FIELD DATA SCANNING UNIT
SE3000 SERIES
SE3000 series scanning units are designed fordata logging with a personal computer. Thebasic unit with 6 input points can be connected
to 7 sub units (6 input points / sub unit) andmaximum 48 analog data can be collected.
Two soft ware packages, “KIDS” for datalogging
and“PASS”
forparameter
programming, are available.
This unit can be used for input point extensionof CHINO’s BR series Graphic Recorders.
Features
MODBUS protocol
Three kinds of serial communications, RS-232C,RS-422A and RS-485, are standard.
The communications protocol is MODBUS for easy
systemconfigurations
andanyspecificcommunications software package is not required.

Data logging software package
Data can be managed by the data logging software
package “KIDS” (separate purchase required) on a
personal computer. The logged data can be utilized
by worksheet applications.

KIDS

MODBUS

SE3000Programdisplayunit

SE3000

System configuration

Parameter programming software package
Through an engineering port, parameters can beeasily set up by the parameter programming
software package “PASS” (separate purchaserequired) from a personal computer.

[KIDS]Up to 5 sets or
Up to 100 channels

PASS

SE3000

Universal input
10 dc voltage/current inputs, 36 thermocouple
inputs and 11 resistance thermometer inputs are
standard, and ranges can be set to each channel
independently.

Upto48pointsincluding temperature,pressure and flow

1

Up
to
48
pointsincluding temperature,pressure and flow

PSEPSE-300

Input specifications
Measuring point:

Basic unit 6 points, sub unit 6 points/unit
Up to 7 sub units can be connected to one basic unit. (Up to 48 points)
Input kinds:
Universal input (dc voltage, dc current, thermocouple, resistance thermometer)
Accuracy ratings:
Refer to the table of measuring ranges.
Measuring interval:
6-point/1 second, 12-point/2 seconds, 18-point/3 seconds, 24-point/4 seconds,30-point/5 seconds, 36-point/6 seconds, 42-point/7 seconds, 48-point/8 seconds
Reference junction compensation accuracy:
K, E, J, T, N, Platinel ….. Lower than ±0.5ºC
R, S, Ni-NiMo, AuFe-Cr, WWRe5-26, WWRe0-26, U, L ….. Lower than ±1.0ºC(The above error is added to the accuracy ratings for the internal reference junctioncompensation.)
Burnout:
The sensor disconnection is detected for the thermocouple input and the resistancethermometer input.
Burnout enable or disable can be selected in each input.

Display specifications (Basic unit)∙ Status display
Display element:
Display content:∙ Data display (option):

Circle type red LED 3 pieces, green LED 1 piece
Run status
Normal
Red LED blinks.
Communications status
Receiving
Green LED lights.
Transmitting
Red LED lights.
Alarm status
Activating
RED LED lights.
5-digit 1 point, Channel shifting/fixed

Programming specifications (basic unit)Communications programming: By DIP switches and sliding switches
Input programming:
By the parameter programming software package “PASS” (separate purchase required) from apersonal computer.
(For the connection to BR graphic recorders, the input programming can be executed by the BR
recorders, too.)

Alarm specifications (Basic unit)
Alarm point:
Alarm:
Alarm output:

Up to 4 points per each channel
High, low, high difference, low difference, high rate-of-change, low rate-of-change
Not provided

Safety specifications
CE: EN61326 A1 Class A, EN61010-1 A2General specifications
Rated power voltage:
Power consumption:
Operating condition:

Casing:Weight:Mounting:

100 to 240VAC, 50/60Hz10VA∙ Normal operating condition …
Ambient temperature/humidity 0 to 50ºC, 20 to 80%RH
Power voltage 90 to 264VAC, Power frequency 50/60Hz ± 2%
∙ Reference operating condition …
Ambient temperature/humidity 21 to 25ºC, 45 to 65%RH
Power voltage 100VAC ± 1%, Power frequency 50/60Hz ± 0.5%
Altitude Left/right 0º, forward tilting 0º, backward tilting 0º
Warming up time More than 1 hour
Resin
Basic unit 0.6kg, sub unit 0.2kg
DIN rail (35mm)

Options
Option
Display unit

Contents
Display unit: 7-segment red LED 7-digit
Display contents: Channel number 2-digit Data 5-digit
Display renewal cycle: 3 seconds

Accessories (separate purchase required)
Accessories
KIDSPASS
Shunt resistor for current input

Contents
Data logging software package Windows 95/98/NT4.0/2000
Parameter programming software package Windows 95/98/NT4.0/2000
For 4 to 20mA 250Ω (1.5W), For 10 to 50mA 100Ω (1.5W)

2

General specifications

Model
Basic unit (6 input points)
SE310 -000

Input kind

Measuring range

DC voltage-13.8

to13.8mVReferencerange±13.8mV-27.6

to27.6mV±27.6mV-69.0

to69.0mV±69.0mV

10µV

-200

to-200.0mV±200.0mV

100µV

-500

to-500.0mV±500.0mV

100µV

-2

to2V±2V

1mV-5

to5V±15V

1mV

-10

to10V

±10V

10mV

-20

to20V±20V

10mV

-50

to50V

±50V

10mV

-200

to300ºC±13.8mV-200

to600ºC±27.6mV-200

to1370ºC±69.0mV

1ºC-200

to200ºC±13.8mV

0.1ºC
0.1ºC

Display unit (option)
0: Not provided
1: Provided
Sub unit (6 input points)
SE3211-000Parameter programming software package

K“PASS” (separate purchase required) Ver. 4.0 or later
The parameter programming software package“PASS” is for programming parameters of theSE3000.
Programming

Channelparameters

EJ

Input kids, range/scale, alarmand burnout
T

Storingofparameters

File/print

Temperature
deadband

unit,alarm

R

S

Storing programmed parameters
to a file (FD, HD), replaying andprinting
Thermocouple

OthersData logging software package“KIDS” (separate purchase required) Ver. 4.0 or later
The data logging software package “KIDS” is forlogging the data measured by the SE3000 andreplaying the stored data.
∙ Data logging by up to 5 sets of SE3000 (up to100 channels)∙ Data logging, replaying, trend graph and report
creation∙ Stored data for worksheet applications

Groupmonitor

Operation

Real timetrend

Monitoring of system configurationand connected instruments±0.1% ± 1 digit

0.1ºC
0.1ºC

-200

to350ºC±27.6mV-200

to900ºC±69.0mV

1ºC-200

to250ºC±13.8mV

0.1ºC
0.1ºC

-200

to500ºC±27.6mV-200

to1200ºC±69.0mV

1ºC-200

to250ºC±13.8mV

0.1ºC

-200

to400ºC±27.6mV

0.1ºC

0to1200ºC±13.8mV

1ºC0to1760ºC±27.6mV

1ºC0to1300ºC±13.8mV

1ºC1ºC

to1760ºC±27.6mV

0to1820ºC±13.8mV

N0to400ºC±13.8mV

0to750ºC±27.6mV

0.1ºC

0to1300ºC±69.0mV

1ºC

WWRe0-260to

2320ºC±69.0mV

WWRe5-260to2320ºC±69.0mV

1ºC±0.15%± 1 digit

0.1ºC

1ºC±0.2%± 1 digit

1ºC

PR5-200to

1800ºC±13.8mV

1ºC

PR20-400to1880ºC±13.8mV

1ºC

Ni-NiMo

0to290ºC±13.8mV

0.1ºC

0to600ºC±27.6mV

0.1ºC

0to1310ºC±69.0mV

1ºC

AuFe-Cr

0to300K±13.8mV

0.1ºK

Platinel-100

to350ºC±13.8mV-100

to650ºC±27.6mV-100

to1390ºC±69.0mV

1ºC-200

to250ºC±13.8mV

0.1ºC
0.1ºC

Pt100

Displayingofdata(10channels/screen) in a trend format

10µV

0L

Displaying of tag, data, alarm status,
etc. of 20 channels per screen in atable format

Displayresolution
10µV

BU

Resistance thermometerSystemmonitor

Accuracy
ratings±0.1% ± 1 digit

Old Pt100

JPt100-200

to500ºC±27.6mV-200

to600ºC±69.0mV-200

to250ºC±13.8mV±0.15%± 1 digit

0.1ºC0.1ºC0.1ºC±0.1%± 1 digit-200

to500ºC±27.6mV-200

to900ºC±69.0mV-140

to150ºC160Ω±0.15% ± 1 digit±0.1% ± 1 digit

0.1ºC
0.1ºC
1ºC0.1ºC-200

to300ºC

220Ω

-200

to850ºC

400Ω

0.1ºC

-140

to150ºC160Ω±0.15% ± 1 digit-200

to300ºC220Ω±0.1% ± 1 digit-200

to649ºC

400Ω

-140

to150ºC160Ω±0.15% ± 1 digit-200to300ºC

220Ω±0.1% ± 1 digit-200

to649ºC

400Ω

0.1ºC

-200

to649ºC

220Ω

0.1ºC

0.1ºC
0.1ºC
0.1ºC
0.1ºC
0.1ºC

Alarm
summary

Displaying and printing of alarm
status log of past 500 data

Historicaltrend

Displaying of stored data (10Pt-Co
4
to374K220Ω±0.15% ± 1 digit
0.1K
channels/screen) in a trend format
Note: The accuracy ratings are under the reference operating conditions.
Trend period: 5 to 300 seconds
For
thermocouple
inputs(internalreference
junctioncompensation), the accuracy ratings do not include the reference
Recalling stored data in a tablejunction compensating accuracy. The detailed description of theformat and displaying in a trend
accuracy ratings is separately prepared.
format* The indication may vary to the value of maximum 2mV or in equivalent
to 25ºC under EMC test environment.
Input kids, range/scale, alarm andburnout

Pt50

Datareplay/conversion
Groupmonitor

30.1ºCTerminal board
The figure shown below is the terminal board for the connection with 3 sets of the sub unit (total 24 inputpoints) and with the display unit (option).
Sub unit(A), (+)
Communications
RS-232C SD
RS-232C RD
RS-232C SG
RS-422A SDA
RS-422A SDB
RS-422A RDA
RS-422A RDB
Power/Ground
100 to 240VAC L
50/60HzH
Protective conductor
terminal(B), (-)(A), (+)

CH1(A), (+)(B), (-)

CH3

CH4

CH5(A), (+)(B), (-)

Basic unit

CH1to
CH6

CH7to
CH12

CH13
to
CH18

CH19toCH24(B), (-)
(B)

(B)

CH6

(B), (-)
(B)

(B)
CH2

(A), (+)(A), (+)(B), (-)(B)(B)
For thermocouple and dc
voltage inputs … (+) (-)
For resistance thermometer
input … (A) (B)Input terminals

Terminal board
For the connection of 1 basic unit and 3 sub units (total 24 input points), and with the display unit

72+36xN (N≤7)
3678138

138

72

100

4
[Basic unit]

Unit: mm[Sub unit]* MODBUS is a registered trademark of Schneider Automation Inc., USA. Windows and Windows NT are registered trademarks of Microsoft
Corporation, USA.
Products and company names shown in this catalog are trademarks or registered trade marks of their respective companies.

Specifications subject to change without notice. Original

32-8, KUMANO-CHO, ITABASHI-KU, TOKYO 173-8632
PHONE: +81-3-3956-2171
FAX: +81-3-3956-0915
E-mail: chinojpn@mb.infoweb.ne.jp

Website: http://www.chino.co.jp

42000.12