Impeller flow meter

FS-C

Flow meter for measuring the fluid flow rate and temperature

Alarm output and Analog output

LCD display



3-color LED*Descriptions are based on Japanese Industrial Standards (JIS)Specifications
Model
Flow range
Fluid
Maximum working pressure

FS-3C

FS-10C

FS-30C

0.5-3 L/min

1.5-10 L/min

5-30 L/min

Industrial water, water
0.97 MPa

Flow temperature

0-60C° (No freezing , No dewing)

Ambient temperature

0-40C° (No freezing , No dewing)

Power supply
Current consumption
Cable

Alarm output

DC 24 V ±10%Max. 40 mAAWM20276, 9 core / 0.2 mm? / 500 mm
redPower supply

blackGround

white

Analog output (flow rate)yellow

Alarm output (A output)

brown

Alarm output (B output)

green

Alarm output cable commonly used

OrangeAnalog output (temperature)

Blue

Spare

Gray

Spare

Max.load current : Less than DC100A
Max.applied voltage : Less than DC40V
Output mode : A or B open collector output

Analog output

Voltage output : DC0-10V

Temp. Measure

Accuracy : ±1C°
Temp. Range : 0-60C°
Output mode : Voltage output (DC0-10V , over 3kΩ load)
Display : 2-3 digits LCD with everi 1C° notch (-10-+100C°)

DisplayLCD, LEDO ringChloroprene or fluoro rubber

Wetted material

POM, polyamide

Adapter size

Rc 3/8" or Rc 1/2"

Adapter material
Accuracy

Stainless steel or brass with nickel plating
FS ±2.5%

Flow direction

FreePositioning

FreeWeight
Protection

Approx. 380 g
Weather resistant electronics

Order form

■ISO screw(Rc) is used.
Flow range

FSï

WettedMaterials

Outputーー

Adapter
MaterialsーAdapter size(ISO)O ringPower supplyーーー

Adapter Materials

Adapter size

O ringBStandard

How to order：FS-10S-0−J−B−3/8−F−24
Flow range

FS-S
type

FS-AR
typeFS3S

0.5〜3／min

10S

1.5〜10／min

0Standard(Polyacetal)

J

4150°C-compatible(SUS304・316)

24

DC24V±10％

٘1/4

V

12SUS

Only on
150°C-compatible
bodiesFluororubber

DC12V±5％Flow range

OutputWetted Materials

Adapter Materials

Adapter size

O ringPower supply

3AR 0.5〜3／min

0〜10V

B

Standard

10AR 1.5〜10／min

J0Standard(Polyacetal)

5〜30／min

3NB

PSUS

Only on
150°C-compatible
bodiesFluororubberDC12V±5％

Puls Output

T
fluorine resin

OutputWetted Materials

Adapter Materials

Adapter size

O ringPower supply0J

Standard(BsBM＋Ni plating)

0〜10VStandard(Polyacetal)

SB

1/2

OutputWetted Materials

JStandard(Polyacetal)

P
Pulse Output

B

150°C-compatible(SUS304・316)

Tfluorine resin

30N 5〜30／min

1/4

3/8
Standard

S

1/2

Adapter Materials

Adapter size

B

Standard

Standard(BsBM＋Ni plating)

1/2

S

1/4

3/8
٘SUS

DC24V±10％٘SUS

0.5〜3／min

10N 1.5〜10／min

24

150°C-compatible(SUS304・316)

5〜30／minFlow range

F

Chloroprenerubber4

0.5〜3／min

10C 1.5〜10／min

3/8Standard(BsBM＋Ni plating)

1〜5V
4〜20mA

30C

FSF

ChloroprenerubberT
fluorine resin

3CFS

S

1/2

4〜20mA

1FS

B

Standard(BsBM＋Ni plating)

3/8

Power supply

30S 5〜30／min

Flow range

FS-N
type

Wetted Materials

0〜10V

30AR

FS-C
type

OutputOnly on
150°C-compatible
bodiesV

12F

Chloroprenerubber24

V

DC24V±10％

O ringPower supply

FluororubberF

Chloroprenerubber

VFluororubber24

DC24V±10％12DC12V±5％Regarding ＜Adapter size＞Regarding ＜Adapter size＞ Regarding ＜Adapter material/O-ring＞

The diameter type of the connection section differs
depending on the type of body material.
Please refer to the table below when filling out yourorder form.

・When selecting "B: 150°C-compatible" as the body material,
do not fill in the adapter material and O-ring fields.
・When selecting "T: Fluororesin" as the body material, do not fillin the O-ring field. We also offer products in which the adapter
material is fluororesin. Please contact our sales department for details.

Connection diameter
Wetted MaterialsStandard type(J)150°C/Compliant(B)fluorine resin type(T)

3/8

1/2

1/4

* 1/4 diameter cannot be selected for 30L specification.

Other① Please contact our sales department for other specifications.
② Please contact our sales department for inquires regarding
date of delivery, etc.
③ Shape and specification may change without notice due toperformance improvements.

10