Prime 95B 22mm™
Scientific
Camera Datasheet
Datasheet
MicroPublisher
6TMCMOSCCD Camera
Scientific CMOS,
Scientific
EMCCDImaging
and CCD
MadeCamerasEasy

The color image solution for
documentation and publication
The Micropublisher 6™:
Big on Field of View
and Accurate Color
Imaging NeedsSolutions• 6 Million Pixels deliver detail and quality

Versatile Pixel Options• Small pixels designed to best match

The MicroPublisher 6 color image solution for documentationand publication. This 6.0 MP camera delivers high resolution
documentation images with a large field of view and accurate
color representation. The camera gives you the freedom tocapture more of what you’re seeing, with more detail andimproved quality than possible with other color cameras.

microscope resolving power• 75% peak QE combined with low noiseelectronics reveals weak signals missed by
Fluorescence Imaging

industrial cameras

Capability• Image capture capability in color or canbe read out as monochrome for tinting andmerging• Large field of view capture - capture twotimes the field of view of a classic color camera• The camera’s large field of view gives you thefreedom to capture more of what you’re seeing• Capture more detail with improved quality
than with other color cameras

The MicroPublisher 6 delivers six million 4.54 micron pixels for
maximum detail capture with low powered objectives. It’s 16mm
diagonal sensor provides twice the field of view of standard color
sensors, giving you the ability to capture more of the sample area
in each image.
The camera also incorporates low read noise electronics, andcooling to remove hot pixels and dark current, making it a suitable
imaging option for high quality fluorescence documentation.
Available upon request, you will receive Ocular™ scientificimage capture software with your MicroPublisher 6 purchasethat provides easy-to-use tools for microscopy and imaging.
Ocular was built from the ground up as a reliable part of daily
research tasks. The software provides a familiar and intuitive user
experience and innovative and easy to use movie modes andsaving functions.

Documentation and Publication

MicroPublisher 6TM CCD

MicroPublisher 6™ Specifications
High Definition Imaging

CCD SensorSensor TypeSony ICX-695 Scientific Interline CCD (Color)

CCD Array2688 x 2200

Pixel Size

4.54μm x 4.54μm

Sensor Dimensions

12.5mm x 10mm (16mm diagonal)• 6.0MP - incredible field of view
• The ideal camera for color documentationand publication• Suitable for fluorescence microscopy

CameraDigital Output

14-bitRead Noise (typical)<6e- RMS• Feature rich - cooled, fast focusing andIntelligent Quantification

Frame Rate

7.1 fps 6 Million Pixels Color
7.1 fps 6 Million Pixels Monochrome
12.81 fps (Bin 2) 1.5 Million Pixels Monochrome• Ocular image acquisition software included
upon request

Dark Current Rate (typical)

0.0007 e/p/s at -12ºC regulated

Sensor Cooling-12ºC stabilized at 22ºC ambient
Thermoelectric cooling with forced air• Service - unparalleled sales and support
personnel• Accelerate discovery - fit more into eachframe

Interfacing
Computer Platforms/
Operating Systems

Windows 7 (64 bit), Windows 8 (64 bit), Windows 10 (64 bit)
Refer to the Teledyne QImaging website for the latest list ofminimum computer recommendations

Digital Interface

USB3.0 (USB2 compatible at reduced max fps)

Included• MicroPublisher 6 Scientific CCD Camera• Power Supply

Mechanical• USB 3.0 Cable

Optical Interface

1”, C-mount optical format

Mounting Hole Thread Size

1/4” - 20 thread, 4 sides• Ocular™ Imaging Software

Camera Dimensions

98.4mm x 76mm x 76mm (length x width x height)• Two Year Limited Warranty

Weight1.55lb, 0.72kg

Power Requirement

7.5V DC, 2.5A

Spectral response

Note: Speci ications are typical and subject to change.
Teledyne Photometrics is a registered trademark, and MicroPublisher 6and Ocular are trademarks of Teledyne Photometrics. All other brandand product names are the trademarks of their respective owners.

www.photometrics.com
info@photometrics.com / tel: +1 520.889.9933
QI_MicroPublisher_6_DS_Rev_A2