SVCam-ECO
ECO-Line CCD-Cameras

SVCam-ECO cameras support the industrial image transfer protocol of Gigabit Ethernet.
Gigabit Ethernet delivers to your application absolutely new dimensions:! Extreme small design! High frame rates
Industrial standard
Cable length up to 100m
Budgetary cost level
SVCam-ECO cameras are available in 8 resolutions from 640 x 480
up to 5 megapixel. These cameras are designed to reach high frame
rates at low signal-to-noise ratios and are enclosed in an utmost smallhousing.

The modular concept of SVCam-cameras provides our customers with afast and low cost way to design customized versions for application
specific requirements. Unique processing of the CCD-signal guarantees an
excellent signal-to-noise ratio. The internal logic allows different ways toadjust exposure time and select trigger modes including:! Synchronisation of image capture to an external event (trigger mode)"Free running" with adjustable frame rateExposure time control via Ethernet interface or over trigger pulse widthExtend the exposure time under low light level conditions. Hi IR sensitivity (eco428+285)lProgressive scan technologyl8/12 Bit video data stream (14 Bit ADC used)l
8 resolution options:leco424640 x 480 pixelleco415782 x 582 pixelleco428768 x 494 pixelleco2041024 x 768 pixelleco2671360 x 1024 pixelleco2851360 x 1024 pixelleco2741600 x 1200 pixelleco6552448 x 2048 pixell1000 MBit Ethernet (Gigabit Ethernet) 100 MBit compatiblelSynchronization:l"free running" (frame rate adjustable)lexternal trigger with internal exposure controllexternal trigger with pulse width exposure controll
Software trigger via PCl2 x 2 binning model
Housing dimensions: e.g. 38mm x 38mm x 33mml
Operating temperature range: -10°C to +40°ClWeight approx. 100gl
Temperature sensor to avoid overheatingl
Monochrome and color sensors (Bayer Pattern) with LUT (4 x)lStatus LEDl
64 MB memory insidel
Full 2 years warrantylSelectable AOIlAdjustable gain/auto gainlStrobe output (e.g. for light flash)lGeneral purpose I/Osl
Time stamp capabilityl
Low offsetlPartial scan mode for higher frame rates(AOI)
lCS-Mount; less space consumingl
10V - 25V DC; e.g. 300mA @ 12V consumption

THE FOCAL P INT OF MACHINE VISION

SVCam-ECO
Camera Types:
Camera Typeeco424

eco415eco428*eco204

Resolution

640 x 480

780 x 580

768 x 494

1024 x 768

12486

60

47

Pixel (µm²)

7.4 x 7.4

8.3 x 8.3

8.4 x 9.8

Exposure Time

3 µs - 2 s

22 µs - 2 s

3 µs - oo
1/3”

Frame Rate (Hz, max.)

eco267

eco285

eco2741360 x 1024 1360 x 102425

eco655*1600 x 1200 2448 x 2048

202084.65 x 4.65

4.65 x 4.65 6.45 x 6.45

4.4 x 4.4

3.45 x 3.45

65 µs - 2 s

17 µs - 2 s

40 µs - 2 s

16 µs - 2 s

26 µs x 2 s

40µs - 1 s

22 µs - oo

65 µs - oo

17 µs - oo

40 µs - oo

16 µs - oo

26 µs - oo

40µs - oo

1/2”

1/2"

1/3"

1/2"

2/3"

1/1.8”

2/3"

internal

Exposure Time
external

CCD-Size Equivalent*PRELIMINARY
Operation Modes:

Configuration Software:

Mode: Free Running

The SVCam-ECO cameras come with our "SVCapture"-software, which allows
easy interactive setup of all camera parameters. The software including a SDK
supports Windows XP including 64 Bit operating system. A LINUX Driver is alsoavailable. The camera can be configured using the XML File stored inside thecamera. This complies also with the international GenICam standard.

In this mode the camera creates all sync signals itself. The frame rate is at its
maximum or less
Exposure
Exposure
Exposure
Exposure(adjustable) and there istime (n-1)
time (n)time (n+1)
time (n+2)
no need to trigger theImage (n-1)
Image (n)Image (n+1)
camera in order to getdata. Exposure time canbe set by using the Ethernet interface of the PC. The enclosed software allows theuser to set the specified values. Change of exposure time is possible from one
frame to the next. The time set stays resident after power off.

Dimensions:

Mode: External Trigger, Internal Exposure Control

In this mode the camera isEXSYNC- Trigger
waiting for an externalExposure
trigger which starts
time
integration and read out.
ImageExposure time can bevaried using the length ofthe TRIGGER pulse (i.e. between the rising and the falling edge). The time
settings in the control software are not activated. This mode is useful inapplications where the light level of the scene changes during operation. Changeof exposure time is possible from one frame to the next.

rear viewside view

38mm

38mm

32,5mmCS-Mount

38mm

Mode: External Trigger, External Exposure Control

26mm

In this mode the camera
EXSYNC Trigger
starts the image acquisition after an externalExposure
trigger event. The expotime
sure time is controlled byImage
the camera. The value forthe exposure time can be
set by using the Ethernet interface of the PC. The trigger signal is fed into thecamera directly by the industrial connector. Change of exposure time is possible
from one frame to the next.

front view

26mm

Ordering Guide:
Monochrome:eco424MVGE
eco415MVGE
eco204MVGE
eco428MVGE
eco267MVGE
eco285MVGE
eco274MVGE
eco655MVGE

Color:
eco424CVGE
eco415CVGE
eco204CVGE
eco428CVGE
eco267CVGE
eco285CVGE
eco274CVGE
eco655CVGE

Options:! Mounting bracket with tripot thread fits on frontplate threaded holes! CS- to C-Mount adapter ring

SVS-VISTEK GmbH
Changes without
notification 09.09.2009

Mühlbachstraße 2082229 Seefeld/Germany

TEL.
+49-(0)8152-99 85-0
FAX+49-(0)8152-99 85-79
E-MAIL info@svs-vistek.com
WEB www.svs-vistek.com