M100

M-SERIES

M100

COMPACTUP TO 3.1 MW RF PEAK
POWER
FULLY COMPUTERIZEDMINIMIZEDCONSUMABLES

THE M-SERIESMAGNETRON MODULATOR UP TO 3.1 MW
ScandiNova's M100 is a stand-alone pulse Modulator
in a very compact enclosure. The Modulator is optimized
for a wide range of different magnetrons with an RF
peak power ranging from 1.0 MW to 3.1 MW. The M100
is also available as an integrated and/or dual version(see M100-i, M100D, and M100D-i).
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers a user-friendly interface. The pulse length,
amplitude and repetition rate can all be changed.
The M100 can be delivered as a pure high-power pulse
modulator or as part of an RF Unit including magnetronand related components. Different levels of service andsupport programmes are available.

ScandiNova's M-series
contains a range of magnetronpulse modulators with highreliability and performance,
a compact design and low
energy consumption.
Our magnetron pulse
modulators require about
one third of the space ofalternative solutions yet still
offer around 30% higherefficiency, two operating
advantages that especiallyappeal to cancer treatment
providers.
Their high reliability andperformance also attract manyindustry applications.

Excellence in pulsed powerwww.scandinovasystems.com

M100

UNIT

VALUE

NOTES

Magnetron RF peak power

MW

1-3.1

Depends on choice of magnetron

Magnetron RF average power

kW

2.8

Maximum, depends on choice of magnetron

Modulator peak power

MW

6.2

Maximum

Modulator average power

kW

8Maximum

Pulse voltage

kV30-52

Pulse current

A30-120

RF pulse length, topμs

0.5-5Depends on max average power

Pulse repetition rate

Hz0-500

Depends on max average power

Modulator Voltage Stability, RMS%0.4

Verified on resistive load (see options)

Water coolingl/min818-40 °C, non-condensing

INTERFACECONNECTOR

DEFAULTOPTION

Mains power, three-phase

Terminal block (4-wire)

400±10 % VAC, 50/60 Hz

208/380/480 VAC

Mains power, single-phaseC20

230±10 % VAC, 50/60 Hz

115 VAC

Control interface

RJ45

Modbus TCP

Water cooling interface

12 mm

Legris Push-in

SwagelockTrig input

BNC

Electrical

OpticalDiagnostics

BNCPulse voltage & current signals

SIZE AND WEIGHT
Weight (incl. oil)

UNITkgVALUE
140

The Standard Modulator Includes

Services

Filament power supply

Training in handling, operationand maintenanceControl system
For more information, visit www.scandinovasystems.com/M100

Control via Modbus TCP
Pulse sensors

515 mm

HV pulse cable

647 mm12-32 mm

DOC-004504-02 / 2024-01-19

SPECIFICATIONS

401 mmService contract

Typical Magnetron Loads

Factory acceptance test certificate

BVERI

GLVAC

VE2110

VE2110A

Manuals

Options
Pulse/RF diagnostics
Enhanced PRF range (1000–2000 Hz)

Digitizer

Information contained in this document is subject to change
without notice.

Installation and start-up

Internal trig generator

Enhanced stability (down to 0.1 %)Contact ScandiNova for any requirements not covered bythis specification.

ShippingGunport

Nisshinbo
M1603

GLM

M1466

GLM5193

MX7640

GLM5810

M1466N

GLM6090

M1466T

Peaking components

Additional System Components
Circulator & RF loads
Directional coupler
Magnet power supply
Waveguide windows
Magnetron

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

VE2098CPIE2V

VMC3109

MG2025

VMC3136

MG5125
MG5193
MG6090MG6493
MG7095

Excellence in pulsed power