Spectral Sensors

New Developments for MCS 500

Spectral Sensors

New Basic Developments
recently introduced• MCS 511 NIR Family• OMK 500 H NIR• Film Control - Low Cost Thickness Measurement System• Measuring Head for Low Reflection Coating 10°/10°

Spectral Sensors

New Software Developments
soon available• MCS 500 Driver for GRAMS® Software Package• Driver to run cascaded MCS501/551
in Double Beam Mode

Spectral Sensors

Recently finished and current Projects• On-Line Measurement of Low-reflection Coatings on CRT
Company LEYBOLDInstallations in Wales( finished) and Japan(Nov. 98)• In-Line Measurement of vacuum-coated foils
Company LEYBOLDInstallation in Switzerland (Nov. 98)• In-Line Measurement and Quality Control of coated Glass
Company LEYBOLDInstallation in France ( Oct. 98)

Spectral Sensors

Recently finished and current Projects• Colour Measurement of PE-Coating on Photo Paper
Company FUJI Netherlands 10/98• Aromatics in Water
Company DOW/PSL Netherlands/Germany• Measurement of Fat, Protein, Moisture in Meat
Company Reichert/Schnell GmbH 10/98

Spectral Sensors

Recently finished and current Projects• On-Line Control of Production of „Bio-Diesel“ / NIR
LUT Germany 11 / 98
• Reflection of Lamp Bulbs ( 380 - 1700 nm)
Auer GmbH / Germany 9/98
• In-Line Measurement of sugar and alcohol in filling plants
Diessel Germany
Test

Spectral Sensors

Spectrometer System MCS 511 NIR• Polychromator MMS NIR• 950 - 1700 nm ( 6nm/diode)• 1400 - 2200 nm ( in Prep.)• 1000 - 2400 nm ( in Prep.)• Compatible with MCS500
Familiy• Cascadeablewith all MCS500

MCS 511 NIR with OMK500

Spectral Sensors
Spectrometer System MCS 511 NIR
Wavelength range

Wavelength accuracy
Wavelength Reprod.
Resolution
Absorption Range
Linear Range
Stray LightNoise950 - 1700 nm ( InGaAs, cooled)
1400 - 2200 nm ( ext. InGaAs, cooled)1000 - 2400 nm ( ext. InGaAs, cooled)< 0,6 nm
< 0,1 nm
6 nm/ diode ( 18 nm acc. Rayleigh)0 ... 3 AU
0 ...2 AU ( With Blocking Filter RG1000)< 0,1 %T measured at water bandwith blocking filter ( RG850)< 1 mAU ( 1sec. Measurement Time)Spectral Sensors
Spectrometer System MCS 511 NIR
AD-Converter

15/16 bitInterface

RS 422 ( serial, 20 Mbit/sec)

Transfer distanceAppr. 80 m ( Standard)
Appr. 2500 m ( Fiber cable)

Housing

19“ Standard incl Lamp19“ Protective housing IP65
19“ Plug-in Rack
9,5‘‘ Housing ( without lamp)

Spectral Sensors

Applications of MCS 511 NIR

Food Industry
Glass Industry
Forage ResearchPharmacie

Spectral Sensors
Spectrometer System MCS 511 NIR
Software

ASPECT PLUS
Customer-tailored Software
GRAMS ® 32
LabView Library
C - LibrarySpectral Sensors
Spectrometer System MCS 511 NIR
Accessories
Integrating Spheres
Reflection Head OMK 500
Diffuse Reflection Probes
Optical Multiplexer ( 2 .. 16 channels)

Spectral Sensors
Spectrometer System MCS 511 NIR
Accessories
SMA Optics
Cell HolderImmersion Probes

Spectral Sensors
Spectrometer System MCS 511 NIR
Reflection Head OMK 500 NIR
Wavelength Range
Geometrie
Light Source
Life Timeof lamp
Power Source

400 - 2400 nm
0°/45°(circular)
Tungsten lamp
> 5000 h
5 V/2A

Fiber Connector

ZEISS

Spectral Sensors

Measurement System for Low-Reflection Coatings

Optimized for CRT
Geometrie 10°/10°
High Attenuation of backReflection
Automatic version withintegrated Calibration Set
available

Spectral Sensors
Reflection Handy Head 10°/10°
Application

Measurement of low reflecting coatings
on glass surfaces

Geometrie

10° / 10°

Wavelength Range

360 - 2000 nm

Measuring Spot2 mmConnector

2 x ZEISSSpectral Sensors

Film Control - Thickness Measurement System

Single Layer Coatings
Range 1 - 30 um
FFT Algorithm
Integrated PC
Tungsten Light Source

Spectral Sensors
Film Control - Thickness Measurement

Thickness Range
Principle
Polychromator
Light Source
Electronic unit
Accessories

1 - 30 um ( optical thickness)
Single Layer
White light interference
MMS1
Tungsten, 5V/8W
Integrated PC
14-bit electronic unit
Y - Light guide
SMA Optic

Spectral Sensors
MultiChannel Dissolution System ZDIS-MUXMCS 551UV / CLD500
8-Channel MUXImmersion Probes

LabView® based Software