universelle Datenlogger

MSR 175 Shock Transportation Data Logger:
Protect your goods!

The MSR 175 shock data logger uses high-resolution sensors to continuously
record what happens to your goods during transportation. The logger data canthen be analysed very accurately – be it in order to ascertain damage or toimprove transport packaging – using the «MSR ShockViewer» software.
The two 3-axis acceleration sensors (±15 g and ±200 g) integrated in the MSR 175
record shocks at a measurement frequency of up to 6400/s. The installed memoryof the logger is capable of storing over 2 million measured values, which is sufficientfor more than 1,000 shocks. In addition, the MSR 175 measures and records temperature profiles from -20° to +65 °C. Additionally, type variants of this data logger areavailable with internal humidity, pressure and light sensors.
Easy-to-use PC software: The software package for the MSR 175 includes 3 programs: MSR 175 Dashboard, MSR ReportGenerator and MSR ShockViewer. TheMSR Dashboard allows you to configure the loggers with just a few mouse clicks. The
data recorded can be quickly transferred to a computer via the USB port. A single clickon the MSR ReportGenerator is sufficient to call up a compact report. For a detailed
measured data analysis, the MSR ShockViewer, which is integrated in the MSR175
software package as a free basic version, is at your disposal.
Technical Data
Medium:Memory capacity:
Key:

Air
Over 2,000,000 measurement values
Start measurement/retrieve status

3-axis-acceleration
sensors:

2 internal 3-axis-acceleration sensors

Working range:
Accuracy:
Measurement/storage rate:±15 g and ±200 g (-20…+65 °C)±15 g sensor: ±0.15 g (0...5 g, +25 °C), ±0.3 g (5...15 g,+25 °C), ±200 g sensor: ±2 g (0...15 g, +25 °C), ±5 g
(15...100 g, +25 °C), ±10 g (100...200 g, +25 °C)
Up to 6400/s (±15 %)Power supply:

LiPo-battery 230 mAh, charged via the USB connection, recording period up to 8 weeks or replaceable LiSOCI2 battery, battery operating time at least 2 years

PC software package:

MSR175 Dashboard, MSR ReportGenerator,MSR ShockViewer (basic version)
USB

Interface:
Operating conditions:
Storage conditions:Temperature -20…+65 °C• +5 °C…+45 °C (ideal storage condition for the battery)• 10…95 % relative humidity, non-condensingStandards:

The MSR 175 complies with EU-Directives RoHS/WEEE.

Housing and battery selection
IPSheat material

Battery

Size (W x H x L) & WeightIP67

PC, encapsulated LiPo battery 230 mAh

47 x 16 x 54 mm, ~ 28 g

PC, encapsulated LiPo battery 230 mAh

47 x 16 x 54 mm, ~ 28 g

IP65

PC, encapsulated Li-SOCI2 battery 3.6 V, 7700 mAh

49 x 39 x 122 mm, ~ 180 g

PC, encapsulated Li-SOCI2 battery 3.6 V, 7700 mAh

49 x 39 x 122 mm, ~ 180 gSensors
The MSR 175 data logger is equipped with two internal 3-axis-acceleration sensors(±15 g and ±200 g) and one internal temperatur sensor. Other type variants feature
three further internal sensors – one light, one humidity and one pressure sensor.
Measuredparameters

Working range

Accuracy

Measurement/
Storage rate

Temperature-20…+65 °C±0.5 °C (-10…+65 °C)

every 10 min.

Rel. humidity

0…100 % rel.
humidity-20…+65 °C±2 % rel. humidity(10…85 %, 0…+40 °C)±4 % rel. humidity(85…95 %, 0…+40 °C)every 10 min.

Air pressure

0…2000 mbar
absolute-20…+65 °C±2.5 mbar(750…1100 mbar
absolute, +25 °C)

every 10 min.

Light

0…83,000 lx

max. sensitivity at
550 nmevery 10 min.

MSR ShockViewer software for analysing the measured data
The MSR ShockViewer analysis software, which specialises in analysing shock
events and jolts, facilitates the graphical presentation and extensive data analysis of
the values recorded by the MSR 175 data logger. The software is capable of quickly
processing millions of data points (or values) relating to dynamic processes. It allows
you to clearly present the extensive measured data, and to quickly analyse and conclusively document critical events.
The MSR ShockViewer analysis software is integrated in the MSR 175 software
package as a free basic version.

MSR Electronics GmbH
Mettlenstrasse 6
CH-8472 Seuzach
Switzerland
Phone +41 52 316 25 55
Fax +41 52 316 35 21
info@msr.ch
www.msr.ch
www.datalogger.shop

More information:
https://www.msr.ch/en/msr175/
MSR ShockViewer for MSR 175

Features• Detailed analysis of shock events

Distributor:

Please contact us for prices and terms of delivery.

0224/ Subject to changes and/or errors.

• Clearly show and organise environmental data