MADEINJAPANTOITU has the longest
history of the development
and manufacturing offetal monitors in Japan.

Reassurance available only with TOITU products
Flexible, satisfactory
support system

Parts are replaceable
if damaged

Smooth connection toour central monitor

TOITU Actocardiograph MT-610
Specifications

Fetal heart rate measurement

Display

FHR input mode

2 channels Doppler ultrasound system

MonitorTFT color liquid crystal display with a touch panel

FHR measuring

Pulse doppler with autocorrelationSize

7-inchtechniqueprocessing

Displayed information

CTG (cardiotocography), fetal/maternal parameter measurements,

FHR measuring range

50-240 bpm

Alarm function

User-selectable high and low FHRand delay time

Fetal movement measurement
FM measuring

Auto measurement by doppler ultrasoundtechniquesystem (2 channels)

Recording method

Spike recording / FM dot-print

Strain gauge

Recording method

Thermal recording by line thermal head (8 dots/mm)

Recording width

FHR: 80 mm (50–210 bpm) or 70 mm (30–240 bpm), UC: 40 mm

Printed information

Year/Month/Day/Hour/Minute, measurement mode,time stamp,
fetal heart rate, fetal movement, UC, UC cycle, manual fetal movement marks,
automatic fetal movement marks, VAS marks, zero-set marks for UC,
mismeasurement marks for FHR

Print speed

10 mm/min, 20 mm/min, and 30 mm/min (changeover),1500 mm/min (for high-speed printing)measurement technique
UC measuring range

0–100 UnitsFHR sound monitor
Doppler soundMT-610Recording unit

Uterine contraction measurement
External UC

Actocardiographdevice status, settings

Power

AC 100-240V, 50/60Hz, 47VA (momentary 66VA)

Battery operating time

Approx. 30 minutes (full charge)

Size

250 (W) × 265 (H) × 205 (D) mm

Weight4.0 kg (excluding accessories)

Always feelthe weeping voice ofthe fetus throughStandard Accessories● Recording paper, color, #0030-026

the waveform● Power supply cable● BasketBasket

LCD touch panel

Y-shaped Doppler & External UC transducer
TR-687-06

Remote marker switch (for single)
SW-527-01

Transducer belt (for single)

Remote marker switch (for twin)
SW-527-02

Multi-cart JC-175

Options

LAN cable
connection port

Tissue holder

Doppler transducer
TR-629-06● USB memory● Analog converter unit [MF-7450]● TR-30 connection cable● Lithium ion batteriesCable hook

Marker switch

Twin set[Doppler transducer, transducer belt (for Doppler),remote marker switch (for twin)]*Some displays on the touch panel in this catalog are just samples.
*The appearance and specifications may change without notice for improvements.
Multi cart JC-175

Approx.

10%

reducedweight*1*1 Compared to MT-516

Distributor:

TOITU CO., LTD.

1-5-10, EBISU-NISHI,SHIBUYA-KU, TOKYO, 150-0021 JAPAN
TEL:+81-3-3463-6381 FAX:+81-3-3463-6380
E-mail:international@toitu.co.jp
http://en.toitu.co.jp/

1639
Ver.5_2024.02

MT-610 is a Highly Considered for
Usability & Work Efficiency andSupport in the Field of Obstetric Medicine !!

More St able Mea su rement● Able to display past

● Perfectly fit the abdominal wall

waveforms by scrollingwith the hole distanceFirmly Measured andMore Comfort

Easy to attach

Color LCD

● Easy to see evenfrom a distanceNew type of transducer

Better sensitivity todetect heart rate

25%

increase!
*Compared to existing device

handle transducer(In case of sweep speed 30 mm/min)● Built-in battery helps maintain temporary power supply during power outage and maternal transport● Temporary back-up during shortage of paper & power outage

Reducedexternal noise● Mini, light and easy to● Able to select from 10/15/40 minute display● Backed-up data print out in a high speed at the time of recovery● Measured data for the last fifteen hours is stored and can be backed up to a USB memory device(sold separately)

Equipped withbuilt-in battery
andmemory functions

Smooth connection
to central monitor