Data Sheet
1032 Sanitary Pressure Gauge
FEATURES
 Can be Clean-in-place (CIP) or Steam-in-place (SIP) Autoclave or sterilize with 3½˝ dial with polysulfone
window option
 Easy Zero™ provides external adjustability of instrument
span (3½˝ dial)
 12-15 RA (microinches) electropolished diaphragmsurface Material certificates and certificate of conformance
supplied as standard to EN 10204:2004 3.1
 PLUS! ™ Performance option
TYPICAL USES Specialized OEM equipment
 Filtration
 Food and dairy
 Pharmaceutical
 Biotech Medical equipment
SPECIFICATIONS                LiquidAccuracy:                Filled(Of Span)             Dry   or XLL
2½˝, 3½˝ All ranges to 100 psi    ±2% ±2.5%2½˝, 3½˝ Pressure over 100 psi    ±1.5% ±2%
4½˝             ±1.5% ±1.5%
Dial Sizes:
2½˝, 3½˝ or 4½˝1032
2½˝, 3½˝, 4½˝ dial sizes

Ranges:Vacuum, compound to 1,000 psi• Electropolished diaphragm and housing

Fill:

Glycerin (USP food grade), Halocarbon®, mineral oil,food grade silicone• Heat lot traceable material certificates

Process Connection
Location:

Lower or lower backProcess Connection:

1½˝ or 2˝ Tri-Clamp®

Pointer:2½˝: Black-painted aluminum, adjustable
3½˝: Black-painted aluminum with ext. adjust
4½˝: Black-painted aluminum, adjustable

Dial:

White with black markings with 3A-74 logo

Clean or Steam inPlace (CIS, SIP):
Case Style:

Process temperature limit of 280 °F (138 °C)

Window Material:

2½˝, 3½˝: Polycarbonate
4½˝: Glass
2½˝, 3½˝: 300 Stainless steel
4½˝: 400 Stainless steel

Movement:
Weather Protection:Autoclave or Sterlize:

Open Front

2½˝ & 3½˝ IP65
4½˝ Dry: IP54
4½˝ Hermetically sealed or liquid filled IP65
3½˝: Temperature limit of 300 °F (149 °C) when
supplied with Polysulfone window (option PS drygauge only)

TM

74-07

KEY BENEFITS• True Zero™ reduces reading errors by using a "zerobox" instead of conventional pins

SPECIFICATIONS (Cont.)
Dampening Options: Liquid fill or PLUS! ™ Performance (Design is notrecommended for gauges expected to be exposed
to temperatuire over 200 °F (93 °C). This includes
gauges subject to autoclaving.
Approvals:
3A compliance to sanitary standard 74, CRN, RoHSWETTED COMPONENTS
Diaphragm

Process Connection

Joints

316L Stainless steel
electropolished
Hastelloy® C276

316L Stainless steel
Hastelloy® C276

WeldedNON-WETTED COMPONENTS
CaseRing

300 Stainless steel polished

300 Stainless steel polishedbayonet, removable

1 of 4All specifications are subject to change without notice. All sales subject to standard terms and conditions.
Ashcroft®, Easy Zero™, Plus!™, True Zero™, Trust the Shield® are trademarks of Ashcroft Inc.
The following non-Ashcroft trademarks are the property of their respective owners: Halocarbon®, Hastelloy®, Neobee®, Tri-Clamp®.
For more information, see Ashcroft Brands & Trademarks ©2023 Ashcroft Inc. 1032_gauge_ds_RevK_11-20-23

ashcroft.com | 1.800.328.8258 | Contact Us

Data Sheet
1032 Sanitary Pressure Gauge
ORDERING CODE

Example:251032

S15L-XC4100#

Dial Size/Model
251032 - 2½˝251032

351032 - 3½˝451032 - 4½˝SystemS - 316L Stainless steel tube and socket, 316L diaphragm and seal housing

SH - 316 Stainless steel tube and socket, Hastelloy C276 diaphragm and seal housing®

Case FillBlank - Dry gauge
L - Liquid fill (Food Grade Glycerine)
Process Connection Size
15 - 1½˝ Tri-Clamp® (2½˝ and 3½˝ dial only)

15

20 - 2˝ Tri-Clamp®

Process Connection Location
LL - Lower
B - Lower BackOptions (If choosing an option(s) must include an “X”)

-X _ _

CF - Halocarbon system fill®

CZ - Food grade silicone system fill
C4 - Traceable calibration certificate
FJ - Distilled water system fill
GX - Halocarbon® case fill

LL - PLUS! ™ performance
MO - Mineral oil case fill
MY - Mineral oil system fill
MQ - Positive material identification (PMI) (diaphragm and top housing only)
NH - Stainless metal tag
NM - Neobee® M-20 system fill

NZ - PLUS! ™ performance silicone freePD - Polycarbonate window (4½˝ only)
PS - Polysulfone window (3½˝ only)
SG - Safety glass
ZF - Food grade silicone case fill
6B - Cleaned for gaseous oxygen service (requires Halocarbon® or distilled water case and systems fill)
6W - Cleaned for gases other than oxygen
Range (Coding examples only, see range table on page 3)
Vacuum (30 IMV)
100# - 100 psi

100#

1000# - 1,000 psi (range required high pressure type clamp, not supplied by Ashcroft)

2 of 4All specifications are subject to change without notice. All sales subject to standard terms and conditions.
Ashcroft®, Easy Zero™, Plus!™, True Zero™, Trust the Shield® are trademarks of Ashcroft Inc.
The following non-Ashcroft trademarks are the property of their respective owners: Halocarbon®, Hastelloy®, Neobee®, Tri-Clamp®.
For more information, see Ashcroft Brands & Trademarks ©2023 Ashcroft Inc. 1032_gauge_ds_RevK_11-20-23

ashcroft.com | 1.800.328.8258 | Contact Us

Data Sheet
1032 Sanitary Pressure Gauge

Positive Pressure

CompoundVacuum

STANDARD PRESSURE RANGES
psibarkPa

kg/cm2

Dual Scale
psi Outer Scale

30IMV

-1BR

100KP

-1KSC

30IMV

30IMV&15#–––––-1&1.5BR-100&150KP-1&1.5KSC

30IMV&20#

30IMV&30#–––––-1&3BR-100&300KP-1&3KSC

30IMV&40#

30IMV&60#–––––-1&5BR-100&500KP-1&5KSC

30IMV&70#

30IMV&100#–––––-1&9BR-100&900KP-1&9KSC

30IMV&125#

15#

100KP

1BR

1KSC

14#(1)

–

1.6BR

160KP

1.6KSC

22#

30#

–––––

2.5BR

250KP

2.5KSC

35#

60#

4BR

400KP

4KSC

55#

–

6BR

600KP

6KSC

85#

100#–––––

10BR

1000KP

10KSC

140#

160#–––

–

200#

–––––

16BR

1600KP

16KSC

220#300#–––––

25BR

2500KP

25KSC

350#

400#––––

600#

40BR

4000KP

40KSC

550#-60BR

6000KP

60KSC

850#––––1000#(1)(1) Requires high pressure clamp

3 of 4All specifications are subject to change without notice. All sales subject to standard terms and conditions.
Ashcroft®, Easy Zero™, Plus!™, True Zero™, Trust the Shield® are trademarks of Ashcroft Inc.
The following non-Ashcroft trademarks are the property of their respective owners: Halocarbon®, Hastelloy®, Neobee®, Tri-Clamp®.
For more information, see Ashcroft Brands & Trademarks ©2023 Ashcroft Inc. 1032_gauge_ds_RevK_11-20-23

ashcroft.com | 1.800.328.8258 | Contact Us

DIMENSIONS in [ ] are milimeters

Data Sheet

2½˝ 1032 SANITARY GAUGE
31.74
1.25

1032 Sanitary Pressure Gauge

49.11
0.36

69.88
2.75

DIMENSIONS in [ ] are milimeters

62.32
2.45

21/2˝ 1032 SANITARY GAUGE

2½˝ 1032 SANITARY GAUGE

76.89
3.03

31.74
1.25

2

Data Sheet

44.70
1.76

9.11
0.36

69.88
2.75

21.37
0.84

"A"

Seal Size

Dim A

1032 Sanitary
Pressure Gauge
1.98
1½[51]DIMENSIONS in [2.52] are milimeters
2[64]

62.32
2.45

76.89
3.03

21.37
0.84

2½˝ 1032 SANITARY GAUGE

21.290.84

31.74
1.25

"A"

44.7
1.7

9.11

69.88
2.75

0.36
4½˝ 1032 SANITARY
GAUGE

52
2.06

120.463½˝ 1032 SANITARY GAUGE
31/2˝ 1032 SANITARY GAUGE
76.89
3.03

31.74
1.25
98.81
3.89

62.32
2.45

132
5.18

21.37
0.84

116
4.56

44.70
1.76

9.11
0.36

"A"

113
4.93

Seal Size

Dim A

1½

1.98
[51]

91.25
3.59

2

91.29
3.59

21.37
0.84

2.50 64

21.2
0.8

24
0.93

2.52
[64]

21.29
0.84

"A"

4½˝ 1032 SANITARY GAUGE

4½˝ 1032 SANITARYGAUGE41/2˝ 1032 SANITARY
GAUGE
12.00
0.46

120.46
52.00
2.06

52
2.06

132
5.18
116
4.56

132.005.18
116.00
4.56

24
0.93

113
4.93
24.00
0.93

113.004.93

All specifications are subject to change without notice.

aAll sales subject to standard terms and conditions.

in©2018 Ashcroft Inc. Ash_Bul_1032_pressure_ds, Rev. A, 02/18

12.50 64
2.50

64.00

4 of 4All specifications are subject to change without notice. All sales subject to standard terms and conditions.
Ashcroft®, Easy Zero™, Plus!™, True Zero™, Trust the Shield® are trademarks of Ashcroft Inc.
The following non-Ashcroft trademarks are the property of their respective owners: Halocarbon®, Hastelloy®, Neobee®, Tri-Clamp®.
For more information, see Ashcroft Brands & Trademarks ©2023 Ashcroft Inc. 1032_gauge_ds_RevK_11-20-23

ashcroft.com | 1.800.328.8258 | Contact Us