LINE LASERDIGEST CATALOGHow to choose line laser
There are four main points to consider when choosing line lasers.
Please check these points in below, and select the most suitable model for your workshops.

1. Visibility

1.0

Choose by laser color and visibility
There are two main types: green lasers that use wavelengths 510 ~ 532 nm,

0.8532nm

0.6510nm

0.4

and red lasers that use wavelengths 635 ~ 650 nm.
The green laser has higher visibility than the red laser, but the price of the0.2

module itself is higher, which affects the product price.

0.0635nm
650nm
350400450500550600650700750

Visibility curve of wavelength

2. Line pattern
Choose an appropriate line pattern depending on the operation
There are several patterns on line laser from simple cross lines with 1V1H line to full lines with 4V4H lines.
If the number of lines is multiple, it’s possible to work efficiently depending on job sites.
However, the number of laser modules reflects directly to the product prices.
Also, turning on unnecessary lines can waste battery power.

4V4H

3. Leveling system
Choose a leveling system that matches the site environment
There are mainly 2 types of leveling system on line laser. One is an electronic
automatic leveling system that adjusts the tilt with a motor, and the other isa magnetic brake system that corrects the pendulum with magnetic force.

Electronic automatic leveling

Pendulum leveling

The electronic automatic leveling system is less susceptible to vibrations and is suitable
for sites with lots of shaking.

4. Accuracy
Determine the quality of work
The accuracy is the most important point, but please note that the inspection standard differs depending on the manufacturer.
Myzox line lasers are listed as “± 1mm / 7.5m” and “± 1mm / 10m”. This means when projected 7.5m or 10m away (reach distance),
the line tolerance created by the inclination falls within ± 1mm. Converted to the length of the projected line (projected distance),
it will be “± 0.81mm / 10m width” at a reach of 7.5m, and “± 0.61mm / 10m width” at a reach of 10m.

10m projection distance

Projection width 10m

r=10m
10m

10m projection distanceaccuracy±1mm /10m

10m
10m

6.1m

1 mm
1 mm
tolerance

10m
3.5m6.1m

Projection width 10m
accuracy±0.61mm /10m width*Same with ±1mm /10m10m
3.5m

0.61 mm0.61 mm
tolerance

1 mm1 mmGreen Line-Laser

Green Laser SR series Debut !!
Green laser SR series comes up from Myzox.
The SR series, which has been changed from the modulation laser to the direct generated laser (Real Green Laser),has a clear line with high visibility and it is easy to see and easy to mark.
Difference between of direct type & modulation type
Direct type

Modulation type

Sharp line

Highest visibility

Wide operable temperature

Low price module
Cannot use below 0℃

Long operating timeLess visibility compared tomodulation type

Fast deterioration of module

G-440SRElectronic Automatic Leveling Real Green Line Laser

Electronic automatic leveling x 4V4H lines

IP54✓ Check4V4H✓ Check

Strong against on-site vibration & shaking

High temperature protection

Compared to the pendulum leveling method, theelectronic automatic leveling method is not affectedby vibration or shaking on building or ground.

Normally, the laser element radiates heat tothe aluminum block to reduce damage to theelement, but it may not be possible to radiate
heat sufficiently under high temperature.
The high temperature protection function isequipped with a temperature sensor, and theelectronic circuit monitors and protects the unit.
To avoid this situation, G-440SR/G-410SR
have heat protection which is auto-off feature.

Electronic automatic leveling

Pendulum leveling✓ Check
3 types of power supply
Rechargeable batteries, alkaline batteries,
AC power can be freely selected from3 types of power supply, so it is safe
even if the battery suddenly runs out.

Specifications
MODELG-440SR

Self-leveling range±2.0°

Laser source

Visible laser diode(Line:Green / Down point:Red)

Operable temperature-5℃ ~ +40℃

IP function

IP54

Wavelength

510 ~ 520nm (Line)

Line width

within 2.5mm / 5m

Power

■ AC adapter

AA battery x 4 (Alkaline/NiMH)
AC adapter■ Detector (GL-RE4)

Laser class

2M (IEC:60825-1:2014)

Operate time (all lines)

approx. 2hours■ Rod clamp (GL-RC)

H & V line accuracy±1mm / 10m

Size

L:152 × W:148 × H:223mm

Leveling system

Electronic system

Weightapprox. 1.7kg (excluding battery)Standard accessories■ Main unit
■ Manual■ Charger■ Carrying case
■ NiMH battery x 4pcs
■ 9V alkaline battery■ Shoulder strap

CODE NO.

MODELREMARKS*Elevating tripod is optional

222852G-440SR

Main unit / Detector set

G-410SR

Real Green Line Laser

High Function & Slim body

IP54

✓ Check

4V1H✓ Check

Line memory function

High temperature protection

Work can be done efficiently by memorizing thesettings when the power is turned off and eliminating
the trouble of resetting when resuming work.

The high temperature protection function isequipped with a temperature sensor, and theelectronic circuit monitors and protects the unit.

✓ CheckON

Abnormal voltage protection
ONOFF

When a high voltage power supply is used, the module
protection function works and the laser is not emitted.

MemorySpecifications
MODEL

G-410SR

Self-leveling range±1.5°

Laser source

Visible laser diode(Line:Green / Down point:Red)

Operable temperature-5℃ ~ +40℃

IP function

IP54

Wavelength

510 ~ 520nm (Line)■ Detector (GL-RE4)

Line width

within 2.5mm / 5m

PowerAA battery x 4 (Alkaline/NiMH)
AC adapter■ Rod clamp (GL-RC)

Laser class

2M (IEC:60825-1:2014)

Operate time (all lines)

approx. 2 hours

H & V line accuracy±1mm / 7.5mSize

L:116 × W:115 × H:195mm

Leveling system

Pendulum system

Weightapprox. 1.25kg (excluding battery)Standard accessories■ Main unit
■ Manual■ AC adapter■ Carrying case
■ AA battery x 4pcs
■ 9V alkaline battery■ Shoulder strap*Elevating tripod is optional

CODE NO.

MODELREMARKS

222886G-410SR

Main unit / Detector set

G-210SR / 110SR

Compact Real Green Line Laser

High Performance Compact Design

IP54✓ Check
Compact design

Leg protection cap (rubber cap)

Achieving compactness
by arranging and utilizing
the maximum internalspace.
Handy size that fits
securely in your hand.

Rubber cap prevent slipping that occur
during rotation the upper of main unit.

165mm

G-210SR : 2V1H

✓ Check

G-110SR : 1V1H
110mm

Specifications
MODEL

G-210SR / G-110SR

Self-leveling range±1.5°

Laser source

Visible laser diode(Line:Green / Down point:Red)

Operable temperature-5℃ ~ +40℃

IP function

IP54 (excluding battery box)

Wavelength

510 ~ 520nm (Line)PowerAA battery x 4

Line width

within 2.5mm / 5m

■ Detector (GL-RES)

Laser class

2M (IEC:60825-1:2014)

Operate time(all lines)

G-210SR : approx. 4 hours
G-110SR : approx. 7 hours■ Rod clamp (GL-RC)

H & V line accuracy±1mm / 7.5m

Size

L:110 × W:115 × H:165mm

Leveling system

Pendulum system

Weightapprox. 0.75kg (excluding battery)Standard accessories■ Main unit
■ Leg protection cap■ Manual■ Carrying case
■ AA battery x 4pcs
■ 9V alkaline battery*Elevating tripod is optional

CODE NO.

MODEL

REMARKS223562223563G-210SR
G-110SR

Main unit (2V1H) / Detector set
Main unit (1V1H) / Detector set

Red Line-Laser

Electronic Automatic Leveling Multi Line Laser

P-440EXElectronic Automatic Leveling x 4V4H lines

IP54Specifications

4V4H

Standard accessories■ Main unit

MODELP-440EX

Self-leveling range±2.0°

Laser source

Visible laser diode(Line/Down point:Red)

Operable temperature-10℃ ~ +40℃

IP function

IP54

Wavelength

635nm (Line)

Line width

within 1.5mm / 5m

PowerAA battery x 4 (Alkaline/NiMH)
AC adapter

Laser class

2M (IEC:60825-1:2014)

Operate time (all lines)

approx. 3 hours■ ChargerH & V line accuracy±1mm / 10m

Size

L:152 × W:148 × H:223mm■ AC adapter

Leveling system

Electronic system

Weightapprox. 1.65kg (excluding battery)■ Manual■ Carrying case
■ NiMH battery x 4pcs
■ Shoulder strap

CODE NO.

MODEL

REMARKS

220400P-440EX

Main unit

P-410EX / 110EX

Compact Line Laser

Lightweight & Compact

IP

54Specifications

P-410EX : 4V1H

P-110EX : 1V1H

Standard accessories■ Main unit
■ Carrying case

MODEL

P-410EX / P-110EX

Self-leveling range±2.0°

Laser source

Visible laser diode(Line / Down point:Red)

Operable temperature-10℃ ~ +40℃

IP function

IP54 (excluding battery box)

Wavelength

635nm (Line)PowerAA battery x 3Line width

within 1.5mm / 5m

Laser class

2M (IEC:60825-1:2014)

Operate time(all lines)

P-410EX : approx. 3.5hours
P-110EX : approx. 6hours

H & V line accuracy±1mm / 7m

Size

L:110 × W:115 × H:165mm

Leveling system

Pendulum system

Weightapprox. 0.9kg (excluding battery)■ AA battery x 3pcs
■ Leg protection cap

CODE NO.

MODEL

REMARKS■ Manual219300
219302

P-410EX
P-110EX

Main unit
Main unit

Optional Accessory

Laser Detector
Specifications

for G-440SR / G-410SR
CODE NO.

MODEL

REMARKS

220910219849221325

GL-RE4
GL-RC
GL-RE4/GL-RC

Detector
Rod clamp
Detector / Rod clamp set

CODE NO.

MODEL

REMARKS

221443219849221444

GL-RES
GL-RC
GL-RES/GL-RC

Detector
Rod clamp
Detector / Rod clmap set

1m ~ 20m

Size

H:150 x W:55 x D:28mm

Weightapprox. 150g (including battery)

MODEL

GL-RES

Operating range

1m ~ 20m

Size

H:150 x W:55 x D:28mm

Weightapprox. 150g (including battery)Specifications

for Red line lasers
CODE NO.

MODEL

REMARKS

217837217838216606

TL-RE
TL-RC
TL-RE/TL-RC

Detector
Rod clamp
Detector / Rod clamp set

MODEL

TL-RE

Operating range

1m ~ 20m

Size

H:150 x W:52 x D:26mm

Weightapprox. 140g (including battery)Specifications

for Red line lasers

56

GL-RE4

Operating range

Specifications

for G-210SR / G-110SR

IP

MODEL

CODE NO.

MODEL

REMARKS

217834217835216255

RE-20IP
RC-2
RE-20IP/RC-2

Detector
Rod clamp
Detector / Rod clamp set

MODEL

RE-20IP

Operating range

1m ~ 20m

Size

H:165 x W:55 x D:26.5mm

Weightapprox. 160g (including battery)G-440SR

MODEL

G-410SR

G-210SR

G-110SR

P-440EX

P-410EX

P-110EX

Laser color

Green (Line) / Red (down point)

Red (line / down point)

Wavelength (Line)

510nm ~ 520nm

635nm

Laser class

2M (JIS : C6802:2014)

Line width (at 5m)within 2.5mm

within 1.5mm

Down point dia. (at 1m)

approx. φ1.5mm

Vertical & Horizontal line

±1mm /10m±1mm / 7.5m

Upper cross (at 5m)±1mm /10m±1mm

Down point (at 1m)
Right angle (at 3m)
Electronic system

Pendulum system

Self-leveling range±2.0°±1.5°±1mm±1mm-Leveling system

Electronic system

IP function

Weight (excluding battery)

1.7kg

1.25kgPendulum system-10 ~ 40℃

IP54 (excluding battery box)
AA battery x 4
AC adaper±2.0°-5 ~ 40℃

Power-±0.5mm±1mm

Operable temperature

±1mm

±1mm

±1mm / 7m*1

AA battery x 4

AA battery x 4
AC adaper

AA battery x 3

0.75kg1.65kg0.9kg*1 : excluding G-210SR / G-110SR / P-410EX / P-110EX

Optional Accessory

Elevating Tripod

Leveling Adapter

LET-B

LET-C

LET-S
*Bracket for G-210SR
G-110SRP-410EXP-110EX

Specifications
MODEL

LET-B

LET-C

LET-S

Elevating range

305mm

240mm

245mm

Maximum height

1,705mm

1,370mm

1,245mm

Minimum height

605mm

525mm

410mm

WeightStandard accessory

1.34kg2.42kg0.61kgCarrying case

Carrying case

Carrying caseSpecifications
MODEL

RL-TB

Size

74 dia. x H:56mm

Weight450g

Screw sizeStandard accessory

CODE NO.

MODEL

REMARKS

213092213696215950

LET-B
LET-C
LET-S

Elevating Tripod for Line Laser (Standard)
Elevating Tripod for Line Laser (Robust type)Elevating Tripod for Line Laser (Light type)

5/8 inch male
Bracket for G-210/110SR, P-410/110EX

CODE NO.

MODEL

213010RL-TBYour Local Authorized Myzox Dealer is :