Pen-sized Electronic Pipettor

More efficient,More convenient,More accurate

WHAT’SNEW

123

Small/Light/Pen-type Grip(Patented)

High accurate multiple
dispensing

Automatical dispensing
volume compensation＋Continuous operation
frequency is 1.5 times
higher thanconventional models
Blowout function
Memory function

Specification● Product No.

MSIC01-03-20

MSIC01-03-250

MSIC01-03-1000● Range of dispensing

0.1-20µL

1-250µL

5-1000µL● Accuracy assurance range

2-20µL20-250µL

100-1000µL

Repeatability（C.V）

20µL ≦ 0.4％
2µL ≦ 2.0％

250µL ≦ 0.15％
20µL ≦ 0.8％1000µL ≦ 0.15％
100µL ≦ 0.5％

Accuracy

20µL ± 1.0％
2µL ± 3.5％250µL ± 0.5％20µL ± 2.5％1000µL ± 0.5％100µL ± 1.5％

Repeatability（C.V）

2µL ≦ 3.5％

20µL ≦ 2.0％

100µL ≦ 1.5％

● Single Dispensing● Multiple Dispensing● Outside dimension（㎜）Accuracy

2µL ± 5.0％

20µL ± 3.0％

100µL ± 2.0％

20 × 54 × 185

20 × 54 × 181

20 × 54 × 181● Weight

Approximately75g（including battery）● PowerAAA battery (Ni-MH/Rechargeable）● Attachable tip

WATSON、GILSON、Eppendorf、Greiner Bio-One、Thermo Fisher Scientiﬁc、RAININ、Compatible with other brand general tips
※ Recommend tips：Thermo Fisher Scientiﬁc（ART Tips, Finn Tips)

1 High accurate dispensing against “Hand Warming”!

Both of manual and electronic pipettes スポット1~27.4℃
スポット2~30.8
1
31.8
have the problem of “Hand Warming”,
that effect to dispensing accuracy,due
2
to hand heat is transmitted inside ofsyringe. As a result of measuring how
24.4
much hand heat effects to the accuracy,
the temperature outside of pipette rose 5.8 ℃ in about 20 minutes,
and the dispensing accuracy changed 1.8%. “pipetty” haveinstalled thermal sensor and is the only pipette in the world that
automatically compensate the dispensing volume according to thetemperature.

pipetty is powered by our micro actuator ofparadox gear mechanism. The high resolution enables the pipetty to control fine liquidcontrol, and high precision dispensing isalways possible regardless of the researcher's technique.

0.00

0.00

−0.50

−0.50

Manual Pipette 1.2％

−1.00

−1.00

−1.50

−1.50

−2.00

−2.00

−2.50
−3.00

Electronic Pipette 2.5％0102030405060

Controlled 0.3％

−2.50
−3.000

Dispensing Times

1020304050

Dispensing Times

Dimension：Φ8×188mm
Resolution：0.0031mm/Step

Manual Pipette vs pipetty

Repeatability（20μL）

Repeatability（2μL）

Repeatability（100μL）

2.5

1.0

0.5

2.0

0.8

0.4

1.5

0.6

0.3

1.0

0.4

0.2

0.5

0.2

0.1

0.0Feature

pipetty

0.50

2 High accurate dispensing

pipetty

Manual Pipette

0.0

Manual Pipette

0.0

pipetty

Manual Pipette

pipetty

3 Reduces the burden of long-time dispensing work

Comparison of workload between pipetty
and manual pipette
Manual pipettes are overloaded with longer
piston strokes as the dispensing volume
increases. On the other hand, the data showedthat the pipetty requires only simple push ofbutton at any volume, and the load is small. Inaddition, because aspiration and dispensing
can be performed at a constant speed, contamination during aspiration and foaming during
dispensing can be suppressed.

Product Customization

Comparison of manual and electronic pipette workloads＊Evaluated by Iwate University

0.3
Average potential [mV]FeatureOther brand pipette

0.50

Accuracy（％）

Feature

EMG sensor measurement position

ManualBase of the thumb

pipetty

0.2

Inside the arm

0.1

020200（250）
Pipette typeGround

1000

This product can be customized to meet your needs. Please feel free to contact us.

【Manufacturer】【Global Distributor】

Icomes Lab Co., Ltd.
2-4-23, Kitaiioka, Morioka-shi, Iwate 020-0857,Japan
TEL:+81-19-601-8228 FAX:+81-19-601-8227
URL:https://www.icomes.co.jp/en/ Mail:globalinfo@icomes.co.jp

TOLIMS Co., Ltd.
2-4-23, Kitaiioka, Morioka-shi, Iwate 020-0857,Japan
TEL:+81-19-601-8157
URL:https://www.tolims.co.jp/ Mail:info@tolims.co.jp

60