IR-CA SERIESHIGH-SPEED RADIATION THERMOMETERThe IR-CA Product Line of Non-Contact
Infrared Thermometers provides broad
selection of units to match yourapplications and requirements for
non-contact temperature measurement.
The product line consists of 15 different
Series grouped into General Purpose
and Application Specific models.

 General Purpose Models
Low Temperature –Long Wavelength

IR-CAB□□□

Page 2IR-CAB Series measures temperatures as low as –50ºC with an accuracy of ±0.8ºC.
IR-CAE Series measures temperature as low as 30ºC with a very fast response timeof 20 milliseconds. Because it operates at a relatively short wavelength, this series isideal for measuring low temperature, unoxidized metals.
IR-CAP Series measures temperature as low as 80ºC, with some models having
measuring spot sizes as small as 1mm at a distance of 300mm. This series is ideal
for measuring metals and measuring through quartz and glass windows.

Low Temperature –Short Wavelength

IR-CAE□□□

Low to Medium
Temperature andSmall Spot Size

IR-CAP□□□

Medium Temperature– Wide Temperature
Range

IR-CAI□□□

IR-CAI Series measures temperature as low as 200ºC, provides temperatures spansas wide as 1300ºC with ultra fast 3 millisecond response times.

Page 3High Temperature –Wide Temperature
Range

IR-CAS□□□

IR-CAS Series measures temperature as low as 500ºC, provides temperatures spansas wide as 2400ºC with ultra fast 3 millisecond response times.

Page 3

Low TemperatureHigh Speed

IR-CAK□□□

IR-CAK Series measures temperature as low as 50ºC with a very fast response timeof 1.5 milliseconds.

Page 4

World’s Widest
Temperature Range
Infrared Thermometer

IR-CAW□□□

IR-CAW Series has an ultra wide temperature range of 20 to 3500ºC in one single
unit.

Page 4

Page 2 & 3
Page 3

 Application Specific Models
IR-CAN Series is designed to measure polyester films as thin as 12.5μm. This unit
operates at a wavelength that matches the PET absorption band. Temperature
measurement can be made without affect of thickness and/or color.
IR-CAM Series is designed to measure polyethylene films as thin as 12.5μm. This
unit operates at a wavelength that matches the Carbon-Hydrogen absorption band.
Temperature measurement can be made without affect of thickness and/or color.

Page 6Polyester Film

IR-CAN□□□

Polyethylene Film

IR-CAM□□□

Measurement
Inside of Furnace

IR-CAR□□□

Glass Temperature

IR-CAG□□□

Semicon/Silicon

IR-CAT□□□

IR-CAT Series is designed to measure low temperature of Silicon wafers without
seeing through the substrate therefore eliminating the interference of heaters/blocks.

Page 6

Semicon/InGaAs

IR-CAU□□□

IR-CAU Series is designed to measure low temperature of InGaAs wafers without
seeing through the substrate therefore eliminating the interference of heaters/blocks.

Page 6

Food Industry

IR-CAFX0□

IR-CAFX0 Series is designed to measure Pasteurization temperatures (60 to
100ºC )in the food industry, with high-speed (10 milliseconds) and high accuracy.

Page 7

Hot Metal
Detector

IR-CADAC01

IR-CADAC01 Series is a HMD that detects the presence of hot metal on a production
line. An Open Collector output is turned ON when hot metal enters the opticalsensing path and exceeds the preset threshold level.

Page 7

IR-CAR Series is designed to look through hot combustion gases inside of a furnace.
Its operating wavelength also minimizes background interference from hotter furnace
walls.
IR-CAG Series is designed to measure glass temperature. This unit utilizes a
Thermoelectrically Cooled MCT IR Detector to provide a fast and stable temperature
measurement.

1

Page 6
Page 6
Page 6

PSE-469C SPECIFICATIONS
Low temperature/long wavelength

IR-CAB□□□

Measuring system: Broadband radiation thermometer
Element:
PE
Measuring wavelength: 8 to 13 μm
Measuring range: -50 to 100ºC or 20 to 1000ºC
Accuracy rating:±0.8ºC (-50 to 100ºC)±2ºC (100 to 200ºC)±0.1% of measured value (200 to 1000ºC)(at ε≒1.0 and reference operating conditions)
Repeatability:
0.2ºC or less (-50 to 100ºC)
1ºC or less (20 to 1000ºC)
Stability:
Temperature drift Lower than 100ºC --- 0.05ºC/ºC
100 to 700ºC --- 0.05%/ºC of measured value
Higher than 700ºC --- 0.025%/ºC of measured
value
At EMC test environment… ±15% of
measuring range
Resolution:
0.1ºC (-50 to 100ºC)
1ºC (20 to 1000ºC)
Response time (95%): 2 sec (-50 to 100ºC)
0.2 sec (20 to 1000ºC)
Optics:Fixed focus lens type
Sighting:
Laser targeting without viewfinder
Lens aperture:
15mm diameter
Power consumption:
Maximum 5VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)

Low temperature/short wavelength IR-CAE□□□
Measuring system: Narrow-band radiation thermometer
Element:
PbSe
Measuring wavelength: 4 μm
Measuring range: 30 to 200ºC
Accuracy rating:±2ºC(at ε≒1.0 and reference operating conditions)
Repeatability:
0.5ºC or lessStability:
Temperature drift 0.15ºC /ºC
At EMC test environment… ±10% of measuring
range
Resolution:
0.1ºC
Response time (95%): 0.02 sec
Optics:Fixed focus lens type
Sighting:
Laser targeting without viewfinder
Lens aperture:
15mm diameter
Power consumption: Maximum 10VA(* The reference operating condition: 23ºC /ºC±5ºC /ºC, 35 to 75%RH)
Relation between measuring distance and diameter
Unit: mm

IR-CAEGIR-CAEH
Relation between measuring distance and diameter
Measuring range: -50 to 100ºC

Unit: mm

IR-CAEJ

IR-CABG

IR-CABH

ModelsLow
temperature/longwavelength

IR-CABJ
Measuring range: 20 to 1000ºC

Low
temperature/shortwavelength

IR-CABG□□
IR-CABH□□
IR-CABJ□□
IR-CABK□□
IR-CABL□□
IR-CABM□□
IR-CABN□□IR-CABP□□
IR-CABZ□□
IR-CAEG□□IR-CAEH□□
IR-CAEJ□□
IR-CAEZ□□

Measuring
diameter/Measuring
distance
Φ37/1000mm
Φ15/400mm
Φ8/200mm
Φ40/2000mm
Φ20/1000mm
Φ10/500mm
Φ4/200mm
Φ2/100mm
SpecialΦ37/1000mm
Φ15/400mm
Φ8/200mm
SpecialMeasuringrangeStandard
sighting-50 to 100℃

20 to1000℃

Laser
targeting(withoutview
finder)Ask CHINO
30 to 200℃

Ask CHINO□ Connection
C : Connector
T : Terminal□ External input/output (option)

IR-CABK

N : None
S : RS485
5 : 4-20mA DC input
J : Contact input (DI)K : Contact output (DO)IR-CABL
IR-CAB,IR-CAE

IR-CABM

IR-CABNIR-CABP

Unit: mm2

Low temperature/short wavelength IR-CAE□□□

Measuring system:
Narrow-band radiation thermometer
Element: PbSe
Measuring wavelength: 4 μm
Measuring range:
100 to 500ºC (distance factor 200)
Accuracy rating:±3ºC(at ε≒1.0 and reference operating conditions)
Repeatability:
1ºC or lessStability:
Temperature drift 0.15ºC /ºC
At EMC test environment… ±10% of measuring range
Resolution:
1ºC
Response time (95%): 0.02 sec
Optics:
Focusable lens type
Sighting:
Direct viewfinder
Lens aperture:
20mm diameter
Power consumption: Maximum 10VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)
Relation between measuring distance and diameter
Measuring distance: 0.5m to ∞Measuring diameter: Measuring distance/distance factor
Measuring
distance(mm)Distancefactor
200

500

1000

2000

φ2.5

φ5

φ10

High temperature IR-CAS□□□

Measuring system:
Narrow-band radiation thermometer
Element:
Si
Measuring wavelength: 0.9 μm
Measuring range:
500 to 2000ºC (distance factor 50)
600 to 3000ºC (distance factor 200 or 300)
700 to 3500ºC (with field diaphragm Φ10, distancefactor 200 or 300)
Accuracy rating:Lower than 1000ºC --- ±5ºC1000 to 1500ºC --- ±0.5% of measured value
1500 to 2000ºC --- ±1% of measured value
More than 2000ºC --- ±2% of measured value(at ε≒1.0 and reference operating conditions)
Repeatability:
0.2ºC or lessStability:
Temperature drift 0.1ºC /ºC or 0.015%/ºC ofmeasured value whichever larger.
At EMC test environment… ±1% of measuring range
Resolution:
0.5ºC
Response time (95%): 0.003 sec
Optics:
Focusable lens type
Sighting:
Direct viewfinder
Lens aperture:
20mm diameter
Power consumption: Maximum 2.4VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)
Relation between measuring distance and diameter* Same as Medium Temperature Model IR-CAI

Low to Medium temperature IR-CAP□□□

Measuring system: Narrow-band radiation thermometer
Element: PbS
Measuring wavelength: 2 μm
Measuring range:
80 to 250ºC (distance factor 50)
150 to 450ºC (distance factor 200)
200 to 800ºC (distance factor 200 or 300)
Accuracy rating:Lower than 500ºC --- ±3ºC
More than 500ºC --- ±5ºC
(at ε≒1.0 and reference operating conditions)
Repeatability:
1ºC or lessStability: Temperature drift
Lower than 500ºC --- 0.15ºC /ºC
Higher than 500ºC --- 0.25%/ºC
At EMC test environment… ±10% of
measuring range
Resolution: 1ºC
Response time (95%): 0.02 sec
Optics:
Focusable lens type
Sighting:
Direct viewfinder
Lens aperture:
20mm diameter
Power consumption: Maximum 10VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)
Relation between measuring distance and diameter
Measuring distance: 0.5m to ∞Measuring diameter: Measuring distance/distance factor

Low
temperature/shortwavelength
Low to medium
temperature

Mediumtemperature

High temperature

Measuring range

IR-CAE2□□□200

100 to 500℃

IR-CAP0□□□50

IR-CAP2□□□

200

IR-CAP3□□□
IR-CAI0□□□
IR-CAI2□□□
IR-CAI3□□□
IR-CAI7□□□

300
50
200
300

80 to 250℃
150 to 450℃
or 200 to 800℃
200 to 800℃
200 to 1000℃

IR-CAS0□□□
IR-CAS2□□□
IR-CAS3□□□
IR-CAS7□□□
IR-CAS8□□□300 to 1600℃

with field diaphragm Φ10, 200
withfield diaphragm Φ10, 300

50
200
300

400 to 2000℃
500 to 2000℃
600 to 3000℃

with field diaphragm Φ10, 200
withfield diaphragm Φ10, 300

700 to 3500℃□ Connection
C : Connector
T : Terminal□ External input/output (option)

500

1000

2000

φ10
φ2.5
φ1.7

φ20
φ5
φ3.4

φ40
φ10
φ6.7

N : None
S : RS485
5 : 4-20mA DC input
J : Contact input (DI)K : Contact output (DO)□ Sighting

Medium temperature IR-CAI□□□

Measuring system:
Narrow-band radiation thermometer
Element: InGaAs
Measuring wavelength: 1.55 μm
Measuring range:
200 to 1000ºC (distance factor 50)
300 to 1600ºC (distance factor 200 or 300)
400 to 2000ºC (with field diaphragm Φ10, distancefactor 200 or 300)
Accuracy rating:Lower than 1000ºC --- ±5ºC1000 to 1500ºC --- ±0.5% of measured value
1500 to 2000ºC --- ±1% of measured value
More than 2000ºC --- ±2% of measured value(at ε≒1.0 and reference operating conditions)
Repeatability:
0.2ºC or lessTemperature drift 0.1ºC /ºC or 0.015%/ ºC ofmeasured value whichever larger.
At EMC test environment… ±1% of measuring range
Resolution:
0.5ºC
Response time (95%): 0.003 sec
Optics:
Focusable lens type
Sighting:
Direct viewfinder
Lens aperture:
20mm diameter
Power consumption: Maximum 2.4VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)
Relation between measuring distance and diameter
Measuring distance: 0.5m to ∞Measuring diameter: Measuring distance/distance factor
Measuring
distance(mm)Distancefactor50
200
300

Distance factor

IR-CAI8□□□

Measuring
distance(mm)Distancefactor50200
300

Models

500

1000

2000

φ10
φ2.5
φ1.7

φ20
φ5
φ3.4

φ40
φ10
φ6.7

(With field diaphragm Φ10)
Measuring
distance(mm)Distancefactor
200
300

500

1000

2000

φ2.5
φ1.7

φ5
φ3.4

φ10
φ6.7

3

Blank : With view finder (standard)
3 : Built-in 300mm close-up lens (option)(190-300mm measuring distance)
6 : Built-in 600mm close-up lens (option)(270-600mm measuring distance)
L : Laser targeting (option)*without view finder

Unit: mm

Widest temperature IR-CAW□□□

Short length/High speed IR-CAK□□□

Measuring system: Broadband/Narrow-band radiation thermometer
Element:
TP/InGaAs/Si
Measuring wavelength: 8-13/1.55/0.9 μm
Measuring range: 20 to 3000ºC
Accuracy rating:Lower than 1000ºC --- ±5ºC1000 to 1500ºC --- ±0.5% of measured value
1500 to 2000ºC --- ±1% of measured value
More than 2000ºC --- ±2% of measured value(at ε≒1.0 and reference operating conditions)
Repeatability:
1ºC or lessStability:
Temperature drift
Lower than 1000ºC --- 0.2ºC /ºC
Higher than 1000ºC --- 0.02%/ºC of measured
value
At EMC test environment… ±1% of measuring
range
Resolution:
1ºC
Response time (95%): 0.1 sec
Optics:Fixed focus lens type
Sighting:
Direct viewfinder
Lens aperture:
30mm diameter
Power consumption: Maximum 2.4VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)

Measuring system: Narrow-band radiation thermometer
Element: PbSe (cooling type)Measuring wavelength: 4 μm
Measuring range: 50 to 400 ºC (displays from 0 ºC but accuracy
rating is guaranteed is from 50 ºC)
Accuracy rating:±3ºC(at ε≒1.0 and reference operating conditions)
Repeatability:
1.0ºC or lessStability:
Temperature drift 0.15 ºC/ ºC
Resolution:
1ºC
Response time (95%):
0.0015 sec
Emissivity ratio setting:
1.9999 to 0.050
Optics:Fixed focus lens type
Sighting:
Laser targeting without viewfinder
Lens aperture:
30mm diameter
Power consumption: Maximum 12VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)

Relation between measuring distance and diameter
IR-CAKL□□

IR-CAKM□□

Relation between measuring distance and diameter

Unit: mm
ModelsMeasuring
Measuring
diameter/Measu
rangering distance

IR-CAWV□□□

φ29/1000mm

20 to 3000℃

IR-CAWZ□□□

SpecialAsk CHINO

IR-CAKN□□

Unit: mmwithview
finder□ Connection
C : Connector
T : Terminal□ External input/output (option)
N : None
S : RS485
5 : 4-20mA DC input
J : Contact input (DI)K : Contact output (DO)□ Sighting
Blank: With view finder (standard)
L: Laser targeting (option)*without view finder

IR-CAKL□□□

Measuring
diameter/Measuring
distance
φ20/1000mm

IR-CAKM□□□

φ10/500mm

ModelsIR-CAKN□□□

φ4/200mm

IR-CAKZ□□□

Optional□ Connection
C : Connector
T : Terminal

IR-CAW□ External input/output (option)
N : None
S : RS485

Unit: mm
Unit: mm4

 CONNECTIVITY

 COMMON SPECIFICATIONS
Display
Emissivity setting
Signalmodulation
Computation
function

Analog output

Parameter
setting key
Self-diagnostic
Working
temperature

Temperature & parameter --- 4-digit LCD
Unit --- ºC or ºF (Key switchable)
1.9999 to 0.050
DELAY --- First-order lag (Time constant: 0.0 to 99.9 sec
with 0.1 sec increment or 0.00 to 9.99 sec with 0.01 sec
increment) Real signal must be set at 0 sec.
PEAK --- Peak tracing (attenuation factor 0, 2, 5, 10ºC
/sec selectable) Peak hold must be set at 0sec.
ZERO/SPAN
adjustment,
automatic
emissivity
computation, output correction
4 to 20mA DC isolated output
Load resistance: Less than 500ΩAccuracy rating: ±0.2% of output range
Resolution: 0.04% of output range
Scaling: Programmable in measuring range
Dummy output: Programmable within 0 to 100% of
analog output
Operator mode --- Emissivity, signal modulation, alarm,othersEngineering mode --- Measuring unit, output scaling,
ZERO/SPAN, reference temperature for automatic
emissivity computation, output correction and otheroptions.
Thermometer temperature abnormal, parameter error Remote setup system* Only IR-CAI/CAS/CAQ/CAW can be connected.
Separate DC power supply is required for other models.

 Data Acquisition Software (option)

0 to 50ºC

This PC software records measuring data for the IR-CA.

24V DC (allowable voltage fluctuation 22 to 28V DC)
Recommended power supply unit
●IR-ZFEP (S82K-01524)●IR-GZ●IR-GCPower supply

Connections
CasingWeightCE marking(connector
connection only) ModelIR-VXC1□
J:Japanese
E:English

Terminal or connector
Aluminum
Approx 1.3Kg
EMC directive EN61326+A1 Emission class A
Immunity Annex A* The product complies when in use of exclusive power
supply unit and connecting cable up to 30m.
(* The reference operating condition: 23ºC±5ºC, 35 to
75%RH)

 Specifications
Environment

 OPTIONS
OptionCommunications
interface*
Analog output*
Contact input*
Contact output*
Laser targeting

FunctionContents
RS485:
Sendingofmeasuringdata,
andsending/receiving of parameters
4-20mA input signal: Selection of emissivity remotesetting or automatic emissivity computation
1 point: Peak hold reset or sample hold. Dry contact oropen collector
1 point: High(low) alarm or error signal. Photo coupler
30VDC 50mA max
Built-in semiconductor laser emitter. 1mW or lower(645nm), class2. No viewfinder model.

Measuringmode* Only one kind of option to be selected.

CAUTIONS FOR LASER TARGETING MODELS- Laser may damage your eyes. Don't stare into a laser beam.
- Make sure to prevent from the reflection when you want tomeasure an object equivalent to mirror surface like a brilliant
metal.

 SETTING/DISPLAY PART

 Connectivity

5

Wimdows 7/10
OS
Hard drive
20MB or more
16MB or more
MemoryFloppy disk drive
Drive
Measuring data display
Data storing, replay, print
1-3 units connectable
Real time trend mode SPECIFICATIONS

Semiconductor IR-CAT□□□, IR-CAU□□□

Film Temperature IR-CAN□□□, CAM□□□
Measuring system: Narrow-band radiation thermometer
Element:
IR-CAN ---PE
IR-CAM ---PbSe
Measuring wavelength:
IR-CAN --- 8 μm
IR-CAM --- 3.43 μm
Measuring range:
IR-CAN ---0 to 300ºC
IR-CAM ---30 to 300ºC
Accuracy rating:Lower than 200ºC ---±2ºC
More than 200ºC --- ±0.1% of measuredvalue(at ε≒1.0 and reference operating
conditions)
Repeatability:
1ºC or lessStability:
Temperature drift
0.15ºC /ºC
At EMC test environment…IR-CAN: ±15%of measuring range
IR-CAM: ±10% of measuring range
Resolution:
1ºC
Response time (95%):
1 sec
Optics:Fixed focus lens type
Sighting:
Laser spot without viewfinder
Lens aperture:
15mm diameter
Power consumption:
IR-CAN --- Maximum 5VA
IR-CAM --- Maximum 10VA(* The reference operating condition: 23ºC
±5ºC, 35 to 75%RH)

Relation between measuring distance and diameter

Measuring system: Narrow-band radiation thermometer
Element: Si
Measuring wavelength:IR-CAT --- 0.6 to 0.96 μm
IR-CAU --- 0.6 to 0.9 μm
Measuring range:
IR-CAT --- 400 to 800ºC (distance factor 100)
500 to 1000ºC (distance factor 200)
600 to 1200ºC (distance factor 200)
IR-CAU --- 400 to 800ºC (distance factor 100)
500 to 1000ºC (distance factor 200)
(at ε≒1.0 and reference operating conditions)
Accuracy rating:Lower than 600ºC ---±3ºC
More than 600ºC --- ±0.5% of measured value
Repeatability:
0.5ºC or lessStability:
Temperature drift
Lower than 700ºC ---0.1ºC /ºC
More than 700ºC --- 0.015%/ºC of measured value
At EMC test environment…±10% of measuring range
Resolution:
0.5ºC
Response time (95%): 0.04 sec
Optics:
Focusable lens type
Sighting:
Direct viewfinder
Lens aperture:
20mm diameter
Power consumption: Maximum 10VA(* The reference operating condition: 23ºC±5ºC, 35 to
75%RH)

Relation between measuring distance and diameter
Measuring distance: 0.5m to ∞Measuring diameter: Measuring distance/distance factor

IR-CA□G

Measuring
distance(mm)Distancefactor100
200

IR-CA□H

IR-CA□J

Measuring wavelength:
Measuring range:
Accuracy rating:Repeatability:
Stability:

Resolution:
Response time (95%):
Optics:Sighting:
Lens aperture:
Power consumption:

1000

2000

φ5
φ2.5

φ10
φ5

φ20
φ10

Glass Temperature IR-CAG□□□

Measurement Inside Furnace object IR-CAR□□□
Measuring system:

500

Narrow-band radiation thermometer
Element: PbSe
3.8 μm
350 to 1100ºC (distance factor 100)
450 to 1300ºC (distance factor 200)
500 to 1500ºC (distance factor 200)
Lower than 1000ºC --- ±5ºC
More than 1000ºC --- ±0.5% of measuredvalue(at ε≒1.0 and reference operating
conditions)
1ºC or lessTemperature drift
Lower than 1000ºC ---0.2ºC /ºC
More than 1000ºC --- 0.02%/ºC ofmeasured value
At EMC test environment…±10% of
measuring range
1ºC
0.02 sec
Focusable lens type
Direct viewfinder
20mm diameter
Maximum 10VA(* The reference operating condition: 23ºC
±5ºC, 35 to 75%RH)

Measuring system: Narrow-band radiation thermometer
Element:
MCT
Measuring wavelength: 5 μm
Measuring range:
100 to 800ºC(distance factor 50)
200 to 1800ºC(distance factor 100)
400 to 2800ºC(distance factor 200)
Accuracy rating:Lower than 1000ºC---±5ºC
1000 to 1500ºC---±0.5% of measured value
1500 to 2000ºC---±1% of measured value
More than 2000ºC--- ±2% of measured value(at ε≒1.0 and reference operating conditions)
Repeatability:
1ºC or lessTemperature drift:
Lower than 1000ºC --- 0.2ºC /ºC
More than 1000ºC --- 0.02%/ºC of measured value
Resolution:
1ºC
Response time (95%): 0.1 sec
Optics:
Focusable lens type
Sighting:
Direct viewfinder
Lens aperture:
20mm diameter
Power consumption: Maximum 10VA(* The reference operating condition: 23ºC±5ºC, 35 to
75%RH)
Relation between measuring distance and diameter
Measuring distance: 0.5m to ∞Measuring diameter: Measuring distance/distance factor
Measuring
distance(mm)Distancefactor50
100
200

Relation between measuring distance and diameter
Measuring distance: 0.5m to ∞Measuring diameter: Measuring distance/distance factor
Measuring
distance(mm)Distancefactor
100
200

500

1000

2000

φ5
φ2.5

φ10
φ5

φ20
φ10

6

500

1000

2000

φ10
φ5
φ2.5

φ20
φ10
φ5

φ40
φ20
φ10

Food industry IR-CAFX0□ (non-CE approval)

 EXTERNAL DIMENSIONS

Measuring system:
Narrow-band radiation thermometer
Element:
PbSe
Measuring wavelength:4μm
Measuring range:
60 to 100ºC
Accuracy rating:70 to 90ºC ---±1.0ºC
Except 70 to 90ºC ---±2ºC(at ε≒1.0 and reference operating conditions)
Repeatability:
0.3ºC
Temperature drift:
0.04ºC /ºC
Resolution:
0.2ºC
Response time (95%): 0.01 sec
Optics:Fixed focus lens type
Sighting:
Laser targeting without viewfinder
Lens aperture:
15mm diameter
Power consumption: Maximum 10VA(* The reference operating condition: 23ºC±5ºC, 35 to 75%RH)

IR-CAN
IR-CAM
IR-CAF
IR-CADA

Relation between measuring distance

IR-CAR
IR-CAG
IR-CAT
IR-CAU

IR-CAFX01

IR-CAFX02

Unit: mm

 Models
Unit: mmPolyester film
ModelsIR-CANG□□
IR-CANH□□
IR-CANJ□□
IR-CANZ□□

Measuring
diameter/Measuring
distance
φ37/1000mm
φ15/400mm
φ8/200mm
Special (Ask CHINO)

MeasuringrangeStandard sighting

0 to 300℃

Laser targeting(without
viewfinder

Polyethylene film
ModelsIR-CAMG□□
IR-CAMH□□
IR-CAMJ□□
IR-CAMZ□□

Measuring
diameter/Measuring
distance
φ37/1000mm
φ15/400mm
φ8/200mm
Special (Ask CHINO)

MeasuringrangeStandard sighting

30 to 300℃

Laser targeting(without
viewfinder

 HMD (Hot Metal Detector)
IR-CADAC01 (non-CE approval)
Output is turned ON
when hot metal enters
the optical sensing pathand exceeds the preset
threshold level.

 Features□ Connection
C : Connector
T : Terminal●Detect luminance temperature of 100 to 550ºC or
equivalent.
●Remote object detection●External detect level setup by 4-20mA DC□ External input/output (option)
N: None
S: RS485
5: 4-20mA DC input
J: Contact input (DI)K: Contact output (DO) ModelIntrafurnace object
ModelsDistance factor

IR-CAR1□□□
IR-CAR2□□□
IR-CAR2□□□

100
200

Measuringrange
350 to1100℃
450 to1300℃
500 to1500℃

IR-CADAC01

Standard sighting

 Connectivity

Direct viewfinder

Glass
ModelsDistance factor

IR-CAG0□□□
IR-CAG1□□□
IR-CAG2□□□

50
100
200

Measuringrange
100 to 800℃
200 to1800℃
400 to 2800℃

Standard sighting
Direct viewfinder Specifications

Semiconductor/Silicon
ModelsDistance factor

IR-CAT1□□□
IR-CAT2□□□
IR-CAT2□□□

100
200

Measuringrange
400 to 800℃
500 to1000℃
600 to 1200℃

Detection system

Standard sighting

Detection

Direct viewfinder

Response time
OutputSemiconductor/InGaAs
ModelsDistance factor

IR-CAU1□□□
IR-CAU2□□□

100
200

Measuringrange
400 to 800℃
500 to1000℃

Detection level

Standard sighting

OpticsMeasuring spot size
TargetingWorking temperaturePower supply

Direct viewfinder□ Connection

C :Connector
T :Terminal□ External input/output (option)
N: None
S: RS485
5: 4-20mA DC input
J: Contact input (DI)K: Contact output (DO)

Accessory□ Sighting

Blank: With view finder (standard)
L: Laser targeting (option) *without view finder

7

Radiation luminance thresholdjudgement
Luminance temperature of 100 to
550ºC or equivalent
0.1 sec
Open collector, normally OFF
Built-in trimmer or external
4-20mA DC
Fixed focus lens type
Φ150mm/15m
Direct viewfinder (reverse view)0 to 50ºC
24V DC (22-28V DC)
Airpurge hood(sold separately)

Communications interface: SETTING DISPLAY UNIT IR-GZA

Connectable number of IR-CA:

Power supply to IR-CA:
Power supply:Power consumption:
Working temperature:
Working humidity:
Casing:
Installation:
Weight:
CE Marking:Setting display unit IR-GZA

Wall-mount box IR-ZGBW

The IR-GZA is combined with the IR-CA with optional
RS485, programs parameters, displays measuring dataand supplies 24V DC power to the IR-CA.

RS485 (Send measuring data, send/receive
each setting parameters, option)
Maximum 1 unit (Up to 2 units for
IR-GZA2□□, prepare separate power supply
for the second unit)
24V DC 830mA
100 to 240V AC, 50/60Hz
100VAC Max. 28VA, 240VAC Max. 36VA
-10 to 50ºC
20 to 90%RH (No dew condensation)
Nonflammable Polycarbonate
Panel mount type
Approx 0.5Kg
EMC directive EN61326+A1
Low voltageENN61010-1+A2
Overvoltage category II,
Pollution level 2
Stability at EMC test environment…±10% External dimensions

 Connectivity

Unit: mm

 ModelWall-mount box IR-ZGBW (Purchase IR-GZA separately)

 SPECIFICATIONSEmissivity (ratio) setting:
Thermometer input:
Signal modulation:

Analog input:

Display:
Analog output:

Output renewal cycle:
Output accuracy ratings:

1.999 to 0.050
RS485
DELAY --- First-order lag (Time constant: 0.0 to
99.9 sec with 0.1 sec increment or 0.00 to 9.99
sec with 0.01 sec increment) Real signal must
be set at 0 sec.
PEAK --- Peak tracing (attenuation degree 0.1
to 10ºC /sec selectable, 0.1s increment)
Remote emissivity setup or Reflection
compensation
4 to 20mA
Temperature, Status display
Output 1: 4 to 20mA DC IR-GZ output (Loadresistance: less than 600Ω)
Output 2: 4 to 20mA DC IR-CA output (Loadresistance: less than 500Ω)
Output 1: 100ms
Output 2: Depending on the model of IR-CA
Output 1: ±0.3% of output range
Output 2: Depending on the model of IR-CA
Event output: 2 points
Select 2 points within "High temperature alarm","High-high temperature alarm", "Lowtemperature alarm”, "Low-low temperature
alarm" and “self diagnostic function”
Relay a-contact: High, High-high, Low, Low-low
Relay b-contact: Self diagnostic function
Contact capacity
240V AC 1.5A
30V DC 1.5A

Unit: mm

 Terminal diagrams8 ACCESSORIES Soft protective case IR-ZCCST (terminal type)The soft protective case IR-ZCCST is an exclusive accessory for the
IR-CA terminal type to protect the thermometer from smoke, dust,
etc. at the installation site. This unit provides airpurge to remove
smoke and dust for keeping the lens clean. Use clean dried air.

 Soft protective case IR-ZCCSC (connector type)The soft protective case IR-ZCCSC is an exclusive accessory for the
IR-CA connector type to protect the thermometer from smoke, dust,
etc. at the installation site. This unit provides airpurge to remove
smoke and dust for keeping the lens clean. Use clean dried air.

 Hard protective case IR-ZCCHT (terminal type)The hard protective case IR-ZCCHT is to protect the IR-CA terminal
type from high-temperature, humidity, smoke, dust, fume, etc. This
unit provides airpurge and water-cooling to operate the thermometer
properly in harsh environment.

 Hard protective case IR-ZCCHC (connector type)The hard protective case IR-ZCCHC is to protect the IR-CA
connector type from high-temperature, humidity, smoke, dust, fume,etc. This unit provides airpurge and water-cooling to operate thethermometer properly in harsh environment.

9 Airpurge Hood IR-ZCAP(for IR-ZCCH□) Front water-cooling plate IR-ZCWC(for IR-ZCCH□)

 Flange mounting unit IR-ZCAF(for IR-ZCCH□)

The airpurge hood is used to disperse
dust and fume for keeping the light path.
It is mounted to the front of the hardprotective case IR-ZCCH□. Use clean
dried air.

The front water-cooling plate is used when
installing the thermometer under highambient temperature. It is mounted to thefront of the hard protective case IR-ZCCH□.
It is applicable when the thermal radiation isintense from the front.

The flange mounting unit is used for fixing at
the front of hard protective case IR-VCCH□.
It is also applicable for mounting the IR-VSW
and IR-ZW□.

Water-cooling flange IR-VSW

Sealing window IR-ZW□
Windowmaterial

Connecting cable

0

Quartz

1

CaF2

2

BaF2

Applicable model
IR-CAI,IR-CAS,IR-CAQ,
IR-CAP,IR-CAU,IR-CAT
IR-CAE,IR-CAG,
IR-CAR,IR-CAN,IR-CAM
IR-CAB,IR-CAW

Close-up lens IR-VAD□□□ (for focusable model)

IR-ZCRC□□□
Example: 1m

001

IR-ZCRT□□□
Example: 1m001

Example: 1m

001

IR-ZCRL□□□

ModelsMeasuring distance

IR-VAD30A

190 to 300 mm

IR-VAD30G

190 to 300 mm

IR-VAD60A

270 to 600 mm

IR-VAD60G

270 to 600 mm

Applicable model
IR-CAI,IR-CAS,IR-CAQ,
IR-CAP,IR-CAU,IR-CAT
IR-CAE(Focusable model),IR-CAG,IR-CAR
IR-CAI,IR-CAS,IR-CAQ,
IR-CAP,IR-CAU,IR-CAT
IR-CAE(Focusable model),IR-CAG,IR-CAR

Mirror IR-ZCLM

Eyepiece filter IR-ZCLF

Angle finder IR-ZCLATripod IR-ZBMT

Universal Head
Simple type IR-VMS

Soft type IR-ZMS

10

Hard type IR-VMH

11

Specifications subject to change without notice. Printed in Japan (I)32-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
PHONE: +81-3-3956-2171
FAX: +81-3-3956-0915
E-mail: inter@chino.co.jp
Website: http://www.chino.co.jp

122018. 12 Recycled Paper