Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

Common Specification for output data of Ophthalmic
Examination Equipment 01 （JOIA Std.)- Common Specification for output data of Refractometer / Keratometer / Tonometerand Lensmeter –1. Data classifications
1-1. Handling of Files
It is possible to handle multiple data classifications such as refractometer, keratometer,
tomometer and lensmeter within one XML file. Each data classification shall only be sent
once in a single file (for example, it is forbidden to send two <LM> data classification in onefile). The data of refractometer, keratometer, tonometer and lensmeter can be handled in
one file or they can be separated in individual files. The file has <Ophthalmology> tag as a
top tag, and tag of each data classification exists in the top tag. The example of dataclassification constitution in the XML file is shown as below.

■Example of data classification constitution in the file
<Common><Common><Common><Measure type=”REF”><Measure type=”REF”><Measure type=”TM”><Measure type=”KM”><Measure type=”KM”><Measure type=”LM”><Measure type=”TM”>

File1<Measure type=”LM”>

File2

File3

* The file has ＜Ophthalmology＞tag as a top tag.
* It is possible to handle all the data classifications in one file, or they can be separated
by files.
* It is also possible to handle the data classifications specified in the common
specification for data output of other Ophthalmic Examination Equipment (02 - ) in one
file.
* Implementations may encounter data classifications that are not included in the JOIA

1／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)specification,

therefore,

implementations

SHALL

ignorenon-recognized

2／31

data

classifications and process the known data classifications accordingly.

1-2. Handling of data classifications
The data classifications of refractometer, keratometer, tonometer and such aremaintained as attribute value (type) of ＜Measure＞ tag.

■Attribute value and classification
Attribute value(type)

Data classification

REF
KM
TM
LM
・

Refractometer data
Keratometer data
Tonometer data
Lensmeter data

・The example with refractmeter, keratometer, tonometer and lensmeter data are shown as
below.
ex) In the case of having refractometer, keratometer,
tonometer and lensmeter data.
<Measure type="REF"><--- Refractometer data

・・・・</Measure><Measure type="KM"><--- Keratometer data

・・・・</Measure><Measure type="TM"><--- Tonometer data

・・・・</Measure><Measure type="LM"><--- Lensmeter data

1-3. Character code
The character code within XML file is UTF-8 or UTF-16.

1-4. Description of style sheet
Description of the style sheet in the XML file is optional.

2. Common data

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

2-1. Handling of common data
Common data such as patient information is included in ＜Common＞ element. One
common data exists in one file. (See “４．Tag definition” on common data elements)■Example of common data in a file
<Common><Company><ModelName><MachineNo>
・・<Patient><No><ID><Name>・・・</Common><Measure type=”REF”>

3／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

4／31

3. Data definition
3-1. Data type definition
Tag elements of each data classification such as refractometer, keratometer, tonometerand lensmeter are defined by XML schema. Data types of values on each tag are also
defined by the schema. Moreover, it is determined by the schema definition whether it is arequired tag or permit overlaps.

3-2. Handling of unit
The unit of the numerical data is described as attribute value "Unit".
ex) Sphere data of refractometer<Sphere unit="D">0.25</Sphere>

4. Tag definition
The items described in the necessity field have following meanings.
Any: Tags are optional. They can be omitted.
No sign: No tags or values are necessary.
△: Tags are necessary even if no value is entered.
○: Tags and values are necessary.

* Indicates that lower tags are necessary when an upper tag exists.

■Ophthalmology tag
Tag name<Ophthalmology>

Description
Ophthalmic data

Data type

Necessity-○

Data type

Necessity■Common data
Tag name<Common>Description
Common tag<Company><ModelName><MachineNo><ROMVersion><Version>

StringCompany name(64 characters)
stringModel name(64 characters)
No. for distinction among the devices
stringof the same model.
(64 characters)
No. for distinction among thestringmodels with the same ROM version. (64 characters)
XML specifications version
string(fixed)(64 characters)

○
○
○
△
△
○

<Date>

Date (year, month, day)

date

○

<Time>

Time (hour, minute, second)

time

○

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<Patient>

5／31

Patient
Patient No.(number of the order
string<No.>of examination)(64 characters)
string<ID>
Patient ID(64 characters)<PatientIDSource> Patient ID Source
String*1(unreliable, manual,auto device,auto patient list)
string<FirstName>
Patient’s first name(64 characters)
string<MiddleName>
Patient’s middle name(64 characters)
string<LastName>
Patient’s last name(64 charactersstring<Sex>
Patient’s sex(M,F,O)○<Age>

Patient’s age

int

△

<DOB>

Patient’s date of birth△<NameJ1>

Patient’s name in other languages<NameJ2>

Patient’s name in other languages

datestring(64 characters)
string(64 characters)<Operator>

Operator<No.>

Operator No.

<ID>

Operator ID

△
○

△
△
△
△

△
△

string(64 characters)
string(64 characters)

Unique tag of each company can be optionally defined here.

△
△

any

*1: For data element < PatientIDSource >, the terms have the following meaning:“unreliable” = not a valid Patient ID (i.e., a device generated exam number)“manual” = Patient ID manually entered on the device“auto device” = Patient ID automatically entered on the device (i.e., barcode scanner, etc.)“auto patient list” = Patient ID selected from a patient list generated by the Patient Source in the clinic (i.e.,
Practice Management System)■Refractometer data
Tag name<Measure type="REF">

Description
Measurement type tag

Data type

Necessity-○<VD unit="mm">Corneal vertex distancedouble○<DiopterStep unit="D">

Diopter increment

double△<AxisStep unit="deg">

Angle increment

int△<CylinderMode>*1

Cylinder mode (-,+,mix)△<CycloplegicMode>

Cycloplegic Modestring(-,+,mix)
string(cycloplegic,
non-cycloplegic)

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

6／31

<REF>
<R>
<List No=" "><Sphere unit="D"><Cylinder unit="D"><Axis unit="deg">

Right eye measurement value
Measurement value(1 to n times)
Spherical refractive
power (D)Cylindrical refractive
power (D)<SE unit="D">

Cylinder axis (°)
Spherical equivalent
refraction (D)<CataractMode>

CAT mode<IOLMode>

IOL mode<ConfidenceIndex>*2

Confidence index
Measurement value(In case of error )<List No=" "><Error>*2

Error Message<Sphere unit="D">Median values
Spherical refractive power(D)
Cylindrical refractive
power (D)<Median>*3<Cylinder unit="D"><Axis unit="deg"><SE unit="D"><L><List No=" "><Sphere unit="D"><Cylinder unit="D"><Axis unit="deg">

Cylinder axis (°)
Spherical
equivalent
refraction (D)Left eye measurement value
Measurement value(1 to n times)
Spherical refractive power(D)
Cylindrical refractive
power (D)<SE unit="D">

Cylinder axis (°)
Spherical
equivalent
refraction (D)<CataractMode>

CAT mode<IOLMode>

IOL mode<ConfidenceIndex>*2

Confidence index
Measurement value(In case of error )<List No=" "><Error>*2

Error Message<Sphere unit="D">Median values
Spherical refractive power(D)
Cylindrical refractive
power (D)<Median>*3<Cylinder unit="D"><Axis unit="deg"><SE unit="D">

Cylinder axis (°)
Spherical equivalent
refraction (D)-○double○double○int○

doublestring(64 characters)
string(64 characters)
string(64 characters)△

string(64 characters)

△
△
△

△

double○double○int○

double△-○double○double○int○

doublestring(64 characters)
string(64 characters)
string(64 characters)△

string(64 characters)

△
△
△

△

double○double○int○

double△Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

<PD>

7／31

<WorkingDistance unit="cm">Pupillary distanceViewing distance used for
Near PD

int△<Distance unit="mm">

PD at optical Infinity

double○<Near unit="mm">

PD at working distancedouble△Unique tag of each company can be optionally defined here.

any*1: For data element <CylinderMode>, the terms have the following meaning:“+” = the cylinder value is always positive“-” = the cylinder value is always negative"mix" = the cylindrical power by positive reading when the refractive power is positive for any meridional
angle and the cylindrical power by negative reading in other cases*2: <ConfidenceIndex> and <Error> data elements are vendor specific strings and JOIA does not define “coded”
values for these data elements.
*3: For the scenario where multiple refractive measurements are sent AND the optional <Median> section is not
sent, the EHR is strongly recommended to calculate the median from the list of measurements and store themedian in its internal data base.

■Keratometer data
Tag name

Description

Data type

Necessity-○<DiopterStep unit="D">

Measurement type tag
Corneal refractive
power increment

double△<AxisStep unit="deg">

Angle increment

int△<CylinderMode>*1

Cylinder mode (-,+)
Cornealrefractiveindex

string(-,+)△

double○<Measure type="KM"><RefractiveIndex ><KM>
Right eye measurement
value
Measurement value(1 to n times)<R><List No=" ">
<R1>

Flat
<Radius unit="mm">

-

○

-

○

double○<Power unit="D">Curvature radiusCorneal refractive
power

double○<Axis unit="deg">*2

Axis angle

int

○

Steep

-

○

double○double○<R2><Radius unit="mm"><Power unit="D">Curvature radiusCorneal refractive
powerCommon Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<Axis unit="deg">*2<Average><Radius unit="mm"><Power unit="D"><Cylinder><Power unit="D"><Axis unit="deg"><List No=" "><Error>*3<Median><R1>

Axis angle

int○

Average of R1 and R2-○

double○double△-○double○int○

Error Message

string(64 characters)△

Median values-

Curvature radiusCorneal refractive
power
Cylinder power
Cylindricalrefractive power
Cylinder axis angle
Measurement value(In case of error )Flat<Radius unit="mm">

8／31

-

○

double○<Power unit="D">Curvature radiusCorneal refractive
power

double○<Axis unit="deg">*2

Axis angle

int

○

Steep

-

○

double○double○<R2><Radius unit="mm"><Power unit="D">Curvature radiusCornealrefractive
power<Axis unit="deg">*2

Axis angle

int○

Average of R1 and R2-○

Curvature radiusCornealrefractive
power

double○double△Cylindrical power
Cylindricalrefractive power-○<Average><Radius unit="mm"><Power unit="D"><Cylinder><Power unit="D"><Axis unit="deg"><L><List No=" "><R1><Radius unit="mm">double○Cylinder axis angle
Left eye measurement
value
Measurement value(1 to n times)

int

○

-

○

Flat

-

○

double○double○int○<Power unit="D">Curvature radiusCorneal refractive
power<Axis unit="deg">*2

Axis angle<Radius unit="mm"><Power unit="D">Curvature radiusCorneal refractive
power<Axis unit="deg">*2

Axis angle
Average of R1 and R2

<R2>

Steep

<Average><Radius unit="mm"><Power unit="D"><Cylinder><Power unit="D">Curvature radiusCorneal refractive
value
Cylinder power
Cylindrical

-

-

○

double○double

○

int

○

-

○

double○double△-○double○Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

9／31

refractive power<Axis unit="deg"><List No=" "><Error>*3<Median><R1><Radius unit="mm">

Cylinder axis angle
Measurement value(In case of error )int○

Error Message

string(64 characters)△

Median values

-

Flat

-

○

double○<Power unit="D">Curvature radiusCorneal refractive
power

double○<Axis unit="deg">*2

Axis angle

int

○

Steep

-

○

double○double○<R2><Radius unit="mm"><Power unit="D">Curvature radiusCorneal refractive
power<Axis unit="deg">*2

Axis angle

int○

Average of R1 and R2-○

Curvature radiusCorneal refractive
power

double○double△-○<Average><Radius unit="mm"><Power unit="D"><Cylinder><Power unit="D">

Cylinder power
Cylindricalrefractive power

double○<Axis unit="deg">

Cylinder axis angle

int○

Pupil size data
Right eye measurement
value

-

Pupil size
Left eye measurement
value

doublePupil size

double<Pupil><R><PupilSize
Lamp="on","off">

unit="mm"<PupilSize
Lamp="on","off">

unit="mm"<L><Corneal><R><CornealSize unit="mm">*4<L><CornealSize unit="mm">
*4

-

-

Corneal size data
Right eye measurement
value

-

Corneal size
Left eye measurement
value

doubleCorneal size

doubleUnique tag of each company can be optionally defined here.

*1: For data element <CylinderMode>, the terms have the following meaning:

△

△

△

△

any

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

10／31“+” = the cylinder value is always positive“-” = the cylinder value is always negative*2: The term “Axis” is used in colloquial clinical parlance. However, the proper optical term for this usage is“meridian” and represents the corneal meridian measured during Keratometry.
*3: <Error> data element is vendor specific strings and JOIA does not define “coded” values for these data
elements.
*4: The data element <CornealSize unit="mm"> is populated by devices in a horizontal measurement direction,
however, it is not specified how the measurement is accomplished.

The most common use case is “from white

to white”, but others may exist.

■Tonometer data
Tag name

Data type

NecessityMeasure type tag-○

Right eye measurement value-

Description<Measure type="TM">

<TM>
<R>
<List No=" ">Measurement value (0 to n times)<IOP_mmHg unit="mmHg"> Intraocular pressure (mmHg)

double○doublestring(64 characters)

△

-

○

<IOP_mmHg unit="mmHg"> Intraocular pressure(mmHg)

double○<IOP_Pa unit="kPa">*2

double△string(64 characters)△<IOP_Pa unit="kPa">*2

Intraocular pressure (kPa) or (hPa)<ConfidenceIndex>*1

Confidence index<Average>

Average value

Intraocular pressure (kPa) or (hPa)△↓In case of error<Error>*1<L><List No=" ">Error Message
Left eye measurement value-

Measurement value (0 to n times)-<IOP_mmHg unit="mmHg"> Intraocular pressure (mmHg)<IOP_Pa unit="kPa">*2

Intraocular pressure (kPa) or (hPa)<ConfidenceIndex>*1

Confidence index<Average>

Average value

double○doublestring(64 characters)

△
△

-

○

<IOP_mmHg unit="mmHg"> Intraocular pressure (mmHg)

double○<IOP_Pa unit="kPa">*2

double△string(64 characters)△

Intraocular pressure (kPa) or (hPa)↓In case of error<Error>*1

Error Message<CorrectedIOP><Formula1 No=" "><R>

Correction Formula1 (0 to n) *3-

Right eye data-

<Param1 unit="mm">Parameter 1(mm)double△<Param2>

Parameter2

double△Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<CCT unit="mm">Central corneal thickness<Measured>Measurement value

11／31

double△-<IOP_mmHg unit="mmHg"> Intraocular pressure (mmHg)

double○<IOP_Pa unit="kPa">*2

double△-○double○double△Intraocular pressure (kPa) or (hPa)<Corrected>

Corrected value
Corrected intraocular pressure<IOP_mmHg unit="mmHg"> (mmHg)
Corrected intraocular pressure<IOP_Pa unit="kPa">*2(kPa) or (hPa)<L>

Left eye data-

<Param1 unit="mm">Parameter 1(mm)double△<Param2>

Parameter2

double△<CCT unit="mm">Central corneal thickness

double△<Measured>Measurement value-<IOP_mmHg unit="mmHg"> Intraocular pressure (mmHg)

double○<IOP_Pa unit="kPa">*2

double△-○double○double△Intraocular pressure (kPa) or (hPa)<Corrected>

Corrected value
Corrected intraocular pressure<IOP_mmHg unit="mmHg"> (mmHg)
Corrected intraocular pressure<IOP_Pa unit="kPa">*2( kPa) or (hPa)Unique tag of each company can be optionally defined here.

any*1: <ConfidenceIndex> and <Error> data elements are vendor specific strings and JOIA does not define codedvalues for these data elements.
*2: Both "kPa" and "hPa" can be used for the unit of intraocular pressure (attribute of ＜IOP_Pa＞tag) listed in
Pascal.
*3: Correction Formula 1 (correction formula of corrected intraocular pressure)
Following correction formula is used for calculation of the corrected intraocular pressure:
Corrected IOP value = Measurement intraocular pressure value + Intraocular pressure correction value
IOP correction value =(Param1×1000 - CCT×1000) × Param2
Correction factor: Param1: Reference central corneal thickness (mm)CCT

： Central corneal thickness (mm)Param2 ：Parameter 2

correction amount adjustment factor (0.0001 to 1.0000)■Lensmeter data
Tag name<Measure type="LM">

Description
Measurement type tag
Measurement mode

<MeasureMode>*1( Normal, Progressive, Contact )<DiopterStep unit="D">

Diopter increment<AxisStep unit="deg">

Cylinder axis angle increment<CylinderMode>*2

Cylinder mode(-,+,mix)

Data type

Necessitystring(64 characters)○

double△int△

string(-,+,mix)△△

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<PrismDiopterStep
unit="pri"><PrismBaseStep
unit="deg">

Prism diopter increment
Prism base increment<PrismMode>*3

Prism mode

<AddMode>*4

Additional power mode

<LensLabel>

Lens Label

12／31

double△int△

string(pb,xy)△

string(add,nsph)
String(64 characters)

△

<LM>
<S>*5

Single measurement value-<Sphere unit="D">

Spherical power (D)double○<Cylinder unit="D">

Cylindrical power (D)double○<Axis unit="deg">

Cylinder axis (° )int○

<SE unit="D">

Spherical equivalent (D)double△<ADD unit="D">

Additional power(D)double△<ADD2 unit="D">Second additional power (D)double△<NearSphere unit="D">double△<NearSphere2 unit="D">Near spherical power (D)Spherical power for second
near portion (D)double△<Prism unit="pri">

Prism diopter (Δ)

double△<PrismBase unit="deg">

Prism base angle (° )int△double△double△int
string(64 characters)△

Horizontal prism diopter (Δ)<PrismX unit="pri" base="out">(base=in, out）
Vertical prism diopter (Δ)<PrismY unit="pri" base="up">(base=up, down)<UVTransmittance unit="%">

UV transmittance (%)<ConfidenceIndex>*6

Confidence index
Single measurement value( In case of error)<Error>*6

Error Message

<S>*5

<R>

Right eye measurement value

string(64 characters)

△

△

-

<Sphere unit="D">

Spherical power (D)double○<Cylinder unit="D">

Cylindrical power (D)double○<Axis unit="deg">

Cylinder axis (° )int○

<SE unit="D">

Spherical equivalent (D)double△<ADD unit="D">

Additional power (D)double△<ADD2 unit="D">Second additional power (D)double△<NearSphere unit="D">double△<NearSphere2 unit="D">Near spherical power (D)Spherical power for second
near portion (D)double△<Prism unit="pri">

Prism diopter (Δ)

double△<PrismBase unit="deg">

Prism base angle (° )int△double△double△int
string(64 characters)△

Horizontal prism diopter (Δ)<PrismX unit="pri" base="out">(base=in, out）
Vertical prism diopter (Δ)<PrismY unit="pri" base="up">(base=up, down)<R><UVTransmittance unit="%">

UV transmittance (%)<ConfidenceIndex>*6

Confidence Index
Right eye measurement value△

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

13／31(in case of error)<Error>*6<L>

Error Message
Left eye measurement value

string(64 characters)△-<Sphere unit="D">

Spherical power (D)double○<Cylinder unit="D">

Cylindrical power (D)double○<Axis unit="deg">

Cylinder axis (° )int○

<SE unit="D">

Spherical equivalent (D)double△<ADD unit="D">

Additional power (D)double△<ADD2 unit="D">Second additional power (D)double△<NearSphere unit="D">double△<NearSphere2 unit="D">Near spherical power (D)Spherical power for second
near portion (D)double△<Prism unit="pri">

Prism diopter (Δ)

double△int△

double△double△int
string(64 characters)△<PrismBase unit="deg">

Prism base angle (°)
Horizontal prism diopter (Δ)<PrismX unit="pri" base="out">(base=in, out）
Vertical prism diopter (Δ)<PrismY unit="pri" base="up">(base=up, down)<UVTransmittance unit="%">

UV transmittance (%)<ConfidenceIndex>*6

Confidence index
Left eye measurement value(in case of error)<Error>*6

Error Message

<L>

<PD>

Pupillary distance<Distance unit="mm"><DistanceR unit="mm">< DistanceL unit="mm">< Near unit="mm">< NearR unit="mm">< NearL unit="mm">double△double△double△Near pupillary distance
Right eye near pupillary
distance
Left eye near pupillary
distance

double△double△double△*1: For data element < MeasureMode >, the terms have the following meaning:“Normal” = Monofocal and Non-Progressive multi-focal“Progressive” = Progressive“Contact” = Contact Lens”*2: For data element <CylinderMode>, the terms have the following meaning:“-” = the cylinder value is always negative△

Far pupillary distance
Right eye far pupillary
distance
Left eye far pupillary
distance

Unique tag of each company can be optionally defined here.

“+” = the cylinder value is always positive

string(64 characters)△anyCommon Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)"mix" = the cylindrical power by positive reading when the refractive power is positive for any meridional
angle and the cylindrical power by negative reading in other cases*3: For data element < PrismMode >, the terms have the following meaning:“pb” = polar (uses data elements <Prism unit="pri"> and <PrismBase unit="deg">)“xy” = Cartesian (uses data elements <PrismX unit="pri" base="out"> and <PrismY unit="pri" base="up">)*4: For data element < AddMode >, the terms have the following meaning:“add” = add (uses data elements <ADD unit="D"> and <ADD2 unit="D">“nsph” = near sphere (uses data elements <NearSphere unit="D"> and <NearSphere2 unit="D">)*5: Data element <S> is used for a single measurement of a lens (i.e., not specific to the left or right eye)
*6: <ConfidenceIndex> and <Error> data elements are vendor specific strings and JOIA does not define codedvalues for these data elements.

4. Namespaces
4-1. Definition of namespace
XML namespace is defined for every data classification.
The prefix added to each data classification is as below.
Data classification
Common data
Refractometer data
Keratometer data
Tonometer data
Lensmeter data
・・

XML Namespace prefix
nsCommon
nsREF
nsKM
nsTM
nsLM
・
・

14／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

5. Schema definition file5-1. Schema file
The tag definition of each data type is described in a schema file. The schema file isprepared for each data type.
Data classification
Common data
Refractometer data
Keratometer data
Tonometer data
Lensmeter data
・・

XML Namespace prefix
Common_schema.xsd
REF_schema.xsd
KM_schema.xsd
TM_schema.xsd
LM_schema.xsd
・
・

15／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)○ XML file sample of refractometer data================================================================<?xml version="1.0" encoding="UTF-8"?><?xml-stylesheet type="text/xsl" href="REF_Stylesheet.xsl" ?><Ophthalmology xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance"
xmlns:nsCommon="http://www.joia.or.jp/standardized/namespaces/Common"
xmlns:nsREF="http://www.joia.or.jp/standardized/namespaces/REF"
xsi:schemaLocation="http://www.joia.or.jp/standardized/namespaces/Common Common_schema.xsd
http://www.joia.or.jp/standardized/namespaces/REF REF_schema.xsd"><nsCommon:Common><nsCommon:Company>ABCD</nsCommon:Company><nsCommon:ModelName>ABC-123</nsCommon:ModelName><nsCommon:MachineNo>1234</nsCommon:MachineNo><nsCommon:ROMVersion>1.0</nsCommon:ROMVersion><nsCommon:Version>1.4</nsCommon:Version><nsCommon:Date>2008-09-01</nsCommon:Date><nsCommon:Time>19:36:28</nsCommon:Time><nsCommon:Patient><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID><nsCommon:FirstName>TARO</nsCommon:FirstName><nsCommon:MiddleName></nsCommon:MiddleName><nsCommon:LastName>TANAKA</nsCommon:LastName><nsCommon:Sex>M</nsCommon:Sex><nsCommon:Age>18</nsCommon:Age><nsCommon:DOB>1975-08-16</nsCommon:DOB><nsCommon:NameJ1>田中 太郎</nsCommon:NameJ1><nsCommon:NameJ2>たなか たろう</nsCommon:NameJ2></nsCommon:Patient><nsCommon:Operator><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID></nsCommon:Operator></nsCommon:Common><nsREF:Measure type="REF"><nsREF:VD unit="mm">12.00</nsREF:VD><nsREF:DiopterStep unit="D">0.25</nsREF:DiopterStep><nsREF:AxisStep unit="deg">5</nsREF:AxisStep><nsREF:CylinderMode>-</nsREF:CylinderMode><nsREF:REF><nsREF:R><nsREF:List No="1">

16／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsREF:Sphere unit="D">-2.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.25</nsREF:Cylinder><nsREF:Axis unit="deg">180</nsREF:Axis><nsREF:SE unit="D"></nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>9</nsREF:ConfidenceIndex></nsREF:List><nsREF:List No="2"><nsREF:Error></nsREF:Error></nsREF:List><nsREF:List No="3"><nsREF:Sphere unit="D">-0.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.50</nsREF:Cylinder><nsREF:Axis unit="deg">90</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>5</nsREF:ConfidenceIndex></nsREF:List></nsREF:R><nsREF:L><nsREF:List No="1"><nsREF:Sphere unit="D">-2.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.50</nsREF:Cylinder><nsREF:Axis unit="deg">180</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>9</nsREF:ConfidenceIndex></nsREF:List><nsREF:List No="2"><nsREF:Sphere unit="D">-2.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.50</nsREF:Cylinder><nsREF:Axis unit="deg">180</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>9</nsREF:ConfidenceIndex></nsREF:List><nsREF:List No="3"><nsREF:Sphere unit="D">-2.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.50</nsREF:Cylinder><nsREF:Axis unit="deg">180</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>9</nsREF:ConfidenceIndex></nsREF:List><nsREF:List No="4"><nsREF:Sphere unit="D">-2.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.50</nsREF:Cylinder><nsREF:Axis unit="deg">180</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>E</nsREF:ConfidenceIndex>

17／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)</nsREF:List><nsREF:List No="5"><nsREF:Sphere unit="D">-0.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.50</nsREF:Cylinder><nsREF:Axis unit="deg">90</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>5</nsREF:ConfidenceIndex></nsREF:List><nsREF:List No="6"><nsREF:Sphere unit="D">+0.50</nsREF:Sphere><nsREF:Cylinder unit="D">1.50</nsREF:Cylinder><nsREF:Axis unit="deg">180</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE><nsREF:CataractMode>on</nsREF:CataractMode><nsREF:IOLMode>on</nsREF:IOLMode><nsREF:ConfidenceIndex>6</nsREF:ConfidenceIndex></nsREF:List><nsREF:Median><nsREF:Sphere unit="D">-2.50</nsREF:Sphere><nsREF:Cylinder unit="D">0.50</nsREF:Cylinder><nsREF:Axis unit="deg">180</nsREF:Axis><nsREF:SE unit="D">-2.75</nsREF:SE></nsREF:Median></nsREF:L></nsREF:REF><nsREF:PD><nsREF:WorkingDistance unit="cm">40</nsREF:WorkingDistance><nsREF:Distance unit="mm">62.5</nsREF:Distance><nsREF:Near unit="mm">60.5</nsREF:Near></nsREF:PD><(Unique tag)>・・・</(Unique tag)></nsREF:Measure></Ophthalmology>

Unique tag of each company is specified
here. Prefix is not necessary because it isout of the namespace definition.

18／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)○ XML file sample of keratometer data==================================================================<?xml version="1.0" encoding="UTF-8"?><?xml-stylesheet type="text/xsl" href="KM_Stylesheet.xsl" ?><Ophthalmology xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance"
xmlns:nsCommon="http://www.joia.or.jp/standardized/namespaces/Common"
xmlns:nsREF="http://www.joia.or.jp/standardized/namespaces/KM"
xsi:schemaLocation="http://www.joia.or.jp/standardized/namespaces/Common Common_schema.xsd
http://www.joia.or.jp/standardized/namespaces/KM KM_schema.xsd"><nsCommon:Common><nsCommon:Company>ABCD</nsCommon:Company><nsCommon:ModelName>ABC-123</nsCommon:ModelName><nsCommon:MachineNo>1234</nsCommon:MachineNo><nsCommon:ROMVersion>1.0</nsCommon:ROMVersion><nsCommon:Version>1.4</nsCommon:Version><nsCommon:Date>2008-09-01</nsCommon:Date><nsCommon:Time>19:36:28</nsCommon:Time><nsCommon:Patient><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID><nsCommon:FirstName>TARO</nsCommon:FirstName><nsCommon:MiddleName></nsCommon:MiddleName><nsCommon:LastName>TANAKA</nsCommon:LastName><nsCommon:Sex>M</nsCommon:Sex><nsCommon:Age>18</nsCommon:Age><nsCommon:DOB>1975-08-16</nsCommon:DOB><nsCommon:NameJ1>田中 太郎</nsCommon:NameJ1><nsCommon:NameJ2>たなか たろう</nsCommon:NameJ2></nsCommon:Patient><nsCommon:Operator><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID></nsCommon:Operator></nsCommon:Common><nsKM:Measure type="KM"><nsKM:DiopterStep unit="D">0.25</nsKM:DiopterStep><nsKM:AxisStep unit="deg">5</nsKM:AxisStep><nsKM:CylinderMode>-</nsKM:CylinderMode><nsKM:RefractiveIndex>1.3375</nsKM:RefractiveIndex><nsKM:KM><nsKM:R><nsKM:List No="1"><nsKM:R1><nsKM:Radius unit="mm">8.05</nsKM:Radius>

19／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsKM:Power unit="D">45.00</nsKM:Power><nsKM:Axis unit="deg">180</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">44.95</nsKM:Power><nsKM:Axis unit="deg">68</nsKM:Axis></nsKM:R2><nsKM:Average><nsKM:Radius unit="mm">7.85</nsKM:Radius><nsKM:Power unit="D">43.35</nsKM:Power></nsKM:Average><nsKM:Cylinder><nsKM:Power unit="D">-2.50</nsKM:Power><nsKM:Axis unit="deg">150</nsKM:Axis></nsKM:Cylinder></nsKM:List><nsKM:List No="2"><nsKM:Error></nsKM:Error></nsKM:List><nsKM:List No="3"><nsKM:R1><nsKM:Radius unit="mm">8.05</nsKM:Radius><nsKM:Power unit="D">45.00</nsKM:Power><nsKM:Axis unit="deg">180</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">44.95</nsKM:Power><nsKM:Axis unit="deg">68</nsKM:Axis></nsKM:R2><nsKM:Average><nsKM:Radius unit="mm">7.85</nsKM:Radius><nsKM:Power unit="D">43.35</nsKM:Power></nsKM:Average><nsKM:Cylinder><nsKM:Power unit="D">-2.50</nsKM:Power><nsKM:Axis unit="deg">150</nsKM:Axis></nsKM:Cylinder></nsKM:List></nsKM:R><nsKM:L><nsKM:List No="1"><nsKM:R1><nsKM:Radius unit="mm">8.04</nsKM:Radius><nsKM:Power unit="D">42.00</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">7.54</nsKM:Radius><nsKM:Power unit="D">44.75</nsKM:Power><nsKM:Axis unit="deg">43</nsKM:Axis></nsKM:R2><nsKM:Average><nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">43.25</nsKM:Power></nsKM:Average>

20／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsKM:Cylinder><nsKM:Power unit="D">-2.75</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:Cylinder></nsKM:List><nsKM:List No="2"><nsKM:R1><nsKM:Radius unit="mm">8.04</nsKM:Radius><nsKM:Power unit="D">42.00</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">7.54</nsKM:Radius><nsKM:Power unit="D">44.75</nsKM:Power><nsKM:Axis unit="deg">43</nsKM:Axis></nsKM:R2><nsKM:Average><nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">43.25</nsKM:Power></nsKM:Average><nsKM:Cylinder><nsKM:Power unit="D">-2.75</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:Cylinder></nsKM:List><nsKM:List No="3"><nsKM:R1><nsKM:Radius unit="mm">7.85</nsKM:Radius><nsKM:Power unit="D">43.75</nsKM:Power><nsKM:Axis unit="deg">180</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">8.25</nsKM:Radius><nsKM:Power unit="D">44.80</nsKM:Power><nsKM:Axis unit="deg">90</nsKM:Axis></nsKM:R2><nsKM:Average><nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">43.50</nsKM:Power></nsKM:Average><nsKM:Cylinder><nsKM:Power unit="D">+2.25</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:Cylinder></nsKM:List><nsKM:List No="4"><nsKM:R1><nsKM:Radius unit="mm">8.50</nsKM:Radius><nsKM:Power unit="D">42.00</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">8.54</nsKM:Radius><nsKM:Power unit="D">42.35</nsKM:Power><nsKM:Axis unit="deg">180</nsKM:Axis></nsKM:R2><nsKM:Average>

21／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">43.25</nsKM:Power></nsKM:Average><nsKM:Cylinder><nsKM:Power unit="D">-0.25</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:Cylinder></nsKM:List><nsKM:List No="5"><nsKM:R1><nsKM:Radius unit="mm">8.04</nsKM:Radius><nsKM:Power unit="D">42.00</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">7.54</nsKM:Radius><nsKM:Power unit="D">44.75</nsKM:Power><nsKM:Axis unit="deg">43</nsKM:Axis></nsKM:R2><nsKM:Average><nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">43.25</nsKM:Power></nsKM:Average><nsKM:Cylinder><nsKM:Power unit="D">-2.75</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:Cylinder></nsKM:List><nsKM:Median><nsKM:R1><nsKM:Radius unit="mm">7.25</nsKM:Radius><nsKM:Power unit="D">42.50</nsKM:Power><nsKM:Axis unit="deg">180</nsKM:Axis></nsKM:R1><nsKM:R2><nsKM:Radius unit="mm">7.54</nsKM:Radius><nsKM:Power unit="D">44.75</nsKM:Power><nsKM:Axis unit="deg">43</nsKM:Axis></nsKM:R2><nsKM:Average><nsKM:Radius unit="mm">7.65</nsKM:Radius><nsKM:Power unit="D">43.25</nsKM:Power></nsKM:Average><nsKM:Cylinder><nsKM:Power unit="D">-2.75</nsKM:Power><nsKM:Axis unit="deg">133</nsKM:Axis></nsKM:Cylinder></nsKM:Median></nsKM:L></nsKM:KM><nsKM:Pupil><nsKM:R><nsKM:PupilSize unit="mm" Lamp="on">3.5</nsKM:PupilSize></nsKM:R><nsKM:L><nsKM:PupilSize unit="mm" Lamp="on">3.0</nsKM:PupilSize>

22／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)</nsKM:L></nsKM:Pupil><nsKM:Corneal><nsKM:R><nsKM:CornealSize unit="mm">2.5</nsKM:CornealSize></nsKM:R><nsKM:L><nsKM:CornealSize unit="mm">2.0</nsKM:CornealSize></nsKM:L></nsKM:Corneal><(Unique tag)>・・・</(Unique tag)></nsKM:Measure></Ophthalmology>

Unique tag of each company is specified
here. Prefix is not necessary because it isout of the namespace definition.

23／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)○ XML file sample of tonometer data==================================================================<?xml version="1.0" encoding="UTF-8"?><?xml-stylesheet type="text/xsl" href="TM_Stylesheet.xsl" ?><Ophthalmology xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance"
xmlns:nsCommon="http://www.joia.or.jp/standardized/namespaces/Common"
xmlns:nsTM="http://www.joia.or.jp/standardized/namespaces/TM"
xsi:schemaLocation="http://www.joia.or.jp/standardized/namespaces/Common Common_schema.xsd
http://www.joia.or.jp/standardized/namespaces/TM TM_schema.xsd "><nsCommon:Common><nsCommon:Company>ABCD</nsCommon:Company><nsCommon:ModelName>ABC-123</nsCommon:ModelName><nsCommon:MachineNo>1234</nsCommon:MachineNo><nsCommon:ROMVersion>1.0</nsCommon:ROMVersion><nsCommon:Version>1.4</nsCommon:Version><nsCommon:Date>2008-09-01</nsCommon:Date><nsCommon:Time>19:36:28</nsCommon:Time><nsCommon:Patient><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID><nsCommon:FirstName>TARO</nsCommon:FirstName><nsCommon:MiddleName></nsCommon:MiddleName><nsCommon:LastName>TANAKA</nsCommon:LastName><nsCommon:Sex>M</nsCommon:Sex><nsCommon:Age>18</nsCommon:Age><nsCommon:DOB>1975-08-16</nsCommon:DOB><nsCommon:NameJ1>田中 太郎</nsCommon:NameJ1><nsCommon:NameJ2>たなか たろう</nsCommon:NameJ2></nsCommon:Patient><nsCommon:Operator><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID></nsCommon:Operator></nsCommon:Common><nsTM:Measure type="TM"><nsTM:TM><nsTM:R><nsTM:List No="1"><nsTM:IOP_mmHg unit="mmHg">24.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">3.2</nsTM:IOP_Pa><nsTM:ConfidenceIndex></nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="2"><nsTM:IOP_mmHg unit="mmHg">21.0</nsTM:IOP_mmHg>

24／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsTM:IOP_Pa unit="kPa">2.8</nsTM:IOP_Pa><nsTM:ConfidenceIndex>*</nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="3"><nsTM:IOP_mmHg unit="mmHg">22.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.9</nsTM:IOP_Pa><nsTM:ConfidenceIndex></nsTM:ConfidenceIndex></nsTM:List><nsTM:Average><nsTM:IOP_mmHg unit="mmHg">22.3</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.97</nsTM:IOP_Pa></nsTM:Average></nsTM:R><nsTM:L><nsTM:List No="1"><nsTM:IOP_mmHg unit="mmHg">24.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">3.2</nsTM:IOP_Pa><nsTM:ConfidenceIndex>*</nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="2"><nsTM:IOP_mmHg unit="mmHg">21.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.8</nsTM:IOP_Pa><nsTM:ConfidenceIndex></nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="3"><nsTM:IOP_mmHg unit="mmHg">22.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.9</nsTM:IOP_Pa><nsTM:ConfidenceIndex></nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="4"><nsTM:IOP_mmHg unit="mmHg">24.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">3.2</nsTM:IOP_Pa><nsTM:ConfidenceIndex></nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="5"><nsTM:IOP_mmHg unit="mmHg">21.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.8</nsTM:IOP_Pa><nsTM:ConfidenceIndex>*</nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="6"><nsTM:IOP_mmHg unit="mmHg">22.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.9</nsTM:IOP_Pa><nsTM:ConfidenceIndex>*</nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="7"><nsTM:IOP_mmHg unit="mmHg">24.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">3.2</nsTM:IOP_Pa><nsTM:ConfidenceIndex></nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="8"><nsTM:IOP_mmHg unit="mmHg">21.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.8</nsTM:IOP_Pa><nsTM:ConfidenceIndex>*</nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="9"><nsTM:IOP_mmHg unit="mmHg">22.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.9</nsTM:IOP_Pa>

25／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsTM:ConfidenceIndex>*</nsTM:ConfidenceIndex></nsTM:List><nsTM:List No="10"><nsTM:IOP_mmHg unit="mmHg">24.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">3.2</nsTM:IOP_Pa><nsTM:ConfidenceIndex>*</nsTM:ConfidenceIndex></nsTM:List><nsTM:Average><nsTM:IOP_mmHg unit="mmHg">22.5</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.99</nsTM:IOP_Pa></nsTM:Average></nsTM:L></nsTM:TM><nsTM:CorrectedIOP><nsTM:Formula1 No="1"><nsTM:R><nsTM:Param1 unit="mm">0.554</nsTM:Param1><nsTM:Param2>0.0450</nsTM:Param2><nsTM:CCT unit="mm">0.588</nsTM:CCT><nsTM:Measured><nsTM:IOP_mmHg unit="mmHg">22.3</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.97</nsTM:IOP_Pa></nsTM:Measured><nsTM:Corrected><nsTM:IOP_mmHg unit="mmHg">20.8</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.77</nsTM:IOP_Pa></nsTM:Corrected></nsTM:R><nsTM:L><nsTM:Param1 unit="mm">0.554</nsTM:Param1><nsTM:Param2>0.0450</nsTM:Param2><nsTM:CCT unit="mm">0.588</nsTM:CCT><nsTM:Measured><nsTM:IOP_mmHg unit="mmHg">22.5</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.99</nsTM:IOP_Pa></nsTM:Measured><nsTM:Corrected><nsTM:IOP_mmHg unit="mmHg">21.0</nsTM:IOP_mmHg><nsTM:IOP_Pa unit="kPa">2.76</nsTM:IOP_Pa></nsTM:Corrected></nsTM:L></nsTM:Formula1></nsTM:CorrectedIOP><(Unique tag)>・・・</(Unique tag)></nsTM:Measure></Ophthalmology>

Unique tag of each company is specified
here. Prefix is not necessary because it isout of the namespace definition.

26／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)○ XML file sample of lensmeter data==================================================================<?xml version="1.0" encoding="UTF-16"?><?xml-stylesheet type="text/xsl" href="LM_Stylesheet.xsl" ?><Ophthalmology xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance"
xmlns:nsCommon="http://www.joia.or.jp/standardized/namespaces/Common"
xmlns:nsLM="http://www.joia.or.jp/standardized/namespaces/LM"
xsi:schemaLocation="http://www.joia.or.jp/standardized/namespaces/Common Common_schema.xsd
http://www.joia.or.jp/standardized/namespaces/LM LM_schema.xsd"><nsCommon:Common><nsCommon:Company>ABCD</nsCommon:Company><nsCommon:ModelName>ABC-123</nsCommon:ModelName><nsCommon:MachineNo>1234</nsCommon:MachineNo><nsCommon:ROMVersion>1.0</nsCommon:ROMVersion><nsCommon:Version>1.4</nsCommon:Version><nsCommon:Date>2008-09-01</nsCommon:Date><nsCommon:Time>19:36:28</nsCommon:Time><nsCommon:Patient><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID><nsCommon:FirstName>TARO</nsCommon:FirstName><nsCommon:MiddleName></nsCommon:MiddleName><nsCommon:LastName>TANAKA</nsCommon:LastName><nsCommon:Sex>M</nsCommon:Sex><nsCommon:Age>18</nsCommon:Age><nsCommon:DOB>1975-08-16</nsCommon:DOB><nsCommon:NameJ1>田中 太郎</nsCommon:NameJ1><nsCommon:NameJ2>たなか たろう</nsCommon:NameJ2></nsCommon:Patient><nsCommon:Operator><nsCommon:No.>123456</nsCommon:No.><nsCommon:ID>ABCDEFG</nsCommon:ID></nsCommon:Operator></nsCommon:Common><nsLM:Measure type="LM"><nsLM:MeasureMode>Progressive</nsLM:MeasureMode><nsLM:DiopterStep unit="D">0.25</nsLM:DiopterStep><nsLM:AxisStep unit="deg">1</nsLM:AxisStep><nsLM:CylinderMode>-</nsLM:CylinderMode><nsLM:PrismDiopterStep unit="pri">0.25</nsLM:PrismDiopterStep><nsLM:PrismBaseStep unit="deg">1</nsLM:PrismBaseStep>

27／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsLM:PrismMode>xy</nsLM:PrismMode><nsLM:AddMode>add</nsLM:AddMode><nsLM:LM><nsLM:S><nsLM:Sphere unit="D">-3.00</nsLM:Sphere><nsLM:Cylinder unit="D">-11.00</nsLM:Cylinder><nsLM:Axis unit="deg">180</nsLM:Axis><nsLM:SE unit="D">-8.50</nsLM:SE><nsLM:ADD unit="D">1.50</nsLM:ADD><nsLM:ADD2 unit="D">2.00</nsLM:ADD2><nsLM:NearSphere unit="D">-1.50</nsLM:NearSphere><nsLM:NearSphere2 unit="D">-1.00</nsLM:NearSphere2><nsLM:Prism unit="pri">0.25</nsLM:Prism><nsLM:PrismBase unit="deg">102</nsLM:PrismBase><nsLM:PrismX unit="pri" base="in">0.00</nsLM:PrismX><nsLM:PrismY unit="pri" base="up">0.25</nsLM:PrismY><nsLM:UVTransmittance unit="%">5</nsLM:UVTransmittance><nsLM:ConfidenceIndex>CYL OVERFLOW</nsLM:ConfidenceIndex></nsLM:S><nsLM:R><nsLM:Sphere unit="D">-3.00</nsLM:Sphere><nsLM:Cylinder unit="D">0.00</nsLM:Cylinder><nsLM:Axis unit="deg">0</nsLM:Axis><nsLM:SE unit="D">-3.00</nsLM:SE><nsLM:ADD unit="D">1.50</nsLM:ADD><nsLM:ADD2 unit="D">2.00</nsLM:ADD2><nsLM:NearSphere unit="D">-1.50</nsLM:NearSphere><nsLM:NearSphere2 unit="D">-1.00</nsLM:NearSphere2><nsLM:Prism unit="pri">0.25</nsLM:Prism><nsLM:PrismBase unit="deg">102</nsLM:PrismBase><nsLM:PrismX unit="pri" base="in">0.00</nsLM:PrismX><nsLM:PrismY unit="pri" base="up">0.25</nsLM:PrismY><nsLM:UVTransmittance unit="%">5</nsLM:UVTransmittance><nsLM:ConfidenceIndex></nsLM:ConfidenceIndex></nsLM:R><nsLM:L><nsLM:Sphere unit="D">-2.00</nsLM:Sphere><nsLM:Cylinder unit="D">-1.00</nsLM:Cylinder><nsLM:Axis unit="deg">176</nsLM:Axis><nsLM:SE unit="D">-2.50</nsLM:SE><nsLM:ADD unit="D">1.75</nsLM:ADD><nsLM:ADD2 unit="D">2.00</nsLM:ADD2><nsLM:NearSphere unit="D">-0.25</nsLM:NearSphere><nsLM:NearSphere2 unit="D">0.00</nsLM:NearSphere2><nsLM:Prism unit="pri">2.50</nsLM:Prism><nsLM:PrismBase unit="deg">90</nsLM:PrismBase><nsLM:PrismX unit="pri" base="out">0.00</nsLM:PrismX><nsLM:PrismY unit="pri" base="up">2.50</nsLM:PrismY><nsLM:UVTransmittance unit="%">5</nsLM:UVTransmittance><nsLM:ConfidenceIndex></nsLM:ConfidenceIndex></nsLM:L></nsLM:LM><nsLM:PD>

28／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)<nsLM:Distance unit="mm">58.5</nsLM:Distance><nsLM:DistanceR unit="mm">29.5</nsLM:DistanceR><nsLM:DistanceL unit="mm">29.0</nsLM:DistanceL><nsLM:Near unit="mm"></nsLM:Near><nsLM:NearR unit="mm"></nsLM:NearR><nsLM:NearL unit="mm"></nsLM:NearL></nsLM:PD><(Unique tag)>・・・</(Unique tag)></nsLM:Measure></Ophthalmology>

Unique tag of each company is specified
here. Prefix is not necessary because it isout of the namespace definition.

29／31

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

30／31

Revision history
Date
VersionContent / Reason
October 21,
JOIA・STD 001-1.0-2009 Establishment
2008
Revision/・Addition of ᇞ to the explanation in the necessity field of
the tag definition.
・Addition of <REF> tag to the refractometer data of the tag
definition.
・Addition of <KM> tag to the keratometer data of the tag
definition.
・Addition of <RefractiveIndex> and <Corneal> tags to theAugust 28,
JOIA・STD 001-1.1-2009
2009
keratometer data of the tag definition.
・Change of definition position of optional tags.
・Addition of explanation paragraph of the namespace.
・ Change of XML file samples to the samples usingnamespaces.
・Change of XML file samples to the samples using optionaltags.
Revision/・Addition of description of tonometer data.
Revision/・ Deletion of the notice of patient’s name in otherlanguages.
・ Addition of character code specification.
December 7,
JOIA・STD 001-1.3-2010 ・ Correction of mistake of name tag.
2010
・ Corrected to no patient number is required.
・ Clarification of method of describing sex
・ Correction of sample according to the above
March
2010

12,

JOIA・STD 001-1.2-2010

Revision/
October 17,
JOIA・STD 001-1.4-2012 ・ Addition of UTF-16 to the character code specification.
2012
・ Addition of the lens meter tag definition.
Revision/
・Change in 1-1. Handling of files description
・ Addition of <PatientIDSource> data element and its
description to Common data.
・Addition of the following items to REF data.
・< CycloplegicMode > data element.
July 10,
・<CylinderMode> data element description.
JOIA Std.001-1.5-2016
2016
・<ConfidenceIndex> and <Error> data elements description.
・<Median> data element description.
・Change in <PD> data element description in REF data.
・Addition of the following items to KM data.
・<CylinderMode> data element description.
・Flat,Steep Description in <R1>,<R2>.

Common Specification for output data of Ophthalmic Examination Equipment 01(JOIA Std.)

31／31

・<Axis unit="deg"> data element description.
・<Error> data elements description.
・Lamp off statement in <CornealSize> data element.
・<CornealSize unit="mm"> data element description.
・Change in <Average> data element description in KM
data.
・Deletion of “mix” at <CylinderMode> in KM data.
・Addition of the following items to NT data.
・<ConfidenceIndex> and <Error> data elements description.
・<Corrected IOP> data element description.
・Change in <PrismDiopterStep> data element in LMdata.
・Addition of the following items to LM data.
・<LensLabel> data element.
・<MeasurementMode> data element description.
・<CylinderMode> data element description.
・<PrismMode> data element description.
・<ADDMode> data element description.
・<S> data element description.
・<ConfidenceIndex> and <Error> data elements description
・ Addition of xml-stylesheet type in XML file sample ofrefractometer data, keratometer data and lensmeter data