M100D-i

M-SERIES

M100D-i

DUAL-ENERGYCOMPACTUP TO 3.1 MW RF PEAK
POWER
FULLY COMPUTERIZEDMINIMIZEDCONSUMABLESMAGNETRON MODULATOR UP TO 3.1 MW
ScandiNova's M100D-i is a compact dual-energy
pulse Modulator designed for system integration. Our
dual-energy models are for applications where there
is a need to switch between two different energy
levels. The Modulator is optimized for a wide range ofdifferent magnetrons with an RF peak power ranging
from 1.0 MW to 3.1 MW. To facilitate system integration,
the Modulator has a split design with a Tank Unit anda Pulse Unit, and can be placed in a very compact
enclosure. The M100D-i is also available as a singleenergy and / or stand-alone version (see M100, M100-i,and M100D).
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers a user-friendly interface. The pulse length,amplitude and repetition rate can all be changed.
The M100D-i can be delivered as a pure high-power
pulse modulator or as part of an RF Unit including
magnetron and related components. Different levels ofservice and support programmes are available.

THE M-SERIES
ScandiNova's M-series
contains a range of magnetronpulse modulators with highreliability and performance,
a compact design and low
energy consumption.
Our magnetron pulse
modulators require about
one third of the space ofalternative solutions yet still
offer around 30% higher
efficiency, two operating
advantages that especiallyappeal to cancer treatment
providers.
Their high reliability andperformance also attract manyindustry applications.

Excellence in pulsed powerwww.scandinovasystems.com

UNIT

VALUE

NOTES

Magnetron RF peak power, high

MW

1-3.1

Depends on choice of magnetron

Magnetron RF peak power, low

MW

0.9–2.5

Depends on choice of magnetron

Magnetron RF average power

kW

2.8Maximum, depends on choice of magnetron

Modulator peak power, high

MW

6.2

Maximum

Modulator peak power, low

MW

5.2

Maximum

Modulator average power

kW8Maximum

Pulse voltage

kV30-52

Pulse current

A30-120

RF pulse length, topμs

0.5-5

Depends on max average power

Pulse repetition rate

Hz0-500

2x250 Hz, depends on max average power

Modulator Voltage Stability, RMS%

0.4

Verified on resistive load (see options)

Water coolingl/min818-40 °C, non-condensing

INTERFACECONNECTOR

DEFAULTOPTION

Mains power, three-phase

Terminal block (4-wire)

400 VAC±10 %, 50/60 Hz

208/380/480 VAC

Mains power, single-phase

C20

230 VAC±10 %, 50/60 Hz

115 VAC

Control interface

RJ45

Modbus TCP

Water cooling interface

12 mm

Legris Push-in

SwagelockTrig input

BNC

Electrical

OpticalDiagnostics

BNC

Pulse voltage & current signals

SIZE AND WEIGHT

UNIT

VALUE

Weight, Pulse Unit

kg68Weight, Tank Unit (incl. oil)kg76

For more information, visit www.scandinovasystems.com/M100-i

The Standard Modulator Includes

Services

Filament power supply

Training in handling, operationand maintenanceControl system
Control via Modbus TCP
Pulse sensors
Cable kit between Pulse Unit and Tank Unit
HV pulse cable

345 mmInternal trig generator
Factory acceptance test certificate
Manuals463.5 mm

407 mm564.4 mm
Pulse Unit

266 mm

M100D-i
DOC-004507-02 / 2024-01-19

SPECIFICATIONS

402 mm

462 mm
528 mmTank Unit

Options

Shipping
Installation and start-up
Service contractTypical Magnetron Loads
BVERI

GLVAC

VE2110

VE2110A

Nisshinbo

VE2098

Pulse/RF diagnostics

M1603

GLM

Enhanced PRF Range (500–1000 Hz)

M1466

GLM5193

Enhanced Stability (down to 0.1 %)MX7640

GLM5810

Digitizer

M1466N

GLM6090

Peaking components

M1466T

Additional System Components

E2V

VMC3109

Circulator & RF loads

MG5193

VMC3136

Directional coupler

MG6090Magnet power supply

MG6493

Waveguide windows

MG7095CPI

Magnetron

Contact ScandiNova for any requirements not covered by this specification.
Information contained in this document is subject to change without notice.

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power