PULSE SCALERSAF SERIESSAFDINSTRUCTION MANUALPreface
Thank you for purchasing the Pulse Scaler SAFD.
This manual contains instructions for the mounting, functions, operations and notes when
operating the SAFD. To prevent accidents arising from the misuse of this instrument, please
ensure the operator receives this manual.

Notes• This instrument should be used in accordance with the specifications described in the manual.
If it is not used according to the specifications, it may malfunction or cause a fire.
• Be sure to follow the warnings, cautions and notices. If they are not observed, serious injury
or malfunction may occur.
• Specifications of the SAFD and the contents of this instruction manual are subject to change
without notice.
• Care has been taken to assure that the contents of this instruction manual are correct, but if there
are any doubts, mistakes or questions, please inform our sales department.
• This instrument is designed to be installed on a DIN rail within a control panel. If it is not, measures
must be taken to ensure that the operator does not touch power terminals or other high voltage
sections.
• Any unauthorized transfer or copying of this document, in part or in whole, is prohibited.
• Shinko Technos CO., LTD. is not liable for any damage or secondary damage(s) incurred as aresult of using this product, including any indirect damage.
SAFETY PRECAUTIONS (Be sure to read these precautions before using our products.)
The safety precautions are classified into categories: “Warning” and “Caution”.
Depending on the circumstances, procedures indicated by
Caution may cause serious
results, so be sure to follow the directions for usage.
which may lead to dangerous conditions and cause
Warning Procedures
death or serious injury, if not carried out properly.

CautionProcedures which may lead to dangerous conditions and cause
superficial to medium injury or physical damage or may degrade or
damage the product, if not carried out properly.

Warning• To prevent an electric shock or fire, only Shinko or qualified service personnel may handle
the inner assembly.
• To prevent an electric shock, fire or damage to instrument, parts replacement may only beundertaken by Shinko or qualified service personnel.

Safety precautions• To ensure safe and correct use, thoroughly read and understand this manual before using this
instrument.
• This instrument is intended to be used for industrial machinery, machine tools and measuring
equipment. Verify correct usage after consulting purpose of use with our agency or main office.
(Never use this instrument for medical purposes with which human lives are involved.)• External protection devices such as protection equipment against excessive temperature rise, etc.
must be installed, as malfunction of this product could result in serious damage to the system or
injury to personnel. Also proper periodic maintenance is required.
• This instrument must be used under the conditions and environment described in this manual.
Shinko Technos Co., Ltd. does not accept liability for any injury, loss of life or damage occurring
due to the instrument being used under conditions not otherwise stated in this manual.
Caution with respect to Export Trade Control Ordinance
To avoid this instrument from being used as a component in, or as being utilized in the manufactureof weapons of mass destruction (i.e. military applications, military equipment, etc.), please
investigate the end users and the final use of this instrument.
In the case of resale, ensure that this instrument is not illegally exported.
2

1. Installation precautions

CautionThis instrument is intended to be used under the following environmental conditions(IEC61010-1): Overvoltage category , Pollution degree 2
Ensure the mounting location corresponds to the following conditions:• A minimum of dust, and an absence of corrosive gases• No flammable, explosive gases• No mechanical vibrations or shocks• No exposure to direct sunlight, an ambient temperature of -5 to 55 (23 to 131 )that does not change rapidly, and no icing• An ambient non-condensing humidity of 35 to 85%RH• No large capacity electromagnetic switches or cables through which large current is flowing• No water, oil or chemicals or where the vapors of these substances can come into
direct contact with the unit
• When installing this unit within a control panel, take note that ambient temperature ofthis unit as well as the control panel must not exceed 55 (131 ). Otherwise the lifeof electronic components (especially electrolytic capacitor) may be shortened.
Note • Avoid setting this instrument directly on or near flammable material eventhough the case of this instrument is made of flame-resistant resin.

2. Wiring precautions

Caution• Do not leave bits of wire in the instrument, because they could cause a fire and malfunction.
• When wiring terminals, use ferrules with an insulation sleeve and crimping pliers madeby Phoenix Contact GMBH & CO. applicable to terminals.
• Tighten the terminal screw using the specified torque.
If excessive force is applied to the screw when tightening, the screw or case may bedamaged.
• This instrument has no built-in power switch, circuit breaker or fuse. It is necessary to
install them near the instrument.
(Recommended fuse: Time-lag fuse, rated voltage 250V AC, rated current 2A)• For wiring of AC power source, be sure to use exclusive terminals as described in thismanual. If AC power source is connected to incorrect terminals, the unit will burn out.
• For a 24V DC power source, do not confuse polarity when wiring.
• Do not apply a commercial power source to the sensor connected to the input terminal
nor allow the power source to come into contact with the sensor, as the input circuit
may burn out.
• Keep the input wire, power line and output wire away from one another.

3. Operation and maintenance precautions

Caution• Do not touch live terminals. This may cause electric shock or problems in operation.
• Turn the power supply to the instrument OFF when retightening the terminal andcleaning. Working or touching the terminal with the power switched ON may result insevere injury or death due to Electric Shock.
• Use a soft, dry cloth when cleaning the instrument.
(Alcohol based substances may tarnish or deface the unit.)• As the display section is vulnerable, do not strike or scratch it with a hard object orpress hard on it.
3Characters used in this manual
Indication
Number,

/

-1

0

1

2

3

4

5

6

7

8

9

A

B

C

D

E

F

G

H

I

J

K

L

M

N

O

P

Q

R

S

T

U

V

W

X

Y

Z

Indication
Alphabet
Indication
Alphabetmeans that no character is indicated (unlit) on the display.

---

CONTENTS ---1. ModelPage
1.1 Model ----------------------------------------------------------------------------- 5
1.2 How to read the model label ------------------------------------------------ 5
2. Name and functions of sections ------------------------------------ 6

3. Mounting

3.1 External dimensions (Scale: mm) ----------------------------------------- 6
3.2 Mounting and removal to/from the DIN rail ----------------------------- 7

4. Wiring

4.1 Recommended ferrules ------------------------------------------------------ 8
4.2 Terminal arrangement and circuit configuration ----------------------- 9
4.3 Wiring of terminals ------------------------------------------------------------ 9
4.3.1 Power source wiring ----------------------------------------------------- 9
4.3.2 Output wiring -------------------------------------------------------------- 9
4.3.3 Input wiring ----------------------------------------------------------------- 9
5. Operation flowchart ------------------------------------------------------ 10

6. Setup

6.1 Indication after power ON ------------------------------------------------- 11
6.2 Basic operation of setup --------------------------------------------------- 12
6.3 Setup of the unit ------------------------------------------------------------- 13
6.3.1 When using this unit as a signal conditioner --------------------- 15
6.3.2 When using the Reverse function ---------------------------------- 15

7. Unit operation

7.1 Indication after power-on -------------------------------------------------- 16
7.2 Unit operation ---------------------------------------------------------------- 16
8. Specifications --------------------------------------------------------------- 17

9. Troubleshooting

9.1 Indication ---------------------------------------------------------------------- 19
9.2 Key operation ---------------------------------------------------------------- 19
9.3 Unit operation ----------------------------------------------------------------- 19
10. Character table ------------------------------------------------------------- 20

4

1. Model
1.1 Model
SAFD - 0
Input

OutputPower supply

0
1
2
0
1
2
0
1

Series name: SAF
Open collector
Voltage pulse
Line driverOpen collector
5V voltage pulse
12V voltage pulse
100 to 240V AC
24V AC/DC(Example) SAFD-000-0 Input frequency: 800Hz, Pulse rate: 1/50
Default value: Input frequency: 9999Hz
Pulse rate: 1/1
1.2 How to read the model label
The model label is attached to left side of the case.
Model (e.g. SAFD-000-0)
Input
OutputSerial number(Fig. 1.2-1)

5

2. Name and functions of sections
1 Power indicator (Green)

Lights when the power to the instrument isturned on.

1

2 Input display (Red)

Indicates the input value during Run mode.
Indicates characters of setting (or adjustment)
item during Setup mode.

2
3

3 Output display (Green)

Indicates output volume during Run mode.
Indicates set (or adjusted) value during Setup
mode.

6

4

7

5

4 Up key ()
Increases the numeric value, or switches theselection items.

5 Down key ()
Decreases the numeric value, or switches theselection items.

6 Mode key ()
Switches the setting mode and registers theset (or selected) value.

(Fig.2.1)

7 Sub-mode key (Unmarked)

If the Mode key is pressed while holding down
this key, the unit proceeds to the Setup mode.

3. Mounting
3.1 External dimensions (Scale: mm)

75

DIN rail

97
22.5

100
(Fig. 3.1-1)
64

3.2 Mounting and removal to/from the DIN rail

Caution• Mount the DIN rail horizontally.
• To remove this instrument, a flat blade screwdriver is required for pulling down
the lever.
Never turn the screwdriver when inserting it into the release lever.
If excessive power is applied to the lever, it may break.
• Be sure to use commercially available fastening plates at both ends of the unit
if it is in a position susceptible to vibration or shock.
Recommended fastening plate
Manufacturer
Omron Corporation
IDEC Corporation
Panasonic Electric Works Co., Ltd.

Model
End plate
PFP-M
Fastening plate BNL6
Fastening plate ATA4806

Mounting to the DIN rail (Fig. 3.2-1)
First, hook 1 of the instrument on the upper side of the DIN rail.
Second, making 1 part of the instrument as a support, fit the lower part 2 of the
instrument to the DIN rail.
The unit will be completely fixed to the DIN rail when a “Click” sound is heard.
Removal from the DIN rail (Fig.3.2-2)
First, insert a flat blade screwdriver into the release lever ( 1 ).
Second, remove the instrument from the DIN rail by pulling down the lever ( 2 ).
Release lever
1

2

2(Fig. 3.2-1) Mounting

1(Fig. 3.2-2) Removal

7

4. Wiring
Warning
Turn the power supply to the instrument off before wiring.
Working or touching the terminal with the power switched on may result in severe
injury or death due to Electric Shock.
4.1 Recommended ferrules
When using ferrules, use the following recommended ferrules and crimping pliers made by
Phoenix Contact GMBH &CO. See (Table 4.1-1)
Take note that screw size and tightening torque differ depending on the terminal number.
(Table 4.1-1)
Terminal Terminal Ferrules with
Conductor
numberscrew insulation sleeve cross sections
1 to 4
M2.6
AI 0.25-8 YE
0.2 to 0.25mm2

5 to 9M2.0

AI 0.34-8 TQ
AI 0.5-8 WH

0.25 to 0.34mm
0.34 to 0.5mm2

AI 0.75-8 GY

0.5 to 0.75mm2

AI 1.0-8 RD
AI 1.5-8 BK

0.75 to 1.0mm2
1.0 to 1.5mm2

AI 0.25-8 YE

0.2 to 0.25mm2

AI 0.34-8 TQ
AI 0.5-8 WH

0.25 to 0.34mm
0.34 to 0.5mm2

8Tightening
torque0.5 to 0.6N•m
2

Crimping pliers
CRIMPFOXZA 3
CRIMPFOXUD 6

0.22 to 0.25N•m
2

4.2 Terminal arrangement and circuit configuration+
Input
Power supply
for sensor+

5

Insulation
circuitInput
circuitCPU

Outputcircuit3

Power
circuit16
7
8

Fixed voltage Display/Setting
key circuit
circuit 12V

9

4

2

+
Output+Power supply-

Input connection example
+

5
6

6

-

12V

7

5

7

8

8

9

9

Open collector

Voltage pulse

12V

5Line driver

6
7
8
9

(Fig. 4.2-1)
4.3 Wiring of terminals

Warning• For 100 to 240V AC, if AC power source is connected to incorrect terminals,
this instrument will burn out.
• For a 24V DC power source, do not confuse polarity when wiring.
4.3.1 Power source wiring
Use terminals 1 (+) and2 (-) for the power supply to the instrument.

4.3.2 Output wiring
Use terminals 3 (+) and4 (-) for the output wiring.

4.3.3 Input wiring
Terminals for wiring differs depending on the input specifications.
Refer to (Fig. 4.2-1).

9

5. Operation flowchart
Power ONWarm-up indication (Approx. 2sec.)

Run mode

Press thekey
while holding down
the Sub-mode key.

IN

Setup modeSet value lock

OUT

IN
OUT

IN
OUT

IN

Press thekey.
Frequency
type/unit

IN

Press thekey.
Frequency
high limit valueIN

Press thekey.
Input 0% value

OUT

IN
OUT

IN
OUT

OUT

IN

Press thekey.
Input 100%
value

IN

Press thekey.
Input decimal
point place

IN

key.

IN
OUT

Press thekey.
Output type/unit

OUT

OUT

Press thePulse ratefrequency

OUT

OUT

IN
OUT

Press thekey.
Output high
limit

IN

Press thekey.
Output 0%
value

IN

Press thekey.
Output 100%
value

IN

Press the

10

key.

OUT

OUT

Output decimal
point place
Press thekey.
One-shot output
pulse width
Press thekey.
OutputNormal/Reverse
Press theDisplay
selection

key.

Press thekey.
Indication time

OUT

Press thekey.

6. Setup
Setup should occur before using this unit, to set the Frequency type/unit, Frequency high limit
value, Input 0% value, Input 100% value, Pulse rate frequency, One-shot output pulse width,
Output 0% value, Output 100% value, Output Normal/Reverse, etc. according to the users’
specifications.
If the users’ specifications are the same as the default value of the instrument, or if setup has
already been completed, it is not necessary to set up the instrument. Proceed to Section “7.
Unit operation”.
(Table 6-1)
Setting item
Default value
Set value lock
UnlockFrequency type/unit
Frequency measurement Hz
Frequency high limit value
9999HzInput 0% value
0
Input 100% value
9999
Input decimal point place
No decimal point
Pulse rate frequency
1
Output type/unit
Frequency measurement Hz
Output high limit value
9999
Output 0% value0
Output 100% value
9999
Output decimal point place
No decimal point
One-shot output pulse width
0ms
Output Normal/Reverse
NormalDisplay selection
Input/Output indication
Indication time
00.00 (Continuous)
6.1 Indication after power-on
After power-on, warm-up status below (Fig. 6.1-1) is indicated for approx. 2sec.
Indicates the model.

Lights whenpower-on.

(Fig. 6.1-1)
After that, the unit switches to the Run mode as shown below (Fig. 6.1-2).
Indicates the input value.
Flasheswhen nothing is connected to the
input terminals.

(Fig. 6.1-2)

Indicates the output volume.
Indicates the pulse rate frequency
corresponding to the input value.
Indicateswhen nothing is connected tothe input terminals.
116.2 Basic operation of setup
Setup is conducted in the Setup mode.
To enter the Setup mode, press the
key while holding down the Sub-mode key in the
Run mode. (Fig. 6.2-1)
To set (select) each item, use theorkey, and register the value with the
key.
(Fig. 6.2-2)(1) Run mode

To enter the Setup mode, press thekey 2 while holding down the
Sub-mode key 1 in the Run mode.

2
1(Fig. 6.2-1)(2) Setup mode

To proceed to each setting item andto register the set (selected) value,
use this key.

To set (or select)each item, use
these keys.
(Fig. 6.2-2)

12

6.3 Setup of the unit
The following shows all setup items. Set up the unit referring to the explanation of each item.
Display
IN

OUT

IN
OUT

IN
OUT

IN
OUT

IN
OUT

IN
OUT

IN
OUT

IN
OUT

Name, Function, Setting range
Default value
Set value lock
UnlockLocks the set values to prevent setting errors.
: Unlock: Lock (None of the set values can be changed.)
Frequency type/unit
Frequency measurement Hz
Sets input frequency type (Low pulse/Frequency) and unit [mHz/Hz(Lowpulse)/Hz/kHz].
: Pulse measurement mHz (Low pulse): Pulse measurement Hz (Low pulse): Frequency measurement Hz: Frequency measurement kHz
Frequency high limit value
9999HzSets input frequency high limit value. (Frequency low limit value is fixed.)
Pulse measurement mHz (Low pulse) : 10 to 9999mHz
Pulse measurement Hz (Low pulse) : 1 to 100Hz
Frequency measurement Hz: 50 to 9999Hz
Frequency measurement kHz: 1 to 15kHz
0
Input 0% value
Sets the value (indicated on the Input display) at 0% input.
Setting range: -1999 to Input 100% value
9999
Input 100% value
Sets the value (indicated on the Input display) at 100% input.
Setting range: Input 0% value to 9999
No decimal point
Input decimal point place
Selects decimal point place for input.
: No decimal point: 1 digit after decimal point: 2 digits after decimal point: 3 digits after decimal point
Pulse rate frequency

1

Sets pulse rate frequency.
Pulse rate is shown in the equation below.
1
1
Pulse rate=
XX
Pulse rate frequency
Pulse unit is selected during Frequency type/unit selection.
Setting range: 1 to 9999
Frequency measurement Hz
Output type/unit
Sets output frequency type (Low pulse/Frequency) and unit [mHz/Hz(Lowpulse)/Hz/kHz].
: Pulse measurement mHz (Low pulse): Pulse measurement Hz (Low pulse): Frequency measurement Hz: Frequency measurement kHz
13

IN

Output high limit value

OUT

Sets output frequency high limit value. (Frequency low limit value is fixed.)
Pulse measurement mHz (Low pulse): 10 to 9999mHz
Pulse measurement Hz (Low pulse) : 1 to 100Hz
Frequency measurement Hz: 50 to 9999Hz
Frequency measurement kHz: 1 to 15kHz
0
Output 0% value

OUT

IN
OUT

IN
OUT

IN
OUT

IN

OUT

IN
OUT

IN

OUT

9999

Sets the value (indicated on the Output display) at 0% output.
Setting range: -1999 to Output 100% value
9999
Output 100% value
Sets the value (indicated on the Output display) at 100% output.
Setting range: Output 0% value to 9999
No decimal point
Output decimal point place
Selects the output decimal point place.
: No decimal point: 1 digit after decimal point: 2 digits after decimal point: 3 digits after decimal point
0ms
One-shot output pulse width
If a pulse is entered, outputs “one-shot output pulse” during one-shot pulse
width setting time.
Not available when setting to 0.
Setting range: 0 to 400ms
NormalOutput Normal/Reverse
Selects either Normal mode (0.0 to 100.0%) or Reverse mode (100.0 to0.0%) for output status.
: Normal: Reverse
Input/Output indication
Display selection
Selects an indication type on the display.
: Input/Output indication: Input indication: Output indication: No indication (Only the power indicator is lit.)
00.00 (Continuous)
Indication time
Sets the indication time of the display after the final key operation.
Not available if No indication (Only the power indicator is lit) is selectedduring Display selection.
After the indication time has elapsed, the displays go off (Only the power
indicator is lit.).
If power is turned on again, or if any of the,,or Sub-mode keys
are pressed while displays are unlit, the displays will light again.
Setting range:
00.00: Continuous
00.01 (1 second) to 60.00 (60 minutes) [Minutes.Seconds]
146.3.1 When using this unit as a standard pulse scaler
Set the Output Normal/Reverse selection to “Normal”. (Fig. 6.3-1)
6.3.2 When using the Reverse function
This function reverses the output (100 to 0%) that corresponds to the input (0 to 100%).
Set the Output Normal/Reverse selection to “Reverse”. (Fig. 6.3-1)
Pulse width accuracy: 1ms
Input
Max. delay
15 s

One-shot output
pulse width setting

Without One-shot

With One-shot

Output(Normal)Output(Reverse)(Fig. 6.3-1)

15

7. Unit operation
7.1 Indication after power-on
After power-on, the following warm-up status is indicated for 2 seconds (Fig. 7.1-1).
Indicates the model.

Lights after power-on.

(Fig. 7.1-1)
7.2 Unit operation
The unit enters the Run mode after approx. 2-second warm-up as shown in (Fig. 7.2-1).
Outputs the input signal with the pulse rate set during Pulse rate frequency setting.
Indicates the input value.
Flasheswhen nothing is connectedto the input terminals.
Indicates the output volume.
Indicates the pulse rate frequency
corresponding to the input value.
Indicateswhen nothing is connectedto the input terminals.
(Fig. 7.2-1)• Indication when input value is 10000 or more
For the indication of 10000 or more, the lower 4 digits of input value are flashing.
(e.g.) Indication of 10020
IN
Flashes.
• Input display indication when pulse is absent
When pulse is absent, 0 (zero) flashes.
• Overrange indication
In case of overrange (1.1 times frequency high limit value), "and Output displays.

" flashes on the Input• Indication time setting
If indication time is set, the displays will go off after the indication time has elapsed.
(Only the power indicator is lit.)
If power is turned on again, or if any of the,,or the Sub-mode keys arepressed while displays are unlit, the displays will light again.

16

8. Specifications
Input specifications
Pulse or frequency measurement via input signals
Open collector
Frequency range: 0.001Hz to15kHz
Minimum pulse width: 5 s or more (for ON and OFF)
Input detection voltage/current : ON : Max. 30mA (30V or less)
OFF: Residual voltage, 0.5V or lessAction input conditions: ON : 200 or less
OFF: 100k or more
Max rated input frequency : 20kHz
Voltage pulse
Frequency range: 0.001Hz to 15kHz
Minimum pulse width: 5 s or more (for High and Low)
Waveform: Rectangular, sine waveform or similar
Detection level: Low: 1V DC or less
High: 2V DC or more
Input impedance: 10k or more
Input amplitude: 2 to 50Vp-p
Max rated input frequency : 20kHzLine driver
AM26LS31 or equivalent
Receiver: AM26LS32 or equivalent
Output specifications
Open collector
Output rating
Max. frequency
Voltage pulse
Output rating
Allowable load resistanceMax. frequency
Performance
Reference accuracy
Display accuracy
Insulation resistanceDielectric strength: 12V DC/30mA: 15kHz: 5V, 12V DC 10%: 500 or more
: 15kHz: Within 0.1%
: Within Reference accuracy 1 digit: Input – Output – Power: 10M or more, at 500V DC: Input – Output – Power: 2.0kV AC for 1 minute17General structure
Case
Front panel
Setting
Displays, Indicator: Flame-resistant resin, Color: Light gray
: Membrane sheet: By the front keypad: Input display : 7 segments Red LED display 4 digits
Character size: 7.4 x 4.0mm (H x W)
Output display : 7 segments Green LED display 4 digits
Character size: 7.4 x 4.0mm (H x W)
Power indicator: Green LED

Installation specifications
Power supply: 100 to 240V AC 50/60Hz, 24V AC/DC 50/60Hz
Allowable voltage range : 85 to 264V AC, 20 to 28V AC/DCPower consumption: Approx. 9VA
Power supply for sensor : 12V DC 5%, 25mA
Ambient temperature: -5 to 55 (23 to 131 )Ambient humidity: 35 to 85%RH (Non-condensing)
Weight: Approx. 120g
Mounting: DIN rail mounting
External dimensions: W22.5 x H75 x D100mm
Attached functions• Power failure countermeasure
The setting data is backed up in the non-volatile IC memory.
• Self-diagnosis
The CPU is monitored by a watchdog timer, and when an abnormal status is foundon the CPU, the unit is switched to warm-up status after turning all outputs off.
• Detecting unconnected sensor
If pulse is not detected for a constant period (Low pulse: 1000sec or 100sec,
Frequency: 1sec), the unit will revert to the initial status (0Hz).

18

9. Troubleshooting
9.1 Indication
ProblemPresumed cause and solution

The Input and Output
displays are flashing"".

• Check whether the input exceeds 1.1 times frequency highlimit value.
Confirm the input signal source.
• Check whether the sensor is securely connected to the input
terminals of this instrument.
Connect the sensor terminals to the input terminals of this
instrument securely.
• Check whether the signal source is correct.
Check the input signal source.
Input value 0 (zero) is• Check if pulse is absent.
flashing.
Confirm the input signal source
The indication of the Input • AC leaks into the sensor circuit.
display is irregular or
Use an ungrounded type sensor.
unstable.
• There may be equipment that interferes with or makes noise
near the unit.
Keep equipment that interferes with or makes noise awayfrom the unit.
9.2 Key operation
Problem
Setting is not possible.

Presumed cause and solution• “Lock” is selected during Set value lock selection.
Select “Unlock”.

9.3 Unit operation
Problem
Input value does notchange.

No output

Presumed cause and solution• The sensor may be out of order. Change the sensor.
• Check whether input and output wires are securely
connected to the I/O terminals of the instrument.
Ensure that input and output wires are securely connectedto the I/O terminals.
• Check whether the wiring of input and output is correct.
• Check whether Output 100% and Output 0% value havebeen set to suitable values.
• Check whether Output Normal/Reverse has been selected
correctly during Output Normal/Reverse selection.

19

10. Character table
All setting items are indicated in the following tables, however, some items will not be
indicated depending on the specifications.
Setup mode
Display
Setting item
Default value
Set value lock
UnlockFrequency type/unit
Frequency measurement: Hz
9999HzFrequency high limit value0
Output 0% value
9999
Output 100% value
Input decimal point place
No decimal point
Pulse rate frequency
1
Output type/unit
Frequency measurement: Hz
9999
Output high limit value
Output 0% value
09999
Output 100% value
Output decimal point place
No decimal point
One-shot output pulse width 0ms
Output Normal/Reverse
NormalDisplay selection
Input/Output indication
Indication time
00.00 (Continuous)

Data******* Inquiries *******For any inquiry about this unit, please contact the vendor where youpurchased the unit or our agency after checking the following.
(e.g.)• Model ........................………. SAFD-0• Serial number ….................... No. xxxxxx
In addition to the above, please let us know the details of malfunction, if any,
and the operating conditions.

SHINKO TECHNOS CO.,LTD.
OVERSEAS DIVISION

Reg. Office : 2-5-1, Senbahigashi, Minoo, Osaka, JapanURL: http://www.shinko-technos.co.jp
Tel : 81-72-727-6100
E-mail: overseas@shinko-technos.co.jp
Fax: 81-72-727-7006
No.SA72E2 2009.04