Prime 95B 22mm™
Scientific
Camera Datasheet
Datasheet
Retiga
LUMOTMCMOSCCD Camera
Scientific CMOS,
Scientific
EMCCDImaging
and CCD
MadeCameras
Easy

Bioluminescence Imaging:
Extreme low light performance
The Retiga LUMO™:
When every photon counts
Great instruments don’t create great science, but they are essential
to tell the story. Teledyne QImaging offers the right tool with theImaging Needs

Solutions• 75% peak QE combined with low noiseelectronics reveals the weak signals missedby industrial cameras

Extreme Low
Light Imaging• Ultra low noise readout mode enables
exposure times of up to 60 minutes.
• Deep sensor cooling and Dynamic Dark
Frame Correction (DDFC) deliver amazing data

Retiga LUMO when that story involves bioluminescence. With theRetiga LUMO from Teledyne Photometrics, you get a 6.0 MP
camera tuned to excel
bioluminescence imaging.

attheuniquechallengesofThe Retiga LUMO is packed with advanced technical featuresthat enable detection and quantification of ultra-low lightluminescence signals. This is accomplished by coupling deep
cooling with FPGA-based intelligent features that correct
defective pixels and remove accumulated dark current. The result
is a camera that outperforms bioluminescence cameras morethan twice the price.

overRapid Findand Focus

those long exposures.

Inside the LUMO camera, Teledyne Photometrics relies on• 50MHz two port readout delivers frame rate

Intelligent Quantification™ - on camera intelligence features that
correct for defective pixels. Fast 50 MHz pixel digitization
increases camera frame rate to give you the speed you need tofind focus, then uses a 650kHz readout mode for ultra-low noisedata collection.

for finding, focusing and imaging samples• Reduce photobleaching and phototoxicity onsamples.
• Intelligent Quantification provides advanced
real-time FPGA algorithms to deliver better
image quality

Flawless Images• Defective Pixel Correction (DPC) and Dynamic

A great camera deserves great software for acquisition - it’s theway you interface with your data. Ocular™ is the all new imaging
platform that’s ready to become the go-to capture program inyour lab. Built around controls you are already familiar with, it will
be love at first click.

Dark Frame Correction (DDFC) correct hot
pixels and remove background for high quality
images over difficult long exposures.

Scientific cameras are the cornerstone of the highest performing
imaging instruments in a lab. Through careful selection ofimage sensors and components, the LUMO will redefine yourexpectations for bioluminescence imaging, as well as moreroutine fluorescence applications. You will not find a morecapable bioluminescence camera on the market for this price.
Call us to demo one today.

Retiga LUMOTM CCD

Bioluminescence Optimized

Retiga LUMO™ Specifications
WHY RETIGA LUMO?

CCD Sensor
Sensor TypeSony ICX-695 Scientific Interline CCD (Monochrome or Color)

CCD Array2688 x 2200

Pixel Size

4.54μm x 4.54μm

Sensor Dimensions

12.5mm x 10mm (16mm diagonal)

Peak Quantum Efficiency

75% at 600nm

Full Well Capacityup to 27,000 e- readout mode dependent• 6.0MP - incredible field of view
• Built for bioluminescence imaginglong exposures, deep cooling, andactive image correction• Proven technology - built on Sony
ICX695 sensor

CameraDigital Output

16-bit with 650kHz readout; 14-bit with 50MHz readout

Digitization Rate

50MHz high frame rate, 650KHz low noise digitization

Read Noise (typical)<4.5e- RMS with 650KHz readout

Frame Rate

7.1 fps (full resolution)
12.8 fps (binned 2x2)

Exposure Time Range

25μs - 60min

Supported Binning Modes

1x1, 2x2, 4x4, 6x6, 8x8, 12x12, 16x16, 24x24

Dark Current Rate (typical)

0.0004 e-/p/s at -20ºC regulated
0.0001 e-/p/s at -20ºC with DPC and DDFC on

Sensor Cooling-20ºC stabilized at 22ºC ambient
Thermoelectric cooling with forced airIntelligent Quantification
Features

DPC - Defect Correction
DDFC - Dynamic Dark Frame Correction• Ocular - powerful and intuitive
capture software• Service - unparalleled sales andsupport personnel• Accelerate discovery - fit more into
each frame

Accessories (Included)• Power Supply• USB 3.0 Cable• Trigger Cable

Interfacing• Ocular Imaging Software

Computer Platforms/
Operating Systems

Windows 7 (64 bit), Windows 8 (64 bit), Windows 10 (64 bit)
Refer to the Teledyne QImaging website for the latest list ofminimum computer recommendations

Digital Interface

USB3.0

Triggering I/O Signals

Trigger In, Expose Out, End-of-Frame, Shutter Out

Supported Triggering Modes

Trigger First, Strobe, Bulb• Access to SDK• Two Year Limited Warranty

Spectral response
Mechanical
Optical Interface

1”, C-mount optical format

Mounting Hole Thread Size

1/4” - 20 thread, 4 sides

Camera Dimensions

98.4mm x 76mm x 76mm (length x width x height)

Weight1.55lb, 0.72kg

Power Requirement

12V, 5A

Note: Specifications are typical and subject to change.
Ocular, Retiga, and Retiga LUMO logo are trademarks of Teledyne
Photometrics Corporation.

Teledyne

Photometrics

isa

registeredtrademark of Teledyne Photometrics Corporation. Other brand andproduct names are the trademarks or registered trademarks of theirrespective owners and manufacturers.

www.photometrics.com
info@photometrics.com / tel: +1 520.889.9933
QI_RETIGA_LUMO_DS_Rev_A3