CALIBRATOR
T5014
INSTRUCTION MANUALPreface
Thank you for purchasing our Calibrator T5014.
This manual contains instructions for the mounting, functions, operations and notes when operating theCalibrator T5014. To prevent accidents arising from the misuse of this instrument, please ensure the operator
receives this manual.

Notes• This instrument should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed, serious injury or malfunction
may occur.
• Specifications, external appearance of the Calibrator T5014 and the contents of this instruction manual are
subject to change without notice.
• Care has been taken to assure that the contents of this instruction manual are correct, but if there are any
doubts, mistakes or questions, please inform our sales department.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos CO., LTD. is not liable for any damage or secondary damage(s) incurred as a result ofusing this product, including any indirect damage.

SAFETY PRECAUTIONS (Be sure to read these precautions before using our products.)
The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on circumstances, procedures indicated by
Caution may be linked to serious results, so be sure
to follow the directions for usage.

Warning
Procedures which may lead to dangerous conditions and cause death or serious injury, if not carriedout properly.

CautionProcedures which may lead to dangerous conditions and cause superficial to medium injury or physicaldamage or may degrade or damage the product, if not carried out properly.

Warning• To prevent an electric shock or fire, only Shinko or qualified service personnel may handle the innerassembly.
• To prevent an electric shock, fire or damage to instrument, parts replacement may only be undertaken by
Shinko or qualified service personnel.

Safety precautions• To ensure safe and correct use, thoroughly read and understand this manual before using this instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring equipment.
Verify correct usage after consulting purpose of use with our agency or main office. (Never use this
instrument for medical purposes with which human lives are involved.)
We strongly recommend periodic maintenance of this equipment by our qualified personnel (standard
charges apply).
• This instrument must be used under the conditions and environment described in this manual. Shinko
Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring due to the
instrument being used under conditions not otherwise stated in this manual.
Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the manufacture of
weapons of mass destruction (i.e. military applications, military equipment, etc.), please investigate the end
users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.
2

1. Usage precautions

CautionThis instrument is intended to be used under the following environmental conditions (IEC61010-1):
Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of 23 5 that does not change rapidly• An ambient non-condensing humidity of 80%RH or less
• No water, oil or chemicals or where the vapors of these substances can come into direct contact withthe unit
Note • Do not install this instrument near flammable material even though the case of this instrument ismade of flame-resistant resin.
Avoid setting this instrument directly on flammable material.

2. Handling precautions

Caution• After power-on, warm up this calibrator for approximately 10 minutes.
• Do not short the input terminals and output terminals.
• Maximum voltage between terminals and between terminal and grounding is 30V.
Therefore, do not apply voltage over 30V. This could cause a malfunction.
• Do not apply voltage to the output terminals, as it could cause a malfunction.
• Do not apply voltage exceeding the measuring range to the input terminals of the calibrator.
Turn the power supply to the unit to be measured OFF before connecting the calibrator.
• Do not apply current (ampere) signal to the input terminals of the calibrator.

3. Maintenance precautions

Caution• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)• As the display section is vulnerable, do not strike or scratch it with a hard object or press hard on it.

3

CONTENTS
1. Unpacking ................................................................................................................................ 5
1.1 Unpacking …………………………………..…………………………………………………………………… 5
1.2 Checking the accessories ……………………………………………………………………………………. 5
1.3 Checking the model T5014 …………………………………………………………………………………… 5

2. Battery loading/replacement ....................................................................... 5
2.1 Battery loading....................................................................................................................................... 5
2.2 Battery removal...................................................................................................................................... 7

3. Name and functions of the sections................................................ 8
3.1 Name and functions of the sections .................................................................................................... 8
3.2 Name and functions of the LCD display……………………………………………………………………. 9

4. External dimensions .............................................................................................. 10
4.1 External dimensions (Unit: mm) .......................................................................................................... 10
4.2 External dimensions of Lead wire CF-36 (with a clip) and name (Unit: mm) ....................................11
4.3 Compensating lead wire K (CT5-K-01, sold separately) external dimensions (unit: mm)………… 12
4.4 Compensating lead wire J (CT5-J-01, sold separately) external dimensions (unit: mm) ............... 12
4.5 RTD 3-wire lead (CT5-P3-01, sold separately) external dimensions (unit: mm) .............................. 12
4.6 RTD 4-wire lead (CT5-P4-01, sold separately) external dimensions (unit: mm) .............................. 12

5. Power ON/OFF ................................................................................................................ 13
5.1 Power ON/OFF ..................................................................................................................................... 13
5.2 Auto-power-OFF .................................................................................................................................. 13

6. Operation flowchart ................................................................................................ 14
7. Output and measurement mode…………………………………………..15
7.1 Output mode ........................................................................................................................................ 15
7.1.1 Wiring ............................................................................................................................................. 15
7.1.2 Operation…………………………………………………………………………………………………... 18
7.1.3 Output reset................................................................................................................................... 19
7.1.4 Automatic cold junction temperature compensating function ………………..………………….19
7.2 Measurement mode …………………………………………………………………………………………... 21
7.2.1 Wiring ……………………………………………………………………………………………………... 221
7.2.2 Operation ....................................................................................................................................... 23
7.2.3 Overrange indication .................................................................................................................... 23
7.2.4 Automatic cold junction temperature compensating function.................................................. 23

8. Specifications .................................................................................................................. 25
9. Troubleshooting ........................................................................................................... 27
10. About calibration…………………………………….……………………………….. ..... 27

4

1. Unpacking
1.1 Unpacking
Upon delivery, unpack and take out a unit and accessories.
1.2 Checking the accessories
Check whether the following accessories are included.
Accessories
Q’ty
Instruction manual
1 copyCarrying case
1 piece
Lead wire CF-36 (with a clip)
1 set
Alkaline battery
1 piece
1.3 Checking the model T5014
Check the model T5014 on the model label at the back of the unit.

2. Battery loading/replacement
Caution• As the material of this calibrator is made of resin, do not use excessive force when turning thelock (,) of the battery cover (), or the lock could be damaged.
2.1 Battery loading(1) Remove the holster.
(2) Release the lock by turning the lock (,) of the battery cover (counterclockwise until it stops, using a flat blade screwdriver.

(Fig. 2.1-1)(3) Remove the battery cover.

(Fig. 2.1-2)
5

at the back of the unit)(4) Load the included alkaline battery.

(Fig. 2.1-3)(5) Set the battery in the battery case.

(Fig. 2.1-4)(6) Mount the battery cover.

(Fig. 2.1-5)

6

(7) To lock the cover, turn the lock (,) of the battery cover (clockwise until it stops, using a flat blade screwdriver.

at the back of the unit)(Fig. 2.1-6)

2.2 Battery removal

Caution• Remove the lead wire, and turn the power supply to the calibrator OFF before replacing the battery.
• Use a new battery: Alkaline battery 9V (ANSI/NEDA 1604A or IEC/JIS 6LR61)

If

is indicated on the LCD display, replace the battery.

(1) Remove the lead wire, and turn the power supply to the calibrator OFF.
(2) Replace the battery with the new one, referring to Section 2.1.

7

3. Name and functions of the sections
3.1 Name and functions of the sections

(7)

(1)

(8)
(9)

(2)

(10)

(3)
(4)

(11)

(5)

(12)

(13)

(6)

(Fig. 3.1-1)(1) LCD display: Indicates action status of the calibrator.
(See Section “3.2 Name and functions of the LCD display” on p.9.)(2) Power key: Turns the power supply to this instrument ON or OFF.
(3) Input/Output key: Switches Measurement or Output mode of the calibrator.
(4) RJ-ON key: Switches Effective/Not effective of the Automatic cold junction temperaturecompensating function.
(5) Shift key: Use this key with the Output value setting key when resetting output valueof the calibrator.
(6) Input terminals : Used in the Measurement mode.
(7) Function key: Selects the unit (V, mV, , ) in the Measurement mode and Output mode.
(8) Range key: Selects thermocouple and RTD type in the Measurement mode and Output mode.
(9), (12) Output value setting key:
Sets output value in the Output mode.
Output value continuously changes by holding down this key for approx. 1 second.
(10), (11) Output value digit selection key:
Selects Output value digit in the Output mode.
(13) Output terminals: Used in the Output mode.
8

3.2 Name and functions of the LCD display(Fig. 3.2-1)
OUTPUT: Indicated during the Output mode.

INPUT: Indicated during the Measurement mode.

RJ-ON: Indicated when Automatic cold junction temperature compensating function
is set to “Effective”.
: Indicated when batteries are low.
: Indicates output value digit in the Output mode.

V, mV,,ON: Indicates the unit (V, mV,,) in the Measurement mode or Output mode.

: Indicated when output signal is ON in the Measurement mode or Output mode.

R, S, K, E, J, T, B, N : Indicates thermocouple type in the Measurement mode or Output mode.
Pt100, Cu50

: Indicates RTD type in the Measurement mode or Output mode.

94. External dimensions
4.1 External dimensions (Unit: mm)(Fig. 4.1-1)

External dimensions when attaching the holster (Unit: mm)
HolsterStand(Fig. 4.1-2)
10

4.2 External dimensions of Lead wire CF-36 (with a clip) and name (Unit: mm)
Plug

Probe

Clip

(Fig. 4.2-1)
Use the probe for a through hole of the PCB or for a pin of the electronic components (Fig. 4.2-2).
Attach the clip to the probe of the lead wire (Fig. 4.2-3).
For the usage of the probe and clip, refer to the drawings below.

(Fig. 4.2-2)(Fig. 4.2-3)

11

4.3 Compensating lead wire K (CT5-K-01, sold separately) external dimensions (unit: mm)(Fig. 4.3-1)

4.4 Compensating lead wire J (CT5-J-01, sold separately) external dimensions (unit: mm)(Fig. 4.4-1)

4.5 RTD 3-wire lead (CT5-P3-01, sold separately) external dimensions (unit: mm)(Fig. 4.5-1)

4.6 RTD 4-wire lead (CT5-P4-01, sold separately) external dimensions (unit: mm)(Fig. 4.6-1)

12

5. Power ON/OFF
5.1 Power ON/OFF
Press thekey to turn the power of the calibrator ON.
After power-on, the calibrator will initiate self-test, and indicates [After that, the instrument proceeds to the Output mode (OUTPUT).

] on the LCD display.

Caution
Do not operate the calibrator while [] is being displayed.

To turn the power supply to the calibrator OFF, Press the

key for approx. 1 second.

CautionWhen turning the power supply to the calibrator OFF and ON again, turn on the power approx. 5
seconds later after power-off, in order to ensure the action of the calibrator.

5.2 Auto-power-OFF
If the instrument has not been operated for 10 minutes after power-on, its power will be automaticallyturned off.
Either Auto-power-OFF “Effective” or “Not effective” can be selected.
(Default: Auto-power-OFF “Effective”)
How to switch Auto-power-OFF Effective/Not effective(1) If the power to the instrument is turned on, turn the power OFF by pressing the
key for approx. 1
second.
(2) Press theandkeys at the same time.
[] or [] will be indicated on the LCD display, and the unit enters the Auto-power-OFF
selection mode.
(3) Using thekey, select “Effective” or “Not effective” of the Auto-power-OFF.
: Auto-power-OFF “Effective”: Auto-power-OFF “Not effective”(4) Press the
key for approximately 1 second.
Auto-power-OFF selection mode will close, and the power supply to the calibrator will be turned OFF.

13

6. Operation flowchart
Explanation of key operationmeans that if the

key is pressed, the unit proceeds to the next setting item.

Power OFF
Pressing theindicates [[Output mode]
OUTPUTkey (Power ON) initiates self-test, and the LCD display].
[Measurement mode]
INPUT

V

mV

mV

R

R

S

S

K

K

E

E

J

J

T

T

B

B

N

N
Pt100

Pt100

Cu50

Cu50

14

7. Output and measurement mode
7.1 Output mode

Caution• Do not short output terminals of the calibrator.
• Maximum voltage between terminals and between terminal and grounding is 30V.
Therefore, do not apply voltage over 30V. This could cause a malfunction.
• Do not apply voltage to the output terminals, as it could cause a malfunction.
In the Output mode, the preset DC voltage or resistance will be output from the output terminals of thecalibrator.
7.1.1 Wiring(1) DC voltage (V, mV), thermocouple wiring
Connect the lead wire plug to the output terminals of the calibrator, and connect the probe to the
input terminals of the unit to be measured.
When using a clip, attach it to the tip of the probe. Refer to (Fig. 4.2-1) to (Fig. 4.2-3).
For thermocouple K, J, the compensating lead wire (plug into the calibrator, and the Y terminal tothe unit to be measured) is sold separately.
Lead wire K: CT5-K-01
Lead wire J: CT5-J-01[DC voltage (V, mV)], [When thermocouple is selected and Automatic cold junction
temperature compensating function of the calibrator is used]
Calibrator

Unit to be measured

Lead wire(Fig. 7.1.1-1)

15

[When thermocouple is selected and Automatic cold junction temperature compensating
function of the calibrator is not used]
Calibrator

Cold junction
compensating unit

Unit to be measured

0

Copper lead wire
Lead wire(Fig. 7.1.1-2)(2) Resistance, RTD wiring
Connect the lead wire pug to the output terminals of the calibrator, and connect the probe to the
input terminals of the unit to be measured.
When using a clip, attach it to the tip of the probe. Refer to (Fig. 4.2-1) to (Fig. 4.2-3).
For the Resistance output, use 3 or 4-wire connection.
When using 2-wire connection, take lead wire error (approx. 0.1 ) into consideration.
If capacitance between calibrator output terminal and the unit to be measured is 0.1 F
or more, a calibration error will be generated.
[2-wire connection]
Calibrator

Unit to be measured

Lead wire(Fig. 7.1.1-3)
16[3-wire connection]
For 3-wire RTD, an exclusive cable (plug into the calibrator, and the Y terminal to the unit to bemeasured) is sold separately.
RTD 3-wire lead: CT5-P3-01
Calibrator

Unit to be measured

RTD 3-wire lead
(Fig. 7.1.1-4)[4-wire connection]
For 4-wire RTD, an exclusive cable (plug into the calibrator, and the Y terminal to the unit to bemeasured) is sold separately.
RTD 4-wire lead: CT5-P4-01
Calibrator

RTD 4-wire lead
(Fig. 7.1.1-5)
17

7.1.2 Operation
When generating thermo-electromotive force of 400 of the thermocouple K
LCD display, key
Operation(1)
Output mode (OUTPUT) confirmation
Ensure that the calibrator is in the Output mode (OUTPUT).
If the calibrator is in the Measurement mode (INPUT), press the
key to enter the Output mode (OUTPUT).

(2)

Output range selection
Select the output unit ( ) with thekey.
For thermocouple and RTD, use thekey for selection (K).
(e.g.) Thermocouple K andhave been selected.

(3)Output value digit selection
Select the Output value digit by pressing the

(4)

or

key.

Output value setting
Set the output value by pressing theorkey.
Holding down this key for 1 second, the output value changes
continuously.
(e.g.) The output value has been set to 400 .

18

7.1.3 Output reset
Output value can be reset by pressing theandkey simultaneously.
Output reset value differs depending on the selected unit and type as follows. (Table 7.1.3-1)(Table 7.1.3-1)
Selected unit and typeOutput reset value
V
0.00000V
mV
000.000mV
000.00
Thermocouple: R, S
0000
Thermocouple: B
400
Thermocouple: K, E, J, T, N
0000.0
RTD: Pt100, Cu50
000.0
7.1.4 Automatic cold junction temperature compensating function
This detects the temperature at the connecting terminal between the thermocouple and the calibrator,
and always maintains the same status as when the reference junction is located at 0 (32 ).
In the case of thermocouple, select “Effective” or “Not effective” of the Automatic cold junction
temperature compensating function of the calibrator.
“Effective” or “Not effective” selection differs depending on the connection method.
• When Automatic cold junction temperature compensating function of the calibrator is used
Calibrator

Unit to be measured

Lead wire(Fig. 7.1.4-1)
Output electromotive force (E):E = Est – Eat
where E : Output electromotive force
Est : Electromotive force corresponding to the setting temperature
Eat : Electromotive force corresponding to ambient temperature
It takes approx. 2 seconds to initiate the Automatic cold junction temperature compensating
function. After that, the function is conducted every 10 seconds.
If ambient temperature is changed, it will take approx. 10 minutes until the internal cold junction
temperature compensation sensor is stabilized.
19

LCD display, key

Operation
Automatic cold junction temperature compensating function“Effective”
Press thekey.
[RJ-ON] is indicated on the LCD display, and Automatic cold junction
temperature compensating function becomes effective.

• When Automatic cold junction temperature compensating function of the calibrator is not used
Calibrator

Cold junction
temperaturecompensating unit

Unit to be measured

0

Copper lead wire
Lead wire(Fig. 7.1.4-2)
LCD display, key

Operation
Automatic cold junction temperature compensating function“Not effective”
Press thekey.
[RJ-ON] is not indicated on the LCD display, and Automatic cold
junction temperature compensating function becomes ineffective.

20

7.2 Measurement mode

Caution• Do not short input terminals of the calibrator.
• Maximum voltage between terminals and between terminal and grounding is 30V.
Therefore, do not apply voltage over 30V. This could cause malfunction.
• Do not apply voltage exceeding the measuring range to the calibrator input terminals.
Turn the power supply to the unit to be measured OFF before connecting the calibrator.
• Do not apply current signal to the calibrator input terminals.
7.2.1 Wiring(1) DC voltage (mV), thermocouple wiring
Connect the lead wire plug to the input terminal of the calibrator, and connect the probe to the output
terminal of the unit to be measured.
When using a clip, attach it to the tip of the probe. Refer to (Fig. 4.2-1) to (Fig. 4.2-3).
For thermocouple K, J, the compensating lead wire (plug into the calibrator, and the Y terminal tothe unit to be measured) is sold separately.
Lead wire K: CT5-K-01
Lead wire J: CT5-J-01[DC voltage (mV)], [In the case of thermocouple input, and when Automatic cold junction
temperature compensating function of the calibrator is used (See p.23, 24)]
CalibratormV generator or thermocouple

Lead wire(Fig. 7.2.1-1)

21

[In the case of thermocouple input, and when Automatic cold junction temperaturecompensating function of the calibrator is not used (See p.23, 24)]
Calibrator

Cold junction temperaturecompensating unitmV generator or thermocouple0

Copper lead wire
Lead wire(Fig. 7.2.1-2)(2) Resistance, RTD wiring
Connect the lead wire pug to the input terminal of the calibrator, and connect the probe to the
output terminal of the unit to be measured.
When using a clip, attach it to the tip of the probe. Refer to (Fig. 4.2-1) to (Fig. 4.2-3).
For the Resistance, RTD measurement mode, only 2-wire connection is available.
Take lead wire error (approx. 0.1 ) into consideration when measuring.

Calibrator

Unit to be measured

Lead wire(Fig. 7.2.1-3)

22

7.2.2 Operation
When measuring DC voltage
LCD display, key
Operation(1)
Measurement mode (INPUT) confirmation
Ensure that the calibrator is in the Measurement mode (INPUT).
If the calibrator is in the Output mode (OUTPUT), press thekey
to enter the Measurement mode (INPUT).

(2)

Measurement range selection
Select the input range with thekey.
For thermocouple and RTD, use thekey for selection.
(e.g.) DC voltage mV range has been selected.
[] is indicated on the LCD display, then measured value
is indicated.

(3)

Measurement
Measured value will be updated for every 0.5 second cycle (forthermocouple, approx. 1 second)(e.g.) Measured value: 400.00mV

7.2.3 Overrange indication
If measured value exceeds the measuring range, [For the thermocouple input, if input signal is open, [] is indicated on the LCD display.
] is indicated.

7.2.4 Automatic cold junction temperature compensating function
This detects the temperature at the connecting terminal between the thermocouple and the calibrator,
and always maintains the same status as when the reference junction is located at 0 (32 ).
For the thermocouple input, Automatic cold junction temperature compensating function
Effective/Not effective can be switched.

23

[In the case of thermocouple input, and when Automatic cold junction temperaturecompensating function of the calibrator is used]
Press thekey.
[RJ-ON] is indicated on the LCD display, Automatic cold junction temperature compensating function
becomes “Effective”.
Indicated temperature will be as shown below.
T = Ttt + Tat
where T : Indicated temperature
Ttt : Thermocouple temperature corresponding to input electromotive force
Tat: Ambient temperature
Calibrator

Thermocouple

Lead wire(Fig. 7.2.4-1)[In the case of thermocouple input, and when Automatic cold junction temperaturecompensating function of the calibrator is not used]
Press thekey.
[RJ-ON] is not indicated on the LCD display, Automatic cold junction temperature compensating
function becomes “Not effective”.
Calibrator

Cold junction temperaturecompensating unit
0
Thermocouple

Copper lead wire

Lead wire(Fig. 7.2.4-2)
24

8. Specifications
Performance
Output performance
The following specifications are applicable at 23
calibration.

5of ambient temperature and within 1 year after

OutputType

DC voltage

1V
100mV

Remarks-0.10000 to 1.10000V 0.01mV
0.02% of setting + 0.01% Max. output
current:-10.000 to 110.000mV 0.001mV of range
2mA

Resistance

400

0.00 to 400.00

0.01

0.02% of setting + 0.02% Energizedof rangecurrent:
1mA(*1,*2)

ThermocoupleR-40 to 1760

1

-20 to 1760

1

K-200 to 1370

0.1

E-200 to 1000

0.1J-200 to 1200

0.1

T
B

-200 to 400
400 to 1800

0.1
1

N

-200 to 1300

0.1

: 1.5
: 1.2
: 1.5
: 1.2
: 0.6
: 0.5
: 0.7
: 0.9
: 0.6
: 0.5
: 0.6
: 0.6
: 0.5
: 0.7
: 0.6
: 2.0
: 1.5
: 1.1
: 1.0
: 0.7
: 0.8

ITS-90 (*3)

S

-40 to 100
100 to 1760
-20 to 100
100 to 1760
-200 to -100
-100 to 400
400 to 1200
1200 to 1370
-200 to -100
-100 to 600
600 to 1000
-200 to -100
-100 to 800
800 to 1200
-200 to 400
400 to 600
600 to 800
800 to 1800
-200 to -100
-100 to 900
900 to 1300

Pt100

-200 to 850

0.1-50 to 150

0.1

: 0.3
: 0.5
: 0.8
: 0.6

Energized
current:
1mA (*1,*2)

Cu50-200 to 0
0 to 400
400 to 850-50 to 150

RTD

Output range

Resolution

Accuracy(*1): Excluding lead wire (accessories included) resistance(*2): Energized current range 0.5 to 2mA, Max output voltage: 2V or less(*3): Excluding an error of the automatic cold junction temperature compensating sensor.
Internal cold junction temperature compensating range: -10 to 50 ,Compensating error: 0.5 or lessTemperature coefficient: 0.005%/ , Temperature range: 0 to 18 , 28 to 50

25

Measurement performance
The following specifications are applicable at 23

5

Input

Type

DC voltage

400mV

0.00 toResistance

400

0.00 to 450.00

ThermocoupleR-40 to 1760

S

-20 to 1760

K-200 to 1370

0.1

E-200 to 1000

0.1J-200 to 1200

0.1

T-200 to 400

0.1B

400 to 1800

1N-200 to 1300

0.1

Pt100

-200 to 850

0.1

Cu50

-50 to 150

0.1

RTD

Input rangeof ambient temperature and within 1 year after calibration.
Resolution
Accuracy
Remarks
Input
0.02% of reading
10 V
resistance:+ 0.02% of range
1G
Measuring
0.02% of reading
0.01
current:+ 0.02% of range
1mA (*1)Input
1-40 to 500: 1.8resistance:
500 to 1760
: 1.5
1G
1
-20 to 500: 1.8
ITS-90 (*2)
500 to 1760
: 1.5

450.00mV-200 to 0
0 to 1370-200 to 0
0 to 1000-200 to 0
0 to 1200-200 to 0
0 to 400
400 to 800
800 to 1000
1000 to 1800
-200 to 0
0 to 1300

: 1.2
: 0.8
: 0.9
: 1.5
: 1.0
: 0.7
: 1.0
: 0.7
: 2.2
: 1.8
: 1.4
: 1.5
: 0.9

-200 to 0
0 to 400
400 to 850-50 to 0
0 to 150

: 0.5
: 0.7
: 0.8
: 0.5
: 0.7

(*1): Excluding lead wire (accessories included) resistance(*2): Excluding an error of the automatic cold junction temperature compensating sensor.
Internal cold junction temperature compensating range: -10 to 50 ,Compensating error: 0.5 or lessTemperature coefficient: 0.005%/ , Temperature range: 0 to 18 , 28 to 50
General specifications
Power supply
Alkaline battery 9V (ANSI/NEDA 1604A or IEC/JIS 6LR61)
Battery life
Approx. 15 hours
Max. allowable voltage
30V (Between terminals, Between terminal and grounding)Operating temperature range 0 to 50 (Accuracy guaranteed at 23 5 )Operating humidity range
80%RH or less
Storage temperature range-10 to 55
Storage humidity range
90%RH or lessWarm-up period
Approx. 10 minutes
External dimensions
98 x 204 x 42mm (W x H x D) (with holster)
Weight550g (with holster)Accessories included
Instruction manual
1 copyCarrying case
1 piece
Lead wire CF-36 (with a clip) 1 set
Alkaline battery
1 piece
Accessories sold separately Lead wire K: CT5-K-01
Lead wire J: CT5-J-01
RTD 3-wire lead : CT5-P3-01
RTD 4-wire lead : CT5-P4-01
26

Measuring
current:
1mA (*1)

9. Troubleshooting
Problem
Output does not occur in the
Output mode.

Error is generated in the TC
electromotive force output
mode.

[] is indicated on the
LCD display.

[] is indicated on the
LCD display.

Power is turned off.

Presumed cause and solution• Check whether the calibrator is in the Output mode (OUTPUT).
• Check whether the lead wire is securely connected to the output
terminals.
• Check wiring.
• Check the output range and output value.
• Check whether the unit to be measured has the automatic cold
junction temperature compensating function.
If it has the automatic cold junction temperature compensating
function, set the automatic cold junction temperature compensating
function of the calibrator to “Effective”.
• Measured value has exceeded the measuring range.
Check the input signal.
• Check whether the lead wire is connected to input terminals• Check wiring.
• For thermocouple input, input signal is open.
Check the input signal.
• Check whether the lead wire is connected to input terminals.
• Check wiring.
• If the calibrator is not operated for 10 minutes after power-on, its
power will be automatically turned off (Auto-power OFF function).
If auto-power-off function is not required, set the Auto-power-OFF
to “Not effective”.

10 About calibration
To maintain accuracy, please calibrate this instrument every year.
We cannot guarantee accuracy if users calibrate this instrument.
For calibration, please contact our main office or dealers.

27

*****

Inquiry*****

For any inquiries about this unit, please contact our agency or theshop where you purchased the unit after checking the following.

(e.g.)• Model------------------------ T5014• Serial number-------------- No. xxxxxxxx

In addition to the above, please let us know the details of the malfunction,
if any, and the operating conditions

SHINKO TECHNOS CO.,LTD.
OVERSEAS DIVISION
Reg. Office: 2-5-1, Senbahigashi, Minoo, Osaka, JapanURL: http://www.shinko-technos.co.jp
E-mail: overseas@shinko-technos.co.jp

Tel : 81-72-727-6100
Fax: 81-72-727-7006
No. T5014E1 2006.04