§Small-sized plug-in transducer§
1 output type

Revolution-speed transducer
FSGT■ Application
Takes input from a taco-generator installed on a
dynamo or suchlike; insulates input and output andconverts it into a DC signal proportional to number of
revolutions (frequency). In addition, pulse input
corresponding to the various sensors also can bemanufactured.
Up to 16 units can be housed in an installation base.

■ Feature
1.
2.
3.
4.
5.

23×76×125mm/160g

Compact and high withstand voltage.
Withstand voltage between input/output/auxiliary
supply/outer case is AC2, 000V (50/60Hz) for 1 min..
Constant voltage/current output type. No need to adjust
the product if it operates within load resistance range.
A LED can confirm status of electric power applied.
Zero/span is adjustable. (±2% adjustable)■ Block Diagram■ Connection diagram (socket)

Input circuit
Zero-cross converter
Frequency-phase-locked loop circuit
One-shot multi-vibrator
Smoothing circuit
Constant voltage circuit
Insulated power source circuit
Oscillating circuit
Pulse width modulation circuit
Photo coupler insulation⑪ Reference voltage⑫ Pulse width demodulation circuit⑬ Output circuit①②③④⑤⑥⑦⑧⑨⑩1§Small-sized plug-in transducer§
1 output type

Revolution-speed transducer■ Specification●How to specify

Type name

FSGT-Specification code

X

0

1:

A:0-33.3Hz

Normal operating
voltage range(input resistance)
Sine wave input

3 :DC0-5V(≧600Ω)F:

Sine wave

B:0-40Hz

1 : 1-25V

4 :DC0-10V(≧2kΩ)

AC/DC80-264V

fuse

±0.4%

input

C :0-50Hz(approx.25kΩ)

5 :DC1-5V(≧600Ω)Rated Voltage

2 :with(within

D :0-55Hz

2 : 2-50V

fuseoutput)

Kind of
input

Input
frequency

1st Output(load resistance)

Auxiliary
supplyPower
fuse

Commonspecification

1 :without

Conversion accuracy:

AC100/110V

10-100%

E:0-60Hz(approx.50kΩ)A :DC0-1mA (≦10kΩ)

50/60HzF :0-65Hz

3:5-110V

B :DC0-5mA (≦2kΩ)AC200/220V

Temperature

G:0-66.6Hz(approx.110kΩ)C :DC0-10mA (≦1kΩ)50/60Hzcharacteristics:

H:0-100Hz

4:10-220V

D :DC0-16mA (≦600Ω)

DC100/110V

0.4%/10℃

I :0-120Hz
J :0-166.6Hz(approx.220kΩ)E :DC1-5mA (≦3kΩ)
5 :DC24V

Response time:A:

K:0-200Hz

Pulse input

Pulse

L :0-333.3Hz

A :5Vp-p(input≧200Hz)

input

M :0-500Hz(approx.5kΩ)≦1.0s/90%

N :0-1kHz

B:10Vp-p(input<200Hz)

F :DC4-20mA (≦750Ω)(DC19-30V)≦0.5s/90%(approx.10kΩ)
C:12Vp-p

Output ripple:(approx.12kΩ)≦1%p-pD :15Vp-p(when input ≧10%)(approx15kΩ)E:24Vp-p

Consumption VA:(approx.24kΩ)At AC110V: 3.5VA
At AC220V: 4.5VA

Z :other

Z :other thanZ :other than thoseZ :other thanAt DC110V: 2W

thanthose above *1

above *1

those above *1

At DC24V: 2.5W

those(See product

Z :other than those(See product range)(See productabove *1range)

above *1

(Seerange)(See product range)

Weight:Without socket:

product

approx.130grange)

With socket:
approx.160g*1 Consult with us for specification other than those indicated in the table above.

Product Range (including special handling)
Input
Kind of input: consult with us for input other than those above.
Input frequency: 30Hz-10kHz *2
Normal operating voltage range: AC10V-250V
5Vp-p-35Vp-p

OutputCurrent output: 1mA-20mA
Voltage output: 4V-10V*2 Input frequency: over 1.1 kHz, but not more than 10 kHz is subject to special handling.

2of