PVR--1500
Ver.11

開放電圧 １５００Ｖ
００Ｖ

内部抵抗 ３ステージ
ステージ 判定もくじPVR-1500 取 扱 説 明 書１．はじめに .......................................................................................................................................................................................... 2
１－１．お問 い合 わせ先 ................................................................................................................................................................ 2
１－２．保 証 について ..................................................................................................................................................................... 2
２．安 全 にお使いいただくために ....................................................................................................................................................... 3２－１．警 告 ..................................................................................................................................................................................... 3
２－２．高 電 圧注 意 ....................................................................................................................................................................... 5３．測 定 に関する注 意事 項 ............................................................................................................................................................... 6４．製 品 について ................................................................................................................................................................................. 7４－１．製 品 の概 要 ........................................................................................................................................................................ 7
４－２．測 定 項目 ............................................................................................................................................................................ 8５．製 品 構 成 ........................................................................................................................................................................................ 9５－１．製 品 内容 ............................................................................................................................................................................ 9
６．各 部 名 称 ...................................................................................................................................................................................... 10６－１．正 面 ................................................................................................................................................................................... 10
６－２．背 面 ................................................................................................................................................................................... 10
７．使 用 方 法 ...................................................................................................................................................................................... 11
７－１．起 動 と停 止 ....................................................................................................................................................................... 11
７-１-１．起 動 .......................................................................................................................................................................... 11
７-１-２．停 止 .......................................................................................................................................................................... 12
７－２．測 定 手順 .......................................................................................................................................................................... 12
７－３．測 定 条件 の確認 ............................................................................................................................................................. 13８．測 定 ............................................................................................................................................................................................... 14
８－１．測 定 準備 .......................................................................................................................................................................... 14
８－２．測 定 ................................................................................................................................................................................... 14
８－３．Ｅｒｏｒｒ表 示 ........................................................................................................................................................................ 15
８－４．再 測 定（ＲＥＴＲＹ） ........................................................................................................................................................... 15
８－５．測 定 データ確 認 ............................................................................................................................................................... 16
９．グループ設 定 ............................................................................................................................................................................... 17１０．判 定値 設 定 ............................................................................................................................................................................... 18
１０－１．設定 画 面 ....................................................................................................................................................................... 18
１０－２．判定 値 設 定 .................................................................................................................................................................. 19
１１．時 計設 定 ................................................................................................................................................................................... 21
１１－１．時計 設 定 ....................................................................................................................................................................... 21
１２．保 存データについて ................................................................................................................................................................. 22
１２－１．ＰＣとの接続 .................................................................................................................................................................. 22
１２－２．保存 データのフォルダ構 成 ......................................................................................................................................... 23
１２－３．保存 データのフォーマット ............................................................................................................................................ 24
１２－４．全保 存 データ消 去 ....................................................................................................................................................... 25
１３．装 置仕 様 ................................................................................................................................................................................... 26
１３－１．測定 仕 様 ....................................................................................................................................................................... 26
１３－２．一般 仕 様 ....................................................................................................................................................................... 27
１４．トラブルシューティング ............................................................................................................................................................. 28

1

Shin-ei Electronic Measuring Co.,Ltd.

１．はじめにPVR-1500 取 扱 説 明 書

このたびは弊社製品をお買い上げいただき、誠にありがとうございます。
ご使用の前に必 ずこの説明書をお読みの上、正しくお使いいただきますようお願い申し上げます。また、本取扱説明書は必ず保管していただき、必要なときにお読みください。
不明点やご質問 、その他お悩みの点がございましたら下記「お問 い合わせ先 」までご連 絡 ください。

１－１．お問い合わせ先

新栄電子計測器株式会社
本社〒252-0816 神奈川県藤沢市遠藤 2636

サポート専用窓口

http://www.shin-ei.ne.jp
TEL：

0466-88-3030（代表）FAX：

0466-87-0627

TEL：

050-5527-4444

FAX：

0466-87-0627

１－２．保証について

本製品の保証期間は、使用の有無に関わらず、購入から１２ヶ月です。



製品を発送する場合は、付属品も含めすべてお送りください。



お客様から弊社へ製品を送付いただく際の送料は、お客様負担となりますので、あらかじめご了承ください。

Shin-ei Electronic Measuring Co.,Ltd2２．安全にお使いいただくために安全にお使いいただくために

PVR-1500 取 扱 説 明 書

この取扱説明書には、本製品を安全に
説明書には、本製品を安全にご使用いただくための注意事項を記載しています。
使用いただくための注意事項を記載しています。
本書をよくお読みになり、使用方法を必ず守りながら正しくお使い
本書をよくお読みになり、使用方法を必ず守りながら正しくお使いください。
さい。
この表 示を無 視 して誤誤 った取 り扱 いをすると、感いをすると、 電 などの警告

けがによる重 症 、または死 亡 を負 う可 能 性 や装 置 の破 損が発生する可能性を示しています。
高電圧が加わる部分です。この表示を無視して誤った取り
この表示を無視して誤った取り

高電圧注意

扱 いをすると、感
感 電 などのけがによる重 症 、または死 亡 を負う可能性や装置の破損の可能性を示しています。

注意

この表示は装置の取扱いまたは動作に関する重要な注意点である事を示しています。（本文中にのみ記載。）

２－１．警告本
本来の目来の目的以外には使用しないでください。
的以外には使用しないでください。



分
分解、改
解、改造などしないでください。
造などしないでください。



定 格電 圧 DC1500
00【V】を上回る回路には、絶対に接
絶対に接 続しないでください。
入力範囲を超えて使用した場合に機器故障や使用者の感電を引き起こす恐れが
あります。



煙 や異臭などが発生した場合どが発生した場合、すぐに電源スイッチを切り、弊社へご連絡ください。
すぐに電源スイッチを切り、弊社へご連絡ください。
異常状態のまま使用
異常状 態のまま使用 すると、すると、装置に触れた時に感電したり、装置が原因したり、装置が原因 となる火災発
生の可能性があります
生の可能性があります。



異 物 や水 などが装置などが装 置 内部 に入 った場 合 、すぐに電
すぐに電 源 スイッチを切 り、弊 社 へご連 絡 ください。
そのまま使用すると、火災や感電の原因となります。
用すると、火災や感電の原因となります。



防防水処理を行っておりません。
水処理を行っておりません。
降雨や降雪の可能性がある場合は使用しないでください。

3

Shin-ei Electronic Measuring Co.,
Co.,Ltd.

２．安全にお使いいただくために

PVR-1500 取 扱 説 明 書



爆
爆発の危
発の危険性がある粉塵、ガス、蒸気、水分のある環境では使用険性がある粉塵、ガス、蒸気、水分のある環境では使用 しないでください。



測測定は、商用電
定は、商用電 気などの気などの交流電力の回路には絶対に接続しないでください。
交流電力の回路には絶対に接続しないでください。

激
激しい振動や落下などの強い衝撃を与えないでください。
しい振動や落下などの強い衝撃を与えないでください。

高高温になる場所に放置しないでください。
温になる場所に放置しないでください。
長時間太陽光にさらされる場所や密閉空間などに放置すると故障の原因になります。



測
測定中は電極からプローブを外さないでください。
定中は電極からプローブを外さないでください。
外れてしまった場合は、測定完了後までプローブを電極に場合は、測定完了後までプローブを電極に再度再度接触しないでください。

保保管場所に注意してください。
管場所に注意してください。
ほこりや粉 塵 の多い場所 、直 射 日 光のあたる場 所 、温 度 湿 度の変 化 が激 しい場 所 、
火気のある場所、水に濡れやすい場所、振動のある場所は避けてください。

仕仕様書に記載のある環境条件の範囲内でご使用ください。
様書に記 載のある環境条件の範囲内でご使用ください。



ベンジンやシンナー、アルコールなど
ベンジンやシンナー、アルコールなどを使用しないでください。



電電池のプラスとマイナスを逆に挿入しないでください。
池のプラスとマイナスを逆に挿入しないでください。
発熱、液漏れ、破裂、発熱、液漏れ、破裂、装置破損の恐れがあります。



長
長期間使用
期間使 用 しない場合、電池を取り外して保管してください。
液漏れ、装置破損の恐れがあります。
破損の恐れがあります。



使使い切った使用済
い切った使用済 み電池
み電池は、すぐに取り出してください。
使
使い切
い切った電
った電 池
池を接
を接続
続 したまま長期間 放置しますと、発 熱、液 漏れ、破 裂 、、装 置 破損の恐れがあります。



電 池は、古い電、古い電 池、銘柄や種類の異なる電池を混ぜて使 用しないでください。
発熱、液漏れ、破裂、発熱、液漏れ、破裂、装置破損の恐れがあります。

Shin-eiei Electronic Measuring Co.,
Co.,Ltd4２．安全にお使いいただくために

PVR-1500 取 扱 説 明 書２－２．高電圧注意
絶 縁用防 護具を着用を着用 して作業を行ってください。
測 定 中 、感感 電 防 止 のため、のため、必 ず絶 縁用 防 護具（絶 縁 ゴム手 袋 や絶や 縁 ゴム長 靴 ）を着用して作業を行ってください。

本
本装置および太陽装置および太陽電池は濡れた手で触れないでください。
電池は濡れた手で触れないでください。
感電や漏電事故の原因となる可能性があります
感電や漏電事故の原因となる可能性があります。



感
感電にご注意ください。
電にご注 意ください。
本 装 置 と太 陽 電 池を接続する場合
池を接 続する場合 は、事前 に必ず接 続する太 陽 電池 をテスターなど
により測定し、感電の恐れがないか確認の上、作業を行ってください。



測 定に付属の測定ケーブル、プローブ以外は使用しないでください。
に付属の測定ケーブル、プローブ以外は使用しないでください。



ご使用前に必ず本装置を点検し、ひびや欠けがないことを確認してください。
ご使用 前に必ず本装置を点検し、ひびや欠けがないことを確認してください。
本装置に損傷がある場合は絶対に使用しないでください。



測測定前の点検確
定前の点検確認
認・ご使用前に必ず測定ケーブル、プローブの損傷・金属の露出がないことを確認してご使用前に必ず測定ケーブル、プローブの損傷・金属の露出がないことを確認してください。損傷がある場合は絶対に使用しないでください。
・太太陽電池
池およびパワーコンディショナを含
およびパワーコンディショナ を含 む周辺 機 器 の電 源 を切り、パワーコンディ
ショナおよび周 辺 機器 を切り離 し、本 装置 以 外が接 続 されていない状 態 にしてください。
・電池カバーおよびーおよび USB キャップを正しく装着してください。

5

測測定中は確実に測定中は確実に測定ケーブル、プローブを太陽電池に接続してください。
定ケーブル、プローブを太陽電池に接続してください。

Shin-ei Electronic Measuring Co.,
Co.,Ltd.

３．測定に関する注意事項
日 射 量



PVR-1500 取 扱 説 明 書

測定時の日射量は 1000【Ｗ/㎡】に近い値（開放電圧が正常に出ている状態）を推奨しています。

測 定 値



結 果 表 示 に表 示 される抵 抗 値 は「推 定 値 」であり、モジュールの種 類 、モジュールの枚 数 、延 長 ケーブルの長 さ、アースケーブルの施 工状態、測定時の設定閾値、日 射量 、外気 温度、地 中 水分量などの諸条件により誤差や誤検知が生じることがあります。

結 晶 系



測 定 可 能 な太 陽 電 池 の構 成 条 件 は、単 一 の直 列 接 続 された結晶系太陽電池モジュールおよび太陽電池ストリングです。

最大１５００Ｖ

太陽電池モジュールの開放電圧が 1500【V】以下です。
上記範囲を超えるストリングでは正しく判定されない可能性があります。機器の故障を招く恐れがありますので、絶対に接続しないでください。

ノ イ ズ



測定時にプローブが確実に接触されておらず接触不良を起こしていた場合、測定した抵抗値が「Ｅｒｒｏｒ」と表示されます。
その場合はＲＥＴＲＹボタンを押して再度計測を行ってください。
数 回 測 定 しても測 定 結 果 が「Ｅｒｒｏｒ」と表 示 される場 合 、太 陽 電
池 の出力のノイズやパワーコンディショナからのノイズの回 り込 みにより測定できない場合があります。

Shin-ei Electronic Measuring Co.,Ltd6

４．製品について

PVR-1500 取 扱 説 明 書

４－１．製品の概要
この製 品 は、太 陽 光 発 電 システムの施 工 業 者、保 守 ・管 理 業 者 、メンテナンス業者 が、施 工時や定期点検時など、太陽光発電システムの動作確認でのご使用に有効です。
故 障 不 具合 を、開放 電 圧 ・内 部 抵 抗 の測 定 で発 見 します。簡 単 な測 定 操 作 での良 否 判 断 、
データ取得を行う、作業 効率の向上を目的とした製品です。


太陽光発電システムの開 放電圧と内部抵抗を測定します。



測定は、ストリング単 位で行います。太陽電池 1 枚も測 定できます。



1 ストリングスあたり 1 秒の高速測定で効率の良い測定ができます。



開放電圧でストリングの発電状況がわかります。



内部抵抗の測定 結果を 3 段階（低抵抗・高抵抗・異常抵 抗）で判定 し、発電不良を知ることができます。

 内部メモリは、２０００ストリングの大容量です。
大規模発電所のデータ管理によるメンテナンスが行えます。

7

Shin-ei Electronic Measuring Co.,Ltd.

４．製品について

PVR-1500 取 扱 説 明 書

４－２．測定項目
 開放電圧値測定（パワーコンディショナに接続しない時のストリング電圧）
太 陽 電 池 が直 列 に接 続されたストリング
パワーコンディショナ

1 枚の太陽電池の開放電圧×直列枚数が測定する開放電圧になります。
例 え ば 開 放 電 圧 ３ ７ ．７Ｖ の太 陽 電 池 を ２ ０枚 接
続 すると、７５４Ｖになります。
太 陽電 池 １

太陽電池２

太 陽 電池 Ｎ

発電不良モジュールがあると開放電圧が低下します。

 内部抵抗測定
ストリングに１ｋＨｚの信号を印加します。
ストリ ングの１ｋＨｚ の出 力 電 圧 Ｖを 印 加
電 流 で割 った値 が、スト リングの内 部 抵
抗です。

ストリング内に発 電不良がある太陽電池が存在すると内部抵抗が上がります。

Shin-ei Electronic Measuring Co.,Ltd8

５．製品構成

PVR-1500 取 扱 説 明 書

５－１．製品内容
本装置には以下の物が梱包されています。
不足、または破損しているものなどがあは破損しているものなどがある場合は、直ちに弊社までご連絡場合は、直ちに弊社までご連絡ください。
番号

名称

個数

説明

１

本体

１台

幅：131mm 奥行き：181mm
mm 高さ：54mm（突起部除く）２

USB ケーブル

１本

３

フック付きストラップ
ストラップ

１本

４

キャリングケース

１個

1.8m

幅：340mm 奥行き：275mm
mm 高さ：70mm（突起部除く）9５単三形アルカリ乾電池
アルカリ乾電池

４本

６

取扱説明書

１部

７

保証書

１部

Shin-ei Electronic Measuring Co.,
Co.,Ltd.

６．各部名称

PVR-1500 取 扱 説 明 書

６－１．正面測定プローブＬＣＤディスプレイ
測 定 結 果 、メ ッセ ージ 、 各 種 メ ニ ュ ーを
表 示 します。また測 定 結 果 に応 して画面の色が変わります。

ＵＳＢ接続端子ＰＣへデータの転送を行います。

Ｆ１・Ｆ２ ボタン
メニュー項 目 の選 択 、設 定 値 の入 力 な
どの操作を行います。

電源 ＯＮ／ＯＦＦ
電源のオン／オフを行います。

ストラップ取り付け
モード ボタン
画面の切り替えや設定を行います。

６－２．背面

電池収納
単三型乾電池が 4 本入っています
本入っています.。

ストラップ
フック付きのストラップです。

Shin-eiei Electronic Measuring Co.,
Co.,Ltd10７．使用方法

PVR-1500 取 扱 説 明 書

７－１．起動と停止※ 必ず太陽電池に接続する前に、装置を起動してください。
※ 必ず太陽電池との接続を切り離した後に、装置を停止してください。

７-１-１．起動
本装置を起動するには、操作パネルの「電源 ＯＮ／ＯＦＦ ボタン」を約１秒間長押しします。
」を約１秒間長押しします。
現在の日時
製 品 のシリアル番
ファームウェアバージョン番号「電源 ＯＮ／ＯＦＦ ボタン
ボタン」を離すと“電池残量”画面が表示されます。
」を離すと“電池残量”画面が表示されます。

“電池残量”画面が３秒間表示されると下記の“ＭＡＩＮ”画面（測定待機）が表示されます。
前回・前々回の測定データ（保 存 Ｎｏ．・内 部 抵 抗 値 ・開 放 電
保存Ｎｏ．

内部抵抗値

開放電圧

圧）を表示します。
現 在 の保 存 件 数 （****/2000）を表（示します。

測定結果が表示されます。
測 定 データ（保 存 Ｎｏ．・内 部 抵 抗値・開放電圧）を表示します。
「Ｆ２ ボタン」：ＲＥＴＲＹ（再測定）を行います。
を行います。
「Ｆ１ ボタン」： “Ｌｏｇ”画面（測定データ確認）に移行します。
に移行します。
「モード ボタン」： “ＭＥＮＵ”画面“ＭＥＮＵ”画面（測定条件設定）に移行します。
に移行します。
11

Shin-ei Electronic Measuring Co.,
Co.,Ltd.

７．使用方法

PVR-1500 取 扱 説 明 書

７-１-２．停止
本装置を停止するには、操作パネルの「電源 ＯＮ／ＯＦＦ ボタン」を約１秒間長押しします。
」を約１秒間長押しします。
また、本装置はオートパワーオフ機能を搭載しています。測定、またはボタン
また、本装置はオートパワーオフ機能を搭載しています。測定、またはボタン操作を２分間操作しなかった場合、自動で電源がオフになります。
、自動で電源がオフになります。

７－２．測定手順


測定はストリング単位で行います。



発 電中の太 陽光 発 電 システムを測定しますので、夜 間や、天候 が悪い時は測定 できません。



接続箱内全ての太陽電池とパワーコンディショナの太陽電池とパワーコンディショナの接続を切ってください。
を切ってください。
電 源ＯＮ
バッテリ残量 の確 認
測 定条 件 の確 認

測 定プローブをストリングに接 続 します。
測 定は、自 動スタートします。
測 定した内 部 抵 抗値 によって表示 が変 わります。

低 抵抗

高 抵抗

異 常抵 抗Ｅｒｒｏｒ

ストリングは正 常です。

ストリング内 に発 電
劣 化 が発 生 しているモジュールがある可能 性があります。

ストリング内 の発 電
劣 化 、もしくは断 線 している箇 所 がある可能 性があります。

プローブの接 触 不 良やノイズにより内 部
抵 抗 が測 定 できません。

再 測定 を行 う場 合は「Ｆ２ ボタン」
（ＲＥＴRY）を押 します。

自 動 測 定 データ保 存

測 定終 了測 定終 了 （電 源ＯＦＦ）

Shin-eiei Electronic Measuring Co.,
Co.,Ltd12

７．使用方法

PVR-1500 取 扱 説 明 書

７－３．測定条件の確認
３つの機能の設定を確認します。
（１）時計の確認※設定方法は１１．時計設定を参照してください。
「電源 ＯＮ／ＯＦＦ ボタン」を押している間に設定された時計を確認してください。

時計を確認してください（２）内部抵抗測定の良否判定の判定値の設定を確認します。
※設定方法は１０．判定値設定を参照してください。
工場出荷時

低抵抗（緑画面）：３００Ω未満

高抵抗（黄画面）：３００Ω以上 １０００Ω未満

異常抵抗（赤画面） ：１０００Ω以上（３）測定データのグループ（Ａ～Ｚ）の設定を確認します。
＊設定方法は９．グループ設定を参照してください。
グループの設定ができます。
接 続 箱 毎 やパワーコンディショナ毎 にグルー
プ番 号を設 定 するとわかりやすいデータになります。

グループを確認してください。

13

Shin-ei Electronic Measuring Co.,Ltd.

８．測定

PVR-1500 取 扱 説 明 書

８－１．測定準備 本体の電源をオンすると“ＭＡＩＮ”画面（測定待機）になります。
 接続箱のブレーカを切り、太陽電池とパワーコンディショナとの接続を切ります。
※接続箱内の全ての系統のブレーカを切ってください。パワーコンディショナからのノイズが回り込み測定できない可能性があります。

８－２．測定（１）測定準備が整いましたら、測定プローブをブレーカの太陽電池側に接続してください。
※プローブの赤はＰ（＋）端子、黒はＮ（－）端子に接続してください。
(２）開放電圧を認識（１０Ｖ以上）するとビープ音が”ピッ”と鳴り内部抵抗測定を開始します。
※測定した内部抵抗値により画面の色とビープ音が変って、判定します。
緑 表示 ：低抵抗
低抵抗判定値＞測定値
ビープ音 ：ピーピピッ

黄 表示 ：高抵抗
低抵抗判定値≦測定値＜異常抵抗判定値
ビープ音 ：ピッピッピッ

赤 表示 ：異常抵 抗
測定値≧異常抵抗判定値
ビープ音 ：ピーピーピー（３）測定プローブを外すと”ＭＡＩＮ“画面になります。

Shin-ei Electronic Measuring Co.,Ltd

14

８．測定

PVR-1500 取 扱 説 明 書

８－３．Ｅｒｏｒｒ表示
開放電圧が１０Ｖ以上のとき測定が開始されますが、内部抵抗測定ができなかったとき「Ｅｒｒｏｒ」表示になります。 内部抵抗値が測定できなかった原因と対応方法です。


測定時にプローブが確 実に接触されておらず接 触 不良を起こしていた場合

対応方法：
・ＲＥＴＲＹ（Ｆ２ ボタン）を押 して再 測 定 を行 ってください。
太 陽 電 池の出 力 に大 きなノイズがあり、内 部 抵 抗 値が測定できない場合

対応方法：
・パワーコンディショナからのノイズが回り込んでいる可能性があります。接続箱内のすべてブレーカを切ってノイズの原因を調査してください。
・太陽電池間の延長ケーブや接続箱への配線の引き回し、施工状態などにより太陽電池の出力に大きなノイズが発生している可能性があります

８－４．再測定（ＲＥＴＲＹ）
再度測定したい時や「Ｅｒｒｏｒ
Ｅｒｒｏｒ」表示がでたとき再測定ができます。
（１）測定信号待機画面の右下に「Ｆ２（ＲＥＴＲＹ））が表示
画面の右下に「Ｆ２（ＲＥＴＲＹ））が表示れています。
（２）「Ｆ２ （ＲＥＴＲＹ）」を押すと前回測定したデータが」を押すと前回測定したデータが削除されます。
（３）測定信号印加で再測定されます。
されます。

前回測定したデータＮｏ．Ａ００４が削除され、次に測定したデータがＡ００４として保存されます。

15

Shin-ei Electronic Measuring Co.,
Co.,Ltd.

８．測定

PVR-1500 取 扱 説 明 書

８－５．測定データ確認“ＭＡＩＮ”画 面（測定待機 ）から「Ｆ１ ボタン」を押すと“Ｌｏｇ”画面（測 定データ確認）に移行 します。

“Ｌｏｇ”画面には以下のデータが表示されます。


データＮｏ．



内部抵抗値［Ω］



開放電圧［Ｖ］「Ｆ１ ボタン」・「Ｆ２ ボタン」でページを切り替えます。

Shin-ei Electronic Measuring Co.,Ltd16

９．グループ設定

PVR-1500 取 扱 説 明 書

本装置は測 定データを最大 で２０００件保存でき、保存するデータをＡ～Ｚまでグループ分 けができます。グループ分 けは測定 現 場 や接続 箱 、ストリング等 で分 けると保 存 されたデータを検索する際に便利です。
“ＭＡＩＮ”画面（測定待機）「モード ボタン」（ＭＥＮＵ）を押すと「Ｆ１ ボタン」でグループ設定を選択し“ＭＥＮＵ”画面に移行します。

「Ｆ２ ボタン」（ＥＮＴＥＲ）で決定します。
“ＭＡＩＮ“画面（測定待機）に戻ります。

※保存時にグループ後につくＮｏ．は連番です。
※保存は各グループ９９９件可能で、合計で最大２０００件保存できます。

「Ｆ１ ボタン」・「Ｆ２ ボタン」でグループを選択します。
「モード ボタン」（ＳＥＴ）で設定してください。

17

Shin-ei Electronic Measuring Co.,Ltd.

１０．判定値設定

PVR-1500 取 扱 説 明 書

本 装 置 は測 定 した内 部 抵 抗 値 より３段 階 の判 定 補 助 機 能を搭 載 しています。内 部 抵 抗 判 定
は「低 抵 抗 」「高 抵 抗 」「異 常 抵 抗 」と判 定 し、それぞれ緑 色 ・黄 色 ・赤 色の３色 で表示 して、診断のアシストを行います。
異常抵抗

内部抵抗
非常に高い

発電異常あり高い

発電不良初期

低い

発電正常

高抵抗

低抵抗

１０－１．設定画面“ＭＡＩＮ”画面から「モード ボタン」を押すと「Ｆ１ ボタン」で選択し、“ＭＥＮＵ”画面に移行します。

「Ｆ2 ボタン」（ＥＮＴＥＲ）で決定してください。

Shin-ei Electronic Measuring Co.,Ltd18

１０．判定値設定

PVR-1500 取 扱 説 明 書

１０－２．判定値設定（１）低抵抗判定値設定

測定終了画面低抵抗を選択すると、低 抵抗判定値設定画面に移行します。

測 定 した内 部 抵 抗 値 が設 定 した判 定 値 未 満
で、画面は緑色表示（低抵抗）になります。
※工場出荷時、判 定値 は３００Ω に設定されています。
「Ｆ１ ボタン」・「Ｆ２ ボタン」で低抵抗判定値を変更し「モード ボタン」で設定してください。

19

Shin-ei Electronic Measuring Co.,Ltd.

１０．判定値設定

PVR-1500 取 扱 説 明 書（２）異常抵抗選択時

測定終了画面異常抵抗を選択すると、異常抵抗判定値設
定画面に移行します。

測 定 した内 部 抵 抗 値 が設 定 した判 定 値 以 上
で、画面は赤色表示（異常抵抗）になります。
※工場出荷 時、判定値 は１０００Ωに設定されています。
「Ｆ１ ボタン」・「Ｆ２ ボタン」で異常抵抗判定値を変更し「モード ボタン」で設定してください。

Shin-ei Electronic Measuring Co.,Ltd20

１１．時計設定

PVR-1500 取 扱 説明 書

本装置は測定したデータの日時を管理するために時計機能を搭載しています。
時計は起動画面で確認でき、時計設定画面で設定・変更できます。

１１－１．時計設定
年設定“ＭＡＩＮ”画面（測定待機）設定を変更します。
モード（ＮＥＸＴ）で次の設定になります。
「モード ボタン」（ＭＥＮＵ）を押すと月・日設定

設定画面に移行します。

設定を変更します。
モード（ＮＥＸＴ）で次の設定になります。
時・分・秒設定「Ｆ１ ボタン」で時計設定を選択し「Ｆ２ ボタン」（ＥＮＴＥＲ）で決定します。
“ＭＡＩＮ“画面に戻ります。

設定を変更します。
モード（ＮＥＸＴ）で次の設定になります。

秒の設定完了で“ＭＥＮＵ”画面に戻ります。

21

Shin-ei Electronic Measuring Co.,Ltd.

１２．保存データについて

PVR-1500 取 扱 説 明 書

測定したデータは自動的に本体内部メモリに保存されます。
本体とパソコンを付属 のＵＳＢケーブルで接続すると《ＵＳＢ大容量記 録装 置》として認 識されます。認識されたＵＳＢドライブ内にある PVR-1500_DATA.csv から必要なデータを抽出してください。


工場出荷時は PVR-1500_DATA.csv はありません。測定を行うと自動生成します。

ＵＳＢケーブル接続時は”ＰＣ 接続中”と表示され本体操作は行うことができません。

ＵＳＢケーブル接 続 時 、《ＵＳＢ大 容 量記 録 装 置》として認 識 するまで時 間 がかかる場
合があります。

１２－１．ＰＣとの接続
本体右にＵＳＢ端子があります。

付属のマイクロＵＳＢケーブルでパソコンと接続します。
ＰＣに接続します。

ＵＳＢ端子蓋をとります。２．６ｍｍのネジです。
本体の電源がＯＦＦでも、ＰＣに接続時本体画面は、下記になります。

本体の測定データは、ＵＳＢメモリのような大規模記憶装置としてＰＣ内でデータ確認できます。
「１２－４．全保存データ消去」の操 作でデータを消去すると、「バックアップ」 フォルダができます。
全消去は１０回でき、最 大 １０ファイル保存 されます。それ以 上は、古 いデータから削 除されます。
「１２－２．保存データ」のフォルダ構成を参照してください。

Shin-ei Electronic Measuring Co.,Ltd22

１２．保存データについて

PVR-1500 取 扱 説 明 書１２－２．保存データのフォルダ構成
ＵＳＢ大容量記録装置のフォルダ構成は以下となっています。
USB 大 容 量 記 録 装 置
Backup
最 大 10 ファイル
PVR-1500_DATA_20231007_01.csv(バックアップデータファイル)
PVR-1500_DATA_20231007_02.csv(バックアップデータファイル)
PVR-1500_DATA_20231009_01.csv(バックアップデータファイル)

消去日

連番

PVR-1500_DATA.csv（保 存データファイル）

Backup フォルダは全保存データ消去を実行すると自動生成されます。

Backup フォルダには全保存データ消去を実行したデータが格納されます。



格納された PVR-1500_DATA は、_消去した日付_連番.csv で格納されています。



バックアップされたファイルは最大 10 件まで保存され、１０件以上は古い順に削除されます。

23

Shin-ei Electronic Measuring Co.,Ltd.

１２．保存データについて

PVR-1500 取 扱 説 明 書

１２－３．保存データのフォーマット
保存データは測定した順に保存されています。
保存項目は横並びのカンマ区切りのｃｓｖデータ（テキストデータ）で保存されます。
保存内容は以下になります。
Ｎo．

：グループを含んだ保存Ｎｏ．



日付

：測定日



時刻

：測定時刻



判定

：Ｌｏ（低抵抗）／Ｈｉ（高抵抗）／Fault（異常抵抗）／Ｅｒｒｏｒ

Ｖｏｃ（Ｖ）：開放電圧

抵抗値（Ω）：内部抵抗値

判定値(低抵抗)：測定時に設定されていた低抵抗判定値



判定値(異常抵抗) ：測定時に設定されていた異常抵抗判定値

Shin-ei Electronic Measuring Co.,Ltd24

１２．保存データについて

PVR-1500 取 扱 説 明 書

１２－４．全保存データ消去
本装置は測定データを２０００件保存できます。
ＰＣへデータ転送後は、本体の測定データを消去してください。

“ＭＡＩＮ”画面（測定待機）“ＭＡＩＮ”画面から「モード ボタン」を押すと「Ｆ１ ボタン」で選択し、“ＭＥＮＵ”画面に移行します。

「Ｆ2 ボタン」（ＥＮＴＥＲ）で決定してください。

「Ｆ２ ボタン」（ＹＥＳ）を押します。
※全消去を行った場合、本体画面から消去したデータは閲覧することはできません。
本体とパソコンをＵＳＢケーブルで接続し、ＰＣ上で閲覧することができます。

「Ｆ２ ボタン」（ＹＥＳ）を押すと内部データは、消去されます。
「モード ボタン」を押すと消去は行われず”ＭＥＮＵ”画面に移行します。
25

Shin-ei Electronic Measuring Co.,Ltd.

１３．装置仕様

PVR-1500 取 扱 説 明 書１３－１．測定仕様
項目
測定対象種類詳細● 単結晶シリコンモジュール（バックコンタクト含む）● 多結晶シリコンモジュール● ヘテロ結合型モジュール

測定対象構成● モジュール単体● 直列接続ストリング

測定項目● 開放電圧値● 内部抵抗値

開放電圧測定範囲

DC 0.0【V】～1500.0【V】

開放電圧表示分解能

0.1【V】

開放電圧測定精度±0.5%【FS】

内部抵抗測定範囲

0【Ω】～1000【Ω】（1000【Ω】より大きいの場合“>1000Ω”と表示）

内部抵抗表示分解能

1【Ω】

内部抵抗測定精度±1%【rdg】±3【dgt】

データ保存件数2000 測定データ

データ保存形式

csv ファイル

データ出力項目「１２－３．保存データのフォーマット」に記載のとおり

通信インタフェース

USB2.0電源

単三形アルカリ乾電池 ４本

消費電力

約 1.5【W】

稼働時間

約 5 時間 ※乾電池によって異なります。

Shin-ei Electronic Measuring Co.,Ltd26

１３．装置仕様

PVR-1500 取 扱 説 明 書１３－２．一般仕様
項目

詳細

本体寸法

幅：131【mm】 奥行き：181【mm】 高さ：54【mm】（突起部除く）本体質量

約 700【g】（電池搭載時）

キャリングケース

幅：340【mm】 奥行き：275【mm】 高さ：70【mm】（突起部除く）27

動作温度範囲0【℃】～+40【℃】（結露なきこと。）

動作湿度範囲80【%】RH 以下（結露なきこと。）

保存温度範囲-10【℃】～+60【℃】（結露なきこと。）

保存湿度範囲90【%】RH 以下（結露なきこと。）

Shin-ei Electronic Measuring Co.,Ltd.

１４．トラブルシューティング

PVR-1500 取 扱 説 明 書

修理やお問い合わせのご連絡をいただく前に、下記項目をご確認ください。
下記項目に当てはまらないトラブルや、技術的質問などは、弊社までご相談ください。
症状
電源が入らない。

詳細
電池が消耗している可能性があります。
新しい電池に交換してください。
交 換 しても症 状 が発 生する場 合 は、ご購 入 先 もしくは弊 社 までご連絡ください。

測 定 を開 始 すると電 源が切電池が消耗している可能性があります。

れる。

新しい電池に交換してください。
交 換 しても症 状 が発 生する場 合 は、ご購 入 先 もしくは弊 社 までご連絡ください。

電池がすぐになくなる。

電池が正しい方向で挿入されているか確認してください。
正しい方向でも症状が発生する場合は、ご購 入 先もしくは弊 社 までご連絡ください。

パソコンとの通 信 ができな

ＵＳＢケーブルが破損していないか確認してください。

い。
本 体 の日 時 データが狂 って内蔵電池が消耗している可能性があります。

しまう。

ご購入先もしくは弊社までご連絡ください。

開放電圧が測定できない。

測定プローブが正しく接続されているか確認してください。
発電していないなど、測定状態を確認してください。

測定結果に異常が多い。

発電していないなど、測定状態を確認してください。
正しい方向でも症状が発生する場合は、ご購 入 先もしくは弊 社 までご連絡ください。

Shin-ei Electronic Measuring Co.,Ltd

28

29

Shin-ei Electronic Measuring Co.,Ltd.

Shin-ei Electronic Measuring Co.,Ltd30

新栄電子計測器株式会社
http://www.shin-ei.ne.jp〒252-0816 神奈川県藤沢市遠藤２６３６
TEL 0466-88-3030
FAX 0466-87-0627

31

Shin-ei Electronic Measuring Co.,Ltd.