SQLC-215-109

INSTRUCTION MANUALPower Line Super Multi-Meter
SQLC-110L[ 1φ2W / 1φ3W / 3φ3W Analog Output ]Hardware Model GSQLC-215-109

Introduction
Thank you for your purchase of our product.
Read this instruction manual carefully before installation, wiring, and using this product.
Keep this instruction manual handy for reference at any time.
Have a contact with us or sales agent in case that this instruction manual is lost or damaged.
<Caution>
Have a contact with us when you have any questions or are aware of missing article.
Safety precaution
In the operation manual, the harm to a user or other persons and damage of property were prevented, andthe important matter for using correctly safely is written in it.
Please read a text well after understanding the following contents (a display and a graphic symbol), andprotect the mentioned item.
Improper use may lead to death or severe injury.
Improper use may possibly lead to death or severe injury.
Improper use may lead to medium injury.
 We are not responsible for the damage caused by following condition (earthquake / fire which is not caused byus, action by third party, other accident, damage caused by our customer, misuse, product usage under abnormal
condition).
 We are not responsible for secondary damage caused by product use / product malfunction(loss of profit, haltof business operation). We are also not responsible for damage caused by false operation in combination withconnecting equipment which is beyond our control.

 Do not disassemble, remodel and repair this product.
If judged as trouble, please contact our company or a distributing agent.
 Do not get this product wet to prevent generation of heat and ignition, product failure.
When this product gets wet, stop using it.
 Metal (wire etc.) other than wiring is not connected to the terminal (metallic portion) of this product.
 Do not work, if combustibles, and an inflammable medicine and inflammable gas are in the circumference.

 Connect specified power supply.
Connecting power supply beyond specification causes fire and product failure.
 When dust has adhered to the terminal, remove dust after turning off the power.
 Follow the below-mentioned procedure when abnormality (fuming / bad odor) happens.
(1) Stop a power supply and input, and stop using.
(2) Contact our company or a distributing agent.

 Do not use this product in a environment of high temperature / high humidity to prevent any damage.
 Do not touch the terminal during operation to prevent electric shock.
 Do not pull and bend connecting cable with force. Cable damage causes heat generation / burn and contact
failure leads to equipment damage.
 Connection or inspection of apparatus are not performed by the wet hand.

1

SQLC-215-109

Cautions on safety

Cautions on safe use
1. Working and storage environments
Don’t mount or store this unit in the following environments. If the unit becomes defective due to the use in anenvironment other than specified, it may be repaired for pay even during its warranty period (one year after thedate of delivery).
① Don’t mount or store the unit at a place where the ambient temperature is other than a range of –10 to +55℃ or
the relative humidity is higher than 85% RH.
② Don’t mount or store the unit at a place where a corrosive gas such as SO2, H2S, etc. is generated.
Corrosive gas = Sulfur dioxide SO2 / Hydrogen sulfide H2S / etc.
Large current bus / saturable reactor / etc.
③ Don’t mount or store the unit at a dusty place.
④ Don’t mount or store the unit at a place subjected to noticeable vibrations or shocks.
⑤ Don’t mount or store the unit at a place subjected to noticeable external noises.
⑥ If this unit directly measures an inverter output of cycle control, SCR phase angle control or PWM control, anerror may increase due to its operation principle.
2. Cautions on use in an outdoor panel
Be careful with the following items when using this unit in an outdoor panel.
① Don’t mount the unit at a place where it is directly exposed to rain or water drops, otherwise this unit may
become defective because of no water-proof or drip-proof structure.
② Don’t mount the unit at a dusty place.
③ Don’t mount the unit at a place exposed to direct sunlight. Avoid exposing the unit to direct sunlight eventhrough a glass window.
If the meter is directly exposed to sunlight, the surface temperature of the meter rises and the case may be
deformed if the temperature exceeds 80℃.
④ If the average temperature around the meter exceeds 40℃, the life of the unit may shorten.
3. About dew condensation
If the temperature and humidity of an installation change rapidly when a product is a non-energization, thewaterdrop by dew condensation may adhere to a display inner side. (The display filter and the LCD surface stickand the pattern of the shape of a circle or an ellipse occur.) This phenomenon is not trouble.
It will disappear, if a control power supply is applied and about 2 hour passes.
4. Mounting and wiring
Mount and connect the unit by a technician while referring to the instruction manual and observing the following
cautions.
 Connect the unit after confirming the connection diagram. An improper connection may cause troubles
such as the generation of a high voltage on the secondary side of its current transformer or burningof the unit or the occurrence of a fire.
 Don’t perform any connection in a hot line without turning off the power supply in advance, otherwise
an electric shock accident, troubles or burning of the unit, a fire, gas explosion, or other very
dangerous accidents may occur.
 The terminal cover is mounted for preventing an electric shock accident. Mount the terminal cover
without fail after the end of work.

2

SQLC-215-109

5. SettingThis unit requires setting and confirmation of the measuring range, etc. before use.
Wrong setting, if any, causes malfunction of the unit. If setting should be wrong, neither measurement nor output
becomes correct. Carefully read the instruction manual before setting the unit.
 Default setting
The default setting values are as specified below at the delivery time. Set them according to the working
conditions. The input circuit of this product is the common use of 3-phase 3-wire (3φ3W), single-phase 2-wire(1φ2W), and single-phase 3-wire (1φ3W). In case an input circuit is designated at the case of an order, it isshipped by the default value of the designated input circuit. And, in case it does not do designation of the input
circuit (with no designation), it is shipped by the default value of 3-phase 3-wire.
The unit will be delivered with your specified setting values, if so specified.
① Voltage, current input (1/2)
№

1

2

3

Setting item
Pattern
Main monitor
Sub-monitor (Left)
Display combination
Sub-monitor (Center)
Sub-monitor (Right)
Bar graph
FactorReset form
Alarm 1
Contact delay time
Alarm
Test
outputFactor１( )Reset form
Alarm 2
Contact delay time
Test
Upper limit value
Demandcurrent Interval
Upper limit value
DemandInterval
Demanddetection
activeOperation form
power

Current

4

Harmonic
detection

VoltagePower-factor operation form
Distortion-factor upper limit
5th conversion content rate upper
limit
Factorn-th content rate
Upper limit
Distortion-factor upper limit
5th conversion content rate upper
limit
Factorn-th content rate
Upper limit

5th conversion detection characteristics

5

6

7

Average value time limit
Instant measurement Voltage upper limit value
detection
Voltage lower limit value
Rated sensitivity current
Factor switching
Leakage detection(１)
Circuit switching
Use ZCT selection
ActionBacklight
Brightness

3-phase 3-wire
110V input 220V input
Pattern 1
A(L2)
V(L1L2)
W
Wh
A(L2)
DA
AUTO
0 second－DA
AUTO
0 second－
80.0A
0 secondOFF
0 second
Operating system
according withbimetallic type.
Instant measurement
OFF

Single-phase 3-wirePattern 1
A(L1)
V(L1N)
W
Wh
A(L1)
DA
AUTO
0 second－DA
AUTO
0 second－
400A
0 secondOFF
0 second
Operating system
according withbimetallic type.
Instant measurement
OFF

Single-phase
110V input 220V input
Pattern 1
A
V
W
Wh
A
DA
AUTO
0 second－DA
AUTO
0 second－
40.0A
0 secondOFF
0 second
Operating system
according withbimetallic type.
Instant measurement
OFF

OFF

OFF

OFF

5th
OFF
OFF

5th
OFF
OFF

5th
OFF
OFF

OFF

OFF

OFF

5th
OFF
Inverse-time-delaymode
0 minute
OFF
OFF
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）

5th
OFF
Inverse-time-delaymode
0 minute
OFF
OFF
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）

5th
OFF
Inverse-time-delaymode
0 minute
OFF
OFF
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）

Note(１) A setting item is not displayed in case there is no corresponding option.

3

SQLC-215-109① Voltage, current input (2/2)
№

8

9

10

11

12

13
14
15

16

Setting item
Voltage range
Current range
Current display peculiar
sensitivity
Active power polarity
Active power range
Reactive power range

3-phase 3-wireSingle-phase 3-wire
110V input 220V input
6600V
220V
110.0V
100.0A
500A

Single-phase
110V input 220V input
3300V
220V
50.0A

100.0A

500A

50.0A

One-side swing
100.0kW50.0kvar
LEAD 0.500‐1.000‐LAG 0.500
45.0‐65.0Hz
A(L1)
V(L1N)
W
cosφ
100.0％
100.0％
50.0％
OFFWh
1kWh/pWh
1kWh/p
Alarm resetMax./Min. reset
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
1φ3W (L1-N-L3)
300V
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

One-side swing
150.0kW10.00kW75.0kvar
5.00kvar
LEAD 0.500‐1.000‐LAG 0.500
45.0‐65.0Hz
A
V
W
cosφ
100.0％
100.0％
50.0％
OFFWh
1kWh/p0.1kWh/pWh
1kWh/p0.1kWh/p
Alarm resetMax./Min. reset
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
ON
1φ2W
110V
220V
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

One-side swing
1200kW
40.0kW
600kvar20.00kvar
LEAD 0.500‐1.000
Power-factor range‐LAG 0.500
Frequency range
45.0‐65.0Hz
Output factor 1
A(L2)
Output factor 2
V(L1L2)
Output factor 3
W
Output factor 4cosφ
Analog output(２)
Current
100.0％
Output peculiar
Active power
100.0％
sensitivity
Reactive power 50.0％
Low input cut
OFFFactorWh
Output 1
Pulse unit
10kWh/p0.1kWh/p
Pulse output(２)
FactorWh
Output 2
Pulse unit
10kWh/p0.1kWh/p
Alarm reset
External operation Input 1 function
input (２)Input 2 function
Max./Min. reset
VoltageON
Current
ON
Active power
ON
Reactive power
ON
Power-factor
ON
FrequencyON
Measurement display
Watt-hour of power receiving
ON
ON/OFF
Watt-hour of power transmission ON
var-hour of power receivingON
var-hour of power transmission
ON
Harmonic current
ON
Harmonic voltage
ON
Current leakage
(２)
ON
3φ3W
Phase line change (４)Input circuit
Input voltage
(５)
110V
220V
Dead band
0.0％
Measurement
Tidal current measurement
General measurement
Analog output specification(２)(３)
1‐5V
Bias adjustment
0.0％
Output 1
Span adjustment
100.0％
Bias adjustment
0.0％
Output 2
Analog output
Span adjustment
100.0％
adjustment
Bias adjustment
0.0％(２)
Output 3
Span adjustment
100.0％
Bias adjustment
0.0％
Output 4
Span adjustment
100.0％
Measurement range

Note(２) A setting item is not displayed in case there is no corresponding option.
The external operation input constitutes default value with an alarm-output option.
It becomes the next function in case there is no alarm-output option.
Input 1 function：Max. / Min. reset, Input 2 function：Measurement factor change.
Note(３) A setting item is not displayed if analog output is except DC0‐5V (or DC1‐5V) specification,
And analog output insulation product does not display a setting item.
Note(４) When the setting of phase line change of an input circuit is changed, it will return to the default valueof phase line which all set value changed.
Note(５) When phase line change setting of an input circuit is set as 3φ3W (or 1φ2W) and the input voltage
setting is changed, the voltage range returns to the default value of the phase line.
(For example：In case of 3φ3W, 6600V at the case of 110V setting, 300V at the case of 220V setting.）

4

SQLC-215-109② Current input
№

1

2

3

4

Setting item
Pattern
Main monitor
Sub-monitor (Left)
Display combination
Sub-monitor (Center)
Sub-monitor (Right)
Bar graph
FactorReset form
Alarm 1
Contact delay time
Alarm
Test
outputFactor６( )Reset form
Alarm 2
Contact delay time
Test
Upper limit value
DemandDemanddetection current IntervalDistortion-factor upper limit
5th conversion content rate upper
Current limit
FactorHarmonic
n-th content rate
detection
Upper limit
5th conversion detection characteristics

5

6
7

8

9
10
11
12
13

14

Average value time limit
Rated sensitivity current
Factor switching
Leakage detection(６)
Circuit switching
Use ZCT selection
ActionBacklight
Brightness
Current range
Measurement range
Current display peculiar
sensitivity
Output factor 1
Output factor 2
Output factor 3
Analog output
Output factor 4(６)
Output peculiar
Current
sensitivity
Low input cut
External operation Input 1 function
input (６)Input 2 function
Current
Measurement display
Harmonic current
ON/OFF
Current leakage (６)Input circuit
Phase line change (８)Dead band
Measurement
Tidal current measurement
Analog output specification(６)(７)
Bias adjustment
Output 1
Span adjustment
Bias adjustment
Output 2
Analog output
Span adjustment
adjustment
Bias adjustment６( )Output 3
Span adjustment
Bias adjustment
Output 4
Span adjustment

3-phase 3-wirePattern 15
A(L2)
A(L1)
A(L3)
－
A(L2)
DA
AUTO
0 second－DA
AUTO
0 second－
80.0A
0 second
OFF

Single-phase 3-wirePattern 15
A(L1)
A(L3)
A(N)
－
A(L1)
DA
AUTO
0 second－DA
AUTO
0 second－
400A
0 second
OFF

Single-phase
Pattern 15
A
－
－
－
A
DA
AUTO
0 second－DA
AUTO
0 second－
40.0A
0 second
OFF

OFF

OFF

OFF

5th
OFF
Inverse-time-delaymode
0 minute
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）
100.0A

5th
OFF
Inverse-time-delaymode
0 minute
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）
500A

5th
OFF
Inverse-time-delaymode
0 minute
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）
50.0A

100.0A

500A

50.0A

A(L2)
A(L1)
A(L3)
－

A(L1)
A(L3)
A(N)
－

A
－
－
－

100.0％

100.0％

100.0％OFF
Alarm resetMax./Min. reset
ON
ON
ON
3φ3W
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％OFF
Alarm resetMax./Min. reset
ON
ON
ON
1φ3W (L1-N-L3)
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％OFF
Alarm resetMax./Min. reset
ON
ON
ON
1φ2W
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

Note(６) A setting item is not displayed in case there is no corresponding option.
The external operation input constitutes default value with an alarm-output option.
It becomes the next function in case there is no alarm-output option.
Input 1 function：Max. / Min. reset, Input 2 function：Measurement factor change.
Note(７) A setting item is not displayed if analog output is except DC0-5V (or DC1-5V) specification,
And analog output insulation product does not display a setting item.
Note(８) When the setting of phase line change of an input circuit is changed, it will return to the default valueof phase line which all set value changed.

5

SQLC-215-109③ Voltage input
№

1

2

3

4

5

6
7

8

9

10

11
12
15

16

Setting item

3-phase 3-wire
110V input 220V input
Pattern 16
V(L1L2)V(L2L3)V(L3L1)Hz
V(L1L2)V
AUTO
0 second－V
AUTO
0 second－
OFF

Pattern
Main monitor
Sub-monitor (Left)
Display combination
Sub-monitor (Center)
Sub-monitor (Right)
Bar graph
FactorReset form
Alarm 1
Contact delay time
Alarm
Test
outputFactor９( )Reset form
Alarm 2
Contact delay time
Test
Distortion-factor upper limit
5th conversion content rate upperOFF
Voltage limit
Factor5thHarmonic
n-th content rate
detection
Upper limit
OFF
Inverse-time-delay
5th conversion detection characteristics
mode
Average value time limit
0 minute
OFF
Instant measurement Voltage upper limit value
detection
Voltage lower limit value
OFF
Rated sensitivity current
0.1A
Factor switching
Io
Leakage detection(９)
Circuit switching
1 phase earthing
Use ZCT selection
Type 0
ActionAUTO
Backlight
Brightness
3（Middle）
Voltage range
6600V
220V
Measurement range
Frequency range
45.0‐65.0Hz
Output factor 1
V(L1L2)
Output factor 2
V(L2L3)
Analog output
Output factor 3V(L3L1)(９)Output factor 4
Hz
Low input cut
OFF
Alarm reset
External operation Input 1 function
input (９)Input 2 function
Max./Min. reset
VoltageON
ON
Measurement display Frequency
ON/OFF
Harmonic voltage
ON
Current leakage
(９)
ON
3φ3W
Phase line change (11)Input circuit
Input voltage
(12)
110V
220V
Dead band
0.0％
Measurement
Tidal current measurement
General measurement
Analog output specification(９)(10)
1‐5V
Bias adjustment
0.0％
Output 1
Span adjustment
100.0％
Bias adjustment
0.0％
Output 2
Analog output
Span adjustment
100.0％
adjustment
Bias adjustment
0.0％(９)Output 3
Span adjustment
100.0％
Bias adjustment
0.0％
Output 4
Span adjustment
100.0％

Single-phase 3-wirePattern 16
V(L1N)
V(L3N)
V(L1L3)Hz
V(L1N)V
AUTO
0 second－V
AUTO
0 second－
OFF

Single-phase
110V input 220V input
Pattern 16
V
－
－
Hz
V
V
AUTO
0 second－V
AUTO
0 second
－
OFF

OFF

OFF

5th
OFF
Inverse-time-delaymode
0 minute
OFF
OFF
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）
110.0V
45.0‐65.0Hz
V(L1N)
V(L3N)
V(L1L3)Hz
OFF
Alarm resetMax./Min. reset
ON
ON
ON
ON
1φ3W (L1-N-L3)
300V
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

5th
OFF
Inverse-time-delaymode
0 minute
OFF
OFF
0.1A
Io
1 phase earthing
Type 0AUTO
3（Middle）
3300V
220V
45.0‐65.0Hz
V
－
－
Hz
OFF
Alarm resetMax./Min. reset
ON
ON
ON
ON
1φ2W
110V
220V
0.0％
General measurement
1‐5V
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％
0.0％
100.0％

Note(９) A setting item is not displayed in case there is no corresponding option.
The external operation input constitutes default value with an alarm-output option.
It becomes the next function in case there is no alarm-output option. Input 1 function：Max. / Min. reset,
Input 2 function：Measurement factor change.
Note(10) A setting item is not displayed if analog output is except DC0-5V (or DC1-5V) specification,
And analog output insulation product does not display a setting item.
Note(11) When the setting of phase line change of an input circuit is changed, it will return to the default valueof phase line which all set value changed.
Note(12) When phase line change setting of an input circuit is set as 3φ3W (or 1φ2W) and the input voltage setting
is changed, the voltage range returns to the default value of the phase line.
(For example：In case of 3φ3W, 6600V at the case of 110V setting, 300V at the case of 220V setting.）
6

SQLC-215-109

6. Operation
Be careful with the following cautions during use.
 Use the input within the rated range. Be careful since negligence of this caution may cause troubles of the unit.
 There is a function to hold the maximum value and the minimum value with a measurement factor in this product.
A blackout is guaranteed and this value isn't also cleared by a power supply reset. However, the minimum value
may be updated in case input is not applied to a power up. For this reason, in order to make the past minimum
value hold by powering on, please apply input within 1 second after switching on a power supply.
 The maximum value, a minimum value measurement factor
Measurement factor
Voltage，Current，Demand current，Active power，Demand active power，Reactive power，Power factor，Frequency
Current leakage ，Harmonic (Distortion factor ，Effective value ，Content)

Maximum value
measurement

Minimum value
measurement

○

○

○

×

 Be careful not to touch any terminal when power is applied to the unit.
 Don’t disassemble or modify this unit without any previous permission of our company, otherwise thewarranty does not apply to the unit any more. Also, modifications may cause troubles, a fire, orother accidents. For specifications change, etc., please contact us.
7. Maintenance and check① Wipe off the surface of the unit with a dry soft cloth. The liquid crystal display (LCD) may light during cleaningon the LCD face. However, this phenomenon is caused by the static electricity that may be produced in the filter,and it does not show any trouble. Leave the unit as it is for a while, and the display goes out due to natural discharge.
Don’t press the LCD face strongly, otherwise it may be broken.
When the filter has been pressed, it may touch the LCD face to stain the LCD face. However, this phenomenon does
not show any trouble, but it is caused by a change of the ambient environment or the like.
The LCD face may be restored to its original condition after a while during the use as it is.
② Check the following items.
◎ Check the unit for damage in appearance.
◎ Check if indications conform to the inputs.
◎ Check if the unit is mounted normally without any loose connections of the terminal board after turning
off the power supply.
If a question has arisen or if the unit seems to be defective, please contact us.
8. Other precautions
 Wipe off dirt on the surface with dry cloth softly. Keep in mind that strong rubbing of nameplate leads tocharacter disappearance. Organic solvent is not appropriate for cleaning.
 Neither mercury parts nor the nickel-cadmium battery is being used for this product.
 This product is disposed as an industrial waste (non-inflammable).
9. Warranty
 Warranty period
It is for one year after product delivery.
 Warranty scope
In the state of the normal use of product-specification within the range according to this instruction manual,
the trouble within the warranty period performs exchange or repair gratuitously.
However, the shipping expenses and the packing cost in the case of shipping obtain as payment on a customer.
And, if it corresponds to the next, it does not warrant.
(1) If it breaks down when converted or repaired except our company.
(2) If it breaks down by use outside the specification range.
(3) If the cause of trouble is based on cause other than this product.
(4) Transportation, movement, damage by falling, and trouble.
(5) A natural disaster, disaster, etc., if it is the trouble which is irresponsible for a payment side(our company or distributing agent).
Only a product simplex is applied with a warranty. We cannot bear the responsibility such as the damage and theloss that occurred by the use of this product and trouble or the loss of profits.
And, a warranty is effective only in Japan.

7

SQLC-215-109

Content

1. Product outline ･･･････････････････････････････････････････････････････････････ 9
1.1 Usage of product ･･･････････････････････････････････････････････････････････ 9
1.2 Features of product ････････････････････････････････････････････････････････ 9
2. The name and function of each part ････････････････････････････････････････････ 10
3. Preparation
3.1 Installation ･･･････････････････････････････････････････････････････････････ 10
3.2 Connections ････････････････････････････････････････････････････････････････ 12
4. Operation ･････････････････････････････････････････････････････････････････････ 17
4.1 The screen change and function by switch operation ･････････････････････････ 18
4.2 The kind of display ････････････････････････････････････････････････････････ 19
4.2.1 Measurement display ･････････････････････････････････････････････････････ 19
4.2.2 Alarm detection display ･････････････････････････････････････････････････ 22
4.2.3 Setting display ･････････････････････････････････････････････････････････ 24
4.3 Operation ･･････････････････････････････････････････････････････････････････ 25
4.3.1 The main monitor display-element change ･････････････････････････････････ 25
4.3.2 Phase (line) display change ･････････････････････････････････････････････ 25
4.3.3 Harmonic measurement display change ･････････････････････････････････････ 26
4.3.4 Watt-hour enlarged display ･･････････････････････････････････････････････ 26
4.3.5 Setting value check ･････････････････････････････････････････････････････ 27
4.3.6 Setting mode ････････････････････････････････････････････････････････････ 28
4.3.7 Reset ･･･････････････････････････････････････････････････････････････････ 29
5. Setting
5.1 Function table ･････････････････････････････････････････････････････････････ 32
5.2 Setting table ･･････････････････････････････････････････････････････････････ 37
5.3 Setting in detail explanation ･･････････････････････････････････････････････ 46
5.3.1 Setting mode 1 ･･････････････････････････････････････････････････････････ 46
5.3.2 Setting mode 2 ･･････････････････････････････････････････････････････････ 58
5.3.3 Setting mode 3 ･･････････････････････････････････････････････････････････ 65
6. Specification
6.1 Specification and intrinsic error ･･････････････････････････････････････････ 69
6.2 Specification, Performance. ････････････････････････････････････････････････ 73
6.3 Option ･････････････････････････････････････････････････････････････････････ 75
7. Maintenance and check7.1 Trouble shooting ･･･････････････････････････････････････････････････････････ 77
7.2 Check ･･････････････････････････････････････････････････････････････････････ 77
7.3 Maintenance ････････････････････････････････････････････････････････････････ 77
7.4 Test ･･･････････････････････････････････････････････････････････････････････ 78
7.5 Storage ････････････････････････････････････････････････････････････････････ 78
7.6 Countermeasures against troubles ･･･････････････････････････････････････････ 78
Appendix table 1 to 3

8

SQLC-215-109

1. Product outline
1.1 Usage of productThis single unit can measure and monitor demand-current ×3, voltage ×3, current ×3, demand active power,
active-power, reactive-power, power-factor, frequency, watt-hour, var-hour, harmonic(voltage, current) andcurrent leakage.
From a low tension circuit to a high tension circuit, it is adapted for various usages, such as a measurementmonitor of a power-receiving circuit, an energy conservation power monitor, a demand current measurement monitor,
and a harmonic monitor, a leakage monitor.
The intensive monitor united with the system is made in an addendum of analog output and a pulse output.

1.2 Features of product A 3-phase 3-wire, single-phase, and single-phase 3-wire common type can be prepared. It's possible to share stock.
 Bar graph 1 measurement and digital 4 measurement are displayed simultaneously.
 Analog output 4 circuit and contact-output 2 circuit can be taken out. (Option)
A contact output can be selected from a pulse output, an alarm output, and a CPU error output. (Please designate
it at an order.) And, An output factor can be selected by setting.
 Two external operation inputs are possible. (Option)
And, selection of reset input and a display change input is possible at setting.
 Power supply is AC85‐264V，DC80‐143V (for both AC and DC uses).
 The mounting method of this unit is compatible with the mounting method of conventional 110 square mechanical meter.
This unit is mounted at 2 diagonal points.
 Integrated value of Wh and varh can indicate expansion to 3rd digit below the decimal point.
 Analog output is with a lower limit limiter.
 A tidal current measurement (output 2 quadrant) change is possible for var and cosφ.
 Current-leakage measurement corresponds to both Io and Igr.
 A backlight function is equipped. Selection of backlight-on, backlight-off, and auto backlight-off and settingof brightness are possible. Automatic turning off the lights at the time of non-operation can be established.
LED：White.
 LCD can be chosen from 2 kinds, the type to see from the top and the type to see from the bottom.
(Please designate it at an order.) The product with analog output can be selected from between analog output insulation type and the analog output
non-insulation type. (Please specify at the time of an order.)9

SQLC-215-109

2. The name and function of each partBar graph display
The measurement value of the main monitor
is indicated by the analog.
(Setting which does bar graph displayof the measurement value of sub-monitor
is also possible.)

Scale markings
Scale markings is automatically set in ameasurement range.
Upper limit (or lower limit) setting index
An upper limit (or lower limit) set point
is displayed.

SQLC-110L

Digital display
Measurement monitor can watch
4 elements at the same time.

Unit display
Unit display is automatically set in ameasurement range.

･Main monitor･Sub-monitor (Right)･Sub-monitor (Center)･Sub-monitor (Left)

Multiplying factor display
Multiplying factor is displayed at the timeof an Wh and varh display.

SETDISPLAY

The switch from which integrated
value of the amount of electric
power is switched to normal display(5 digits of integer) and expansion
indication (integer 2 digits ＋ below
decimal point, 3 digits) variously.
If it is not operated for 10 minutes after adisplay change, it will usually return to adisplay. It is used also as a switch whichchanges to setting mode. If it continues
pushing 3 seconds or more, it will change tosetting mode.
In setting mode, it is used as a switch that
determines a set point.

This is a switch which changes the phase(line) display of current (voltage).
If it is not operated for 10 minutes after adisplay change, it returns to the original
phase (line) display. In setting mode, it isused as a switch that terminates setting mode.

－＋

The switch to which measurement displays
element of main monitor is changed.
If it is not operated for 10 minutes after a display
change, it returns to the original measurement display
factor. In set mode, it is used as a switch thatchanges a setting value.

RESET/SHIFT

MAX/MIN

This switch resets various alarms.
And, in the maximum and the minimum measurement
display, it is used as a switch that resets the
maximum and the minimum value.
In setting mode, it is used as a switch to which
a setting item is shifted.

The switch to which general measurement display (usually)
and the maximum minimum measurement display are changed.
MODE

The switch to which general measurement display (usually) and theharmonic measurement (voltage and current) display are changed.
In set mode, it is used as a switch that changes a setting item.

3. Preparation
3.1 Installation
Mount the unit by the attached M5 nuts to a panel of thinner than 10mm, referring to the following external
dimensions drawing and panel cutout. Fasten these nuts with tightening torque 2.75 to 3.82N･m.
● Dimension diagram(1) Multi-meter

10

SQLC-215-109(2) Correspondence ZCT (Our company recommendation) ZCT is not attached with leakage measurement.
50A

OTG-LA21 (OMRON Corporation)

100A OTG-LA30 (OMRON Corporation)

200A OTG-LA42 (OMRON Corporation)

4
2400A OTG-LA68 (OMRON Corporation)

kt

k

l

600A OTG-LA82 (OMRON Corporation)

kt

lt

k

l

lt

100A(For outdoor) OTG-LA30W (OMRON Corporation)＊ Installation pitch

This product also corresponds to the next ZCT.
TAIWA ELECTRIC INDUSTRIES CO.,LTD.：Zero-phase current
transformer for low voltage (Through-type) TYPE：ZB-□
Hitachi, Ltd.：Through-type ZCT ZR series

TYPE：ZR-□

Please consult, in case you use other ZCT.

● Caution on handling
Mount the LCD to obtain an optimum angle, since the contrast changes according to the monitoring angle.
(1) For upper case installation(2) For lower case installation

11

SQLC-215-109● Installation(1) A product is put in a cut hole of a panel from a front.
A body is inserted until it exceeds retaining stopperof the lower base.
(2) Please fix a product certainly with attached M5 flange
nut for installation. Please give a tightening
torque as 2.75 to 3.82N･m.

3.2 Connections
Please perform connection after referring to the following wiring diagram.
● Connection drawing (16)(1) 1φ2W(2) 1φ3W
L1 N L3

L1 L2

VT

CT

LOAD

AUX.SUPPLY

AUX.SUPPLY
1
2

8
9

17

12
13

23

19
21

5
6
7

25
27
29

16
18
20
22
24
26

LOAD

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15
CONTACT OUTPUT2 ( )

1
2

8
9

17

12
13
14
15

23

19
21

5
6
7

28

( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2(3) 3φ3W (2VT，2CT)(4) 3φ3W (2VT，3CT)

L1 L2 L3 VT

25
27
29

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15
CONTACT OUTPUT2 ( )28

L1 L2 L3 VT

CT

CT

AUX.SUPPLY
LOAD

1
2

AUX.SUPPLY
8
9

17
19
21

5
6
7

( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2

12
13
14
15

23
25
27
29

16
18
20
22
24
26

LOAD

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15( )CONTACT OUTPUT2

1
2
3
4
5
6
7

28

( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2

12

8
9

17
19
21

12
13
14
15

23
25
27
29

16
18
20
22
24
26
28

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15( )CONTACT OUTPUT2SQLC-215-109(5) Current input 3φ3W (2CT)(6) Current input 3φ3W (3CT)

L1 L2 L3

L1 L2 L3

CT

CT

AUX.SUPPLYLOAD

AUX.SUPPLY
17

1
2

19
21

5
6
7

12
13
14
15

23
25
27
29

16
18
20
22
24
26

LOAD

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15
CONTACT OUTPUT2 ( )

1
2
3
4
5
6
7

28

( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2(7) Current input 1φ3W

(8) Current input 1φ2W

L1 N L3

17
19
21

12
13
14
15

23
25
27
29

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15
CONTACT OUTPUT2 ( )

28

L1 L2

CT

CT

LOAD
AUX.SUPPLY

AUX.SUPPLY

LOAD
17

1
2

19
21

5
6
7

12
13
14
15

23
25
27
29

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15( )CONTACT OUTPUT228( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2(9) Voltage input 3φ3W

(10) Voltage input 1φ3W

L1 L2 L3 VT

12
13
14
15

23

8
9
10

17

25
27
29

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15( )CONTACT OUTPUT228

L1 N L3

AUX.SUPPLY

AUX.SUPPLY

LOAD
1
2

8
9
10

17
19
21
23

5
6
7

25
27
29

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15
CONTACT OUTPUT2 ( )

1
2

5
6
7

28

19
21
23
25
27
29

( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15
CONTACT OUTPUT2 ( )28( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2(11) Voltage input 1φ2W
L1 L2

19
21

5
6
7

( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2

LOAD

17

1
2

VT

LOAD
AUX.SUPPLY
1
2

8
9

17
19
21

5
6
7

23
25
27
29

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 13 )ANALOG OUTPUT4CONTACT OUTPUT1 15
CONTACT OUTPUT2 ( )28( 13 ) EXT.INPUT1( 14 ) EXT.INPUT2

Note(13) Analog output, contact output, external operation
input is an option.
Note(14) Can change to external reset function or
external display change function by setting.
Note(15) Can choose the contact output among the pulse output,
the alarm output, the CPU error
output. (Designation)
・Combination of contact output
Pulse＋Alarm
Alarm×2
Pulse×2
Pulse＋CPU error
Alarm＋CPU error

Contact output 1
Pulse output
Alarm output 1
Pulse output 1
Pulse output
Alarm output

Contact output 2
Alarm output
Alarm output 2
Pulse output 2
CPU error output
CPU error output

Note(16) In case of low-voltage circuit, secondary sidegrounding of VT and CT is unnecessary. And, VT is
unnecessary in case it used 110V or direct 220V.

13

SQLC-215-109● Schematics in the leakage monitor of low-voltage circuit (20)(22)(1) 1φ2W (N-phase earth)(2) 1φ3W (N-phase earth)

L1 N

L1 N L3
21

21

CT

CT

AUX.SUPPLY

AUX.SUPPLY
17
19
21

ZCT

23
25
27
29

LOAD

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 17 )ANALOG OUTPUT4CONTACT OUTPUT1 19
CONTACT OUTPUT2 ( )

17
19
21

ZCT

23
25
27

28

29

LOAD

( 17 ) EXT.INPUT1( 18 ) EXT.INPUT2

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 17 )ANALOG OUTPUT4CONTACT OUTPUT1 19
CONTACT OUTPUT2 ( )28( 17 ) EXT.INPUT1( 18 ) EXT.INPUT2(3) 3φ3W (L2-phase earth)(4) 3φ3W (No earth)

L1 L2 L3

L1 L2 L3
21
21

EVT

CT

CT

AUX.SUPPLY

AUX.SUPPLY
17

17

19
21

ZCT

23
25
27
29

LOAD

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 17 )ANALOG OUTPUT4CONTACT OUTPUT1 19
CONTACT OUTPUT2 ( )

17

17

19
21

ZCT

23
25
27

28

LOAD

29

16
18
20
22
24
26

ANALOG OUTPUT1ANALOG OUTPUT2ANALOG OUTPUT3( 17 )ANALOG OUTPUT4CONTACT OUTPUT1 19
CONTACT OUTPUT2 ( )28( 17 ) EXT.INPUT1( 18 ) EXT.INPUT2( 17 ) EXT.INPUT1( 18 ) EXT.INPUT2

Note(17) Analog output, contact output, external operation input is an option. Zero-phase current input productwill be only with leakage current measurement.
Note(18) Can change to external reset function or external display change function by setting.
Note(19) Can choose the contact output among the pulse output, the alarm output, the CPU error output. (Designation)
・Combination of contact output
Pulse＋Alarm
Alarm×2
Pulse×2
Pulse＋CPU error
Alarm＋CPU error

Contact output 1
Pulse output
Alarm output 1
Pulse output 1
Pulse output
Alarm output

Contact output 2
Alarm output
Alarm output 2
Pulse output 2
CPU error output
CPU error output

Note(20) In case of low-voltage circuit, secondary side earthing of VT and CT is unnecessary. And, VT is unnecessary
in case it used 110V or direct 220V.
Note(21) Voltage input is needed at the time of Igr measurement in a short circuit electric current.
Note(22) A voltage input and a leakage input are not insulated. Please use it combining ZCT.

14

SQLC-215-109● In case of installing ZCT to ground wire by leakage monitor.
L1 N L3 (1φ3W, N-phase grounding)L1 L2 L3 (3φ3W, L2-phase grounding)

LOAD
L
ZCT
K

3
4

<Caution> When doing Igr measurement by a leakage monitor,
please be careful about polarity of ZCT.

● Caution on connection(1) Mount the terminal cover without fail for safety after the end of connections.
(2) Separate the input wiring and output wiring from each other without fail, and take a preventive measure against
malfunction due to external noises.
(3) Connect the grounding terminal E (No. 7 terminal) to the ground without fail for enhancing the shield effect.
Keep the grounding resistance between the grounding terminal and the ground to be lower than 100Ω.
(4) Keep a distance of more than 30cm between this unit and the circuit breaker as well as between this unit and therelay contact signal line.
(5) No protection is necessary for this unit even if the transmission line may be affected by an induced lightning
surge or the like when transmitting an analog output to the receiver. Mount an about 500V arrester or the likebetween the line surge protector and the ground as well as between the transmission line and the ground on thereceiver side for the purpose of protecting the devices on the receiver side.
(6) It is recommended to mount a surge killer outside when connecting an inductive load to the pulse output and alarm
output. If no surge killer is mounted, the contact life may shorten.
(7) Please do not earth the output terminal of ZCT.
(8) Please shorten wiring for this product from secondary of ZCT as much as possible.
becomes close to other large current circuits, please use shielding wire.

And, in case secondary wiring(9) In case it measures a leakage current Igr, it is necessary to measure correctly the phase angle of a voltage input
and a zero phase-current input. Wiring of ZCT (primary, secondary) and wiring for this product should check
schematics.

15

SQLC-215-109● The distinction method of the primary side polarity of ZCT(1) Product of OMRON Corporation
The direction (near side) which can read correctly the character of the name plate currently stuck on ZCT is "K".

(2) Product of TAIWA ELECTRIC INDUSTRIES CO.,LTD.
① ZB-30M，ZB-58M
A side with " k " and " l " of the terminal for secondary wiring connection is "L".

② ZB-70M
The " 検 " mark side of ZCT is "L".

③ ZB-90M
It is printed by ZCT.
(3) Product of Hitachi, Ltd.
① ZR-30B，ZR-58B
A side with " k " and " l " of the terminal for secondary wiring connection is "L".

② ZR-65，ZR-80
The name plate side of ZCT is "K".

16

SQLC-215-109
4. Operation● The function of switch

SwitchFunction

SET

The integrated value of electric energy is changed to the usual display and an enlarged display.
If it continues pushing 3 seconds or more, it will change to setting mode.
In setting mode, it is used for the determination of a set point.

RESET/SHIFT

Various kinds of alarms are reset. The maximum value and the minimum value are reset in the
maximum minimum measurement display. In setting mode, it is used for movement of a setting item.
The usual general measurement display and harmonic measurement (voltage, current) display arechanged. In setting mode, it is used for the change of a setting item.

MODE
MAX/MIN

The usual measurement display and maximum value or minimum value display are changed.

＋ ，－

The measurement display element of the main monitor is changed.
In setting mode, it is used for change of a set point.

DISPLAY

A phase (between lines) display of current (voltage) is changed. It is used in case it terminatessetting mode. And, it is used in case it returns the display combination of a measurement factor.

● Convenient functions(1) In case a measurement change or a phase change is performed and the original screen composition is not clear
anymore, DISPLAY is pushed for more than 3 seconds or it's no-operation for 10 minutes and returns to original
screen structure.
(2) Even if it stops operation with setting mode, it returns to the display mode in 10 minutes.
● 7 segment displays
This product shows the guidance in various setting using 7 segment displays besides a display of a measurement
value. A digital readout and 7 segment displays corresponding to each alphabet are shown in the following.

A

B(b)

C

D(d)

E

F

G

H

I

J

K

L

M

Y(y)

Z

Non-dis Non-dis
play
play

N(n)

O(o)

P

Q(q)

R(r)

S

T(t)

U(u)

V

W

X
Non-dis
play

0

1

2

3

4

5

6

17

7

8

9

SQLC-215-109
4.1 The screen change and function by switch operation
This product changes various screens by switch operation. Here, the change step of the screen by switch operation is
explained.
Display mode
DISPLAY

Instant measurement display
Generalmeasurement
display
MODE

Current harmonicmeasurement
displaySETRESET/SHIFT

Electric-energy
integrated-value
enlarged display

Push
1 seconds

SET
or no operation,
10 minute
SET

All alarm
reset

No operation,
10 minute＋Bar graphdisplay factor
change，－No operation,
10 minute

Main monitor
display-element change

RESET/
＋ －
SHIFT

MODE

Voltage harmonicmeasurement
display

Phase (line)
display change

SET

Set point check

Bar graphdisplay factor
changeDISPLAY or
No operation,
10 seconds

MODE

SET
Push 3 seconds
MAX/MIN
Maximum and the minimum
measurement display

General measurement
maximum and minimum
display

Setting mode 1
RESET/SHIFT
Push 1 seconds

MODE

RESET/＋ ＋
SHIFT

Current harmonicmeasurement maximum
display

Push 1 seconds

Maximum / Minimum
reset(Display factor)

DISPLAY or
No operation,
10 minute
SET ＋

RESET/SHIFT

Push 3 seconds
Setting mode 2
Maximum / Minimum
reset(All factor)

MODE

DISPLAY or
No operation,
10 minute
SET ＋ DISPLAY
Push 3 seconds

Voltage harmonicmeasurement maximum
display

Setting mode 3
DISPLAY or
No operation,
10 minute

MODE

18

SQLC-215-109
4.2 The kind of display
4.2.1 Measurement display
A measurement value display has the three following types of displays.
The change of the measurement display element of the main monitor by switch operation and the change of thephase / line display of current / voltage is possible (temporarily).
In a general measurement display, if switch operation is not performed for 10 minutes after changing a display
element, it returns to the original measurement display element automatically.
① General measurement display
Measurement factors, such as current, voltage,
and active power, are displayed.
The measurement value of four factors is
displayed at the maximum.
Setting which always displays a measurement
factor is possible. And, it is possible tochange to a display of the maximum value and theminimum value, about the measurement factor
which performs holding of the maximum value andthe minimum value by switch operation.
These maximum values and the minimum value arereset by switch operation (it updates to theinstantaneous value at the time).
In addition, as for the maximum value and theminimum value, power-supply reset is not cleared
either. And, this display is held by even after
10 minutes of switch non-operation.

Range of minimum to maximum
is displayed in bar graph.

A(L2) display

V(L1L2)
displayWdisplayWhdisplay

General measurement display

Minimum
display

Instant
display

Maximum
display

General measurement maximum
and minimum display

・The example of a measurement display of each measurement factor (Main monitor) ･･･ Harmonic measurement is excluded.
Measurement
factorExample of display

Measurement
factorNote

Voltage

Current
leakage

Example of display

Note

Current“LEAK” isdisplay

Watt-hour(Powerreceiving)

Demandcurrent“DEMAND” isdisplay

Watt-hour“－” isdisplay(Powertransmission)

Demand active
power

Active power

19“DEMAND” isdisplaySQLC-215-109

Measurement
factorExample of display

Measurement
factorNote

Example of display

Note

Reactive
power

LAG
or
LEAD
display

Power factor

LAG
or
LEAD
display

var-hour(Powerreceiving，
LAG)

“LAG”
display

var-hour(Powerreceiving，
LEAD)“LEAD”
display

var-hour(Powertransmission,
LAG)“LAG” and”－”display

var-hour(Powertransmission,
LEAD)“LEAD” and ”－” display

Frequency

・ About a current-leakage measurement display
The current leakage of this product can select a factor from the next two
types.
The system by which Io where an electric current for the capacity by a earth
capacitance (Ic) and an electric current for the resistance by insulation
degradation (Igr) are synthesis is measured.
The method which measures only the resistance part current (Igr) by insulation
degradation etc.
By the Igr method, a current leakage is calculated from a voltage input anda zero phase-current input. In case of 3-phase 3-wire (1-phase earthing),
in the range whose phase angle (leading-phase angle of the zero-phase current
by reference voltage) of these inputs is 150 to 350°,
it will become the outside of a measuring range and a measurement displayof a zero-phase current will be "----".

The measurement display besides
an Igr measuring range② Current harmonic measurement display
It is a measurement factor display of the distortion factor of current, relative harmonic content, harmonic effective
value, etc. The measurement value of three factors is displayed at the maximum.
About sub-monitor (center) and sub-monitor (right), it is possible to change to the factor that always indicates bymeasurement. And, it is possible to change to a display of the maximum value by switch operation.
These maximum values can be reset by switch operation (it updates to the instantaneous value at the time).
In addition, as for the maximum value, even power-supply reset is not cleared. And, a display is held, even after
elapsing for 10 minutes without operating a switch.
Measurement display element)
Main monitor：Distortion factor
Sub-monitor (Left) ：5th conversion or harmonic order (n)Sub-monitor (Center)：Harmonic 5th conversion content or harmonic nth content (n=3,4,5,7,9,11,13,15)
Sub-monitor (Right) ：Fundamental-wave effective value , 5th harmonic conversion effective value or harmonic
nth effective value (n=3,4,5,7,9,11,13,15)

20

SQLC-215-109③ Voltage harmonic measurement display
It is a measurement factor display of the distortion factor of voltage, relative harmonic content, harmonic
effective value, etc. About a function, it is the same as a current harmonic measurement display.

Distortion-factor display

Effective-value display

Distortion-factor maximum
display

Content display

Content maximum
display

Order display

Order display

Current harmonic measurement display.

Distortion-factor display

Effective-value maximum
display

Current harmonic measurement maximum display.

Effective-value display

Distortion-factor maximum
display

Content display

Content maximum
display

Order display

Order display

Voltage harmonic measurement display.

Effective-value maximum
display

Voltage harmonic measurement maximum display.

21

SQLC-215-109

4.2.2 Alarm detection display
The alarm value setting is a possible measurement factor (demand current and harmonic, etc.), it displays in casean input exceeds a set point.
Besides the usual measurement display, the detected factor is displayed on a screen upper case.
In addition, in case setting OFF (not use) as measurement factor, it does not detect.
And in the case of with an alarm-output option, it is possible to do an alarm output (relay make contact) to the outside
simultaneously with a screen display.
Alarm display possible factor) Demand current, Demand, Harmonic distortion factor (current, voltage),
Harmonic 5th conversion content (current, voltage),
Harmonic nth content (current, voltage n＝3, 4, 5, 7, 9, 11, 13, 15),
Voltage, Current leakage
Alarm detection(Demand current upper limit )A(L2) display
Alarm
detection
W display

V(L1L2) display
Wh display
General measurement display.

General measurement display (Alarm detection)

・The example of a display at the case of the detection in each alarm factor.
In case the alarm factor is indicating by measurement at the main monitor or the sub-monitor, a measurement value
constitutes a blinking display.
The displays after an alarm return.
In case a return method is automatic reset setting：It returns to the usual measurement display.
In case a return method is manual reset setting：A detection display and an alarm output hold (in case setting as an alarm output of applicable factor).
The return in this case needs alarm reset operation. Please refer to "4.3.7 Reset" about alarm reset.
Alarm factor

Demandcurrent

Upper
limit
(H)

Example of a display

Alarm factor

Detection display
Alarm setting value

Voltageharmonic
5th
conversion
content

Example of a display

Upper
limit
(H)
Alarm setting
Value

Detection
display

Setting as 5th conversion content ofbar graph (It distinguishes in anunderbar)

22

SQLC-215-109

Alarm factor

Demand

Upper
limit
(H)

Example of a display

Alarm factor

Detection display
Alarm setting value

Voltageharmonic
nthcontent

Example of a display

Upper
limit
(H)

Alarm setting
Value

Detection
display

Setting as 11th content of bar graph(Distinguishes in an underbar)

Current
leakage

Voltagedistortion
factor

Upper
limit
(H)

Upper
limit
(H)

Detection display
Rated sensitivity
current value
Detection display
Alarm setting Value

Current
harmonic
5th
conversion content

Upper
limit
(H)

Voltage

Upper
limit
(H)

Upper
limit
(H)

Voltage

Alarm setting
valueUpper limit (H)alarm set point

Lower limit (L) alarm set point

Setting as 5th conversion content
of bar graph (Distinguishes in anunderbar)

Current
harmonic
nthcontent

Detection
Display

Lower
limit
(L)

Detection
DisplayUpper limit (H)alarm set point

Lower limit (L) alarm set point

Detection
display

Setting as 3rd content of bar graph(Distinguishes in an underbar)

23

SQLC-215-109

Alarm factor

Current
distortion factor

Example of a display

Upper
limit
(H)

Detection display
Alarm setting value

4.2.3 Setting display
The display of various setting. There are three types of setting modes according to the contents of a setting.
Refer to "5. Setting" for the operation in setting mode, and the detailed contents of a setting.
① Setting mode 1
Setting of a measurement display element, an alarm
output, and an alarm value is mainly performed.
And, an alarm output can be tested in this settingmode.

New setting
Current setting
Setting No.

Setting mode 1 (No.111 Display pattern)② Setting mode 2
Setting of measurement range, analog output, pulse
output, and measurement display ON/OFF is mainly
performed.
And, initialization of a setting value and reset ofwatt-hour integrated value can be performed in thissetting mode.

New setting
Current setting
Setting No.
Setting factor
Setting mode 2 (No.211 Voltage range)③ Setting mode 3
Setting of an input circuit and tidal current
measurement is mainly performed.
And, analog output can be adjusted in this settingmode.

New setting
Current setting
Setting No.

Setting mode 3(No.311 Input circuit phase line change)

24

SQLC-215-109

4.3 Operation
4.3.1 The main monitor display-element change
The measurement display element of the main monitor is changed. A change is performed by ＋ －.
This operation can be performed also except a general measurement display (harmonic measurement display, maximum
display and minimum display). However, in a harmonic measurement display, a sub-monitor changes with a harmonicorder. (The main monitor is distortion-factor fixation.)
After changing a measurement display element, if a switch is not operated for 10 minutes, it will return to theoriginal measurement display element automatically.
In a harmonic measurement display and maximum display and minimum display, even if a switch is not operated for
10 minutes, it does not return to the original display.
Setting can perform same operation in external operation input. Please refer to "5.3.2 Setting mode 2, (4)external operation input setting" about the setting method. Please refer to "6.3 Option specification" about
external operation input.
A(L2) display

No operation 10 minute

－

－

＋

＋

＋

＋

－

－

V(L1L2)
display
Main monitor A(L1)W
displayWhdisplay

Main monitor A(L2)(Default display)

Main monitor A(L3)4.3.2 Phase (line) display change
A phase (line) display of current or voltage is changed. (Everything which is being indicated)
A change is performed by DISPLAY . This operation can be performed also except a general measurement display(harmonic measurement display, maximum display and minimum display).
In addition, after changing a phase (line) display, if a switch is not operated for 10 minutes, it will returnto the original phase (line) display automatically.
In a harmonic measurement display and maximum display and minimum display, even if a switch is not operated for
10 minutes, it does not return to the original display.
Setting can perform same operation in external operation input. Please refer to "5.3.2 Setting mode 2, (4)external operation input setting" about the setting method. Please refer to "6.3 Option specification" about
external operation input.
A(L2) display

No operation 10 minute

DISPLAY

V(L1L2)
display

DISPLAYDISPLAYWh display

W display

A(L2)，V(L1L2) (Default display)

A(L3)，V(L2L3)

25

A(L1)，V(L3L1)

SQLC-215-109

4.3.3 Harmonic measurement display change
A general measurement display and a harmonic measurement display are changed. A change is performed by MODE .
Whenever it pushes a switch, it changes as follows. General measurement display → Current harmonic measurement
display → Voltage harmonic measurement display → General measurement display ･･･This operation can also perform the maximum display and minimum display.
In that case, it changes in order of a next. General measurement maximum and the minimum display → Current harmonicmeasurement maximum display → Voltage harmonic measurement maximum display → General measurement maximum and theminimum display ･･･
In addition, about this operation, even if it does not do switch operation for 10 minutes, it does not return to theoriginal display.
Distortion-factor
displayA(L2) display

Distortion-factor
display

MODE

V(L1L2)
display

Orderdisplay

Wh display
W display

General measurement display.

MODE

MODE

Content
display

Effective-value
display

Orderdisplay

Content
display

Effective-value
display

Current harmonic measurement display. Voltage harmonic measurement display.

4.3.4 Watt-hour enlarged display
In case electric energy is being displayed by the general measurement display, an electric-energy display is usuallychanged to a display (5 digits of integers), and an enlarged display (2 digits integer + below decimal point, 3 digits).
A change is performed by SET .
After an enlarged display, if a switch is not operated for 10 minutes, it returns to the usual display automatically.
<Caution > If it continues pushing SET 3 seconds or more, it will become the setting mode 1. (An electric-energy
display does not change)
A(L2) display

No operation 10 minute

SET

SET

W display
V(L1L2) display

Watt-hour enlarged display

Wh display

General display

Enlarged display

26

SQLC-215-109

4.3.5 Setting value check
A voltage range (VT ratio), a current range (CT ratio), and an alarm-output set point are checked.
Check is RESET/SHIFT and － are pushed simultaneously and performed.
The change of a set point is carries out by ＋ and － .
This operation can be performed also except a general measurement display (harmonic measurement display, maximum
display and minimum display). DISPLAY is pushed in case it returns to the original measurement display.
And, if a switch is not operated for 10 seconds after a set point check, it will return to the original measurement
display automatically.
A(L2) display

No operation
10 minute

VT ratioRESET/
SHIFT

＋ －

＋

－

DISPLAY

W display
V(L1L2) display
Wh display

Factor display

General display＋＋
Currentrange(CT ratio)

Input-rating
voltage display

Set point check
Voltage range (VT ratio)－－

Alarm output 2
Time interval，Contact delay
time

＋

－

－
Alarm output 2
factor，
Return method＋27

Alarm output 1
Time interval，Contact delay
time＋

Alarm output 1
factor ，Return method－

SQLC-215-109

4.3.6 Setting mode
Various kinds of setting are performed. Setting mode is three types, and operations are different.
DISPLAY is pushed in case it returns to the original measurement display. And, if a switch is not operated
for 10 minute after a set point check, it will return to the original measurement display automatically.
Operation and the contents of setting (detail) in setting mode, please refer to "5 Setting".
・Setting mode 1：Press SET for longer than 3 seconds.
・Setting mode 2：Press SET and RESET/SHIFT together for longer than 3 seconds.
・Setting mode 3：Press SET and DISPLAY together for longer than 3 seconds.
<Reference> This operation can be performed also except a general measurement display (harmonic measurement display,
maximum display and minimum display).
A(L2) display

Setting No.

New setting

Current setting

SET 3 seconds
SET ＋ RESET/SHIFT 3 seconds

SET ＋ DISPLAY 3 seconds

V(L1L2) display

Setting mode 1

Wh display

W display

Setting factor
Current setting
New setting

Measurement display

DISPLAY
Return tomeasurement
display
Setting No.
Setting mode 2
Setting No.

Current setting

New setting

Setting mode 3

28

SQLC-215-109

4.3.7 Reset
Various kinds of reset are performed. The kind of reset is as follows and operations are different, respectively.
Reset of watt-hour integrated value (zero clear)，
Reset of maximum value and minimum value (it updates to the instantaneous value at the time)，
Alarm-output reset (OFF of an alarm output (at the case of manual reset setting)).
And, the operation from each measurement display constitutes conditions at each reset.
(1) Watt-hour integrated-value reset
It resets by package about the integrated value of various watt-hour. Watt-hour reset is performed in the setting
mode 2. In detail explanation in the setting mode 2, please refer to "5.3.2 Setting mode 2".
① Press SET and RESET/SHIFT together for longer than 3 seconds. It goes into the setting mode 2.
② MODE is pushed until setting is set to No.271. Further RESET/SHIFT is pushed once and it is made awatt-hour-reset display.
③ Press SET for longer than 3 seconds.
④ DISPLAY is pushed and it returns to a measurement display.
A(L2) display

New setting

SET ＋ RESET/SHIFT
3 second

V(L1L2) display

W display

Setting factor

Wh display

Measurement display

New setting

Current setting

No.211MODE

DISPLAYRESET/SHIFT×ｎ(To setting No.272)

Setting No.

SET 3 second
Wh and varh
display byturns.

Wh and varh
display by turns.

Setting No.

It displays by
clear completion.
No.272（Clear completion）

No.272（Before clear）

Setting mode 2

29

SQLC-215-109(2) Reset of maximum value and minimum value
Reset of the various measurement values of maximum value and minimum value is performed.
This reset has two types of methods. (How to perform according to a measurement factor individual.
How to reset all maximum values and minimum values by package.)a) Individual reset
Reset of only a certain differential maximum value or the minimum value is performed.
Other maximum values and minimum values are not reset by this operation.
① A measurement factor to reset is displayed. (General measurement maximum value, minimum value, orCurrent, voltage harmonic measurement maximum display)② Press RESET/SHIFT for longer than 1 seconds.
<Caution> An alarm output will be reset if this operation is performed by instant measurement display.
Please be sure to perform this operation after displaying the maximum value and a minimum value
measurement factor to make it reset.
Instant display

The range of minimum to maximum isdisplayed in bar graph.

RESET/SHIFT
1 second

Minimum display

Maximum display

Maximum, Minimum measurement display(General measurement maximum and minimum display)

Minimum display(Reset)Instant display

Maximum display(Reset)Reset completionb) Package reset
Reset of all the maximum values and minimum value is performed.
In addition, setting can perform same operation in external operation input.
Please refer to "5.3.2 Setting mode 2 (4) external operation input setting" for the setting method,
Please refer to "6.3 Option" for the external operation input,① Press RESET/SHIFT and ＋ together for longer than 1 seconds. By the maximum and the minimum measurement
display (The general measurement maximum and minimum display or current, voltage harmonic measurement maximum
display).
The range of minimum to maximum isdisplayed in bar graph.
Instant display

RESET/SHIFT ＋ ＋
1 second

Minimum display

Maximum display

Maximum, Minimum measurement display(General measurement maximum and minimum display)

Minimum display(Reset)Instant display

Maximum display(Reset)Reset completion<Caution> In the general measurement maximum and minimum value reset, all the factors of general measurement
are reset by package. (The harmonic measurement maximum value is not reset.)
By harmonic measurement maximum value reset, a current factor and a voltage factor are reset by package.
(The general measurement maximum value and the minimum value are not reset.)
30

SQLC-215-109(3) Alarm reset
In case an alarm return method is set to “HOLD (manual return)”, an alarm output is reset (output OFF).
(With an alarm-output option)
However, an output is not turned off by this operation, in case an alarm continues and it has caused.
And, this operation is unnecessary in case setting as "AUTO (automatic return)” in alarm return method. (By which
an output is also OFF according to an alarm return.)
In case the number of alarm outputs is two, both outputs are reset (output OFF) by this operation. (Return operationthat comes out individually cannot be performed.)
In addition, setting can perform same operation in external operation input.
Please refer to "5.3.2 Setting mode 2 (4) external operation input setting" for the setting method,
Please refer to "6.3 Option" for the external operation input,① By instant measurement display (a general measurement display or current, voltage harmonic measurement
display), RESET/SHIFT is pushed 1 second or more.
<Caution> If this operation is performed by the maximum and the minimum measurement display, the maximum value
and minimum value of the measurement factor currently displayed will be reset.
Please be sure to perform this operation in the state of an instant measurement display.
Alarm detection (Hold)RESET/SHIFT
1 second

Instant measurement display(General measurement display)

Reset completion

31

SQLC-215-109

5. Setting< Caution >When changing the input circuit setting, please be sure to perform a setup from an input circuit setting inthe setting mode 3. After changing the other setting, when the input circuit setting is changed the set value
returns to default value (default value of a changed input circuit).

5.1 Function table
This product has each function setting with a front switch.
<Caution> In case the input circuit is not designated at the case of an order, it is shipping in the default valueof 3-phase 3-wire.
Setting mode 1. Function table (1)Setting
No.

Function

Functional description

Current Voltage
input input

111

Display pattern

Set the display combination patternof the digital 4 displays and bar
graph display.

○

○

112

Main monitor

Set the display factor of digital main
monitor.

○

○

113

Sub-monitor(Left)

Set the display factor of digital
sub-monitor (left).

○

○

114

Sub-monitor(Center)

Set the display factor of digital
sub-monitor (center).

○

○

115

Sub-monitor(Right)

Set the display factor of digital
sub-monitor (right).

○

116

Bar graph

Set the display factor of bar graph.

121AL
(23)

Alarm 1 factor

Default setting

○

47-51

○

47-51

○

47-51

W

○

47-51

○

Wh

○

47-51

○

○

3φ3W
1φ3W
1φ2W

○

47-51

Set the output factor of alarm 1.

○

○

DA

○

52

Alarm 1 return
methodAlarm 1 mask time(Contact delaytime)

Set the output action at the caseof reset of alarm 1.

○○

Automatic reset52

Set the contact delay time of alarm 1.

○○

0 second

52

124AL
(23)

Alarm 1 test

Output test of alarm 1 is performed.

○

○

－

52

125AL
(23)

Alarm 2 factor

Set the output factor of alarm 2.

○

○

DA

Alarm 2 return
methodAlarm 2 mask time(Contact delaytime)

Set the output action at the caseof reset of alarm 2.

○○

Automatic reset52

Set the contact delay time of alarm 2.

○○

0 second

52

128AL
(23)

Alarm 2 test

Output test of alarm 2 is performed.

○

○

－

52

131H

Demand current
upper limit

Sets the high-alarm value of demandcurrent.

○

80%(Full scale＝
100%)

132

Demand current timeinterval

Set time interval of demand current.

○

0 second

133H

Demand active power
upper limit

Set the high-alarm value of demand
active power.

122AL
(23)
123AL
(23)

126AL
(23)
127AL
(23)

Pattern 1

ImportantPage
setting

3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

A(L2)
A(L1)
A

OFF (No operation)

Note(23) A setting item is not displayed in case there is no corresponding option.

32

A(L2)
A(L1)
A
V(L1L2)V(L1N)
V

○

52

53,54
○

53,54
53,54

SQLC-215-109

Setting mode 1. Function table (2)Setting
No.

Function

Functional description

134

Demand active power
time interval

Set the time interval of demand active
power.

135

Demand active power
operating method

Set the operating method of demand
active power.

Power-factor
operating method
Current
distortion-factor
upper limit
Current 5th
conversion content
upper limit
Current n-thcontent factor

Set the operating method of
power-factor measurement.

136
141H

142H
143
144H
145H

146H
147
148H
149
14A
151H
152L
161
(24)

Current Voltage
input input

Default setting
0 second
Operating system
according withbimetallic type.
Instant
measurement

ImportantPage
setting
○

53,54

○

53,54
53,54

Set the high-alarm value of current
distortion factor.

○

OFF (No operation)

54,55

Set the high-alarm value of current 5th
conversion content.

○

OFF (No operation)

54,55

Set the order of n-th current content.

○

5th

54,55

Current n-thcontent upper limit
Voltagedistortion-factor
upper limit
Voltage 5th
conversion content
upper limit
Voltage n-thcontent factor

Set the high-alarm value of current
n-th content.

○

OFF (No operation)

54,55

Set this high-alarm value of voltage
distortion factor.

○

OFF (No operation)

54,55

Set the high-alarm value of voltage 5th
conversion content.

○

OFF (No operation)

54,55

Set the order of n-th voltage content.

○

5th

54,55

Voltage n-thcontent upper limit
5th conversion
detection
characteristics
Average value timeinterval

Set the high-alarm value of voltage
n-th content.

○

OFF (No operation)

54,55

Instant measurement
voltage upper limit
Instant measurement
voltage lower limit
Leakage detection
rated sensitivity
current value

Set the detection characteristic of 5thconversion content.

○○

Inverse-timedelay mode

54,55

Set the average value detection timeinterval of harmonic.

○○

0 minute

54,55

Set the high-alarm value of instantvoltage.

○OFF (No operation)

55

Set the low-alarm value of instantvoltage.

○OFF (No operation)

55

○

○

0.1A

56

○

○

Io

56

○

○

One phase earth

56

○

○

TYPE 056

Set the rated sensitivity current valueof current leakage.

164
(24)

Leakagedetect-circuitry
changeSelection of ZCTused

Set the factor of the current
leakage that performs leakage
detection.
Set the circuit configuration at thetime that 3-phase 3-wire performs
leakage detection.
Sets the ZCT used by current-leakage
measurement.

171

Backlight action

Set the ON/OFF of backlight.

○○

AUTO OFF

57

172

Backlight
brightness

Set the brightness of backlight.

○○3（Middle）

57

Leakage detection
162(24)(25) factor change
163
(24)

Note(24) A setting item is not displayed in case there is no corresponding option.
Note(25) Current input product are only Io detection. Igr detection cannot be performed.

33

SQLC-215-109

Setting mode 2. Function table (1)Setting
No.

Function

Functional description

Current Voltage
input input

211

Voltage range

Set the voltage-measurement range(VT ratio).

212

Current range

Set the current-measurement range(CT ratio).

○213

Current display
peculiar
sensitivity

Set the full scale of current meter.

○

214

Active power
polarity

Set the swing display of active power
meter.

215

Active power range

Set the full scale of active power
meter.

216

Reactive power
range

Set the full scale of reactive power
meter.

217

Power-factor range

218

Frequency range

221A
(26)
222A
(26)
223A
(26)
224A
(26)
225A
(26)
226A
(26)
227A
(26)
228A
(26)

○

3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

Set the full scale of power-factor
meter. And, set this output rangeof analog output.
Set the full scale of frequency meter.
And, set this output range of analog
output.

○AO1 output factor○○

AO2 output factor

Set the output factor of AO2 (analog
output 2).

○

○

○

○

○

○

AO4 output factor
Current output
peculiar
sensitivity
Active power-output
peculiar
sensitivity
Reactive power
output peculiar
sensitivity
Low input cut

Set the output factor of AO3 (analog
output 3).
Set the output factor of AO4 (analog
output 4).
Set the analog output sensitivity(％ of output for input) of current.
Set the analog output sensitivity(％ of output for input) of active
power.
Set the analog output sensitivity(％ of output for input) of reactive
power.
Set the function which makes a lower
limit the output at the case of aminute input (adequate to 0.5％ orless) in analog output.

3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

○

58-60

○

58-60

58-60

○

○

○

58-60

1200kW (28)100.0kW150.0kW (28)600kvar (28)50.0kvar75.0kvar(28)

58-60

58-60

0.500‐1.000‐0.500

58-60

45.0‐65.0Hz

58-60

3φ3W
1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

A(L2)
A(L1)
A
V(L1L2)V(L1N)
V

○

61,62

○

61,62

W

○

61,62

cosφ

○

61,62

100.0％

61,62

100.0％

61,62

50.0％

61,62

OFF (No operation)

61,62

Note(26) A setting item is not displayed in case there is no corresponding option.
Note(27) In 220V input, it becomes "220V".
Note(28) In 220V input, It becomes "40.0kW" and "20.00kvar" (3φ3W).
In 220V input, It becomes "10.00kW" and "5.00kvar" (1φ2W).

34

6600V (27)110.0V
3300V (27)100.0A
500A
50.0A
100.0A
500A
50.0A

ImportantPage
setting

One-side swing

Set the output factor of AO1 (analog
output 1).

AO3 output factor

Default setting

SQLC-215-109

Setting mode 2. Function table (2)Setting
No.

241P
(29)

FunctionPO1 factor

Functional description
Set the output factor of PO1 (pulse
output 1).

242P
PO1 pulse unit
(29)(30)

Set the output pulse unit of PO1(pulse output 1).

243P
(29)

Set the output factor of PO2 (pulse
output 2).

PO2 factor

244P
PO2 pulse unit
( )(30)29

Current Voltage
input input

Default setting
Wh

○

3φ3W

10kWh/p

1φ3W
1φ2W

1kWh/pWh

Set the output pulse unit of PO2(pulse output 2).

ImportantPage
setting

62

○

3φ3W

10kWh/p

1φ3W
1φ2W

1kWh/p

62

62

62

251
(29)

External operation
input 1 function

Set the function of the external
operation input 1.

○○

Alarm reset

○

63

252
(29)

External operation
input 2 function

Set the function of the external
operation input 2.

○○

Maximum / Minimum
reset

○

63

261

Voltage ON/OFF

Set the ON/OFF of voltage measurement
display.

○

ON

64

262

Current ON/OFF

Set the ON/OFF of current measurement
display.

ON

64

263

Active power ON/OFF

Set the ON/OFF of active power
measurement display.

ON

64

264

Reactive power
ON/OFF

Set the ON/OFF of reactive power
measurement display.

ON

64

265

Power-factor ON/OFF

Set the ON/OFF of power-factor
measurement display.

ON

64

266

Frequency ON/OFF

Set the ON/OFF of frequency measurement
display.

ON

64

267

Power-receiving
watt-hour ON/OFF

Set the ON/OFF of power-receiving
watt-hour measurement display.

ON

64

Power transmission
watt-hour ON/OFF
Power-receiving
var-hour (LAG,LEAD)
ON/OFF
Power transmission
var-hour (LAG,LEAD)
ON/OFF
Harmonic current
ON/OFF

Set the ON/OFF of power transmission
watt-hour measurement display.
Set the ON/OFF of power-receiving
var-hour (LAG,LEAD) measurement
display.
Set the ON/OFF of power transmission
var-hour (LAG,LEAD) measurement
display.
Set the ON/OFF of harmonic current
measurement display.

ON

64

ON

64

ON

64

ON

64

26C

Harmonic voltage
ON/OFF

Set the ON/OFF of harmonic voltage
measurement display.

○

ON

64

26D
(29)

Current leakage
ON/OFF

Set the ON/OFF of current-leakage
measurement display.

○

○

ON

64

271

set value
initialization

All set values are initialized.
(Return to default setting)

○

○

－

64

272

Watt-hour reset

Integrated value is cleared by package
about each electric energy.

－

64

268
269

26A
26B

○

○

○

Note(29) A setting item is not displayed in case there is no corresponding option.
Note(30) In 220V input, it becomes "0.1kWh/p". (3φ3W、1φ2W)

35

SQLC-215-109

Setting mode 3. Function table
Setting
No.

Function

311

Input circuit phaseline change

Functional description

Current Voltage
input input

Default setting
3φ3W

312

Input voltage

321

Measurement dead
band

322

Tidal current
measurement

Analog output
331(31)(32) specification

Set the input circuit or phase line.

○

Set the input voltage or phase-voltage
full scale.
Set the dead band of measurement
display.
Set the general measurement or tidal
current measurement which wasconscious of power transmission / powerreceiving, in measurement of reactive
power and power-factor.

○

○

○

○

1φ3W
1φ2W
3φ3W
1φ3W
1φ2W

ImportantPage
setting

3φ3W
1φ3W(L1-N-L3)
1φ2W
110V
300V
110V

○

66

○

66

0.0％

67

Generalmeasurement67

Set the analog output specification.

○○

DC1‐5V or DC0‐5V(Designation )

68

341
(31)

AO1 BIAS adjustment

Set the BIAS value of AO1 (Analog output
1).

○

○

0.0％

68

342
(31)

AO1 SPAN adjustment

Set the SPAN value of AO1 (Analog output
1).

○

○

100.0％

68

343
(31)

AO2 BIAS adjustment

Set the BIAS value of AO2 (Analog output
2).

○

○

0.0％

68

344
(31)

AO2 SPAN adjustment

Set the SPAN value of AO2 (Analog output
2).

○

○

100.0％

68

345
(31)

AO3 BIAS adjustment

Set the BIAS value of AO3 (Analog output
3).

○

○

0.0％

68

346
(31)

AO3 SPAN adjustment

Set the SPAN value of AO3 (Analog output
3).

○

○

100.0％

68

347
(31)

AO4 BIAS adjustment

Set the BIAS value of AO4 (Analog output
4).

○

○

0.0％

68

348
(31)

AO4 SPAN adjustment

Set the SPAN value of AO4 (Analog output
4).

○

○

100.0％

68

Note(31) A setting item is not displayed in case there is no corresponding option.
Note(32) A setting item is not displayed if analog output is except DC0‐5V (or DC1‐5V) specification,
And analog output insulation product does not display a setting item.

36

SQLC-215-109

5.2 Setting table
A setting item changes by the specification of a product, or the existence of an option.
(1) Important setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the measurement
range of voltmeter(211)

Press SET and RESET/SHIFT together for longer than 3 seconds(211)
Select a measurement range by ＋ and －Press SET
Selected measurement range is entered

Set the measurement
range of ammeter(212)

Press DISPLAY

58-60

Returns to display mode.

Press RESET/SHIFT
Press SET and RESET/SHIFT together for longer than 3 seconds
(211)
(212)
Select a measuring range by ＋ and －Press SET
Selected measuring range is entered

Set the display
combination(111)

Press DISPLAY

Page

58-60

Returns to display mode.

Select the display combination by ＋ and －Press SET for longer than 3 seconds(111)
Press SET
Selected display combination is entered
Press DISPLAY

47-51

Returns to display mode.
Set the output
factor of analog
output 1 (AO1).
(221A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Select an output factor by ＋ and －Press SET
Selected output factor is entered

Press DISPLAY

61,62

Returns to display mode.

Set the output
factor of analog
output 2 (AO2).
(222A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Select an output factor by ＋ and －Press SET(222A)
Selected output factor is entered
Press DISPLAY
Returns to display mode.

61,62

Set the output
factor of analog
output 3 (AO3).
(223A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Select an output factor by ＋ and －(222A)(223A)
Press SET
Selected output factor is entered
Press DISPLAY
Returns todisplay mode.

61,62

Set the output
factor of analog
output 4 (AO4).
(224A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)
Select an output factor by ＋ and －Press SET
Selected output factor is entered

Set the output
factor of pulse
output 1 (PO1).
(241P)

Set the output
factor of pulse
output 2 (PO2).
(243P)

Press DISPLAY

Returns to display mode.

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Select an output factor by ＋ and －Press SET
Press MODE
(241P)
Selected output factor is entered
Press DISPLAY
Returns to display mode.
Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press RESET/SHIFT
Press RESET/SHIFT(241P)(242P)(243P)
Select an output factor by ＋ and －Press SET
Selected output factor is entered

Press DISPLAY

37

61,62

Returns to display mode.

62

62

SQLC-215-109

Items
Set the factor ofalarm output 1.
(121AL)

Setting and operation procedures
Press MODE
Press SET for longer than 3 seconds(111)(121AL)
Select an output factor by ＋ and －Press SET
Selected factor is entered
Press DISPLAY

Set the factor ofalarm output 2.
(125AL)

Page

Returns to display mode.

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(123AL)(124AL)(125AL)
Select an factor by ＋ and －Press SET
Selected factor is entered
Press DISPLAY

52

52

Returns to display mode.

Set the external
operation input 1
function.
(251)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press MODE
Select an function by ＋ and －Press SET
(241P)(251)
Selected function is entered
Press DISPLAY
Returns to display mode.

63

Set the external
operation input 2function.
(252)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press MODE
Press RESET/SHIFT
Select an function by ＋ and －(241P)(251)(252)
Press SET
Selected function is entered
Press DISPLAY
Returns todisplay mode.

63

Set the time intervalof demand current.
(132)

Press MODE
Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(131H)
Select an time interval by ＋ and －Press SET
(132)
Selected time interval is entered
Press DISPLAY
Returns to display mode.

53,54

Set the time intervalof demand active
power.
(134)

Press MODE
Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Select an time interval by ＋ and －(132)(133H)(134)
Press SET
Selected time interval is entered
Press DISPLAY
Returns todisplay mode.

53,54

Set the operation
method of demand
active power.
(135)

Press MODE
Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(132)(133H)(134)(135)
Select an operation method by ＋ and －Press SET
Selected operation method is entered

Set the phase wire of
input circuit.
(311)

Returns to display mode.

Press SET and DISPLAY together for longer than 3 seconds(311)
Select an phase wire by ＋ and －Press SET
Selected phase wire is entered
Press DISPLAY

Set the input voltage(phase-voltage fullscale).
(312)

Press DISPLAY

66

Returns to display mode.

Press RESET/SHIFT
Press SET and DISPLAY together for longer than 3 seconds
(311)
(312)
Select an input voltage (In case of 1φ3W, it is phase-voltage full scale) by ＋ and －Press SET

53,54

Selected input voltage is entered

38

Press DISPLAY

Returns to display
mode.

66

SQLC-215-109(2) A combination except a display pattern.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the display
factor of main
monitor.
(112)

Set the display
factor of
sub-monitor (left).
(113)

Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
Select an display factor by ＋ and －Press SET
Selected display factor is entered

Press DISPLAY

47-51

Returns to display mode.

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Select an display factor by ＋ and －Press SET
Selected display factor is entered

Page

Press DISPLAY

47-51

Returns to display mode.

Set the display
factor of
sub-monitor(center).
(114)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Select an display factor by ＋ and －Press SET
Press RESET/SHIFT(114)
Selected display factor is entered
Press DISPLAY
Returns to display mode.

47-51

Set the display
factor of
sub-monitor (right).
(115)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Press RESET/SHIFT
Press RESET/SHIFT
Select an display factor by ＋ and －
(114)
(115)
Press SET
Selected display factor is entered
Press DISPLAY
Returns todisplay mode.

47-51

Set the display
factor of bar graph.
(116)

Press RESET/SHIFT
Press RESET/SHIFT
Press SET for longer than 3 seconds
(111)
(112)
(113)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(114)
(115)
(116)
Select an display factor by ＋ and －(If a sub-monitor is selected, an underbar will be 47-51
displayed on the bottom of a digital display.)
Press SET

Selected display factor is entered

Press DISPLAY

Returns todisplay mode.

(3) Setting of active power polarity and measurement range.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

Setting of
measurement rangeof active power.
(214), (215)

Page

Press RESET/SHIFT
Press SET and RESET/SHIFT together for longer than 3 seconds
(211)
(212)
Press RESET/SHIFT
Press RESET/SHIFT
(213)
(214)
Select an one-side(P)/both-side(-) deflection by ＋ and －Press SET
58-60
Selected deflection is entered

Press RESET/SHIFT(215)

Select a measuring range by ＋ and －Press SET

Selected measuring range is entered

Press DISPLAY

39

Returns to display mode.

SQLC-215-109(4) Setting of reactive power measurement range.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

Setting of
measurement range ofreactive power.
(216)

Press RESET/SHIFT
Press SET and RESET/SHIFT together for longer than 3 seconds
(211)
(212)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(213)
(214)
(215)
(216)
Select a measuring range by ＋ and －Press SET
Selected measuring range is entered

Press DISPLAY

Set the pulse unitof pulse output 2(Po2).
(244P)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press RESET/SHIFT
Select a output pulse unit by ＋ and －(241P)(242P)
Selected output pulse unit is entered
Press DISPLAY
Returns to
Press SET
display mode.
Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press MODE
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(241P)(242P)(243P)(244P)
Select a output pulse unit by ＋ and －Press SET
Selected output pulse unit is entered

Press DISPLAY

58-60

Returns to display mode.

(5) Setting of Wh (varh) output pulse unit.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the pulse unitof pulse output 1(Po1).
(242P)

Page

Page

62

62

Returns to display mode.

(6) Setting of active power-factor and frequency measurement range
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

Page

Set the measurement
range of active
power-factor.
(217)

Press RESET/SHIFT
Press SET and RESET/SHIFT together for longer than 3 seconds
(211)
(212)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(213)
(214)
(215)
(216)
Press RESET/SHIFT
Select a measuring range by ＋ and －Press SET
(217)
Selected measuring range is entered
Press DISPLAY
Returns to display mode.

Set the measurement
range of frequency.
(218)

Press RESET/SHIFT
Press SET and RESET/SHIFT together for longer than 3 seconds
(211)
(212)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
(213)
(214)
(215)
(216)
58-60
Press RESET/SHIFT
Press RESET/SHIFT
Select a measuring range by ＋ and －
(217)
(218)
Press SET
Selected measuring range is entered
Press DISPLAY
Returns todisplay mode.

(7) Setting of current display intrinsic sensitivity.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the display
sensitivity (％ ofa display to an
input) of current.
(213)

58-60

Page

Press RESET/SHIFT
Press SET and RESET/SHIFT together for longer than 3 seconds
(211)
(212)
Press RESET/SHIFT
Select a display sensitivity by ＋ and －Press SET
58-60
(213)
Selected display sensitivity is entered
Press DISPLAY
Returns to display mode.

40

SQLC-215-109(8) Setting of analog output.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the analog
output sensitivity(％ of an output toan input) of current.
(225A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
Select a output sensitivity by ＋ and －Press SET
Selected output sensitivity is entered

Set the analog
output sensitivity(％ of an output toan input) of active
power.
(226A)

Set the analog
output sensitivity(％ of an output toan input) of reactive
power.
(227A)

Press DISPLAY

Page

61,62

Returns to display mode.

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
Press RESET/SHIFT
Select a output sensitivity by ＋ and －Press SET
(226A)
Selected output sensitivity is entered
Press DISPLAY
Returns to display mode.
Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
Press RESET/SHIFT
Select a output sensitivity by ＋ and －Press RESET/SHIFT(226A)(227A)
Press SET
Selected output sensitivity is entered
Press DISPLAY

61,62

61,62

Returns to display mode.

Set the output cut
function at thecase in minute input(0.5％ or less).
(At the case ofanalog output)(228A)

Press MODE
Press SET and RESET/SHIFT together for longer than 3 seconds(211)(221A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(222A)(223A)(224A)(225A)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
61,62(226A)(227A)(228A)
Select a low Input cut ON/OFF by ＋ and －Press SET
Selected action is entered

Press DISPLAY

41

Returns to display mode.

SQLC-215-109(9) Setting of alarm output.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the return
method of alarm
output 1.
(122AL)Set the contact
delay time of alarm
output 1.
(123AL)Set the return
method of alarm
output 2.
(126AL)

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Select a return method by ＋ and －Press SETThe selected return method is entered

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(123AL)(124AL)(125AL)(126AL)
Press SETThe selected return method is entered
Select a return method by ＋ and －52

52

Returns to display mode.

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(123AL)(124AL)(125AL)(126AL)
Press RESET/SHIFT
Select an contact delay time by ＋ and －Press SET
(127AL)The selected contact delay time is entered
Press DISPLAY
Returns to display mode.

(10) Demand measurement (current, active power) setting.
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the high-alarm
value of demandcurrent.
(131H)

52

Returns to display mode.

Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(122AL)
Press RESET/SHIFT
Select an contact delay time by ＋ and －Press SET
(123AL)The selected contact delay time is entered
Press DISPLAY
Returns to display mode.

Press DISPLAY

Set the contact
delay time of alarm
output 2.
(127AL)

Press DISPLAY

Page

Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Press DISPLAY

52

Page

53,54

Returns to display mode.

Set the high-alarm
value of demand
active power.
(133H)

Press MODE
Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Select a high-alarm value by ＋ and －Press SET
(132)
(133H)
Selected high-alarm value is entered
Press DISPLAY
Returns to display mode.

53,54

Set the operation
method ofpower-factor
measurement.
(136)

Press MODE
Press MODE
Press RESET/SHIFT
Press SET for longer than 3 seconds(111)(121AL)(131H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(132)(133H)(134)(135)
Select a operation method by ＋ and －Press SET
(136)
Selected operation method is entered
Press DISPLAY
Returns to display mode.

53,54

42

SQLC-215-109(11) Harmonic measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the high-alarm
value of current
distortion factor.
(141H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Press DISPLAY

Page

54,55

Returns to display mode.

Set the high-alarm
value of 5th current
conversion content.
(142H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Select a high-alarm value by ＋ and －Press SET
(142H)
Selected high-alarm value is entered
Press DISPLAY
Returns to display mode.

54,55

Set the order ofn-th current
content.
(143)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Select a order by ＋ and －Press SET
(142H)(143)
Selected order is entered
Press DISPLAY
Returns to display mode.

54,55

Set the high-alarm
value of n-th
current content.
(144H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Set the high-alarm
value of voltage
distortion factor.
(145H)

Press DISPLAY

Returns to display mode.

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Select a high-alarm value by ＋ and －Press SET
Selected high-alarm value is entered

Press DISPLAY

54,55

54,55

Returns to display mode.

Set the high-alarm
value of voltage 5th
conversion content.
(146H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Select a high-alarm value by ＋ and －Press SET
(146H)
Selected high-alarm value is entered
Press DISPLAY
Returns to display mode.

54,55

Set the order ofn-th voltage
content.
(147)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Select a order by ＋ and －Press SET
(146H)(147)
Selected order is entered
Press DISPLAY
Returns to display mode.

54,55

Set the high-alarm
value of n-th
voltage content.
(148H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
54,55(146H)(147)(148H)
Select a factor by ＋ and －Press SET
Selected factor is entered
Press DISPLAY

Returns to display mode.

43

SQLC-215-109

Items

Setting and operation procedures

Set the detectedcharacteristics of
5th conversion
content.
(149)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(146H)(147)(148H)(149)
Select a detected characteristics by ＋ and －Press SET
Selected detected characteristics is entered

Set the averagetime interval.
(14A)

Press DISPLAY

Page

54,55

Returns to display mode.

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(142H)(143)(144H)(145H)
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT
Press RESET/SHIFT(146H)(147)(148H)(149)
Press RESET/SHIFT
Select a time interval by ＋ and －Press SET
(14A)
Selected time interval is entered
Press DISPLAY
Returns to display mode.

(12) Instant measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

54,55

Page

Set the high-alarm
value of instantvoltage.
(151H)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Select a high-alarm value by ＋ and －Press SET
(151H)
Selected high-alarm value is entered
Press DISPLAY
Returns to display mode.

55

Set the low-alarm
value of instantvoltage.
(152L)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press RESET/SHIFT
Select a low-alarm value by ＋ and －(151H)(152L)
Press SET
Selected low-alarm value is entered
Press DISPLAY
Returns todisplay mode.

55(13) Current-leakage measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the rated
sensitivity current
value of current
leakage.
(161)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Select a rated sensitivity current by ＋ and －(151H)(161)
Press SET
Selected rated sensitivity current is entered
Press DISPLAY

Page

56

Returns to display mode.

Set the detectedfactor of current
leakage.
(162)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Press RESET/SHIFT(151H)(161)(162)
Select a detected factor by ＋ and －Press SET
Selected detected factor is entered
Press DISPLAY

Returns to display mode.

44

56

SQLC-215-109

Items
Set the leakage
detected circuit
configuration.
(by 3-phase 3-wire)(163)

Setting and operation procedures
Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Press RESET/SHIFT
Press RESET/SHIFT(151H)
(161)
(162)
(163)
Select a circuit by ＋ and －Press SET
Selected circuit is entered
Press DISPLAY

Set the ZCT usedby current-leakage
measurement.
(164)

Page

Returns to display mode.

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Press RESET/SHIFT
Press RESET/SHIFT(151H)
(161)
(162)
(163)
Press RESET/SHIFT
Select a ZCT type by ＋ and －Press SET
(164)
Selected ZCT type is entered
Press DISPLAY
Returns to display mode.

(14) Backlight setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures

Set the action of
backlight.
(171)

Set the brightnessof backlight.
(172)

Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Press MODE
Select a backlight action by ＋ and －(151H)(161)(171)
Press SET
Selected backlight action is entered
Press DISPLAY
Returns todisplay mode.
Press MODE
Press MODE
Press MODE
Press SET for longer than 3 seconds(111)(121AL)(131H)(141H)
Press MODE
Press MODE
Press MODE
Press RESET/SHIFT(151H)
(161)
(171)
(172)
Select a brightness by ＋ and －Press SET
Selected backlight brightness is entered

Press DISPLAY

Set the tidal
current measurementof reactive power
and power-factor.
(322)

Press MODE
Press SET and DISPLAY together for longer than 3 seconds
(311)
(321)
Select of dead band value in measurement display by ＋ and －Press SETThe value of dead band in display is entered

Press DISPLAY

Page

57

57

Page

67

Returns to display mode.

Press MODE
Press SET and DISPLAY together for longer than 3 seconds
(311)
(321)
Press RESET/SHIFT
Select a tidal current measurement ON/OFF by ＋ and －(322)
Press SET
Selected action is entered
Press DISPLAY
Returns to display mode.

45

56

Returns to display mode.

(15) Other, measurement setting
Each parenthesized number shows a setting number and this number is displayed on the setting screen.
Items
Setting and operation procedures
Set the dead bandof measurement
display.
(321)

56

67

SQLC-215-109

5.3

Setting in detail explanation

5.3.1 Setting mode 1SET
3 seconds
DISPLAY

111 to 116
Display setting
MODE

DISPLAY

121AL to 128AL
Alarm output setting
MODE

DISPLAY

131H to 136
Demand detection setting
MODE

Display mode
(Measurement
display)

DISPLAY

141H to 14A
Harmonic detection setting
MODE

DISPLAY

151H to 152L
Instant measurement detection setting
MODE

DISPLAY

161 to 164
Leakage detection setting
MODE

DISPLAY

171 to 172
Backlight setting

Setting mode 1

MODE

Setting mode 1 is selected by pressing SET switch for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.

< Caution >If setting change should have been mistaken, an alarm output is not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

46

SQLC-215-109(1) 111 to 116 Display combination setting 【All models】● 3-phase 3-wire (Voltage, current input) (33)(34)
Sub-monitor№
Pattern №
Main monitor(Left)
1
Pattern 1
A(L2)
V(L1L2)
2
Pattern 2
A(L2)
V(L1L2)
3
Pattern 3
A(L2)
V(L1L2)
4
Pattern 4
DA(L2)A(L2)5
Pattern 5
DA(L2)A(L2)6
Pattern 6
DA(L2)
V(L1L2)
7
Pattern 7
W
V(L1L2)
8
Pattern 8W
V(L1L2)
9
Pattern 9W
V(L1L2)
10
Pattern 10
DW
V(L1L2)
11
Pattern 11
DW
V(L1L2)
12
Pattern 12
A(L2)
cosφ
13
Pattern 13
A(L2)
var
14
Pattern 14
W
cosφ
15
Pattern 15
A(L2)
A(L1)
16
Pattern 16
V(L1L2)V(L2L3)
17
Pattern 17
A(L1)Io/Igr (35)18
Pattern 18
A(L1)Io/Igr (35)Sub-monitor(Center)
W
W
W
V(L1L2)
V(L1L2)
W
A(L2)
A(L2)
A(L2)
W
A(L2)
W
W
var
A(L3)
V(L3L1)
V(L1L2)
V(L1L2)Sub-monitor(Right)
Wh
cosφ
Hz
W
Wh
cosφ
Wh
cosφ
Hz
Wh
cosφ
Wh
Wh
Wh
Wh
Hz
W
Wh

Bar graph
A(L2)
A(L2)
A(L2)
DA(L2)
DA(L2)
DA(L2)
W
W
W
DW
DW
A(L2)
A(L2)
W
A(L2)
V(L1L2)
A(L1)
A(L1)

Note(33) Voltage input is pattern 16 only.
Note(34) Current input is pattern 15 only. (However, sub-monitor (right) is displayed blank.)
Note(35) Io/Igr is only with leakage measurement.
● Displays set factor (3-phase 3-wire)
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，var，cosφ，Hz，Main monitor
Io/Igr，Wh，-Wh，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)，Distortion factor(A,V)
Sub-monitor
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，W，var，cosφ，Io/Igr(Left)
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，var，Sub-monitor
varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)，Harmonic 5th conversion content(A,V)，(Center)
Harmonic nth content(A,V)
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，cosφ，Hz，Sub-monitor
Wh，-Wh，Fundamental-wave effective value(A,V)，Harmonic 5th conversion effective value(A,V)，(Right)Harmonic nth effective value(A,V)
V(L1L2)，V(L2L3)，V(L3L1)，A(L1)，A(L2)，A(L3)，DA(L1)，DA(L2)，DA(L3)，W，DW，var，cosφ，Hz，
Io/Igr，Distortion factor(A,V)，Harmonic 5th conversion content(A,V)，Harmonic nth content(A,V)，
Bar graph
Fundamental-wave effective value(A,V)，Harmonic 5th conversion effective value(A,V)，
Harmonic nth effective value(A,V)● Phase (line) change (3-phase 3-wire) ● Measurement factor change (3-phase 3-wire, Measurement display mode)(36)
V(L1L2) V(L2L3) V(L3L1)
V(L1L2) V(L2L3) V(L3L1)
A(L1)
A(L2)
A(L3)
DA(L1)
A(L2)

A(L3)

A(L1)

Note(36) Press DISPLAY , Voltage
and current are replacedat the same time.

DA(L2)

DA(L3)W

DW

Io/IgrWh-Whvarh(LAG)-varh(LAG)-varh(LEAD)● Measurement factor change (3-phase 3-wire, Harmonic measurement display mode）
Main monitor：Distortion factor (Fixation)
Sub-monitor (Left) ：Harmonic order ｎ
5th conversion 1 3 4 5 7 9Sub-monitor (Center)：Harmonic nth content (Fixation)
Sub-monitor (Right) ：Harmonic nth effective value (Fixation)

47

var

cosφ

Hz

varh(LEAD)
Nothing

11

13

15

SQLC-215-109● Single-phase 3-wire (Voltage, current input) (37)(38)Sub-monitor№
Pattern №
Main monitor(Left)
1
Pattern 1
A(L1)
V(L1N)
2
Pattern 2
A(L1)
V(L1N)
3
Pattern 3
A(L1)
V(L1N)
4
Pattern 4
DA(L1)A(L1)
5
Pattern 5
DA(L1)A(L1)
6
Pattern 6
DA(L1)
V(L1N)
7
Pattern 7W
V(L1N)
8
Pattern 8W
V(L1N)
9
Pattern 9W
V(L1N)
10
Pattern 10
DW
V(L1N)
11
Pattern 11
DW
V(L1N)
12
Pattern 12
A(L1)
cosφ
13
Pattern 13
A(L1)
var
14
Pattern 14
W
cosφ
15
Pattern 15
A(L1)
A(L3)
16
Pattern 16
V(L1N)
V(L3N)
17
Pattern 17
A(L1)Io/Igr (39)18
Pattern 18
A(L1)Io/Igr (39)Sub-monitor(Center)
W
W
W
V(L1N)
V(L1N)
W
A(L1)
A(L1)
A(L1)
W
A(L1)
W
W
var
A(N)
V(L1L3)V(L1N)
V(L1N)

Sub-monitor(Right)
Wh
cosφ
Hz
W
Wh
cosφ
Wh
cosφ
Hz
Wh
cosφ
Wh
Wh
Wh
Wh
Hz
W
Wh

Bar graph
A(L1)
A(L1)
A(L1)
DA(L1)
DA(L1)
DA(L1)
W
W
W
DW
DW
A(L1)
A(L1)
W
A(L1)
V(L1N)
A(L1)
A(L1)

Note(37) Voltage input is pattern 16 only.
Note(38) Current input is pattern 15 only. (However, sub-monitor (right) is displayed blank.)
Note(39) Io/Igr is only with leakage measurement.
● Displays set factor (Single-phase 3-wire)
V(L1N)，V(L3N)，V(L1L3)，A(L1)，A(L3)，A(N)，DA(L1)，DA(L3)，DA(N)，W，DW，var，cosφ，Hz，Io/Igr，
Main monitor
Wh，-Wh，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)，Distortion factor(A,V)
Sub-monitor
V(L1N)，V(L3N)，V(L1L3)，A(L1)，A(L3)，A(N)，W，var，cosφ，Io/Igr(Left)
V(L1N)，V(L3N)，V(L1L3)，A(L1)，A(L3)，A(N)，DA(L1)，DA(L3)，DA(N)，W，DW，var，Sub-monitor
varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)，Harmonic 5th conversion content(A,V)，(Center)
Harmonic nth content(A,V)
V(L1N)，V(L3N)，V(L1L3)，A(L1)，A(L3)，A(N)，DA(L1)，DA(L3)，DA(N)，W，DW，cosφ，Hz，Sub-monitor
Wh，-Wh，Fundamental-wave effective value(A,V)，Harmonic 5th conversion effective value(A,V)，(Right)Harmonic nth effective value(A,V)
V(L1N)，V(L3N)，V(L1L3)，A(L1)，A(L3)，A(N)，DA(L1)，DA(L3)，DA(N)，W，DW，var，cosφ，Hz，Io/Igr，
Distortion factor(A,V)，Harmonic 5th conversion content(A,V)，Harmonic nth content(A,V)，
Bar graph
Fundamental-wave effective value(A,V)，Harmonic 5th conversion effective value(A,V)，
Harmonic nth effective value(A,V)● Phase (line) change (40)(Single-phase 3-wire)
V(L1N)
V(L3N)
V(L1L3)
A(L1)

A(L3)

A(N)

Note(41) Press DISPLAY , Voltage
and current are replacedat the same time.

● Measurement factor change (3-phase 3-wire, Measurement display mode)
V(L1N)

V(L3N)

V(L1L3)

A(L1)

A(L3)

A(N)

DA(L1)

DA(L3)

DA(N)

W

DW

var

cosφ

Hz

Io/IgrWh-Whvarh(LAG)-varh(LAG)-varh(LEAD)

varh(LEAD)
Nothing● Measurement factor change (Single-phase 3-wire, Harmonic measurement display mode）
Main monitor：Distortion factor (Fixation)
Sub-monitor (Left) ：Harmonic order ｎ
5th conversion 1 3 4 5 7 9 11Sub-monitor (Center)：Harmonic nth content (Fixation)
Sub-monitor (Right) ：Harmonic nth effective value (Fixation)

48

13

15

SQLC-215-109● Single-phase 2-wire (Voltage, current input) (41)(42)Sub-monitor№
Pattern №
Main monitor(Left)
1
Pattern 1
AV2
Pattern 2
AV3
Pattern 3
AV4
Pattern 4DAA5Pattern 5DAA6Pattern 6DAV
7Pattern 7WV
8Pattern 8WV
9Pattern 9WV10Pattern 10
DWV11
Pattern 11
DWV12
Pattern 12
A
cosφ
13
Pattern 13
A
var14
Pattern 14
W
cosφ
15
Pattern 15
A－
16
Pattern 16
V－17
Pattern 17
A
Io/Igr (43)18
Pattern 18
A
Io/Igr (43)Sub-monitor(Center)
W
W
W
V
V
W
A
A
A
W
A
W
W
var
－
－
V
V

Sub-monitor(Right)
Wh
cosφ
Hz
W
Wh
cosφ
Wh
cosφ
Hz
Wh
cosφ
Wh
Wh
Wh
Wh
Hz
W
Wh

Bar graph
A
A
A
DA
DA
DA
W
W
W
DW
DW
A
A
W
A
V
A
A

Note(41) Voltage input is pattern 16 only.
Note(42) Current input is pattern 15 only. (However, sub-monitor (right) is displayed blank.)
Note(43) Io/Igr is only with leakage measurement.
● Displays set factor (Single-phase 2-wire)
V，A，DA，W，DW，var，cosφ，Hz，Io/Igr，Wh，-Wh，varh(LAG)，varh(LEAD)，-varh(LAG)，
Main monitor-varh(LEAD)，Distortion factor(A,V)
Sub-monitor
V，A，W，var，cosφ，Io/Igr(Left)
Sub-monitor
V，A，DA，W，DW，var，varh(LAG)，varh(LEAD)，-varh(LAG)，-varh(LEAD)，(Center)
Harmonic 5th conversion content(A,V)，Harmonic nth content(A,V)
Sub-monitor
V，A，DA，W，DW，cosφ，Hz，Wh，-Wh，Fundamental-wave effective value(A,V)，(Right)Harmonic 5th conversion effective value(A,V)，Harmonic nth effective value(A,V)
V，A，DA，W，DW，var，cosφ，Hz，Io/Igr，Distortion factor(A,V)，
Harmonic 5th conversion content(A,V)，Harmonic nth content(A,V)，
Bar graph
Fundamental-wave effective value(A,V)，Harmonic 5th conversion effective value(A,V)，
Harmonic nth effective value(A,V)● Measurement factor change (Single-phase 2-wire, Measurement display mode)
V
A
DA
W
DW
var
cosφ
Hz

varh(LAG)

varh(LEAD)-varh(LAG)-varh(LEAD)

Io/Igr49-Wh

Nothing● Measurement factor change (Single-phase 2-wire, Harmonic measurement display mode）
Main monitor：Distortion factor (Fixation)
Sub-monitor (Left) ：Harmonic order ｎ
5th conversion 1 3 4 5 7 9 11Sub-monitor (Center)：Harmonic nth content (Fixation)
Sub-monitor (Right) ：Harmonic nth effective value (Fixation)

Wh

13

15

SQLC-215-109
Display combination setting
RESET/SHIFT
111
Combination
display

RESET/SH
IFT

112
Main monitor

Pattern 1

A(L2)：5＋＋－

RESET/SH
IFT

113
Sub-monitor(Left)RESET/SH
IFT

V(L1L2)：1

－

＋

114
Sub-monitor(Center)

RESET/SH
IFT

＋

－

＋

A(L3)：6

V(L2L3)：2

DW：11-Wh：15

＋

＋

＋

＋

＋

DA(L1)：7

Pattern 3

＋
＋

－

－

＋

＋

－

＋

－

＋

－
＋

＋

A(L1)：4－

Pattern 17

＋

＋

＋

DA(L3)：9

A(L2)：5＋＋－

Pattern 18
＋

－

－

W：10

－

Manual setting
value：0

＋

＋

－

＋

－

＋

＋

－

＋

－

＋

Nothing：0＋－

cosφ：9
＋

＋

＋

－

＋

－

－

－

－

A(L2)：5－

V(L1L2)：1

＋
－

＋

－

＋

－

Io/Igr：10

V(L2L3)：2

DA(L1)：7

＋

＋

＋

＋

Io/Igr：15＋－－

Nothing：0－－－

V(L3L1)：3

DA(L2)：8

＋

＋

－

－

Wh：16

A(L1)：4

DA(L3)：9

＋

＋

＋

－

－

-Wh：17

A(L2)：5＋＋－varh(LAG)：18＋－

varh(LEAD)：19＋－－

V(L1L2)：1＋－

V(L2L3)：2＋－

V(L3L1)：3
＋

DW：11
－

DA(L1)：7

cosφ：12

＋

＋

－

－

Hz：13

－

Default setting－

Nothing：0＋－

DA(L3)：9-varh(LEAD)：21
＋

＋

＋

＋

－

－

＋

－

At the time of
3-phase 3-wire

W：10

A(L3)：6

DA(L2)：8-varh(LAG)：20＋－－

A(L3)：6

Hz：14－

Sub-monitor：RightA(L1)：4

－

var：8

－

＋

－

＋

－

A(L1)：450－

V(L3L1)：3-varh(LEAD)：16

var：12＋＋-varh(LAG)：15

－

＋

－

V(L2L3)：2

－

＋

－

＋

－

varh(LEAD)：14

W：7

cosφ：13

＋

＋

－

＋

DW：11

At the cases other thana display combination
pattern, it becomesmanual setting.

－A(L3)：6

V(L1L2)：1

varh(LAG)：13－

Sub-monitor：Center－
DA(L2)：8－

Sub-monitor：Left

Nothing：0

var：12V(L3L1)：3－－

Main monitor－

Pattern 2－

116
Bar graphfactorRESET/SH
IFT

Wh：14

W：10

－

115
Sub-monitor(Right)－SQLC-215-109◆ 111 Combination display
Select the factors to be measured and monitored by 4 digital
displays out of combination patterns.
Set values are updated by SET .

Setting №◆ 112 to 115 Main monitor, Sub-monitor (left),
Sub-monitor (center), Sub-monitor (right)
Set these items for a display configuration other than
combined patterns. Set values are updated by SET .
◆ 116 Bar graph factor
A factor being monitored in the main monitor is basicallydisplayed by a bar graph.
Set this item for displaying a factor being monitored on a
sub-monitor by bar graph.
An underbar is attached to the digital display of thesetting sub-monitor.
Set values are updated by SET .

Setting №
Underbar

51

New setting
Current setting

SQLC-215-109(2) 121AL to 128AL Alarm output setting 【With an alarm output option】
Various setting and an output test are performed about an alarm output.
In case contact outputs 1 and 2 are alarm-output specifications, the corresponding alarm output is setting.
121AL
Alarm 1 factor

RESET/SHIFTRESET/SHIFT
128AL
Alarm 2 test

RESET/SHIFT

122AL
Alarm 1
return method

RESET/SHIFT

127AL
Alarm 2 contact
delay timeRESET/SHIFT

123AL
Alarm 1 contact
delay timeRESET/SHIFT

126AL
Alarm 2
return method

RESET/SHIFT

124AL
Alarm 1 test
RESET/SHIFT
125AL
Alarm 2 factor◆ 121AL Alarm 1 factor setting，125AL Alarm 2 factor setting.
Set the output factor of alarms 1 and 2. Selection by ＋ and － , set value is updated by SET .
Default value ：1(DA：Demand current)
OFF：Nothing＋－＋

1：Demandcurrent＋－－
10：Voltage－＋＋

2：Demandpower

9：Harmonic nth
content(Voltage)－＋－

3：Io/Igr
Current
leakage

8：Harmonic 5th
conversion
content(Voltage)

－
＋

＋
－

4：Distortion
factor(Current)

7：Distortion
factor(Voltage)

－
＋

＋
－

5：Harmonic 5th
conversion
content(Current)

6：Harmonic nth
content(Current)－＋◆ 122AL Alarm 1 reset method setting,
126AL Alarm 2 reset method setting
Output action at the case of a reset of alarms 1 and 2 can be selectedfrom AUTO (automatic reset) and HOLD (manual reset).
In "AUTO (automatic reset)", an alarm output also constitutes OFF
according to a reset of an alarm. In "HOLD (manual reset)", even afteran alarm reset, an output holds ON. The reset in this case (output OFF)is performed in RESET/SHIFT .
Selection by ＋ and － , set value is updated by SET .
Default setting：AUTO (automatic reset)
New setting
Current setting
Setting №◆ 123AL Alarm 1 contact delay time, 127AL Alarm 2 contact delay time
Set the contact delay time of alarms 1 and 2. The setting range is 0 to 300 seconds (1-second step).
Selection by ＋ and － , set value is updated by SET .
Default setting：0 second (With no contact delay)◆ 124AL Alarm 1 test, 128AL Alarm 2 test
The output of alarms 1 and 2 is tested. While pushing SET , an output is ON, and if it detaches, an output
switches OFF. In addition, while pushing SET , the test input exceeding rated sensitivity current value isapplied to the zero-phase current input circuit inside a product, and if the time which is pushing SET exceeds
the operating time (time-delay type,
Setting No.
Status of output
Setting No.
Status of output
0.1 second ＜ 2 second) of a current
leakage, an alarm will occur.
(In case an alarm-output factor isa current leakage (Io/Igr))And, test input will become OFF if
SET is detached. Please apply rated
SET
voltage to a voltage-input terminal(between P1-P2) at this time.
In addition, about a reset of an alarm,
it becomes action united with reset
method setting (automatic reset /manual reset) of each alarm output.
(Output OFF)

(Output ON)
Alarm 1 test

52

SQLC-215-109(3) 131H to 136 Demand detection setting 【Except voltage input】
The following operation method is setting.
Demand current, action of demand, high-alarm value, time-interval, power-factor, demand active power.
131H
Demand current
upper limit

RESET/SHIFT

136
Power-factor
operation method

RESET/
SHIFT

132
Demand current
time interval

RESET/
SHIFT

135
Demand power
operation method

RESET/SHIFT

133H
Demand power
upper limit

RESET/
SHIFT

134
Demand power
time interval

RESET/SHIFT◆ 131H Demand current upper limit. 133H Demand active power upper limit.
Set the high-alarm value of demand current(DA) and demand active power(DW).
The setting range is 5 to 100％ (1％ step). Selection by ＋ and － , set value is updated by SET .
Default setting：80％ (Demand current),
OFF [Non use] (Demand active power)◆ 132 Demand current time interval. 134 Demand active power time interval.
Set the time interval (95％ time interval) of demand current(DA) anddemand active power(DW). Selection by ＋ and － , set value is updated
by SET .

New setting

Current setting

Default setting：0 second (demand current, demand active power)
0sec＋(Nothing)

－
－
＋

5sec

－

4min

5min
－
＋

＋

30min

－
＋
＋
－
－
＋

10sec

3min

6min

25min

＋
－
－
＋
＋
－
－
＋

20sec

2min

7min

20min

＋
－
－
＋
＋
－
－
＋

30sec

1min

8min

15min

＋
－
－
＋
＋
－
－
＋

40sec

＋
－

50sec

9min

＋
－

Setting factor
10min

Setting No.

Demand current time interval◆ 135 Demand active power operation method
d：Operating system
The operating system of demand (DW) can be selected fromaccording withd (operating system according with bimetallic type：bimetallic typedemand) and A (Averaging arithmetic in demand timedelay：average).
Selection by ＋ and － , set value is updated by SET .
Default setting：d (Operating system according with bimetallic type)＋＋－

A：Averaging
arithmetic indemand time
delay

－

● Demand time delay characteristic (Demand current, Demand active power)
100
95
80

Arithmetic method accordingwith bimetallic type.
(Indication time to 95％
of a final constant value)

Time delay 10 minutes

Demandvalue 60（％）

Averaging arithmetic indemand time delay.
(Averaging time in demand
time delay.)

40

20
0
0

10

20

30

Time（min）

Arithmetic method
Demand current measurement：Arithmetic method according with bimetallic type.
Demand active power measurement：Arithmetic method according with bimetallic type (Default value).
Or the averaging arithmetic in a demand time interval.
One is selected by setting.
100％ indication time is about 3 times the time delay at the case of the arithmetic method according withbimetallic type.(In case of 10 minutes/95％ of time-interval, time to reach to 100％ is about 30 minutes.)
Demand measurement is measured to the 2 times of the rated current, and the 2 time of a rated active power.
53

SQLC-215-109◆ 136 Power-factor operation method
The operation method of power-factor measurement can be selected from 0(instant measurement) and 1 (averaging operator in a demand timeinterval).
In case it is set as "1 (averaging operator in a demand time interval)",power-factor measurement is calculated from the operation method of apower demand time interval and a demand active power meter.
Selection by ＋ and － , set value is updated by SET .
Default setting：0 (Instant measurement )＋0：Instantmeasurement＋

New setting

1：Averaging operator－

in a demand timeinterval

Current setting
Setting No.
Power-factor operation method－(4) 141H to 14A Harmonic detection setting 【All models】
Set the high-alarm value of each harmonic (current, voltage) measurement factor, factor, 5th conversion detection
characteristics, and average time interval is performed.
141H
Current
distortion-factor
upper limit

RESET/SHIFTRESET/SHIFT

142H
Current 5th
conversion content
upper limit
147
Voltage nthcontent factor

RESET/SHIFT

148H
Voltage nthcontent upper
limit

RESET/SHIFT

144H
Current nth
content upper
limit

RESET/SHIFT

146H
Voltage 5th
conversion content
upper limit

RESET/SHIFT

145H
Voltagedistortion-factor
upper limit

RESET/
SHIFT

149
5th conversion
detection
characteristics

RESET/SHIFTRESET/SHIFT

143
Current nth
content factor◆ 141H Current distortion-factor upper limit,
145H Voltage distortion-factor upper limit

RESET/SHIFT

14A
Average timeinterval

Setting factor

New setting

Set the high-alarm value of distortion factor (current, voltage).
Setting range. Current：5 to 100％ (1％ step)
Voltage：1.0 to 20.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF [Non-use] (Current, Voltage)◆ 142H Current 5th conversion content upper limit,
146H Voltage 5th conversion content upper limit.
Set the high-alarm value of 5th conversion content (current, voltage).
Setting range. Current：5 to 100％ (1％ step)
Voltage：1.0 to 20.0％ (0.1％ step)
Current setting
Setting No.
Selection by ＋ and － , set value is updated by SET .
Current distortion-factor upper limit
Default setting：OFF [Non-use] (Current, Voltage)◆ 143 Current nth content factor, 147 Voltage nth content factor.

Setting factor

New setting

Set the factor (order) of nth content (current, voltage).
An order can be selected from n= 3, 4, 5, 7, 9, 11, 13, 15.
Selection by ＋ and － , set value is updated by SET .
Default setting：5th(Current, Voltage)◆ 144H Current nth content upper limit,
148H Voltage nth content upper limit.
Set the high-alarm value of nth content (current, voltage).
Setting range. Current：5 to 100％ (1％ step)
Voltage：1.0 to 20.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF [Non-use] (Current, Voltage)

Setting No.

Current setting

Voltage nth content factor

54

SQLC-215-109◆ 149 5th conversion detection characteristics.
Alarm output, Inverse-time-delay characteristics
In case of high-limit-setting value 35％(voltmeter 3.5％)
10000

The detection characteristics of 5th conversion content canbe selected from A (average mode) and I (inverse-time-delaymode).
In case of "A (average mode)", when average measured value(the average of instantaneous value in average timeinterval) exceeded the upper limit alarm value, it detects.
And in case of "I (inverse-time-delay mode)", when exceeding
the upper limit warning value by anti-time limit
characteristic of instantaneous value, it detects.
Selection by ＋ and － , set value is updated by SET .
Default setting：I(inverse-time-delay mode)

1000

＋

A：Average＋

I：inverse-time-

mode

－

delay mode

100
Time
(sec）

－

New setting

10

Setting factor
Setting No.

1(Ammeter) 0(Voltmeter) 0

Current setting

20
2

40
4

80 100 120 140 160 180 200
8 10 12 14 16 18 20

60
6

5th conversion content (％)

5th conversion detection◆ 14A Average time interval
Set the average time interval of each harmonic measurement.
Selection by ＋ and － , set value is updated by SET .
In addition, in case the detection characteristics of 5th conversion content are set as inverse-time-delay mode,in order that 5th conversion content may operate in inverse-time-delay characteristics, this time interval isdisregarded. Default setting：0 minute (With no average)
0 minute(Nothing)＋－

1 minute＋－

2 minute＋－

5 minute＋－

10 minute＋－15 minute＋－30 minute－＋(5) 151H to 152L Instant measurement detection setting 【Current input is excluded.】
An upper limit low-alarm value is set by instant measurement (voltage factor).
151H
Instant measurement
voltage upper limit

RESET/SHIFT

152L
Instant measurement
voltage lower limit

New setting

Current setting

Setting No.
RESET/
SHIFT◆ 151H Instant measurement voltage upper limit,
152L Instant measurement voltage lower limit.
Set the high-alarm value and low-alarm value of instant measurement(voltage).
Setting range is 30 to 150％(1％ step).
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF〔Non-use〕(Upper limit, lower limit )55

Instant measurement
voltage upper limit

SQLC-215-109(6) 161 to 164 Leakage detection setting 【With leakage measurement】
The rated sensitivity current value of leakage measurement, a detection factor, the circuit configuration at thecase of 3-phase 3-wire, and setting of ZCT to be used are performed.
161
Leakage detection
rated sensitivity
current value

RESET/SHIFT

162
Leakage detection
factor change

163
RESET/SHIFT

Leakage detectcircuitry change

164
RESET/SHIFT

Use ZCT selection

RESET/SHIFT◆ 161 Leakage detection rated sensitivity current value
Set the rated sensitivity current value of current leakage.
Setting value can be selected from 0.03A, 0.05A, 0.1A, 0.2A, 0.4A, 0.8A.
Selection by ＋ and － , set value is updated by SET .
Default setting：0.1A

New setting

Current setting◆ 162 Leakage detection factor change
Measurement of current leakage and detection factor can be selectedfrom Io and Igr.
Selection by ＋ and － , set value is updated by SET .
However, a current input product constitutes Io fixation and cannot carryout Igr detection.
Default setting：Io<Caution> Please apply a voltage input, in case you measure and detect
Igr. In case the voltage input is not applied, measurement or
detection of Igr cannot be performed correctly.

Setting No.
Leakage detection
sensitivity current value

＋
＋
－

Io

Igr

－

◆ 163 Leakage detect-circuitry change
The circuit configuration of the current-leakage detection in 3-phase 3-wire can be selected from 1-phase earthing,
1-phase earthing (negative phase sequence), and non-earthing.
Selection by ＋ and － , set value is updated by SET . Default setting：A (1 phase earthing)<Caution> If actual connection is different from this setting, a current leakage Igr cannot measure or detectcorrectly. Setting of "1-phase earthing (negative phase sequence)" is setting for making Igr measureand detect correctly, in case a circuit is a negative phase sequence. In the case of the right usualphase sequence, it is not necessary to make it this setting.
＋＋

A：1-phase

earthing－

B：1-phase earthing(Negative phasesequence)＋－

C：non-earthing－◆ 164 Use ZCT selection
ZCT used by current-leakage measurement can be selected.
Selection by ＋ and － , set value is updated by SET .
Default setting：0(TYPE 0)<Caution> If ZCT which you actually use is different from this setting, the error of current leakages Io and
Igr will become large. In case you use ZCT(s) other than our company recommendation goods, please
consult in advance.
＋

0：TYPE 0＋－

1：TYPE 1

TYPE 0：Our company recommendation product (Standard).
TYPE 1：Except our company recommendation product.

－

56

SQLC-215-109(7) 171 to 172 Backlight setting 【All models】(However, the brightness of backlight is only white backlight products)
It sets action and brightness of backlight.
171
Backlight action

RESET/
SHIFT

172
Backlight brightness

RESET/SHIFT

New setting

Current setting◆ 171 Backlight action
It can select from ON (always-on), AUTO (auto off), and OFF (always-off)about action of backlight.
If 5 minutes elapses without operating a switch in case it is set as "AUTO(auto off)", backlight will go out automatically.
After that, backlight will be turned on if either of switches is operated.
Selection by ＋ and － , set value is updated by SET .
Default setting：AUTO (Auto off)
AUTO：
Auto off

＋

OFF：

＋

ON：

－

Always-off－

Always-on－＋◆ 172 Backlight brightness
It can select the brightness of backlight as five steps of 1 to 5.
Backlight becomes the darkest if it is set as "1".
Backlight becomes the brightest if it is set as "5".
Selection by ＋ and － , set value is updated by SET .
Default setting：3 (Middle)

57

Setting No.
Setting factor
Backlight

Setting54321Brightness
BrightDark

SQLC-215-109

5.3.2 Setting mode 2
SET andRESET/SHIFT
3 seconds
211 to 218
Measurement-range setting

DISPLAY

MODE
221A to 228A
Analog output setting

DISPLAY

MODE
241P to 244P
Pulse output setting

DISPLAY
Display mode

MODE

(Measurement
display)

251 to 252
External operation input setting

DISPLAY

MODE
261 to 26D
Measurement ON/OFF setting

DISPLAYMODE
DISPLAY

271 to 272
Initialization of watt-hour and setting value.

Setting mode 2

MODE

Setting mode 2 is selected by pressing SET and RESET/SHIFT switches continuously for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
< Caution >If setting change should have been mistaken, a display and output of measurement are not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

(1) 211 to 218 Measurement-range setting 【All models】
Setting of measurement range in each measurement factors.
211
Voltage range
RESET/SHIFT

212
RESET/SHIFTRESET/
SHIFT

217

218
Frequency range

Current range

RESET/SHIFTPower-factor
range

213
Current display
intrinsic
sensitivity

214
RESET/SHIFT216
RESET/SHIFT

58

Reactive power
range

Power polarity

215
RESET/SHIFTPower range

RESET/SHIFTSQLC-215-109◆ 211 Voltage range
Set the voltage range (VT ratio). Change of this setting also sets the measurement range of active power and reactive
power automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：6600V(3φ3W)、110.0V(1φ3W)、3300V(1φ2W)
Voltage measurement range (34 range)
Setting factor
150.0V(110V)
150V (110V)
300.0V(220V)
300V (220V)
500V (380V)
600V (440V)
600V (460V)
600V (480V)
1200V (880V)
1500V (1100V)
2400V (1650V)
3000V (2200V)
3.00kV (2200V)

4500V (3300V)
4.50kV (3300V)
9000V (6600V)
9.00kV (6600V)
15.00kV (11kV)
18.00kV (13.2kV)
18.00kV (13.8kV)
24.00kV (16.5kV)
25.00kV (18.4kV)
30.0 kV (22kV)
45.0 kV (33kV)
90.0 kV (66kV)
120.0 kV (77kV)

New setting

150.0kV (110kV)
180.0kV (132kV)
210.0kV (154kV)
270.0kV (187kV)
300.0kV (220kV)
400.0kV (275kV)
500.0kV (380kV)
750.0kV (550kV)

Setting No.
Current setting
Voltage range◆ 212 Current range
Set the current range (CT ratio). Change of this setting also sets the measurement range of active power andreactive power automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：100.0A(3φ3W)、500A(1φ3W)、50.0A(1φ2W)
Current-measurement range (76 range)
5.00A
6.00A
7.50A
8.00A
10.00A
10.0A
12.00A
12.0A
15.00A
15.0A

20.00A
20.0A
25.00A
25.0A
30.00A
30.0A
40.0A
50.0A
60.0A
75.0A

80.0A
100.0A
100A
120.0A
120A
150.0A
150A
200.0A
200A
250.0A

250A
300.0A
300A
400A
500A
600A
750A
800A
900A
1000A

1.00kA
1200A
1.20kA
1500A
1.50kA
1600A
1.60kA
1800A
1.80kA
2000A

2.00kA
2500A
2.50kA
3000A
3.00kA
4000A
4.00kA
5000A
5.00kA
6000A

6.00kA
7500A
7.50kA
8000A
8.00kA
9.00kA
10.00kA
10.0kA
12.00kA
12.0kA

15.00kA
15.0kA
20.00kA
20.0kA
30.00kA
30.0kA◆ 213 Current display intrinsic sensitivity
Set the full scale of current meter.
The setting range is 40 to 120％ of CT ratio. And, it can select from the following values.
Selection by ＋ and － , set value is updated by SET .
Setting factor
New setting
Default setting：100.0A(3φ3W), 500A(1φ3W), 50.0A(1φ2W)
The measurement range (current, active
power, reactive power) which can be set.
(×10n)
1.□
1.0／1.2／1.4／1.5／1.6／1.8
2.□
2.0／2.4／2.5／2.8
3.□
3.0／3.2／3.6
4.□
4.0／4.2／4.5／4.8
5.□
5.0／5.66.□
6.0／6.4
7.□
7.2／7.58.□
8.0／8.49.□
9.0／9.6

Example )In case of CT ratio＝100.0A.
･40％ of 100A is 40A.
･120％ of 100A is 120A.
A measurement range can beselected within the limitsof 40 to 120A.
Therefore, a left table
A measurement range can beselected from
40/42/45/48/50/56/60/64/72/
75/80/84/90/96/100/120A.

Setting No.

Current setting

Current display intrinsic
sensitivity

59

SQLC-215-109◆ 214 Active power polarity
A swing display of active power meter can be selected from P (one side swing) and － (both swings).
Selection by ＋ and － , set value is updated by SET .
Default setting：P (One-side swing)
New setting＋P：
One-side swing＋－

Both swing display－：Both swings－

Current setting
Setting No.
One-side swing display
Power polarity◆ 215 Active power range
Set the full scale of active power meter. The setting range is 40 to 115％ of range of a VT ratio × CT ratio.
And it can select from "213 current display intrinsic sensitivity" tables.
Selection by ＋ and － , set value is updated by SET .
Setting factor
New setting<Caution> VT ratio：In case of 220V direct input. Calculates by VT
ratio＝2.
CT ratio：In case of 1A input. Calculates by CT ratio÷5.
Default setting：3φ3W：1200kW ，1φ3W：100.0kW ，1φ2W：150.0kW◆ 216 Reactive power range
Set the full scale of reactive power meter.
The setting range is 30 to 115％ of range of a VT ratio × CT ratio.
And it can select from "213 current display intrinsic sensitivity" tables.
Selection by ＋ and － , set value is updated by SET .
<Caution> VT ratio：In case of 220V direct input. Calculates by VT
ratio＝2.
CT ratio：In case of 1A input. Calculates by CT ratio÷5.
Default setting：3φ3W：600kvar ，1φ3W：50.0kvar ，1φ2W：75.0kvar

Setting No.

Current setting

Reactive power range◆ 217 Power-factor range
A power-factor measurement range can be selected from 0.500‐1.000‐0.500/0.000‐1.000‐0.000.
Change of this setting also sets the analog output range of a power-factor automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：0.500‐1.000‐0.500＋

0.500‐1.000‐0.500＋

0.000‐1.000‐0.000－－◆ 218 Frequency range
A frequency measurement range can be selected from 45.0‐55.0Hz/
55.0‐65.0Hz/45.0‐65.0Hz. Change of this setting also sets theanalog output range of a frequency automatically simultaneously.
Selection by ＋ and － , set value is updated by SET .
Default setting：45.0‐65.0Hz
50：
45.0‐55.0Hz

＋
－

60：
55.0‐65.0Hz＋－

Setting factor

New setting

55：
45.0‐65.0Hz－＋
Setting No.
Current setting
Frequency range

60

SQLC-215-109(2) 221A to 228A Analog output setting 【With an analog output option】
Various setting of analog output is performed.
221A
Ao1 output
factorRESET/SHIFT

222A
Ao2 output
factorRESET/SHIFT

227A
Reactive
power output
intrinsic
sensitivity

RESET/SHIFT
228A
Low input cut

RESET/SHIFTRESET/SHIFT

223A
Ao3 output
factor226APower-output
intrinsic
sensitivity

New setting
Setting No.
Current setting

RESET/SHIFT

224A
Ao4 output
factorRESET/SHIFT

225A
Current
outputintrinsic
sensitivity

RESET/
SHIFT◆ 221A to 224A AO(analog output)1 to 4 output factor.
Set the output factor of each analog outputs.
Selection by ＋ and － , set value is updated by SET .
Default setting：AO1 A(L2) (3φ3W)、A(L1) (1φ3W)、A (1φ2W)
AO2 V(L1L2) (3φ3W)、V(L1N) (1φ3W)、V (1φ2W)
AO3 W (3φ3W/1φ3W/1φ2W)
AO4 cosφ (3φ3W/1φ3W/1φ2W)
OFF：Nothing＋

1：V(L1L2)－－

9：DA(L3)＋－

8：DA(L2)

＋

10：W

＋

＋

19：Current
harmonic 4thcontent
20：Current
harmonic 5thcontent－＋

29：Current
harmonic 4th
effective value
30：Current
harmonic 5th
effective value－＋

39：Voltage
harmonic 4thcontent
40：Voltage
harmonic 5thcontent－＋

49：Voltage
harmonic 4th
effective value
50：Voltage
harmonic 5th
effective value

－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－

2：V(L2L3)

－

11：DW

＋

7：DA(L1)

21：Current
harmonic 7thcontent
28：Current
harmonic 3rd
effective value
31：Current
harmonic 7th
effective value
38：Voltage
harmonic 3rdcontent
41：Voltage
harmonic 7thcontent
48：Voltage
harmonic 3rd
effective value
51：Voltage
harmonic 7th
effective value

－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－

3：V(L3L1)－

12：var＋

6：A(L3)

22：Current
harmonic 9thcontent
27：Current
fundamental-wave
effective value
32：Current
harmonic 9th
effective value
37：Voltage
harmonic 5th
conversion
t t
42：Voltage
harmonic 9thcontent
47：Voltage
fundamental-wave
effective value
52：Voltage
harmonic 9th
effective value

－
＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－

13：cosφ－＋－

5：A(L2)

＋

14：Hz

－
16：Current
distortion
factor
23：Current
harmonic
11th content
26：Current harmonic
5th conversion
effective value
33：Current
harmonic 11th
effective value
36：Voltage
distortion
factor
43：Voltage
harmonic
11th content
46：Voltage harmonic
5th conversion
effective value
53：Voltage
harmonic 11th
effective value－＋

61

4：A(L1)＋－
17：Current
harmonic 5th
conversion content

＋
－

＋

－
18：Current
harmonic
3rd content

＋
－

＋

－
－

＋
－

Ao1 output factor－＋－

15：Io/Igr

＋
＋
－
－
＋
＋
－
－
＋
＋
－
－
＋
＋
－

24：Current
harmonic
13th content＋－

25：Current
harmonic
15th content
34：current
harmonic 13th
effective value＋－

35：Current
harmonic 15th
effective value
44：Voltage
harmonic
13th content＋－

45：Current
harmonic
15th content
54：Voltage
harmonic 13th
effective value
55：Voltage
harmonic 15th
effective value＋－

SQLC-215-109
Current setting
Setting No. New setting◆ 225A Current output intrinsic sensitivity,
226A Active power output intrinsic sensitivity,
227A Reactive power output intrinsic sensitivity.

Output intrinsic sensitivity (％ of an output to an input) is set
about each analog output of current, active power, and reactive power.
The setting range can be selected from the following.
Current：40.0 to 120.0％ (0.1％ step)
Active power ：40.0 to 115.0％ (0.1％ step)
Reactive power：30.0 to 115.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：100.0％(Current, Active power)，
50.0％(Reactive power)
Power-output intrinsic
Example) 3φ3W, 9000V, 100.0A, 1200kW range, Analog output DC4‐20mA
sensitivity,･Set 100.0％. Analog output is DC20mA in 1200kW of primary inputs.
･Set 80.0％. Analog output is DC20mA in 1200kW×0.8＝960kW of primary input.
◆ 228A Low input cut
It is the function which makes analog output a lower limit at the caseof a minute input (input which corresponds to 0.5％ or less).
A function can be selected from ON (Use) and OFF (Non-use).
Selection by ＋ and － , set value is updated by SET .
Default setting：OFF (Non-use)
New setting
Setting No.
Setting factor
Current setting
Low input cut(3) 241P to 244P Pulse output setting 【With a pulse output option】
Various setting of a pulse output is performed.
241P
Po1 output factor

RESET/SHIFT

242P
Po1 pulse unit

RESET/SHIFT

243P
Po2 output factor

RESET/SHIFT

244P
Po2 pulse unit

RESET/SHIFTNew setting Current setting◆ 241P PO(pulse output) 1 output factor,
243P PO(pulse output) 2 output factor

Setting No.

Set the output factor of each pulse output.
Selection by ＋ and － , set value is updated by SET .
Default setting：Wh (PO1，PO2)
OFF：
Nothing

＋

1：Wh

－

＋

2：-Wh

－

＋

3：varh
(LAG)

－

－
－
＋

6：-varh(LEAD)－＋

5：-varh
(LAG)

－

＋

4：varh(LEAD)＋

Po2 output factor
New setting Current setting◆ 242P PO(pulse output) 1 pulse unit,
244P PO(pulse output) 2 pulse unit

Setting No.

Set the pulse unit of each pulse output.
A pulse unit can be selected from four types.
The pulse unit that can be selected is decided with full-load power.
Default setting：10kWh/p [3φ3W] (PO1，PO2),
1kWh/p [1φ3W/1φ2W] (PO1，PO2)

Po1 pulse unit
62

SQLC-215-109(4) 251 to 252 External operation input setting 【With an external operation input option】
Various setting of external operation input is performed.
251
External operation
input 1 function

RESET/SHIFT

252
External operation
input 2 function

Setting factor

New setting
ALARM and RESET aredisplayed by turns.
(Alarm reset)

RESET/SHIFT◆ 251 External operation input 1 function,
252 External operation input 2 function

Measurement factors (W,cosφ, Hz, etc.) and DISP
are displayed by turns.
(Measurement factor
change)

The function of each external operation input(alarm reset, maximum / minimum reset,
measurement factor change, phase change) canbe selected.
Selection by ＋ and － , set value is
Setting No.

updated by SET .

Current setting

Default setting (With an alarm-output option)
Alarm reset (External operation input 1)
Maximum / minimum reset (External operation input 2)
Default setting (With no alarm-output option)
Maximum / minimum reset (External operation input 1)
Measurement factor change (External operation input 2)
ALARM RESET:
Alarm reset＋－

MAX./MIN. RESET：
Maximum / minimum reset＋－

Measurement factor DISP：
Measurement factor change＋－

L1,L2,L3 DISP：
Phase change－＋

・About the setting display in an external operation input function
Function
Alarm reset
Maximum /minimum reset
Measurement
factor change
Phase change

The contents of a display at the case of function setting"Current setting"display point"New setting"display point"ALARM" and "RESET" are displayed by turns by 7 segment
displays of the main monitor.
"RESET" is displayed by 7 segment displays of the main
monitor. And, "MAX" and "MIN" display by turns.
"DISP" is displayed by 7 segment displays of the main
monitor. And, each measurement factor (units of A, V,
W, etc.) displays by turns.
"DISP" is displayed by 7 segment displays of the main
monitor. And, each phase (L1, L2, L3, N) displays byturns.

Sub-monitor(right)Main monitor

63

SQLC-215-109(5) 261 to 26D Measurement ON/OFF setting 【All models (However, a current leakage is with leakage measurement)】
Measurement display ON/OFF setting of each measurement factor is performed.
Selection by ＋ and － , set value is updated by SET .
Default setting：ON (All measurement factors)
New setting
Setting No.
Current setting
261
262
264
263
Voltage

RESET/SHIFTRESET/SHIFTRESET/SHIFT

Current

Power
266

RESET/SHIFT

267
Power-receiving
watt-hourRESET/SHIFT

Frequency

268
Power
transmission
watt-hourRESET/SHIFT

269
Power-receiving
var-hour(LAG,LEAD)

26D
Current
leakage

RESET/SHIFTRESET/SHIFT

26C
Harmonic
voltage

RESET/
SHIFT

Reactive
power
265

RESET/SHIFTPower-factor
26A
Power transmission
var-hour(LAG,LEAD)RESET/SHIFTRESET/SHIFT

26B
Harmonic
current

RESET/
SHIFT

Voltage ON/OFF(6) 271 to 272 Initialization of watt-hour and setting value. 【All models 】
Package reset of watt-hour is performed. And, each set value is initialized (returns to default setting).
271
Setting value isinitialized.

RESET/SHIFT

272
Watt-hour reset

RESET/
SHIFT◆ 271 Initialization of setting value
Each set value is initialized (returns to default setting).
Pushing SET for 3 seconds initializes all set values.

SET
3 seconds

Setting No.

Setting No.

Completion of
initializing(Before initialization)(Initialization completion)

Initialization of setting value◆ 272 Watt-hour reset
Integrated value of each electric-energy display is cleared (＝0). By pushing SET for 3 seconds, all integrated
value (Wh, -Wh, var(LAG), -var(LAG), var(LEAD), var(LEAD)) is cleared by package.
Setting No.

Setting No.
Wh and varh aredisplayed byturns.

Wh and varh aredisplayed by turns.

SET
3 seconds

After clear
completion

Clear completion(Before clear)(Clear completion)
Watt-hour reset

64

SQLC-215-109

5.3.3 Setting mode 3
SET and DISPLAY
3 seconds
DISPLAY

311 to 312
Input circuit setting
MODE

DISPLAY

321 to 322
Measurement setting

Display mode
MODE

(Measurement
display)
DISPLAY

331
Analog output specification change
MODE

DISPLAY

341 to 348
Analog output adjust
MODE

Setting mode 3

Setting mode 3 is selected by pressing SET and DISPLAY switches continuously for longer than 3 seconds.
Pushing MODE switch performs movement of setting item.
The present mode can be returned to the display mode by pressing DISPLAY switch.
< Caution >If setting change should have been mistaken, a display and output of measurement are not obtained correctly.
Therefore, users must not set. The setting item without the corresponding option is not displayed.

65

SQLC-215-109(1) 311 to 312 Input circuit setting 【All models】
Set the input circuit and phase wire and input voltage / phase-voltage full scale.
311
Input circuit andphase wire change

312

RESET/SHIFT

Input voltage

RESET/
SHIFT◆ 311 Input circuit and phase wire change
Set the input circuit and phase wire (1φ3W).
Selection by ＋ and － , set value will be updated if SET ispushed for 3 seconds.
Default setting：3φ3W (3φ3W, No designation),
1φ3W [L1-N-L3] (1φ3W), 1φ2W (1φ2W)<Caution>
・When this setting is changed, it will become the default setting of the
input circuit after all set values changing.
・The right measurement cannot be performed if setting of actual connection
and phase wire are different.
3φ3W

＋
－

1φ3W
(L1-N-L3)

＋
－

1φ3W
(L1-N-L2)

＋
－

1φ3W
(L2-N-L3)＋－

New setting
Current setting
Setting No.

1φ2W

Input circuit andphase wire change－＋◆ 312 Input voltage
Set the input voltage (3φ3W, 1φ2W) or phase-voltage full scale (1φ3W).
In 3φ3W/1φ2W and 1φ3W, the contents of a setting are different.
Selection by ＋ and － , set value is updated by SET .
Default setting：300V (1φ3W)
Default setting：110V (3φ3W，1φ2W or no designation)
・3φ3W，1φ2W

・1φ3W
＋

110V

＋

＋

150V

220V

＋
300V

－

－

－

－

Setting of 300V, Output is AC0‐150V/DC4‐12mA.
Setting of 150V, Output is AC0‐150V/DC4‐20mA.
New setting
Setting No.
Current setting

L1N
Setting No.

New setting

L3N alternate display(Phase display L1-L3-N)

Current setting

Input-voltage rating (3φ3W)

Phase-voltage full scale (1φ3W)

66

SQLC-215-109(2) 321 to 322 Measurement setting 【All models】
Set the dead band of measurement display, and with or without of tidal current measurement.
321
Dead band of
measurement322
Tidal current
measurement

RESET/SHIFT

New setting
Current setting

RESET/SHIFT◆ 321 Measurement dead band
Set the dead band of measurement display.
By this setting, variation of less than this set value isdisregarded by voltage, current, active power, and reactive
power measurement display.
Setting range：0.0 to 2.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
Default setting：0.0％ (Nothing)◆ 322 Tidal current measurement
By measurement of reactive power and a power-factor,
0 (general measurement) or 1 (tidal current measurement
which was conscious of power transmission/power receiving)can be selected.
Selection by ＋ and － , set value is updated by SET .
Default setting：0 (General measurement)
0：Generalmeasurement＋＋

1：Tidal current－

measurement

Setting No.
Setting factor

Dead band of measurement

New setting
Current setting
Setting No.

Tidal current measurement－(General measurement)
Display

Analog output[mA] LEAD 0‐LAG 0/4‐20mA

LEAD 0‐LAG 0/LEAD 0‐LAG 0

LAG 0

20

cosφ=1

12

LEAD 04

Input
φ=-90° φ=0° φ=90°φ=180°
φ=270°[cosφ]
LAG
cosφ LEAD
LEAD cosφ
1
0
1
0
0

Input
φ=-90° φ=0° φ=90°φ=180°
φ=270°[cosφ]LEAD cosφ
LAG
cosφ LEAD
0
1
0
1
0

Power receiving

Power receiving

Power transmission

Power transmission(Tidal current measurement)
Display

LEAD 0‐LAG 0/LEAD 0‐LAG 0

Analog output[mA]

LAG 0

LEAD 0‐LAG 0/4‐20mA

20

cos180°=1
LEAD
0
LAG

12

cos0°=1
4

LEAD 0Input
φ=-90° φ=0° φ=90°φ=180°
φ=270°[cosφ]LEAD cosφ LAG,LEAD cosφ LAG
1
0
0
1
0
0

Input
φ=-90° φ=0° φ=90°φ=180°φ=270°[cosφ]LEAD cosφ LAG,LEAD cosφ LAG
1
0
0
1
0
0

Power receiving

Power receiving

Power transmission

67

Power transmission

SQLC-215-109(3) 331 Analog output specification setting 【With analog output (DC0‐5V or DC1‐5V)】
Set the specification of analog output (DC0‐5V／DC1‐5V).
Selection by ＋ and － , set value is updated by SET .
Default setting：DC1‐5V or DC0‐5V (Designation)
New setting
＋

＋

0‐5V
－

Current setting

1‐5V

Setting No.

－<Note> This setting does not exist at the time of ananalog output insulation product.

Setting factor

Analog output specification(4) 341 to 348 Analog output adjustment 【With analog output】
The bias and span of each analog output are adjusted.

RESET/SHIFT

341
Ao1 BIAS
adjustment

RESET/SHIFT

348
Ao4 SPAN
adjustment

RESET/SHIFT

342
Ao1 SPAN
adjustment

RESET/SHIFT

347
Ao4 BIAS
adjustment

RESET/SHIFT

343
Ao2 BIAS
adjustment

RESET/SHIFT

346
Ao3 SPAN
adjustment

RESET/SHIFT

344
Ao2 SPAN
adjustment
345
Ao3 BIAS
adjustment

RESET/
SHIFT◆ 341 AO(Analog output)1 bias adjustment,
343 AO(Analog output)2 bias adjustment,
345 AO(Analog output)3 bias adjustment,
347 AO(Analog output)4 bias adjustment.
The bias of each analog output is adjusted.
Setting range：±10.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .
◆ 342 AO(Analog output)1 span adjustment,
344 AO(Analog output)2 span adjustment,
346 AO(Analog output)3 span adjustment,
348 AO(Analog output)4 span adjustment.

Ao1 span adjustment

The span of each analog output is adjusted.
Setting range：±10.0％ (0.1％ step)
Selection by ＋ and － , set value is updated by SET .

68

SQLC-215-109

6. Specification
6.1

Specification and intrinsic error.

Input circuit
3-phase 3-wireSingle-phase 2-wireSingle-phase 3-wireZero-phase current

Measurement
item

Input
AC110V，220V common use.
AC 5A or AC 1A 50/60Hz Designation
AC100-200V (44)AC 5A or AC 1A 50/60Hz Designation
Rated sensitivity current value.
0.03A / 0.05A / 0.1A / 0.2A / 0.4A / 0.8A

Measurement range /Display specification

Current Voltage
input
input

Intrinsic error (45)
Maximum Minimum
Analog output
measur- measurDigital
Pulse output
ement
ement
display(46)

Current

Maximum demand (55), Demand,
Instant
AC5.00A to 30.0kA (76 range)

Active power

Maximum demand (55), Demand,
Instant
200W to 1000MW (Range select)
In voltage and current range.
One-side swing or both sideswings can be setting.

±1.0％

±0.5％

○

○

Reactive
power

LEAD,LAG 150var to 1000Mvar(Range select)
In voltage and current range.

±1.0％

±0.5％

○

○

Power factor

LEAD 0.500‐1.000‐LAG 0.500
or
LEAD 0.000‐1.000‐LAG 0.000
Range select

±2.0％

±2.0％

○

○

Frequency

45.0‐55.0Hz or
55.0‐65.0Hz or
45.0‐65.0Hz
Range select

○

±0.5％

±0.5％

○

○

○

±2.5％
(52)

±2.5％
(52)

○

○

±1.0％

±2.5％

○

±2.5％

±2.5％

○

±1.5％

±1.5％

○

Digital display
is ％ to a voltage
range.

±1.5％

±1.5％

○

Digital display
is ％ to a current
range.

Io method,
Igr method

Current leakage
AC0.03A to 0.8A (6 range)

Voltage
Distortion
factorCurrent

Harmonic nth
Voltageeffective
value
Fundamental-wave
effective
value

Current

0.0 to 20.0％ (Second
to 15th harmonic)
L1L2-L2L3 (53)0.0 to 100.0％ (Second
to 15th harmonic)
L1-L3 (54)AC150V to 750.0kV(34 range)
n=3,4,5,7,9,11,13,15,
and fundamental wave.
L1L2-L2L3 (53)AC5.00A to 30.0kA(76 range)
n=3,4,5,7,9,11,13,15,
and fundamental wave.
L1-L3 (54)

○

○

○

○

69

±0.5％

○

○

±1.0％

±0.5％

○

○

L1L2-L2L3-L3L1 line
change (47)
L1-L2-L3 phasechange (48)
Possible to set a
range of indicationand output apartfrom the CT ratio.

AC150V to 750.0kV (34 range)

○

±1.0％

Note

Voltage

Current
leakage (51)○

Only with leakage measurement

Range of an analog
output can be set as
indication
independently.
(49)(50)
Range of an analog
output can be set as
indication
independently. (50)
In case input isbelow 20％ of
voltage range or
below 2％ of current
range：cosφ=1.
(Output is cosφ=1
equivalence)
0.0Hz in case input
is below 20％ of
voltage range.
Output is a lower
limit value. (Lower
limit value -1％：％
for output span)％ for rated
sensitivity current
value.
An input (ZCT
primary) 3mA or lessbecomes zerodisplay.
Digital display
is ％ to 100％ of
distortion factors.

SQLC-215-109

Measurement
item

Measurement range /Display specification

Voltage
Harmonic nth
content
Current

Harmonic 5th Voltage
conversion
effective
value
Current
Harmonic 5th Voltage
conversion
content
Current

Watt-hour

var-hour

0.0 to 20.0％
n＝3,4,5,7,9,11,13,15
L1L2-L2L3 (53)0.0 to 100.0％
n＝3,4,5,7,9,11,13,15
L1-L3 (54)AC150V to 750.0kV(34 range)
L1L2-L2L3 (53)AC5.00A to 30.0kA(76 range)
L1-L3 (54)0.0 to 20.0％
L1L2-L2L3 (53)0.0 to 100.0％
L1-L3 (54)Current Voltage
input
input

○

○

○

○
○
○

Intrinsic error (45)
Maximum Minimum
Analog output
measur- measurDigital
Pulse output
ement
ement
display
(46)
±1.0％

±2.5％

○

±2.5％

±2.5％

○

±1.5％

±1.5％

○

±1.5％

±1.5％

○

±1.0％

±2.5％

○

±2.5％

±2.5％

○

Power
factor 1：±2.0％

Power
factor 1：±2.0％

Display：Integer, 5 digit.
Multiplier：Integral number times of 10.
Possible to indicate it to 3 decimal place.
Electric power is integrated. (Power receiving， PowerPower transmission)
factor 0.5：±2.5％
Display：Integer, 5 digit.
Multiplier：Integral number times of 10.
Possible to indicate it to 3 decimal place.
Integrating reactive power of power receiving.
(LAG･LEAD)
Integrating reactive power of powertransmission. (LAG･LEAD)

Power
factor 0.5：±2.5％

Power
factor 0：±2.5％

Power
factor 0：±2.5％

Power
factor 0.87：±2.5％

Power
factor 0.87：±2.5％

Note

Digital display is％ to 100% ofcontent.
Digital display is％ to a voltage
range.
Digital display is％ to a current
range.
Digital display is％ to 100% ofcontent.
Conformity withnormal watt-hourmeter.
Setting range ofpulse output unit
(kWh/pulse) isreferred to optionspecification.
Setting range ofpulse output unit
(kvarh/pulse) isreferred to optionspecification.

Note (44) The rated voltage of each phase and L3 phase is 100V. However, the full scale of a bar graph is 300V.
Note (45) If this unit directly measures an inverter output of cycle control, SCR phase angle control or PWM, an error
may increase due to its operation principle.
As for harmonic 5th conversion effective value and nth harmonic effective value, a display will be zero by
0.2％ or less of the measurement range. And, analog output constitutes lower limit output. At this time,a distortion factor, harmonic 5th conversion content rate, and nth harmonic content also become 0％ (analog
output is lower limit output). As for a distortion factor, harmonic 5th conversion effective value / content,
and nth harmonic effective value / content, fundamental-wave effective value constitutes zero by 3％ or lessof the measurement range.
Note (46) Analog output and pulse output are options.
Analog output of harmonic distortion factor and harmonic content：It becomes an upper limit output from a lower
limit output, for current 0 to 100％ and voltage 0 to 20％.
The analog output of current leakage constitutes an upper limit output to leakage sensitivity current value.
(Only with leakage measurement)
Note (47) Single-phase 3-wire：L1N-L3N-LIL3, Single-phase 2-wire：With no phase display.
Note (48) Single-phase 3-wire：L1-L3-N, Single-phase 2-wire：With no phase display.
Note (49) At the case of one side swing setting of bar graph. Digital meter measures reverse power to -15% full scale.
(Instantaneous power)
Note (50) Active power, reactive power display digit.
A full scale display is below 4000：4 digits display. A full scale display is 4000 or more：3 digits display.
Example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW
Please refer to an appendix table for details.
Note (51) Only with leakage measurement. However, current input product constitute only Io method.
Note (52) The error of ZCT is not included. However, leakage detection sensitivity current below 0.1A constitutes±0.0025A (ZCT primary).
Note (53) Single-phase 3-wire：L1N-L3N，single-phase 2-wire：With no phase display, line display.
Note (54) Single-phase 3-wire：L1-L3，single-phase 2-wire：With no phase display
Note (55) The maximum value (maximum demand current, others) minimum value can be checked in the maximum and theminimum measurement mode.

70

SQLC-215-109

Item
Bar graphdisplay

OperatingmethodIntervalsetting

The factor
in whichdisplaysetting ispossible

OptionSpecification
Bar graph display of the main-monitor factor is done. (Watt-hour and var-hour exclude)
A display of a sub-monitor factor can also be set.
Current, Voltage ：Effective value computing type.
Demand ammeter：Arithmetic method according with bimetallic type.
Demand active power meter：Arithmetic method according with bimetallic type, or average value within the demand
time
limit. (One side is selected by setting.)
Active power, Reactive power, Watt-hour, var-hour：Time-division multiplication method.
Power factor：Average value within instant measurement or the demand time limit. (One side is selectedby setting.) Calculates for active power and reactive power.
Frequency：Zero cross cycle computing type.
Current leakage ：Fundamental-wave effective-value arithmetic method.
Harmonic：FFT computing type.
0 second / 5 seconds / 10 seconds / 20 seconds / 30 seconds / 40 seconds / 50 seconds /Demand current
1 minute / 2 minutes / 3 minutes / 4 minutes / 5 minutes / 6 minutes / 7 minutes /8 minutes / 9 minutes / 10 minutes / 15 minutes / 20 minutes / 25 minutes / 30 minutes
Demand active(95％ time limit)
power
Harmonic
Average time limit：0 minute / 1 minute / 2 minutes / 5 minutes / 10 minutes
measurement/ 15 minutes / 30 minutes Average measurement.
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Demand active power, Reactive power, Power factor, Frequency, Current leakage,
Main monitor
Watt-hour(Power receiving, Power transmission),var-hour(Power receiving LAG/LEAD, Power transmission LAG/LEAD), Distortion factor(A, V)
Sub-monitor
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Active power, Reactive power, Power factor,(Left)
Current leakage
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Sub-monitor
Demand active power, Reactive power, var-hour(Power receiving LAG/LEAD,(Center)
Power transmission LAG/LEAD), Harmonic 5th conversion content(A, V), Harmonic nth content(A, V)
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Sub-monitor
Demand active power, Power factor, Frequency, Watt-hour(Power receiving, Power transmission),(Right)Fundamental-wave effective value(A, V), Harmonic 5th conversion content(A, V),
Harmonic nth effective value(A, V)
Voltage(L1L2-L2L3-L3L1), Current(L1-L2-L3), Demand current(L1-L2-L3), Active power,
Demand active power, Reactive power, Power factor, Frequency, Current leakage,
Bar graph
Distortion factor(A, V), Harmonic 5th conversion content(A, V), Harmonic nth content(A, V),
Fundamental-wave effective value(A, V), Harmonic 5th conversion effective value(A, V),
Harmonic nth effective value(A, V)
Analog output (4 sets)
Contact output (2 sets. Select of pulse output or alarm output or CPU error output.)
External operation change input (2 sets)● About power and reactive power full scale range selection
The power range and reactive power range is automatically decided in a current range and voltage range.
The full scale range of a bar graph can be selected out of the following values within a 40 to 115％ (range of power)and 30 to 115％ (range of reactive power), assuming that the rated power (VT ratio × CT ratio)(56) is 100％.
1.0 / 1.2 / 1.4 / 1.5 / 1.6 / 1.8 / 2.0 / 2.4 / 2.5 / 2.8 / 3.0 / 3.2 / 3.6 / 4.0 / 4.2 / 4.5 / 4.8 / 5.0 / 5.6 / 6.0 / 6.4 / 7.2 /7.5 / 8.0 / 8.4 / 9.0 / 9.6 ×10n
Example ) In case of VT ratio × CT ratio ＝ 1200kW
A full scale range can be selected from the following.
480 / 500 / 560 / 600 / 640 / 720 / 750 / 800 / 840 / 900 / 960 / 1000 / 1200
Note (56) Assume VT ratio is “2” for calculation in case of 220V input specifications and also assume CT ratio is
CT ratio ÷ 5 in case of 1A input specifications.

71

SQLC-215-109● Measurement is possible range.
Input (57)

Measurement factor
VoltageAC0‐150V [AC0‐300V]

Current

AC0‐5A[AC0‐1A]

0‐1kW (0‐200W)[0‐2kW (0‐400W)]
LEAD 1‐0‐LAG 1kvar(LEAD 200‐0‐LAG 200var)[LEAD 2‐0‐LAG 2kvar(LEAD 400‐0‐LAG 400var)]

Power

Reactive power

Measurement is possible range
Display
Analog output
101％ of meter full
101％ of output
scale.
span.
120％ of output
120％ of input rating.
span.
-1％ and 120％ of
120％ of input rating.
output span,
120％ of input rating.

LEAD 0‐1‐LAG 0

LEAD 0.000‐1‐LAG 0.000

LEAD 0.5‐1‐LAG 0.5

LEAD 0.490‐1‐LAG 0.490

45‐55Hz55‐65Hz45‐65HzAC0 to Rated sensitivity
current value

44.9‐55.1Hz
54.9‐65.1Hz
44.8‐65.2Hz

Power factor

Frequency
Current leakage
Effective
value

120％ of input rating.

Current

AC0‐5A[AC0‐1A]

VoltageAC0‐150V [AC0‐300V]

101％ of meter fullscale.

Current

0 to 100％120％

Voltage0 to 20％

100％

Harmonic
Rate ofcontent
Note (57) [120％ of 0.8A

] is the 300V input case. () is the 1A input case.

72-1％ and 120％ of
output span,
0％ and 100％ of
output span,-1％ and 101％ of
output span,-1％ and 101％ of
output span,
120％ of output
span.
120％ of output
span.
101％ of output
span.
120％ of output
span.
120％ of output
span.

Remarks

Instant,
DemandInstant,
DemandSQLC-215-109

6.2

Specification, Performance.

Item
Accuracy
Accuracy of bar graph
Influence by temperature
Conformity technicalstandard
Display updating time

Display device
Display composition

LCD view angle

Backlight

Auxiliary supply

Rush current(Time constant)

Input consumption VA

Overload capacity

Insulation resistance
JIS C 1102-1(58)
JIS C 1111

Withstand voltage (58)
JIS C 1102-1
JIS C 1111

Impulse withstandvoltage
JIS C 1111

Specification
Reference to measure specification and accuracy±10％ （％ for span）
23±10℃ within accuracy.
JIS C 1102-1, -2, -3, -4, -5, -7：1997 ，JIS C 1111：1989 ，JIS C 1216：1995 ，
JIS C 1263：1995 ，JIS C 8374：1991
About 1 second (Bar graph：0.25 seconds)(Current leakage measurement：Digital and bar graph is 2 seconds or less.
Harmonic measurement：Digital and bar graph is 10 seconds or less.)
Main monitor
5 digit, character height 11mm
Sub-monitor (Left)
4 digit, character height 6mm
LCD
Sub-monitor (Center),(Right)
5 digit, character height 6mm
Bar graph
20 dots
For upper installation
Upper view angle 10°，Lower view angle 60°，
Standard(For lower view)
Right and left view angle 60°
For lower installation
Upper view angle 60°，Lower view angle 10°，Special(For upper view)
Right and left view angle 60°
LED backlight：White
Always-on, Auto off (after 5 minutes without operating), Always-off. Setting is possible.
Backlight can select brightness from five steps of 1 to 5.
(1) AC85‐264V 50/60Hz 10VA (Rated voltage, AC100/110V，200/220V)
DC80‐143V
6W (Rated voltage, DC100/110V) for both AC and DC uses
(2) DC20‐56V
6W (Rated voltage, DC24/48V)
Rated voltage AC110V 2.2A or less (About 3.6ms)
Rated voltage AC220V 4.4A or less (About 3.6ms)
Rated voltage DC110V 1.6A or less (About 3.6ms)
Rated voltage DC24V
5.0A or less (About 2.0ms)
Rated voltage DC48V
9.9A or less (About 2.0ms)
Voltage circuit
0.1VA or less (110V) ，0.2VA or less (220V)
Current circuit
0.1VA or less (5A,1A)
Voltage circuit
2 times 10 seconds, 1.2 times continuation of rated voltage.
40 times 1 second, 20 times 4 seconds, 10 times 16 seconds,
Current circuit
1.2 times continuation of rated current.
1.5 times 10 seconds, 1.2 times continuation of rated voltage.
Auxiliary supply In case of DC110V, 1.5 times 10 seconds, 1.3 times continuation of rated
voltage.
Between electric circuits and case (Earth).
Between input and output and auxiliary supply.
Between analog output and pulse output andAbove 50MΩ at DC500V
alarm output.
Between pulse outputs.
Between alarm outputs.
Non-insulation (minus common)
Between analog outputs.
Designation
Above 50MΩ at DC500V megger
Between electric circuits and case (Earth).
AC2000V (50/60Hz) 1 minuteBetween input and output and auxiliary supply.
Between analog output and pulse output andalarm output.
AC1500V (50/60Hz) 1 minuteBetween pulse outputs.
Between alarm outputs.
Non-insulation (minus common)
Between analog outputs.
Designation
AC500V (50/60Hz) 1 minuteBetween auxiliary supply and case (Earth).
7kV 1.2/50μs Both positive and negative(Only with leakage measurement)
polarities, for each 3 time.
Between electric circuits and case (Earth).
6kV 1.2/50μs Both positive and negative(An analog output is excluded)polarities, for each 3 time.
5kV 1.2/50μs Both positive and negative
Between analog output and case (Earth).
polarities, for each 3 time.

Note (58) A voltage input and a leakage input are not insulated. Please use it combining ZCT.

73

SQLC-215-109

Item

Noise-capacity
ANSI C37.90a
JEA B-402
IEC801-2

Vibration, Shock
JIS C 1102-1
JIS C 0040，JIS C 0041
Construction
Material
Color
MassBlackout guarantee
Operating temperatureand humidity limits
Storage temperature
limitsSpecification(1) Oscillatory surge voltage
If a vibration damping waveform (1 to 1.5MHz, Peak voltage：2.5 to 3kV) is repeated andadded, a measurement error should be within 10％. And, there needs to be no malfunction.
Voltage input circuit (Normal / Common),Current input circuit (Common. Zero-phase current input circuit is excluded.)
Auxiliary supply circuit (Normal / Common)(2) Square wave impulse noise
If a noise (1μs, 100ns width) is repeated and added, a measurement error should bewithin 10％. And, there needs to be no malfunction.
Auxiliary supply circuit (Normal / Common) Over 1500V
Voltage input circuit (Normal / Common) Over 1500V
Current input circuit (Common) Over 1500V (Zero-phase current input circuit is excluded)
Pulse output (Common) Over 1000V
Alarm output (Common) Over 1000V
Operation input (Common) Over 1000V
Analog output circuit (Induction) Over 1000V(3) Electric wave noise
If intermittence irradiation of the electric wave of a 150,400,900MHz band is done by(5W, 1m), a measurement error should be within 10％.
And, there needs to be no malfunction.
(4) Electrostatic noise
It is within 10％ of errors in 8kV electrostatic noise at the case of an energization.
There needs to be no 10kV damage at the case of a non-energization.
Condenser charge form.
Vibration：Single amplitude 0.15mm, 10 to 55Hz, Each minute octave in 5 times sweep
Shock：490m/s２ It is each 3 times to the X,Y,Z direction.
Dimension：110mm(Width)×110mm(Height)×103.5mm(Depth) Body diameter：99mmφ
With terminal coverCase，Cover：ABS(V-0) ，Terminal block：PBT ，Terminal cover：Polycarbonate
Black (Munsell N1.5)
Approx. 600g
Maximum value, Minimum value, Integrates value and each setting value.
Data hold by nonvolatile memory.
-10 to +55℃, 30 to 85％ RH, Non condensing.
-25 to +70℃

74

SQLC-215-109

6.3 Option
Item

Specification
4 circuits
Analog output non-insulation type
Analog output insulation typeDC4‐20mA(Below 550Ω)
DC4‐20mA (Below 550Ω)
DC0‐1mA(Below 10kΩ)DC1‐5V (Over 600Ω)
OutputDC0‐5V/1‐5V (Over 600Ω)One of them is designated by identicalspecification
DC0‐10V(Over 2kΩ)
ratings.
One of them is designated by identicalratings.
Voltage(L1L2-L2L3-L3L1)，Current(L1-L2-L3)，Demand current(L1-L2-L3)，Active power，Demand active power，Reactive power，Power factor，Frequency，Current leakage (Only with leakage measurement)，Distortion factor(A，V)，
Fundamental-wave effective value (Maximum value of each A/V phase)，
Output factor
Harmonic 5th conversion content (Maximum value of each A/V phase)，
Harmonic 5th conversion effective value (Maximum value of each A/V phase)，
Harmonic nth content (Maximum value of each A/V phase)，
nth harmonic effective value (Maximum value of each A/V phase)
1 second or less (Time within ±1％ of final constant value.)Response time
Leakage measurement is 2 seconds or less. Harmonic measurement is 10 seconds or less.
Output ripple
Within the double precision of accuracy (％ for output span)
Possible output ：Watt-hour or var-hour.
Output form
：Optical MOS-FET relay. 1a contact
Contact capacity：AC，DC125V, 70mA (Resistance load, inductive load)
Pulse width：250±10ms (Output pulse width when the output pulse period of rated active power
constitutes speed more than 2 pulse/second by setting of an voltage
measurement range, a current-measurement range, and an output pulse unit
is set to 100 to 130ms.)
Setting of output pulse unit is possible by the next range.
■ 3-phase 3-wire：Full load power (kW,kvar)＝√3×Rated voltage(V)×Rated current(A)×10-3■ Single-phase 3-wire：Full load power (kW,kvar)＝2×Rated voltage(V)×Rated current(A)×10-3■ Single-phase 2-wire：Full load power (kW,kvar)＝Rated voltage(V)×Rated current(A)×10-3
Full load power (kW,kvar)
Output pulse unit. kWh(kvarh)/pulse
Multiplying factor
Below 1
0.1
0.01
0.001
0.00010.01 (59)Over 1
Below 10
1
0.1
0.01
0.001
0.1
Over 10
Below 100
10
1
0.1
0.01
1
Over 100
Below 1,000
100
10
1
0.1
10
Over 1,000
Below 10,000
1,000
100
10
1
100
Over 10,000 Below 100,000
10,000
1,000
100
10
1,000
Over 100,000 Below 1,000,000
100,000
10,000
1,000
100
10,000
Number of output

Analog output

Pulse output(60)

Note(59) Although multiplying factor is 0.01, a multiplying factor display will be 0.1.
(The place by the integer is 4 digits display. An enlarged display is 4 digits below decimal point.)
Note(60) 2 outputs combination is possible, for pulse output and alarm output and CPU error output.
(CPU error output is 1 point only)

75

SQLC-215-109

Item

Specification
Alarm factor：Demand current, Demand active power, Current leakage，harmonic 5th conversion content,
Harmonic nth content, Distortion factor, Voltage, Alarm OFF.
Possible to setting one of them.
Reset form ：Automatic reset or Manual reset (Setting)
Output contact ：No-voltage a contact (OR of each phase detection)
Contact capacity：AC250V 8A, DC125V 0.3A (Resistance load) AC250V 2A, DC125V 0.1A (Inductive load)
Alarm factor
Item
Specification
Demand measurement value ≧ Upper limit setting value,
Demand current, Function
Alarm display, Alarm output.
Demand active
Setting accuracy±1.0％ (％ for full scale )power
Setting range
5 to 100％ to the maximum scale. (1％ step)Sensitivity current 50 to 100％ (％ for rated sensitivity current value)
Current leakage Rated sensitivity
0.03A / 0.05A / 0.1A / 0.2A / 0.4A / 0.8A
current value
Only withOperating time
Time-delay type (It exceeds 0.1 seconds and is 2 or less seconds.)
leakageThefunctional test of the current-leakage detection by the test
measurement
Test function
mode is possible.
Measurement value ≧ Upper limit setting value,
Alarm output
Function
Alarm display, Alarm output. (Detects by the maximum phase)(61)
Setting accuracy
Current：±2.5％, Voltage：±1.0％, ％ to 100％ of content rate.
Harmonic 5th
Harmonic 5th conversion content, Harmonic nth content
conversion
Current (n=3,4,5,7,9,11,13,15), Distortion factor.
content
5 to 100％ (1％ step)
Setting range
Harmonic 5th conversion content, Harmonic nth content
Harmonic nth
Voltage (n=3,4,5,7,9,11,13,15), Distortion factor.
content
1.0 to 20.0％ (0.1％ step)
Average value mode：It will detect, if a par measurement value
Distortion
becomes beyond the above setting.
factor
Detection
Inverse-time-delay mode：It detects in the inverse-time-delay
characteristics
characteristics of instantaneous value (only a harmonic 5th
conversion content is possible).
Measurement value ≧ Upper limit setting value,
Alarm display, Alarm output. (Detects by the maximum phase)
Function
Measurement value ≦ Lower limit setting value,
Voltage
Alarm display, Alarm output. (Detects by the minimum phase)
Setting accuracy±1.0％ (％ for full scale)Setting range
Using a full scale as 150％. 30 to 150％ (1％ step)
Detection item (Self-diagnostics item)
Contact composition
Contact capacity(1) Watchdog timer (Inside, outside)
CPU error(2) RAM check error
AC250V 5A, DC125V 0.2A
output(3) A/D-conversion error(Resistance load)b contact(61)
At the case of error detection and not applying for
AC250V 1.5A, DC125V 0.1A
auxiliary supply, an output is ON.
(Inductive load)
It becomes OR output of a detection item.
Number of circuits
2 circuits, A function (four types) is changed by setting.
Four types of following functions can be operated by adding a voltage signal
Functionfrom the outside in addition to switch operation.
Alarm reset
Alarm output is reset (output OFF).
Maximum / Minimum
The maximum/minimum value is reset (it updates to the instantaneous value atvalue reset
the time).
Measurement factor
External
The measurement display element of the main monitoring is changed.
changeoperation
Phase change
A phase/lines display of all the current/voltage currently displayed is changed.
input
Minimum operation pulse
300ms ，Continuation applying is possible.
width
Input rating becomes the same as that of auxiliary supply.
(1) AC100/110V 0.4VA，AC200/220V 1.4VA，DC100/110V 0.4W AC DC two ways.
Rated input
Contact capacity：About 3mA (AC,DC100/110V) ，About 6mA (AC200/220V)(2) DC24V 0.3W，DC48V 1.2W
Contact capacity : About 10mA (DC24V) ，About 20mA (DC48V)
Note (61) 2 outputs combination is possible, for pulse output and alarm output and CPU error output.
(CPU error output is 1 point only)
76

SQLC-215-109● Caution on the use of external display selection input (option)
External power consumption is 0.4VA at AC110V or 1.4VA at AC220V or 0.4W at DC110V.
In case a relay or a switch is used for power-supply supply, please use the thing of about
1mA of the minimum application loads.

P N
5
6
28
29

7. Maintenance and check7.1 Trouble shooting
Symptoms

Does not display.

Measuring value error
is noticeable
Measurement displayof current leakage Igrwill be "----".
Analog output is not
outputted.
Pulse output is not
outputted.
Alarm output does not
return.

Possible causes
The power supply is not supplied.
(Not connected. or voltage is low)
Measurement display ON/OFF setting is set toOFF.
Trouble of device.
Range is not set correctly.
Out of range in rated frequency (45‐65Hz).
Cycle control, SCR phase angle control, PWM,
or other inverter output is measured.
The direction of the connection to ZCT or thisproduct is reverse.
The cable run constitutes a negative phasesequence.
Analog output is set to OFF or measurement
factor is set to OFF.
Pulse output is set to OFF or measurement
factor is set to OFF.

Remedial measures
Check the auxiliary supply.
Again, a power supply is supplied.

The return method is a "manual reset".

Please check a setting

Setting check.
Replace the device.
Please set again.
Cannot be used.
Cannot be used.
Please do connection correctly.
Please change a setting (No.163) into “1：one phase earth (negative phase sequence)”.
Please check a setting
Please check a setting

7.2 CheckBe careful with the following items periodically.
(1) Check if the measuring values, scale digits, units, etc. are displayed correctly.
(2) Check if LCD is free of a color change, breakage of case, or other defects.
(3) Check the unit for loose wiring and loose mounting screws.
(4) Please remove, if dust has adhered to the product.
7.3 Maintenance(1) Please do not make a wiring change of the input and auxiliary supply in an energization.
(2) In case you check an output in an energization, please warn to be unable to touch output wiring and a humanbody to an input and an auxiliary supply terminal. (Please do not short-circuit a voltage output.)(3) Please check a connection diagram, in case you check the voltage of input, output, and auxiliary supply.
(4) If a name-plate is wiped with solvents (an alcoholic nature), a display item may disappear.
Please wipe off with the dry cloth.

77

SQLC-215-109

7.4 Test
In case this product is tested, a special setting or operation is not needed fundamentally.
However, the following test should operate it along with each process.
(1) Alarm-output test
Even if this product does not have input, it can perform ON/OFF test of an alarm output (relay-contact output).
Operation is performed by alarm 1 test and alarm 2 test in the setting mode 1.
Please refer to "5.3.1 Setting mode 1 (2) alarm-output setup" about the details of operation.
In case an alarm-output factor is a current leakage (Io/Igr).
While pushing SET, the test input exceeding rated sensitivity current value is applied to the zero-phasecurrent input circuit inside a product. If the time that is pushing SET exceeds the operating time (it exceeds
time-delay form 0.1 seconds, and is 2 or less seconds) of a current leakage, an alarm will occur.
And, test input will become off if SET is detached. Please be sure to apply rated voltage to voltage input
terminal (between P1-P2) at this time.
In addition, an alarm return constitutes action corresponding to a return method setting (automatic reset /manual reset) of each alarm output.
< Operating process >・Alarm 1 test (Setting № 124)
(111)
(121)
(124)
Press SET for 3 seconds → Press MODE → Press RESET/SHIFT for 3 times → While pushing SET , thealarm 1 output is ON.
・Alarm 2 test (Setting № 128)
(111)
(121)
(128)
Press SET for 3 seconds → Press MODE → Press RESET/SHIFT for 7 times → While pushing SET , thealarm 2 output is ON.
・In case an alarm-output factor is current leakage (Io/Igr) ･･･ Example ) ALARM 2
(111)
(121)
(128)
Press SET for 3 seconds → Press MODE → Press RESET/SHIFT for 7 times
If it continues pushing SET for a long time from the current-leakage operating time,
alarm 2 output is ON.
(2) Harmonic time-interval test
Please test by the following operation about the time interval of a harmonic. An error becomes large, incase it tests without performing the following operation.
Operation is performed with the average time interval in the setting mode 1.
Please refer to "5.3.1 setting mode 1 (4) harmonic detection setting" for operation details.
< Operating process > (Setting № 14A)
(111)
(141)
(14A)
Press SET for 3 seconds → Press MODE for 3 times → Press RESET/SHIFT for 9 times
Please press SET simultaneously with a test start.
Please press DISPLAY and return to a measurement screen.

7.5 Storage(1) Please avoid storage in the next space. Low temperature, high temperature, high humidity, and sunny place.
(2) The aluminum electrolytic capacitor is used for a product. Please do the energization of the power supply
within one year after shipment.
7.6 Countermeasures against troubles
As our principle, we recall product in question and repair it. If judged as product failure, have a contact withus or sales agent for repairing work (Also have a contact with us or sales agent for specification change).
Product failure which we are not responsible for (When responsibility in manufacturing process is not recognized,
when product is disassembled/remodeled, in case of false operation by customer, etc.) is beyond our warranty.

78

SQLC-215-109
Appendix table 1-1
Active power range, watt-hour multiplier rate table (3-phase 3-wire)
V range
Multiplierrate
A range
5A[W]
5.00 M

6A

6.00 M

7.5A

×100

8.00 M

10A

10.00 M

12A

12.00 M

15A

15.00 M

20A

20.00 M

25A

25.00 M

30A

30.00 M

40A

40.0 M

50A

50.0 M

60A

60.0 M

75A

75.0 M
80.0 M

100A

100.0 M

120A

120.0 M

150A

150.0 M

200A

200.0 M

500.0kV

375.0kV

300.0kV

250A

250.0 M

300A

300.0 M

400A

400 M

500A

500 M

600A

600 M

[W]
3600 k(3455)
4.20 M(4.15)
5.60 M(5.18)
5.60 M(5.53)
7.20 M(6.91)
8.40 M(8.29)
10.00 M
(10.36)14.00 M
(13.82)18.00 M
(17.27)20.00 M
(20.73)28.00 M
(27.64)36.00 M
(34.55)42.0 M(41.5)
56.0 M(51.8)
56.0 M(55.3)
72.0 M(69.1)
84.0 M(82.9)
100.0 M
(103.6)140.0 M(138.2)180.0 M
(172.7)200.0 M
(207.3)280.0 M
(276.4)360.0 M
(345.5)
420 M
(415)
560 M
(518)
560 M
(553)
640 M
(622)
720 M
(691)
840 M
(829)

[W]
2500 k[W]
2000 k

3000 k2400 k

4.00 M(3.75)
4.00 M

3000 k

5.00 M

4.00 M

6.00 M

4.80 M

7.50 M

6.00 M

10.00 M

8.00 M

14.00 M
(12.50)15.00 M

10.00 M
12.00 M

20.00 M

16.00 M

25.00 M

20.00 M

30.00 M

24.00 M

40.0 M(37.5)
40.0 M

30.00 M

3200 k

32.00 M

50.0 M

40.0 M

60.0 M

48.0 M

75.0 M

60.0 M

100.0 M

80.0 M

140.0 M
(125.0)150.0 M

100.0 M
120.0 M

200.0 M

160.0 M

250.0 M

200.0 M

300.0 M

240.0 M

400 M
(375)
400 M

300.0 M

450 M

360.0 M

500 M

400 M

600 M

480 M

1500A

750 M

600 M

1600A

800 M

640 M

900 M

720 M

1000 M

800 M

750A
800A
900A
×100007.50 M

8A

80A
×1000

750.0kV

1000A
1200A

1800A
2000A
2500A
3000A

255.0kV

210.0kV

180.0kV

150.0kV

105.0kV

90.0kV

45.0kV

30.0kV(VT550000/110V) (VT380000/110V) (VT275000/110V) (VT220000/110V) (VT187000/110V) (VT154000/110V) (VT132000/110V) (VT110000/110V) (VT77000/110V) (VT66000/110V) (VT33000/110V) (VT22000/110V) Multiplier

750 M
800 M
900 M
1000 M

320.0 M

1000 M

[W]
1800 k(1700)
2000 k(2040)
2500 k(2550)
3000 k(2720)
3600 k(3400)
4.20 M(4.08)
5.60 M(5.10)
7.20 M(6.80)
9.00 M(8.50)
10.00 M
(10.20)14.00 M
(13.60)18.00 M
(17.00)20.00 M
(20.40)28.00 M
(25.50)28.00 M
(27.20)36.00 M
(34.00)42.0 M(40.8)
56.0 M(51.0)
72.0 M(68.0)
90.0 M(85.0)
100.0 M
(102.0)140.0 M
(136.0)180.0 M
(170.0)200.0 M
(204.0)280.0 M
(255.0)280.0 M
(272.0)320.0 M
(306.0)360.0 M
(340.0)
420 M
(408)
560 M
(510)
560 M
(544)
640 M
(612)
720 M
(680)
900 M
(850)

[W]
1400 k[W]
1200 k[W]
1000 k1800 k(1680)
2400 k(2100)
2400 k(2240)
2800 k1500 k(1440)
1800 k1200 k2000 k(1920)
2400 k1600 k3600 k(3360)
4.20 M

3000 k(2880)
3600 k2400 k

5.60 M

4.80 M

4.00 M

7.20 M(7.00)
8.40 M

6.00 M

5.00 M

7.20 M

12.00 M
(11.20)14.00 M
18.00 M
(16.80)24.00 M
(21.00)24.00 M
(22.40)28.00 M

15.00 M
(14.40)18.00 M

12.00 M

20.00 M
(19.20)24.00 M

16.00 M

36.00 M
(33.60)42.0 M

30.00 M
(28.80)36.00 M

24.00 M

56.0 M

48.0 M

40.0 M

72.0 M(70.0)
84.0 M

60.0 M

50.0 M

72.0 M

120.0 M
(112.0)140.0 M
180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

150.0 M
(144.0)180.0 M

120.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

160.0 M

360.0 M
(336.0)420 M

300.0 M
(288.0)360.0 M

240.0 M

450 M
(448)
560 M
(504)
560 M

400 M
(384)
450 M
(432)
480 M

320.0 M

720 M
(700)
840 M

600 M

500 M

720 M

600 M

[W]
600 k

[W]
300.0 k720 k360.0 k240.0 k

900 k

450 k

300.0 k[W]
200.0 k

960 k

480 k

320.0 k1200 k

600 k

400 k

1800 k(1680)
2400 k(2100)
2800 k1500 k(1440)
1800 k

720 k

480 k

900 k

600 k

2400 k1200 k800 k3000 k1500 k1000 k

6.00 M

3600 k(3500)
4.20 M

3600 k1800 k1200 k

9.60 M

8.00 M

5.60 M

4.80 M

2400 k1600 k12.00 M

10.00 M

7.20 M(7.00)
8.40 M

6.00 M

3000 k2000 k

7.20 M

3600 k2400 k12.00 M
(10.50)12.00 M
(11.20)14.00 M

9.00 M

4.50 M

3000 k

9.60 M

4.80 M

3200 k2000 k

3000 k

15.00 M

12.00 M

6.00 M

4.00 M

15.00 M
(14.40)18.00 M

7.20 M

4.80 M

9.00 M

6.00 M

24.00 M

12.00 M

8.00 M

30.00 M

15.00 M

10.00 M

60.0 M

36.00 M
(35.00)42.0 M

36.00 M

18.00 M

12.00 M

96.0 M

80.0 M

56.0 M

48.0 M

24.00 M

16.00 M

120.0 M

100.0 M

72.0 M(70.0)
84.0 M

60.0 M

30.00 M

20.00 M

72.0 M

36.00 M

24.00 M

120.0 M
(105.0)120.0 M
(112.0)140.0 M
(126.0)140.0 M

90.0 M

45.0 M

30.00 M

5000A
6000A
7500A

20.00 M

30.00 M

150.0 M

180.0 M
200.0 M

300.0 M

360.0 M
400 M

×100

96.0 M

48.0 M

32.00 M

120.0 M(108.0)120.0 M

56.0 M(54.0)
60.0 M

36.00 M

180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

150.0 M
(144.0)180.0 M

72.0 M

48.0 M

90.0 M

60.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

96.0 M

64.0 M

120.0 M(108.0)120.0 M

80.0 M

360.0 M
(350.0)420 M

300.0 M

150.0 M

100.0 M

360.0 M

180.0 M

120.0 M

40.0 M

72.0 M

×1000

800 M

560 M

480 M

240.0 M

160.0 M

1000 M

720 M
(700)
840 M

600 M

300.0 M

200.0 M

720 M

360.0 M

240.0 M

900 M

450 M

300.0 M

960 M

480 M

320.0 M

560 M
(540)
600 M

360.0 M

12000A

720 M

480 M

15000A

900 M

600 M

8000A
9000A
10000Arate

×10

18.00 M
(16.80)24.00 M
(21.00)28.00 M

960 M

4000A

1500 k

[W]
720 k
(700)
900 k
(840)
1200 k(1050)
1200 k(1120)
1400 k

400 M

800 M

20000A

×1000030000A<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW79<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a four-step setup
is possible) is set as the fastest.
The output pulse width of a pulse output is
set to 100 to 130ms. (Usually 240 to 260ms)

SQLC-215-109
Appendix table 1-2
Active power range, watt-hour multiplier rate table (3-phase 3-wire)
V range
Multiplierrate
A range
5A
6A
7.5A
8A
10A
12A
15A
20A
25A

×10

30A
40A
50A
60A
75A
80A
100A
120A
150A
200A
×100

250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A

×1000

2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10000A
12000A
15000A

×1000020000A
30000A

25.00kV

24.00kV

18.00kV

18.00kV

15.00kV

9000V

4500V

3000V

2400V

1500V

1200V

600V

(VT18400/110V) (VT16500/110V) (VT13800/110V) (VT13200/110V) (VT11000/110V) (VT6600/110V) (VT3300/110V) (VT2200/110V) (VT1650/110V) (VT1100/110V) (VT880/110V) (VT480/110V) Multiplier[W]
180.0 k(167)200.0 k(201)
280.0 k(251)
280.0 k(268)
360.0 k
(335)
420 k
(401)
560 k
(502)
720 k
(669)
840 k
(836)
1000 k(1004)
1400 k(1338)
1800 k(1673)
2000 k(2007)
2800 k(2509)
2800 k(2676)
3600 k(3345)
4.20 M(4.01)
5.60 M(5.02)
7.20 M(6.69)
8.40 M(8.36)
10.00 M
(10.04)14.00 M
(13.38)18.00 M
(16.73)20.00 M
(20.07)28.00 M
(25.09)28.00 M
(26.76)32.00 M
(30.11)36.00 M
(33.45)42.0 M(40.1)
56.0 M(50.2)
56.0 M(53.5)
64.0 M(60.2)
72.0 M(66.9)
84.0 M(83.6)
100.0 M
(100.4)140.0 M
(133.8)180.0 M(167.3)200.0 M
(200.7)280.0 M
(250.9)280.0 M
(267.6)320.0 M
(301.1)360.0 M
(334.5)
420 M
(401)
560 M
(502)
720 M
(669)

[W]
150.0 k180.0 k240.0 k(225.0)240.0 k300.0 k360.0 k
450 k
600 k
750 k
900 k
1200 k1500 k1800 k2400 k(2250)
2400 k
3000 k3600 k
4.50 M
6.00 M
7.50 M
9.00 M
12.00 M
15.00 M
18.00 M
24.00 M
(22.50)24.00 M
28.00 M
(27.00)30.00 M
36.00 M
45.0 M
48.0 M
56.0 M(54.0)
60.0 M
75.0 M
90.0 M
120.0 M
150.0 M
180.0 M
240.0 M
(225.0)240.0 M
280.0 M
(270.0)300.0 M
360.0 M
450 M
600 M
900 M

[W]
140.0 k(125.5)160.0 k(150.5)200.0 k(188.2)200.0 k(200.7)280.0 k(250.9)320.0 k(301.1)
400 k
(376)
560 k
(502)
640 k
(627)
800 k
(753)
1000 k(1004)
1400 k(1255)
1600 k(1505)
2000 k(1882)
2000 k(2007)
2800 k(2509)
3200 k(3011)
4.00 M(3.76)
5.60 M(5.02)
6.40 M(6.27)
8.00 M(7.53)
10.00 M
(10.04)14.00 M
(12.55)16.00 M
(15.05)20.00 M(18.82)20.00 M
(20.07)24.00 M
(22.58)28.00 M
(25.09)32.00 M
(30.11)40.0 M(37.6)
42.0 M(40.1)
48.0 M(45.2)
56.0 M(50.2)
64.0 M(62.7)
80.0 M(75.3)
100.0 M
(100.4)140.0 M
(125.5)160.0 M
(150.5)200.0 M
(188.2)200.0 M
(200.7)240.0 M
(225.8)280.0 M
(250.9)320.0 M
(301.1)
400 M
(376)
560 M
(502)
800 M
(753)

[W]
120.0 k[W]
100.0 k[W]
60.0 k[W]
30.00 k[W]
20.00 k[W]
15.00 k[W]
10.00 k150.0 k(144.0)180.0 k120.0 k72.0 k36.00 k24.00 k18.00 k12.00 k9.60 k

150.0 k90.0 k

45.0 k30.00 k15.00 k12.00 k200.0 k(192.0)240.0 k160.0 k96.0 k

48.0 k

32.00 k24.00 k(22.50)24.00 k16.00 k200.0 k120.0 k60.0 k

40.0 k30.00 k20.00 k14.00 k(12.80)16.00 k240.0 k150.0 k(144.0)180.0 k72.0 k

48.0 k36.00 k24.00 k90.0 k

60.0 k

45.0 k30.00 k20.00 k(19.20)24.00 k300.0 k(288.0)360.0 k300.0 k[W]
8.00 k

480 k

400 k

240.0 k120.0 k80.0 k

60.0 k

40.0 k

32.00 k

600 k

500 k

300.0 k150.0 k100.0 k75.0 k

50.0 k

40.0 k

720 k

600 k

360.0 k180.0 k120.0 k90.0 k

60.0 k

48.0 k

960 k

800 k

480 k

240.0 k160.0 k120.0 k80.0 k

64.0 k1200 k1000 k600 k300.0 k200.0 k150.0 k100.0 k80.0 k1500 k(1440)
1800 k1200 k720 k360.0 k240.0 k180.0 k120.0 k96.0 k1500 k

900 k

450 k

300.0 k150.0 k120.0 k2000 k(1920)
2400 k1600 k

960 k

480 k

320.0 k240.0 k(225.0)240.0 k160.0 k2000 k1200 k

600 k

400 k

300.0 k200.0 k140.0 k(128.0)160.0 k3000 k(2880)
3600 k2400 k

720 k

480 k

360.0 k240.0 k3000 k1500 k(1440)
1800 k

900 k

600 k

450 k

300.0 k200.0 k(192.0)240.0 k4.80 M

4.00 M

2400 k1200 k

800 k

600 k

400 k

320.0 k6.00 M

5.00 M

3000 k1500 k1000 k

750 k

500 k

400 k

7.20 M

6.00 M

3600 k1800 k1200 k

900 k

600 k

480 k

9.60 M

8.00 M

4.80 M

2400 k1600 k1200 k

800 k

640 k

12.00 M

10.00 M

6.00 M

3000 k2000 k1500 k1000 k800 k15.00 M
(14.40)18.00 M

12.00 M

7.20 M

3600 k2400 k1800 k1200 k960 k15.00 M

9.00 M

4.50 M

3000 k1500 k1200 k

20.00 M
(19.20)24.00 M
(21.60)24.00 M

16.00 M

9.60 M

4.80 M

3200 k2400 k(2250)
2400 k1600 k18.00 M

12.00 M
(10.80)12.00 M

5.60 M(5.40)
6.00 M

3600 k1800 k

4.00 M

2800 k(2700)
3000 k1400 k(1280)
1500 k(1440)
1600 k30.00 M
(28.80)36.00 M

24.00 M

15.00 M
(14.40)18.00 M

7.20 M

4.80 M

3600 k2400 k

9.00 M

20.00 M

30.00 M

2000 k2000 k(1920)
2400 k

6.00 M

4.50 M

3000 k

9.60 M

6.40 M

4.80 M

3200 k12.00 M
(10.80)12.00 M

7.20 M
8.00 M

5.60 M(5.40)
6.00 M

3600 k

40.0 M

20.00 M
(19.20)24.00 M
(21.60)24.00 M

4.00 M

2800 k(2560)
3000 k(2880)
3200 k

50.0 M

30.00 M

15.00 M

10.00 M

7.50 M

5.00 M

4.00 M

72.0 M

60.0 M

36.00 M

18.00 M

12.00 M

9.00 M

6.00 M

4.80 M

96.0 M

80.0 M

48.0 M

24.00 M

16.00 M

12.00 M

8.00 M

6.40 M

120.0 M

100.0 M

60.0 M

30.00 M

20.00 M

15.00 M

10.00 M

8.00 M

150.0 M
(144.0)180.0 M

120.0 M

72.0 M

36.00 M

24.00 M

18.00 M

12.00 M

9.60 M

150.0 M

90.0 M

45.0 M

30.00 M

24.00 M
(22.50)24.00 M

15.00 M

12.00 M

16.00 M

14.00 M(12.80)15.00 M
(14.40)16.00 M

40.0 M(38.4)
45.0 M(43.2)
48.0 M

32.00 M

60.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

36.00 M

160.0 M

96.0 M

48.0 M

32.00 M

180.0 M

120.0 M(108.0)120.0 M

56.0 M(54.0)
60.0 M

36.00 M
40.0 M

28.00 M
(27.00)30.00 M

72.0 M

48.0 M

36.00 M

24.00 M

90.0 M

60.0 M

45.0 M

30.00 M

20.00 M
(19.20)24.00 M

200.0 M
240.0 M

18.00 M
20.00 M

300.0 M
(288.0)360.0 M

300.0 M

150.0 M
(144.0)180.0 M

480 M

400 M

240.0 M

120.0 M

80.0 M

60.0 M

40.0 M

32.00 M

720 M

600 M

360.0 M

180.0 M

120.0 M

90.0 M

60.0 M

48.0 M<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW

80

rate
[W]
4.50 k
(4.36)
5.60 k(5.24)
7.20 k(6.55)
7.20 k
(6.98)
9.00 k(8.73)
12.00 k(10.47)×0.1
14.00 k(13.09)18.00 k(17.45)24.00 k(21.82)28.00 k(26.18)36.00 k(34.91)45.0 k
(43.6)
56.0 k
(52.4)
72.0 k
(65.5)
72.0 k(69.8)
90.0 k(87.3)
120.0 k(104.7)×1
140.0 k(130.9)180.0 k(174.5)240.0 k(218.2)280.0 k(261.8)360.0 k(349.1)
450 k
(436)
560 k
(524)
720 k
(655)
720 k
(698)
800 k
(785)
900 k
(873)
1200 k(1047)
×10
1400 k(1309)
1400 k(1396)
1600 k(1571)
1800 k(1745)
2400 k(2182)
2800 k(2618)
3600 k(3491)
4.50 M(4.36)
5.60 M(5.24)
6.40 M(6.55)
7.20 M(6.98)
8.00 M(7.85)
8.40 M(8.73)
10.00 M
(10.47)×100
14.00 M
(13.09)18.00 M
(17.45)28.00 M
(26.18)×1000<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a four-step setup
is possible) is set as the fastest.
The output pulse width of a pulse output is
set to 100 to 130ms. (Usually 240 to 260ms)

SQLC-215-109
Appendix table 1-3
Active power range, watt-hour multiplier rate table (3-phase 3-wire)
V range
Multiplierrate
A range
5A
6A
7.5A
8A
10A
×0.1

12A
15A
20A
25A
30A
40A
50A
60A
75A
80A
100A
120A

×1

150A
200A
250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A

×10

1500A
1600A
1800A
2000A
2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10000A×100

12000A
15000A
20000A×1000

30000A

600V
600V
500V
300V
(VT460/110V) (VT440/110V) (VT380/110V) (VT220/110V)
[W]
[W]
[W]
[W]
4.20 k
4.00 k3600 k
2000(4.18)(3455)
5.60 k4.80 k
4.20 k
2400(5.02)(4.15)
6.40 k
6.00 k
5.60 k
3000(6.27)(5.18)
7.20 k
6.40 k
5.60 k
3200(6.69)(5.53)
8.40 k
8.00 k
7.20 k
4.00 k
(8.36)(6.91)
10.00 k9.60 k
8.40 k4.80 k
(10.04)(8.29)
14.00 k12.00 k10.00 k6.00 k(12.55)(10.36)18.00 k16.00 k14.00 k8.00 k
(16.73)(13.82)24.00 k20.00 k18.00 k10.00 k(20.91)(17.27)28.00 k24.00 k20.00 k12.00 k(25.09)(20.73)36.00 k32.00 k28.00 k16.00 k(33.45)(27.64)42.0 k
40.0 k36.00 k20.00 k(41.8)(34.55)56.0 k
48.0 k
42.0 k
24.00 k(50.2)(41.5)
64.0 k
60.0 k
56.0 k30.00 k(62.7)(51.8)
72.0 k
64.0 k
56.0 k
32.00 k(66.9)(55.3)
84.0 k
80.0 k
72.0 k
40.0 k(83.6)(69.1)
100.0 k96.0 k
84.0 k
48.0 k(100.4)(82.9)
140.0 k120.0 k100.0 k60.0 k(125.5)(103.6)180.0 k160.0 k140.0 k80.0 k(167.3)(138.2)240.0 k200.0 k180.0 k100.0 k(209.1)(172.7)280.0 k240.0 k200.0 k120.0 k(250.9)(207.3)360.0 k320.0 k280.0 k160.0 k(334.5)(276.4)420 k
400 k
360.0 k200.0 k(418)(345.5)
560 k
480 k
420 k
240.0 k
(502)
(415)
640 k
600 k
560 k
300.0 k
(627)
(518)
720 k
640 k
560 k
320.0 k
(669)
(553)
800 k
720 k
640 k
360.0 k
(753)
(622)
840 k
800 k
720 k
400 k
(836)
(691)
1000 k
960 k
840 k
480 k
(1004)(829)
1400 k1200 k1000 k600 k(1255)(1036)
1400 k1400 k1200 k640 k(1338)(1280)(1105)
1600 k1500 k1400 k720 k(1505)(1440)(1244)
1800 k1600 k1400 k800 k(1673)(1382)
2400 k2000 k1800 k1000 k(2091)(1727)
2800 k2400 k2000 k1200 k(2509)(2073)
3600 k3200 k2800 k1600 k(3345)(2764)
4.20 M
4.00 M
3600 k2000 k
(4.18)(3455)
5.60 M
4.80 M
4.20 M
2400 k
(5.02)(4.15)
6.40 M
6.00 M
5.60 M
3000 k(6.27)(5.18)
7.20 M
6.40 M
5.60 M
3200 k
(6.69)(5.53)
8.00 M
7.20 M
6.40 M
3600 k
(7.53)(6.22)
8.40 M
8.00 M
7.20 M
4.00 M(8.36)(6.91)
10.00 M
9.60 M
8.40 M
4.80 M(10.04)(8.29)
14.00 M
12.00 M
10.00 M
6.00 M(12.55)(10.36)18.00 M
16.00 M
14.00 M
8.00 M(16.73)(13.82)28.00 M
24.00 M
20.00 M
12.00 M
(25.09)(20.73)150V(110V)[W]

Multiplier
rate

1000
×0.01
1200
1500
1600
2000
2400
3000
4.00 k
5.00 k
6.00 k
8.00 k
10.00 k×0.1
12.00 k15.00 k16.00 k20.00 k24.00 k30.00 k40.0 k
50.0 k
60.0 k
80.0 k
100.0 k×1
120.0 k150.0 k160.0 k180.0 k200.0 k240.0 k300.0 k320.0 k360.0 k
400 k
500 k
600 k
800 k
1000 k×101200 k1500 k1600 k1800 k2000 k2400 k
3000 k
4.00 M
6.00 M
×100<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW81<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a four-step setup
is possible) is set as the fastest.
The output pulse width of a pulse output is
set to 100 to 130ms. (Usually 240 to 260ms)

SQLC-215-109
Appendix table 2
Active power range, watt-hour multiplier rate table (Single-phase 3-wire)
V range

150V(110V)[W]

A range
5A
6A
7.5A
8A
10A
12A
15A
20A
25A
30A
40A
50A
60A
75A
80A
100A
120A
150A
200A
250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A
2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10000A
12000A
15000A
20000A
30000A

Multiplier
rate

1000
1200
1500
1600
2000
2400
3000
4.00 k
5.00 k
6.00 k
8.00 k
×0.1
10.00 k12.00 k15.00 k16.00 k20.00 k24.00 k30.00 k40.0 k
50.0 k
60.0 k
80.0 k×1
100.0 k120.0 k150.0 k160.0 k180.0 k200.0 k240.0 k300.0 k320.0 k360.0 k
400 k
500 k
600 k
800 k
×10
1000 k1200 k1500 k1600 k1800 k2000 k2400 k
3000 k
4.00 M
6.00 M
×100<Note 1>
Parenthesis is primary active power (reactive power) value in /1kW(1kvar).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW82<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a four-step setup
is possible) is set as the fastest.
The output pulse width of a pulse output is
set to 100 to 130ms. (Usually 240 to 260ms)

SQLC-215-109
Appendix table 3-1
Active power range, watt-hour multiplier rate table (Single-phase 2-wire)
300.0kV
255.0kV
210.0kV
750.0kV
500.0kV
180.0kV
150.0kV
45.0kV
30.0kV
105.0kV
90.0kV
375.0kVV range(VT550000/110V) (VT380000/110V) (VT275000/110V) (VT220000/110V) (VT187000/110V) (VT154000/110V) (VT132000/110V) (VT110000/110V) (VT77000/110V) (VT66000/110V) (VT33000/110V) (VT22000/110V) MultiplierMultiplier
rate
rate
[W]
[W]
A range
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
5A

2500 k

6A

3000 k

7.5A

4.00 M(3.75)
4.00 M

8A
10A
12A
×100

7.50 M

20A

10.00 M

25A
30A

14.00 M
(12.50)15.00 M

40A

20.00 M

50A

25.00 M

60A

30.00 M

75A

40.0 M(37.5)
40.0 M

100A

50.0 M

120A

60.0 M

150A

75.0 M

200A

100.0 M

250A
300A

140.0 M
(125.0)150.0 M

400A

200.0 M

500A

250.0 M

600A

300.0 M

750A

400 M
(375)
400 M

800A
900A

450 M

1000A

500 M

1200A

600 M

1500A

750 M

1600A
×100006.00 M

15A

80A

×1000

5.00 M

1800A
2000A
2500A
3000A
4000A
5000A
6000A

800 M
900 M

1800 k(1727)
2000 k(2073)
2800 k(2591)
2800 k(2764)
3600 k(3455)
4.20 M(4.15)
5.60 M(5.18)
7.20 M(6.91)
9.00 M(8.64)
10.00 M
(10.36)14.00 M
(13.82)18.00 M
(17.27)20.00 M
(20.73)28.00 M
(25.91)28.00 M
(27.64)36.00 M
(34.55)42.0 M(41.5)
56.0 M(51.8)
72.0 M(69.1)
90.0 M(86.4)
100.0 M
(103.6)140.0 M(138.2)180.0 M
(172.7)200.0 M
(207.3)280.0 M
(259.1)280.0 M
(276.4)320.0 M
(310.9)360.0 M
(345.5)
420 M
(415)
560 M
(518)
560 M
(553)
640 M
(622)
720 M
(691)
900 M
(864)

1400 k(1250)
1500 k1000 k2000 k(1875)
2000 k1500 k2500 k2000 k

3000 k2400 k

4.00 M(3.75)
5.00 M

3000 k

6.40 M(6.25)
7.50 M

5.00 M
6.00 M

10.00 M

8.00 M

14.00 M
(12.50)15.00 M

10.00 M

20.00 M
(18.75)20.00 M

15.00 M

1200 k1600 k

4.00 M

12.00 M

16.00 M

25.00 M

20.00 M

30.00 M

24.00 M

40.0 M(37.50)50.0 M

30.00 M

64.0 M(62.5)
75.0 M

50.0 M

100.0 M

80.0 M

140.0 M
(125.0)150.0 M

100.0 M

200.0 M
(187.5)200.0 M

40.0 M

60.0 M

120.0 M
150.0 M
160.0 M

240.0 M
(225.0)250.0 M

180.0 M
200.0 M

300.0 M

240.0 M

400 M
(375)
400 M

300.0 M

450 M

360.0 M

500 M

400 M

640 M
(625)
750 M

500 M

320.0 M

600 M
800 M

900 k
(850)
1000 k(1020)
1400 k(1275)
1400 k(1360)
1800 k(1700)
2000 k(2040)
2800 k(2550)
3600 k(3400)
4.50 M(4.25)
5.60 M(5.10)
7.20 M(6.80)
9.00 M(8.50)
10.00 M
(10.20)14.00 M(12.75)14.00 M
(13.60)18.00 M
(17.00)20.00 M
(20.40)28.00 M
(25.50)36.00 M
(34.00)45.0 M(42.5)
56.0 M(51.0)
72.0 M(68.0)
90.0 M(85.0)
100.0 M
(102.0)140.0 M
(127.5)140.0 M
(136.0)160.0 M
(153.0)180.0 M
(170.0)200.0 M
(204.0)280.0 M
(255.0)280.0 M
(272.0)320.0 M
(306.0)360.0 M
(340.0)
450 M
(425)
560 M
(510)
720 M
(680)
900 M
(850)

720 k
(700)
840 k

600 k

500 k

720 k

600 k

1200 k(1050)
1200 k(1120)
1400 k

900 k

750 k

960 k

800 k

1200 k1000 k1800 k(1680)
2400 k(2100)
2800 k1500 k(1440)
1800 k1200 k2400 k2000 k3600 k(3500)
4.20 M

3000 k2500 k3600 k

3000 k

5.60 M

4.80 M

4.00 M

7.20 M(7.00)
8.40 M

6.00 M

5.00 M

7.20 M

6.00 M

12.00 M
(10.50)12.00 M
(11.20)14.00 M

9.00 M

7.50 M

9.60 M

8.00 M

1500 k

360.0 k(350.0)420 k

300.0 k150.0 k100.0 k360.0 k180.0 k120.0 k

560 k
(525)
560 k

450 k

150.0 k480 k240.0 k(225.0)240.0 k

720 k
(700)
840 k

600 k

300.0 k200.0 k720 k360.0 k240.0 k1200 k(1050)
1400 k

160.0 k

900 k

450 k

300.0 k1200 k

600 k

400 k

1500 k

750 k

500 k

1800 k

900 k

600 k

2400 k1200 k800 k3600 k(3500)
4.20 M

3000 k1500 k1000 k3600 k1800 k1200 k

5.60 M(5.25)
5.60 M

4.50 M
4.80 M

2400 k(2250)
2400 k1600 k

7.20 M(7.00)
8.40 M

6.00 M

3000 k2000 k

7.20 M

3600 k2400 k12.00 M
(10.50)14.00 M

9.00 M

4.50 M

3000 k12.00 M

6.00 M

4.00 M

15.00 M

7.50 M

5.00 M

18.00 M

9.00 M

6.00 M

24.00 M

12.00 M

8.00 M

36.00 M
(35.00)42.0 M

30.00 M

15.00 M

10.00 M

36.00 M

18.00 M

12.00 M

56.0 M(52.5)
56.0 M

45.0 M

15.00 M

1800 k(1750)
2400 k(2100)
2800 k×10

12.00 M

10.00 M

18.00 M
(16.80)24.00 M
(21.00)28.00 M

15.00 M
(14.40)18.00 M

12.00 M
15.00 M

24.00 M

20.00 M

36.00 M
(35.00)42.0 M

30.00 M

25.00 M

36.00 M

30.00 M

56.0 M

48.0 M

40.0 M

72.0 M(70.0)
84.0 M

60.0 M

50.0 M

72.0 M

60.0 M

120.0 M
(105.0)120.0 M
(112.0)140.0 M
(126.0)140.0 M

90.0 M

75.0 M

18.00 M
(17.50)24.00 M
(21.00)28.00 M

1500 k×100

96.0 M

80.0 M

120.0 M(108.0)120.0 M

90.0 M
100.0 M

180.0 M
(168.0)240.0 M
(210.0)240.0 M
(224.0)280.0 M
(252.0)280.0 M

150.0 M
(144.0)180.0 M

120.0 M

200.0 M
(192.0)240.0 M
(216.0)240.0 M

200.0 M

360.0 M
(350.0)420 M

300.0 M

250.0 M

360.0 M

300.0 M

560 M

480 M

400 M

720 M
(700)
840 M

600 M

500 M

720 M

600 M

900 M

750 M

150.0 M
160.0 M
180.0 M

48.0 M

24.00 M
(22.50)24.00 M

64.0 M(63.0)
72.0 M(70.0)
84.0 M

56.0 M(54.0)
60.0 M

28.00 M
(27.00)30.00 M

20.00 M

72.0 M

36.00 M

24.00 M

120.0 M
(105.0)120.0 M
(112.0)140.0 M
(126.0)140.0 M

90.0 M

45.0 M

30.00 M

96.0 M

48.0 M

32.00 M

120.0 M(108.0)120.0 M

56.0 M(54.0)
60.0 M

36.00 M

150.0 M

75.0 M

50.0 M

180.0 M
(175.0)240.0 M
(210.0)280.0 M

16.00 M
18.00 M

40.0 M

180.0 M

90.0 M

60.0 M

240.0 M

120.0 M

80.0 M

300.0 M

150.0 M

100.0 M

360.0 M

180.0 M

120.0 M

450 M

150.0 M

×1000

7500A

800 M

8000A

360.0 M
(350.0)
420 M
560 M
(525)
560 M
640 M
(630)
720 M
(700)
840 M

9000A
10000A
12000A

480 M

240.0 M
(225.0)240.0 M

560 M
(540)
600 M

280.0 M
(270.0)300.0 M

180.0 M

720 M

360.0 M

240.0 M

900 M

160.0 M

200.0 M

450 M

300.0 M

20000A

600 M

400 M

30000A

900 M

600 M

15000A

×10000<Note 1>
Parenthesis is primary active power (reactive power) value in /500W(500var).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW83<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a four-step setup
is possible) is set as the fastest.
The output pulse width of a pulse output is
set to 100 to 130ms. (Usually 240 to 260ms)

SQLC-215-109
Appendix table 3-2
Active power range, watt-hour multiplier rate table (Single-phase 2-wire)
24.00kV
18.00kV
18.00kV
15.00kV
3000V
2400V
1500V
1200V
600V
9000V
4500V
25.00kVV range(VT18400/110V) (VT16500/110V) (VT13800/110V) (VT13200/110V) (VT11000/110V) (VT6600/110V) (VT3300/110V) (VT2200/110V) (VT1650/110V) (VT1100/110V) (VT880/110V) (VT480/110V) MultiplierMultiplier
rate
rate
A range
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
[W]
5A

×1

6A
7.5A
8A
10A
12A
15A
20A
25A
30A
40A

×10

50A
60A
75A
80A
100A
120A
150A
200A
250A
300A
×100

400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A
2500A
3000A

×1000

4000A
5000A
6000A
7500A
8000A
9000A
10000A
12000A
15000A
20000A

×1000030000A

84.0 k(83.6)
100.0 k(100.4)140.0 k(125.5)140.0 k(133.8)180.0 k(167.3)200.0 k(200.7)280.0 k(250.9)360.0 k(334.5)
420 k
(418)
560 k
(502)
720 k
(669)
840 k
(836)
1000 k(1004)
1400 k(1255)
1400 k(1338)
1800 k(1673)
2000 k(2007)
2800 k(2509)
3600 k(3345)
4.20 M(4.18)
5.60 M(5.02)
7.20 M(6.69)
8.40 M(8.36)
10.00 M
(10.04)14.00 M
(12.55)14.00 M
(13.38)16.00 M
(15.05)18.00 M
(16.73)20.00 M
(20.07)28.00 M
(25.09)28.00 M
(26.76)32.00 M
(30.11)36.00 M
(33.45)42.0 M(41.8)
56.0 M(50.2)
72.0 M(66.9)
84.0 M(83.6)
100.0 M
(100.4)140.0 M
(125.5)140.0 M
(133.8)160.0 M
(150.5)180.0 M(167.3)200.0 M
(200.7)280.0 M
(250.9)360.0 M
(334.5)
560 M
(502)

75.0 k
90.0 k
120.0 k(112.5)120.0 k150.0 k180.0 k240.0 k(225.0)300.0 k400 k(375.0)
450 k
600 k
750 k
900 k
1200 k(1125)
1200 k1500 k1800 k2400 k(2250)
3000 k
4.00 M(3.75)
4.50 M
6.00 M
7.50 M
9.00 M
12.00 M
(11.25)12.00 M
14.00 M
(13.50)15.00 M
18.00 M
24.00 M
(22.50)24.00 M
28.00 M
(27.00)30.00 M
40.0 M(37.5)
45.0 M
60.0 M
75.0 M
90.0 M
120.0 M
(112.5)120.0 M
140.0 M
(135.0)150.0 M
180.0 M
240.0 M
(225.0)300.0 M
450 M

64.0 k
(62.7)
80.0 k(75.3)
96.0 k(94.1)
100.0 k(100.4)140.0 k(125.5)160.0 k(150.5)200.0 k(188.2)280.0 k(250.9)320.0 k(313.6)400 k(376.4)
560 k
(502)
640 k
(627)
800 k
(753)
960 k
(941)
1000 k(1004)
1400 k(1255)
1600 k(1505)
2000 k(1882)
2800 k(2509)
3200 k(3136)
4.00 M(3.76)
5.60 M(5.02)
6.40 M(6.27)
8.00 M(7.53)
9.60 M(9.41)
10.00 M
(10.04)12.00 M
(11.29)14.00 M
(12.55)16.00 M
(15.05)20.00 M(18.82)24.00 M
(20.07)24.00 M
(22.58)28.00 M
(25.09)32.00 M
(31.36)40.0 M(37.64)56.0 M(50.2)
64.0 M(62.7)
80.0 M(75.3)
96.0 M(94.1)
100.0 M
(100.4)120.0 M
(112.9)140.0 M
(125.5)160.0 M
(150.5)200.0 M
(188.2)280.0 M
(250.9)
400 M
(376)

60.0 k

50.0 k30.00 k15.00 k10.00 k7.50 k

5.00 k

4.00 k

72.0 k

60.0 k36.00 k18.00 k12.00 k9.00 k

6.00 k4.80 k

90.0 k

75.0 k

45.0 k

15.00 k7.50 k

6.00 k

8.00 k

6.40 k

96.0 k

80.0 k

48.0 k

24.00 k(22.50)24.00 k16.00 k12.00 k(11.25)12.00 k120.0 k100.0 k60.0 k30.00 k20.00 k15.00 k10.00 k8.00 k

150.0 k(144.0) k
180.0 k120.0 k72.0 k36.00 k24.00 k18.00 k12.00 k9.60 k

150.0 k90.0 k

45.0 k30.00 k12.00 k200.0 k120.0 k60.0 k

40.0 k

24.00 k(22.50)30.00 k15.00 k240.0 k20.00 k16.00 k300.0 k250.0 k150.0 k75.0 k

50.0 k25.00 k20.00 k360.0 k300.0 k180.0 k90.0 k

60.0 k

40.0 k
(37.5)
45.0 k30.00 k24.00 k

480 k

400 k

240.0 k120.0 k80.0 k

60.0 k

40.0 k

32.00 k

600 k

500 k

300.0 k150.0 k100.0 k75.0 k

50.0 k

40.0 k

720 k

600 k

360.0 k180.0 k120.0 k90.0 k

60.0 k

48.0 k

900 k

750 k

450 k

150.0 k60.0 k

800 k

480 k

120.0 k(112.5)120.0 k75.0 k960 k240.0 k(225.0)240.0 k80.0 k

64.0 k

160.0 k1200 k1000 k600 k300.0 k200.0 k150.0 k100.0 k80.0 k1500 k(1440)
1800 k1200 k720 k360.0 k240.0 k180.0 k120.0 k96.0 k1500 k

900 k

450 k

300.0 k120.0 k2000 k1200 k

600 k

400 k

240.0 k(225.0)300.0 k150.0 k2400 k

200.0 k160.0 k3000 k2500 k1500 k

750 k

500 k

250.0 k200.0 k3600 k

3000 k1800 k

900 k

600 k

400 k
(375.0)450 k

300.0 k240.0 k4.80 M

4.00 M

2400 k1200 k

800 k

600 k

400 k

320.0 k6.00 M

5.00 M

3000 k1500 k1000 k

750 k

500 k

400 k

7.20 M

6.00 M

3600 k1800 k1200 k

900 k

600 k

480 k

9.00 M

7.50 M

4.50 M

1500 k600 k8.00 M

4.80 M

1200 k(1125)
1200 k750 k

9.60 M

2400 k(2250)
2400 k

800 k

640 k

12.00 M
(10.80)12.00 M

9.00 M

2800 k(2700)
3000 k1800 k720 k2000 k1400 k(1350)
1500 k

900 k10.00 M

5.60 M(5.40)
6.00 M

1000 k800 k15.00 M
(14.40)18.00 M

12.00 M

7.20 M

3600 k2400 k1800 k1200 k960 k15.00 M

9.00 M

4.50 M

3000 k1200 k

16.00 M

9.60 M

4.80 M

3200 k2400 k(2250)
2400 k1500 k

20.00 M
(19.20)24.00 M
(21.60)24.00 M

1600 k18.00 M

12.00 M
(10.80)12.00 M

5.60 M(5.40)
6.00 M

3600 k

20.00 M

30.00 M

25.00 M

15.00 M

7.50 M

5.00 M

36.00 M

30.00 M

18.00 M

9.00 M

48.0 M

40.0 M

24.00 M

12.00 M

1600 k2800 k(2700)
3000 k1800 k2000 k1400 k(1280)
1500 k(1440)
1600 k2500 k2000 k

6.00 M

4.00 M(3.75)
4.50 M

3000 k2400 k

8.00 M

6.00 M

4.00 M

3200 k

4.00 M

60.0 M

50.0 M

30.00 M

15.00 M

10.00 M

7.50 M

5.00 M

4.00 M

72.0 M

60.0 M

36.00 M

18.00 M

12.00 M

9.00 M

6.00 M

4.80 M

90.0 M

75.0 M

45.0 M

15.00 M

12.00 M
(11.25)12.00 M

7.50 M

6.00 M

8.00 M

6.40 M

9.00 M

7.20 M

96.0 M

80.0 M

48.0 M

24.00 M
(22.50)24.00 M

120.0 M(108.0)120.0 M

90.0 M

28.00 M
(27.00)30.00 M

18.00 M

100.0 M

56.0 M(54.0)
60.0 M

20.00 M

14.00 M
(13.50)15.00 M

10.00 M

8.00 M

120.0 M

72.0 M

36.00 M

24.00 M

18.00 M

12.00 M

9.60 M
12.00 M

16.00 M

150.0 M
(144.0)180.0 M

150.0 M

90.0 M

45.0 M

30.00 M

200.0 M

120.0 M

60.0 M

40.0 M

24.00 M
(22.50)30.00 M

15.00 M

240.0 M

20.00 M

16.00 M

360.0 M

300.0 M

180.0 M

90.0 M

60.0 M

45.0 M

30.00 M

24.00 M

<Note 1>
Parenthesis is primary active power (reactive power) value in /500W(500var).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW

84

2400
(2182)
2800(2618)
3600(3273)
3600(3491)
4.50 k
(4.36)
5.60 k(5.24)
7.20 k(6.55)
9.00 k(8.73)
12.00 k(10.91)14.00 k(13.09)18.00 k(17.45)24.00 k(21.82)28.00 k(26.18)36.00 k(32.73)36.00 k(34.91)45.0 k
(43.6)
56.0 k
(52.4)
72.0 k
(65.5)
9.00 k(87.3)
120.0 k(109.1)140.0 k(130.9)180.0 k(174.5)240.0 k(218.2)280.0 k(261.8)360.0 k(327.3)360.0 k(349.1)
400 k
(393)
450 k
(436)
560 k
(524)
720 k
(655)
720 k
(698)
800 k
(785)
900 k
(873)
1200 k(1091)
1400 k(1309)
1800 k(1745)
2400 k(2182)
2800 k(2618)
3600 k(3273)
3600 k(3491)
4.00 M(3.93)
4.50 M(4.36)
5.60 M(5.24)
7.20 M(6.55)
9.00 M(8.73)
14.00 M
(13.09)

×0.1

×1

×10

×100
×1000

<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a four-step setup
is possible) is set as the fastest.
The output pulse width of a pulse output is
set to 100 to 130ms. (Usually 240 to 260ms)

SQLC-215-109
Appendix table 3-3
Active power range, watt-hour multiplier rate table (Single-phase 2-wire)
600V
600V
500V
300V
V range(VT460/110V) (VT440/110V) (VT380/110V) (VT220/110V)
Multiplier
rate
[W]
[W]
[W]
[W]
A range
5A
6A
7.5A
8A
10A
12A
15A
×0.1

20A
25A
30A
40A
50A
60A
75A
80A
100A
120A
150A
200A

×1

250A
300A
400A
500A
600A
750A
800A
900A
1000A
1200A
1500A
1600A
1800A
2000A

×10

2500A
3000A
4000A
5000A
6000A
7500A
8000A
9000A
10000A
12000A
15000A×100
×1000

20000A
30000A

2400(2091)
2800(2509)
3200(3136)
3600(3345)
4.20 k
(4.18)
5.60 k
(5.02)
6.40 k(6.27)
8.40 k
(8.36)
12.00 k(10.45)14.00 k(12.55)18.00 k(16.73)24.00 k(20.91)28.00 k(25.09)32.00 k(31.36)36.00 k(33.45)42.0 k(41.8)
56.0 k(50.2)
64.0 k
(62.7)
84.0 k(83.6)
120.0 k(104.5)140.0 k(125.5)180.0 k(167.3)240.0 k(209.1)280.0 k(250.9)320.0 k(313.6)360.0 k(334.5)400 k(376.4)
420 k
(418)
560 k
(502)
640 k
(627)
720 k
(669)
800 k
(753)
840 k
(836)
1200 k(1045)
1400 k(1255)
1800 k(1673)
2400 k(2091)
2800 k(2509)
3200 k(3136)
3600 k(3345)
4.00 M(3764)
4.20 M(4.18)
5.60 M(5.02)
6.40 M(6.27)
8.40 M(8.36)
14.00 M
(12.55)

2000
2400
3000
3200
4.00 k4.80 k
6.00 k
8.00 k
10.00 k12.00 k16.00 k20.00 k24.00 k30.00 k32.00 k40.0 k
48.0 k
60.0 k
80.0 k
100.0 k120.0 k160.0 k200.0 k240.0 k300.0 k320.0 k360.0 k
400 k
480 k
600 k
640 k
720 k
800 k
1000 k1200 k1600 k2000 k2400 k
3000 k3200 k3600 k
4.00 M
4.80 M
6.00 M
8.00 M
12.00 M

1800
(1727)
2000(2073)
2800(2591)
2800(2764)
3600(3455)
4.20 k
(4.15)
5.60 k
(5.18)
7.20 k
(6.91)
9.00 k
(8.64)
10.00 k(10.36)14.00 k(13.82)18.00 k(17.27)20.00 k(20.73)28.00 k(25.91)28.00 k(27.64)36.00 k(34.55)42.0 k(41.5)
56.0 k
(51.8)
72.0 k(69.1)
90.0 k
(86.4)
100.0 k(103.6)140.0 k(138.2)180.0 k(172.7)200.0 k(207.3)280.0 k(259.1)280.0 k(276.4)320.0 k(310.9)360.0 k(345.5)
420 k
(415)
560 k
(518)
560 k
(553)
640 k
(622)
720 k
(691)
900 k
(864)
1000 k(1036)
1400 k(1382)
1800 k(1727)
2000 k(2073)
2800 k(2591)
2800 k(2764)
3200 k(3109)
3600 k(3455)
4.20 M(4.15)
5.60 M(5.18)
7.20 M(6.91)
10.00 M
(10.36)150V(110V)
[W]

1000

500

1200

600

1500

750

1600

800

2000

1000

2400

1200

3000

1500

4.00 k

2000

5.00 k

2500

6.00 k

3000

Multiplier
rate

×0.01

8.00 k

4.00 k

10.00 k5.00 k12.00 k6.00 k

15.00 k7.50 k

16.00 k8.00 k

20.00 k10.00 k24.00 k12.00 k30.00 k15.00 k40.0 k

20.00 k50.0 k25.00 k60.0 k30.00 k×0.1

80.0 k

40.0 k

100.0 k50.0 k

120.0 k60.0 k

150.0 k75.0 k

160.0 k80.0 k

180.0 k90.0 k

200.0 k100.0 k240.0 k120.0 k300.0 k150.0 k320.0 k160.0 k×1

360.0 k180.0 k400 k200.0 k500 k250.0 k600 k300.0 k

800 k

400 k

1000 k500 k1200 k600 k1500 k750 k1600 k800 k1800 k900 k2000 k1000 k2400 k1200 k

3000 k1500 k

4.00 M

2000 k

6.00 M

3000 k×10×100<Note 1>
Parenthesis is primary active power (reactive power) value in /500W(500var).
In the blank, setting is impossible.
About active power and reactive power range.
Full scale display, less than 4000：4 digit. More than 4000：3 digit.
An example) 4800kW → 4.80MW
40kvar → 40.0kvar
20kW → 20.00kW85<Note 2>
In case the voltage range and the currentof an upper table
range were set as
and an output pulse unit (a four-step setup
is possible) is set as the fastest.
The output pulse width of a pulse output is
set to 100 to 130ms. (Usually 240 to 260ms)

SQLC-215-109

Tokyo Office : 11-13, Hitotsuya 1-chome, Adachi-ku, Tokyo, 121-8639, JAPAN.
TEL：+81-3-3885-2411 , Fax：+81-3-3858-3966
Kyoto Office : 1-19, Ichinobe-Nishikawahara, Jyoyou-shi, Kyoto, 610-0114, JAPAN.
TEL：+81-774-55-1391 , Fax：+81-774-54-1353
Revision A, DATE：November 7, 201586