Turbine flow meter

FS-AR


Relay output
Analog output or Pulse output (Select when ordering)
LCD display

3-color LED*Descriptions are based on Japanese Industrial Standards (JIS)Specifications
Model
Flow range
Fluid
Maximum working pressure

FS-3AR

FS-10AR

FS-30AR0.5-3 L/min

1.5-10 L/min

5-30 L/min

Industrial water, water
0.97 MPa

Flow temperature

0-60C° (No freezing , No dewing)

Ambient temperature

0-40C° (No freezing , No dewing)

Power supply

DC 12 V ±5% or DC 24 V ±10%Current consumption
Cable

Relay output

Max. 50 mA (DC1-5V,4-20mA:Max.70mA)
AWM20276, 9 core / 0.2 mm? / 500 mm
redPower supply

black

Groundwhite

Relay output commonly usedgreen

Relay output (a)yellow

Relay output (b)brownAnalog voltage output

orangeAnalog current output

Blue

Pulse output

Gray

Spare

Max.load current : Less than DC2A
Max.applied voltage : Less than DC30V
Output mode : a or b contact output

Analog output

Voltage output : DC0-10V or DC1-5V
Current output : DC4-20mA

Pulse output

Max.load current : Less than DC100mA
Max.applied voltage : Less than DC40V
Output mode : A or B open collector
Frequency output FS-3AR : 0-43Hz
Frequency output FS-10AR : 0-93Hz
Frequency output FS-30AR : 0-121Hz

Display

LCD, LEDO ring

Chloroprene or fluoro rubber

Wetted material

POM, polyamide

Adapter size

Rc 3/8" or Rc 1/2"

Adapter material
Accuracy

Stainless steel or brass with nickel plating
FS ±2.5%

Flow direction

FreePositioning

Free

WeightProtection

Approx. 380 g
Weather resistant electronics

Order form

■ISO screw(Rc) is used.
Flow rangeFSïWettedMaterialsOutputーーAdapterMaterials

ーAdapter size(ISO)

O ringPower supplyーーー

Adapter Materials

Adapter size

O ringBStandard

How to order：FS-10S-0−J−B−3/8−F−24
Flow range

FS-S
type

FS-AR
type

FS

3S

0.5〜3／min

10S

1.5〜10／min

0Standard(Polyacetal)

J

4150°C-compatible(SUS304・316)

24

DC24V±10％

٘1/4

V12SUS

Only on
150°C-compatible
bodiesFluororubber

DC12V±5％Flow range

OutputWetted Materials

Adapter Materials

Adapter size

O ringPower supply

3AR 0.5〜3／min

0〜10VB

Standard

10AR 1.5〜10／min

J0Standard(Polyacetal)

5〜30／min

3N

B

P

SUS

Only on
150°C-compatible
bodiesFluororubber

DC12V±5％

Puls Output

T
fluorine resin

OutputWetted Materials

Adapter Materials

Adapter size

O ringPower supply0J

Standard(BsBM＋Ni plating)

0〜10VStandard(Polyacetal)

S

B

1/2

OutputWetted Materials

J

Standard(Polyacetal)

P
Pulse Output

B

150°C-compatible(SUS304・316)

Tfluorine resin

30N 5〜30／min

1/4

3/8
Standard

S

1/2

Adapter Materials

Adapter size

B

Standard

Standard(BsBM＋Ni plating)

1/2

S

1/4

3/8
٘SUS

DC24V±10％٘SUS

0.5〜3／min

10N 1.5〜10／min

24

150°C-compatible(SUS304・316)

5〜30／minFlow range

F

Chloroprenerubber

40.5〜3／min

10C 1.5〜10／min

3/8Standard(BsBM＋Ni plating)

1〜5V
4〜20mA

30C

FS

F

ChloroprenerubberT
fluorine resin

3CFS

S

1/2

4〜20mA

1FS

B

Standard(BsBM＋Ni plating)

3/8

Power supply

30S 5〜30／min

Flow range

FS-N
type

Wetted Materials

0〜10V30AR

FS-C
type

OutputOnly on
150°C-compatible
bodies

V12F

Chloroprenerubber24V

DC24V±10％

O ringPower supply

FluororubberF

Chloroprenerubber

VFluororubber

24

DC24V±10％12DC12V±5％Regarding ＜Adapter size＞

Regarding ＜Adapter size＞ Regarding ＜Adapter material/O-ring＞

The diameter type of the connection section differs
depending on the type of body material.
Please refer to the table below when filling out yourorder form.

・When selecting "B: 150°C-compatible" as the body material,
do not fill in the adapter material and O-ring fields.
・When selecting "T: Fluororesin" as the body material, do not fillin the O-ring field. We also offer products in which the adaptermaterial is fluororesin. Please contact our sales department for details.

Connection diameter
Wetted Materials
Standard type(J)150°C/Compliant(B)fluorine resin type(T)

3/8

1/2

1/4

* 1/4 diameter cannot be selected for 30L specification.

Other① Please contact our sales department for other specifications.
② Please contact our sales department for inquires regarding
date of delivery, etc.
③ Shape and specification may change without notice due toperformance improvements.

10