More flexibility when choosing your sensors:

Wireless Data Logger MSR 145W2Dwith WiFi and FlexSensors

Wireless data logger (WiFi) for monitoring temperature, humidity, fluid pressure, air pressure, light, acceleration, strain gauges and more.
The MSR 145W2D stores over one million measured values. By integration with aWLAN/WiFi network, the data logger can be used to flexibly implement practically any
kind of measurement applications.

• integration either into an existing WiFi or via an access point, by means of mobilenetwork access where required, into a separate wireless network• WiFi range can be extended with a commercially available repeater• data storage is guaranteed even if the transmitter fails• MSR SmartCloud: retrieve, analyse and export measured data online• alarm message if limit value is exceeded• MSR WLAN software for graphical visualization, components for LabVIEW available• FlexConnector for interchangeable FlexSensors
General Technical Data of the MSR 145W2D WiFi Data Logger
Memory capacity:

Over 1 000 000 measured values.

Measurement/storage
1/s up to every 12 hrate:
High-contrast, brilliant OLED colour display to show data
OLED display:
and graphical charts, resolution 96 x 64 pixels.
Interface:USB, WiFi
MSR Setup, Reader, Viewer and Online software
PC software:
MSR WLAN Dashboard and Viewer
Operating conditions:Temperature: -20...+65 °C

Storage conditions:Temperature:
Humidity:+5...+45 °C (ideal storage conditions for the battery)
10...95 % relative humidity, non-condensingStandards:

The MSR 145W2D complies with EU-Directives RoHS/WEEE.

Housing and Battery Selection
Sheat material

Battery

Size & Weight (W x H x L)

Standard
IP 60

PC, not encapsulated

230 mAh 35 x 17 x 57 mm, approx. 27 g

PC, not encapsulated

900 mAh 35 x 25 x 57 mm, approx. 43 g

Waterproof
IP 67

PC, encapsulated

230 mAh 35 x 17 x 57 mm, approx. 36 g

PC, encapsulated

900 mAh 35 x 25 x 57 mm, approx. 62 g~23
Ø5Housingext. temperature sensor

Selection of Sensors and Connections
You can select various internal sensors as well as up to five external sensors or connectors. The intelligent FlexConnector plug connection for interchangeable FlexSensors guarantees you the greatest possible flexibility. Info: www.msr.ch/en/msr145w2d
Working range

Accuracy(max. deviation)

Temperature

ext.: -40...+125 °C±0.2°C (-10...+50 °C)±1°C (-40...+125 °C)±0.2°C (-10...+50 °C)

int.: -20…+65 °C

Ø6Measured parameters~23

ext. humidity sensor~23-200...+1350 °C

Relative humidity
with integrated
temperature

0…100 % rel. humidity
ext. -40...+125 °C
int. -20...+65 °C±1.8 % rel. humidity(10…85 %, 0…+40 °C)±4 % rel. humidity(85…95 %, 0…+40 °C)

Air pressure
absolute, withintegratedtemperature

10...2000 mbar absoluteext.: -40...+85 °C
int.: -20...+65 °C
0...14 bar absolute-20...+65 °C±2 mbar(750…1100 mbar
absolute, +25 °C)±50 mbar(1...10 bar abs., +25 °C)

Fluid pressure withintegrated temperature-1000...3000 mbar relative-20...+85 °C±30 mbar (1 % FS)

Material in contact withmedia:• Stainless SteelAISI 316L(DIN1.4404/1.4435)• O-Rings:
Viton® 70° Shore

0...3000 mbar absolute-20...+85 °C±30 mbar (1 % FS)

0...30 bar absolute-20...+85 °C±0.3 bar (1 % FS)

0...400 bar absolute-20...+85 °C
0...1000 bar absolute-20...+85 °C±4 bar (1 % FS)

ext. fluid pressure sensor±10 bar (1 % FS)

3-axis-acceleration(static)-15...+15 g-20...+65 °C±0.15 g(0...5 g, +25 °C)

Light

0…65 000 lx

max. sensitivity at
500 nmMSR Electronics GmbH
Mettlenstrasse 6CH-8472 Seuzach
Switzerland
Phone +41 52 316 25 55
Fax +41 52 316 35 21
info@msr.ch
www.msr.chØ5Connector for thermal element of TypeK (without sensor)

Connection for strain full bridge • rated output 1mV / V • bridge voltage 3 Vgauges
Analogue inputs
Resolution: 12 bit

4 or 8 analogue inputs with freely selectable input
configuration: 0…20 mA; 4…20 mA; 0…3.0 V; 0…5.0V;0…10.0 V; 0…12.0 V; 0…24.0 V; 0…36,0 V; 0...48,0 VControl Terminal

alarm output • input for starting and stopping data recording • 3 V voltage output

FlexConnector

Please find suitable FlexSensors on our website.

ext. light sensor

Ø6Ø19~57

ext. air pressure sensor

G¹/86kt.SW17

Ø16,6~51

Distributor:09.20

Please contact us for prices and terms of delivery.