the WLAN.
à If the Wi-Fi device is connected to the same router, the camera appears in Labscope.

Info
If the WLAN list is empty or does not contain the one you want to connect to, wait for a few
seconds and click again to refresh.

Info
Not all special characters are supported when typing in a password.
allowed: A~Z a~z 0~9 @ # % * .
not allowed: ！ 。 ， ； ？ / 、 \ & （） ‘’ ` - …… ￥5.6 Connecting the camera to a certified PC
The USB port can also be used for data transfer to the PC.
Prerequisite ü The camera is powered via mains supply or microscope.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

24

5 Connecting the Camera | 5.7 Function indicator signals

ZEISS

1. Insert the USB 3.0 cable's Type-C connector
into the corresponding socket on the camera.

2. Insert the USB 3.0 cable's Type-A connector into the corresponding socket on the PC.

5.7 Function indicator signals
For the definitions of the LED color signals, refer to the list below:

SignalDescription

Pink

Camera starting

Blue

Power supplied and camera readyBlue flash

Snapping/recording in process, and saving data to USB flash drive

Red flashFirmware updating/factory resetting

Pink flash

No USB flash drive or the drive is fullOff

No power supplied

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

25

6 On Screen Display (OSD) Menu | 6.1 Opening the OSD menu

ZEISS

6 On Screen Display (OSD) Menu
Info
Certain functions of the OSD menu are only available with compatible microscope stands i.e.
Axioscope 5/7 or Axiolab 5. For more information, please refer to the relevant microscope’s
manual.

6.1 Opening the OSD menu
When your camera is powered on and connected to a display device via HDMI, you can open or
close the OSD menu by pressing the Menu button on the camera.
To operate the OSD, plug the USB hub (included in the package) into the camera’s USB port. Then
you can connect your mouse/keyboard (not included in the package) and the USB flash drive (included in the package) via the hub.

Info
A USB flash drive used should be of FAT32 format and have enough free space for storing thedata.

6.1.1 Opening menu via camera's operator panel
Prerequisite ü The camera is connected to your display device via HDMI cable.
1. To open the OSD menu, press the Menu button on the camera's operator panel.

The OSD menu opens on your display device.

6.1.2 Opening menu via microscope stand buttons
Prerequisite ü The camera is connected to your display device via HDMI cable.
ü The camera is mounted on a Axioscope 7 stand.
1. To open the OSD menu, press the microscope stand's Snap button and Stage control
button simultaneously.
The OSD menu opens on your display device.

Info
For more information, please refer to the Axioscope 5/7/Vario operating manual.

6.1.3 Opening menu via network browser
Prerequisite ü The camera is connected to the same network as your display device (i.e. iPad or PC).

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

26

6 On Screen Display (OSD) Menu | 6.2 Home menu

ZEISS

1. On the display device, open the network browser.
2. Insert the camera's IP address into the browser window's address line and press Enter.
The OSD menu opens on your display device.

6.2 Home menu
The Home menu gives you basic imaging controls to capture your images with minimum amountof effort.
The controls for selected options are highlighted blue.

1

8

2

9

3
4

10

5
6

11

7

12
13

14

Fig. 7: Home menu

No.

Name

Description

1

Home icon

Opens the Home menu.

2

Snap button

Snaps a single image.

3

Record button

Starts a video recording. A time counter is displayed.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

27

6 On Screen Display (OSD) Menu | 6.2 Home menu

No.

Name

ZEISS

Description
When pushing the button the following controls expand:§ The recording can be stopped by clicking the redsquare symbol above the time display.

4

Auto Exposure button

Ensures consistent brightness of the image by continuously calculating the correct exposure time based on
the current light intensity.

5

Target Intensity controls

In Auto Exposure mode, the light intensity can be finetuned using the slider or the input field.

6

Auto White Balance
button (for Axiocam 208
color)

Ensures a consistent color temperature of the image bycontinuously calculating the white balance.

7

White button (for Axiocam 208 color)If active, in the Auto White Balance mode the color
temperature is calculated based on the assumptionthat the light source is white.

8

Settings icon

Opens the Settings menu.

9Multi-Snap button (for
Axioscope 5/7 and Axiolab 5)

Performs a multi-channel acquisition.

Manual Exposure button

Enables you to manually re-adjust the exposure time,e.g. when the Auto Exposure mode doesn't deliver appropriate results.

10

Please note:
A proper functioning of the Auto White Balance mode
requires enough empty area in the field of view.

Please note:
The Multi-Snap function requires that the camera isconnected to a compatible microscope stand (correct
light source and appropriate multi-band filter set). See
the corresponding microscope user manual for details.

When pushing the button, the following controls expand:Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

28

6 On Screen Display (OSD) Menu | 6.2 Home menu

No.

Name

ZEISS

Description§ The exposure time can be fine-tuned using the upper slider or input field. For exposure time range, refer to Axiocam 208 color [} 10] and Axiocam 202
mono [} 12].
§ The gain value can be set using the lower slider or
input field.

11

Manual White Balance
button (for Axiocam 208
color)

Enables you to manually re-adjust the color temperature, e.g. when the Auto White Balance mode doesn'tdeliver appropriate results.
When pushing the button, the following controls expand:§ The color temperature can be set warmer (more red)or colder (more blue) using the slider.
§ The white balance can be calculated once by clicking
the Push button.

12

Eyepieces button (forAxiocam 208 color)If active, in the Auto White Balance mode the color
temperature of the image is matched to the lightsource's color temperature. The colors in the camera
image, thus, resemble the colors seen through the eyepieces.

13

Multi-Snap Setup expander (for Axioscope 5/7
and Axiolab 5)

Enables you to adjust the settings for the Multi-Snapfunction.
When pushing the button, the following controls expand:§ The fluorescence channels to be captured can be selected and unselected by clicking the corresponding
buttons (UV, B, G, and R).
§ The expander can be closed by clicking the Hidebutton.

14

Thumbnail area

Shows a thumbnail of the last image taken, the first
frame of the last video captured, or the merged imageof the last multi-channel acquisition with overlaid
pseudo-colors.
§ The thumbnail can be closed by clicking the Close
symbol.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

29

6 On Screen Display (OSD) Menu | 6.3 Settings menu

No.

Name

ZEISS

Description§ The thumbnail can be maximized to the full display
by clicking the Expand symbol.

Tab. 5: Home menu, control elements

6.3 Settings menu
The Settings menu gives you more options for image settings, the opportunity to configure yourmicroscope, as well as to define some basic operating system settings.

1

5

2

3

4
Fig. 8: Settings menu

No.

Name

Description

1

Image Settings

Opens the Image Settings menu.

2

Operating System Settings

Opens the Operating System menu.

3

System lnformation
buttonShows the status of the entire camera/microscope system.

4

Back button

Opens the previous menu.

5

Microscope System Settings

Opens the Microscope System menu.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

30

6 On Screen Display (OSD) Menu | 6.3 Settings menu

ZEISS

6.3.1 Image Settings menu
The Image Settings menu gives you more advanced controls to optimize your image. It also
gives you some options to show different image information and how snapped images areshown.
The controls for selected options are highlighted blue.

1

2

8

3

9

4

5
6

7

Fig. 9: Image Settings menu

No.

Name

Description

Gamma controls

Enable you to adjust the gamma value.
§ The gamma value can be fine-tuned using the slider
or the input field.
§ The gamma value can be reset to default by clicking
the Reset button.
Please note:
For Axiocam 208 color the default gamma value is
0.45.
For Axiocam 202 mono the default gamma value is
1.2.

Optimize Pixel Correction button

Corrects dead and hot pixels in the image.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

31

6 On Screen Display (OSD) Menu | 6.3 Settings menu

No.

ZEISS

Name

Description

Optimize Denoise button

Reduces noise.

Image Orientation controls

Sets the image orientation.
§ 0°: original image orientation§ Mirror vertical: image mirrored in vertical direction§ Mirror horizontal: image mirrored in horizontal direction§ 180°: image rotated by 180°Show Scale Bar button

Shows a scale bar on the live display.
Please note:
The scale bar will also appear in the captured image.

Show Image after SnapbuttonIf activated, the image is shown on the full display for 5
seconds after being captured. The full display image
can be closed or minimized to a thumbnail, if required.

Smart Snap Mode button (for Axioscope 5/7
and Axiolab 5)

Activates the triggering of a multi-channel acquisition
by pushing the snap button on the camera or microscope stand.
Please note:
The Multi-Snap function requires that the camera isconnected to a compatible microscope stand (correct
light source and appropriate multi-band filter set). See
the corresponding microscope user manual for details.

Optimize HDR button(for Axiocam 208 color)

Converts several images with different exposure times
into a high contrast image or High Dynamic Range(HDR) image in order to show large differences inbrightness (e.g. reflective samples) in detail.

Optimize Sharpen button

Sharpens details.

6.3.2 Microscope System menu
The Microscope System Settings menu provides access to some microscope configurations tobe adjusted for easier imaging. For example, the microscope information can be updated if a newpart is installed. If required, a configuration wizard guides through the entire configuration setup.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

32

6 On Screen Display (OSD) Menu | 6.3 Settings menu

ZEISS

1
2
3
4
5

Fig. 10: Microscope System Settings menu

No.

Name

Description

1

Reflector / FL Modules
button (for Axiolab 5, Axioscope 5/7 )Opens a drop-down list to define the item installed at
the reflector turret's current position.

2

Objectives button (forAxiolab 5, Axioscope 5/7)

Opens a drop-down list to define the item installed at
the objective turret's current position.

3

Shading Correction button

Opens a menu to define the shading correction foreach combination of reflector, objective and lightsource, see Applying Shading Correction [} 33].

4

Configuration Wizard
button (for Axiolab 5, Axioscope 5/7)

Opens a wizard to support you in setting up all microscope configurations listed above.

5

Camera Adapter button(for Axiolab 5, Axioscope
5/7)

Opens a menu to define the camera adapter being currently used.
Please note:
For recommended adapters, see Mounting the camera
to the microscope [} 20].

6.3.2.1 Applying Shading Correction
Follow the subsequent instructions to define the shading correction for a combination of reflector, objective and light source:

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

33

6 On Screen Display (OSD) Menu | 6.3 Settings menu

ZEISS

1

2

Fig. 11: Shading correction menu
1. Move in the reflector module and the objective to be configured.
2. Turn on the light source to be configured.
à The current position of the reflector turret is indicated by 1 .
à The current position of the objective turret and the light source (RL: reflected light, TL:
transmitted light) are indicated by 2 .
3. Click the highlighted button to enter the shading correction menu for the current combination.
4. Follow the instructions on the display to perform the shading correction configurations.

Info
Shading correction may take a few seconds to take effect.

6.3.3 Operating System menu
The Operating System menu gives you options such as setting the language, defining the filename format or updating the system firmware.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

34

6 On Screen Display (OSD) Menu | 6.3 Settings menu

ZEISS

1
2
3
4
5
6
7

Fig. 12: Operating System menu

No.

Name

Description

1

Language button

Opens a menu to set English, Chinese or German as
operating system language.

2

Date & Time button

Opens a menu to set date and time.

3

File Options button

Opens a menu to set the filename format and the file
type for captured images, see File Options menu[} 35].

4

HDMI Resolution button(for Axiocam 208 color)

Opens a menu to set the desired HDMI resolution forlive view.
Please note:§ Switching from 1080p to 4K must be confirmed bypressing the Confirm button.
§ The camera restarts automatically for the new resolution to take effect.
§ If the connected monitor does not support 4K it automatically switches back to 1080p after 20 seconds.

5

Wi-Fi Settings button(for USB Wi-Fi adapter)

Opens a workflow to set up a wireless connection, seeConnecting the camera to a network [} 22].

6Bright Pixel Correction
buttonOpens a workflow to perform Bright Pixel Correction,
see Bright Pixel Correction menu [} 36].

7

Firmware Update button

Starts a firmware update when the update file is available via the USB interface, see Updating the firmware[} 43].

6.3.3.1 File Options menu
The File Options menu allows you to predefine a template for file naming.
The controls for selected options are highlighted blue.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

35

6 On Screen Display (OSD) Menu | 6.3 Settings menu

ZEISS

1

2

3

4

Fig. 13: File Options menu

No.

Name

Description

1

Filename format example

Shows the template of the filenames based on the currently selected components.

2

Filename format components

Allows you to adjust the filename template.
§ Static text components (e.g. for sample names etc.)
can be added by typing text into the Custom fields.
Empty fields will be removed from the template.
§ The date and time format can be set.
§ A counter number is appended to each filename bydefault.

3

Set Filename after eachSnap button

Activates a prompt after each image capturing to manually set the filename.

4

File type selection

Allows you to set the file type (JPEG or TIFF) for captured images. For Axiocam 202 mono, you can additionally choose to save a TIFF image as 8-bit or 12-bit.

6.3.3.2 Bright Pixel Correction menu
The Bright Pixel Correction menu allows you to apply the Bright Pixel Correction procedure.
This procedure corrects newly developed bright (or hot) pixels due to long exposure times, highgain settings or cosmic events.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

36

6 On Screen Display (OSD) Menu | 6.3 Settings menu

ZEISS

1

2

Fig. 14: Bright Pixel Correction menu

No.

Name

Description

1

Correct button

Initiates Bright Pixel Correction.

2

Reset button

Restores the factory default Bright Pixel Correction configuration.

Applying Bright Pixel Correction
Prerequisite ü The C-mount port is closed so that no light will reach the image sensor during the procedure.
You can close the C-mount port by either closing the light path of the microscope stand or
detaching the camera from the stand and protect sensor from light by screwing the protective
cap onto the camera's c-mount thread.
1. Tap the Correct button.
à Do not operate the camera during the procedure.
à If light reaches the sensor, the procedure fails.
Either cover the camera properly or reset the Bright Pixel configuration before restarting
the procedure.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

37

6 On Screen Display (OSD) Menu | 6.3 Settings menu

ZEISSThe Bright Pixel Correction procedure is applied.
The camera reboots after a successful procedure.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

38

7 Installing Software and Drivers | 7.1 Installing software on PC

ZEISS

7 Installing Software and Drivers
7.1 Installing software on PC
To acquire images with the camera(s) on a PC, you must install ZEISS software (e.g. ZEN or Labscope ). You will find the installation files on the USB flash drive delivered. The camera drivers arealso installed during the software installation. The latest version of ZEISS software products can be
downloaded from our web page: https://portal.zeiss.com/download-center/softwares/mic
Prerequisite ü The camera is connected to the PC.
1. Install the software according to the installation guide delivered with the software.
2. To install the camera drivers, confirm the corresponding prompts that are displayed during
the software installation process.
3. When the installation is finished, restart your PC.
4. In the device manager of your PC, verify that the installation was successful.

Info
For information on how to acquire images with the specific software, please refer to the respective software manual.

7.2 Installing software on iPad
1. From your iPad, go to the App Store and search for “Labscope” by ZEISS.
2. Follow the instructions on your iPad to install the app.

7.3 Installing TWAIN plugin on PC
The TWAIN plugin for Axiocam 208 color / 202 mono is a standardized software interface to callup and control basic camera functions via a TWAIN-compatible non-ZEISS app.
1. Go to https://portal.zeiss.com/download-center/softwares/mic.
2. Select TWAIN from the list.
3. Click on the Download Button.
à The installation file is downloaded.
4. Open your Downloads folder and unzip the TWAIN installation file.
5. Double-click the TWAIN installation file (.exe).
6. Follow the instructions of the wizard.
For more information see the Quick Guide ZEISS TWAIN for Axiocam 202/208. You can findthe PDF document in the TWAIN download folder.

7.4 Installing TWACKER DEMO Application
To demonstrate the image acquisition with the TWAIN plugin you can use the TWACKER application. TWACKER is not mandatory for operating the TWAIN plugin. If your laboratory software
supports the TWAIN standard, you don't need to install TWACKER.
1. Go to https://portal.zeiss.com/download-center/softwares/mic.
2. Select TWAIN from the list.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

39

7 Installing Software and Drivers | 7.4 Installing TWACKER DEMO Application

ZEISS

3. Click on the Download Button.
à The installation file is downloaded.
4. Open your Downloads folder and unzip the TWAIN installation file.
5. Double-click the TWACK_32.msi installation file.
6. Follow the instructions of the wizard.
For more information see the Quick Guide ZEISS TWAIN for Axiocam 202/208. You can findthe PDF document in the TWAIN download folder.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

40

8 Acquiring Images and Videos | 8.1 Introduction

ZEISS

8 Acquiring Images and Videos
8.1 Introduction
The Axiocam 208 color and the Axiocam 202 mono are high definition cameras for color andmonochromatic imaging, respectively. They are suitable for use as accessories for educational androutine microscopy in laboratory environments and for use by trained laboratory personnel. The
cameras have been designed to be used in the field of light microscopy for general observation,
routine work, and simple applications in which a sufficient amount of light is available.

8.2 Basic procedure using operator panel
Prerequisite ü An USB flash drive has been inserted into the camera's USB port or into the USB hub connected to the camera.
1. Position your specimen on the microscope and adjust the microscope to see a focused image through the eyepieces.
2. Set the microscope light path to divert the image to the camera (e.g. adjust the microscope's tube slider to 50% camera and 50% eyepieces).
3. To take a single image, briefly press the Capture button on the camera.
à The image is saved to the USB flash drive in either JPEG or TIFF format.
4. To start video recording, press the Capture button on the camera for about 5 seconds.
5. To finish video recording briefly press the Capture button on the camera once again.
à The video is saved to the USB flash drive in MP4 format.

8.3 Basic procedure using OSD menu
Prerequisite ü The USB flash drive and mouse/keyboard have been inserted into the USB hub connected tothe camera.
ü The camera is connected to a monitor via HDMI.
ü The OSD menu has opend by pressing the menu button on the camera
1. Position your specimen on the microscope and adjust the microscope to see a focused image on the monitor.
2. To take a single image, click on the Snap button in the OSD menu.
à The image is saved to the USB flash drive in either JPEG or TIFF format.
3. To start video recording, click on the Record button in the OSD menu.
4. To finish video recording click on the Stop button in the OSD menu.
à The video is saved to the USB flash drive in MP4 format.

8.4 Image acquisition with Labscope
Upon first starting Labscope, each screen starts with an overlaying information screen explaining
the functions. Refer to the displayed information for using the software. Disable or re-enable theinformation screens in the software`s Settings menu on your PC monitor iPad.

Info

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

41

8 Acquiring Images and Videos | 8.4 Image acquisition with Labscope

ZEISS

For support in using Labscope, visit our support forum under https://forums.zeiss.com/microscopy/community/viewforum.php?f=34. Please check the Labscope threads for problemsolving notes.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

42

9 Care and Maintenance | 9.1 Optical system

ZEISS

9 Care and MaintenanceTo ensure the optimum performance of the device, preventive maintenance work should be performed at regular intervals.

Time interval

Component

Activity

As required

Infrared filter or protective
glass

Cleaning [} 43]

As required

FirmwareUpdate [} 43]

Tab. 6: Maintenance plan

9.1 Optical system
The internal optical components of the camera should always be protected. If no lens, or camera
adapter with optics, is screwed into the camera's C-Mount thread, the camera's sensor and protective glass must be protected by screwing the protective cap onto the camera's C-Mountthread.

9.2 Cleaning the infrared filter or protective glass

NOTICE
Sensitive optical parts
An inadequate handling of optical components may damage the components or decrease thedevice's imaging quality. The unauthorized handling of device components leads to the loss of
warranty.
4 Do not remove the filter or the protective glass.
4 Do not clean the sensor directly.
4 Do not use tap water to clean the IR filter.

Tool / part

Quantity

Soft brush1

Cotton1

Cleaning fluid for optics

1

Tab. 7: Tools and parts
1. Use a soft brush or cotton to wipe away dry dust from the front side of the infrared filter or
the protective glass.
2. Use cotton and cleaning fluid for optics to wipe away more serious contamination from theinfrared filter.

9.3 Updating the firmware
Follow the subsequent instructions to update the camera's firmware:
1. Download the latest firmware from the webpage http://www.zeiss.com/microscopy/int/
downloads.html.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

43

9 Care and Maintenance | 9.3 Updating the firmware

ZEISS

2. Save the latest firmware update package (".tar" file) to the root folder of the USB flash drive(contained in the scope of delivery).
3. Insert the flash drive into the connected USB hub.
4. At the OSD menu, navigate to Settings > Operating System.
5. Open the Firmware Update menu and press
the update button.

à Note that the update procedure takes several minutes.
à Do not operate the camera or unplug the USB flash drive during the update procedure.
The firmware is updated.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

44

10 Troubleshooting | 10.1 ZEN software

ZEISS

10 Troubleshooting
10.1 ZEN software
Symptom

Cause

MeasureCamera does not appear in
the menu for selectable
cameras.

The camera is not properly connected.

Check and adjust the camera connections to PC and power supply, if necessary.

The software and the drivers are notproperly installed.

Make sure you installed the softwareand the drivers with administrative
rights and according to the instructions in this manual.

Incompatible accessories (e.g. USB
adapters, cables etc.) were used, so
the camera was brought into wrong
mode and failed to be recognized.

1. Connect the camera to the PC
with the accessories provided
in the delivery package.
2. Restart the camera.

The PC's USB driver is outdated.

Upgrade to the latest driver provided
by the PC manufacturer.

The camera is not properly connected.

1. Make sure, the camera's status LED constantly lights blue.
[} 25]
2. If necessary, check and adjust
the camera connections to PC
and power supply.
3. Restart the camera.

The illumination reaching through tothe camera is not sufficient.

1. Check the light path settings ofthe microscope.
2. If necessary, check and adjust
the position of the beam splitter between the ocular and thecamera port.
3. If necessary, check and adjust
the setting of the aperture diaphragm of the microscope, if
necessary.
4. Execute an automatic exposure
measurement.

Inappropriate display settings areused.

Check and adjust the display device's
settings for live imaging, if necessary.

The color matching is inappropriate.

Set the color temperature.

No camera image visible on
the screen.

The color of the displayed
image taken by Axiocam
208 color does not correspond to the image seen
through the eyepieces.

Check the monitor's color temperature setting. If necessary, reduce thecolor temperature to the lowest value
possible.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

45

10 Troubleshooting | 10.2 Camera

ZEISS

Symptom

Cause

MeasureThe color of the displayed
image taken by Axiocam
202 mono does not correspond to the image seen
through the eyepieces.

Unsuitable overlay colors are used torepresent fluorescence dyes.

Select alternative overlay colors.

Symptom

Cause

MeasureThe LED indicator is off.

The camera is not powered on properly.

In case of Axioscope 5/7 and Axiolab
5, make sure the microscope is powered on and the camera is connected
to microscope via Micro-D power cable.

10.2 Camera

Alternatively, make sure the camera is
powered by the plug-in power supply.
The LED indicator flashes
red.

The camera is updating firmware or
being reset.

NOTICE! Do not switch off thepower supply.

The image / video cannot besaved to the USB flash drive.

USB flash drive is not in correct format.

Format the USB flash drive to FAT32
format on a PC.

USB flash drive has not enough freememory.

Make sure there is enough free memory on the flash drive.

USB flash drive cannot be recognized.

Restart the camera.

USB flash drive is not in correct format.

Format the USB flash drive to FAT32
format on a PC.

USB flash drive does not enough freememory.

Make sure there are at least 200 MBof free memory on the USB flash
drive.

USB flash drive cannot be recognized.

Restart the camera.

The firmware cannot be found.

Make sure the latest firmware isstored to the root folder of the USB
flash drive.

The firmware is not uploaded properly.

Restart the firmware update process
and exactly follow the instructions in
the firmware update menu.

Date and time are not set correctly.

1. At the OSD menu, navigate to
Settings > Operating System.
2. Tap the Date & Time button.
3. Set the camera date and time.

The buffer battery is empty.

Please contact your local ZEISS service
organization to change the battery.
The expected battery lifetime is approx. 4-5 years.

The firmware update does
not function.

The camera date and time is
not correct.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

46

10 Troubleshooting | 10.3 Labscope

ZEISS

Symptom

Cause

MeasureThe image has severe noise.

The amplification (gain) is set too
high.

Manually reduce the gain value.

The exposure time is set too low.

Manually adjust the exposure time.

The light intensity is set too low.

Increase the light intensity. Activate
denoise function in Image settingmenu.

The image is too dark or too
bright.

Automatic exposure time has notbeen activated.

Activate the automatic exposure settings or manually adjust the exposure
settings to the current light situation.

The camera settings are notsaved after a camera
restart.

The camera has been powered off toearly after changing the settings.

For the settings to be automaticallystored, wait at least 5 seconds after
changing the settings before you
power off the camera.

Monitor connected via
HDMI does not display an
image.

The camera is not delivering a signal,or signal is not compatible with themonitor.

Ensure that the camera has beenswitched on for at least 30 seconds
and the LED indicator is blue. Check
the plug connections on the cameraand monitor.

For Axiocam 208 color,when switching to 4K, themonitor becomes black andthen reverts to 1080p.

The monitor does not support 4K.

Use a monitor supporting 4K resolution (e.g. Monitor TFT 32" HP Z32,
Order No. 410350-3201-000), listed
on ZEISS micro-shop website.

The image appears distorted
on full screen monitor.

The monitor's image aspect ratio is
not set to 16:9.

Set the monitor's aspect ratio to 16:9.

The image is blurred on thescreen, but the sample is in
focus through the eyepieces.

The focus plane of the camera is different from that of the eyepieces.

1. Focus the sample through theeyepieces.
2. Calibrate the camera adaptor
until image is in focus on themonitor.

Buttons do not react whenbeing pressed.

The buttons are temporarily locked,
as the camera is being accessed exclusively from the PC or network.

Control the camera via PC software or
ZEISS app, or close the PC software or
ZEISS app.

The Camera otherwise behaves abnormally.

The camera may have been brought
into a non-intended state.

Press the Camera factory reset button on the camera.

10.3 LabscopeInfo
For support in using Labscope, visit our support forum under https://forums.zeiss.com/microscopy/community/viewforum.php?f=34. Please check the Labscope threads for problemsolving notes.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

47

11 Disposal and Recycling​

ZEISS

11 Disposal and Recycling
This product has been developed, tested and manufactured in accordance with the applicable environmental provisions and directives of the European Union:§

The product and its accessories comply with EU directives 2015/863/EU (RoHS) and 2012/19/
EU (WEEE), insofar as these apply to this product.

§

ZEISS has implemented a take-back and recycling process that ensures that proper recycling iscarried out in accordance with the aforementioned EU directives.

§

Please contact your ZEISS sales/service organization for details relating to disposal and recycling.

§

This product must not be disposed of with domestic waste or using municipal waste disposal
services. In the event of resale, the seller must inform the buyer of the need to dispose of the
product appropriately.

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 | 426570-7011-001

48

Instruction Manual ZEISS Axiocam 208 color / 202 mono | en-US | Rev.5 |Modifications reserved.
Carl Zeiss Microscopy GmbH
Carl-Zeiss-Promenade 10
07745 Jena
Germany

phone: +49 3641 64 3161
fax: +49 3641 64 3439
info.microscopy@zeiss.com
www.zeiss.com/microscopy