Machine Vision Cameras

EoSens ® 2.0CXP2Engineered for highest
Performance

EoSens ® 2.0CXP2 Key Features:• 2,240 fps at 2 megapixel resolution• Up to 170,500 fps at reduced resolution• High sensitivity• Fast CoaXPress-12 interface• 80 x 80 x 53 mm small

APPLICATION EXAMPLES• Melt pool monitoring• Ball grid inspection• PIV measurements• Roboter positioning• Additive Manufacturing• Sorting• Inspection• Microscopy

Featuring a high-speed CMOS sensor, the EoSens ®
2.0CXP2 offers full HD 1,920 x 1,080 pixel resolution
at an out-standing speed of 2,240 fps. The camera
takes advan-tage of the new CoaXPress Standard
Version 2.0 transmitting data at twice the data rateof Version 1.1. The CXP-12 interface supports datatransmission of up to 4 lanes at 12.5 Gpbs.

High Sensitivity
The camera’s extreme sensitivity of 20 V/lux*s @ 550
nm delivers high-contrast images even in very low light
conditions. This allows to integrate it in applications wherelighting is the crucial point.

MIKROTRON EoSens ® 2.0CXP2Technical Data

Compact Design
Contained with an 80 x 80 x 53 mm in a fanless metal
housing and an integrated CoaXPress interface, theEoSens ® 2.0CXP2 is suited for use in a wide range ofindustrial applications.

Frame rates(More detailed specifications are available on request)

EoSens ® 2.0CXP2monochrome / color
Resolution

2 MpixActive pixels

1,920 x 1,080 px

Interface

CoaXPress® @ 4 x 12.5 Gbit/s

Resolution

Frame rate

1,920 x 1,080 px

2,240 fps

1,024 x 768 px

3,175 fps

640 x 480 px

5,020 fps

Sensor type

128 x 128 px

18,170 fps

Sensor format

1,920 x 8 px

170,575 fpsPixel size

10 x 10 μm

Max. 170,575

Sensitivity

20 V/lux*s @ 550 nm

Frame rate (8 bit)2,240 fpsSensor

LUX19HSCMOS global shutter
4/3’’

Pixel data width

10 / 8 bit

Smart Features

Dynamic range

60 dB• CoaXPress lanes switchable

Shutter time (steps)

1 µs

Shutter time (range)

2 µs – 1 s• Gamma Correction• FPN Correction

1, 2 or 4• Link speed switchable

Max. Jitter via CXP Trigger

down to 3,125 Gbps
(CoaXPress 1.1 compatible)±4 ns

Mount options

C-Mount / F-Mount

Dimensions(W x H x L w/o mount)

80 x 80 x 53 mm

Weight (C-Mount)

450 gPower consumption

11 W

Power supply

12 – 24 VDC

Camera body temperature+5 °C … +50 °C

Shock / Vibration proof
Conformity

70 g / 7 grms
CoaxPress / GenICam / CE / RoHS

Camera Body Dimensions
54,38081,881,81845,580

45,5Micro BNC 75OHM ST JACK BULKHEAD

Hirose HR 10A-10R-12Pb

C-Mount

active sensor plane

F-Mount

active sensor plane

04/2021

MIKROTRON GmbHMIKROTRON GmbH provides a full range of high-speed
imaging solutions for challenging applications in industry,engineering, science and sports. The company‘s extremeslow-motion recording solutions enable customers to optimize manufacturing processes, improve product design,
revolutionize quality management and analyze motion.

Germany
Landshuter Str. 20-22
85716 Unterschleissheim+49(0)89-726342-00
info@mikrotron.de
www.mikrotron.de

All trademarks are properties of their respective owners. MIKROTRON reserves the right of change without notice.
MIKROTRON is not liable for harm or damage incurred by information contained in this document.

North America
14032 Hermosillo Way
US-Poway, CA 92064+1-858-774-1176
steve.ferrell@mikrotron.de
www.mikrotron.de

G-Mount

active sensor plane