IR-BAT SERIESHIGH-SPEED COMPACT RADIATION THERMOMETERThe IR-BA Series is a compact, fast response time,
non-contact Infrared Thermometer. In spite of it's compact
design (about the size of a deck of cards) the IR-BA
internally handles all the signal processing necessary toprovide a 4 to 20mA DC linear output over its measuring
range. The compact size allows this on-line unit to beinstalled in tight spaces and on small production equipment.
This full-featured product line provides 50ms response times
and spot sizes as small as 5mm. The IR-BA's infrared
spectral selectivity allows it to be used on application
specific process such as; glass, semiconductor processing
and measurement of thin plastics (down to 25mm thick)such as PE & PET.

 FEATURES Fast response time of 50ms
 Compact and light-weight 4 to 20mA DC linear output
 Accurate emissivity setup
 Abundant mountings & accessories

 SPECIFICATIONS FOR GENERAL PURPOSE
MODELS MODELS General Purpose Models
IR- BATSpot size & distance
1: Standard Φ40/500mm
2: Small spot/short distance Φ5/80mm
S: Small spot/long distance Φ40/1000mm
Measuring range & response time
A: 0 to 300°C, 100ms
B: 0 to 600°C, 100ms*
M: 0 to 300°C, 50ms
P: 0 to 600°C, 50ms**Only available for standard and small spot/long distance models

Cable
Blank: Standard 2m
E:
Extension cable connectable model Application Specific Models
IR- BAX
Application
G1: Glass Temperature
L1: Lamp Anneal
F1: Polyethylene Film
F2: Polyester Film
H1: Combustion Gas
Cable
Blank: Standard 2m (5m for IR-BAXH1)
E:
Extension cable connectable model Extension cable
IR- ZBRA
Cable lengeth (m)Measuring system:Broadband radiation thermometer
Measuring wavelength: 8 to 14μm
Element:
Thermopile
Resolution:
0.2°C (at 0 to 300°C)
0.5°C (at 0 to 600°C)
Repeatability:±0.2°C (at 0 to 300°C)±0.3°C (at 0 to 600°C)
Optics:GE lens
Emissivity compensation:
ε=1.99 to 0.10 (digital switch)
Analog output:
4 to 20mA DC (load resistance 280Ω or less)Working temperature: 0 to 50°C
Power supply:
12 to 24V DC
Current consumption: 60mA or less
Cable length:
Φ4.5 4-core shielded cable
Non-CE approval --- Max 200m withextension cable
CE approval --- Standard 2mMax length 30m on order (indoor use only)
IR-ZBRA is not applicable.
Casing:Aluminum die-castProtection:
IPX2Mounting:M4 screws (2 pcs) or tripod
Weight:Approx 220g
CE approval:
EMC directive EN61326+A1 Class A,EN61326+A1 Annex A1Notes) 1. Connecting cable up to 30m(IR-ZBRA is not applicable.)
2. One-by-one DC power supply unit
must be used.
Stability:±10°C under EMC test environment

 SPECIFICATIONS FOR APPLICATION SPECIFIC
MODELS (Non-CE approval)
Measuring system:
Narrow band radiation thermometer
Measuring wavelength: 4.6 to 5.2μm (IR-BXG1)
3.43μm (IR-BAXL1, IR-BAXF1)
7.6 to 8.4μm (IR-BAXF2)
4.3μm (IR-BAXH1)
Element:
Thermopile
Resolution:
0.5°C (0.2°C for IR-BAXG1)
Repeatability:
0.6°C (0.3°C for IR-BAXG1)
Optics:GE lens (Ge/Si lens for IR-BAXH1)
Cable:
Φ4.5 4-core shielded cableStandard length 2m, 5m for IR-BAXH1
Max 200m with extension cable*Other specifications are same as General Purpose Models.

: Max 200m

 CONNECTIVITY

1

PSE-412F

 SPECIFICATIONS
ModelsMeasuring range

IR-BAT1A

0 to 300°C

IR-BAT1B

0 to 600°C

IR-BAT2A

0 to 300°C

IR-BATSA

0 to 300°C

IR-BATSB

0 to 600°C

IR-BAT1M

0 to 300°C

StandardGeneralmodels

Small spot &short distance
Small spot &long distanceStandard

Response time
(95% response)

Spot size/measuring
distance

100ms

Φ40/500mm

100ms

Φ5/80mm100ms

Φ40/1000mm

50ms

Φ40/500mm

50ms

Φ5/80mm50ms

Φ40/1000mm

Accuracy rating

Measuring
wavelength

300°C or less: ±3°C
300°C or more: ±1% ofmeasured value(at ε=1 under reference
operation conditions)

8 to 14μm

IR-BAT1P

0 to 600°C

IR-BAT2M

0 to 300°C

IR-BATSM

0 to 300°C

IR-BATSP

0 to 600°C

Glass Temperature

IR-BAXG1100 to 400°C

1s

Φ20/150mm±4°C

4.6 to 5.2μm

Lamp Anneal

IR-BAXL1

400 to 1300°C

1s

Φ40/500mm

1% of measured value

Polyethylene Film

IR-BAXF1

80 to 250°C

1s (63% response)Φ40/500mm±4°C

3.43μm(half-value width120mm)

Polyester Film

IR-BAXF2

50 to 150°C

1s

Φ40/500mm±4°C

7.6 to 8.4μm

Combustion Gas

IR-BAXH1500 to 1300°C

10s

Φ100/1000mm1.5% of measured value

4.3μm

High-speedmodels

Small spot &short distance
Small spot &long distanceNote) Normal operation condition: Temperature --- 23°C±5°C, Humidity --- 35-75%RH

 RELATION BETWEEN MEASURING DIAMETER AND DISTANCE

 WIRING

2 TERMINAL DIAGRAMSINDICATOR WITH POWER SUPPLY
IR-GBHCLower terminal connections (input / power supply) MODELS
IR- GBHC
Alarm output
K: With Alarm
N: None SPECIFICATIONSUpper terminal connections (external control /analog output)

Input signal:
Display:

4 to 20mA DCRed and Green 7 segments LED
Character height 18mm
Analog output: 4 to 20mA DC isolated output
Load resistance 550Ω or less
Alarm output: Photo coupler open collector output (NPN) 3 points,
30VDC, 50mA or less (option)
Power supply to IR-BA:
12V DC 100mA
Working temperature: -5 to 50˚C
Working humidity:
35 to 85%RH (no dew condensation)
Power supply: 100 to 240V AC ±10%, 50/60HzPower consumption:
Max. 12VA at 100V AC
Max. 15VA at 240V AC
Weight:Approx. 250g

 EXTERNAL DIMENSIONS
Middle terminal connections (alarm output,option model IR-GBHCK)

Unit: mm3 EXTERNAL DIMENSIONS Radiation Thermometer
IR-BAT Protective case
General type IR-ZBCSH

Water cooling protective case IR-ZBCWH
IR-BATE

Sanitary case IR-ZBCAH Extension cableIR-ZBRA Mounting metal
IR-ZBML

Unit: mm

Specifications subject to change without notice. Printed in Japan (I) 2017. 3 Recycled Paper

32-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
PHONE: +81-3-3956-2171
FAX: +81-3-3956-0915
E-mail: inter@chino.co.jp

Website: http://www.chino.co.jp

4