Page 40

FATIGUE RATED STAINLESS STEEL(FRS) SERIES FORCE TRANSDUCERSFATIGUE RATED STAINLESS STEEL(FRS) SERIES FORCE TRANSDUCERSFEATURES• Stainless Steel Construction• Welded Diaphragm• Optional Hermetically Sealed Connector• Guaranteed Maximum Off-Center Load and MomentError• Low Creep• Low Susceptibility to Magnetic Fields• Temperature and Barometric Pressure Compensation• 5pt Tension & Compression Calibration• Proprietary Modulus Compensated Strain Gages
Model

FRS10

FRS20

U.S. lbf.

2.5K, 5K

12.5K, 25K

Metric kN

12.5, 2550, 125Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Off-Center Load Sensitivity, %/inch
Side Load Sensitivity, %Zero Balance, % Rated Output+/-0.050+/-0.050+/-0.060+/-0.010+/-0.025+/-0.250+/-0.250+/-1.0+/-0.050+/-0.050+/-0.060+/-0.010+/-0.025+/-0.250+/-0.250+/-1.0Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ ° F
Effect on Zero, % Rated Output/ ° C+15 to +115
-10 to +45-65 to +200
-55 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-65 to +200
-55 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027

Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms

220350 + 40/-3.5350 +/-3.5PARAMETERSACCURACYTEMPERATURE

ELECTRICAL

FRS10
2.5K,5K lbf
inch

FRS10M
12.5,25 kN
mm

FRS20
12.5K, 25Klbf
inch

2.51
R 2.53
4.12
1.38
1.25

63.5
R 64.3
104.8
34.9
31.8

6.

1.29

32.8

7.
8.

3.50
22.5°
.28
8 Places
5/8 - 18
1.12 Deep
0.18
1.13
0.03
1.25
5/8 - 18.88 Deep

88.9
22.5°
7.1
8 Places
M16 x 228.5 Deep.
4.7
28.6
0.8
31.8
M16 x 222.4 Deep

3.51
88.9
R 3.50R 88.9
6.06
154.0
1.75
44.5
1.63
41.3
2.34 12.5K 59.4 50 kN
2.58 25K 65.6 125 kN
5.13
130.3
15°
15°.41
10.3
12 Places
12 Places
1 1/4 - 12M33x 2
1.50 Deep 38.1 Deep
0.29
7.5
1.75
44.5
0.03
0.8
2.25
57.2
1 1/4 - 12
M33 x 21.50 Deep 38.1 Deep

Capacity
Dimensions
1.
2.
3.
4.
5.
Note: Shown with Optional Connector and Connector Box

10.
11.
12.
13.
14.
15.

FRS20M
50,125 kNmm

OPTIONS
Additional bridges
Output adjustment to standard
Compression overload protection
Hermetically sealed connectorConnector Protectors
Integral cable

220350 + 40/-3.5350 +/-3.5
5000 @ 50
5000 @ 50 VDC
VDC
PT02E-10-6P (MS3112E-10-6P)

Insulation Resistance, Megohm
Bayonet Connector

MECHANICALModel

9.

SPECIFICATIONSCapacity

DIMENSIONS

Safe Overload Range, % Rated Capacity
Weight lb
Weight kg
Weight w/base lb
Weight w/base kg
Flexure Material

ACCESSORIESMating connector
Loading fixtures
Interconnect cable+/-300+/-300
2.9
9.1
1.3
4.1
6.5
21.5
3.0
9.8
Stainless SteelTOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Shunt calibration resistor
Instrumentation

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 41

Page 40

FATIGUE RATED STAINLESS STEEL(FRS) SERIES FORCE TRANSDUCERSFATIGUE RATED STAINLESS STEEL(FRS) SERIES FORCE TRANSDUCERSFEATURES• Stainless Steel Construction• Welded Diaphragm• Optional Hermetically Sealed Connector• Guaranteed Maximum Off-Center Load and MomentError• Low Creep• Low Susceptibility to Magnetic Fields• Temperature and Barometric Pressure Compensation• 5pt Tension & Compression Calibration• Proprietary Modulus Compensated Strain Gages
Model

FRS10

FRS20

U.S. lbf.

2.5K, 5K

12.5K, 25K

Metric kN

12.5, 2550, 125

Static Error Band, % Rated Output
Nonlinearity, % Rated Output
Hysteresis, % Rated Output
Nonrepeatability, % Rated Output
Creep, % in 20 minutes
Off-Center Load Sensitivity, %/inch
Side Load Sensitivity, %Zero Balance, % Rated Output+/-0.050+/-0.050+/-0.060+/-0.010+/-0.025+/-0.250+/-0.250+/-1.0+/-0.050+/-0.050+/-0.060+/-0.010+/-0.025+/-0.250+/-0.250+/-1.0Range, Compensated, °F
Range, Compensated, °C
Range, Operating, °F
Range, Operating, °C
Effect on Sensitivity, % Rdg/ °F
Effect on Sensitivity, % Rdg/ °C
Effect on Zero, % Rated Output/ ° F
Effect on Zero, % Rated Output/ ° C+15 to +115
-10 to +45-65 to +200
-55 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027+15 to +115
-10 to +45-65 to +200
-55 to +90+/-0.0008+/-0.0015+/-0.0015+/-0.0027

Rated Output, mV/V, Nominal
Excitation Voltage, VDC Maximum
Input Resistance, Ohms
Output Resistance, Ohms

220350 + 40/-3.5
350 +/-3.5

PARAMETERS

ACCURACYTEMPERATURE

ELECTRICAL

FRS10
2.5K,
5K lbfinch

FRS10M
12.5,
25 kN
mm

FRS20
12.5K, 25Klbf
inch

2.51
R 2.534.121.38
1.2563.5R 64.3
104.8
34.9
31.8

6.

1.29

32.8

7.
8.

3.50
22.5°.28
8 Places
5/8 - 18
1.12 Deep
0.18
1.13
0.03
1.25
5/8 - 18.88 Deep

88.9
22.5°
7.1
8 Places
M16 x 228.5 Deep.
4.7
28.6
0.8
31.8
M16 x 222.4 Deep

3.51
88.9
R 3.50R 88.9
6.06
154.0
1.75
44.5
1.63
41.3
2.34 12.5K 59.4 50 kN
2.58 25K 65.6 125 kN
5.13
130.3
15°
15°.41
10.3
12 Places
12 Places
1 1/4 - 12
M33x 21.50 Deep 38.1 Deep
0.29
7.5
1.75
44.50.03
0.8
2.25
57.2
1 1/4 - 12
M33 x 21.50 Deep 38.1 Deep

Capacity
Dimensions
1.
2.
3.
4.
5.
Note: Shown with Optional Connector and Connector Box

10.
11.
12.
13.
14.
15.

FRS20M50,
125 kNmm

OPTIONS
Additional bridges
Output adjustment to standard
Compression overload protection
Hermetically sealed connector
Connector Protectors
Integral cable

220350 + 40/-3.5
350 +/-3.5
5000 @ 50
5000 @ 50 VDC
VDC
PT02E-10-6P (MS3112E-10-6P)

Insulation Resistance, Megohm
Bayonet Connector

MECHANICAL

Model

9.

SPECIFICATIONSCapacityDIMENSIONS

Safe Overload Range, % Rated Capacity
Weight lb
Weight kg
Weight w/base lb
Weight w/base kg
Flexure Material

ACCESSORIESMating connector
Loading fixtures
Interconnect cable+/-300+/-300
2.9
9.1
1.3
4.1
6.5
21.5
3.0
9.8
Stainless SteelTOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Shunt calibration resistor
Instrumentation

TOVEY ENGINEERING, INC. • 1619 W Knudsen Drive • Phoenix, AZ 85027
Phone: (623) 434-5110 • Fax: (623) 434-5130
Email: tei_sales@toveyengineering.com • Web: www.toveyengineering.com© 2015 Tovey Engineering, Inc.

Page 41