TVM

5

TRIMOS |  HEIGHT GAUGES

TVM
introduction
The TVM instruments are perfect for height measurements
and scribing work and reliable in all kinds of workshop conditions. The result of an unique, compact design of the instrument is great mobility and extreme stability.
Because of its advanced measuring system, the self-contained working time of the TVM is one year. Therefore it canbe used for multiple applications in the workshop area. The
simplicity of manipulation allows constant change of the user
without any time-consuming training.
The success of the TVM among competitive gauges is outstanding. More than 14’000 instruments are in use worldwide.
Three models are available : with application range from 300
to 1000 mm. A large range of accessories for all kinds ofmeasuring applications is available.

Robust and compact construction
Perfect for use in production area
Accurate and reliable measurements
User friendly
Opto-RS232 data output
Rechargeable battery packTwo models : cast iron base or a granite base

6

DESCRIPTION
Surfaces
2 references
Internal / external diameters
Tolerances
Min/Max/Delta

Column made of hard
chromed steel,precision ground

Digital display

Interchangeable
accessories

Handwheel for the
displacement with precision
measuring carriage

Cast iron base offunctional design or granite
plate (on request)

7

TRIMOS |  HEIGHT GAUGES

TVM
display / software
The clearly defined functions of the display unit allow to collect all height measuring values.

Min / max function indicator
2 references
End of battery life warning indication
PRESET selection
Measuring unit (mm / in) indicatorMM

Tolerance mode indicator

SET

technical specifications
TVM
Measuring range
Max. permissible errors
Repeatability (2s)1)1)

Frontal squareness deviation
Resolutions
Measuring force
Max. manual carriage displacement speed
Autonomy

302

602

1002

mm (in)320 (12)620 (24)1020 (40)

mm

0.02

0.03

0.04

mm
mm

0.005
0.02

0.03

mm (in)0.01 / 0.001 (.0005 / .00005)

N

3mm / s

1500

h

2000

Data output

Opto RS232

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Weight (cast iron base)

kg

6

10

Weight (granite base)

kg

14

18

1)

8

0.03

Values valid with measuring insert and holder (TVM1 / 1.1 / 2) at temperature of 20 ± 0.5°C and relative humidity of 50 ± 5 %.

24

513 / 813 / 1275

diagram

TRIMOS

MODE

ON/OFF

180*

102*

standard instrument
The TVM instruments are supplied as follows :TVM 300- 600

TVM1000

Instrument according to specifications

Instrument according to specifications

Measuring insert and holder (TVM1 / 1.1 / 2)

Measuring insert and holder (TVA1)

Lithium battery, 3 V (BAT-TVM.OPTO)

Lithium battery, 3 V (BAT-TVM.OPTO)

Protection cover (TVM.HO300/ 600)

User’s manual (750 50 0018 03)

User’s manual (750 50 0018 03)

Test certificate

Test certificate

code numbers
Cast iron baseGranit base

TVM302 700 102 10 11

TVM302G 700 102 10 14

Measuring range 300 mm

TVM602 700 102 20 11

TVM602G 700 102 20 14

Measuring range 600 mm

TVM1002 700 102 30 11

Measuring range 1000 mm* TVM 302 and 602 only

9

TRIMOS |  HEIGHT GAUGESTVMApplications

Height and depth measurements(TVM1/1.1/2)

Diameter measurements (TVM4)

Scribing (TVM1/TVM3)

Ease of handling

10

Measuring of centerline distances(TVM5)

V / V Plus

11

TRIMOS | HEIGHT GAUGES

V / V PlusINTRODUCTION
Talking about the V and V Plus series means looking at aheight measuring instrument of universal use.
The new ergonomic and compact design has been well accepted from the industry worldwide. Ease of use and quick
performance of all measuring tasks are given by the clearly
defined function keys.
The updated measuring system combined with a powerfuldisplay unit guarantees high precision and maintains an incomparable self-containment of use in any manufacturingconditions.
The adjustable measuring force of V Plus models between
0.7 N and 1.6 N (0.3 N on request) allows the performance ofmeasurements on components of delicate material.
The result of a modular instrument concept is a complete series with application ranges from 300 mm to 1235 mm at very
competitive prices.
A wide range of accessories makes it possible tosolve all required measuring tasks.
Perfect for use in the workshop area
Extremely easy to operate
Autonomous use with rechargeable
battery packAir cushion standard on C version (V PLUS)
Clear definition of all measuring functions
Extremely large display of values
Acceptance of a measured value by
acoustic signal
Direct RS232C data output (bi-directional)
Adjustable measuring force

12

DESCRIPTION
Surfaces
Internal / external diameters
Centerline distances
Min/Max/Delta
2 references
Squareness measurements
Calculation mode

Interchangeable
measuring inserts

Integrated electronic
display unit

Handwheel for the
displacement of themeasuring carriage

Operating handle
for displacementof the instrument

V Plus: with locking
and fine adjustment

Button to activate
the air cushion (option
only available on V Plus)

13

TRIMOS |  HEIGHT GAUGES

V / V PlusDISPLA / SOFTWARE
The clearly defined functions of the display unit allow quick
performance of all required height measurements.
Surface and centerline distance measurements
Zero setting of the display or preset
value setting
Data transfer (RS 232)
Selection of measuring unit mm or inch
Selector key to obtain a Min / Max / Delta value
Measuring using two references
Adjustment of the acoustic signal 

TECHNICAL SPECIFICATIONSV
Measuring range
Application range
Max. permissible errors

1)

300

600

mm (in)300 (12)600 (24)mm (in)535 (21)835 (32)µm

8Repeatability (2s) 1)µm

Frontal squareness deviationµm

Resolutions
Measuring force

10

mm (in)15
0.01/0.001 (.0001/.00005)

N

1

mm/s

1500

h

100

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Weightkg

Max. manual carriage displacement speed
Autonomy
Data output

RS232

V Plus

10

12

300

600

1000

600 (24)1000 (40)835 (32)

1235 (48)

Measuring range

mm(in)

300 (12)

Application range

mm(in)

535 (21)Max. permissible errors 1)µm

2.5 + L (mm) / 300

Repeatability (2s) 1)µm

2

Frontal squareness deviationµm

Resolutions

5

8

mm(in)

0.01 / 0.001 (.0001/.00005)N

0.7 ÷ 1.6 (V305: 0.3 ÷ 1.2)mm / s1500h100 / 30

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Weightkg

Measuring force (adjustable)
Max. manual carriage displacement speed
Autonomy (without/with air cushion)
Data output

1)

10
2 (Ø:4)

RS232

10

12

Values valid with standard ball probe (TA-MI-101) at temperature of 20 ± 0.5°C and relative humidity of 50 ± 5 %.

14

12

15

235

565 / 865 / 1275

SCHEMA

STANDARD INSTRUMENT

242.7

229

The V instruments are supplied as follows :Instrument according to specifications

User’s manual (750 50 0004 00)

Charging unit (TA-EL-131)

Test certificate

Measuring insert with ruby ball Ø 4 mm (TA-MI-101)

Hex screw driver 2 mm (TA-TO-002)

Setting gauge (TA-MG-103)

Hex screw wrench 2.5 mm (V-50.15)

The V Plus instruments are supplied as follows :Instrument according to specifications

User’s manual (750 50 0004 00)

Charging unit (TA-EL-131)

Test certificate

Measuring insert with ruby ball Ø 4 mm (TA-MI-101)

Hex screw driver 2 mm (TA-TO-002)

Setting gauge (TA-MG-103)

Hex screw wrench 2.5 mm (V-50.15)

Protection cover (V.HO300 / V.HO600 / V.HO1000)

CODE NUMBERsV

V PlusWithout air cushion

Without air cushion

With air cushion

VL300 700 105 10 41

V304 700 105 10 01

V304C 700 105 10 02

Measuring range 300 mm

V305 700 105 10 10

V305C 700 105 10 11

Measuring range 300 mm 1)

V604C 700 105 20 02

Measuring range 600 mm

V1004C 700 105 30 02

Measuring range 1000 mm

VL600 700 105 20 41

1)

V604 700 105 20 01

The instruments V305 and V305C are with low measuring force (0.3-1.2 N)

15

TRIMOS |  HEIGHT GAUGES

V / V Plus
APPLICATIONSChecking of squareness deviation

Tiny Surface, diameter and centerline
distance measurements (V-5/V50.9)

Depth measurement(TA-IH-109/V-50.5/V-50.4)

Checking of surface distances(TA-MI-101)

Thickness measurements (TA-MI-101)

Min/max measurements (TA-MI-101)

16

VECTRATOUCH

17

TRIMOS |  HEIGHT GAUGESVECTRA-TOUCH
INTRODUCTION
The "Vectra-Touch" height measuring instruments are of universal use in any production field. They are the result of 40
years of experience in development and manufacturing. This
experience makes it possible to introduce a product thatcombines ergonomic design with new technology. Its reliable
concept is a reference for durability. Ease of use and quick
performance of all required measuring tasks allow unlimited
applications in any manufacturing area.
The distinctive design of the Vectra-Touch, as well as its exceptional functionalities, offer the user an ease of use unequalled on the market. The display unit is identical for allmodels and gathers all assets for a rational use in the workshop. A limited number of keys (19) makes the instrument accessible to any user, whatever his level of training. The robust
colour touch screen offers great ease of use. Only essential
information will be provided to the user. A graphic assistanceguides the execution of all measurements.
Ease of use has been an absolute priority in the requirements,
thus resulting in functions, usually considered as complex,
such as 2D, programming, statistics, becoming child’s play.
Clear display with large digits
Graphic help for measuring
Printing of reports on USB printerBackup on external support (USB key)
Measuring range from 300 mm to 2000 mm
Available in manual or motorized version
Working on accumulator
Large range of accessories

18

DESCRIPTION
Surfaces
Internal / external diameters
Centerline distances
Graphic help “EasyScan“
Min/Max/Delta
9 references
Sqareness measurements
Angle measurements
Calculation mode
Tolerances

2D 2D measuring mode
Display unit withcolour touch screen

Measurement sequences
Statistical analysis
Temperature compensation
Online help
USB ports
RS232 ports
Wireless communication

Interchangeable
measuring inserts
Handwheel for the
displacement of the measuring
carriage (manual or motorized)

Automatic displacement

Trigonal shaped basefor optimum stability

Displacement operating
handle of ergonomic
design with 2 programmablefunction keys. Air cushion
activation key

19

TRIMOS |  HEIGHT GAUGESVECTRA-TOUCHDISPLAY / SOFTWARE
The colour display facilitates the reading of all measuring
parameters. The functions are easily and directly accessible.
The display unit is identical for all models.
Very simple graphic interface
Date and time display

Graphic display ofsquarenessDisplay of environmental temperature
Acoustic and graphic probing indicator
Graphic display of squarenessColor touch screen
Touch-screen with intuitive interface

Measuring in 2D mode

Statistical analysis ofmeasurement results

Programming of measuring sequences
Statistical analysis
Temperature compensation feature
Integrated on line help
Buffer with 999 values Graphic help for eachfunction

TECHNICAL SPECIFICATIONS
Vectra-Touch

300

600

1000

1500

2000

mm(in)305 (12)610 (24)1016 (40)1524 (60)2034 (80)With second probe holder

mm(in)

567 (22)872 (34)

1278 (50)1786 (70)2296 (90)Max. permissible errorsµm

Measuring range
1)2+ L(mm)/400
1 (Ø:2)mm(in)

0.01/0.001/0.0001 (.001/.0001/.00001)

Manual carriage displacement speed

mm / s1000

Motorized carriage displacement speed

mm / s150

Measuring force (adjustable)N

0.5 ÷ 1.8

Frontal squareness deviationµm

AutonomyhResolutions

4

6

10

25

2 x RS232 C and 2 x USB (A et B)

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Weight

kg

22

25

29

Values valid with standard ball probe (TA-MI-101) at temperature of 20 ± 0.5°C and relative humidity of 50 ± 5 %.

20

15

>8

Data output

1)

2.5 + L(mm)/300µm

Repeatability (2s) 1)

39

44

SCHEMA

STANDARD INSTRUMENTThe Vectra-Touch instruments are supplied as follows :Instrument according to specifications

User’s manual (750 50 0002 03)

Charging unit (TA-EL-130)

Test certificate

Measuring insert with ruby ball Ø 4 mm (TA-MI-101)

Hex screw driver 2 mm (TA-TO-002)

Setting gauge (TA-MG-101)

Hex screw wrench 5 mm (TA-TO-001)

Protection cover (TA-TO-109 / 110 / 111 / 112 / 113)

Touch screen pen (LABC-30.3)

CODE NUMBERs
ManualMotorized

VT300MA 700 107 10 01

VT300MO 700 107 10 02

Measuring range 300 mm

VT600MA 700 107 20 01

VT600MO 700 107 20 02

Measuring range 600 mm

VT1000MA 700 107 30 01

VT1000MO 700 107 30 02

Measuring range 1000 mm

VT1500MA 700 107 40 01

Measuring range 1500 mm

VT2000MA 700 107 50 01

Measuring range 2000 mm

21

TRIMOS |  HEIGHT GAUGESVECTRA-TOUCHAPPLICATIONS

Diameter measurement with graphic
help "EasyScan"(TA-MI-101/TA-AD-101)

Squareness mechanically adjusted on
all instruments

Measurement of angles and cones
with graphic help display(TA-MI-101/TA-AD-101)

Measurement with automatic displacement in programming mode

Checking of squareness using an
additional electronic probe and colour
print-out of the curve (TA-MS-101)

Measurement using temperaturecompensation system with optional
external sensor (TA-EL-120)

Storage of all data to an external
memory, e.g. USB Memory Stick(TA-EL-140)

Wireless communication by Bluetoothor Wi-Fi (optional) (TA-EL-123/4)

Standard probes up to 320 mm.
(here optional 550 mm)

22

MESTRA +
TOUCH

23

TRIMOS |  HEIGHT GAUGESMESTRA +TOUCH
INTRODUCTION
The "Mestra" height measuring instruments are classified
among the most accurate ones worldwide. This new instrument line is the result of 40 years of experience, which makes
it possible to offer a product having advanced technologies
combined with ergonomic design.
The result of the well designed instrument concept incorporating an entire new measuring system is an optimum accuracy level. The extreme ease of use allows unrestricted
application in production areas as well as in measuring laboratory environment.
Two specific models are available, "Mestra" and "MestraTouch". The difference between these two models is thelayout of the display unit as well as the available measuringfunctions. The "Mestra" has a display unit incorporating all
basic functions such as checking of heights, depths, diameters and centerline distances, squareness deviation, angles
and tolerance limits indication. In addition the "Mestra-Touch"offers a colour touch screen and functions such as the 2-coordinate system, programming of measuring sequences, statistical analysis of memorized values and display of environmental temperature.
Extreme accuracy level
Graphic help for measuring
Printing of reports on USB printerBackup on external support (USB key)
Measuring range from 300 mm to 1000 mm
Available in manual or motorized version
Working on accumulator
Large range of accessories

24

DESCRIPTION
Mestra + mestra-touch
Surfaces
Internal / external diameters
Centerline distances
Graphic help “EasyScan“
Min/Max/Delta
9 references
Sqareness measurements
Angle measurements
Calculation mode

Interchangeable
measuring inserts

Tolerances
USB ports

Display unit with colour
screen (Touch screen for
Mestra-Touch)

RS232 ports
Wireless communication
Automatic displacement
Mestra-Touch

2D 2D measuring mode
Measurement sequences
Statistical analysis
Handwheel for the displacement of the measuring carriage(manual or motorized)

Temperature compensation
Online help

Trigonal shaped basefor optimum stability

Displacement operating
handle of ergonomic
design with 2 programmablefunction keys. Air cushion
activation key

25

TRIMOS |  HEIGHT GAUGESMESTRA +TOUCHDISPLAY / SOFTWARE
The colour display facilitates the reading of all measuring
parameters. The functions are easily and directly accessible.

Interactive help
Date and time display

Easy identification ofeach measuring function

Acoustic and graphic probing indicator
Measuring of values with tolerance indication
Graphic display of squarenessTouch-screen with intuitive interface(Mestra-Touch)

Touch screen (Mestra-Touch)
Measuring in 2D mode (Mestra-Touch)

Graphical help for
diameter measurements

Measuring sequences (Mestra-Touch)Statistical analysis (Mestra-Touch)
Temperature compensation feature(Mestra-Touch)Integrated on line help (Mestra-Touch)

Clear indication of part
conformity

TECHNICAL SPECIFICATIONS
Mestra/Mestra-Touch

300

600

1000

mm (in)305 (12)610 (24)1016 (40)With second probe holder

mm (in)567 (22)872 (34)

1278 (50)

Max. permissible errorsµm1.2+ L(mm)/1000µm

0.5 (Ø:1)

mm (in)0.01/0.001/0.0001 (.001/.0001/.00001)

Manual carriage displacement speed

mm / s1000

Motorized carriage displacement speed

mm / s150

Measuring force (adjustable)N

0.5 ÷ 1.8

Frontal squareness deviationµm

AutonomyhMeasuring range
1)

Repeatability (2s) 1)
Resolutions

4

2 x RS232 C and 2 x USB (A et B)

Operational temperature°C+10 ÷ +40

Temperature of storage°C-10 ÷ +40

Weightkg

22

Values valid with ball probe TA-MI-119 at temperature of 20 ± 0.5°C and relative humidity of 50 ± 5 %.

26

10

>8

Data output

1)

6

25

29

263

645 / 950 / 1357

SCHEMA

325

330

STANDARD INSTRUMENTThe Maestra and Maestra-Touch instruments are supplied as follows :Instrument according to specifications

User’s manual (750 50 0002 03)

Charging unit (TA-EL-130)

Test certificate

Measuring insert with ruby ball Ø 5 mm (TA-MI-119)

Hex screw driver 2 mm (TA-TO-002)

Setting gauge (TA-MG-101)

Hex screw wrench 5 mm (TA-TO-001)

Protection cover (TA-TO-109 / 110 / 111)

Touch screen pen (LABC-30.3)

CODE NUMBERs
MestraMestra-Touch

ManualMotorized

ManualMotorized

M300MA 700 107 10 03

M300MO 700 107 10 04 MT300MA 700 107 10 05 MT300MO 700 107 10 06 Measuring range 300 mm

M600MA 700 107 20 03

M600MO 700 107 20 04 MT600MA 700 107 20 05 MT600MO 700 107 20 06 Measuring range 600 mm

M1000MA 700 107 30 03 M1000MO 700 107 30 04 MT1000MA 700 107 30 05 MT1000MO 700 107 30 06 Measuring range 1000 mm

27

TRIMOS |  HEIGHT GAUGESMESTRA +TOUCHAPPLICATIONSHeight measurement of a small board(TA-IH-109/V-50.5/TA-MI-118)

Checking of a small diameter using
the adequate measuring insert(V-5/V-50.9)2-coordinate measurement withstandard graphic help
(TA-MI-101/TA-AD-101)

Checking of squareness deviation using
an additional electronic probe and colour print-out of the curve (TA-MS-101)

Measurement of a groove using a 90°
angled measuring insert holder and an Lshaped insert (TA-IH-105/V.50.5/V-50.3)

Automatic measurement of an external
diameter performed with "TRIMOS® NC
movement" (TA-MI-101/TA-AD-101)

The complete digital measuring system
offers an extremely high accuracy(TA-MI-101/TA-AD-101)

An USB data output allows the storageof all data on an USB Memory Stickas well as the printing of protocols(TA-EL-140)

Wireless communication by Bluetoothor Wi-Fi (optional) (TA-EL-123/4)

28

TV2S2

Cone Ø 0-15 mm•Cone Ø 13-20.5 mm•

Cone Ø 17-24.5 mm•

Cone Ø 23-30.5 mm•

Cone Ø 26-35.5 mm•

Cone Ø 32-39 mm•

Cone Ø 36-45 mm•

Cone Ø 41-50 mm•

Cone Ø 46-55 mm•

Cone Ø 51-60 mm•

Ball insert Ø 0.5 mm•

Ball insert Ø 1 mm•

Ball insert Ø 2 mm•

Ball insert Ø 3 mm•

Ball insert Ø 4 mm•

Ball insert Ø 5 mm•

Ball insert Ø 6 mm•

Ball insert Ø 7 mm•

Ball insert Ø 8 mm•

279 901007 010

TV2S3
279 901007 003

TV2S4
509 05 20 0001

TV2S5
279 901007 005

TV2S6
279 901007 006

TV2S7
279 901007 007

TV2S8
279 901007 008

TV2S9
279 901007 009

TV2S10279 901007 010

TV2S11
279 901007 011

TVM2.9509 05 20 0001

TVM2.8509 05 20 0003

TVM2.4509 05 20 0009

TVM2.3509 05 20 0014

TVM2.2509 05 20 0025

TVM2.6509 05 20 0031

TVM2.1509 05 20 0037

TVM2.7509 05 20 0045

TVM2
509 05 20 0051

M/MT

VT

V/ V PlusTVM

ACCESSOries

29

TRIMOS | HEIGHT GAUGESTVM

V / V Plus

VT

M/MT

TVM2.5

Disc shaped insert Ø 8 mm•

Scriber•

Scriber for TVM1000•

509 05 20 0059

TVM3
609 12 003

TVM1003609 12 002

TV7.1

Ball insert Ø 3 mm•

Pin chaped insert Ø 2 mm•

Disc shaped insert Ø 5 mm•barrel-shaped insert M3-M16• • •barrel-shaped insert M6-M48• • •barrel-shaped insert M12-M150• • •Measuring insert with tungst. carb. ball, Ø 1 mm

• • •Measuring insert with tungst. carb. ball Ø 2 mm

• • •Measuring insert with tungst. carb. ball Ø 2mm• • •Measuring insert with ruby ball Ø 2 mm

• • •609 05 020

TV7.4609 05 019

TVA9.1509 05 20 0011

TVA9.2509 05 20 0029

TVA9.3509 05 20 0062

TA-MI-111
509 05 20 0078

TA-MI-110
509 05 20 0077

TA-MI-104
509 05 20 0080

TA-MI-103
509 05 20 0076

30

M/MT

•

609 05 022

TV7.3

VT

Knife-edge insert

609 05 023

TV7.2

V/ V PlusTVM

ACCESSOries

TA-MI-102

Measuring insert with tungst. carb. ball Ø 4 mm

• • •Measuring insert with ruby ball Ø 5 mm

• • •Measuring insert with ruby ball Ø 4 mm

• •279 918011 004

TA-MI-101
509 05 20 0074

TA-MI-119

Measuring insert with ruby ball Ø 5 mm

SP279 918011 005

TA-MI-107• • •Measuring insert with parallel faces• • •Measuring insert with ruby ball Ø 10 mm• • •Measuring insert with ruby ball Ø 3 mm

• • •Measuring insert with ruby ball Ø 2 mm

• • •Knife-edge insert• • •barrel-shaped insert Ø 3 x 5 mm

• • •Measuring insert Ø 1 mm with holder• • •Pin, 1 hemispheric and 1 plane face

• • •509 05 20 0082

TA-MI-108279 918011 002

TA-MI-109
279 918011 003

V-50.9279 918011 001

V-50.10279 918007 001

V-50.11279 918013 001

V-50.12279 918012 001

TA-MI-114
506 22 20 0063•

Measuring insert with pin Ø 1.5 mm

509 05 20 0081

TA-MI-112

M/MT

• • •

509 05 20 0079

TA-MI-106

VT

Measuring insert with tungst. carb. ball Ø 4 mm

509 05 20 0075

TA-MI-105

V/ V PlusTVM

ACCESSOries

31

TRIMOS | HEIGHT GAUGESTVM

V / V Plus

VT

M/MT

M/MT

• • •

Measuring insert M1.4 with ball Ø 2 mm

• • •Measuring insert M1.4 with ball Ø 3 mm

• • •Measuring insert M2,5 with ball contact• • •279 901003 001

Measuring insert M2.5 with 4 interchangeable pins
L=16/26/36/46 mm• • •V-50.2.1

Disc-shaped inserts M2.5, Ø 7.7 mm• • •Disc-shaped inserts M2.5, Ø 11.5 mm• • •Disc-shaped inserts M2.5, Ø 18 mm• • •Set of 3 disc-shaped inserts M2.5

• • •Corner insert M2.5

• • •Measuring insert M2.5 with ruby ball Ø 3 mm

• • •279 901001 001

TA-MI-116

øVT

Measuring insert M1.4 with ball Ø 1 mm

TA-MI-115

12.5

V/ V PlusTVM

ACCESSOries

279 901001 002

TA-MI-117
279 901001 003

V-50.4279 918010 001
L

TA-MI-118
ø 1,5

R1,5
9

279 918005 002

V-50.2.2
279 918005 003

V-50.2.3
279 918005 004

V-50.2279 918005 001

V-50.3279 918008 001

V-50.1
279 918009 001

TVM1.1

Clamping device Ø 8 mm•

Standard insert holder for TVM, L=154 mm•

Insert holder, L=250 mm•

Insert holder, L=300 mm•

Insert holder, L=350 mm•

502 02 10 0015

TVM1
612 11 007

TVM1/L250
612 11 061

TVM1/L300
612 11 062

TVM1/L350
612 11 063

32

TVM1/1.1/2

Measuring insert + holder (TVM1 + 1.1 + 2)•

Measuring insert + holder for TVM1000•

Bi-directional probe Ø4 mm•

Bi-directional probe Ø2 mm•

Support pour touche bidirectionnelle•

Measuring insert Ø2 mm for bi-directional probe•

Measuring insert Ø4 mm for bi-directional probe•

Cone holder•

Set of cones with holder (TV2S2/3/4/5 + TVMS2)•

Holder Ø8 mm for measuring inserts M2.5•

Holder Ø8 mm for measuring inserts 4-48•

609 05 061

TVA1
612 11 028

TVM4
609 05 074

TVM4.1/4.2
609 05 075

TVM4.1
612 11 030

TVM4.2609 05 021

TVM4.3609 05 024

TVM5.1609 05 054

TVM5709 05 054

TVM6512 11 20 0018

TVM6E
512 11 20 0019

TA-IH-101

612 11 046

M/MT

• • •

Measuring insert holder 4-48

• • •Measuring insert holder M2.5

• • •612 11 047

TA-IH-102

VT

Measuring insert holder M2.5

612 11 045

TA-IH-103

V/ V PlusTVM

ACCESSOries

33

TRIMOS | HEIGHT GAUGESTVM

V / V Plus

VT

M/MT

TA-IH-104

Swivel holder Ø 1/4 "•Swivel holder Ø 4 and 8 mm• • •Swivel holder Ø 1/4"

• • •Swivel holder Ø 4 and 8 mm

• • •Insert holder 90°, Ø 8 mm

• • •Insert holder 90°, Ø 8 mm

• • •Insert holder 90°, Ø 4 mm

• • •Insert holder 90°, Ø 4 mm

• • •Insert holder 30°, Ø 4 mm

• • •Insert holder 90°, Ø 4 mm, L=150 mm• • •Insert holder 90°, Ø 4 mm, L=300 mm• • •Insert holder with pin Ø 2 x 20 mm• • •Insert holder with pin Ø 2 x 20 mm, L=200 mm• • •Insert holder with pin Ø 2 x 20 mm, L=300 mm• • •612 11 051

V-2E/D1/4/L150
612 11 013

TA-IH-106
612 11 039

TA-IH-107
612 11 055

TA-IH-108
612 11 052

TA-IH-109
612 11 054

TA-IH-110
279 918103 001

V-4
612 11 011

TA-IH-111
612 11 041

TA-IH-112
612 11 042

TVA4
609 05 007

TA-IH-113
612 11 043

TA-IH-114
612 11 044

34

M/MT

• • •

612 11 023

TA-IH-105

VT

Swivel holder Ø 4 and 8 mm

612 11 053

V-2E

V/ V PlusTVM

ACCESSOries

V-50.6

Measuring insert holder M2.5, L=124 mm• • •Measuring insert holder M2.5, L=200 mm• • •Insert holder 90°, M2.5

• • •Insert holder 90°, Ø 4 mm

• • •Measuring insert holder Ø 1 x 10 mm• • •Holder for measuring inserts M2.5, M1.6, M1.4

• • •Holder for measuring inserts 4-48, M1.6, M1.4

• • •279 918002 003

V-50.7279 918001 002

V-50.8279 918001 001

TA-IH-116
279 918001 003

TA-IH-117
512 11 20 0012

TA-IH-118
512 11 20 0013

TA-IH-126

Holder for squareness probe• •Squareness probe for VT/M/MT 300/600• •Squareness probe for VT/M/MT 1000

• •Squareness probe for VT1500•

Squareness probe for VT2000•

612 07 006

TA-MS-101
276 940001 001

TA-MS-102
276 940001 002

TA-MS-103
609 02 020

TA-MS-104
609 02 021

TVM1010

Reduction sleeve Ø 20 / 15 mm (TVM1000)

603 12 005

V-5
612 11 012

M/MT

• • •

279 918002 001

TA-IH-115

VT

Measuring insert holder M2.5

279 918002 002

V-50.5

V/ V PlusTVM

ACCESSOries

Reduction Ø 8 / 4 mm

•
• • •
35

TRIMOS | HEIGHT GAUGESTVM

V / V Plus

VT

M/MT

TA-AD-101

Positioning holder for measuring insert V+

603 11 002

TA-AD-104•• •Mobile support for diplay VT/MT

SP612 05 014 01

V-7/H100

Extension for probe holder 100 mm•

Quick clamp for insert holder•

Printer holder for V+•

612 11 025

V-20
606 00 029

V-32
612 06 001

V-32/SP

Printer holder for VT/MT (factory mounted only)• •SP612 06 001 01

TA-SE-101

Set of accessories "Macro" (16 pcs)

• • •Set of accessories "Micro" (20 pcs)

• • •Setting gauge•

605 01 007

TA-SE-102
605 01 008

TA-MG-103
609 01 026

TA-MG-101

Setting gauge25 mm• •Setting gauge 1"• •609 01 024

TA-MG-102
609 01 025

V-60

Wooden accessories support• • •Wrench for measuring inserts• • •612 12 045

V-50.13290 918001 001

36

M/MT

• •

Positioning holder for measuring insert VT/MT

603 11 001

TA-AD-102

VT

V/ V PlusTVM

ACCESSOries

V-50.15

M/MT

VT

•

Allen key 2.5 mm

290 000911 005

TA-TO-001

V/ V PlusTVM

ACCESSOries• •Allen key 5 mm

290 000911 009

TA-TO-002• • •Hex Screwdriver 2 mm

290 920001 001

LABC-30.3• •Touch screen pen

3708 0002

TVM.HO300

Protection cover for TVM301, TC401•

Protection cover for TVM601, TC601•

Protection cover for TVM1000•

505 05 10 0009

TVM.HO600
505 05 10 0011

TVM.HO1000
505 05 10 0013

V.HO300

Protection cover for V300•

Protection cover for V600•

Protection cover for V1000•

505 05 10 0020

V.HO600
505 05 10 0021

V.HO1000
505 05 10 0022

TA-TO-109

Protection cover for VT/M/MT 300• •Protection cover for VT/M/MT 600• •Protection cover for VT/M/MT 1000

• •Protection cover for VT 1500•

Protection cover for VT 2000•

505 05 10 0005

TA-TO-110
505 05 10 0006

TA-TO-111
505 05 10 0007

TA-TO-112
505 05 10 0008

TA-TO-113
505 05 10 0037

37

TRIMOS | HEIGHT GAUGESTVM

V / V Plus

VT

M/MT

BAT-TVM.OPTO

Battery

3705 0002

TELMA31

Foot pedal (VT/MT)• •Laser printer (USB)• •Inkjet printer (USB)• •356 0010

TA-EL-030
356 0016

V-30

• • •

Printer (RS232)

356 0007

TVM.O-PC/AT.9P

Cable Opto-PC / AT 9 P/F 2 m

333 9 0003•

Cable for RS232 printer•

332 01 0001

Cable RS232 m/f, 1.8 m(VT/M/MT: Printer & V / V Plus / H: PC)• • •TA-EL-110

Cable RS232 f/f, 1.8 m (PC)• •USB A-B connection cable, L=1.8 m

• •

V-31
333 0 0003

CABL.RS.1/1-9P

332 01 0012

TA-EL-013
332 02 0001

TA-EL-011

USB-RS232 Converter

358 0006

TA-EL-131
334 0020

38

M/MT

•

Foot pedal

3706 0006

LABC-40VT•3706 0002

TA-EL-101

V/ V PlusTVM

ACCESSOries

Universal AC adapter, 9 V (without cable)• • • ••

TA-EL-130

Power cable, 2 poles, Europe• • •Power cable, 2 poles, USA/Japan• • •Power cable, 2 poles, Australia• • •Power cable, 2 poles, UK• • •Power cable, 2 poles, Korea• • •Paper rolls (5 pces)•

332 10 0011

TA-EL-002
332 10 0013

TA-EL-003
332 10 0016

TA-EL-004
332 10 0014

TA-EL-005
616 20 003

V-30.7788 000001 001

TA-EL-120

Temperature compensation system•

Wireless communication system "Wi-Fi"• •Wireless communication system "Bluetooth"• •Temperature compensation system "Wi-Fi"•

Temperature compensation system "Bluetooth"•USB memory stick• •609 50 005

TA-EL-121
616 70 001

TA-EL-122
616 70 002

TA-EL-123
609 50 006

TA-EL-124
609 50 007

TA-EL-140
358 0008

TA-SW-001
394 1 0050

M/MT

• •

Universal AC adapter, 28 V (without cable)

357 0100

TA-EL-001

VT

V/ V PlusTVM

ACCESSOries

Software for data transfer WinDDE• • • •
39

TRIMOS | HEIGHT GAUGESTVM

V / V Plus

VT

M/MT

M/MT

VT

V/ V PlusTVM

ACCESSOries

Set of accessories Macro (16 pieces):

TA-SE-101
605 01 007

TA-MI-105 Measuring insert with tungst. carb. ball Ø4 mm, L=140 mm
TA-MI-110 Measuring insert with tungst. carb. ball Ø2 mm, L=80 mm
TA-MI-107 Measuring insert with pin Ø1.5 mm, L=78 mm
TA-IH-105 Swivel holder Ø4 and Ø8 mm, L=150 mm
TA-IH-109 Insert holder 90°, Ø4 mm
TVA4 Insert holder with pin Ø2 x 20 mm
TA-IH-101 Measuring insert holder M2.5, L=103 mm
V-50.5 Measuring insert holder M2.5, L=124 mm
TA-IH-115 Measuring insert holder M2.5, L=200 mm
V-50.4 Measuring insert M2,5 with ball contact
V-50.2.3 Disc-shaped inserts M2.5, Ø18 mm
TA-MI-115 Measuring insert M1.4 with ball Ø1 mm, L=12.5 mm
TA-MI-116 Measuring insert M1.4 with ball Ø2 mm, L=12.5 mm
TA-MI-117 Measuring insert M1.4 with ball Ø3 mm, L=12.5 mm
V-50.13 Wrench for measuring inserts
V-50.15 Allen key 2.5 mm• • •Set of accessories Micro (20 pieces):

TA-SE-102
605 01 008

40

V-50.4 Measuring insert M2,5 with ball contact
TA-MI-118 Measuring insert M2.5 with 4 interchangeable pins
V-50.2.1 Disc-shaped inserts M2.5, Ø7.7 mm
V-50.2.2 Disc-shaped insert M2.5, Ø11.5 mm
V-50.3 Corner insert M2.5
TA-MI-115 Measuring insert M1.4 with ball Ø1 mm, L=12.5 mm
TA-MI-116 Measuring insert M1.4 with ball Ø2 mm, L=12.5 mm
TA-MI-117 Measuring insert M1.4 with ball Ø3 mm, L=12.5 mm
TA-IH-117 Holder for measuring inserts M2.5, M1.6, M1.4
V-50.6 Measuring insert holder M2.5
V-50.8 Insert holder 90°, Ø4 mm
TA-IH-116 Measuring insert holder, Ø1 x 10 mm, L=80 mm
V-50.9 Measuring insert with ruby ball Ø2 mm, L=88 mm
V-50.12 Measuring insert Ø1 mm with holder, L=86 mm
V-50.10 Knife-edge insert
V-5 Reduction Ø 8 / 4 mm
TA-IH-104 Swivel holder Ø4 and Ø8 mm
V-50.13 Wrench for measuring inserts
TA-TO-003 Allen key 1.5 mm
V-50.15 Allen key 2.5 mm• • •