GoldeyeCL-032 TEC1• Camera Link InGaAs camera• VGA resolution• Compact industrial design• No fan• Simple setup by GenCP• High dynamic range

See the invisible

Short-wave infrared (SWIR) cameras with InGaAs sensortechnology
Goldeye CL-032 TEC1 with InGaAs FPA 636 × 508 runs 100.0 frames per second at 0.3 MP resolution.
Goldeye cameras are equipped with InGaAs sensor technology making them sensitive in the shortwave infrared spectrum ranging from 900 nm to 1,700 nm. Some models have extended sensitivity
in the visible spectrum down to 400 nm. All Goldeye SWIR cameras can be operated at very highframe rates and capture outstanding low-noise images. They are the perfect choice for industrial
and scientific applications beyond the visible spectrum. All Goldeye models are available witheither a Camera Link or a GigE Vision interface.
Easy software integration with Allied Vision’s Vimba Suite and compatibility to the most popular
third party image-processing libraries.
AcquireControl adds extensive image analysis functions, such as:• Pseudo color LUT with several color profiles• Auto contrast• Auto brightness• Analyze multiple regions (rectangular, circle) within the image• Real-time statistics and histogram display
The Modular Concept offers various options for lens mount, housing variants, optical filters, casedesign, and more. See the Customization and OEM Solutions webpage for additional options.

Goldeye CL-032 TEC1Page 1/6

V1.0.0, 2022-Jul-18

Specifications
Interface

Camera Link BaseResolution

636 (H) × 508 (V)Spectral range

900 nm to 1700 nm

SensorInGaAs FPA 636 × 508

Sensor type

InGaAsSensor size

No standard size

Pixel size25 µm × 25 µm

Lens mounts (available)

C-Mount

Max. frame rate at full resolution

100 fps

ADC

14 BitImage buffer (RAM)256 MByte

Cooling temperature+5 °C, +20 °C (default and calibrated), +35 °C, +50 °C, oruser-configurable

Dark current

380 keˉ/s (at +20 °C FPA temperature)

Temporal dark noise

400 eˉ (Gain0), 170 eˉ (Gain1)

Saturation capacity

1.9 Meˉ (Gain0), 39 keˉ (Gain1)

Dynamic range

73 dB (Gain0), 47 dB (Gain1)

OutputBit depth

8 - 14 Bit

Monochrome pixel formats

Tap Geometry 1X 1Y: Mono8, Mono12, Mono14; Tap Geometry 1X2 1Y: Mono8, Mono12

General purpose inputs/outputs (GPIOs)
TTL I/OsLVTTL I/Os: 1 input, 1 output

Opto-isolated I/Os

1 input, 2 outputs

RS232

115 200 Baud, 8N1 (adjustable)

Operating conditions/dimensions
Operating temperature-20 °C to +55 °C (case)Power requirements (DC)10.8 V to 30.0 V

Goldeye CL-032 TEC1Page 2/6

V1.0.0, 2022-Jul-18

Power consumption

10 W at 12 VDC

Mass

340 g (with C-Mount adapter)

Body dimensions (L × W × H in mm)

78 × 55 × 55

Regulations

CE: 2014/30/EU (EMC), 2011/65/EU, incl. amendment
2015/863/EU (RoHS); FCC Class B

Quantum efficiency

Goldeye CL-032 TEC1Page 3/6

V1.0.0, 2022-Jul-18

Features
Image control: Auto
• Auto contrast• Auto exposure

Image control: Other• Background correction• Binning• Decimation• DPC (defect pixel correction)• LUT (look up table)• Multiple ROIs (regions of interest)• NUC (non-uniformity correction)• ROI (region of interest)

Camera control• Acquisition frame rate
• Bandwidth control• Event channel• Firmware update in the field• I/O and trigger control• Image chunk data
• Stream hold
• User sets

Sensor temperature control• Temperature management by TEC• Temperature specific events• Temperature status indicator

Goldeye CL-032 TEC1Page 4/6

V1.0.0, 2022-Jul-18

Technical drawing
68636

1/4-20 UNC - 813

4.5 - 1.5
93.2

9

M4 - 6 (4x)

86.9

3678

19.1
14.8

9.6 9.6

C-Mount

M4 - 6 (4x)29.531365591.755

M4 - 6 (4x)43.9

M4 - 6 (4x)Applications
Goldeye cameras are very sensitive in the SWIR spectrum. They can be used in an extended operating
temperature range. Thanks to temperature stabilization and integrated image correction, Goldeye
cameras achieve an outstanding image quality with little noise and a high dynamic range. They are
well-suited for many typical SWIR applications in various industry branches:• Semiconductor industry: solar cell and chip inspection• Recycling industry: plastics sorting• Medical imaging, sciences: hyper- and multi-spectral imaging, microscopy, optical coherence
tomography (OCT)• Metal and glass industry: thermal imaging of hot objects (250 °C to 800 °C)• Agriculture industry: airborne remote sensing• Printing industry: banknote inspection

Goldeye CL-032 TEC1Page 5/6

V1.0.0, 2022-Jul-18

• Electronics industry: laser beam profiling• Surveillance and security: vision enhancement (for example, seeing through fog)
White Paper To learn more about typical application fields for SWIR cameras, download our White
Paper: Seeing beyond the visible – short-wave infrared (SWIR) cameras offer new application fields
in machine vision

Goldeye CL-032 TEC1Page 6/6

V1.0.0, 2022-Jul-18