Pen-type Electronic Pipette pipetty (MSIC01-03)
Pen-type Electronic Pipette with wireless communication function pipetty Pro (MSIC04-01)

pipetty Series Instruction Manual

Instruction Manual Ver. 1.0.3
2022.4.22

Contents

1

1

1.1

Introduction ……………………………………………………………………… 1
1.1
1.2

About the Instruction Manual ………………………………………………………
Cautions on Using the Pipette ………………………………………………………
1.2.1 Pipetty Main Unit …………………………………………………………
1.2.2 Battery ………………………………………………………………………
1.2.3 Operation ……………………………………………………………………
1.2.4 Wireless Communication …………………………………………………

1
1
1
1
2
2

2

Product Description …………………………………………………………… 3

3

Preparation ……………………………………………………………………… 5

2.1
2.2
2.3
2.4
2.5

3.1
3.2
3.3
3.4
3.5

Appearance ……………………………………………………………………………
Display …………………………………………………………………………………
Package Contents ……………………………………………………………………
Product Features ………………………………………………………………………
Parts Materials …………………………………………………………………………

Charging Battery ………………………………………………………………………
Installing and Removing the Battery ………………………………………………
Checking Model Information …………………………………………………………
Turn On / Turn Oﬀ ……………………………………………………………………
Attach and Release the Tip ……………………………………………………………

3
3
4
4
4

5
5
5
5
5

4

How to Operate ………………………………………………………………… 6

5
6
7
8
9
10

Maintenance…………………………………………………………………… 11
Troubleshooting ……………………………………………………………… 11
Storage ………………………………………………………………………… 12
Product Speciﬁcation ………………………………………………………… 12
Warranty Policy ……………………………………………………………… 13
Disposal ………………………………………………………………………… 13

4.1
4.2
4.3
4.4
4.5
4.6
4.7
4.8

Selecting the Mode …………………………………………………………………… 6
M (Multi) Mode………………………………………………………………………… 6
S (Single) Mode (Mixing) …………………………………………………………… 7
MC (M-Mode Calibration) Mode …………………………………………………… 8
SC (S-Mode Calibration) Mode ……………………………………………………… 9
SEL (Select) Mode …………………………………………………………………… 10
SM (Super Multi) Mode ……………………………………………………………… 10
Memory function ……………………………………………………………………… 10

Introduction
About the Instruction Manual

Read this instruction manual before using this product for the ﬁrst time.

It is recommended that you print this instruction manual and keep it near the product.
This instruction manual corresponds to software version Pn/PP/PS-6. **.

1.2

Cautions on Using the Pipette1.2.1 Pipette Main Unit
【WARNING】●This product cannot be used in a potentially explosive environment. Do not use explosive substances.
●Do not use radioactive substances or liquid containing radioactive substances.
●Do not use a liquid such as a medical solution that directly enters the human body.
●Do not disassemble or modify this product.
【CAUTION】●Although this product has excellent chemical resistance, do not use strong acidic or strong alkaline
substances or liquids such as acetone and trichloroethylene, as they may cause damage to the main unit.
●This product is not waterproof. Please note that adhesion of water droplets including conductive liquid may
cause malfunction.
●Be sure to thoroughly investigate applicable environmental laws and regulations, such as the RoHS Directivethat regulates the use and inclusion of speciﬁc substances, when using the product so that it complies withsuch laws and regulations.

1.2.2 Battery【WARNING】●Use the battery speciﬁed by our company for this product.
●Do not disassemble or modify the battery.
●Do not use leaked or damaged batteries. Dispose of them in accordance with regulations.
【CAUTION】●If the battery is used for the ﬁrst time or if it has not been used for a long period of time, the performance ofthe battery cannot be fully exhibited by charging alone and the number of operations may be reduced,
however, discharging the battery to indicator scale 1 as a guide and then charging it for a couple of times
may solve the problem.
●If the battery is repeatedly charged before running out, a phenomenon called the “memory eﬀect”, whichdrops the discharge voltage, occurs and the number of operations decreases. In this case, the problem may
be solved by discharging the battery to indicator scale 1 as a guide and then charging it for a couple of times.
●The battery has a life cycle, and the number of operation times decreases as charging and discharging arerepeated. If so, replace the battery with a new one.
●The standard operation times of the battery which reached the life cycle is about 60% of the case of using a newbattery.
●Recommend the use of operation guaranteed battery for this device. (AAA Nickel Hydride Battery)

Copyright © Icomes Lab Co., Ltd.
Icomes Lab, Icomes Lab logo, and pipetty are the registered trademarks of Icomes Lab Co., Ltd. Other company names and product names are trademarks or registeredtrademarks of their respective companies.

− 1 −1.2.3 Operation【WARNING】●Do not dispense toward the human body. Some liquid can be harmful to the human body.
●If harmful substances adhere to the main unit, take appropriate detoxiﬁcation measures before use.
●Do not eject the tip toward the human body.
●Remove the battery when the display is oﬀ, or when the mode selection display is displayed, or when START
is ﬂashing.
【CAUTION】●Use a tip that meets the speciﬁed capacity.
●Do not turn the tip upwards or vibrate while the liquid is in the tip. This may cause the liquid to enter the unit
and damage to the main unit.
●The tip must be changed after each dispensing. Repeated use may cause a decrease in accuracy.
●Use the tip ejector to remove the tip. If removed by hand, the nozzle will loosen, and the dispensing accuracy
will decrease due to leakage.
●When dispensing, keep the liquid, pipette body, and tip all at the same temperature. If the liquid temperature
diﬀers between the pipette body and the tip during dispensing, accuracy may be reduced.
●The accuracy guarantee range diﬀers for each model. When working with settings outside the guaranteed
accuracy range, please note that the dispensing volume may not be accurate.

2
2.1

Appearance
❼

❷

❺

❶
❽

❻

1.2.4 Wireless Communication (only models with wireless function: pipetty Pro (MSIC04-01)/pipetty Smart (MSIC12-01))

This product incorporates radio equipment that has been certified for construction type as a radio
station for low-power data communication systems based on the Radio Law. EYSHCN: 001-A10745【WARNING】●Do not use this product near medical devices such as cardiac pacemakers and hearing aids. Electromagnetic
interference may cause life-threatening danger.
●The Bluetooth module installed in this product has received technical conformity certiﬁcation (certiﬁcate ofconstruction type) as a radio station radio equipment for low-power data communication systems based on
the Radio Law. Therefore, you do not need permission from a radio station to use this product, however, if you
do the following, you may be punished by law.
・Modify the installed Bluetooth module.
・Peeling oﬀ the certiﬁcation label attached to the installed Bluetooth module.
【CAUTION】●The frequency band of the Bluetooth module installed in this product is compatible with local radio stations
for identifying mobile units and speciﬁc low-power radio stations used in such as medical equipment,
microwave ovens, industrial/scientiﬁc instruments, and production lines in factories. Since they are in thesame frequency band, communication interference due to radio wave interference may occur.
●If wireless communication is performed with the battery indicator scale set is less than 2, a communication
failure may occur. In this case, charge the battery before operating.
●In an environment with a device that generates electromagnetic noise around, regardless of the remaining
battery level, communication disturbance may occur due to the eﬀect of electromagnetic noise.

Product Description❸❹❶ PUSH button …………………… Power on, aspirate and dispense operations. Same function as the BACK button.
❷ UP or DOWN button …………… Switch between number and mode.

To turn oﬀ the power at anarbitrary timing, press and hold the DOWN button❸ MODE／ENTER button ………… Set the number and mode.

❹ LCD ……………………………… The set number and mode are displayed.
❺ BACK button…………………… Power on, aspirate and dispense operations. Same function as the PUSH button.

❻ Nozzle …………………………… Attach the tip.
❼ Tip ejector……………………… Slide to eject a tip.

❽ Battery cover …………………… Remove it to replace the battery.

2.2

Display
Battery indicatorModes●M（Multi）Mode ……………… Equal volume continuous dispensing●S（Single）Mode (Mixing) ……… Quantitative dispensing, mixing, blowout●SM（Super Multi）Mode ……… Continuous dispensing of diﬀerent volumes *1●MC（Multi-calibration）Mode …… M mode calibration●SC（Single Calibration）Mode …… S mode calibration●SEL（Select）Mode …………… Speed/sound settings
Dispensing volume
Mixing volume or dispensing times

START (Flash when dispensing is ready)*1 This mode is dedicated to the application of pipetty with wireless function. It cannot be used with the pipetty (MSIC01-03).

− 2 −

− 3 −

2.3

2.4

Package Contents(1) pipetty main unit
(2) Quick start guide(3) Inspection report(4) Warranty

3

1
1
1
1

Product Features

2.5

3.1

Charging Battery

3.2

Installing and Removing the Battery

3.3

Checking Model Information

The pen-type electronic pipette pipetty series, equipped with Icomes Lab micro-actuator (driving unit), was developed
as *the world's smallest and lightest electronic pipette that can be used like holding a pen. Adopting the air displacment
method, in addition to quantitative dispensing (forward pipetting) and equal volume continuous dispensing,mixing (pipetting) is also available. Moreover, continuous dispensing of diﬀerent volumes is possible for models with awireless communication function. Not only there is an automatic volume correction function for hand warming tostabilize dispensing accuracy, but also it is possible to operate even with the conventional grip style so that you canselect the gripping method, which can reduce the workload.
* As of February 2020, based on our research

Parts Material

The main components used in the pipetty series are as follows.
Item

Materials

Case

ABS

Tip ejector

ABS

BACK button

ABS

Battery coverABS

PUSH button, UP button, DOWN button, MODE/ENTER button

PET

NozzlePOM

Preparation●The battery is not fully charged at the time of purchase. Use it after charging with the dedicated charger.
●Charge the battery after using it for a certain period of time. Frequent charging may reduce the number of operations
or cause unstable communication due to the memory eﬀect of the battery.
●pipetty : The standard time to charge the battery is when it begins to show “2” to “1” on the scale.
pipetty Pro : The standard time to charge the battery is when it begins to show “3” to “2” on the scale to stabilize thewireless communication.

Remove the battery cover and attach the battery. Remove the battery when the display is oﬀ, the mode selection
display, or when START is ﬂashing.
*Use our designated battery and do not use other dry batteries.

After the battery is installed the model information is displayed on the upper line of the LCD, and the software version
is displayed on the lower line. (Fig. 1)

Model information display
Pn: Pen-type electronic pipette
pipetty (MSIC01-03)
PP: Pen-type electronic pipette with wireless communication function

pipetty Pro (MSIC04-01)

Software version display example 700: Ver7.00（Fig. 1）

3.4

Power On /Power Off○Power ON
When the screen display is oﬀ, pressing the PUSH button or the BACK button turns on the power and performs theinitialization operation automatically.
○Power OFF
The power is automatically turned oﬀ when the mode select screen or the indicate START (*1) on the screen is left on for
1 minute.
To turn oﬀ the power at an arbitrary timing, press and hold the DOWN button when the mode select screen or theindicate START (*1) on the screen.
*1 When the START indicate while the Mixing or Blowout state, the power is not turned oﬀ because the operation is not
yet completed.

3.5

− 4 −

Attach and Release the Tip

Attach the tip to the nozzle. Refer to “8. Product Speciﬁcations” for compatible tips.
Use a tip ejector to remove the tip. If removed by hand, the nozzle will loosen, and dispensing accuracy will decrease.

− 5 −

4
4.1

How to Operate

4.3

S (Single) Mode (Mixing)

S (Single) Mode performs quantitative dispensing. This mode dispenses the aspirated volume at
once.
1.
2.
3.
4.
5.

Selecting the Mode

The display flashes M. Use the UP or DOWN button to switch the mode and use the MODE/ENTER button todetermine the mode.

Flash

Flash

Flash
（Fig. 1）

4.2

（Fig. 2）

●You can switch the dispensing volume by pressing and holding either the UP
button when the START flashes.
●Press and hold the MODE/ENTER button when START flashes to return to themode selection display.
●If an error (Err004) is displayed during aspirating, it indicates that the pipette unit
has exceeded the normal operating temperature range. Suspend the work andplace the pipette unit in a cool place for a certain period of time or reduce thedispensing volume and set.

M (Multi) ModeIn the M (Multi) Mode, the equal volume continuous dispensing is performed. In this mode,
the aspirated volume is dispensed at the set volume for the set times.
1.
2.
3.
4.
5.
Flash

6.
Flash

（Fid. 1）

（Fig. 2）

（Fig. 3）

Use the UP or DOWN button to change the dispensing volume. (Fig. 1)
When the desired volume is reached, press the MODE/ENTER button to confirm.
Use the UP or DOWN button to change the dispensing times. (Fig. 2)
When the desired times is reached, press the MODE/ENTER button to confirm.
The setting is completed when START flashes and dispensing preparation iscompleted. (Fig. 3)
Aspirate by pressing the PUSH or BACK button.
After aspirating, press the PUSH or BACK button to discharge the set times. Whenthe set times of dispense is completed, the dispensing volume and the dispening
times will flash, then press the PUSH or BACK button twice. After discharging theremaining liquid, it automatically operates, START flashes, and returns to thestate when dispensing preparation is completed.

●All dispense will be performed halfway if the MODE/ENTER button is pressed before
the dispensing times reaches “0”. Alter all dispense is performed halfway, press the
PUSH or BACK button as the dispensing volume and the dispensing times will flash.
START flashes and returns to the state when dispensing preparation is completed.
●You can change the dispensing volume and dispensing times by pressing andholding the UP button when START is flashing.
●To return to the mode selection display, press and hold the MODE/ENTER button
when START is flashing.
●The automatic correction function for hand warming may change the dispensing
times or limit the times.
●If an error (Err004) is displayed when setting the dispensing volume, it indicates thatthe pipette unit has exceeded the normal operating temperature range. Suspend
the work and place the pipette unit in a cool place for a certain period of time or
reduce the dispensing volume and set.
●It may take several seconds for the dispensing times to be displayed due to thecalculation of the number of dispenses.

（Fig. 3）

Use the UP or DOWN button to change the dispensing volume. (Fig. 1)
When the desired volume is reached, press the MODE/ENTER button to confirm.
Use the UP or DOWN button to change the mixing volume. (Fig. 2)
When the desired volume is reached, press the MODE/ENTER button to confirm.
The setting is completed when START flashes and dispensing preparation iscompleted. (Fig. 3)
6. Aspirate by pressing the PUSH or BACK button.
7. After aspirating, dispense by pressing the PUSH or BACK button.
8. After dispensing, it operates automatically, START flashes, and returns to thestate when dispensing preparation is completed.
*Be careful not to aspirate the dispensed liquid since the suction operation is
performed automatically after discharging.

Mixing repeats the aspirating and dispensing operation.

1. Pressing the UP button while START is flashing, and it switches to the Mixing Mode. (Fig. 4)
2. To perform mixing, press the PUSH or BACK button.
3. To end mixing, press the PUSH or BACK button again during mixing.

（Fig. 4）●After mixing is completed, it automatically returns to the START flashing.
●After switching the mode by pressing the UP button, press the DOWN button to return to the START
flashing.
●If an error (Err004) is displayed during aspirating, it indicates that the pipette unit has exceeded thenormal operating temperature range. Suspend the work and place the pipette unit in a cool place for acertain period of time or reduce the mixing volume and set.

Blowout is an operation to efficiently expel the liquid adhering to the inner wall of
the tip.
【Blowout】
1. In the mode select screen, press the MODE/ENTER button and the PUSH button simulta
neously to switch between ON and OFF. (Fig. 5)
2. The S mode screen changes to "bo (Blowout)" after aspirating and dispensing the set
dispensing volume. (Fig. 6)
3. Press the PUSH button or the BACK button to perform Blowout and exit.
●After Blowout is complete, the display automatically returns to START.
●When Mixing is performed with Blowout ON, the indication automatically switches to"bo (Blowout)" after Mixing is finished. (Fig. 6)（ﬁg. 5）（ﬁg. 6）【Additional Blowout】
If dispense out the remaining liquid in the tip, additional Blowout can be performed by thefollowing operation regard less of the Blowout ON/OFF setting.
1. In S mode, after completion of dispensing, press the DOWN button to switch to additional
Blowout when the indicate START. (Fig. 6)
2. Additional Blowout is performed by pressing the PUSH button or the BACK button.
For more additional Blowout, press the PUSH button or the BACK button to repeatedly
perform an additional Blowout.
3. Press UP button to end additional Blowout.
●After the additional Blowout is completed, the display automatically returns to the START.

− 6 −

− 7 −

4.4

MC (M-Mode Calibration) Mode

4.5

SC (S-Mode Calibration) Mode

MC (M-Mode Calibration) Mode is for calibrating the dispensing volume in

SC (S-Mode Calibration) Mode is a mode for calibrating the dispensing volume in

M (Multi) mode and SM (Super Multi) mode. Calibration is performed at the time

S (Single) Mode. Calibration is performed at the time of shipment.

of shipment.

●When selecting the MC (M-Mode Calibration) Mode, press and hold the MODE/ENTER button.
●To perform calibration, prepare an electronic balance that can measure to the third decimal place.
●If you select a wrong mode and wish to cancel it, press the MODE/ENTER button three times and press and hold the
MODE/ENTER button while START is flashing to return to the mode selection display.

1.
2.
03.
4.

（Fig. 1）

（Fig. 2）

5.
6.
7.
8.

9.

10.
（Fig. 3）（Fig. 4）

11.

12.
13.

The previous calibration set volume on the maximum volume side is displayed.
(Fig. 1 Example: 19.801μl)
Press the MODE/ENTER button to display the previous calibration set volume for theminimum volume. (Fig. 2)
Press the MODE/ENTER button to display the temperature at the calibration. (Fig. 3)*The temperature for calibration is automatically measured and input.
By pressing the MODE/ENTER button, it switches to the calibration setting for the
maximum volume, the dispensing volume is displayed as “Cr1”, the dispensing times
are displayed as “5”. Then START flashes, and preparation for dispensing is completed.
(Fig. 4)
Aspirate by pressing the PUSH or BACK button.
After aspirating, dispense by pressing the PUSH or BACK button.
After dispensing, press the PUSH or BACK button to discharge the remaining liquid.
After dispensing, START flashes, the dispensing times decrease by one, and returns tothe state when dispensing preparation is completed automatically.
*Be careful not to aspirate the dispensed liquid since the suction operation is performed
automatically after discharging.
Repeating steps 4 to 8 five times will switch to the input display. Switch the display volume
with the UP or DOWN button to match the average volume of the dispensing volume.
(Fig. 5)
When the desired volume is reached, press the MODE/ENTER button and enter thecalibration volume for the maximum volume.
It switches to the calibration setting on the minimum volume side, the dispensing volume
is displayed as “Cr2” and the dispensing times are displayed as “5”. START flashes and the
preparation for dispensing is completed. (Fig. 6)
Thereafter, repeat steps 5 to 9 to set the calibration volume on the minimum
volume side.
Press the MODE/ENTER button to input the calibration on the minimum volume side andreturn to the mode selection display.

●When selecting the SC (S-Mode Calibration) Mode, press and hold the MODE/ENTER button.
●To perform calibration, prepare an electronic balance that can measure to the third decimal place.
●If you select a wrong mode and wish to cancel it, press the MODE/ENTER button three times and press and hold the
MODE/ENTER button while START is flashing to return to the mode selection display.

1.
2.
3.
4.

（Fig. 1）

5.
6.
7.

8.

（Fig. 3）（Fig. 4）

9.
10.

11.
12.

（Fig. 5）

（Fig. 5）

（Fig. 2）The previous calibration set volume on the maximum volume side is displayed.
(Fig. 1 Example: 19.801μl)
Press the MODE/ENTER button to display the previous calibration set volume for theminimum volume. (Fig. 2)
Press the MODE/ENTER button to display the temperature at the calibration. (Fig. 3)*The temperature for calibration is automatically measured and input.
By pressing the MODE/ENTER button, it switches to the calibration setting for the
maximum volume, the dispensing volume is displayed as “CF1”, the dispensing times isdisplayed as “5”. Then START flashes, and preparation for dispensing is completed.
(Fig. 4)
Aspirate by pressing the PUSH or BACK button.
After aspirating, dispense by pressing the PUSH or BACK button.
After dispensing, START flashes, the dispensing times decrease by one, and returns to the state
when dispensing preparation is completed automatically.
*Be careful not to aspirate the dispensed liquid since the suction operation is performed
automatically after discharging.
Repeating steps 4 to 7 five times will switch to the input display. Switch the display volume
with the UP or DOWN button to match the average volume of the dispensing volume.
(Fig. 5)
When the desired volume is reached, press the MODE/ENTER button and enter thecalibration volume for the maximum volume.
It switches to the calibration setting on the minimum volume side, the dispensing volume
is displayed as “CF2” and the dispensing times is displayed as “5”. START flashes and the
preparation for dispensing is completed. (Fig. 6)
Thereafter, repeat steps 5 to 8 to set the calibration volume on the minimum
volume side.
Press the MODE/ENTER button to input the calibration on the minimum volume side andreturn to the mode selection display.

（Fig. 6）

（Fig. 6）

− 8 −

− 9 −

4.6

SEL (Select) Mode
This mode changes the settings for dispensing/aspirating speed and buzzer.
1.

The dispensing speed setting display appears, “SPd” flashes, and the set speed flashes. “Hi”
at the initial setting. (Fig. 1)
2. Press the UP or DOWN button to switch and press the MODE/ENTER button to confirm.
3. The aspirating speed setting display appears, “SPA” flashes, and the set speed flashes. “Hi”
at the initial setting. (Fig. 2)
4. Press the UP or DOWN button to switch and press the MODE/ENTER button to confirm.
5. The buzzer setting display appears, “bEP” flashes, and the setting status flashes. “Hi” at theinitial setting. (Fig. 3)
6. Press the UP or DOWN button to switch and press the MODE/ENTER button to confirm,then it returns to the mode selection display.

Flash

Flash

Flash

（Fig. 1）

（Fig. 2）

（Fig. 3）

4.7

SM (Super Multi) Mode
(only models with wireless function. pipetty Pro (MSIC04-01))
The SM (Super Multi) Mode performs continuous dispensing of different volumes
and dispenses the aspirating volume according to the set dispensing volume.

●This is an application-only mode and cannot be operated without wireless communication.
●This mode cannot be used with pipetty (MSIC01-02).
●See the application manual for the operation method.

4.8

MaintenanceIf the pipette case gets dirty, wipe it off with a cloth impregnated with 70 vol% ethanol, containing isopropanol as
an additive.
Please note that using a chemical solution other than those specified may result in damage to the main unit.

6

Troubleshooting
Contact your dealer or our customer service if the problem cannot be solved by the following method or otherabnormalities that occur. We will analyze and repair as necessary.
Problem

Err002 is displayed

Err004 is displayed

Memory function
The past dispensed history can be selected in S mode and M mode.
1.
2.
3.

Flash

The past dispensed history is displayed by press PUSH button or BACK button while S or M isblinking. (Fig. 1)
Press MODE/ENTER to decide the dispensing volume from the dispensing history.(fig. 2）
Setting is complete and ready for dispensing when indicate START. (Fig. 3)●The dispensed history can be recalled up to 6 times in the past.
●The speed of aspiration and dispensing is not recorded.
Flash（ﬁg. 1）

5（ﬁg. 2）

Solution

Motor operation error

Press the PUSH or BACK button

Insufficient battery power

Replace with a charged battery

Battery degradation

Replace with a new battery that has been
chargedThe pipette unit has exceeded thenormal operating temperature range.

uspend the work and place the pipetteunit in a cool place.

Loose nozzle

Retighten the nozzle

Tip is not installed correctly.

●Check if the tip is applicable.
●Reattach the tip.

There is residual liquid in the tip in
S (Single) mode.

●Discharge all the remaining liquidin the tip.
●Replace the tip.

Tip is not in contact with liquid at
the time of dispensing in M (Multi) modeor SM (Super Multi) mode.

Dispense with the tip in contact
with the liquid.

ispensing with settings outside theguaranteed accuracy range.

Dispense with settings within theguaranteed accuracy range.

The temperature of the pipette unit,
liquid, and tip differ greatly.

Leave the pipette unit, liquid, and tip in
the same environment and work at
a constant temperature.

Using a liquid with different physicalproperties such as density from water.

Re-calibrate.

Battery has reached a life cycle.

Replace with a new battery. Refer to 8.
Product Specifications: Battery

Abnormal noise andmalfunction

Using in an environment outside thespecification range.

Use in an environment between
10°C and 30°C.

Power does not turn on

Battery mounting direction is reversed.

Install the battery correctly.

Liquid leak from tip/
incorrect dispensing
volume（ﬁg. 3）− 10 −

Possible reason− 11 −

7

9

StorageRemove the battery and store it if the product will not be used for a long time.
If the product is stored with the battery attached, it may be over-discharged, and the battery may not be able toexhibit its performance.

8

Warranty policy
1.

The repair will be free of charge if the unit breaks down within the warranty period under the usage conditions
according to the “Cautions on Using the Pipette” in the instruction manual. If any problems occur, contact
your distributor.

2.

In accordance with Article 25 of the Ministry of Health, Labor and Welfare Ordinance No.169, we request the issuanceof a safety confirmation letter for the purpose of ensuring the safety of technicians and related persons in contact
with the target equipment when requesting repair including free repair. Hand over the product, safety confirmation,
and warranty card to the retailer from whom you purchased the product and let us know the location of the failure.

3.

Please contact our customer service or distributor in your country for repair requests in case of relocation.

4.

Repairs will be charged in principle even within the warranty period for the following cases.
●Failure or damage due to incorrect use, improper repair or modification.
●Failure or damage due to transportation including dropping after purchase.
●Failure or damage due to fire, earthquake, flood, lightning, or other natural disasters.
●Failure or damage caused when loading on vehicles, ships, etc.
●Without warranty card submission.
●When the date of purchase, customer name, retailer name is not entered in the warranty, or the letters are
rewritten.
●In case of malfunction due to the consumption of parts.

5.

International warranty varies by each country. Please contact the distributor which you have purchased the product.

6.

The warranty will not be reissued. Please keep it nearby.

Product Specifications
Product namePen-type electronic pipette pipetty (MSIC01-03)
Pen-type electronic pipette with wireless communication function
pipetty Pro (MSIC04-01)

Dispensing range

0.1−20μl

1−250μl5−1000μl

Accuracy guarantee range

2−20μl20−250μl

100−1000μl

Reproducibility() indicates continuous
dispensing

20μl≦0.4%2μl≦2.0%（2μl≦3.5%）250μl≦0.15%20μl≦0.8%（20μl≦2.0%）1000μl≦0.15%
100μl≦0.5%（100μl≦1.5%）20μl±1.0%2μl±3.5%（2μl±5.0%）250μl±0.5%20μl±2.5%（20μl±3.0%）

1000μl±0.5%100μl±1.5%（100μl±2.0%）

WATSON

10μl pipette tip (207)200μl pipette tip (703)300μl pipette tip (505)1000μl pipette tip (806)1200 pipette tip (706)

GILSON−

DL300 20−300µl

DL1000 100−1000µl

epT.I.P.S.LoRetention
2-200µl

epT.I.P.S.LoRetention
50-1000µl

pipette tip
0.5-20μl

pipette tip
300μl

pipette tip
1250μl★ART micro point tip
0.1-20μl(model number: 3512-05-HR)

ART micro point tip
200μl

ART micro point tip
1000μl−★FINNTIP250Universal
0.5-250μl(model number: 9400250)★FINNTIP
100-1000μl(model number: 9401030)−

UNV RCC-250
2-250μl

UNV RCC-1000
100-1000μl

20×54×185

20×54×181

20×54×181

Accuracy() indicates continuous
dispensing

eppendorf

Applicable tip

Greiner bio-one

Thermo Fisher
Scientiﬁc

RAININ
External dimensions

epT.I.P.S

0.5−20µl

Product weight

Approx. 75g including rechargeable battery

Operating environment

10℃ to 30℃ 30 to 60% RH

Driving system

Micro actuator φ8

Speed setting

5 steps each for aspiration and dispensing

Buzzer setting

ON/OFF switchable

Power

Recommended battery：AAA Ni-MH rechargeable battery（1.2Vdc）

Wireless communication

Bluetooth5.0 (pipetty Smart (MSIC12-01)/pipetty Pro (only MSIC04-01)★ Recommended tip.
*Specifications are subject to change without notice.

− 12 −*The above warranty rules promise free repair under the conditions and conditions specified.
*The above warranty rules do not limit your legal rights.
*If you are uncertain about the period of maintenance and repair parts after the warranty period has expired, pleasecontact our customer service or your retailer.

10 Disposal
This device cannot be recycled as unsorted municipal waste in accordance with theEuropean Directive WEEE (2012/19/EU) on the disposal and reduction of dangerous
goods in electrical and electronic equipment. Dispose of this device separately
according to local recycling regulations.
Battery should be disposed of in accordance with local regulations and should not be
disposed of with household waste.

Customer support and inquiries for pipette products:
Icomes Lab Co., Ltd.

2-4-23, Kitaiioka
Morioka-shi, Iwate, Japan
Postal code: 020-0857
Phone: +81-19-601-8228 FAX: +81-19-601-8227
E-mail: globalinfo@icomes.co.jp
URL: https://www.icomes.co.jp/en− 13 −