Technical SheetGL SPECTIS 1.0 TOUCH
The world’s first smart spectrometer. Often copied but never duplicated this isa reliable and versatile device for accurate absolute spectral measurement.
GL SPECTIS 1.0 Touch is the world’s first intuitively operated touch screen
version of our successful SPECTIS 1.0 product line. If you need to measure lux,lumen, CRI, CCT, color, mWatt and much more, our highly portableand precise GL SPECTIS 1.0 Touch is the perfect solution.
Features: Completely portable device
 Color LCD Touch screen
 Communication features: USB cable, SDcard slot
 Android based operating system Up to 4 hours on battery
APPLICATION
Application

Natural light, LEDs, halogen light, etc.

LED MEASUREMENT
Illuminance (lux)*

10 – 100 000 lx (for white LED)

Luminance [cd/m2]

Available with optional GL OPTI PROBE

Luminous flux [lm]Available with optional GL OPTI SPHERE

Luminous intensity [cd]Calculated in GL SPECTROSOFT

Irradiance [W/m ]0.03 – 600 W/m² (for white LED)

Illuminance classClass B – DIN 5032-7
Class AA – JIS C 1609-1:2006

Tolerance – cosine response (f2‘)< 3 % (1,9 %)Spectral range**

340 – 780 nm (UVa – VIS)
640 – 1050 nm (VIS – NIR)
340 – 750 nm (UVa – VIS)

2

CALCULATED VALUESCRI – Color rendering index
according to CIE

Ra, R1 – R14

CRI according to TM-30-15

R15

CCT – Correlated color temperatureaccording to CIE 13.3✓

Color peak✓

Color dominantoptional with GL SPECTROSOFT

Color position coordinates [x,y]according to CIE 1931✓

Color position coordinates [u‘,v‘]
according to CIE 1976✓

Color position coordinates [u, v]
according to CIE 1960✓PAR/PPFD✓Color coordinate error

optional with GL SPECTROSOFT

Metameric index

optional with GL SPECTROSOFT

Binning

optional with GL SPECTROSOFT

Assessment in accordance with ISO 3664

✓GL OPTIC Polska Sp. z o.o. Sp.kul. Poznańska 70, 62-040 Puszczykowo, Poland
Phone: +48 61 819 40 03 | E-mail: office@gloptic.com
www.gloptic.com

SPECTIS 1.0 Touch UVa – VIS
SPECTIS 1.0 Touch VIS – NIR
SPECTIS 1.0 Touch LS

Technical SheetGL SPECTIS 1.0 TOUCH
PHOTOMETRY / RADIOMETRYSensor

CMOS image sensor

Number of pixels

256

Physical resolution / datapoint interval~ 1.7 nmWavelength reproducibility

0.5 nmIntegration time

10 ms – 10 s

A/D converter

16 bitSignal-to-noise ratio

1000:1

Stray light

2*10 E-3

Optical resolution / FWHM

10 nm

Radiometric accuracy*** ****

5 % within range 340 – 500 nm
4 % within range 500 – 1050 nm

Flicker compensation✓

Temperature sensor anddark current compensation✓

Uncertainty of color coordinates***0.0015

Automatic accessory detection✓/GENERAL PROPERTIES
Operating SystemAndroid

Power supply via USB connector< 640 mA

Power adapter

Power supply unit 100…240 V (50/60 Hz) 0,15 A

Battery / Power packLi-ion battery 3500 mAh

Automatic shut-off✓Battery life

up to 6 h*****

Operating temperature5 – 35 °C

Dimensions [H x W x D]

74.5 mm x 145.5 mm x 36.6 (with standard diffusor)

Weight349 g

Tripod adapter✓INTERFACE & MEMORYUSB

USB 2.0Trigger

Open collector, minijack 3.5mm, 3-pin

SD Card slot

microSD

Measurement result storage

Auto / 4 GB microSD

Data format

XML

Fiber optic connector

Optional SMA905D

DISPLAY & OPERATION
Display

3.5“ color LCD (240 x 320px)

Operation

Touch Screen, PC / Notebook

SOFTWARE
SoftwareOptional GL SPECTROSOFT Basic / Pro / Lab

GL OPTIC Polska Sp. z o.o. Sp.kul. Poznańska 70, 62-040 Puszczykowo, Poland
Phone: +48 61 819 40 03 | E-mail: office@gloptic.com
www.gloptic.com

Technical SheetGL SPECTIS 1.0 TOUCH
ORDERING INFORMATIONCase✓

Battery✓

USB cable✓

Power supply

✓

Leash

✓

Display protection foil✓

4GB microSD card✓

Part number

GLX 1.0t no. 106260
GLX 1.0t no. 200862
GLX 1.0t no. 202032

SPECTIS 1.0 touch UVa – VIS
SPECTIS 1.0 touch VIS – NIRSPECTIS 1.0 Touch LS*

Dynamic range is spectrum related and should be calculated separately for any light source. Estimated dynamic range for typical 4000 K white LED.
Range estimated for optical system made to default specification. Alterations of that are often possible. Please consult technical support if you are looking for specific parameters.

**

Spectral range of the sensor. Actual spectral range of system may be reduced due to limitations of used optical accessory.

***

Absolute measurement uncertainty immediately after calibration. The expanded uncertainty corresponds to a coverage probability of 95 % and the coverage factor k = 2.
Parameters valid in laboratory conditions 25deg C, relative humidity 45%.

**** Applies only within the spectral range of the given model.
***** In moderate use – continuous measurements and WiFi significantly increase energy consumption.
Note: Instrument, firmware and software specification are subject to change without prior notice. All information included in GL OPTIC datasheets and product information available in any form are carefully prepared and included
information believed to be true. Please note that discrepancies may occur due to text and/or other errors or changes in the available technology. We advise to contact GL Optic before the use of the product to obtain the latest productspecification.

GL OPTIC Polska Sp. z o.o. Sp.kul. Poznańska 70, 62-040 Puszczykowo, Poland
Phone: +48 61 819 40 03 | E-mail: office@gloptic.com
www.gloptic.com