No.CRE-52-01

PDF Chart

Graphic Recorder

K R 3D0 0 s e r ie s
KR3D00 series are advanced paperless Graphic Recordersthat oﬀers high performance and excellent operating
functions along with color LCD display.

Graphic Recorder KR3D00 creates chart paper as a PDF
ﬁle adapting to needs of chart paper. Chart paper
storage, search and sharing are also available.

KR3D00 can create 100, 180 and 250mm widthchart paper in PDF format and output in thesame size as the chart paper.

Chart paper scale is selectable.

Conventional chart recorder

KR3D00 series

Where is the chart paper ofday
monthyear??

**Chart paper**Storage (**years)

Transfers data to server and cloud automatically**Graphic recording
andPDF chart

Easy to retrieve
andaccess to detailed data.

■MODELS

KR3D□□―□□□―□□□
Measurement point/sampling rate
21: Input 12 points/1 sec.
41: Input 24 points/1 sec.
61: Input 36 points/1 sec.
81: Input 48 points/1 sec.
Communications interface
G : Ethernet+high/low order communication (RS485)(Standard)
E : Ethernet
B : Ethernet+low order communications (RS232C D-Sub 9 pin) *1*Barcode reader exuclusive specification
N : None
Alarm output, Contact input (option)
0 : None
2 : Merchanical alarm relay output (4 point ‘c’ contact)
7 : Digital input (4 points)
8 : Mechanical alarm relay output (2 points ‘c’ contact)+Digital input (2 points)
Installation type
A : Panel mounting type
T : Portable type (Grip and rubber feet attached)*1

Data is able to be extracted only in required interval.
PDF password protection function prevents falsiﬁcation.
Able to print the information linked to recorders such as
product name and lot number.
Information input by bar code and printing are also possible.
Equipped with network function (FTP, email) and others.
Analog trend recording, list printing andﬁxed-time digital printing are also possible.

Others (option)-NNN : None
-1NN : Custom graphic screen-2NN : High Accuracy Tepmerature Converter KT-M input(Com. interface G)-3NN : Custom graphic screen + High Accuracy Temperature
Converter KT-M input (Com. interface G)-N1N : Barcode reader specifications(Barcode reader/others are sold separately)(Com. interface B) *1-N2N : Barcode recipe specifications(Barcode reader/others are sold separately)(Com. interface B) *1-NNP : Past profile replay*1 Non-conformance to CE■SPECIFICATIONS
Measuring points: 12 points, 24 points, 36 points and 48 points
Input types:
Universal
DC voltage… ±13.8mV to ±50V
Thermocouple… B, R, S, K, E, J, T, N, PtRh40-Ptrh20, Nimo-Ni,WRe5-Wre26, W-WRe26, CR-AuFe, Platinel II, U, L
Resistance thermometer… Pt100, JPt100, Pt50, Pt-Co
Sampling rate:
Approximately 1sec. / All points
Internal memory: Flash memory (8MB)External memory: CF card (128MB to 8GB)
Recording cycle: 1, 2, 3, 5, 10, 15, 10, 20 s
1, 2, 3, 5, 10, 15, 20, 30, 60min
Logging data:
Measured data>>>file name (group name), time of day,month and year of recording start, tag,measured data, alarm status/types, others
Setting parameter
Rated power supply voltage:
100 to 240 V AC (universal power supply), 50/60Hz
Maximum power consumption:
60 VA
Installation method:
Panel mount
Weight:Approximately 5.6kg(Same dimensions for the unit with options)

PANEL CUTOUT AND MINIMUM CLEARANCE* The names of companies and products mentioned in this catalog are trademarks or registered trademarks of those companies.

Cautions on Safety● This product was designed and manufactured as a general industrial measuring device.
● In Installing, connecting and using this product, read the manual sufficiently and then use the product correctly.
● The written contents may be changed without advance notice because of improvement in the performance etc. Please understand it in advance.
● The contents in this catalog are as of November, 2019.

Cat.No.CXE-52-01 Nov.’19 ⒷPDF
UPL : https://www.chino.co.jp