www.thermotekusa.com

T257P

210 Watt Thermoelectric Chiller
The new T257P thermoelectric recirculating chiller from ThermoTek is our newest offering in our high performance lineof solid state chillers. This new peltier
chiller is the next generation design of ourindustry renowned T255P chiller and has
both pump performance and design improvements over the T255P. The T257P
provides up to 210 watts of cooling andhas a 21.75 psi variable-drive centrifugal
pump. It also has integrated Fan-Speed
Control and boasts a user-friendly touchscreen interface as well as RS-232 and USB
connectivity. Coupled with a 12% reduction in size and 20% reduction in weight,
and designed with the same ThermoTek
reliability and performance you have cometo expect, this chiller will satisfy your mostdemanding cooling applications.

210 Watts cooling capacity
 21.75 psi / 4 LPM variable drive
centrifugal pump Integrated Fan-speed control
 Weighs only 17 lbs (7.7 kg)
 Touch screen interface
 Low noise and vibration
 Integrated carry handle
 Universal power supply Leak-free QD fittings
 USB and RS-232 connectivityCOMPANY PROFILE
ThermoTek, Inc. has been a leader inthe medical device industry for morethan twenty three years. With ourinnovative designs for precision thermal management solutions and numerous patents on our technology,
we continue to push the envelopewith new and creative offerings.

SYSTEM PERFORMANCE
Thermal Performance:
Stability:
Flow Rate Availability:
Pressure Availability:
Temperature Range:
Input Voltage:
Input Power:

up to 210 Watts cooling @ 0° DT+/- 0.1°C @ >= 0.6 LPM flow at constantload & constant ambient temp
up to 4.0 LPM (open flow water)
up to 1.5 Bar (21.75 psi) max pressure-5°C to +45°C
100—240 VAC 50 / 60 Hz
600 Watts max

ThermoTek

TMCorporate Offices: 1200 Lakeside Parkway, Flower Mound, TX 75028 * Call 1-877-242-3232, or see us on the web at www.thermotekusa.com

www.thermotekusa.com

T257P

210 Watt Thermoelectric Chiller
Model #T257P-20

Cooling Capacity(at 0° Delta T)

210 watts(730 Btu/hr)

Pump Pressure (max)

1.5 bar(21.75 psi)

Pump Max Flow rate

4.0 LPM
Variable Drive Centrifugal

Pump Type(Pump Drive Settable from 30% — 100%)

Temperature Stability+/- 0.1°C

Temperature Range-5° to +45°C

Communication

RS-232 & USB

Heating Capability

Yes

Input Voltage

100-240 VAC
50/60 HzMax Input Power

600 watts

Noise Level

65 dBa* (55 dBa)**

Dimensions

12.2” x 7.125” x 13.4”

Weight17 lbs (7.7 kg)

Mating quick disconnect male fittings for use with 3/8” ID tubing included* Integrated Fan Speed Control disabled and pump drive set to 80%** Integrated Fan Speed Control active and Pump Drive set to 80%

ThermoTek

TMCorporate Offices: 1200 Lakeside Parkway, Flower Mound, TX 75028 * Call 1-877-242-3232, or see us on the web at www.thermotekusa.com

www.thermotekusa.com

T257P

210 Watt Thermoelectric Chiller
SYSTEM PERFORMANCE
Thermal Performance:
Stability:
Flow Rate Availability:
Pressure Availability:
Temperature Range:
Input Voltage:
Input Power:

T257P Typical Thermal Capacity

250 W

Thermal Performance

up to 210 Watts cooling @ 0° DT+/- 0.1°C @ >=0.6 LPM flow at constant load & constant ambient temp
up to 4.0 LPM (open flow water)
up to 1.5 Bar (21.75 psi) max pressure-5°C to +45°C
100—240 VAC 50 / 60 Hz
600 Watts max

200 W
150 W

100 W
50 W0W-20 -18 -16 -14 -12 -10 -8-6-4-202

Set Temperature minus Ambient Temp (Delta T)

T257P Pump Curves
1.6

System Pressure (Bar)

1.4
1.2

100%
Pump
Drive

1.0

0.8
30%
Pump
Drive

0.6
0.4
0.2

0.0

0.0

1.0

2.0

3.0

4.0

Process Flow (LPM)

ThermoTek

TM

Corporate Offices: 1200 Lakeside Parkway, Flower Mound, TX 75028 * Call 1-877-242-3232, or see us on the web at www.thermotekusa.com