P5500 Series
Hydraulic andpneumatic comparison test pumps,
media separators and accessories

Technical Data

Hydraulic comparison test pumps
Fluke Calibration hydraulic test pumps are used for checking pressure measuring instruments
against master test gauges, indicators or transducers. These cost effective instruments, which
include several features from our popular line of hydraulic deadweight testers, are capable of easily
generating high pressures and provide precise control for important calibration requirements.

Model P5515

Model P5514• Built-in pumpfor large
volume applications andsystem priming• High quality
screw press forfine pressure
control• Acrylic reservoir forvisibility of fluid level and quality• Test port adapters which require no PTFE tape orwrenches• Built-in drain plug for removal of old fluid• Sturdy carrying case with lid• Compatible with a wide range of fluids including
sebacate, mineral oil, and distilled water.

• Bench mount design• Screw press forfine pressureadjustments• Test connections
can be madewithout the needfor PTFE tape orwrenches• Compatible with awide range of fluids
including sebacate,mineral oil, and distilled water.
The standard O-ring seals are Viton. Ethylene
Propylene seals are available for use with phosphateester-based fluids.
ModelO-ring

ModelO-ring

P5515-140M

Viton

P5514-70M

Viton

P5515-140M-EP

Ethylene Propylene

P5514-70M-EP

Ethylene Propylene

Hydraulic comparison test pumps
Model P5515

Model P5514

Pressure range

0 psi to 20,000 psi (140 MPa)0 psi to 10,000 psi (70 MPa)

Test port adapters

M20 X 1.5, M14 X 1.5, 1/8, 1/4, 3/8, and 1/2 NPT and BSP(plus 1 additional 1/4 NPT adapter)

M20 X 1.5, M14 X 1.5, 1/8, 1/4, 3/8, and 1/2 NPT and BSP(plus 1 additional 1/4 NPT adapter)

WeightDimensions (W x D x H)

16 kg (36 lb)

5 kg (11 lb)

440 mm x 300 mm x 215 mm (17.5 in x 12 in x 8.5 in)

279.4 mm x 279.4 mm x 254 mm (11 in x 11 in x 10 in)

Reservoir volume

150 cm3 (9.2 in3)

75 cm3 (4.6 in3)

Screw press displacement

5.5 cm3 (0.34 in3)

20 cm3 (1.2 in3)

Priming Pump displacement

4.7 cm3 (0.293 in) per stroke

Pneumatic comparison test pumps
These pneumatic comparison test pumps are designed for testing pressure measuring instruments
against master test instruments. These cost effective instruments provide precise control for important calibration requirements and include many features found in our popular line of pneumatic
deadweight testers.

Model P5510

Model P5513• Dual pressure/vacuum capability• Bench mount design• Built-in handpump
as pressure/vacuum
source• High quality needle
valve for fine control• Test port adapters which require
no PTFE tape orwrenches• High pressure
pneumaticoperation• Screw press forfine pressureadjustments• High quality
needle valves
for fine control• Test port
adapters which
require no PTFE tape or wrenches• Sturdy carrying case with lid
Note: P5513 requires a pressure source, such as a nitrogen bottle, for operation.

Pneumatic comparison test pumps
Pressure range
Vacuum range

Model P5510-2M

Model P5513-20M

0 psi to 300 psi (2 MPa)0 psi to 3,000 psi (21 MPa)

0 in to 27 in Hg (915 mbar 90% vacuum)

Test port adapters

M20 X 1.5, M14 X 1.5, 1/8, 1/4, 3/8, and 1/2 NPT and BSP(plus 1 additional 1/4 NPT adapter)

WeightDimensions (W x D x H)

M20 X 1.5, M14 X 1.5, 1/8, 1/4, 3/8, and 1/2 NPT and BSP(plus 1 additional 1/4 NPT adapter)3.5 kg (7.5 lb)

8 kg (17 lb)

240 mm x 330 mm x 180 mm (9.5 in x 13 in x 7 in)

440 mm x 300 mm x 215 mm (17.5 in x 12 in x 8.5 in)Gas supply connection

1/4 NPT female

2700G Reference
Pressure Gauge• Provides best-inclass 0.02% FSaccuracy• Available in ranges
up to 10,000 psi(70 MPa)
• Can be used withP5500 Series Comparison Test Pumps
for a complete pressure calibration solution
Note: Detailed product information about the 2700G, including a product datasheet, is available on the Fluke Calibration website at www.flukecal.com/2700G.

2 Fluke Calibration

P5500 Series hydraulic and pneumatic comparison test pumps, media separators and accessories

Media separators
Liquid/liquid and liquid/gas media separators areprovided as a barrier between the deadweight tester
operating fluid and that of the device under test.

Dirt/moisture traps

Liquid to liquidDirt/moisture traps provide an effective barrier against the transfer of moisture and dirt
from an instrument under test to the sensitive piston/cylinder assembly of a pneumatic
deadweight tester. Unexpected particle
contamination or fluid inside the device under
test will be prevented from entering thedeadweight system, avoiding downtime formaintenance or repair. Designed for operationin the vertical position these traps are simple
to dismantle and easy to clean.

Model P5521/5522
These liquid-to-liquid separators connect directly to the test
port of a hydraulic deadweighttester or comparison test pump.
A flexible diaphragm separates
the fluids, protecting the calibrator from contamination andallows calibration of the device
in its specific working fluid.

Liquid to gas
interface

Model P5531
Designed to mount directlyon the deadweight tester andfeaturing an acrylic chamber
for visibility of contaminates
this unit utilizes the standard
test port adapters for easy
instrument connections.

Model P5523
For high pressure pneumatic
calibrations thisunit interfaces withhydraulic deadweighttesters providing a rateless liquid-to gas separation. Driven
by a pneumatic pressure source (nitrogen bottle) andcontrolled by needle valves, high pressure gas is balanced
against the liquid in a hydraulic deadweight tester througha series of fluid traps.
Note: since fluid mist may be transferred during operation we do not recommend thisdevice for oxygen- safe instruments using an oil operated deadweight tester.

Media separators/interfaces
Liquid to liquid
Model P5521/5522

Liquid to gas
Model P5523

Pressure range

Model P5521:
0 psi to 10,000 psi (70 MPa)
Model P5522:
0 psi to 7,000 psi (48 MPa) for
phosphate-ester-based fluids

0 psi to 3,000 psi (21 MPa)

Test port adapters

Uses adapters supplied withcalibrator

Uses adapters supplied withcalibrator

3.5 kg (7.5 lb)

11 kg (24 lb)(W x H)
100 mm x 150 mm(4 in x 6 in)(W x D x H)
440 mm x 300 mm x 215 mm(17.5 in x 12 in x 8.5 in)

WeightDimensions

Calibrator connection

3/8 BSP male(mating adapter supplied)

Gas supply connection
Aluminum bronze

Diaphragm

5521- Viton® 5522-Ethylene
Propylene

Reservoir material

3 Fluke Calibration

Hard-anodized aluminumThis high pressure version isprovided with an aluminum
central chamber for safety
and utilizes the standard test
port adapters.

Dirt/moisture traps
Model P5531

1/4 NPT female

Body material

Model P5532

Pressure range

Model P5532

0 psi to 500 psi (3.5 MPa) 0 psi to 3,000 psi (21 MPa)
Stainless steel

Stainless steel

Chamber material

Body material

AcrylicAluminum

O-Rings

NitrileNitrileTest port
connection

As supplied withcalibrator

As supplied withcalibrator

Deadweight
connection

3/8 in BSP male(mating adaptersupplied)

3/8 in BSP male(mating adaptersupplied)

P5500 Series hydraulic and pneumatic comparison test pumps, media separators and accessories

Accessories
P5544-STD Two gauge standThis adapter mounts directly to the test
port of the calibrator and allows for thecalibration of two instruments at thesame time or the connection of a reference test instrument. Maximum working
pressure is 10,000 psi (70 MPa).

Adapter sets
P5540 Metric: M14 x 1.5, M20 x 1.5, 1/8 in and 1/4 in BSP
P5541 BSP: 1/8 in, 1/4 in, 3/8 in and 1/2 in BSP
P5542 NPT: 1/8 in, 1/4 in, 3/8 in and 1/2 in NPT
PK-P3000-ADPTR-SD: M14 x 1.5, M20 x 1.5, 1/8, 14/, 3/8,
and 1/2 in BSP and NPT

Model P5543 Angle adapter

Fluke Calibration.	Precision, performance, confidence.™

To calibrate gauges with the pressureconnection on the rear (e.g. panel
mount gauges) In their correct operating
position, an angle adaptor should be used.
The angle adapter uses the standard
gauge adapters and positions the gauges
at 90°. The maximum working pressureof this Unit is 10,000 psi (70 MPa).

Model P5551 Pointer remover/punchThis tool is designed to quickly remove and consistently
refit the pointer of a pressure gauge.

Accessories
Model P5544

Model P5543

Brass and stainless steel

Stainless steel

Viton®

Viton®

Test port adapters

M14 x 1.5, M20 x 1.5, 1/8, 14/, 3/8, and
1/2 in BSP and NPT

As supplied with calibrator

Calibrator connection

1/2 in BSP male (mating adapter supplied)

3/8 in BSP male (mating adapter supplied)

Body materialO-Rings

株式会社 大 手 技 研
本社：茨城県つくば市観音台1-25-12 関西営業所：兵庫県明石市松の内2-1-8 6F
Tel: 029-839-0777 Fax: 029-839-2288 Tel: 078-926-1178 Fax: 078-926-1180
https://www.ohtegiken.co.jp/
4 Fluke Calibration

main.sales@ohtegiken.co.jp

P5500 Series hydraulic and pneumatic comparison test pumps,
media separators and accessories

Fluke Calibration
PO Box 9090, Everett, WA 98206 U.S.A.
Fluke Europe B.V.
PO Box 1186, 5602 BD
Eindhoven, The Netherlands
For more information call:In the U.S.A. (877) 355-3225 orFax (425) 446-5116
In Europe/M-East/Africa +31 (0) 40 2675 200 or
Fax +31 (0) 40 2675 222
In Canada (800)-36-FLUKE or
Fax (905) 890-6866
From other countries +1 (425) 446-5500 or
Fax +1 (425) 446-5116
Web access: http://www.flukecal.com©2010-2013 Fluke Calibration.
Specifications subject to change without notice.
Printed in U.S.A. 7/2015 3833524d-en
Pub-ID 11746-eng
Modification of this document is not permitted
without written permission from Fluke Calibration.