PULSE-ANALOG TRANSMITTER

SGJ
SGJW
SGJL
INSTRUCTION MANUALPreface

Thank you for purchasing our SGJ, SGJW or SGJL, Pulse-Analog Transmitter. This manualcontains instructions for the mounting, functions, operations and notes when operating theSGJ, SGJW or SGJL. To prevent accidents arising from the misuse of this instrument, please
ensure the operator receives this manual.

Notes• This instrument should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed, serious injury
or malfunction may occur.
• The contents of this instruction manual are subject to change without notice.
• Care has been taken to ensure that the contents of this instruction manual are correct, but if
there are any doubts, mistakes or questions, please inform our sales department.
• This instrument is designed to be installed on a DIN rail within a control panel. If it is not,measures must be taken to ensure that the operator does not touch power terminals or otherhigh voltage sections.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos Co., Ltd. is not liable for any damage or secondary damage(s) incurred as aresult of using this product, including any indirect damage.
SAFETY PRECAUTIONS (Be sure to read these precautions before using our products.)
The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on circumstances, procedures indicated by
Caution may result in serious
consequences, so be sure to follow the directions for usage.
which may lead to dangerous conditions and cause
Warning Procedures
death or serious injury, if not carried out properly.

CautionProcedures which may lead to dangerous conditions and cause
superficial to medium injury or physical damage or may degrade
or damage the product, if not carried out properly.

Warning• To prevent an electrical shock or fire, only Shinko or qualified service personnel may
handle the inner assembly.
• To prevent an electrical shock, fire, or damage to instrument, parts replacement may
only be undertaken by Shinko or qualified service personnel.

Safety Precautions• To ensure safe and correct use, thoroughly read and understand this manual before using this
instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring
equipment. Verify correct usage after purpose-of-use consultation with our agency or main
office. (Never use this instrument for medical purposes with which human lives are involved.)• External protection devices such as protective equipment against excessive temperaturerise, etc. must be installed, as malfunction of this product could result in serious damage tothe system or injury to personnel. Proper periodic maintenance is also required.
• This instrument must be used under the conditions and environment described in this manual.
Shinko Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring
due to the instrument being used under conditions not otherwise stated in this manual.

Caution with Respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the
manufacture of weapons of mass destruction (i.e. military applications, militaryequipment, etc.), please investigate the end users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.
- 2 -Installation Precautions

CautionThis instrument is intended to be used under the following environmental conditions(IEC61010-1): Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of -10 to 55 (14 to 131 )that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85 %RH• No large capacity electromagnetic switches or cables through which large current is flowing• No water, oil or chemicals or the vapors of these substances can come into directcontact with the unit.
• When installing this unit within a control panel, please note that ambient temperature ofthis unit – not the ambient temperature of the control panel – must not exceed 55(131 ). Otherwise the life of electronic components (especially electrolytic capacitor)
may be shortened.
Note: Avoid setting this instrument directly on or near flammable material even though
the case of this instrument is made of flame-resistant resin.

Wiring Precautions

Caution• Do not leave bits of wire in the instrument, because they could cause a fire andmalfunction.
• When wiring, use a crimping pliers and a solderless terminal with an insulation sleevein which an M3 screw fits.
• Tighten the terminal screw using the specified torque. If excessive force is applied tothe screw when tightening, the screw or case may be damaged.
• This instrument does not have a built-in power switch, circuit breaker and fuse. It isnecessary to install a power switch, circuit breaker and fuse near the instrument.
(Recommended fuse: Time-lag fuse, rated voltage 250 V AC, rated current 2 A)
• For wiring of the AC power source, be sure to use terminals as described in this manual.
If the AC power source is connected to incorrect terminals, the unit will be burnt out.
• Do not apply a commercial power source to the sensor which is connected to the
input terminal nor allow the power source to come into contact with the sensor.
• When using DC voltage and current input, do not confuse polarity when wiring.
• Keep the input/output wires and power line separate.

Operation and Maintenance Precautions

Caution• Do not touch live terminals. This may cause an electrical shock or problems in operation.
• Turn the power supply to the instrument OFF when retightening the terminal or cleaning.
Working on or touching the terminal with the power switched ON may result in severe
injury or death due to electrical shock.
• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)• As the display section is vulnerable, be careful not to put pressure on, scratch or strike
it with a hard object.
- 3 -Characters used in this manual
Indication
Number, /Indication
Alphabet
Indication
Alphabet[: No character is indicated (unlit).]

-1

0

1

2

3

4

5

6

7

8

9

°C

°F

A

B

C

D

E

F

G

H

I

J

K

L

M

N

O

P

Q

R

S

T

U

V

W

X

Y

Z

Contents
Page
1. Model .................................................................................................................................... 5
1.1
Model.......................................................................................................................... 5
1.2
How to Read the Model Label ................................................................................. 6
2. Name and Functions ........................................................................................................... 7
2.1
Front Panel ................................................................................................................ 7
2.2
Display Section ......................................................................................................... 8
3. Mounting .............................................................................................................................. 9
3.1
External Dimensions (Scale: mm) .............................................................................. 9
3.2
Mounting to, and Removal from the DIN Rail ........................................................... 10
4. Wiring ................................................................................................................................. 11
4.1
Lead Wire Solderless Terminal ............................................................................. 11
4.2
Circuit Configuration .............................................................................................. 11
4.3
Terminal Arrangement............................................................................................ 13
4.4
Wiring ....................................................................................................................... 15
5. Display Mode ..................................................................................................................... 17
6. Setting Mode ...................................................................................................................... 19
6.1
Display Transition in Setting Mode ........................................................................... 19
6.2
Input Setting Mode ................................................................................................. 21
6.3
Output 1 Setting Mode ........................................................................................... 23
6.4
Output 2 Setting Mode ........................................................................................... 25
6.5
Instrument Setting Mode ....................................................................................... 27
6.6
Communication Setting Mode ............................................................................... 28
6.7
Custom Display Setting Mode ............................................................................... 30
6.8
Manual Mode ........................................................................................................... 30
7. Adjustment......................................................................................................................... 31
7.1
Basic Operation of Adjustment ............................................................................. 31
7.2
Adjustment .............................................................................................................. 31
7.2.1
Output 1 Adjustment ....................................................................................... 31
7.2.2
Output 2 Adjustment ....................................................................................... 31
8. Operation ........................................................................................................................... 32
8.1
Indication after Power-on ...................................................................................... 32
8.2
Operation ................................................................................................................. 32
8.2.1
Input Indication Range.................................................................................... 32
8.2.2
Indication Range of Output 1 and Output 2.................................................. 32
8.2.3
Indication Time ................................................................................................ 32
8.2.4
Detecting Unconnected Sensor ..................................................................... 32
9. Specifications .................................................................................................................... 33
10. Troubleshooting .............................................................................................................. 36
10.1 Indication ................................................................................................................. 36
10.2 Key Operation ......................................................................................................... 36
10.3 Operation ................................................................................................................. 37
11. Character Table................................................................................................................ 38- 4 -1. Model
1.1 Model
SGJ
SGJ–F□□–0–□
Option0: No option needed
1: Multi-rotation trimmer
2: Moisture-proof treatment
3: Multi-rotation trimmer + Moisture-proof treatment
Power supply
100 to 240 V AC
Output *2
Input *1
Series name

SGJW
SGJW-F□□□ -0-□

Option0: No option needed
1: Multi-rotation trimmer
2: Moisture-proof treatment
3: Multi-rotation trimmer + Moisture-proof treatment
Power supply
100 to 240 V AC
Output 2 *2
Output 1 *2
Input *1
Series name
SGJL
SGJL–F□□–0-□
Option0: No option needed
1: Multi-rotation trimmer
2: Moisture-proof treatment
3: Multi-rotation trimmer + Moisture-proof treatment
Power supply
100 to 240 V AC
Output *2
Input *1
Series name (with Serial communication)- 5 -*1: Input
Code
F0
F1
F2
F3

Input Type0.001 Hz to 100 kHz
0.001 Hz to 100 kHz
0.001 Hz to 30 Hz
0.001 Hz to 100 kHz

Open collector
Voltage pulse
Mechanical contact
Line driver*2: Output 1, Output 2
Code
Output Type
Code
Output Type1
4 to 20 mA
A
0 to 10 mV*1
2
0 to 20 mA*1
B
0 to 100 mV*1
3
C
0 to 1 V*1Current output 0 to 16 mA*1
4
2 to 10 mA
D
0 to 5 V*1
Voltage output
1
5
0 to 10 mA*E
1 to 5 VF
0 to 10 V*1
G-5 to 5 V*2*1: 0 V or less: Out of base accuracy*2: Not available for the SGJW.
1.2

How to Read the Model Label

The model label is attached to the left side of the case.

Terminal arrangement
Model (e.g. SGJW-F011-0-0)
Input (Enter a type selected in [Input type].)
Output 1 (Enter a type selected in [Output 1 type].)
Output 2 (Enter a type selected in [Output 2 type].)
Serial number(Fig. 1.2-1)- 6 -2. Name and Functions
2.1

Front Panel
SGJ, SGJL

SGJW

①

①

②

②

③

③

⑦

④

④

⑧

⑤

⑦

⑤

⑨

⑥

⑧

⑥

⑩

(Fig. 2.1-1)① Display section② Mounting screw③ DISP key④ MODE key⑤ Up key⑥ DOWN key⑦ Output 1 Zero
⑧ Output 1 Span
⑨ Output 2 Zero*⑩ Output 2 Span*

Indicates setting contents, input value, output value, etc.
Used for fixing the instrument to the socket or removal from it.
Switches the displays, and moves to the next setting item.
In Manual mode, Output 1 and Output 2 setting can be switched.
Releases the lock status of the DISP key by pressing for 3 seconds.
Selects either a setting mode or a display mode.
Shifts the digit for the Custom Display.
Enters the setting mode by pressing and holding for 5 seconds.
Increases the numerical value.
Contents of Multi-Display A and B can be changed alternately when
Default Display is RUN display mode 1, 2*, 3, 4*, 5 and 6*.
Decreases the numerical value.
Enters Manual mode by pressing for 3 seconds.
Adjusts the value of Output 1 Zero.
Adjusts the value of Output 1 Span.
Adjusts the value of Output 2 Zero.
Adjusts the value of Output 2 Span.

* For the SGJW only
- 7 -

2.2

Display Section
SGJ，SGJL
①
②
③
④
⑤
⑥
⑦
⑧
⑨
⑩

SGJW
⑪

⑬
⑭
⑮
⑯

⑱

①
②
③
④
⑤
⑥
⑦
⑧
⑨
⑩

⑪
⑫
⑬
⑭
⑮
⑯
⑰
⑱

(Fig. 2.2-1)① Setting display
indicator A

Lights up in Manual mode.

② Input indicator A

Lights up when Multi-Display A indicates an input value.

③ Alarm indicator A

Lights up if an input error or input disconnection occurs in adisplay mode except All unlit display mode.

④ Output indicator A Lights up when Multi-Display A indicates an output value.
⑤ mA indicator

Lights up when mA is selected in [Indication unit].

⑥ % indicator

Lights up in Manual mode or when % is selected in [Indication unit].

⑦ Setting display
indicator B

Lights up for the setting display.
For the SGJW, lights up for the setting display or in Manual mode.

⑧ Input indicator B

Lights up when Multi-Display B indicates an input value.

⑨ Alarm indicator B

Lights up if an input error or input disconnection occurs while
Multi-Display B indicates an input value.

⑩ Output indicator B Lights up when Multi-Display B indicates an output value.
⑪ 1 indicator A

Lights up in Manual mode or when Multi-Display A indicates

Output 1 value.
⑫ 2 indicator A

Lights up when Multi-Display A indicates Output 2 value.

⑬ Multi-Display A

Indicates the following in accordance with the display indication:
Input value, output value, custom characters, setting item

⑭ V indicator

Lights up when V is selected in [Indication unit].

⑮ °C indicator

Lights up when °C is selected in [Indication unit].

⑯ 1 indicator B

Lights up when Multi-Display B indicates Output 1 value.

⑰ 2 indicator B

Lights up when Multi-Display B indicates Output 2 value.

⑱ Multi-Display B

Indicates the following in accordance with the display indication:
Input value, output value, custom characters, setting value

Output indicators A and B, Alarm indicators A and B: Red
Other indicators: White- 8 -3. Mounting
3.1

External Dimensions (Scale: mm)

8P socket (SGJ, SGJL)

11P socket (SGJW)DIN rail(Fig. 3.1-1)- 9 -DIN rail3.2

Mounting to, and Removal from the DIN Rail

Caution• Mount the DIN rail horizontally.
• To remove the socket, a flat blade screwdriver is required.
Never turn the screwdriver when inserting it into the Lock lever. If excessive power
is applied to the lever, it may break.
• If the instrument is mounted in a position susceptible to vibration or shock, mountcommercially available fastening plates at both ends of the instrument.
Recommended Fastening Plate
Manufacturer
Model
End plate
PFP-M
Omron Corporation
Fastening plate BNL6
IDEC Corporation
Fastening plate ATA4806
Panasonic Electric Works Co., Ltd.
Mounting to the DIN rail (Fig. 3.2-1)① Separate the instrument from the socket by loosening the mounting screw on thefront panel.
② Make sure the lock lever of the socket is located in the lower part of the socket.
Hook the upper side of the socket onto the DIN rail, then fit the lower part of thesocket onto the DIN rail (A clicking sound should be heard when done properly).

Caution• Before inserting the instrument to the socket, make sure the cable is wired properly.
(Refer to “4. Wiring”.)• When inserting or removing the socket, make sure the socket is oriented vertically.
If force is applied in any other direction than vertically, a malfunction may occur.
• If the mounting screw is fastened too tightly, a malfunction may occur.
③ Insert the SGJ into the socket.
④ Fasten the mounting screw by turning it clockwise, to secure the SGJ onto thesocket. Tighten the screw lightly.
Removal from the DIN rail (Fig. 3.2-2)① Turn the power to the instrument OFF.
② Separate the instrument from the socket by loosening the mounting screw on thefront panel.
③ Insert a flat blade screwdriver into the Lock lever (lower part of the socket), andremove the socket from the DIN rail while pulling the lever down.

DIN rail

DIN rail(Fig. 3.2-1)(Fig. 3.2-2)- 10 -

Lock lever4. Wiring
Warning
Turn the power supply to the instrument off before wiring or checking.
Working on or touching the terminal with the power switched on may result insevere injury or death due to electrical shock.
4.1 Lead Wire Solderless Terminal
Use a solderless terminal with an insulation sleeve in which an M3 screw fits as shown
below. The torque should be 0.63 N•m.
Solderless
Manufacturer
Model
Terminal
Nichifu Terminal Industries Co., Ltd.
TMEV1.25Y-3
Y-typeJapan Solderless Terminal MFG Co., Ltd. VD1.25-B3A
Nichifu Terminal Industries Co., Ltd.
TMEV1.25-3
Ring-type
Japan Solderless Terminal MFG Co., Ltd. V1.25-3
Y-type(Scale: mm)
Ring-type(Scale: mm)(Fig. 4.1-1)
4.2 Circuit Configuration
SGJ, SGJL(Fig. 4.1-2)

Front
trimmerInsulation circuit+
Input -

+

①

Input
circuitOutputcircuit

CPU

④
⑤
⑬
⑭

Fixed
voltagecircuit 12VPower
circuit

Display/
Setting key
circuitInsulation circuit
Communication
circuitFor SGJL

Modularjack

When inputting line driver, 1 and 4 terminals are line receiver and 5 terminal issignal ground.
(Fig. 4.2-1)- 11 -

+
⑨
⑫-

SGJW
Front
trimmerInsulation circuit+①
Input -+

Input
circuitCPU

Output 1
circuit

+
⑦
⑧

②
④
⑩
⑪

Fixed
voltagecircuit 12VPower
circuitInsulation circuit

Display/
Setting key
circuitOutput 2
circuit+③⑥-

When inputting line driver, 1 and 2 terminals are line receiver and 4 terminal issignal ground.
(Fig. 4.2-2)- 12 --

4.3 Terminal Arrangement
SGJ
When inputting line driver(Fig. 4.3-1)(Fig. 4.3-2)

SGJL
When inputting line driver(Fig. 4.3-3)(Fig. 4.3-4)- 13 -SGJW
When inputting line driver(Fig. 4.3-5)
PWR
OUT(OUT1)OUT2ININ PWR
RS-485(Fig. 4.3-6)
Power supply 100 to 240 V AC
Output or Output 1 (for SGJW)
Output 2 (for SGJW)
Input
Power for sensor 12 V DC
Serial communication (for SGJL)- 14 -

4.4

WiringWarning• For 100 to 240 V AC, if the AC power source is connected to incorrect terminals,
the instrument will be burnt out.
(1) Power Source Wiring
SGJ, SGJL: Use terminals ⑬, ⑭ for the power supply to the instrument.
SGJW: Use terminals ⑩, ⑪ for the power supply to the instrument.
(2) Output Wiring
SGJ, SGJL: Use terminals ⑨(+), ⑫(-) for the output wiring.
SGJW: Output 1: Use terminals ⑦(+), ⑧(-) for Output 1 wiring.
Output 2: Use terminals ③(+), ⑥(-) for Output 2 wiring.
(3) Input Wiring
SGJ, SGJL: Use terminals ①, ④, ⑤ for the input wiring.
SGJW: Use terminals ①, ②, ④ for the input wiring.
Open collector
SGJ, SGJL

SGJW

①
④
⑤

＋
－
＋

①

＋

3kΩ
②
12V
④

(Fig. 4.4-1)

－
＋

3kΩ
12V

(Fig. 4.4-2)

Voltage pulse
SGJ, SGJL

SGJW
①
④
⑤

＋

①

－
＋

②
12V
④

(Fig. 4.4-3)＋－＋(Fig. 4.4-4)- 15 -

12V

Mechanical contact
SGJ, SGJL
①
④
⑤

SGJW

＋
－
＋

①
3kΩ
②
12V
④

(Fig. 4.4-5)－＋(Fig. 4.4-6)

Line driver
SGJ, SGJL
120Ω

＋

SGJW
120Ω

＋

＋

－

－

(Fig. 4.4-7)(Fig. 4.4-8)(4) Communication Wiring
For the SGJL, connect the SGJL to SGJL using the provided cable.

(Fig. 4.4-9)- 16 -

3kΩ
12V

5. Display Mode
MODE key+
DISP keyfor 3 sec

RUN display
mode 1

MODE key+
DISP keyfor 3 sec

MODE key+
DISP keyfor 3 sec

DISP key
RUN display
mode 2

MODE key+
DISP keyfor 3 sec

MODE key+
DISP keyfor 3 sec

DISP key
RUN display
mode 3

MODE key+
DISP keyfor 3 sec

DISP key

MODE key+
DISP keyfor 3 sec

RUN display
mode 4

MODE key+
DISP keyfor 3 sec

DISP key

MODE key+
DISP keyfor 3 sec

RUN display
mode 5

MODE key+
DISP keyfor 3 sec

DISP key

MODE key+
DISP keyfor 3 sec

RUN display
mode 6

MODE key+
DISP keyfor 3 sec

DISP key

MODE key+
DISP keyfor 3 sec

Custom display
mode 1

Custom display
mode 2

DISP key
Custom display
mode 3
DISP key
Custom display
mode 4
DISP key
Unlit display
mode
DISP key
All unlit
display mode
DISP key
Model display
mode
DISP key

DISP key

Default Display•: Available only for the SGJW.

Default Display:

If the MODE and DISP keys (in that order) are pressed together
for approx. 3 seconds in any display mode, the display mode will
become the Default Display.
Once the Default Display is set, the DISP key will be in lockstatus.
• If the DISP key is pressed for approx. 3 seconds on the Default
Display, the key lock status will be cancelled.
If the DISP key is pressed while the DISP key is in lock status,
Multi-Display A indicates.
- 17 -

RUN display mode 1:
RUN display mode 2:
RUN display mode 3:
RUN display mode 4:
RUN display mode 5:
RUN display mode 6:
Custom display mode 1:
Custom display mode 2:
Custom display mode 3:
Custom display mode 4:
Unlit display mode:All unlit display mode:

Model display mode:

Multi-Display A indicates an input value, and Multi-Display B
indicates Output 1 value.
Multi-Display A indicates an input value, and Multi-Display B
indicates Output 2 value.
Multi-Display A indicates an input value, and Multi-Display B
is unlit.
Multi-Display A indicates Output 1 value, and Multi-Display B
indicates Output 2 value.
Multi-Display A is unlit, and Multi-Display B indicates Output 1
value.
Multi-Display A is unlit, and Multi-Display B indicates Output 2
value.
Multi-Display A indicates characters set in [Multi-Display A].
Multi-Display B indicates characters set in [Multi-Display B].
Multi-Display A indicates an input value. Multi-Display B
indicates characters set in [Multi-Display B].
Multi-Display A indicates an Output 1 value. Multi-Display B
indicates characters set in [Multi-Display B].
Multi-Display A indicates an Output 2 value. Multi-Display B
indicates characters set in [Multi-Display B].
Multi-Display A and B are unlit, and the Input indicator A lights
up.
Alarm indicator A lights up if they are under the conditions oflighting.
All displays and indicators are unlit.
Alarm indicator A and B do not light up even if they are under theconditions of lighting.
Multi-Display A indicates a model name, and Multi-Display B
indicates an input code and output code.

- 18 -

6. Setting Mode
6.1••

Display Transition in Setting Mode

: Available only for the SGJW.
: Available only for the SGJL.
• If the MODE key is pressed and held down for approx. 5 seconds in each setting mode,
the unit will move to the Default Display.
Default Display

DOWN
key
for 3 sec

MODE
key
for 5 sec
5 秒押し

Manual mode
(See p.28.)
MODE key, or
after Manualmode auto
return time has
elapsed.

Input setting
mode(See pp.19, 20.)

DISP
key
for 3 sec
Frequency range
group

MODE
key

Output 1
setting mode
(See pp.21, 22.)

DISP
key
for 3 sec

MODE
key

Output 2
setting mode
(See pp.23, 24.)

DISP
key
for 3 sec

Output 1 typeOutput 2 type

DISP
key

DISP
key

Frequency
high limit

Output 1

Output 2

DISP
key

DISP
key

DISP
key

DISP
key

Returns to theDefault Display.

Indication value
at output 0%
DISP
key

DISP
key
Indication value
at output 100%

DISP
key

DISP
key
Shutdown
threshold value

Indication value
at output 0%
DISP
key
Indication value
at output 100%
DISP
key
Shutdown
threshold value

DISP
key

DISP
key

DISP
key

Indication unit

Output value

Output value

at shutdown

at shutdown

DISP
key

DISP
key

DISP
key
Save settings

Output 1
Normal/Reverse
DISP
key

DISP
key

Save settings

Save settings

DISP
key

DISP
key

DISP
key
Returns to theDefault Display.

Output 2
Normal/Reverse

Returns to theDefault Display.

- 19 -

Returns to theDefault Display.

MODE
key

Instrument
setting mode
(See p.25.)
DISP
key
for 3 sec

MODE
key

Communication
setting mode
(See pp.26, 27.)
DISP
key
for 3 sec

Indication time

DISP
key
Auto/Manual
DISP
key
Manual mode
auto return time
DISP
key

Returns to theDefault Display.

DISP
key
for 3 sec
Multi-Display A

DISP
key
Communication
speed

DISP
key
Multi-Display B

DISP
key

DISP
key

Data bit/Parity

Save settings

DISP
key

DISP
key
Returns to theDefault Display.

Save settings
DISP
key

MODE
key

Custom display
setting mode
(See p.28.)

DISP
key
Responsedelay time
DISP
key
Save settings
DISP
key

Returns to theDefault Display.

- 20 -MODE
key

To Input setting mode

6.2

Input Setting Mode

Frequency Range group
Selects the frequency range group.
Setting Range

Indication
Multi-Display A Multi-Display B

Ultra-low frequency
Low frequency
Frequency

Factory Default
Low frequency

Frequency High Limit
Sets the frequency high limit.
Setting Range

Indication
Multi-Display A Multi-Display B

Ultra-low frequency
10 to 9999 mHz
Low frequency 1 to 9999 Hz
Frequency
1 to 100 kHz

Factory Default
9999 HzSet value

Decimal Point Place
Selects the decimal point place.
Setting Range

Indication
Multi-Display A Multi-Display B

No decimal point
1 digit after decimal point
2 digits after decimal point
3 digits after decimal point

Factory Default
No decimal point

Indication Value for Input Low Limit
Sets the indication value for input low limit.
Setting Range

Indication
Multi-Display A Multi-Display B

Factory Default
0-1999 to
Indication value for input high limit

Set value

Indication Value for Input High Limit
Sets the indication value for input high limit.
Setting Range

Indication
Multi-Display A Multi-Display B

Indication value for input low limit
to 9999

Set value- 21 -

Factory Default
Frequency highlimit

Indication Unit
Selects the unit for indication.
Setting Range

Indication
Multi-Display A Multi-Display B

No unit%mA
V°C

Factory Default
No unit

Save Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
Save

Save
Not save- 22 -

6.3

Output 1 Setting Mode

Output 1 Type
Selects Output 1 type.
Setting Range

Indication
Multi-Display A Multi-Display B

4 to 20 mA
0 to 20 mA
0 to 16 mA
2 to 10 mA
0 to 10 mA
0 to 10 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V
0 to 10 V-5 to 5 V ** Not available for the SGJW.

Factory Default

4 to 20 mA

Output 1 Decimal Point Place
Selects a decimal point place for Output 1.
Setting Range

Indication
Multi-Display A Multi-Display B

No decimal point
1 digit after decimal point
2 digits after decimal point
3 digits after decimal point

Factory Default
2 digits after
decimal point

Indication Value at Output 0%
Sets an indication value at the time of output 0%. [See (Table 6.3-1) on p.20.]
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

4.00-1999 to
Indication value at output 100%

Set value

Indication Value at Output 100%
Sets an indication value at the time of output 100%.
[See (Table 6.3-1) on p.20.]
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
20.00

Indication value at output 0% to9999

Set value- 23 -

(Table 6.3-1)
Output Range
4 to 20 mA
0 to 20 mA
0 to 16 mA
2 to 10 mA
0 to 10 mA
0 to 10 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V
0 to 10 V-5 to 5 V

Indication value at
output 0%
4.00
0.00
0.00
2.00
0.00
0.0
0.0
0.00
0.00
1.00
0.00
-5.0

Indication value at
output 100%
20.00
20.00
16.00
10.00
10.00
10.0
100.0
1.00
5.00
5.00
10.00
5.0

Shutdown Threshold Value
Sets the input value for the shutdown threshold.
If an input value is lower than this point, the value set in [Output value at shutdown] will
be output.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

00 to 10% FS

Set value

Output Value at Shutdown
If any inputs lower than the [Shutdown Threshold Value] are received, the value
set here will be output.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

0%-5 to 105%

Set value

Output 1 Normal/Reverse
Selects either Normal mode or Reverse mode for Output 1 status.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
NormalNormalReverseSave Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
Save

Save
Not save- 24 -

6.4 Output 2 Setting Mode
Available only for the SGJW.
Output 2 Type
Selects Output 2 type.
Setting Range

Indication
Multi-Display A Multi-Display B

4 to 20 mA
0 to 20 mA
0 to 16 mA
2 to 10 mA
0 to 10 mA
0 to 10 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V
0 to 10 V

Factory Default

4 to 20 mA

Output 2 Decimal Point Place
Selects the decimal point place for Output 2.
Setting Range

Indication
Multi-Display A Multi-Display B

No decimal point
1 digit after decimal point
2 digits after decimal point
3 digits after decimal point

Factory Default
2 digits after
decimal point

Indication Value at Output 0%
Sets an indication value at the time of output 0%.
[See (Table 6.4-1) on p.22.]
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
4.00-1999 to
Indication value at output 100%

Set value

Indication Value at Output 100%
Sets an indication value at the time of output 100%.
[See (Table 6.4-1) on p.22.]
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
20.00

Indication value at output 0% to9999

Set value- 25 -

(Table 6.4-1)
Output Range
4 to 20 mA
0 to 20 mA
0 to 16 mA
2 to 10 mA
0 to 10 mA
0 to 10 mV
0 to 100 mV
0 to 1 V
0 to 5 V
1 to 5 V
0 to 10 V

Indication value at
output 0%
4.00
0.00
0.00
2.00
0.00
0.0
0.0
0.00
0.00
1.00
0.00

Indication value at
output 100%
20.00
20.00
16.00
10.00
10.00
10.0
100.0
1.00
5.00
5.00
10.00

Shutdown Threshold Value
Sets the input value for the shutdown threshold.
If an input value is lower than this point, the value set in [Output value at shutdown] will
be output.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

00 to 10% FS

Set value

Output Value at Shutdown
When any inputs lower than the [Shutdown Threshold Value] are received, the value
set here will be output.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

0%-5 to 105%

Set value

Output 2 Normal/Reverse
Selects either Normal mode or Reverse mode for Output 2 status.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
NormalNormalReverseSave Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
Save

Save
Not save- 26 -

6.5

Instrument Setting Mode

Indication Time
Sets duration from no operation until indication (of Multi-Display A, Multi-Display B,
and each action indicator) turns off.
They remain lit during setting mode or in the event of an input error.
When set to 00.00, they remain lit.
After indication time has elapsed, and if any key is pressed while they are unlit,
they will light up again.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

00 : 00 to 60 : 00 (Minutes : Seconds)
00 : 00 .................. Continuous
00 : 01 to 60 : 00 ..Indication time

Set value

30 : 00(Minutes :Seconds)

Auto/Manual
If AUTO is selected, the output value will be output corresponding to the input value.
When MANUAL is selected, the unit can enter Manual mode. The output value set in
Manual mode will be output.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

ManualAuto
ManualManual Mode Auto Return Time
Sets duration from manual mode until the unit automatically returns to the Default
Display.
If set to 0 (zero), auto return will not occur.
Indication
Factory Default
Setting Range
Multi-Display A Multi-Display B

30 minutes
0 to 60 minutes

Set value

Save Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
Save

Save
Not save- 27 -

6.6 Communication Setting Mode
Available only for the SGJL.
Instrument Number
Sets an instrument number.
Setting Range

Indication
Multi-Display A Multi-Display B

Factory Default
1

1 to 247

Set value

Communication Speed
Selects the communication speed.
Setting Range

Indication
Multi-Display A Multi-Display B

9600 bps
19200 bps
38400 bps

Factory Default
38400 bps

Data bit/Parity
Selects data bit and parity.
Setting Range

Indication
Multi-Display A Multi-Display B

8 bits/No parity
8 bits/Even
8 bits/Odd

Factory Default
8 bits/Odd

Stop Bit
Selects stop bit.
Setting Range

Indication
Multi-Display A Multi-Display B

Factory Default
1 bit

1 bit
2 bitsResponse Delay Time
Response from the instrument can be delayed after receiving command from the hostcomputer.
Indication
Setting Range
Factory Default
Multi-Display A Multi-Display B

10 ms
0 to 1000 ms

Set value- 28 -

Save Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
Save

Save
Not save- 29 -

6.7 Custom Display Setting Mode
Customizes characters to be indicated on the Multi-Display A and B (*).
Use alphanumeric characters and symbols.
(e.g.) FLOW, TEMP, No.1, No.2(*) Number of characters which can be indicated differs depending on the display
mode.
Refer to Section ‘5. Display Mode’ (pp.13, 14).
• If Custom display mode 1 is selected:
Up to 8 characters can be displayed in total for both Multi-Display A and B.
• If any of Custom display mode 2 to 4 is selected:
Up to 4 characters can be displayed on the Multi-Display B.
Can be set from the thousands digit of the display.
Digits can be selected with the MODE key.
Multi-Display A
Characters for the Multi-Display A can be customized.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
AAAA

A to Z, 0 to 9, /, –，. , (Blank)

Set value

Multi-Display B
Characters for the Multi-Display B can be customized.
Indication
Setting Range

Multi-Display A Multi-Display B

Factory Default
AAAA

A to Z, 0 to 9, /, –，. , (Blank)

Set value

Save Settings
Selects whether the settings are saved (registered) or not.
Indication
Setting Range

Multi-Display A Multi-Display B

Save

Factory Default
Save

Not save
6.8

Manual Mode
If MANUAL is selected in [Auto/Manual] in Instrument setting mode, press the DOWN
key for 3 seconds. The unit can enter Manual mode.
At this time, Multi-Display A indicates Output 1 value, and Multi-Display B indicates
Output 2 value.
The output value can be set by the UP or DOWN key.
The output value will be lit while setting.
By pressing the DISP key, the desired output can be selected. and the desired output
flashes.
By pressing the MODE key in Manual mode, or after Manual mode auto return time
has elapsed, the unit returns to the Default Display, and outputs the output value
corresponding to the input value.
- 30 -

7. Adjustment
Performs the output zero and span adjustments.
For this instrument, the output adjustment has already been completed when shipped.
If the instrument is used with the ordered Input/Output spec, the adjustment is not required.
However, for calibration, or for the fine adjustment of the SGJ to which any equipment isconnected, perform the adjustment.
Connect a pulse generator to the input terminals of this instrument.
Connect a digital multimeter to the output terminals.
7.1

Basic Operation of Adjustment

Use the following trimmers on the front panel for adjustment.
Output 1 Zero: Adjusts the value of Output 1 Zero.
Output 1 Span: Adjusts the value of Output 1 Span.
Output 2 Zero: Adjusts the value of Output 2 Zero. (for SGJW)
Output 2 Span: Adjusts the value of Output 2 Span. (for SGJW)
7.2

Adjustment

7.2.1 Output 1 Adjustment
The following outlines the procedure for Output 1 adjustment.
① Enter the value corresponding to output 0%, and adjust the value using the‘Output 1 Zero’ trimmer while viewing the output value (on the digital multimeter).
② Enter the value corresponding to output 100%, and adjust the value using the‘Output 1 Span’ trimmer while viewing the output value (on the digital multimeter).
③ Enter the value corresponding to output 0% again, and confirm the output value(on the digital multimeter).
④ If the value corresponding to output 0% is not at 0%, repeat steps ① to ③ again.
7.2.2 Output 2 Adjustment
The procedure for Output 2 adjustment is the same as that of Output 1 adjustment.
Use Output 2 Zero and Span trimmers for adjustment.

- 31 -

8. Operation
8.1 Indication after Power-on
After the power is turned on, the instrument is switched to warm-up status for 3
seconds. Multi-Display A indicates a model name, and Multi-Display B indicates the
input code (for the thousands and hundreds digits), output 1 code (for the tens digit) andoutput 2 code (for the ones digit).
(e.g.) SGJW-F011-0-0
Multi-Display A:
Multi-Display B:
A value corresponding to input 0% will be output for Output 1 and Output 2.
8.2 Operation
After warm-up status, the unit enters display mode.
The input signal selected in [Input type] will be converted to the output selected in[Output 1 type] and [Output 2 type].
8.2.1 Input Indication Range
The measured value is indicated within the following range:[Indication value for Input low limit] to[Indication value for Input high limit + (Indication value for Input high limit –Indication value for Input low limit) x 10%]
For a value higher than 10000, the lower 4 digits will flash.
When exceeding the indication range,
will flash.
If pulse is absent, 0 (zero) will flash.
The placement of the decimal point follows the selection.
8.2.2 Indication Range of Output 1 and Output 2
The output value is indicated within the following range:[Indication value at output 0% – (Indication value at output 100% – Indication
value at output 0%) x 10%] to[Indication value at output 100% + (Indication value at output 100% – Indication
value at output 0%) x 10%]
However, for a value higher than (and including) 10000, the lower 4 digits will flash.
(The placement of the decimal point follows the selection.)
8.2.3 Indication Time
After the preset indication time has elapsed, Multi-Display A, Multi-Display B andeach action indicator are turned OFF. They light up again if any key is pressed.
They remain lit in setting mode or in the event of an input error.
If the indication time is set to 00:00, they will remain lit.
8.2.4 Detecting Unconnected Sensor
If pulses are not detected during the detection time-out below, the input will default
to the initial status (0 Hz).
Multi-Display A or B will flash 0 (zero) when it indicates an input value.
Detection time-out: Ultra-low frequency: 1000 seconds
Low frequency:
100 seconds
Frequency:
1 second- 32 -

9. Specifications
Input Specifications
Open collector

Frequency range:
0.001 Hz to 100 kHz
Minimum pulse width: 4 µs minimum. (for both ON and OFF)
Detecting voltage/current: Approx. 12 V/4 mA
Detecting level:
At ON: 200 Ω max./0.8 V max.
At OFF: 100 kΩ minimum./11 V minimum.

Voltage pulse

Frequency range:
0.001 Hz to 100 kHz
Minimum pulse width: 4 µs minimum. (for both ON and OFF)
Waveform:
Square, sine waveform or similar
Detecting level:
Low level 1 V DC max.
High level 2 V DC minimum.
Input impedance:
100 kΩ minimum.
Input amplitude:
2 V to 50 Vp-p
At OFF: 100 kΩ minimum.

Mechanicalcontact

Frequency range:
0.001 Hz to 30 Hz
Minimum pulse width: 10 ms minimum. (for both ON and OFF)
Action input condition: At ON: 200 Ω max.
At OFF: 100 kΩ minimum.

Line driver

Frequency range:
0.001 Hz to 100 kHz
Receiver:RS-422 compliant
Minimum pulse width: 5 µs minimum. (for both ON and OFF)
Waveform:
SquareOutput 1 Specifications
Direct current

Allowable load
Zero adjust- Span adjustresistancement range ment range
4 to 20 mA
750
max.
0 to 20 mA*
750max.
0 to 16 mA*
900
max. -5 to 5%
95 to 105%
2 to 10 mA
1500
max.
0 to 10 mA*
1500
max.
* 0 mA or less: Out of base accuracy
Output Range

DC voltage

Allowable load
Zero adjust- Span adjustresistancement range ment range
0 to 10 mV *1
10 k minimum.
0 to 100 mV *1
100 k minimum.
*1
0 to 1 V
1000
minimum.
0 to 5 V *1
5000
minimum.
-5 to 5%95 to 105%
1 to 5 V
5000
minimum.
0 to 10 V *1
10 k minimum.
-5 to 5 V *2
10 k minimum.
*1: 0 V or less: Out of base accuracy*2: Not available for the SGJW.
Output Range- 33 -

Output 2 Specifications
Direct current
Output Range

Allowable load
Zero adjust- Span adjustresistancement range ment range
4 to 20 mA
750
max.
0 to 20 mA*
750max.
0 to 16 mA*
900
max. -5 to 5%
95 to 105%
2 to 10 mA
1500
max.
0 to 10 mA*
1500
max.
* 0 mA or less: Out of base accuracy

DC voltage

Allowable load
Zero adjust- Span adjustresistancement range ment range
0 to 10 mV *10 k minimum.
0 to 100 mV *100 k minimum.
0 to 1 V *1000
minimum.
-5 to 5%95 to 105%
0 to 5 V *5000
minimum.
1 to 5 V
5000
minimum.
0 to 10 V *10 k minimum.
* 0 V or less: Out of base accuracy
Output Range

Performance
Base accuracy (at 25°C)

0.1% of each input span

Temperature coefficient

0.015 %/°C

Response time(250 ms + Pulse cycle + Output value update cycle +Frequency sampling period) max.

Output value update
cycle

125 msIndication accuracy

Base accuracy 1 digit

General Structure
Dimensions

22.5 x 89 x 70 mm (W x H x D)

Weight1 output: Approx. 80 g (excluding socket)
2 outputs: Approx. 90 g (excluding socket)

Mounting

DIN railCase

Flame-resistant resin, Color: Black

Front panel

Polycarbonate

Installation Specifications
Power supply

100 to 240 V AC 50/60 Hz

Allowable voltage range

85 to 264 V AC

Power consumption

SGJ: Approx. 9 VA
SGJW: Approx. 12 VA
SGJL: Approx. 10 VA

Ambient temperature-10 to 55°C (Non-condensing, no icing)Ambient humidity

35 to 85 %RH (Non-condensing)- 34 -

Serial Communication (for SGJL)
Operation from anexternal computer

Reading and setting of various set values
Reading of the input value and action status
Function change

Communication line EIA RS-485
Communication
methodHalf-duplex communication

Communication
speed

9600, 19200, 38400 bps (Selectable by keypad)(Factory default: 38400 bps)

Synchronization
methodStart-stop synchronization

Communication
protocolModbus RTU

Start bit

1 bit

Data bit

8 bitsParityEven/Odd/No parity (Selectable by keypad)(Factory default: Odd)

Stop bit

1 bit or 2 bits (Selectable by keypad)(Factory default: 1 bit)

Response delaytime

Response from the instrument can be delayed after
receiving command from the host computer.
0 to 1000 ms (Factory default: 10 ms)

Standard Function
Power failurecountermeasure

The setting data is backed up in the non-volatile IC memory.

Self-diagnosis

The CPU is monitored by a watchdog timer, and if an abnormal
status occurs, the instrument is switched to warm-up status,
turning all outputs OFF.

- 35 -

10. Troubleshooting
10.1 Indication
Problemflashes when
Multi-Display A or B

Possible Cause

Solution

Input signal source may bedisconnected.

Check the input signal source.

The sensor may bedisconnected.
Check whether the sensor
is securely mounted to the
input terminals of this
instrument.

Replace with a new sensor.

If pulses are not detected for a
fixed time,
will flash.
Input signal source may bedisconnected.

Check the input signal source.

If a value higher than (andincluding) 10000 is entered,
the lower 4 digits will flash.
Multi-Display A or B is
There may be equipment
irregular or unstable when that interferes with or makes
it indicates an input
noise near the instrument.
value.
Displays and indicators
The Indication Time (p.25) isare unlit.
set to any value other thanIf any key is pressed,
00 : 00.
they will light up.
(Factory default is 30 : 00.)

Check the input signal source.

indicates an input value.

Multi-Display A or B
flasheswhen
Multi-Display A or B
indicates an input value.

4-ditgits are flashing.

10.2 Key Operation
ProblemIf the DISP key ispressed, Multi-Display A
indicates, and the display
mode cannot be switched.

Possible Cause
The DISP key is in locked
status.

- 36 -

Connect the sensor
terminals to the instrument
input terminals securely.

Keep the instrument clear ofany potentially disruptive
equipment.
To indicate continuously,
set the Indication Time
(p.25)
to “00 : 00”.

Solution
Press the DISP key for approx.
3 seconds to release the key
lock.

10.3 Operation
Problem
When Multi-Display A or B
indicates an input value,
the input value does notchange.

No output

Possible Cause
The sensor may be out oforder.
Check whether input andoutput wires are securely
connected to the I/O
terminals of the instrument.
Check whether the wiring of
input and output are correct.
Selections in [Output 1 type(p.21)], [Output 1 Normal/
Reverse (p.22)], [Output 2
type (p.23)] or [Output 2
Normal/Reverse (p.24)]
may be incorrect.

- 37 -

Solution
Replace with the newsensor.
Ensure that input and output
wires are securely
connected to the I/O
terminals of the instrument.
Wire them correctly.
Make a correct selection in[Output 1 type (p.21)],[Output 1 Normal/Reverse(p.22)], [Output 2 type (p.23)]
or [Output 2 Normal/Reverse(p.24)].

11. Character Table
Please use the following factory default values for your reference.
Display mode
Setting Item
Default Display mode
RUN display mode 1
RUN display mode 2 *Multi-Display A
Multi-Display B
Follows currently indicated display mode.
Input value
Output 1 value
Output 2 value
Input value

RUN display mode 3
RUN display mode 4 *RUN display mode 5

Input value
Output 1 value
Unlit

Unlit
Output 2 value
Output 1 value

RUN display mode 6 *Custom display mode 1
Custom display mode 2
Custom display mode 3
Custom display mode 4 *Unlit display mode

Unlit

Output 2 value

Input value
Output 1 value
Output 2 value
Unlit

Unlit

All unlit display mode

Unlit

Unlit

Model display mode
* Available only for the SGJW.

Model

Input, Output codes

Setting mode
Setting Item
Input setting mode
Output 1 setting mode

Multi-Display A

Multi-Display B
Unlit
Unlit

Output 2 setting mode *Unlit

Instrument setting mode

Unlit

Communication setting mode

Unlit

Custom display setting mode
* Available only for the SGJW.

Unlit

Input setting mode
Setting Item
Frequency range group
Frequency high limit
Decimal point place
Indication value for Input low
limit
Indication value for Input highlimit
Indication unit
Save settings

Multi-Display A- 38 -

Multi-Display B

Data

Data

Data

Output 1 setting mode
Setting Item
Output 1 typeOutput 1 decimal point place
Indication value at output 0%
Indication value at output 100%
Shutdown threshold value
Output value at shutdown
Output 1 Normal/Reverse
Save settings

Multi-Display A

Multi-Display B

Data

Multi-Display A

Multi-Display B

Data

Multi-Display A

Multi-Display B

Data

Communication setting mode (SGJL)
Setting Item
Multi-Display A
Instrument number

Multi-Display B

Data

Multi-Display B

Data

Output 2 setting mode
Setting Item
Output 2 typeOutput 2 decimal point place
Indication value at output 0%
Indication value at output 100%
Shutdown threshold value
Output value at shutdown
Output 2 Normal/Reverse
Save settings
Instrument setting mode
Setting Item
Indication time
Auto/Manual
Manual mode auto return time
Save settings

Communication speed
Data bit/Parity
Stop bit
Response delay time
Save settings
Custom display setting mode
Setting Item
Multi-Display A
Multi-Display B
Save settings

Multi-Display A- 39 -

***** Inquiries *****For any inquiries about this unit, please contact our agency or thevendor where you purchased the unit after checking the following.
[Example]• Model ------------------------------ SGJW-F011-0-0• Serial number -------------------- 154F05000
In addition to the above, please let us know the details of themalfunction, or discrepancy, and the operating conditions.

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office: 2-5-1, Senbahigashi, Minoo, Osaka, 562-0035, Japan[URL]
https://shinko-technos.co.jp/e/Tel: +81-72-727-6100[E-mail]overseas@shinko-technos.co.jp
Fax: +81-72-727-7006

No. SGJ11E4 2024.02