EL3000 SERIES 100mm CHART

ANALOG RECORDER (DOT PRINTING TYPE)
EL3000 series is a dot printing type analog
recorder sized 144x144mm with 100mm widthchart.
The unit starts recording as soon as the power
supply and input are connected and it is alsoeasy to operate.
Scale plate, input range and function of therecorder can be selected for various purpose
and applications as many kinds of options areprepared.

■FEATURES■MODELS●Universal power supply
Universal power supply with voltage range of 100 to240V AC (50/60Hz) is applied.
●Linear temperature scale
Temperature scale of thermocouple and resistancethermometer input is a linear scale that is excellent inreading indication value.
●Standard 6 chart speeds
6 chart speeds (5,10,20,40,80,160mm/h) areswitchable as standard.
5 chart speed and hour/minute change are preparedas option.
●Alarm setting (common alarm) as standard
Higher and lower limit alarm can be programmed forevery point. Alarm value is easy to be programmed bypointer location.
You can check the alarm by front LED lighting.
Alarm output is prepared as option.
●CE marking
The EL recorder is conformed to the rules of safety
standards of CE.
●Unit structure and light-weightLight-weight (50% of the previous unit weight) wasrealized by easy maintenance unit structure.
●Employing removable type terminal board
Employing easy connecting removable type terminal
board.

EL3D□□−□□□
Input point
1 : 1 point
2 : 2 points
3 : 3 points
6 : 6 points
Input signals
5 : Thermocouple/DC voltage
7 : Resistance thermometer
Thermocouple with burnout/
DC voltage
Built-in voltage divider input (option)*1
Input and scale plate(option)*2
0 : Standard input+ standard scale plate
1 : Non-standard input(Including current input, and built-involtage divider) + standard scale plate
2 : Standard input + non-standard scale plate
3 : Non-standard input(Including current input, and built-involtage divider) + Non-standard scale plate
Alarm output
0 : None
1 : 2 alarm outputs
Chart speed and burnout*3
0 : Standard 6-speed + burnout disabled
1 : Standard 6-speed + up-scale burnout
2 : Standard 6-speed + down-scale burnout
A : Standard 5-speed hour / minute change+ burnout disabled
B : Standard 5-speed hour / minute change+ up-scale burnout
C : Standard 5-speed hour / minute change+ down-scale burnout*1 : Optional built-in voltage divider and thermocouple / resistance thermometerburnout input is only type "7".

*2 : Double scale is available.
Input and scale selection are needed for non-standard input
and non-standard scale plate.
*3 : Burnout on all channels is programmed together
for thermocouple / resistance thermometer input.



PSE-350B

EL3000 Series■INPUT SPECIFICATIONS■OPERATION / PROGRAMMING SPECIFICATIONS

Measurement point:Reference range and types:

Switches:

1,2,3 and 6 points
DC voltage --- ±13.8mV, ±27.6mV,±69mV, 200mV, ±500mV,±2V, ±5V
Built-in voltage divider; ±10V, ±25V, ±50V
DC current --- External installation of shunt
resistor(250Ω) is applied (option)
Thermocouples --- K, E, J, T, R, and B (option)
Resistance thermometer --- Pt100(1997)(Measured current; 1mA)
* Linear scale for thermocouple and resistancethermometer
Input designation:
Single scale (standard), double scale (option)
Accuracy rating:±0.5% of input span (except for some input
under standard operating condition)
Refer to the table of standard range andminimum width of scale for non-standard input
Indicating deadband:
0.3% of input span
Reference junction compensation accuracy:
K,E,J,T --- ±1.0°C or less (23°C±10°C)±2.0°C or less (0 to 50°C)(For internal reference junction compensation,
the errors above are added to the accuracyrating)Temperature drift:±0.02%/°C (Converted into reference ranges)
Measurement cycle:
6 seconds/point
Indicating resolution:
Approximately 1 / 2,000
Burnout (option):On thermocouple or resistance thermometer
input, disconnection of signal can be detected.
(Up-scale and down scale burnout on all
channels can be programmed.)
Burnout detection --- Voltage application
method (approximately 8V, 1mA)
Allowable signal source resistance:
Thermocouple inputs, DC voltage inputs (±5Vor less)--- 1kΩ(burnout disabled) or less
DC Voltage inputs (input more than ±5V)--- 100Ω or less
Resistance thermometer inputs --- per wire
10Ω or less (Same resistance for 3 wires)
Input resistance:
Thermocouple inputs, DC voltage inputs (±5Vor less) --- Approximately 8MΩ
DC voltage inputs (more than ±5V) --Approximately 1 MΩ
Maximum input voltage:
Thermocouple inputs, DC voltage inputs --±10V DC or less
DC voltage inputs (Voltage divider built-in) --±60V DC or less
Resistance thermometer --- ±6V DC or lessMaximum common mode voltage: 30V AC
Common mode rejection ratio: 120dB or more (50/60Hz±0.1%)
Normal mode rejection ratio:
50dB or more (50/60Hz±0.1%)

Indication:■GENERAL SPECIFICATIONS
Rated power voltage:

100 to 240V AC, 50/60Hz (Universal power supply)with power supply switch
Power consumption:
Maximum 12VA (100V AC)
Maximum 20VA (240V AC)
Environmental conditions: Reference operation condition--- Ambient temperature range: 21 to 25°C
Ambient humidity range: 45 to 65%RH
Power voltage: 100V AC ±1%
Power frequency: 50/60Hz ±0.5%
Attitude: left/right 0°, forward tilting 0°,
backward tilting 0°
Warm-up time: longer than 30 minutes
Normal operation condition--- Ambient temperature range: 0 to 50°C
Ambient humidity range: 20 to 80%RH
Power voltage: 90 to 264V ACPower frequency: 50/60Hz ±2%
Attitude: left/right 0 to 10°, forward tilting 0°,
backward tilting 0 to 20°
Transportation condition (at the packed condition
on shipment from our factory)--- Ambient temperature range: -20 to 60°C
Ambient humidity range: 5 to 90%RH (No dewcondensation)
Vibration: 10 to 60Hz, 4.9m/ S2 (0.5G) or less
Impact: 392m/S2 (40G) or less
Storage condition--- Ambient temperature range: -20 to 60°C
Ambient humidity range: 5 to 90%RH (No dewcondensation)
Insulation resistance:
Secondary terminals and protective conductorterminals --- 20MΩ or more at 500V DC
Primary terminals and protective conductor
terminals --- 20MΩ or more at 500V DC
Primary and secondary terminals --- 20MΩ
or more at 500V DC
Notes: Primary terminals --Power (L,N), Alarm terminals (mechanicalrelay)
Secondary terminals --- Measurement input
terminals
Dielectric strength:
Secondary terminals and protective conductor
terminals --- 1 minute at 500V AC
Primary terminals and protective conductor terminals--- 1 minute at 1500V AC
Primary and secondary terminals --- 1 minute at2300V ACNotes; Primary terminals --Power(L,N), Alarm terminals(mechanical relay)
Secondary terminals --- Measurement input
terminals
Case:
Door (frame) --- ABS resin, window --- glass
Case --- ABS resin
Color:
Door (frame) --- Black (equivalent to Munsell N1.5),
window --- Transparent
Case --- Black (equivalent to Munsell N1.5)
Mounting:
Panel mounting
Weight:Approximately 1.7kg
Power voltage fluctuation: Indication fluctuation 0.2% or less (conversed into
reference ranges at 90 to 264V AC)■RECORDING SPECIFICATIONS
Recording accuracy:
Recording system:
Balancing time:
Recording color:
Chart paper:

Chart speed:Chart speed accuracy:±0.5% of recording span
Inkpad dotting
Input span movement --- approximately 2seconds
1: red 2: blue 3: green 4: violet 5: purple
6:brown
Fan-fold type: total width of 114mm,
total length of 10m, effective chart width of100mm
6-speed change, 5,10,20,40,80,160mm/h(standard)±0.1% or less (It is based on the chart scale.)■INDICATING SPECIFICATIONS
Analog indication:
Scale plate:

POWER --- ON/OFF the recorder power supply
AUTO CH --- Switching automatic channels change and fixedchannel (Chart feed stops when 1 point indication modeselected)
CHART SPEED --- Selecting chart speed (Chart feed stops
when all switches are OFF)SET-RUN --- Switching alarm setup/normal operation mode--- Moves pointer for alarm setup and calibration
LED (green) --- Power ON monitor
LED (red) --- Alarm monitor (All channels or output)

Scale plate and pointerSingle scale or double scale (minimum scale
division: 80)■ALARM SPECIFICATIONS■STANDARDS

Alarm display:

CE marking:Alarm types:
Alarm programming:

Alarm deadband:
Alarm output (option):

Pointer and alarm-point seal pasted on scale.
Alarm LED lamp lightens for alarming(All channels OR output)
Higher and lower-limit alarm
Individual setting for higher and lower-limit
value(Programming percentage of input span byindicating pointer, input resolution 0.5%)
0.4% of input span
1a contact and 2 outputs (common)
Maximum contact capacity:
2A (resistive load), 0.5A (inductive load)

EMC directive, low voltage directive conformity
EN61326+A1+A2+A3, EN61010-1* Under EMC directive test condition, indication equivalent tomaximum 500mV fluctuates in case

■ MAINTENANCE
Input correction:
Memory reset:

2Zero/span correction for all channels
Initializes indication adjustment value (User maintenancearea)

■OPTION SPECIFICATIONS
Options

Input typeContents

Alarm output

DC current input
Built-in voltage
divider

Non-standard input

Non-standard
scale plate●Standard range and minimum width of scale

Alarm contact output is available
Alarm relay --- Mechanical relay 1a contact,
2 outputs (common)
Maximum contact rating --250V AC 2A, 30V DC 2A(resistive load)
250V AC 0.5A, 30V DC 0.5A (inductive load))250Ω of shunt resistor is applied to measure voltage
input
Built-in voltage divider(1/1000) measures input in therange of ±5V to ±50V (input type "7" only)
Refer to the table of standard range andprogrammable minimum width of scale
Minimum width of scale --DC voltage: 10mV DC width or more
Thermocouple:
K; 250°C width or more
E,J,T; 200°C width or more
R; 800°C width or more
Resistance thermometer: 100°C width or more

DC voltage

KE

Scale plate for non-standard input

Measures input with 2 types of scales (each scale isonly serial channel)
Function for detecting disconnection for sensor withthermocouple or resistance thermometer input.
Up-scale and down scale burnout on all channels canBurnout
be programmed(Input type "7" only), parallel operation is not possible
5-speed change, 5,10,20,40,80mm/minute, hour
Chart speed
Hour / minute change change
16m chart paper
Maximum length 15.6m

T/CJT

DC voltage

KT/CE
JT

R

RTD

ScalesChart paper Nos. Minimum scales

10mV

EM-008

0.2

M1

RTD

0 to20mV

EM-5190.5

M850mVEL42003

1M91 to5V

EL420100.05

V60 to 250℃

EL05096

5K20 to 300℃

EL05010

5K30 to 400℃

EL05009

10K40 to 600℃

EL05168

10K6Input types

0 to 800℃

EL05121

10K8

K,E,J,T

0 to 1000℃

EL05157

20

KA

0 to 1200℃

EL05116

20

KC0 to 200℃

EL05047

5E2EL05010

5E30 to 300℃

EL050105J30 to 400℃

EL05009

10J40 to 200℃

EL05047

5T2T30 to 300℃

EL050105-50 to 150℃

EL05007

5T50 to 1400℃

EL05137

20R40 to 1600℃

EL05147

20R60 to 100℃

EL05052

2310 to 150℃

EL05034

23A

0 to 200℃

EL05047

5320 to 300℃

EL050105330 to 500℃

EL050481035-20 to80℃

EL05035

238-50 to50℃

EL05006

23E-50 to 150℃

EL0500753B

35mV-200 to 200mV

100mV-500 to 500mV

250mV-2 to 2V

1V

-5 to 5V

2.5V

-10 to 10V

5V-25 to 25V

13 V-50 to 50V

25 V

410mAto 20mA

-200 to 330℃

250℃

-200 to 660℃

400℃

-200 to 1370℃

700℃-200to 200℃

200℃

-200 to 380℃

250℃-200 to 720℃

380℃

-200 to 900℃

720℃

-200 to 250℃

200℃

-200 to 500℃

300℃

-200 to 1200℃

500℃-200 to 280℃

200℃

-200 to 400℃

300℃

0to 1240℃

800℃

0 to 1760℃

1480℃

0 to 1820℃

900℃-140 to 150℃

150℃

-200 to 300℃

200℃

-200 to 650℃

400℃

K,E,J,T,R : IEC584,JIS C 1602-1995
Pt100 : IEC751,JIS C 1604-1997
Programmable minimum width of scale: DC voltage --- 10mV DC width or more
Thermocouple --- K:
250°C width or more
E,J,T: 200°C width or more
R:
800°C width or moreResistance thermometer --- 100°C width or more

0 to0 to 300℃17mV-69 to 69mV

B

Input code

0 to10mV-27.6 to 27.6mVR●Standard input and chart paper Nos.

Minimum width of scale-13.8 to 13.8mV

DC current

Double scale

Input type

Standard range●Exceptions of accuracy ratings
Measuring range

Accuracy ratings±1.0% of measuring range-200to-50℃

B

0to400℃

NoneR

0to400℃±1.0% of measuring rangeNote) The accuracy ratings are converted into the measuring range

K、E、J、T、R：IEC584. JIS C1602-1995Pt100：IEC751. JIS C1604-1997

3

■TERMINAL BOARD

No. ALM No.

1 C 2Thermocouple
and DC voltage

Alarm output terminals(option)（＋）（A）Terminals（−）（B）Terminals（B）Terminals
Resistance thermometer

Power/protective
conductor terminals

Measurement input terminals■DIMENSIONS●Panel cutout

60

138 ＋10149
137

144

57

138 ＋10153●Minimum clearance for plural installation
200
144

127.3
170

200

24

Unit : mm

Specifications subject to change without notice. Printed in Japan (I) 2014. 432-8 KUMANO-CHO,ITABASHI-KU,TOKYO 173-8632
Telephone : +81-3-3956-2171
Facsimile : +81-3-3956-0915
E-mail : inter@chino.co.jp
Website : www.chino.co.jp/"