IMADA CO., LTD.
Standard Model Digital Force-Gauge ZTS Series

Standard Model Digital Force Gauge ZTS Series-Fast sampling for both Peak Values and Continuous Data, achieving high reproducibility-Clear OLED display with a variety of significant capabilities-Measurement benefit increased by external output and practical functions-Allows easy Data acquisition and management with the included software

Product Package withincluded Accessories

High Load Model
ZTS-2500N~5000N

ZTS-2N~1000N[Evolution to the Next Series]
Next Series is a generic term for IMADA measurement devices with upgraded functions, performances with theessential modifications. The devices enable to install additional functions from the user support site, IMADAConnected. On the website, related data such as software and instruction manuals can also be downloaded.
New Series: ZTS Series Firmware Ver.5 or later.

New Features-Increased measurement stability achieved fromfurther noise reduction to the measurement circuit.

-Force gauge program (firmware) online updates
available.
-Various software and additional functions
downloadable to increase the measurement
conveniences.
-Various software and functions to increase themeasurement conveniences downloadable on
IMADA-Connected.

Analog output comparison
Yellow: Next series/Green: Old Model

IMADA Connected(https://www.imada-connected.com/)

1 / 7
2023/12/27

IMADA CO., LTD.
Standard Model Digital Force-Gauge ZTS Series[Features]
Fast sampling for both Peak Values and Continuous Data, achieving high reproducibility
High-speed sampling of up to 2000 times/second updates measured
values for more accurate peak-value readings.
By using the optional software sold separately (refer to P.5), graphed
as continuous Data with the sampling data of up to 2000 times/second,and even frictional Load changes visualized and evaluated.

*The Test Results vary depending on the sample and measurement conditions.

Destructive test comparison of the same
sample by sampling speed: -Left Sampling speed 2000Hz-Right Sampling speed 100Hz
The less variation shown in the left results

Clear OLED display with a variety of significant capabilities
The OLED provides excellent contrast and high visibility, which reduces
errors in reading measurement results.
In addition, the display screen can be divided into three sections, allowing
to customize the display contents for liking, such as calendar, bar graphand Comparator setting values on the top and bottom.
In addition, it has various display capabilities such as a multi-language
setting menu and overload warning display.

Measurement benefit increased by external output and practical functions
By providing a wide range of data output options such as USB, wireless,serial communication, and analog output, it can be used in a wide rangeof situations from data management on a PC to interlocking with variousdevices.
The system used to control external devices, such as stopping thethe machine at a specified load value, or to create an inspection device linked
to a work process in order to improve the efficiency of the inspection process.

The warning light and otherdevices may be used todetect measurement
abnormalities.

Allows easy Data acquisition and management with the included software
USB Cable and software Force Logger included in the original package toproceed with the data management when purchased.
With the software, displayed values and internal memory data stored in a force
gauge and continuous data acquired (10 data per second).
It also allows to change and save the Force-Gauge settings from facilitating themanagement of measurement conditions. (Please refer to P.4 for the system
requirements.

2 / 7
2023/12/27

IMADA CO., LTD.
Standard Model Digital Force-Gauge ZTS Series[Specifications of ZTS series]
ModelZTSFeatureStandard Model with high functionality and usability.

Accuracy+/-0.2%F.S.+/-1digit

Measurement Unit

N, kgf, lbf (*1)

Display

4-digit Organic EL

Display update

16 / sec

Sampling Rate

2000 data / sec at maximum

Battery

6.5 hours (2 hours for re-charging)

Safe overload rating

Approx.200%F.S.

Operating environment

Temperature: 0 to +40 degree Celsius, Humidity: 20 to 80%RH

Function

Customized display (header and footer), Peak hold (tension and compression), Internal
1000 points data memory, Comparator (judgment of OK or NG), Reversible display, Sign
inversion, Zero clear timer, +NG alarm, Off timer (auto power off), Dumping, Timedisplay, Setting lock

OutputUSB, RS232C / ±2VDC analog output (D/A),
Comparator 3 steps (-NG/OK/+NG), Overload alarm

Overload warning

Approx.110%F.S. (Warning message and alarm)

External connecting switch

Power ON / OFF / Send and Hold / Zero reset / Peak and Real-time switching

WeightFrom 2N to 1000N Model: Approx.490g (*2)
2500N, 5000N: Approx.1100g (*2)

Dimensions

Please refer to dimensions.

Accessory

AC adapter, Inspection certificate, CD driver (including data logging software),Attachments (The set of attachments varies according to range.),
Handles (only for 2500N/5000N range), USB cable, Carrying case.

(*1) The measurement Unit: mN displayed for models of 5N or less, and kN displayed for models of 1000N or more.
In addition, possible to display the unit kg (g: 5N or less range) as a conversion value.
(*2) The Weight differs according to the Model range.

[Option]OptionDescription

Analog output: RAWModelThe raw analog data is output without digital processing-AN-For this option, add the code (-AN) to the end of the Product Model Number. (Example: ZTS-500N-AN)-This option requires a dedicated cable CB-118 sold separately.
-Data responsiveness is higher than standard analog output (D / A), with the disadvantages of Zero reset disable andhigh noise components.

[ZTS series models]ModelCapacity

DisplayResolution

Measuring shaft

Included nut

ZTS-2NZTS-5NZTS-20N
ZTS-50N
ZTS-100N
ZTS-200N
ZTS-500N
ZTS-1000N

2N
5N
20N
50N
100N
200N
500N
1000N

2.000N(2000mN)
5.000N(5000mN)
20.00N
50.00N
100.0N200.0N
500.0N
1000N (1.000kN)

0.001N (1mN)
0.001N (1mN)
0.01N
0.01N
0.1N
0.1N
0.1N
1N (0.001kN)

M6 P1.0 (Coarse thread)
Thread length: 11.6mm

Hexagon nut
Material: Resin
Height: 5mm

ZTS-2500N

2500N

2500N (2.500kN)

1N (0.001kN)

ZTS-5000N

5000N

5000N (5.000kN)

1N (0.001kN)

M10 P1.5 (Coarse thread)Thread length: 15mmHexagon nut
Material: SteelHeight: 6mm3 / 7
2023/12/27

IMADA CO., LTD.
Standard Model Digital Force-Gauge ZTS Series[Included Accessories]-Force Logger: Data Acquisition Software
Main Functions-Easy Importing of the Measurement Data
-Display Acquired Data Value Statistics: -Maximum / Minimum /Average Values.
-Data saved in CSV format-Continuous Data acquisition up to 10 times per second.
-Force-Gauge Function Settings.
Operating environment- OS: 8.1/10/11- Hardware: CPU 1GHz or more recommended
Memory 2GB or more recommended
Hard disk 10GB (Data storage area) or more
- Platform: .NET Framework4.8 or later-Standard Force-Gauge Attachments (Accessories) : For Axis size M6Force-Gauge Capacity 100N or less：A-Types (Aluminum)
Force-Gauge Capacity 200N -1000N：S-Types (Steel)
A-1/S-1*SR-1

A-2/S-2

A-3/S-3

A-4/S-4

A-5/S-5

A-6/S-6

A-7/S-7

A-8/S-8

Small Hook

Disk Tip

Conical Tip

Chisel Tip

Notched Tip

Extension RodLarge Hook

Y HookTensionCompression

Compression

Compression

Compression

Axis Length
Adjustment

TensionTension*Small Hook S-1 for Capacity 200N – 500N, and SR-1 for Capacity 1000N.

-High-Capacity Force Gauge Attachments (Accessories)：For Axis size M10Force-Gauge Ranges 2500N-5000N: B-Types (Steel)
B-1

B-2

B-3

B-4

B-5

B-6

B-7

B-8

Small Hook

Disk Tip

Conical Tip

Chisel Tip

Notched Tip

Extension RodHandle

Mounting Bolts

TensionCompression

Compression

Compression

Compression

Axis Length
Adjustment

For High Capacity
ModelsFor High Capacity
Models4 / 7
2023/12/27

IMADA CO., LTD.
Standard Model Digital Force-Gauge ZTS Series[Related Products]
Downloadable Force-Time (Torque-Time) Graphing Software: Force Recorder Next Standard *1- High sampling rate of max. 2000Hz enables accurate force-time graphing.
- Test condition preset functions increase efficiency of recording andanalyzing measurement results.
- Various data editing functions such as up to 10-graph overlaying
enable analyzing measurement result.
- Adding Images and Comments functions are useful for recording thedetails of measurement.
- The Print Command allows users to output of measurement data as
PDF/Word, Excel, and images and drawing up reports easily.
Force Recorder Next Standard* CD version software Force Recorder Standard is also available. Some specification suchas operating environment is different from Downloadable version and CD version.
* Please refer to the individual specification sheets for details.

Wireless Data Transmit System: WL01 Series-Wireless communication allows the Data received from remotelocations.
-The Data reception detection and interference prevention functions
support stable communication.
-The communication format opened to the public and incorporated into
the available facilities possible.
Note

WL01-USB and WL01-ADP-Compatible with firmware Ver3.10 or later.
-Restrictions on the countries and regions for use.
-Please refer to [Wireless System] Specifications for details.

Handle FOH-1 (*2)

Hand Grip Belt HGB-1

Rechargeable Battery: BP-308

For max. capacity 200N

Attach to the Force Gauge forTo hold and stabilize theRechargeable Battery
Firm-hold and stable operation with
Force-Gauge for single-handfor replacement
both hands
operation*1 To use this product, user & product registration on IMADA Connected and downloading the software are required. Only
Next Series products (For the ZTS series, firmware versions 5.00 or later) are available for the product registration. AnInternet connection is required in process of user & product registration and downloading the software.
*2 High-Capacity Type ZTS-2500N/5000N is not mountable. Handles for the high-capacity type is included in ZTS-2500N/5000N.

[Cables]ModelsCB-108
CB-118
CB-208CB-508
CB-528
CB-908

Names
Analog Cable
Analog cable (For -AN option)
RS232C Cable
Test stand option cable（For MXSeries）
Test stand option Cable（MX2 / MX Series）
Open End Connection Cable

Descriptions
To connect to a multi-meter oscilloscope.
To connect to a multi-meter oscilloscope.
To connect to a PC having its systemTo connect to a test stand to enable force control or overload
prevention functions.
To connect to a test stand to enable force control or overload
prevention functions.
Output cable for loose wire 37 pin(Useful for connection with unique equipment.)

5 / 7
2023/12/27

IMADA CO., LTD.
Standard Model Digital Force-Gauge ZTS Series[Other Options]
Force gauge ZTA Series

Sensor-separate force gauge
ZTS/ZTA Series

eZ-Connect Series: Sensorinterchangeable Amplifier eZTHigh-Functional Model withdisplacement input/output function
added to ZTS series.

External sensor type for use in therestricted environment (narrowspace etc.) or installing into thespecific facilities.

The amplifier enables to connect tothe various eZ-Connect Series LoadCells.

[Dimensions: ~1000N]

Unit: mm

6 / 7
2023/12/27

IMADA CO., LTD.
Standard Model Digital Force-Gauge ZTS Series[Dimensions: 2500N~5000N]

Unit: mm[ISO Calibration Service and Calibration Certificate]-We are the accredited ISO / IEC 17025: 2017 Calibration Organization with the in-house Calibration Service facility.
-The Calibration Service and Issuing of the Calibration Certificates available at cost.
-For further information, please kindly contact the local authorized distributor or us.

[Caution]-Information in this document is subject to change without prior notice.
-This product is designed for force measurement purposes only.
-Do not copy and use this content without authorization.
-Do not apply force more than its capacity and from incorrect direction to the measuring shaft.
-Note that the allowable values may vary slightly depending on the unit selected on display. Please contact us formore information and details.

IMADA CO., LTD.
99, Jinnoshinden-Cho, Aza, Kanowari, Toyohashi,
Aichi 441-8077, JAPANTel: +81-(0)532-33-3288
Fax: +81-(0)532-33-3866
E-mail: info@forcegauge.net
Website: https://www.forcegauge.net/en/

Visit our website for more
information on wide
product specifications,
measurement
applications and videos.

7 / 7
2023/12/27