Certificate of Compliance
Certificate:1125003

Master Contract:

206947

Project:

80085854

Date Issued:

2021-09-10

Issued To:

PR Electronics A/S
Lerbakken 10, Lerbakken 2Ronde, South Denmark, DK-8410
Denmark
Attention: Peter Bergmann

The products listed below are eligible to bear the CSA Mark shown withadjacent indicators 'C' and 'US' for Canada and US or with adjacentindicator 'US' for US only or without either indicator for Canada only.

Issued by:

Daniel Ergezi

PRODUCTS
CLASS 2258 04 - PROCESS CONTROL EQUIPMENT - Intrinsically Safe Entity - For Hazardous Locations
CLASS 2258 84 - PROCESS CONTROL EQUIPMENT - Intrinsically Safe Entity - For Hazardous Locations Certified to US Standards
Class I, Division 1, Groups A, B, C and D T6…T4
Ex ia IIC T6…T4 Ga
Class I, Zone 0, AEx ia IIC T6…T4 Ga
Models 5331D, 5332D, 5333D, 5335D, 5336D, 5337D, 6331B2a, 6332B1a, 6333B1a, 6335D2a and 6337D2a
temperature transmitters; input rated 8-30V dc, 4-20mA; Input entity parameters for 5331D, 5333D, 5335D, 5336D,5337D, 6331B2a, 6335D2a, and 6337D2a: Ui(Vmax)=30V, Ii(I max)=120mA, Pi=0.84W, Li=10μH, Ci=1nF;
Output entity parameters for models 5335D, 5336D, 5337D, 6335D2a and 6337D2a: Uo(Voc)=9.6V,

DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 1

Certificate: 1125003
Project: 80085854

Master Contract: 206947
Date Issued: 2021-09-10

Io(Isc)=28mA, Po(Pmax)=67.2mW, Co(Ca)=3.5μF, Lo(La)=35mH); Input entity parameters for 6333B1a:Ui(Vmax)=30V, Ii(I max)=120mA, Pi=0.84W, Li=10μH, Ci=6.2nF and Output for models for 5331D, 5333D,6331B2a and 6333B1a: Connect to passive or non-energy storing devices such as RTD’s Resistors andThermocouples only.
Intrinsically safe when connected according to Installation Drawings:
5331D, 5333D, 5335D, 5336D, 5337D: 533XQC03
5332D:5332QC01
6331B2a: 6331QC01
6332B1a:6332QC01
6333B1a: 6333QC01
6335D2a and 6337D2a: 6335QC02
Temperature Code: T6 for maximum ambient temperature of 60C and T4 for maximum ambient temperature of85C.
Where:

a = A (Single Channel) or B (Double Channel)

Note: The units are certified as a component for use within an enclosure where the suitability of the final
combination is to be determined by authority having jurisdiction.

DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 2

Certificate: 1125003
Project: 80085854

Master Contract: 206947
Date Issued: 2021-09-10

CLASS 2258 03 - PROCESS CONTROL EQUIPMENT - Intrinsically Safe and Non-Incendive Systems - For
Hazardous Locations
Class I, Division I, Groups A, B, C and D T6…T4:
Model 5333Z input rated 8-28V dc and Model 5335Z, input rated 8-30V dc, 4-20mA temperature transmitters,
intrinsically safe when connected according to Installation Drawing No. 5333ZQC01 and Installation Drawing No.
5335ZQC02; Temperature Code: T6 for maximum ambient temperature of 60C and T4 for maximum ambient
temperature of 85C.
Note: The units are certified as a component for use within an enclosure where the suitability of the final
combination is to be determined by authority having jurisdiction.
Class 2258 02 - PROCESS CONTROL EQUIPMENT - For Hazardous Locations
Class 2258 82 - PROCESS CONTROL EQUIPMENT - For Hazardous Locations - Certified to US Standards
Class I, Division 2, Groups A, B, C, D T6…T4
Ex nA[ic] IIC T6…T4
Class I Zone 2, AEx nA[ic] IIC T6…T4
Model 5331A, 5332A, 5333A, 5335A, 5337A, and 6331A2a, 6332A2a, 6333A1a, 6335A2a, 6337A2a Series ofTemperature Transmitters Suitable for Class I, Division 2, Groups ABCD & Class I, Zone 2, Ex nA IIC and withClass I, Division 2 & Class I, Zone 2, Ex nA IIC. Wiring Practices and with Non-Incendive Field Wiring Outputs
for Class I, Division 2 & Class I, Zone 2, Ex nA [ic] IIC. Ambient temperature -40C to +85C. Install per NonIncendive Installation drawings:
5331A, 5332A 5331QC025333A 5333QC025335A, 5337A 5337QC026331A2A, 6331A2B 6331QC02
6333A1A, 6333A1B 6333QC02
6335A2A, 6335A2B6337QC02
6337A2A, 6337A2B 6337QC02
Where:

a = A (Single Channel) or B (Double Channel)

Model 6331, 6332, 6333, 6335 and 6337 are DIN rail versions of the sensor head mounted 5331, 5332, 5333, 5335
and 5337 versions
5331a, 6331abc:
5332a, 6332abc:
5333a, 6333abc:
5335a, 6335abc:
5337a, 6337abc:

2-wire programmable transmitter
2-wire programmable RTD transmitter
2-wire programmable transmitter
2-wire transmitter with HART 5 protocol
2-wire transmitter with HART 7 protocol

a: A = Zone 2 / Div2 approved
DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 3

Certificate: 1125003
Project: 80085854

Master Contract: 206947
Date Issued: 2021-09-10

b: 1 = Standard, 2 =Galvanic isolatedc: A = One Channel, B= Two Channel
53xx: Terminal 1,2 Vmax= 35VDC
63xx: Terminal 11,13 / 21,23 Vmax= 35VDC
T4: -40°C ≤ Ta ≤85°C
T6: -40°C ≤ Ta ≤60°C
Type
5331
5332
5333
5335
5337
6331
6332
6333
6335
6337

Terminal
3,4,5,6
3,4,5,6
3,4,6
3,4,5,6
3,4,5,6
41,42,43,44 /
51,52,53,54
41,42,43,44 /
51,52,53,54
41,42,43/
51,52,53
41,42,43,44 /
51,52,53,54
41,42,43,44 /
51,52,53,54

Sensor
Uo: 9.6VDC, Io:25mA, Po:60mW, Lo:33mH, Co:2.4µF
Uo: 9.6VDC, Io:25mA, Po:60mW, Lo:33mH, Co:2.4µF
Uo: 5.0VDC, Io:4mA, Po:20mW, Lo:900mH, Co:1000µF
Uo: 9.6VDC, Io:28mA, Po:67mW, Lo:45mH, Co:28µF
Uo: 9.6VDC, Io:28mA, Po:67mW, Lo:45mH, Co:28µF
Uo: 9.6VDC, Io:25mA, Po:60mW, Lo:33mH, Co:2.4µF
Uo: 9.6VDC, Io:25mA, Po:60mW, Lo:33mH, Co:2.4µF
Uo: 5.0VDC, Io:4mA, Po:20mW, Lo:900mH, Co:1000µF
Uo: 9.6VDC, Io:28mA, Po:67mW, Lo:45mH, Co:28µF
Uo: 9.6VDC, Io:28mA, Po:67mW, Lo:45mH, Co:28µF

Notes:Div 2 / Zone 2 (Ex nA) and Non-Incendive Field Wire Wiring Practices Installation instructions:
See Installation drawings:
5331A, 5332A 5331QC025333A 5333QC025335A, 5337A 5337QC026331A2A, 6331A2B 6331QC02
6333A1A, 6333A1B 6333QC02
6335A2A, 6335A2B6337QC026337A2A, 6337A2B 6337QC02

Conditions of Acceptability:

DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 4

Certificate: 1125003
Project: 80085854

Master Contract: 206947
Date Issued: 2021-09-10

The transmitter must be installed in an enclosure providing a degree of protection of at least IP54 according toIEC60529 that is suitable for the application and is correctly installed. Cable entry devices and blanking elements
shall fulfil the same requirements.
If the enclosure is made of non-metallic materials or of painted metal, electrostatic charging shall be avoided.
Use supply wires with a rating of at least 5 K above the ambient temperature.
The Temperature transmitters require connecting to Class 2 Power Supply with Transient protection or equivalent.

APPLICABLE REQUIREMENTS
CAN/CSA C22.2 No. 0-10
CSA Std C22.2 No. 142M1987(R2009)
CSA Std C22.2 No. 157-92 (R2012)
CAN/CSA C22.2 No. 60079-0:11
CAN/CSA C22.2 No. 60079-11:11
CAN/CSA C22.2 No. 60079-15
UL Std No. 913, Ed. 8

UL Std No. 916, Ed. 4
UL 60079-0, Ed 5
UL Std No. 60079-11, Ed. 6
UL 60079-15 Ed 4General Requirements-Canadian Electrical Code, Part IIProcess Control Equipment
Intrinsically Safe and Non-Incendive Equipment for Use in Hazardous
Locations
Explosive atmospheres — Part 0: Equipment — General requirement
Electrical Apparatus for Explosive Gas Atmospheres – Part 11:
Intrinsic Safety "i"Explosive atmospheres — Part 15: Equipment protection by type ofprotection “n”
Intrinsically Safe Apparatus and Associated Apparatus for use inClassI, II, and III, Division 1, Hazardous (Classified) Locations
Energy Management Equipment
Explosive atmospheres — Part 0: Equipment — General requirement
Electrical Apparatus for Explosive Gas Atmospheres - Part 11:
Intrinsic Safety "i"Explosive atmospheres — Part 15: Equipment protection by type ofprotection “n”

MARKINGS
The manufacturer is required to apply the following markings:• Products shall be marked with the markings specified by the particular product standard.
• Products certified for Canada shall have all Caution and Warning markings in both English and French.
Additional bilingual markings not covered by the product standard(s) may be required by the Authorities Having
Jurisdiction. It is the responsibility of the manufacturer to provide and apply these additional markings, whereapplicable, in accordance with the requirements of those authorities.

DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 5

Certificate: 1125003
Project: 80085854

Master Contract: 206947
Date Issued: 2021-09-10

The products listed are eligible to bear the CSA Mark shown with adjacent indicators 'C' and 'US' for Canada andUS (indicating that products have been manufactured to the requirements of both Canadian and U.S. Standards) orwith adjacent indicator 'US' for US only or without either indicator for Canada only.

Nameplate adhesive label material approval information:Markings are on CSA accepted adhesive labels. Refer to the Descriptive Documents for details.

DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 6

Supplement to Certificate of Compliance
Certificate: 1125003

Master Contract: 206947

The products listed, including the latest revision described below,
are eligible to be marked in accordance with the referenced Certificate.

Product Certification History
ProjectDate

Description

80085854

2021-09-10

80005719

2019-06-20

Update to report 1125003 to address the non-conformance issues noted inFIR (FC# 207080; Factory ID: 0004650388; Date: April 14, 2021) andother minor miscellaneous documentation corrections.
Evaluation to update report 1125003 to add Ex nA IIC T4…T6 Gc andClass I, Division 2, Groups A, B, C, D; Vmax: 35VDC to existing ratings.
Add applicable standards. Add new class number. Add new models.
Evaluate related drawings. Evaluate national differences between IECEx
and UL/CSA.

000070200926

2018-10-05

Follow up for 70197608 to correct typos in report 1125003.

000070197608

2018-09-28

Evaluation to update report 1125003 (last project 70125202) to included
new type numbers 5332D and 6332B1a and add potting compound SILGEL
612.

000070125202

2017-05-22

Evaluation to update Report 1125003; per only a paperwork review.
Additional funds to be requested for any testing determined required.

000070071303

2016-04-28

Evaluation to update report 1125003 to include PB5010 terminal blocks as
replacement of PB5001. PB5010 can be SMD installed on 6331-1-07 and6333-1-02 and 6335-1-02 PCB’s. No changes are required to criticalcomponents.

000070054592

2016-01-15

Evaluation to update Report 1125003 to include the update of PCB andchange of supplier of enclosure. Revision update of the DescriptiveDocuments. No Ex components in parts list apart from the PCB andenclosure has been changed.

000070012457

2014-10-21

Evaluation for update of report 1125003 to update to include an update tothe existing listing to include 1 and 2 Channel versions and add the 6331Band 6333B which is the DIN rail version of the CSA approved 5331D and5333D . Quote assumes only an evaluation to the submitted New andRevised Drawings. Additional funds and samples may be requested if
testing is determined required.

DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 1

0002708500

2014-03-11

Update to report 1125003 to update labels with IECEx and Inmetro
approvals, update pcb layouts and schematics to reflect componentschanges.

0002643173

2013-08-22

Update to report 1125003 to add alternate enclosure for 63xx.

0002327767

2011-09-14

Update of report 1125003 to include 4 new models, drawing and otherminor report updates.

0002243864

2009-12-09

Update of report 1125003 to include revised Label Drawing, minor
component change and removal of old model number.

0002057809

2008-07-11

Update report to include revised drawings.

0001800201

2006-06-15

Update report to include revised drawings per project 5331-1-15.

0001785800

2006-04-28

Alternate enclosure material Witcom-PC/2S-FR-ECO-S

0001754524

2006-01-23

Update Report to include revised Entity Parameter of 30 Volts

0001553888

2004-05-11

Update to add alternate potting compound

0001501857

2003-12-02

Report update to include Division 1 Listing for D Models.

DQD 507 Rev. 2019-04-30© 2018 CSA Group. All rights reserved.

Page 2