DATA SHEETLP Series – Analog
LP Series - Analog is a surface mountable pressure sensor package with acompensated analog output suitable for ultra-low pressure sensing applications.
COMPANY: Merit Sensor is a leader in piezoresistive

Merit Sensor is based in Salt Lake City, Utahpressure sensing and partners with clients to create highperforming solutions for a variety of applications and industries.

SENTIUM: Merit Sensor products incorporate a proprietary
Sentium® technology developed to provide superior stability.
TECHNOLOGY: Merit Sensor utilizes a piezoresistive Wheatstonebridge in a design that anodically bonds glass to a chemically etchedsilicon diaphragm. All products are RoHS compliant.

CAPABILITIES: Merit Sensor designs, engineers, fabricates, dices,assembles, tests and sells die and packaged products from a state-ofthe-art facility near Salt Lake City, Utah

FEATURES
Pressure
Range

0.04 to 15 psi (2.5 mbar to 1 bar; 250 Pa to
100 kPa; 1 in H2O to 415 in H2O)

OutputAmplified Analog

TypeGage, Differential and AbsoluteMedia

Clean, Dry Air and Non-corrosive Gases

Packaging

Tape and Reel

Customization Supply Voltage, Temperature Calibration Range, 		Output Range, Accuracy Specification,
Update Rate, etc.

BENEFITS
Performance Enjoy best-in-class performance due to Merit’s
proprietary Sentium technology

Sales +1 801.208.4722 · Customer Service +1 801.208.4700 · Fax +1 801.208.4798 · sales@meritsensor.com · www.MeritSensor.com
1600 W. Merit Parkway · South Jordan, Utah · 84095 · USA

Cost

Save money over time with high-performing die

Security

Feel confident doing business with an experienced
company backed by a solid parent company(NASDAQ: MMSI)

SpeedGet to market quickly with creative andflexible solutions

ServiceExperience prompt, personal andprofessional support

1410 Family Part Number Configurator
1410-XXX X -XX- XXPressure
P04 = 250 Pa
P07 = 500 Pa
P15 = .15psi
P30 = .30psi
1P0 = 1.0psi
15P = 15psi
Reference
D = Differential
G = Gage
A = Absolute
Calibrated Supply
Voltage
1 = 5.0V
2 = 3.3V

Pin Type
1 = J-leadPort
1 = Dualhorizontal, facingsame direction
2 = Single Horizontal

Output Range
2 = 10 to 90 %Vs

Sales +1 801.208.4722 · Customer Service +1 801.208.4700 · Fax +1 801.208.4798 · sales@meritsensor.com · www.MeritSensor.com
1600 W. Merit Parkway · South Jordan, Utah · 84095 · USA

1410seriesDS.009

LP Series – Analog

LP Series – Analog

SPECIFICATIONSMinimum

TypicalMaximum

Units

Supply Voltage (Vs)

4.5

5

5.5

V

Supply Voltage (Vs)

3.0

3.3

3.6

VNotesDIMENSIONS FOR STANDARD OPTIONS (in millimeters)

Electrical

1.25

2

Output Current

Min Output Load Resistance

2.4mA

1.9mA

5kΩ

Operating Temperature-4085°C

Storage Temperature-55100°C

2.5Device Pinout(1)Notes:(1) @ 5V input voltage(2) Must be added at thepoint of use(3) Over 0°C to 60°C
(4) Applicable if Vs = ±5% ofcalibrated supply voltage(5) Full scale pressure

Performance
DAC Resolution

12Ratiometric Output Range (Vout)

Bit

10 to 90%Vs

Accuracy-1.5

1.5%FS

Lifetime Drift-0.5

0.5%FS
Depending on the part configuration

Analog Update Time

25ms

Depending on the part configuration

10X•

Vout – Vmin
Vmax – Vmin

7WherePpsi
PMaxPMin
Vmin
VmaxVout+ Pmin

(For Use With Non-corrosive Dry Gasses
Solder temperature: max 250 °C, 5 seconds max

CROSS SECTION FOR DIFFERENTIAL AND GAGE

655
2.1

P3 = N/C

13.4

P4 = Ground

12.1

11.4

P5 = N/C
P6 = Vout

21

P7 = N/C

2.5

P8 = N/C

1

23

1.9

4

34

10.3

SMT Landing Pad Layout

2.5

2.5
2.5

Example 1: 0.0 to 0.15 PSI Gage 0-60°C

Media Compatibility

Part: 1410-P15G-12-11
5= Measured Pressure in PSI= Maximum Pressure= Minimum Pressure= Minimum Volatage (Usually 10% Vs)= Maximum Volatage (Usually 90% Vs)= Output voltage (pin 6)

Part: 1410-P15G-12-114
3.5

Vout =2.5 V
VminCompV = 0.5 V, VmaxCompV =4.5 V

CROSS SECTION FOR ABSOLUTESoft die attach material

Calibrated Range (Pressure)

4.5

Pmin =0.0 psi, Pmax =0.15 psi

MEMS sensing element(differential pressure)
Cover/mechanical protection

6(5)Transfer Function FormulaPpsi = ( Pmax – Pmin

78.5msBurst Pressure

P2 = N/C

1.3

65X

8(3) (4)Startup Time
Proof Pressure8

P1 = Vs(2)

3.7

2.5

2.5

Ppsi = ( Pmax – Pmin •Vout – Vmin
Vmax – Vmin

PSI =(0.15–0.0) •2.5 –0.5+04.5 –0.5Voltage

Supply Current

Dimensions for reference only. Engineering drawings (with tolerance) available upon order.
Depending on calibrated supply voltage(Parameter3
2.5
2+ Pmin

1.5
10.5
0-0.01875

00.01875

0.03750.05625

PSI =.075

0.075

0.09375

0.11250.131250.15

0.16875

0.131250.15

0.16875

PSI

Integrated signalConditionioning(ASIC)Media inlet ports

Part: 1410-P15D-12-11

Example 2: -0.15 to 0.15 PSI Differential 0-60°C

Filter capacitor

5Part: 1410-P15D-12-114Pmin =-0.15 psi, Pmax =0.15 psi
Note: Power supply decoupling and output filtering included

4.0mm

Vout =3.25 VVminCompV =0.5 V, VmaxCompV =4.5 V

12.0mmVsPpsi = ( Pmax – Pmin •(Analog outSSC
Vout

32.0mm6PSI= .05625
Gnd
4Vout – Vmin
Vmax – Vmin

PSI=(0.15–(–0.15) •(PACKAGINGAND SHIPPING

16.0mm

3.5+ Pmin

3.25 –0.5+(–0.15)
4.5 –0.5Voltage

ELECTRICALElectrical connections

Calibrated Range (Pressure)

4.5

3
2.5
2
1.5
10.5
0-0.01875-0.15-0.1125-0.075-0.0375

0PSI

0.03750.1125LP Series – Analog

LP Series – Analog

SPECIFICATIONSMinimum

TypicalMaximum

Units

Supply Voltage (Vs)

4.5

5

5.5

V

Supply Voltage (Vs)

3.0

3.3

3.6

V

Notes

DIMENSIONS FOR STANDARD OPTIONS (in millimeters)

Electrical

1.25

2

Output Current

Min Output Load Resistance

2.4mA

1.9mA

5kΩ

Operating Temperature-4085°C

Storage Temperature-55100°C

2.5

Device Pinout(1)
Notes:(1) @ 5V input voltage(2) Must be added at thepoint of use(3) Over 0°C to 60°C
(4) Applicable if Vs = ±5% ofcalibrated supply voltage(5) Full scale pressure

Performance
DAC Resolution

12Ratiometric Output Range (Vout)

Bit

10 to 90%Vs

Accuracy-1.5

1.5%FSLifetime Drift-0.5

0.5%FS
Depending on the part configuration

Analog Update Time

25ms

Depending on the part configuration

10X•Vout – Vmin
Vmax – Vmin

7WherePpsi
PMax
PMin
Vmin
VmaxVout+ Pmin(For Use With Non-corrosive Dry Gasses
Solder temperature: max 250 °C, 5 seconds max

CROSS SECTION FOR DIFFERENTIAL AND GAGE

6552.1P3 = N/C

13.4

P4 = Ground

12.1

11.4

P5 = N/C
P6 = Vout

21

P7 = N/C

2.5

P8 = N/C

123

1.943410.3

SMT Landing Pad Layout

2.5

2.5
2.5

Example 1: 0.0 to 0.15 PSI Gage 0-60°C

Media Compatibility

Part: 1410-P15G-12-11
5= Measured Pressure in PSI= Maximum Pressure= Minimum Pressure= Minimum Volatage (Usually 10% Vs)= Maximum Volatage (Usually 90% Vs)= Output voltage (pin 6)

Part: 1410-P15G-12-11

4
3.5

Vout =2.5 V
VminCompV = 0.5 V, VmaxCompV =4.5 V

CROSS SECTION FOR ABSOLUTESoft die attach material

Calibrated Range (Pressure)

4.5

Pmin =0.0 psi, Pmax =0.15 psi

MEMS sensing element(differential pressure)
Cover/mechanical protection

6(5)Transfer Function Formula

Ppsi = ( Pmax – Pmin

7

8.5ms

Burst Pressure

P2 = N/C

1.365X

8(3) (4)Startup Time
Proof Pressure8

P1 = Vs(2)3.7

2.5

2.5

Ppsi = ( Pmax – Pmin •Vout – Vmin
Vmax – Vmin

PSI =(0.15–0.0) •2.5 –0.5+04.5 –0.5

Voltage

Supply Current

Dimensions for reference only. Engineering drawings (with tolerance) available upon order.
Depending on calibrated supply voltage(Parameter3
2.5
2+ Pmin

1.5
10.5
0-0.01875

00.01875

0.03750.05625

PSI =.075

0.075

0.09375

0.11250.131250.15

0.16875

0.131250.15

0.16875

PSI

Integrated signalConditionioning(ASIC)Media inlet portsPart: 1410-P15D-12-11

Example 2: -0.15 to 0.15 PSI Differential 0-60°C

Filter capacitor

5

Part: 1410-P15D-12-114Pmin =-0.15 psi, Pmax =0.15 psi
Note: Power supply decoupling and output filtering included

4.0mm

Vout =3.25 V
VminCompV =0.5 V, VmaxCompV =4.5 V

12.0mmVsPpsi = ( Pmax – Pmin •(Analog outSSC
Vout

32.0mm6PSI= .05625
Gnd
4

Vout – Vmin
Vmax – Vmin

PSI=(0.15–(–0.15) •(PACKAGINGAND SHIPPING

16.0mm

3.5+ Pmin

3.25 –0.5+(–0.15)
4.5 –0.5

Voltage

ELECTRICALElectrical connections

Calibrated Range (Pressure)

4.5

3
2.5
2
1.5
10.5
0-0.01875-0.15-0.1125-0.075-0.0375

0PSI

0.03750.1125DATA SHEETLP Series – Analog
LP Series - Analog is a surface mountable pressure sensor package with acompensated analog output suitable for ultra-low pressure sensing applications.
COMPANY: Merit Sensor is a leader in piezoresistive

Merit Sensor is based in Salt Lake City, Utahpressure sensing and partners with clients to create highperforming solutions for a variety of applications and industries.

SENTIUM: Merit Sensor products incorporate a proprietarySentium® technology developed to provide superior stability.
TECHNOLOGY: Merit Sensor utilizes a piezoresistive Wheatstonebridge in a design that anodically bonds glass to a chemically etchedsilicon diaphragm. All products are RoHS compliant.

CAPABILITIES: Merit Sensor designs, engineers, fabricates, dices,
assembles, tests and sells die and packaged products from a state-ofthe-art facility near Salt Lake City, Utah

FEATURES
Pressure
Range

0.04 to 15 psi (2.5 mbar to 1 bar; 250 Pa to100 kPa; 1 in H2O to 415 in H2O)

OutputAmplified Analog

TypeGage, Differential and AbsoluteMedia

Clean, Dry Air and Non-corrosive Gases

Packaging

Tape and Reel

Customization Supply Voltage, Temperature Calibration Range, 		Output Range, Accuracy Specification,
Update Rate, etc.

BENEFITS
Performance Enjoy best-in-class performance due to Merit’s
proprietary Sentium technology

Sales +1 801.208.4722 · Customer Service +1 801.208.4700 · Fax +1 801.208.4798 · sales@meritsensor.com · www.MeritSensor.com
1600 W. Merit Parkway · South Jordan, Utah · 84095 · USA

Cost

Save money over time with high-performing die

Security

Feel confident doing business with an experienced
company backed by a solid parent company(NASDAQ: MMSI)

Speed

Get to market quickly with creative andflexible solutions

ServiceExperience prompt, personal andprofessional support

1410 Family Part Number Configurator
1410-XXX X -XX- XX
Pressure
P04 = 250 Pa
P07 = 500 Pa
P15 = .15psi
P30 = .30psi
1P0 = 1.0psi
15P = 15psi
Reference
D = Differential
G = Gage
A = Absolute
Calibrated SupplyVoltage
1 = 5.0V
2 = 3.3V

Pin Type
1 = J-leadPort
1 = Dualhorizontal, facingsame direction
2 = Single Horizontal

Output Range
2 = 10 to 90 %Vs

Sales +1 801.208.4722 · Customer Service +1 801.208.4700 · Fax +1 801.208.4798 · sales@meritsensor.com · www.MeritSensor.com
1600 W. Merit Parkway · South Jordan, Utah · 84095 · USA

1410seriesDS.009