Machine Vision Cameras

EoSens ® 4CXP
High-Speed CMOS Camera

Engineered for Performance

EoSens ® 4CXP Key Features:• 563 fps at 4 megapixel resolution• Up to 225,000 fps at reduced resolution• High-performance CoaXPress® interface• Extended feature set• 80 x 80 x 48 mm small

APPLICATION EXAMPLES• Laser triangulation• PIV measurements• Rail inspection• 3D inspection• Pick and place processing• PCB inspection• Flaw detection• Ballistics and explosives

Featuring a 4/3’’ CMOS sensor, the EoSens ® 4CXP offers
2,336 x 1,728 pixel resolution at an outstanding speed of563 frames per second. Higher frame rates can be achieved
by defining up to three independent regions of interest.
Combined with a 4-channel CXP-6 CoaXPress® output, theEoSens ® 4CXP transfers megapixel images in real time.

Powerful Features
The EoSens ® 4CXP boasts an extended feature set. Image
processing functions allow you to correct noise and gamma
settings, bringing unsurpassed image quality and fidelity toyour application. The sequencer function enables you to define
four regions of interest, each with a different exposure time.

EoSens ® 4CXPTechnical Data

Compact Design
Contained within an 80 x 80 x 48 mm metal housing, theEoSens ® 4CXP is suited for use in a wide range of scientificand industrial applications.
The camera is designed for operation in rough environments
and has passed the IEC test for vibration, shock, temperature and humidity. It is furthermore MIL-STD-810G certified
allowing it to be utilized in High-G automotive safety andmilitary testing.

(More detailed specifications are available on request)

EoSens ® 4CXP
(monochrome / color)

Resolution

4 MpixActive pixels

2,336 x 1,728 px

InterfaceCoaXPress® @ 4 x 6.25 Gbit/s

Frame rate (8 bit)

563 fpsSensor

Tested according to IEC 60068-2-X

AM41Sensor type

CMOS global shutter
4/3''

Storage temperature-40 °C … +80 °C

Sensor format

Operating temperature-20 °C … +40 °C

Active sensor area (H x V)

Operating humidity

96 % / 93 % relative humidity

Shock resistanceIEC 60068-2-27: 70 g

Sensitivity (mono)

Vibration resistanceIEC 60068-2-6 /IEC 60068-2-64: 7 grms

Color depth

16.35 x 12.10 mmPixel size

7 x 7 μm
11 V/lux*s @ 550nm
10 / 8 bit

Dynamic range

Frame rates

60 dB

Shutter time (steps)

1 µs

Shutter time range

1 µs – 1 s

563 fpsMax. Trigger frequency

300 kHz

1,920 x 1,080 px

900 fpsMax. Jitter±4 ns

1,024 x 768 px

1,260 fps

640 x 480 px

2,020 fps

128 x 128 px

7,490 fps

Resolution
2,336 x 1,728 px

Frame rate

max. 225,000 fps

DIN 1.0 / DIN 2.3

Mount options

C-Mount / F-Mount

Dimensions(W x H x L w/o mount)

80 x 80 x 48 mm

Weight (C-Mount)

Smart Features• Gamma Correction• FPN Correction• Noise Reduction (PRNU)• Multiple ROI

Connector options

450 gPower consumption• PoCXP• Sequencer Function• Skipping Mode
• GPIO

10 W

Power supply

12 – 24 V DCConformity

CE / RoHS / GenICam / CoaXPress®

MIL-STD-810G certified

EMVA1288 reports



Camera Body Dimensions

11/2020MIKROTRON GmbHMIKROTRON GmbH provides a full range of high-speed
imaging solutions for challenging applications in industry,engineering, science and sports. The company‘s extremeslow-motion recording enables customers to optimize
manufacturing processes, improve product design,
revolutionize quality management and analyze motion.

Germany
Landshuter Str. 20-2285716 Unterschleissheim+49(0)89-726342-00
info@mikrotron.de
www.mikrotron.de

All trademarks are properties of their respective owners. MIKROTRON reserves the right of change without notice.
MIKROTRON is not liable for harm or damage incurred by information contained in this document.

North America
14032 Hermosillo Way
US-Poway, CA 92064+1-858-774-1176
steve.ferrell@mikrotron.de
www.mikrotron.de