PERRY JOHNSON LABORATORY
ACCREDITATION, INC.

Certificate of Accreditation
Perry Johnson Laboratory Accreditation, Inc. has assessed the Laboratory of:

NOISE LABORATORY CO., LTD.
Customer Service Center
1-4-4 Chiyoda, Chuo-ku, Sagamihara-shi, Kanagawa 252-0237(Hereinafter called the Organization) and hereby declares that Organization is accredited
in accordance with the recognized International Standard:

ISO/IEC 17025:2017
This accreditation demonstrates technical competence for a defined scope and theoperation of a laboratory quality management system(as outlined by the joint ISO-ILAC-IAF Communiqué dated April 2017):

Electrical Calibration(As detailed in the supplement)
Accreditation claims for such testing and/or calibration services shall only be made from addresses referenced within thiscertificate. This Accreditation is granted subject to the system rules governing the Accreditation referred to above, and theOrganization hereby covenants with the Accreditation body’s duty to observe and comply with the said rules.

For PJLA:

Tracy Szerszen
President
Perry Johnson Laboratory
Accreditation, Inc. (PJLA)
755 W. Big Beaver Rd., Suite 1325
Troy, Michigan 48084

Initial Accreditation Date:

Issue Date:

Expiration Date:

February 25, 2018

February 26, 2024

February 28, 2026

Accreditation No.:

Certificate No.:96653

L24-161

The validity of this certificate is maintained through ongoing assessments based on acontinuous accreditation cycle. The validity of this certificate should be
confirmed through the PJLA website: www.pjlabs.com

Page 1 of 4

Certificate of Accreditation: Supplement
NOISE LABORATORY CO., LTD.
Customer Service Center
1-4-4 Chiyoda, Chuo-ku, Sagamihara-shi, Kanagawa 252-0237
Contact Name: Takeru Kijima Phone: 042-712-2021

Accreditation is granted to the facility to perform the following calibrations:Electrical
MEASURED INSTRUMENT,QUANTITY OR GAUGERANGE OR NOMINALDEVICE SIZE AS
APPROPRIATECALIBRATION AND
MEASUREMENTCAPABILITY EXPRESSEDAS AN UNCERTAINTY (±)

Electrostatic discharge immunity test F
ESD Simulator
Contact Discharge
Current (1 kV to 30 kV) 0.07 A to 148 A
4.0 % of reading
Time Interval
Rise Time
Voltage Indication

0.5 ns to 1.2 ns
0.2 kV to 30 kV

5.2 % of reading
0.44 % of reading

CALIBRATIONEQUIPMENTAND REFERENCE
STANDARDS USED

“Calibration procedures for ESD
Simulator based on IEC 61000-4-2
Ed.1” (TPS011-2-2)“Calibration procedures for EDS
Simulator based on IEC 61000-4-2
Ed.2 or ISO 10605 Ed.2” (TPS011-23)“Calibration procedures for DC highvoltage” (TPS011-7-1)
On basis of:
IEC 61000-4-2 Ed.1/Ed.2 and ISO
10605 Ed.2

Electrical fast transient/burst immunity test FO
EFT/Burst Simulator
0.01 kV to 5.5 kV
4.4 % of reading

VoltageTime Interval
Rise Time
Pulse Width
Burst Duration

3 ns to 7 ns
30 ns to 160 ns
0.5 ms to 20 ms

4.9 % of reading
3.5 % of reading
1.2 % of reading

Burst Period
Frequency
Phase Shift

200 ms to 400 ms
2 kHz to 120 kHz
0 to 360 Degree

0.14 % of reading
0.34 % of reading
0.69 % of reading

Oscilloscope
Current Target
Attenuator
High Voltage Voltmeter“Calibration procedures for EFT/B
Simulator based on IEC 61000-4-4:
Ed.2+A1” (TPS021-7-2)“Calibration procedures for EFT/B
Simulator based on IEC 61000-4-4
Ed.3” (TPS021-7-3,4)“Calibration procedures forsynchronous phase angle
measurement” (TPS001-1)
On basis of:
IEC 61000-4-4 Ed.2+A1/Ed.3
Oscilloscope
Attenuator (50 Ω,1000 Ω load)
High Voltage Probe

Issue: 02/2024

This supplement is in conjunction with certificate #L24-161

Page 2 of 4

Certificate of Accreditation: Supplement
NOISE LABORATORY CO., LTD.
Customer Service Center
1-4-4 Chiyoda, Chuo-ku, Sagamihara-shi, Kanagawa 252-0237
Contact Name: Takeru Kijima Phone: 042-712-2021

Accreditation is granted to the facility to perform the following calibrations:Electrical
MEASURED INSTRUMENT,QUANTITY OR GAUGE

Voltage
Current
Time IntervalFront TimeTime to Half Value
Phase Shift

RANGE OR NOMINALDEVICE SIZE AS
APPROPRIATECALIBRATION AND
MEASUREMENTCAPABILITY EXPRESSEDAS AN UNCERTAINTY (±)

Surge immunity test FO
Lightning Surge Simulator
0.001 kV to 17 kV
3.4 % of reading
0.001 kA to 8 kA
2.7 % of reading
0.8 μs to 13 μs
14 μs to 900 μs
0 to 360 Degree

6.3 % of reading
6.3 % of reading
0.69 % of reading

CALIBRATIONEQUIPMENTAND REFERENCE
STANDARDS USED

“Calibration procedures for Lightning
Surge Simulator based on IEC 610004-5 Ed.2” (TPS031-2-2,4,6)“Calibration procedures for Lightning
Surge Simulator based on IEC 610004-5 Ed.3” (TPS031-2-3,5,7)“Calibration procedures forsynchronous phase angle
measurement” (TPS001-1)
On basis of:
IEC 61000-4-5 Ed.2/Ed.3

Impulse Noise SimulatorFO
VoltageTime Interval
Rise Time
Pulse Width
Pulse Repetition
Phase Shift

0.01kV to 5.0 kV

5.8 % of reading

0.1 ns to 4 ns
8 ns to 1200 ns
0.8 ms to 1200 ms
0 to 360 Degree

2.6 % of reading
1.8 % of reading
2.4 % of reading
0.69 % of reading

OscilloscopeHigh Voltage Probe
Rogowski coil“Calibration procedures for Impulse
Noise Simulator” (TPS041-2)“Calibration procedures forsynchronous phase anglemeasurement” (TPS001-1)
On basis of:Specifications of Impulse Noise
Simulator manufactured by NOISELABORATORY CO., LTD. based on
JEM-TR 177:2007 and JEM-TR 177:
2020 (Addendum 1)
Oscilloscope
Attenuator
High voltage probe

1.

The CMC (Calibration and Measurement Capability) stated for calibrations included on this scope ofaccreditation represents the smallest measurement uncertainty attainable by the laboratory when performinga more or less routine calibration of a nearly ideal device under nearly ideal conditions. It is typically
expressed at a confidence level of 95 % using a coverage factor k (usually equal to 2). The actual
measurement uncertainty associated with a specific calibration performed by the laboratory will typicallybe larger than the CMC for the same calibration since capability and performance of the device being
calibrated and the conditions related to the calibration may reasonably be expected to deviate from ideal tosome degree.

Issue: 02/2024

This supplement is in conjunction with certificate #L24-161

Page 3 of 4

Certificate of Accreditation: Supplement
NOISE LABORATORY CO., LTD.
Customer Service Center
1-4-4 Chiyoda, Chuo-ku, Sagamihara-shi, Kanagawa 252-0237
Contact Name: Takeru Kijima Phone: 042-712-2021

Accreditation is granted to the facility to perform the following calibrations:2.

The laboratories range of calibration capability for all disciplines for which they are accredited is theinterval from the smallest calibrated standard to the largest calibrated standard used in performing thecalibration. The low end of this range must be an attainable value for which the laboratory has or has accessto the standard referenced. Verification of an indicated value of zero in the absence of a standard iscommon practice in the procedure for many calibrations but by its definition it does not constitutecalibration of zero capacity.

3.

The presence of a superscript F means that the laboratory performs calibration of the indicated parameter at
its fixed location. Example: Outside MicrometerF would mean that the laboratory performs this calibration
at its fixed location.

4.

The presence of a superscript FO means that the laboratory performs calibration of the indicated parameterboth at its fixed location and onsite at customer locations. Example: Outside MicrometerFO would meanthat the laboratory performs this calibration at its fixed location and onsite at customer locations.

5.

Measurement uncertainties obtained for calibrations performed at customer sites can be expected to belarger than the measurement uncertainties obtained at the laboratories fixed location for similar calibrations.
This is due to the effects of transportation of the standards and equipment and upon environmental
conditions at the customer site which are typically not controlled as closely as at the laboratories fixedlocation.

Issue: 02/2024

This supplement is in conjunction with certificate #L24-161

Page 4 of 4

ペリージョンソン ラボラトリー
アクレディテーション インク

認定証

ペリージョンソン ラボラトリー アクレディテーション インクは、下記の試験所を審査しました。

株式会社ノイズ研究所
カスタマサービスセンター〒252-0237 神奈川県相模原市中央区千代田 1-4-4
ここに本組織が、以下の認知された国際規格に基づき、認定されたことを証します。

ISO/IEC 17025:2017
本認定により、以下の範囲及び試験所品質マネジメントシステムの運営における技術的能力を実証するものとします。(2017 年 4 月発行 ISO-ILAC-IAF 共同コミュニケに準ずる)

電気的校正(詳細は付属書に記述)
上記試験及び／又は校正サービスに対する認定資格は本認定証内で言及された住所のみを対象とする。本認定は、上記規格の認定を管理するシステム規定に従い授与され、組織はその規定を遵守し、認定機関の任務を尊重することをここに誓約する。
PJLA
初回認定日
2018 年 2 月 25 日___________________

認定番号96653

発行日
2024 年 2 月 26 日

認定証有効期限2026 年 2 月 28 日

認定証番号L24-161

トレーシー サーツェン
プレジデントPerry Johnson Laboratory
Accreditation, Inc. (PJLA)
755 W. Big Beaver Rd., Suite 1325
Troy, Michigan 48084

この認定証の有効性は、持続された認定に基づく継続審査を通して維持されています。
PJLA ウェブサイト (www.pjlabs.com)でご確認いただけます。

尚、本認定証は日本語翻訳版であり、英文の認定証を正式のものとする。
1/3 頁

認 定 証付 属 書株式会社ノイズ研究所
カスタマサービスセンター〒252-0237 神奈川県相模原市中央区千代田 1-4-4
木嶋 豪 Tel: 042-712-2021
本認定を、上記組織の実施する下記校正について授与する。
電気的校正
校正を受けた計量機器
または計測器Current（1 kV to 30 kV）
Time Interval
Rise Time
Voltage Indication

範囲または必要に応じて基準装置サイズ不確かさとして表現された校正測定能力(+/-)

静電気試験器 F
ESD Simulator
Contact Discharge
0.07 A to 148 A

4.0 % (無単位)

0.5 ns to 1.2 ns
0.2 kV to 30 kV

5.2 % (無単位)0.44 % (無単位)

ファスト・トランジェント/
バースト試験器 FO

VoltageTime Interval
Rise Time
Pulse Width
Burst Duration
Burst Period
Frequency
Phase Shift

EFT/Burst Simulator
0.01kV to 5.5 kV

4.4 % (無単位)3 ns to 7 ns30 ns to 160 ns
0.5 ms to 20 ms
200 ms to 400 ms
2 kHz to 120 kHz
0 to 360 Degree

4.9 % (無単位)
3.5 % (無単位)
1.2 % (無単位)0.14 % (無単位)0.34 % (無単位)0.69 % (無単位)使用された校正機器
および基準、規格IEC61000-4-2 Ed.1/Ed.2 及びISO10605 Ed.2 に基づく
「IEC61000-4-2 Ed.1 に対応した静電気試験器の校正手順書(TPS011-2-2)」及び「IEC61000-4-2 Ed.2
またはISO10605 Ed.2 に対応した静電気試験器の校正手順(TPS011-2-3)」、「直流高電圧の校正手順書(TPS011-7-1)」
Oscilloscope
ESD Current Target
Attenuator
High Voltage Meter
IEC61000-4-4 Ed.2+A1/Ed.3 に基づく
「IEC61000-4-4：Ed.2＋A1 に対応した
EFT/B 試験器の校正手順書(TPS021-7-2)」及び「IEC61000-4-4 Ed.3に対応したEFT/B 試験器の校正手順書(TPS021-7-3,4)」、｢位相角同期測定の校正手順書(TPS0011)｣
Oscilloscope
Attenuator(50 Ω,1000 Ω load)
High Voltage Probe

2024 年 2 月発行

本付属書は、認定証番号 L24-161 の認定証を補足するものである。

2/3 頁

認 定 証付 属 書

株式会社ノイズ研究所
カスタマサービスセンター〒252-0237 神奈川県相模原市中央区千代田 1-4-4
木嶋 豪 Tel: 042-712-2021
本認定を、上記組織の実施する下記校正について授与する。
電気的校正
校正を受けた計量機器
または計測器VoltageCurrent
Time Interval
Front TimeTime to Half Value
Phase Shift

範囲または必要に応じて基準装置サイズ不確かさとして表現された校正測定能力(+/-)

雷サージ試験器 FO
Lightning Surge Simulator
0.001 kV to 17 kV
3.4 % (無単位)0.001 kA to 8 kA
2.7 % (無単位)
0.8 μs to 13 μs
14 μs to 900μs
0 to 360 Degree

6.3 % (無単位)
6.3 % (無単位)0.69 % (無単位)使用された校正機器
および基準、規格IEC61000-4-5 Ed.2/Ed.3 に基づく
「IEC61000-4-5 Ed.2 に対応した雷サージ試験器の校正手順書(TPS031-2-2,4,6)」及び「IEC61000-4-5 Ed.3に対応した雷サージ試験器の校正手順書(TPS031-2-3,5,7)」、｢位相角同期測定の校正手順書(TPS0011)｣
OscilloscopeHigh Voltage Probe
Rogowski coil

VoltageTime Interval
Rise Time
Pulse Width
Pulse Repetition
Phase Shift

インパルスノイズ試験器 FO
Impulse Noise Simulator
0.01kV to 5.0 kV
5.8 % (無単位)
0.1 ns to 4 ns
8 ns to 1200 ns
0.8 ms to 1200 ms
0 to 360 Degree

2.6 % (無単位)
1.8 % (無単位)
2.4 % (無単位)0.69 % (無単位)

JEM-TR 177:2007 及び JEM-TR
177: 2020(追補 1)に基づくノイズ研究所社製インパルスノイズ試験器の仕様に基づく｢インパルス ノイズ試験器の校正手順書(TPS041-2)｣、｢位相角同期測定の校正手順書(TPS001-1)｣
Oscilloscope
Attenuator
High voltage probe

1．

この認定範囲を含む校正に対して記載された CMC（校正測定能力）は、ほぼ理想的な条件下でほぼ理想的な機器をおおよそ定められた方法で校正している試験所であれば、達成しうる最小測定不確かさを表している。それは、包含係数 k=2 を用いて 95%の信頼水準で表される。校正されている機器の能力や性能及び校正に関連する条件は、適度にある程度理想から逸脱しうるので、試験所が行っている特定の校正に関する実際の測定不確かさは、通常同じ校正に対する CMC より大きい。

2.

校正を認定する場合、校正機関の校正能力の範囲は校正を実施する際に用いる参照標準、標準物質等の最小値から最大値に起因される。従って、校正範囲の最低下限は校正機関が入手できる最低到達可能値でなければならない。
標準がない場合、手順や方法によって校正された０（ゼロ）の値を検証することによって、“0 点は校正ではない”とする定義を除けば、校正方法は手順に起因する。ただし、この場合、0 点の校正が全くできないとする定義は成立しないこともある。

3.

上付き文字“F”は、試験所が固定された位置で示されたパラメータの校正を実行することを意味している。
(例：“Outside MicrometerF”は、試験所が固定された位置でこの校正を行うことを明確にしている。)

4.

上付き文字”FO”は、試験所が固定された位置、及び顧客の施設内で示されたパラメータの校正を実行することを意味している。
(例：“Outside MicrometerFO”は、試験所が固定された位置、及び顧客の施設内でこの校正を実行することを明確にしている。)

5.

客先訪問して校正する校正機関で実施された校正から得られた測定の不確かさは、試験所・校正機関の所定の場所で同様の校正に対し求められた測定の不確かさよりも大きくなることが 見込まれる。これは、標準及び機器の移送の結果、並びに試験所・校正機関の所定の場所ほど厳重に管理されていないお客様のサイトの環境条件によるものである。

2024 年 2 月発行

本付属書は、認定証番号 L24-161 の認定証を補足するものである。

3/3 頁