TOUCH SCREENPROGRAMMABLE CONTROLLER

PCT-200
INSTRUCTION MANUALPrefaceThank you for purchasing our PCT-200, Touch Screen Programmable Controller.
This manual contains overview, functions, operation methods and notes for operating the PCT-200.
For the mounting and wiring, refer to the instruction manual of each connected instrument.
To ensure safe and correct use, thoroughly read and understand this manual and each manual of theconnected instruments before using this instrument.
To prevent accidents arising from the misuse of this controller, please ensure the operator receives thismanual and each manual of the connected instruments.
Abbreviations used in this manual
SymbolTerm
PV
Process variable
SV
Desired valueMV
Output manipulated variableAT
Auto-tuning

Caution• This instrument should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow all of the warnings, cautions and notices in this manual and each manual of the connected
instruments. If they are not observed, serious injury or malfunction may occur.
• The contents of this manual and each manual of the connected instruments are subject to change without
notice.
• Care has been taken to assure that the contents of this manual and each manual of the connected
instruments are correct, but if there are any doubts, mistakes or questions, please inform our sales
department.
• Measures must be taken to ensure that the operator cannot touch power terminals or other high voltage
sections.
• Any unauthorized transfer or copying of this document and each manual of the connected instruments, inpart or in whole, is prohibited.
• Shinko Technos Co., Ltd. is not liable for any damage or secondary damage(s) incurred as a result of using
this product, including any indirect damage.

Safety Precautions (Be sure to read these precautions before using our products.)
The safety precautions are classified into 2 categories: “Warning” and “Caution”. Depending on thecircumstances, procedures indicated by
Caution may cause serious results, so be sure to follow thedirections for usage.

Warning

Procedures which may lead to dangerous conditions and cause death or serious
injury, if not carried out properly.

CautionProcedures which may lead to dangerous conditions and cause superficial tomedium injury or physical damage or may degrade or damage the product, if not
carried out properly.

3Warning• To prevent an electric shock or fire, only Shinko or other qualified service personnel may handle the innerassembly of each instrument.
• To prevent an electric shock, fire or damage to the instruments, parts replacement may only be undertaken
by Shinko or other qualified service personnel.

SAFETY PRECAUTIONS• To ensure safe and correct use, thoroughly read and understand this manual and each manual of theconnected instruments before using this instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring equipment.
Verify correct usage after purpose-of-use consultation with our agency or main office. (Never use this
instrument for medical purposes with which human lives are involved.)• External protection devices such as protective equipment against excessive temperature rise, etc. must
be installed, as malfunction of this product could result in serious damage to the system or injury to
personnel. Also proper periodic maintenance is required.
• This instrument and the connected instruments must be used under the conditions and environment
described in this manual and each manual of the connected instruments. Shinko Technos Co., Ltd. does
not accept liability for any injury, loss of life or damage occurring due to the instruments being used underconditions not otherwise stated in this manual and each manual of the connected instruments.

Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the manufacture of
weapons of mass destruction (i.e. military applications, military equipment, etc.), please investigate the end
users and the final use of this instrument. In the case of resale, ensure that this instrument is not illegally
exported.

4

This manual describes the Touch Screen Programmable Controller (PCT-200), consisting ofTouch Screen (LT3300) and Temperature Control Module (WCL-13A).
• Touch Screen LT3300: Made by Digital Electronics Corp.
• Temperature Control Module WCL-13A: Made by Shinko Technos Co., Ltd.

Contents
1. Overview ---------------------------------------------------------------------------------------------- 6
2. External Dimensions (Scale: mm) ------------------------------------------------------------ 7
3. Wiring -------------------------------------------------------------------------------------------------- 9
4. Communication Setting -------------------------------------------------------------------------- 9
5. Creating Pattern Data
5.1 USB Memory Stick --------------------------------------------------------------------------- 10
5.2 Editing CSV File ------------------------------------------------------------------------------- 10
5.3 Application Software
5.3.1 Converting to the Binary File -------------------------------------------------------- 12
5.3.2 Converting to the CSV File ---------------------------------------------------------- 13
6. Touch Screen Monitoring Display
6.1 DisplayConfiguration ------------------------------------------------------------------------- 14
6.2 Opening Display ------------------------------------------------------------------------------ 16
6.2.1 Interface Language -------------------------------------------------------------------- 16
6.2.2 WCL-13A Spec Modification Display --------------------------------------------- 17
6.2.3 Menu Window -------------------------------------------------------------------------- 17
6.3 Operation --------------------------------------------------------------------------------------- 18
6.4 Operation Monitoring ------------------------------------------------------------------------ 19
6.5 Program Setting
6.5.1 Temperature, Time Settings -------------------------------------------------------- 20
6.5.2 PID, Wait, Alarm Settings ----------------------------------------------------------- 21
6.5.3 Time Signal Setting ------------------------------------------------------------------- 22
6.6 Block Setting
6.6.1 PID Block Setting ---------------------------------------------------------------------- 23
6.6.2 Time Signal Block Setting ----------------------------------------------------------- 23
6.6.3 Wait Block Setting --------------------------------------------------------------------- 24
6.6.4 Alarm Block Setting ------------------------------------------------------------------- 24
6.7 Engineering Setting -------------------------------------------------------------------------- 25
6.8 Lock Selection, Selection after Power Restoration ---------------------------------- 26
6.9 Zone Control Function
6.9.1 SV Bias Setting ------------------------------------------------------------------------ 27
6.9.2 Monitoring all WCLs ------------------------------------------------------------------ 27
6.10 Pattern Selection and USB Management -------------------------------------------- 28
7. Logging Function -------------------------------------------------------------------------------- 29

5

1. Overview
When delivered, please check the contents of package to confirm that all items you have ordered arecontained as follows.
Model
PCT-20Temperature control
moduleControl output--

1 to 9Number of connected WCL-13A units
Relay contact output(WCL-13A-RA/MM PCT)

R
S
A

Temperature control module
Socket1
2

Non-contact voltage output (WCL-13A-SA/MM PCT)
Direct current output(WCL-13A-AA/MM PCT)For Y-type terminals only; Finger-safe (ASK-001-1)
For Y-type and ring terminals (ASK-002-1)

Contents of Package
Model
Touch Screen

LT3300-S1-D24-K
WCL-13A-RA/MM PCT
WCL-13A-SA/MM PCT
WCL-13A-AA/MM PCT

Temperature control module (*1)
Connection cable(Between Touch Screen - WCL-13A)
Connection cable (Between WCL-13As)
Switching power supply 24 V DC
Cable for front mounted USB port
EX module
MIL socket connector harness
USB memory stickCWD-C5

Quantity1
1 to 9
(*2)
1

CDD
(*3)
PS3X-D24AFG
1
CA5-USBEXT-01
1
EXM-DDO16UK
1
HIFS-SY-SB-20-5
1
U3C-HP16G
1
ASK-001-1
Socket (*1)(*3)
ASK-002-1(*1) Specify one when ordering. When WCL units are connected, their specifications should be the same.
(*2) For Instrument Number setting for the WCL-13A, start from 1 (one) and progress chronologically.
(*3) When multiple WCL units are ordered, corresponding amount of cables and sockets are included.
Configuration Example

A maximum of 9 unitsof WCL-13A

Touch Screen
DIO connector???

Connection cable
CWD-C5(Touch Screen - WCL)Power connector(included)+

When connecting multiple WCL units,
use included cables (CDD).
For details, refer to the Instruction Manual
for the WCL-13A.

Connection cable CDD(Between WCL units)-

Cable for front mounted USB port
(CA5-USBEXT-01)
Switching power supply
PS3X-D24AFG

Socket for WCL
8

7

4

3

6

-

2

14

5

+

13 Power

supply100 to 240 V AC
24 V DC

12- 11-CH1 input

1

+
10

+-

EX module is mounted to the back of Touch
Screen.
EX module

MIL socket connector
harness

9+

Trans- CH1
mission output
outputTouch Screen

62. External Dimensions (Scale: mm)Touch Screen
Panel Cutout
Panel thickness: 1.6 to 5.0 mm

Switching Power Supply
5-M3.5 terminal screw
2-M3 (Depth 2.5 - 4.0)LED

9.5 (Terminal pitch)

Terminal cover

LED
Potentiometerfor changing
output voltage

2-M3 (Depth 2.5 - 4.0)

7

Cable for Front Mounted USB Port

Panel Cutout
Panel cutout (A) is recommended.
Panel cutout (B) can be used, however, mounting may become loose, causing drip-proof rating

to become invalid.
(A)

(B)

Temperature Control Module WCL-13A- A/MM PCT

Connection Cable CWD-C5 (Between Touch Screen - WCL-13A)

Connection Cable CDD (Between WCL-13A - WCL-13A)

8

3. Wiring
Digital Electronics Corp. Proface Touch Screen LT3300
DIO Connector
Pin Connection

Pin No.
Signal Name
Pin No.
A1
IN1
BCD 100-2
B1
A2
IN3
BCD 100-8
B2
A3IN5
BCD 101-2
B3
A4IN7
BCD 101-8
B4
A5
IN9
STOP
B5
A6
IN11
FAST
B6
A7
IN13
B7
A8
IN15
B8
A9
NC
B9
A10
Sink output type: NC
B10
A11
Sink output type: 0 V
B11
A12
OUT1
AL1
B12
A13
OUT3
AL3
B13
A14
OUT5
TS1
B14
A15
OUT7
TS3
B15
A16
OUT9
TS5
B16
A17
OUT11
TS7
B17
A18
OUT13
TS9
B18
A19
OUT15
TS11
B19
(*) Contact CLOSED: RUN, Contact OPEN during RUN: HOLD
EX Module Connector
Pin Connection

Pin No.
Signal Name
20
Q0
TS12
18
Q1
TS13
20
19
16
Q2
TS14
14
Q3
TS15
12
Q4
TS16
10
Q5
TS17
8
Q6
TS18
6
Q7
TS19
4
COM(-)1
2
2+V
For details, refer to [LT3300 Series Hardware Manual].

Pin No.
19
17
15
13
11
9
7
5
3
1

Signal Name
IN0
BCD 100-1
IN2
BCD 100-4
IN4
BCD 101-1
IN6
BCD 101-4
IN8
RUN/HOLD(*)
IN10
ADV
IN12
IN14
COM
Sink output type: +24 V
Sink output type: 0 V
OUT0
P.END
OUT2
AL2
OUT4
AL4
OUT6
TS2
OUT8
TS4
OUT10
TS6
OUT12
TS8
OUT14
TS10

Signal Name
Q8
TS20
Q9
Q10
Q11
Q12
Q13
Q14
Q15
COM(-)+V

4. Communication Setting
Instrument Numbers should be individually set from the second WCL-13A and all following units.
Set the Instrument Numbers from 2 (two) via the keypad, progressing chronologically.
For detailed usage and options, refer to the Instruction Manual for the WCL-13A.
(See Section 7. Key Operation Flowchart.)
Instruction Manual for the WCL-13A can be downloaded from Shinko Website as follows.
http://www.shinko-technos.co.jp/e/
Support & Downloads
Downloads
Manuals
Character (*)

Name, Function, Setting Range
Factory Default
Instrument number
1• When using more than one WCL-13A in Serial communication, sets the instrument
number to each unit individually.
• Available when Serial communication (C5 option) is specified when ordering.
• Setting range: 0 to 95(*) Characters are indicated on the CH1 PV/SV display of the WCL-13A.

9

5. Creating Pattern Data
5.1 USB Memory Stick
The following files are included in the USB memory stick provided.
CSV.Convert_V2.01.exe (Application software)
PARA.csv(Pattern data file)
Data folder(Data folder)
SAMP01 folder(Logging folder)ja folder(Folder for Japanese OS)
ja-JP folder(Folder for Japanese OS)

In the Data folder, the following files are included.
B_DATA.bin
Each block data file of PID, Wait, Alarm and Time signal
PTN001.bin to PTN100.bin
Data file of 100 steps per pattern[Note] Never change the file name of B_DATA.bin, and PTN001.bin to PTN100.bin.
The Touch Screen will not be able to recognize them.
Logging data will be saved in the SAMP01 folder.
Csv.Convert_V2.01.resources.dll (File for Japanese OS) is included in the ja folder.
Csv.Convert_V2.01.resources.dll (File for Japanese OS) is included in the ja-JP folder.
5.2 Editing CSV FileIn the PARA.csv file, a maximum of 100 patterns 100 steps/pattern of data (temperature, time and eachblock number) can be created.
Factory default: 0 (for all)
Edit the PARA.csv file using the commercially available spreadsheet software.
[Note] When creating data, enter English letters and numbers.

Program No.: Pattern number (Editing impossible)
STEP:
Step number (Editing impossible)
Temp:
Temperature
Setting range: Depends on the input type of the WCL-13A (p.17)
Time:
Time
Setting range: 0 to 9959 (0 to 99 hr 59 min)PID:
PID block number
Setting range: 0 to 9
Wait:
Wait block number Setting range: 0 to 9
Alarm:
Alarm block number Setting range: 0 to 9
TS1 to TS20: Time signal 1 to 20 Time signal block setting range: 0 to 15 (Hexadecimal: 0 to F)

10

Editing Pattern Data
Create pattern data by editing the PARA.csv file using the commercially available spreadsheet
software.
When creating data, enter English letters and numbers.
For numbers with a decimal point, ignore the decimal point when setting temperature.
(e.g.) For setting 100.0 : Enter 1000. (Decimal point is automatically inserted.)
For setting time of 1 hour 20 minutes: Enter 120.
For time signal block numbers 10 to 15, use hexadecimal figures A to F.
(e.g.) Temperature rises to 100
Pattern
No.

STEP
No.

Temperature

1
1

1
2

1000
1000

for 30 minutes, and is maintained at 100

Time

PID
Block
No.

Wait
Block
No.

Alarm
Block
No.

30
60

1
2

1
0

1
2

Time Signal
1

1
1

2

3

10
2

11
15

Contents of CSV file

[Note] For the PARA.csv, its file name can be changed.
Always back up data.
Shinko Technos Co., Ltd. is not responsible for loss of data.

11

for 60 minutes
4005.3 Application Software
5.3.1 Converting to the Binary File
Load the CSV file edited, then convert it to the binary file.
Touch Screen cannot recognize the CSV file, so be sure to convert it to the binary file.
Start the application by double-clicking CSV.Convert_V2.01.exe,
then select [Convert to *.bin] tab.
Convert to *.bin tab

(1)
(2)
(3)
(4)

(5)
No.
(1)
(2)
(3)

Item
CSV file name
BIN save folder
Select an item

Description
Selects the CSV file.
Selects a location for the BIN file folder.
Creating all patterns Creates data for 1 to 100 patterns.
Specifying patternnumbers(4)(5)

Create *.bin
Close

Creates data for the specified pattern number
range.

Specifying one
Creates data for one specified patternpattern number
number.
Creates the binary files.
Closes the application.

Display example after the file is converted

125.3.2 Converting to the CSV File
Binary files saved on the Touch Screen can be converted to the CSV file.
Start the application by double-clicking CSV.Convert_V2.01.exe,
then select [Convert to *.csv] tab.
Convert to *.csv tab

(1)
(2)
(3)
(4)

(5)
No.
(1)
(2)
(3)

Item
CSV file name
BIN read folder
Select an item

Description
Selects the CSV file.
The binary files in the indicated folder are loaded.
Creating all patterns
Creates data for 1 to 100 patterns.
Specifying patternnumbers(4)(5)

Create *.csv
Close

Creates data for the specified patternnumber range.

Specifying one
Creates data for one specified patternpattern number
number.
Creates the CSV file.
Closes the application.

Display example after the file is converted

136. Touch Screen Monitoring Display
6.1 Display Configuration
Touch Screen monitoring displays are shown below.
Opening display is indicated only when the power is turned ON.
After power-on, make a selection in the following order.
(1) Interface language, (2) WCL-13A spec modification, (3) Various settings from the menu button(An arrow) means that the display moves to the next page.

[Opening display][WCL-13A spec modification]
Turn the power OFF,then ON again.

Moves to the following displays by pressing the menu button
See P.16.
[Interface language]
EnglishSee P.18.
[Operation]

See P.19.
[Operation
monitoring].

See P.27.
[Monitoring all WCLs]

See P.20.
[Program setting]Temperature, Time

[Interface language ]Chinese[Program setting]
PID, Wait, Alarm[Interface language]
Korean[Program setting]Time signal

14

See P.23.
[Block setting]
PID

See P.25.
[Engineering setting][Block setting]Time signal[Alarm 1 (type, hys)
setting]
The same applies to
Alarms 2 to 4

See P.26.
[Setting lock selection,
Selection after restoration][Block setting]
Wait[Block setting]
Alarm

15

See P.27.
[SV bias setting]See P.27.
[Pattern selection &USB management]

6.2 Opening Display
Opening display is indicated only when the power is turned ON.
(1)

(2)
(4)
(3)

No
(1)
(2)

(3)
(4)

Item
Menu button
WCL-13A spec
modification (press 3
sec)
Interface language

Description
Opens the menu window.
Changes the input of the WCL-13A.

Number of connected
WCL-13A units

Sets the number of connected WCL-13A units.

Selects a language to be indicated on the display.

6.2.1 Interface Language
Japanese, English, Chinese, Korean languages are available, and they can be switched by pressing
the [Interface language (表示言語)] button.

[Japanese][English][Korean][Chinese]166.2.2 WCL-13A Spec Modification Display

(1)

(2)

(3)
No
(1)
(2)
(3)

Item
Input type
Input type selection
OK

Input No.
0
1
2
3
4
5
6
7
8
9
10
11
12
13
14
15
16
17

K
K
J
R
S
B
E
T
N
PLC
Pt100
JPt100
Pt100
JPt100K
K
J

Description
Indicates current input type of the WCL-13A.
Changes the input type of the WCL-13A. Enter an input number.
Press [OK] after the input type selection is finished.
[Note] If an input type is changed, turn the power to the Touch
Screen main unit OFF, then ON again.
The changed input type will then become valid.

Input Type-200 to 1370
-199.9 to 400.0-200 to 1000
0 to 1760
0 to 1760
0 to 1820-200 to 800-199.9 to 400.0-200 to 1300
0 to 1390
0 to 2315-199.9 to 850.0-199.9 to 500.0-200 to 850-200 to 500-320 to 2500
-199.9 to 750.0-320 to 1800

Input No.
18
19
20
21
22
23
24
25
26
27
28
29
30
31
32
33
34
35

R
S
B
E
T
N
PLC
Pt100
JPt100
Pt100
JPt1004 to 20mA DC
0 to 20mA DC
0 to 1V DC
0 to 5V DC
1 to 5V DC
0 to 10V DC

6.2.3 Menu Window
By touching the Menu button, the menu window appears as follows.

17

Input Type
0 to 3200
0 to 3200
0 to 3300-320 to 1500
-199.9 to 750.0-320 to 2300
0 to 2500
0 to 4200-199.9 to 999.9-199.9 to 900.0-300 to 1500
-300 to 900-1999 to 9999
-1999 to 9999
-1999 to 9999
-1999 to 9999
-1999 to 9999
-1999 to 9999

6.3 Operation
(5)
(1)

(2)

(3)
(4)

No.
(1)

Item
Operation
buttons

Description
Controller is operated by pressing buttons.
Item
Description
CONTINUE
Lights when RUN is pressed after power restoration.
Logging
Selects logging Start/Stop.
RUN
Starts program.
STOP
Stops currently performing program.
PVst/SVst
Selects PV start/SV start.
ADV
During program control, interrupts performing step, andproceeds to the next step.
FAST
During program control, speeds up step time progression
60 times faster than usual.
HOLD
During program control, pauses time progression.
By touching again, HOLD is released.

(2)

StatusMonitoring

Shows the following status of programmable controller.
Item
Description
Pattern No.
Indicates currently selected pattern No.
STEP No.
Indicates currently performing step No.
PV
Indicates current PV.
SV
Indicates current SV.
R.TIME
Indicates Remaining Time of currently performing step.
TS.OUT
Indicates Time Signal Output status.
ALM
Indicates Alarm output status.

(3)
(4)

AT buttons
REPEAT

Performs/Cancels AT (auto-tuning).
Use this button to perform program control repeatedly.
No repetition occurs when set to 0 (zero).
Number of repetitions: 0 to 99(5)
Menu buttons Opens the menu window.
[Note] The status of program control (during RUN) is the same as Lock status. See p.26.
PV Start/SV Start Selection
PV start
When program control starts, SV and step time are advanced to the PV, thenprogram control starts.
SV start
When program control starts, control starts from 0 (zero) of SV.
[Note] Selection of PV start/SV start will be maintained even if power is turned OFF.
18

6.4 Operation Monitoring
(6)
(1)

(2)

(4)
(5)

(3)

No.
(1)

Item
Statusmonitoring

Description
Indicates the following status of programmable controller.
Item
Description
PTN No.
Indicates currently performing pattern No.
STEP No.
Indicates currently performing step No．
PV
Indicates current PV.
SV
Indicates current SV.
MV
Indicates current MV.
TIME
Indicates remaining time of currently performing step.
TS.OUT
Indicates time signal output status.
ALM
Indicates alarm output status.

(2)

Pattern graph(3)

Logging
statusSTEP display
arrowsPWR
Menu button

Pattern temperatures are plotted in graph.
During automatic operation, currently performing step is indicatedby flashing grid lines for that step.
Indicates [Possible] while logging, and [Impossible] while logging is
stopped.
Selects steps to be displayed.

(4)
(5)
(6)

Flashes when power is restored.
Opens the menu window.

196.5 Program Setting
6.5.1 Temperature, Time Settings
(3)
(8)
(1)

(4)

(2)

(5)

(6)

(7)
No.
(1)
(2)
(3)
(4)
(5)
(6)

(7)
(8)

Item
PTN No.
STEP
STEP No.
STEP display
arrowsPattern graph
Pattern setting 1

Pattern setting items
selection
Menu button

Description
Indicates pattern No.
Indicates step No.
Sets step No.
Selects steps to be displayed.
Indicates a pattern in accordance with the SV.
Sets temperature and time to each step.
[Setting range]• Temperature: Depends on the scaling setting of the WCL-13A.
• Time: 0 hr 0 min to 99 hr 59 min
Selects pattern setting items.
Opens the menu window.

20

6.5.2 PID, Wait, Alarm Settings
(3)
(8)
(1)

(4)

(2)

(5)

(6)

(7)

No.
(1)
(2)
(3)
(4)
(5)
(6)

(7)
(8)

Item
PTN No.
STEP
STEP No.
STEP display
arrowsPattern graph
Pattern setting 2

Pattern setting items
selection
Menu button

Description
Indicates pattern No.
Indicates step No.
Sets step No.
Selects steps to be displayed.
Indicates a pattern in accordance with the SV.
Sets block No. (PID, Wait, Alarm) to each step.
[Setting range]• PID block: 0 to 9
• Wait block: 0 to 9
• Alarm block: 0 to 9
Selects pattern setting items.
Opens the menu window.

21

6.5.3 Time Signal Setting
(8)
(1)

(3)
(4)

(2)

(5)

(6)

(7)

No.
(1)
(2)
(3)
(4)
(5)
(6)

(7)
(8)

Item
PTN No.
STEP
STEP No.
STEP display arrows
Pattern graph
Pattern setting 3

Pattern setting items
selection
Menu button

Description
Indicates pattern No.
Indicates step No.
Sets step No.
Selects steps to be displayed.
Plots a pattern in accordance with the SV.
Sets Time signal block No. to each step.
[Setting range]• Time signal block: 0 to F
Time signal number can be set from 1 to 20 (TS1 to TS20).
Selects pattern setting items.
Opens the menu window.

226.6 Block Setting
6.6.1 PID Block Setting
(4)

(1)

(2)

(3)

No.
(1)
(2)

Item
Block No.
PID parameters

(3)
(4)

Block selection
Menu button

Description
Sets PID block No.
Sets PID parameters.
[Setting range]
Proportional band: 0 to 9999 (Decimal point depends on the
input range.)
Integral time:
0 to 1000
Derivative time: 0 to 300
ARW:
0 to 100
Selects block setting items.
Opens the menu window.

6.6.2 Time Signal Block Setting
(4)

(1)

(2)

(3)

No.
(1)
(2)

(3)
(4)

Item
Block No.
Time signal blockparameters

Block selection
Menu button

Description
Sets Time signal block No.
Sets time signal parameters.
[Setting range]
ON time: 0 hr 0 min to 99 hr 59 min
OFF time: 0 hr 0 min to 99 hr 59 min
Selects block setting items.
Opens the menu window.
23

6.6.3 Wait Block Setting
(4)
(1)
(2)

(3)

No.
(1)
(2)

Item
Block No.
Wait blockparameter(3)(4)

Block selection
Menu button

Description
Sets Wait block No.
Sets the Wait parameter.
[Setting range]
Wait: 0 to 1000 (Decimal point depends on the input range.)
Selects block setting items.
Opens the menu window.

6.6.4 Alarm Block Setting
(4)
(1)
(2)

(3)

No.
(1)
(2)

Item
Block No.
Alarm blockparameters

(3)
(4)

Block selection
Menu button

Description
Sets Alarm block No.
Sets alarm parameters.
[Setting range]
Alarm 1 value: Depends on the input range and Alarm 1 type.
Alarm 2 value: Depends on the input range and Alarm 2 type.
Alarm 3 value: Depends on the input range and Alarm 3 type.
Alarm 4 value: Depends on the input range and Alarm 4 type.
Selects block setting items.
Opens the menu window.

24

6.7 Engineering Setting
(5)

(1)

(2)
(3)
(4)

No.
(1)

(4)

Item
Alarm type andhysteresis setting
Proportional cycle
setting
Sensor correction
setting
Save set value(5)

Menu button

(2)
(3)

Description
Indicates each alarm type and hysteresis setting display.
Sets proportional cycle.
Setting range: 1 to 120 seconds
Sets the sensor correction value.
Setting range: -100.0 to 100.0
If this key is pressed after values (alarm type, alarm hysteresis,
proportional cycle, sensor correction value) are changed, they will
be written in the non-volatile memory of the WCL-13A.
If this key is not pressed, those values will be discarded whenpower to the WCL-13A is turned OFF.
Opens the menu window.

Alarm 1 setting display (The same applies to Alarm 2, 3, 4.)
(3)

(1)
(2)

No.
(1)

Item
Alarm type selection(2)

Alarm hysteresis
setting
OK(3)

Description
Selects an alarm type.
Setting range: Alarm types 0 to 9
Sets the alarm hysteresis.
Setting range: 0.1 to 100.0
Returns to the Engineering display.

25

6.8 Setting Lock Selection, Selection after Power Restoration
(7)
(3)

(3)
(1)

(2)
(6)

(4)

(6)

(5)
When [Lock] is pressed.
When RUN after power restoration is selected.
No.
(1)
(2)
(3)
(4)
(5)
(6)
(7)

Item
Lock
UnlockUnlocked/Lockedindicator
Restore-RUN
Restore-STOP
Restore-RUN/
Restore-STOP
indicator
Menu button

Description
Locks the set values (program setting, block setting, proportional
cycle, sensor correction) to prevent change.
Releases the lock.
Indicates current Unlock/Lock status.
The status after power restoration is RUN.
The status after power restoration is STOP.
Indicates current status (RUN or STOP) after power restoration.

Opens the menu window.

[Note] The status of program control (during RUN) is the same as Lock status.

26

6.9 Zone Control Function
6.9.1 SV Bias Setting
Sets the SV bias value for Zones 2 to 9.
For Zones 2 to 9, each SV bias value is added to Zone 1 SV, which becomes the SV.
For Zone 1 (instrument No. 1), set the SV.
From Zones 2 to 9, set the SV Bias value (SV adds the SV of Zone 1 to the SV Bias value.)
(2)
(1)

No.
(1)

Item
SV bias value(2)

Menu button

Description
Sets SV bias value for the connected WCL-13A units.
Setting range: -500 to 500 (Decimal point depends on the input
range.)
Opens the menu window.

6.9.2 Monitoring all WCLs
(4)

(1)
No.
(1)
(2)

PV
SV

(3)
(4)

MV
Menu button

(2)

(3)

Item

Description
Indicates PV of each zone.
For Zone 1, indicates SV.
For Zones 2 to 9, indicates SV bias value.
(See Section 6.9.1 SV bias setting.)
Indicates MV of each zone.
Opens the menu window.

27

6.10 Pattern Selection and USB Management
(4)
(1)

(2)

(3)

No.
(1)

Item
Pattern selection(Pattern loading)

(2)

USB
management(Pattern saving)

(3)

USB
management(Block data)
Menu button(4)

Description
Loads the selected pattern data from the USB memory stick.
Select the pattern number with the UP or DOWN Key, or enter thepattern number directly by pressing the number. (If the pattern number
is pressed, the onscreen numerical keyboard appears.)
Program control is performed for the pattern number selected here.
The pattern number cannot be changed while running.
Saves the selected pattern data in the specified pattern number on the
USB memory stick.
Select the pattern number with the UP or DOWN key, or enter thepattern number directly by pressing the number. (If the pattern number
is pressed, the onscreen numerical keyboard appears.)
By changing the pattern number, data can be copied.
Loads block data (PID, Wait, Alarm, Time signal) from the USB memory
stick, or saves it on the USB memory stick.
Opens the menu window.

[Note]
Buttons (Load, Save, Pattern selection) will beeffective by pressing them twice.
Confirm the message after pressing the button once,
then press the button again.

Error message will appear if data cannot be loaded.
Data may be non-existent.
• If a new pattern data is loaded, the pattern data selected on the Touch Screen will be erased.
Save the pattern data selected on the Touch Screen if necessary. (If a set value is changed on
the screen and is not saved on the USB memory stick, the data will be lost.)• Data SAVE means that the binary data saved in the data folder will be updated.
The CSV file cannot be saved via Touch Screen.

28

7. Logging Function
Using the sampling function of the LT3300, the registered data can be logged in a constant period.
The logged data will be saved in the USB memory stick in the CSV file format.
Logging Conditions
Item
Data acquisition period
Timing to execute File SAVE

Output file save destination
Output file name

Description
10 seconds
Every 3 hours
When automatic operation
is finished.
SAMP01
SAxxxxx.csv

Remarks
During automatic operation only
Only when logging Effective isselected.

A file is created when logging starts,and the file is saved (SA00001.csv)
when logging stops.
Every time logging stops, a new filewith a serial number will be created.
(The file number is entered at xxxxx
section.)When power is turned ON, the filenumbered 00001 will be used, anddata will be saved in it. However,previous data will be retained and the
new data will be added to it.

Output File Format
When PV, SV are 1.0 (with a decimal point).
Output status 0: OFF, 1: ON
TS item output status: HexadecimalDate
2008/6/13
2008/6/13
2008/6/13
2008/6/13

Time PTN STEP PV SV ALM1 ALM2 ALM3 ALM4 TS1-16 TS17-20
9:08:20
1
1
10
10
1
1
1
1
C000
000F
9:08:30
1
1
10
10
1
1
1
1
C000
000F
9:08:40
1
1
10
10
1
1
1
1
C000
000F
9:08:50
1
1
10
10
1
1
1
1
C000
000F

29

30

31

SHINKO TECHNOS CO., LTD.
OVERSEAS DIVISION
Head Office : 2-5-1, Senbahigashi, Minoo, Osaka, Japan
http://www.shinko-technos.co.jp/e/
URL:
Tel: +81-72-727-6100
E-mail:overseas@shinko-technos.co.jp
Fax: +81-72-727-7006
No. PCT21E6 2017.11