M100-i

M-SERIES

M100-iCOMPACTUP TO 3.1 MW RF PEAK
POWER
FULLY COMPUTERIZEDMINIMIZEDCONSUMABLESMAGNETRON MODULATOR UP TO 3.1 MW
ScandiNova's M100-i is a compact pulse Modulator
designed for system integration. The Modulator is
optimized for a wide range of different magnetrons
with an RF peak power ranging from 1.0 MW to 3.1 MW.
To facilitate system integration, the Modulator has asplit design with a Tank Unit and a Pulse Unit, and canbe placed in a very compact enclosure. The M100-i isalso available as a stand-alone and/or dual version (seeM100, M100D, and M100D-i).
All interlocks and essential diagnostics are fully
integrated, and ScandiNova's modern control system
offers a user-friendly interface. The pulse length,
amplitude and repetition rate can all be changed.
The M100-i can be delivered as a pure high-power
pulse modulator or as part of an RF Unit including
magnetron and related components. Different levels ofservice and support programmes are available.

THE M-SERIES
ScandiNova's M-series
contains a range of magnetronpulse modulators with highreliability and performance,a compact design and low
energy consumption.
Our magnetron pulse
modulators require about
one third of the space ofalternative solutions yet still
offer around 30% higher
efficiency, two operating
advantages that especiallyappeal to cancer treatment
providers.
Their high reliability andperformance also attract manyindustry applications.

Excellence in pulsed powerwww.scandinovasystems.com

M100-i

UNIT

VALUE

NOTES

Magnetron RF peak power

MW

1-3.1

Depends on choice of magnetron

Magnetron RF average power

kW

2.8

Maximum, depends on choice of magnetron

Modulator peak power

MW6.2

Maximum

Modulator average power

kW

8Maximum

Pulse voltage

kV30-52

Pulse current

A30-120

RF pulse length, topμs

0.5-5

Depends on max average power

Pulse repetition rate

Hz0-500

Depends on max average power

Modulator Voltage Stability, RMS%0.4

Verified on resistive load (see options)

Water coolingl/min818-40 °C, non-condensing

INTERFACECONNECTOR

DEFAULTOPTION

Mains power, three-phase

Terminal block (4-wire)

400±10 % VAC, 50/60 Hz

208/380/480 VAC

Mains power, single-phaseC20

230±10 % VAC, 50/60 Hz

115 VAC

Control interface

RJ45

Modbus TCP

Water cooling interface

12 mm

Legris Push-inSwagelock

Trig input

BNC

Electrical

OpticalDiagnostics

BNCPulse voltage & current signals

SIZE AND WEIGHT

UNIT

Weight, Pulse Unit

VALUEkg

Weight, Tank Unit (incl. oil)

46kg

76

For more information, visit www.scandinovasystems.com/M100-i

The Standard Modulator Includes

Services

Filament power supply

Training in handling, operationand maintenanceControl system
Control via Modbus TCP
Pulse sensors
Cable kit between Pulse Unit and Tank Unit

221 mm

HV pulse cableInternal trig generator
Factory acceptance test certificate

462 mm

402 mm

402 mmManuals

Options

Pulse Unit

266 mm

DOC-004505-03 / 2024-01-19

SPECIFICATIONSTank Unit

Contact ScandiNova for any requirements notcovered by this specification.

Installation and start-up
Service contract

Typical Magnetron Loads
BVERI

GLVAC

VE2110

VE2110A

Nisshinbo

VE2098

Pulse/RF diagnostics

M1603

GLM

Enhanced PRF Range (1000–2000 Hz)

M1466

GLM5193

Enhanced Stability (down to 0.1 %)MX7640

GLM5810

Digitizer

M1466N

GLM6090

GunportM1466T

Peaking components

E2V

VMC3109

Additional System Components

MG5193

VMC3136

Circulator & RF loads

MG6090

Directional coupler

MG6493Magnet power supply

MG7095462 mm
528 mmShipping

CPI

Waveguide windows
Magnetron

Information contained in this document is subject
to change without notice.

HEADQUARTERS

CONTACT

ScandiNova Systems AB

Tel: +46 (0)18 480 59 00

Typsnittsgatan 15

E-mail: info@scandinovasystems.com

SE-754 54 Uppsala, Sweden

www.scandinovasystems.com

Excellence in pulsed power