huiHandy Logger

MODELMR2041-EU
SERIES

FOUR-CHANNELS HANDY LOGGER
Easy to Operate, Logging Data At Anytime
Handy logger MR2041 series is a palm size
data logger, it is capable of storing up to10000 data per each channel. Power supplyof logger is selectable from AA battery or ACadaptor, so it is usable as a portable typelogger or fixing type logger.
By using the exclusive software, recording /recalling of data and programming of logging
parameters can be executed with logger orthrough a personal computer.

 FEATURES• Input signals: thermocouple K, E, J, T,
0 to 5V; 0 to 20mA;• Simultaneous 4 channels, measurementand display.
• High/low alarms for each channel.
• Calculation function: maximum / minimum /average.
• Logging method: log mode and tag-mode.
• Data process-able through a personalcomputer.
• CE marking, IP 64 dust tight and splash
proof enclosure.
• AA battery supply.

 APPLICATION• Logging and storing data in research andlaboratories• Calorimetric measurement• Temperature monitoring for HACCP, ISO• Control industrial process• Temperature monitoring for production line
• Temperature monitoring in food andtransportation.
 LOGGING SOFTWARE (OPTION)

 ACCESSORIES
Name

Model

Note

MR9407
MR9408USB cable 1.5m
attachedRange 0 to 20mA DC
Range 0 to 5V DC

Software
Current-input adaptor
Voltage-input adaptor

 MODEL
Communication TypeUSB

Model
MR2041-EU

Logging and storing measured value.
Recalling of stored data; recalling of the set
programs; computing the measured data.

1PSE-483C

 SOFTWARE SPECIFICATION

 COMMON SPECIFICATION

Input point: 4 points
Input signals: Multi-range
T/C … K, E, J, T
DC voltage…0 to 5V (by optional adaptor)
Save file: Save each log data or tag data to the PC by
DC current…0 to 20mA (by optional adaptor)
text file or exclusive format.
Measuring range: T/C … K –200.0 to 1370.0°C
E –200.0 to 700.0°C
Write logger: Write the recording settings (20) from theJ–200.0 to 700.0°C
PC into the logger.
T –200.0 to 400.0°C
DC voltage… 0 to 5V (scale –9999 to 9999)
Data printing: Prints the read data as a table format.
DC current… 0 to 20mA (scale –9999 to 9999)
Graph display (Log mode only): Display selected Accuracy rating:
T/C…±(0.1% of measuring value+ 0.3°C) (>-100°C)
logged data as in graph format.
… ±(0.1% of measuring value +0.6°C) (<-100°C)
DC voltage…±(0.1% of measuring value+ 0.2%of range)
Working environment: 23°C±3°C
 SOFTWARE SYSTEM REQUIREMENTS
Reference junction compensation accuracy:±0.4°C (at ambient temperature: 15°C to 35°C)
Communication Cable Shape: USB cable length 1.5m,±0.7°C (at ambient temp.: -10°C to 15°C, 35°C to 50°C)
Series A.
Temperature drift: ±0.01%/°C of measuring range
OS: DOS-V, Windows XP SP3/ Vista/ 7/ 8/ 8.1 (32/64 bit). Allowable signal source resistance: 100Ω (burnoutover upper limit)
Inputresistance:
DC voltage inputs … over 1MΩ
Communication port: USB 1.0/2.0
Resistingvoltage
between channels: 400V AC
compatible communication port (COM).
Measuring cycle: 1 second / 4 channels
Recording interval: 1 second to 30 seconds (1s step)
Hard disk drive: At least 10 MB.
1 minute to 24 hours (1min. step)
Loggingdata:
Log-mode…10000
data / channel.
Free memory capacity: At least 64MB (128MB or more5500
data/ channel.
Tag-mode…
of memory recommended).
Display: Liquid crystal display with backlight.
Dot-matrix 128×64 dot (view area 41×29 mm)
Operation
key: 18-key including numeric key
 PC CONGIURATION EXAMPLE
Logging method: Log-mode …logging data per a cycle(20 points) in log-area.
Tag-mode…logging data (20 points)
then press the trigger in tag-area.
Display contents: measure value (4,2,1ch), alarm,
maximum / minimum / average value,
computation value. etc.
Other functions: Clock (year, month, day/ hour, minute,UnitPCsecond); Auto power-off; reading correction;
backlight; buzzer; residual battery; residualmemory; alarm; calculation; key lock; etc.
Communicate: USB
Battery: AA×4pieces or AC (100 to 240V AC)
Battery life: About 500 hours (at 25°C, 1min interval)
Backup battery: Lithium battery for data logging, clock
Backup battery life: Over 5 years (at 25°C)
Outside dimensions/weight: 70×159×33mm/300g
Material/color: Nylon ABS
Working environment: -10 to 50°C, 10 to 80%rh
CE-marking: EN-61326, Class B (battery operation)
Housing structure: IP64 (S)Accessory: Wall-holder, AA×4, AC adaptor (option)
Read logger: Read the log or tag (20) data and display
on the screen.

2 DIMENSIONSADAPTORVOLTAGE / CURRENT INPUT ADAPTER• Connectable
Dimensions ofSM connector(option)• Voltage input adapter: MR9408
Input range: 0 to 5V DC
Input resistant: 500kΩ
Input diameter wire:
Single wire 0.14 to 1.5 mm2Twisted wire 0.14 to 1mm2
Conversion accuracy: ±0.1%
Operation temperature accuracy: -10 to 50°C

Unit: mmadaptor: MR9407• Current input adapter: Special sensor with water-proof connector• General sensor

Model
Measuring range:
MR9401A021-60 to 260°C
MR9401A051
Response(90%):
About 5.5seconds
In 0°C to 90°C water.

T/C
K
K20mA
DCDC
Input range: 0 to 5V
Input resistant: 250Ω
Input diameter wire:
2Single wire 0.14 to 1.5 mm
2Twisted wire 0.14 to 1mm
Conversion accuracy: ±0.1%
Operation temperature accuracy: -10 to 50°C

Lead wire
2m5m

Unit: mm

• Tapering sensor
Model
Measuring range:
MR9401D021-60 to 260°C
MR9401D051
Response (90%):
About 1.2seconds MR9402D021
In 0°C to 90°C water. MR9402D051

T/CK
KT
T

Lead wire
2m5m
2m5m

Unit: mm

• Room temperature sensor, MR9402R (T/C T)
Measuring range:-20 to 60°C
Response (90%):
About 52seconds
In 20°C to 0°C air.
• Teflon covered sensor
Measuring range:-60 to 200°C
Response (90%):
About 20seconds
In 0°C to 90°C water.

Unit: mm

Model
MR9401T021

T/CK

Lead wire
2m

MR9401T051

K5m

MR9402T021

T2m

MR9402T051

T5m
Unit: mm

Specifications subject to change without notice. Printed in Japan (I) 2017.03

32-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
PHONE: +81-3-3956-2171
FAX: +81-3-3956-0915
E-mail: inter@chino.co.jp
Website: http://www.chino.co.jp

3