Data sheet
testo 890

Thermal imager
testo 890 – Thermography for thehighest demands

Infrared resolution 640 x 480 pixels
testo SuperResolution technology to 1280 x 960 pixels
Thermal sensitivity < 40 mK
Flexibility thanks to rotatable handle and fold-out, rotatabledisplay
Exchangeable lenses°C

Special measurement mode for mould-risk areas%RH

High temperature measurement up to 1,200 °C
Panorama image assistant
SiteRecognition technology
Fully radiometric video measurement and image sequence
capturing

The thermal imager testo 890 offers outstanding image

measurement objects, e.g. on industrial plants, can bequality for the highest thermographic demands. Thankssafely recorded thermographically in best image quality andto the high-quality infrared measurement system with ahigh resolution. Even thermal processes can be precisely

640 x 480-pixel detector, thermal images can be recorded

analyzed over time using the fully radiometric video

in megapixel quality (1290 x 960) using the SuperResolution

measurement: all measurement points of the thermal image

technology. This means that even the smallest measurement

are available, accurate to the pixel, at all times.

objects such as electronic components or far distant

www.testo.com

testo 890

Ordering data
testo 890 thermal imagers

Part no.

testo 890 thermal imager with integrated testo SuperResolution and one lens (choice between 42° standard lens,
25° lens and 15° telephoto lens) in a robust case, including pro software (free download), SD card, USB cable,carrying strap, lens-cleaning cloth, mains unit, Li-ion rechargeable battery and headset

0563 0890 X1

Thermal imager testo 890 with integrated testo SuperResolution and super-telephoto lens in a robust caseincl. pro software (free download), SD card, USB cable, carrying strap, lens-cleaning cloth, mains unit,
Li ion rech. battery, headset

0563 0890 X4

testo 890 sets with yourselection of lenses
testo 890 thermal imager with integrated testo SuperResolution in arobust case, including pro software (free download), SD card,
USB cable, carrying strap, lens-cleaning cloth, mains unit, Li-ion
rechargeable battery, lens protection glass, spare rechargeable battery,
fast charger, headset and lens case. Choice between 42° standard lens,25° lens and/or 15° telephoto lensPart no.
testo 890 set with two lenses – see above for further set components

0563 0890 X2

testo 890 set with three lenses – see above for further set components

0563 0890 X3

testo 890 set with SuperTele and one lens – see above for further set components

0563 0890 X5

testo 890 set with SuperTele and two lenses – see above for further set components

0563 0890 X6

Accessories
Code1)(Initial equipment)

Part no.
(Retrofit)

included indelivery

0554 7806

Lens protection glass. Special Germanium protection glass for optimum protection of thelens from dust and sctratching

F10554 0289

Additional battery.
Additional lithium-ion rechargeable battery for extending the operating time.

G10554 8852

Fast battery charger. Desktop charging station for two rechargeable batteries for theoptimization of the charging time.

H10554 8851

High temperature measurement up to +1,200 °C

I12)

Humidity measurement 3)

E12)

Telephoto lens 15° x 11°

D12)

25° Lens

O12)

Super-telephoto lens 6.6° x 5°

T22)

Process analysis package:
image sequence capturing in instrument and fully radiometric video measurement

V10554 8902

FeverDetection

J1SuperResolution.
Four times more measurement values for even more detailed analysis of the thermal images

Emission adhesive tape. Adhesive tape, e.g. for reflective surfaces (roll, L.: 10 m, W.: 25 mm),
ε = 0.95, temperature resistant to +250 °C
PC software testo IRSoft for data analysis and reporting

0501 8809

ISO calibration certificates;
Calibration points at 0 °C, +25 °C, +50 °C

0520 0489 4)

ISO calibration certificates;
Calibration points at 0 °C, +100 °C, +200 °C

0520 0490 4)

ISO calibration certificates;
Freely selectable calibration points in the range -18 to +250 °C

0520 0495 4)

 hen ordering as first equipment, you receive the accessories directly in the case.
W
Example: testo 890 incl. lens protection glass and SuperResolution:
Order no. 0563 0890 X1 F1 S12)
Please contact our customer service
1)–0554 0051

 ireless humidity probes only in the EU, Norway, Switzerland, USA, Canada,WColombia, Turkey, Brazil, Chile, Mexico, New Zealand, Indonesia.
Per lens5)
Plus installation3)

4)

testo 890

Technical data

Infrared image output

Imager equipment

Infrared resolution

640 x 480 pixels

Digital camera4

Thermal sensitivity (NETD)< 40 mK at +30 °C

Lens version

Field of view/min. focus
distance(Lens version)

42° x 32° / 0.1 m (Standard)
25° x 19° / 0.2 m (25° Lens)
15° x 11° / 0.5 m (Telephoto)
6.6° x 5° / 2 m (Super-telephoto)

42° x 32° (Standard)
25° x 19° (25° Lens)
15° x 11° (Telephoto)
6.6° x 5° (Super-telephoto)

Geometric resolution(IFOV)(Lens version)

1.13 mrad (Standard)
0.68 mrad (25° Lens)
0.42 mrad (Telephoto)
0.18 mrad (Super-telephoto)

SuperResolution(pixel / IFOV)(Lens version)

Image refresh rate

1280 x 960 pixels / 0.71 mrad (Standard)
1280 x 960 pixels / 0.43 mrad (25° Lens)
1280 x 960 pixels / 0.26 mrad (Telephoto)
1280 x 960 pixels / 0.11 mrad
(Super-telephoto)
33 Hz*

Focus

auto / manualSpectral range

7.5 to 14 µm

Image output visual
Image size /min. focus distance

3.1 MP / 0.5 m

Image display

4.3“ LCD touchscreenwith 480 x 272 pixels

Digital zoom

1- to 3-fold

Display options

IR / real image

Video output

USB 2.0, Micro HDMI

Colour palettes

9 (iron, rainbow, rainbow HC, cold-hot,blue-red, grey, inverted grey, sepia,
Testo)

Measurement

AccuracyHigh temperature
measurement - optional
Accuracy-30 to +100 °C /0 to +350 °C (switchable)
0 to +650 °C (switchable)±2 °C, ±2 % of measurement value(larger value applies)(±3 °C of m.v. at -30 to -22 °C)+350 to +1200 °C (not in connection withthe super-telephoto lens)±2 °C, ±2 % of m.v.

Emissivity / reflected
temperature

0.01 to 1 / manual

Transmission correction(atmosphere)

4

4

Panorama image assistant4

Laser (laser classification
635 nm, Class 2)***

Laser marker

Voice recording

Bluetooth****/
wired headset

Video measurement(via USB)

up to 3 measurement points

Process analysis package:
image sequence capturing
in instrument andfully radiometric video
measurement(4)

FeverDetection(4)

Interface

Image presentation

Measuring range

SiteRecognition (measurement site recognition withimage management)

LabVIEW, interface description downloadon the Testo homepage

Image storage
File format single image.bmt; Exportmöglichkeit in .bmp, .jpg,.png, .csv, .xls

File format video (via USB).wmv, .mpeg-1 / Testo format(fully radiometric video)

Storage device

SD cart 2GB(approx. 1500 - 2000 images)

Power supply
Battery typeFast-charging, Li-ion battery can bechanged on-site

Operating time

4.5 hours

Charging optionsin instrument / in charger (optional)

Mains operation

4Ambient conditions
Operating temperaturerange-15 °C to +50 °CStorage temperature range
Air humidity-30 to +60 °C
20 to 80 % RH non-condensing

Housing protection class(IEC 60529)

IP54

Measuring functions

Vibration (IEC 60068-2-6)

2G

Display of surface moisture distribution (usingmanual input)

4

Physical specifications

Humidity measurementwith radio humidity probe(automatic measurement
value transfer in realtime)**(4)

Solar mode

4

Analysis function***

Weight1630 gDimensions (L x W x H)

253 x 132 x 111 mm

Tripod mounting

1/4“ - 20UNC

HousingABS

PC software
System requirements
up to 10 measurement points,Hot/Cold Spot Recognition,
up to 5 x area measurement(min/max & average),
Isotherm and alarm values

inside the EU, outside 9 Hz
Wireless humidity probes only in the EU, Norway, Switzerland, USA, Canada,Colombia, Turkey, Brazil, Chile, Mexico, New Zealand, Indonesia*** excepting USA, China and Japan**** Bluetooth only in the EU, Norway, Switzerland, USA, Canada, Colombia,Turkey, Japan, Russia, Ukraine, India, Australia

Windows 11, Windows 10, Windows
Vista, Windows 7 (Service Pack 1),
Windows 8, interface USB 2.0

Standards, tests
EU Directive
4 included in delivery

2004 / 108 / EC(4) optional

testo 890

Overview of variants

testo 890

testo 890 Set

640 x 480 pixels

Thermal sensitivity (NETD)< 40 mK

Measuring range-30 to +650 °C

Image refresh rate

33 Hz*

SuperResolution

4425° x 19° Lens

(4)

(4)

Telephoto lens 15° x 11° *****(4)

4

Super-telephoto lens 6.6° x 5° *****(4)

4Auto focus

44High temperature measurement up to 1,200 °C

(4)

(4)

Panorama image assistant

44

SiteRecognition(measurement site recognition with image management)

4

4

Laser marker**

44

Display of surface moisture distribution (by manual input)

44

Humidity measurement with wireless humidity probe***(automatic measurement value transfer in real time)

(4)

(4)

HDMI interface

44

FeverDetection

(4)

(4)

Process analysis package: image sequence capturing in
instrument and fully radiometric video measurement

(4)

(4)

Voice recording using the headset****4

4

Solar mode

4

4

Lens protection glass(4)4Additional battery(4)4

Fast battery charger(4)40981 8854/msp/I/02.2024

Infrared resolution

4 included in delivery(4) optional******

inside the EU, outside 9 Hz
excepting USA, China and Japan
Wireless humidity probes only in the EU, Norway,
Switzerland, USA, Canada, Colombia, Turkey,Brazil, Chile, Mexico, New Zealand, Indonesia**** Bluetooth only in the EU, Norway, Switzerland,USA, Canada, Colombia, Turkey, Japan, Russia,Ukraine, India, Australia***** depending on the selected set

www.testo.com

Subject to change without notice.

Features