RF POWER AMPLIFIER■ All Solid-State Amplifier■ Broadband Frequency:200MHz～1000MHz■ Output Power(CW) :1kW(min.) @1dB Comp.

R&K-A201M102-6060R

APPLICATIONEMC(Electromagnetic Compatibility)
Accelerator Application
Telecommunication Test and Measurement Application
Etc...

SPECIFICATIONS ＠＋25℃: 200MHz ～ 1000MHz
Frequency Range: ＋60.0dB (min.)
Small Signal Gain: ±2.5dB (max.)
Gain Flatness: 1kW (min.) @1dB Comp.
Output Power (CW)
: ＋66.0dBm (typ.)Output I.C.P.
: －20.0dBc (max.) @Po=1kWHarmonics: －70.0dBc (max.) @Po=1kW
Spurious: 50Ω
Impedance: 2.0 (max.)
Input VSWR
: 3.0 (max.)
Output VSWR
Maximum RF Input Power : ＋3.0dBm: AC200V±15％／3φ, 50／60Hz, 20kVA
AC Supply Input
0℃ to ＋35℃
Operating Temperature :: －15℃ to ＋55℃
Storage Temperature: RF-IN
N -FEMALE @REAR PANEL
Connectors
RF-OUT 7/16-FEMALE @REAR PANEL: (W)560mm×(D)1000mm×(H)1276mm
Size(Excluding Projection): 600kg (max.)
Weight: Forced Air Cooling
Cooling: Over Temperature Protection
Protection Circuits
Power Supply Voltage Protection
Output Over Power ProtectionOutput Over Reflection Power Protection: R&K Multi Monitoring System
Other Functions
GP-IB(IEEE-488)
Safety Interlock(EMERGENCY)

TYPICAL PERFORMANCE (Temp @+25℃)
Small Signal Gain

80

Input / Output VSWR

3.0
2.8

62

2.4

VSWR

60

60

2.2
2.0

58

1.8
1.6

50

56

1.4
1.2200

66

400

600

800

Frequency (MHz)

1000

200

Output Power 1dB Gain Comp.

600

6058
56
5452

1000

Output Harmonics

0

Output Harmonics (dBc)

62800

Frequency (MHz)Output Power = 1kW-1064

Output Power (dBm)400

Output Power (dBm)

Gain (dB)

Input VSWROutput VSWR

2.670

Output Power Linearity

64545250

48-20-30

46

Frequency
200MHz400MHz600MHz800MHz
1000MHz-40
44-50-60

42-70
200

400600800

Frequency (MHz)1000

200400600

800

Frequency (MHz)R&K reserves the right to make changes in the specifications of or discontinue products at any time
without notice. R&K products shall not be used for or in connection with equipment that requires anextremely high level of reliability and safety such as aerospace uses or medical life support equipment
Further, The export of R&K products from Japan may be subject to an export license by the governmentof Japan, based on Japan's "Foreign Exchange and Foreign Trade Law".

1000-15-13-11-9-7-5-3

Input Power (dBm)-1

13

RF POWER AMPLIFIERTYPICAL PERFORMANCE (Temp @+25℃)■Output Power Stability

R&K-A201M102-6060R

Output Power vs Time

61.50

Output Power (dBm)

61.00

60.50

60.00

59.50

59.00

Measured at 500MHz0102030

4050

6070

80

90

100120110

Time (minutes)

OUTLINE DRAWING※IN MILLIMETERS

SIDE VIEW

560

1000

REAR VIEW

56

FRONT VIEWREF MON

FWD MON-60dB Coupling-60dB CouplingJ2101

REF MON IN

FWD MON IN

RF OUT

J2103

J2104

J2102

RF IN

DC+12V OUTADDRESS

A201M102-6060R

GP-IB CONTROL

EXTERNAL MON-A

RF POWER AMPLIFIERCONTROL UNIT

ST-BY

CONTROL-A

1 1
LOCAL

REMOTERF ON/OFFP01
ON
1.0kW
100W

RESET

POWER

Page Up

J2121
CONTROL-B

J2123

J2125

RF SW
FOWARD
REVERSE

J2111
PS ON/OFFAC OUT-1

EMERGENCY IN
J2122
EXTERNAL MON-B

LO=ON
J2106

J2105
J2124

AC OUT-4

AC OUT-3

J2112
AC OUT-2

J2115

J2114

J2113

LINE INPUT
AC200V，3Φ，50/60Hz

Page Down

RF INPUT

A201M102-6060R
RF POWER AMPLIFIERADDRESS
1DRIVER AMP／PD UNIT

Debug

CONTROL-A

2J223

J2201

J2221
CONTROL-B

RF OUT-1

RF OUT-3

RF OUT-5
J2222
AC200V IN

J2202

J2204

J2206

RF OUT-2

RF OUT-4

RF OUT-6

J2203

J2205

J2207

Debug

CONTROL-A

POWER

A201M102-6060R
RF POWER AMPLIFIERFINAL AMP.-1 UNIT

J2211

CONTROL-B

ADDRESS
12AC200V IN
RF OUT-1

RF OUT-2

RF IN-1

RF IN-21276

POWER

A201M102-6060R

Debug

RF POWER AMPLIFIERFINAL AMP.-2 UNITCONTROL-A

CONTROL-B

ADDRESS
12AC200V IN
RF OUT-1

RF OUT-2RF IN-1

RF IN-2

POWER

A201M102-6060RDebug

RF POWER AMPLIFIERFINAL AMP.-3 UNITCONTROL-A

CONTROL-B

ADDRESS
1BBW360

LINE INPUT
AC200V，3Φ 50/60Hz
R93

POWER

R&K reserves the right to make changes in the specifications of or discontinue products at any time
without notice. R&K products shall not be used for or in connection with equipment that requires anextremely high level of reliability and safety such as aerospace uses or medical life support equipment
Further, The export of R&K products from Japan may be subject to an export license by the governmentof Japan, based on Japan's "Foreign Exchange and Foreign Trade Law".

S RF OUT-1
TGND

LINE INPUT
AC200V，1Φ 50/60Hz
RF OUT-2

RF IN-1

AC200V IN
RF IN-22