Product Specifications
Reliability Creativity
ServiceVer. 6

Gas Flow Management & Control
Turbine Meter ATZTA TBX(Built-in battery）

1/4

TBX [Capacity] [Connection type] / [Flowdirection] [Connection diameter (Code) ]Model

1. Specifications
 ModelTBX [Capacity] [Connection type] / [Flow direction] [Connection diameter (Code)]
Capacity

Connection
type

Connection
diameter(Code)

Flowdirection/Description

□30
□100
□15030 (Screw type only)
100 ((Screw type and flange type)
150 (Flange type only)□ Not indicated□F

Screw type
Flange type/□L□R□U□DLeft inlet (Left to right)
Right inlet (Right to left)
Bottom inlet (Bottom to top)
Top inlet (Top to bottom)□3□432A (Rc1-1/4)
40A (Rc1-1/2)

Only TBX100F

Only TBX30

Connection diameter
Model

TBX30

TBX100Connection diameter(Code)

34Connection
diameter

Rc1.1/4

Rc1.1/2

Rc2

TBX100F50A flange (JIS 10K)

Flow rate range
Model
Flow rate range

TBX30
4 to 30 m3/h

TBX100
TBX100F
10 to 100 m3/h

TBX150F
12.5 to 150 m3/h

Accuracy: ±1% F.S.

Maximum working pressure: 100kPa

Pressure lossModel

TBX30

TBX1000.3kPa(*)With air at a gauge pressure of 2kPa
Pressure loss

Items with "" are optional.

TBX100F

TBX150F

TBX150F
0.40kPa

Product Specifications
Reliability Creativity
ServiceVer. 6

Gas Flow Management & Control
Turbine Meter ATZTA TBX(Built-in battery）

2/4

TBX [Capacity] [Connection type] / [Flowdirection] [Connection diameter (Code) ]Model

Installation position: Horizontal and vertical
Applicable fluid: Limited to only clean and dry gases (City gas, LP gas, air, nitrogen, etc.).
Durability: 7 years (When used at room temperature with the load of the maximum flow rate of 50% maximumfolw-rate)
Use environment: -10 to +60 C, max 90%RH (No dew condensation)
Storage environment: -10 to +60 C, max 90%RH (No dew condensation)
Display:

Accumulated flow volume, instantaneous flow-rate, trip accumulated flow volume, setting values, decimal point,
and pilot are displayed on the LCD. Changeover them by using "FLOW RATE switch" and "START switch".
Display

TBX30

TBX100

Accumulated 8-digit display
flow volume 999999.99 m3
Trip6-digit displayaccumulated
9999.99 m3
flow volume
Instantaneous 3-digit display
U 99.9 m3/h
flow-rate

TBX100F

TBX150F

8-digit display
999999.99 m3
6-digit display
9999.99 m3

8-digit display
9999999.9 m3
6-digit display
99999.9 m3

4-digit display
U 999.9 m3/h

3-digit display
U 999 m3/h

Power source: Built-in lithium battery [battery life: 7 years (When used at room temperature)]
The battery is not replaceable.
Pulse output
Electrical specifications
High-density pulse output
Specifica(Synchronized with theUnit pulse output
tionsrotation of the vane wheel)
MethodOpen drain
Maximum rating
24VDC
ON current
20 mA or less
10 mA or lessON resistance50Ω or less100Ω or lessOFF resistance100Ω or more

Output unit
Model
TBX30
TBX100
TBX100F
TBX150F

Unit pulse output
Standard

High-density pulse output *(Vary according to individual difference ofthe flow measurement portion)

10 L/P (Pulse output width: 40 ms)

Approx. 110cm3/P

10 L/P (Pulse output width: 40 ms)

Approx. 250cm3/P

Approx. 470cm3/P* Duty ratio is 0.45 to 0.55 (At a constant flow rate).

100 L/P (Pulse output width: 40 ms)

Product Specifications

Ver. 6

Gas Flow Management & Control
Turbine Meter ATZTA TBX(Built-in battery）

Reliability Creativity
Service

3/4

TBX [Capacity] [Connection type] / [Flowdirection] [Connection diameter (Code) ]Model

Pulse output setting conditions
Pulse
output width

Pulse
configuration
allowed /not allowed

10000L/P(10m3/P)

40ms
120ms
40ms
120ms
40ms
120ms
40ms
120ms
40ms
120ms

○ (Allowed)
x (Not allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)

Pulse
output unit

Pulse
output width

Pulse
configuration
allowed /not allowed

40ms
120ms
40ms
120ms
40ms
120ms
40ms
120ms
40ms
120ms

x (Not allowed)
x (Not allowed)○ (Allowed)
x (Not allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)

Pulse
output unit

Model

1L/P
10L/P
TBX30

100L/P
1000L/P(1m3/P)

Model

1L/P
10L/P
TBX150F

100L/P
1000L/P(1m3/P)
10000L/P(10m3/P)

Model

Pulse
output unit

1L/P
10L/P
TBX100
TBX100F

100L/P
1000L/P(1m3/P)
10000L/P(10m3/P)

Pulse
output width

Pulse
configuration
allowed /not allowed

40ms
120ms
40ms
120ms
40ms
120ms
40ms
120ms
40ms
120ms

x (Not allowed)
x (Not allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)○ (Allowed)

Maximum extension distance: Varies according to input specifications of a remote counter.

WeightModel
WeightTBX30
0.9 kgTBX100
1.6 kgTBX100F
3.1 kgTBX150F
2.5 kgComponents
Part name

TBX30

Material or component parts
TBX100
TBX100F
Aluminum die casting

Ethylenevinyl
alcohol

Polyacetal resin

Meter casingvane wheel
MagnetBearing

Rare earth magnet
Stainless steel, PTFE resin

TBX150F
Ethylenevinyl
alcohol
FerriteProduct Specifications
Reliability Creativity
ServiceGas Flow Management & Control
Turbine Meter ATZTA TBX(Built-in battery）

Ver. 6Model

4/4

TBX [Capacity] [Connection type] / [Flowdirection] [Connection diameter (Code) ]Accessories: Instruction manual Output cable (Option) □ Attached □ Not attached4-core cableCable length: 2 m
Wire connection:
High-density pulse (-) --- Black
High-density pulse (+) ----- White
Unit pulse (-) ----------------- Blue
Unit pulse (+) ----------------- Red
Relay terminal box (4 terminals)
Items with "" represent selection items.

2. Precautions in handling
Installation environment: Avoid areas with much electromagnetic noise, corrosive atmosphere, or high humidity liable tocause dew condensation.
Since this turbine meter is designed for indoor installation, install it at a place not exposed tosplash of rainwater.
It is not compliant with the ATEX Directive (2014/34/EU). Do not use in flammable gas or otheratmospheres.
Piping conditions:
Straight pipes having a length of 10D (i.e. 10 times the pipe diameter) or greater must be
provided prior and subsequent to this turbine meter.
The specified accuracy may not be satisfied in environments where a sudden reduction in
flow-rate or pulsations of flow, etc., occurs. Therefore, it is recommended to be installed at aplace where a sudden reduction in flow-rate, pulsations of flow, etc., are little..