medi | mate INTERCONNECT

Better interconnect solutions
for better patient care
E-mail: Info@AmphenolAlden.com | Phone: 1(508) 427-7000 | www.AmphenolAlden.com

medi | mate INTERCONNECT
MEDICAL PUSH-PULL CONNECTORSAt Amphenol Alden Products Company our goal is to help enable our OEM partner’s products deliver better patient care.
We achieve this by providing cable assemblies and connectors that are efficient, effective and reliable. For our partners
this means lower costs, greater reliability and shortened development cycles due to the unmatched value and service weprovide.
Whether your application requires disposable or reusable medical interconnect (or both), Amphenol Alden’s medi | mateseries of connectors and cable assemblies combine the familiarity of a circular push-pull interconnect with ourunparalleled commitment to quality products. Sterilizable, reliable, and robust, medi | mate is available in multiple
keying and color options making for fast and safe product identification wherever multiple interconnects are used.
Leverage our expertise in design and manufacturing to turn your connector into a finished cable assembly.

FEATURES & BENEFITS
Push-pull actuation – Familiar, instinctive & effective••••Self-latching
SecureTactile mating confirmation
Smooth low-insertion force

Versatile Product Offering•••••Standard contact configurations from two (2) to 14 poles
Cable-mount and panel-mount receptacles standard
Seven (7) standard color combinations
Six (6) standard keying options
Available as an overmolded cable assembly

Environmental Resilience•••••

Safe, recessed contacts endure repeated, high frequency mating
Impact resistant circular housings
Flex, strain, & impulse resistant cable protection
IP50 rated* (when mated)
Reprocess via steam or gas* IP64 rated optional (contact manufacturer)

APPLICATIONSElectrosurgical Devices

Patient Monitoring••••Single-use soft tissue cutting tools
Ablation needles for tumor termination

SpO2
Electrocardiography (ECG)

Electrophysiology

Therapy••••

Mapping catheters
Diagnostic catheters

Infusion pump
Gastrointestinal ablation

For additional information about medi | mate or any other product,
Phone: 1(508) 427-7000
E-mail: Info@AmphenolAlden.com please contact an Amphenol Alden Product Specialist today.

www.AmphenolAlden.com

Page 2 of 4

medi | mate INTERCONNECTSPECIFICATIONS
Material

Environmental

Mechanical•

••
•

••

•

Connector Housing:◦ Polysulfon (PSU)◦ Polyetherimide (PEI)
Molded Core Insulator:◦ Polyetheretherketon (PEEK)
Contacts:◦ Pins—Brass, 20µ” Gold over 50µ” Nickel◦ Sockets—Phosphor Bronze, 20µ” Goldover 50µ” Nickel
Flammability Rating: UL94 V-0•••

Operating Temp.:◦ PSU: -50 °C to +150 °C◦ PEI: -50 °C to +170 °C
IP-rating: IP-50 (when mated)*
Sterilizable**:◦ Autoclave◦ EtOChemical Resistance:
common or cleaning solvents

Mating cycles: 2,000
Mating Force (without contacts):◦ Mate: 8N (avg.)◦ Un-mate: 8N (avg.)◦ Unintended extraction: 90N (min.)* IP64 rated optional…contact manufacturer** Dependent on housing and over-mold materials

ELECTRICAL CHARACTERISTICS
Number ofPoles

Contact Size Pin φ(mm)Wire Gauge, max.

Test Voltage

Operating Current,
max. (A)*Mated Contact
Resistance (mΩ)

140.5

28600 V RMS

2.0

<8.5

100.5

28

850 V RMS

3.0

<8.5

90.5

28

850 V RMS

3.0

<8.580.726

1,050 V RMS

5.0

<6.5

70.7

26

1,050 V RMS

5.0<6.560.726

1,050 V RMS

6.0<6.550.9

221,050 V RMS

7.0<4.540.9221,200 V RMS

8.0

<4.52

1.3

20

1,200 V RMS

10.0<3.5*30C Heat Rise Limit

BASIC DIMENSIONSPlug with locking nutKeying
configurations

Panel-mount receptacle with sockets

Plug with nut for bend relief

0°

40°

60°

80°

170°

Cable-mount receptacle withsockets and optional bend relief

205°

For additional information about medi | mate or any other product,Phone: 1(508) 427-7000
E-mail: Info@AmphenolAlden.com please contact an Amphenol Alden Product Specialist today.

www.AmphenolAlden.com

Page 3 of 4

medi | mate INTERCONNECT
HOW TO ORDER
Configurable by design, medi | mate connectors utilize an intelligent part numbering system to streamline theproduct identification and ordering process. Simply pick the options you need and the table to the right will guide
you to the correct part number.
Having trouble? Let us help configure the right product for your application. Call or email us at +1 (508) 427-7000
or Info@AmphenolAlden.com.
PRODUCTNUMBER OF
CONTACTSMPP = medi | mateconnector

02
04
05
06
07

BODY STYLE

02

CONTACT
TYPE

COLOR
CODE

P = Plug –cable mount0 = 0⁰C = Receptacle –cable mount
1 = 40⁰
R = Receptacle –front panel mount2 = 60⁰
U = Receptacle –front or back panel mount 3 = 80⁰
Y = Disposable receptacle –cable mount
4 = 170⁰

M = Pin –solder cup
40 = GreyF = Socket –solder cup
41 = BlackS = Socket –90⁰ PCB42 = BlueP = Socket –Straight PCB 43 = Red

5 = 205⁰

45 = Yellow
46 = White08091014MPPKEYINGC0COLLETTYPE

COLLET SIZE – HOUSINGCABLE OD
MATERIAL

A = Locking nut 1 = 2.7 - 3.9 mm
B = Collet forbend relief *2 = 4.0 - 5.2 mm
0 = No collet**

3 = 5.3 - 6.5 mm
0 = no collet

0 = PSU –Grey
1 = PEI –Grey
2 = PSU –Black
3 = PEI –Black

44 = GreenF40

A

10*Bend relief sold separately** Panel-mount receptacles only

Configured part number example:
medi | mate 2-pole cable-mount receptacle, Key 0°, solder-cup socket contacts,
Grey locking nut cable collet for cable OD 3.9mm max., Grey PSU material

PRODUCTMPS = medi | matestrain relief

CABLE BUNDLE OD(mm)
1 = 2.7 – 3.9
2 = 4.0 – 5.2
3 = 5.3 – 6.5

MPS

1COLOR
40 = Grey
41 = Black
42 = Blue
43 = Red
44 = Green
45 = Yellow
46 = White
40

Configured strain relief example:
medi | mate strain relief, max. cablebundle OD 3.9mm, Grey

For additional information about medi | mate or any other product,
Phone: 1(508) 427-7000
E-mail: Info@AmphenolAlden.com please contact an Amphenol Alden Product Specialist today.
Rev. 1017

www.AmphenolAlden.com

Page 4 of 4