EVSE ADAPTER

KEW 8602

Dedicated adapter for EVSE(Electric Vehicle Supply Equipment)
installation inspections,maintenance and troubleshooting
Compact adapter for complete testingof Mode 3 AC EVSE
● EVSE tests under various simulations
・CP State / PP State simulation
・CP Error / PE Error simulation● Touch pad for PE voltage check● L1, L2, L3, N, PE measuring terminals
for electrical tests of EVSE
● CP signal terminals
for CP signal monitoring● Mains socket forload current test (MAX. 10A)
● CATⅡ 300V rating

What you can do with

KEW 8602

Measuringterminals
PE, N, L1 (Single phase)
PE, N, L1, L2, L3 (Three phase)Live LED
Lights up when the voltage is LIVE.

CP signal output terminals
Terminals for measuring CP signals
with oscilloscope, etc.

PE PRE-TEST
Touch pad to test for dangerous
voltages present on the PE.

CP error simulation button
The case of an earth fault in the CP line
can be simulated. While this button ispressed, the EVSE output is stopped.

PE error simulation button
This button can be used to simulate thecase of a broken earth wire. While thisbutton is pressed, the EVSE output isstopped.

Actual Size

CP(Control Pilot)
state selector

PP(Proximity Pilot)
state selector

By operating this selector,
the connection state of thevehicle can be simulated.

This selector can be used tosimulate the rated capacity of thecable in the Untethered EVSE.

Test overview for EVSE using theKEW 6516 series*Can be used with other insulation resistance testers, DMMs, etc.

Tests conducted underdead-line conditions (CP STATE A)
Insulation test (for cable)

By connecting the test leads to the adapter terminal, the insulation
resistance of cables can be measured for both single phase and three phase
EVSE. (*Insulation measurement between wires (N, L1, L2, L3) other than PE
is not possible.)

Earth Continuity test (200mA)

It is possible to check continuity between the PE terminal of theadapter and the outer metal part or the earth of the electrical circuit.

Earth test (3-wire & 2-wire)

The resistance of the earth to which the EVSE is connected can be measured.

Tests conducted underlive line conditions (CP STATE C, D)
Voltage

Voltage/frequency between each terminal can be measured.

Phase rotation

Phase rotation of three phase power supply can be measured.

Loop Impedance (Loop ATT function)

Loop impedance between Line-Earth can be measured.
Typical measuring instruments are designed to make Loop impedance
measurements on circuits where RCDs are installed, at currents that do nottrip the RCD, which is rated at 30mA.
However, the 6mA DC RCDs built into the EVSE often trip even at thiscurrent, so the KEW 6516 series has a dedicated EVSE range thatmeasures Loop impedance at even lower currents.

RCD test

The EVSE's built-in 6mA DC RCD can be tested for operation.
Polarity (+,-), x1/2, x1 and x5 tests can be auto tested.
Type AC, A, B and F RCDs can also be tested.

Connection toType 1 EVSE
The EVSE of Type 1 canbe tested by using theoptional conversion adapter(KEW 8603).

Mains socket
Load current tests up to
10A can be carried out
with this socket.
*Sockets are available in
EU and UK types.

● KEW 8602 Specification● Accessories

Plug

IEC 62196-2 Type 2
AC 250V Max.(Single-phase)
Rated voltage
AC 430V Max.(Three-phase)
Rated Frequency
50/60Hz
Rated voltage/
AC 10A/250V
current of mains socket※8602(EU):Type E socket, 8602(UK):Type BF socket
Fuse rating
AC 10A/250V, φ5×20mm
Operating temp. & humidity range 0 – 40℃, RH 80% or less (no condensation)
Storage temp. & humidity range -10 – 50℃, RH 80% or less (no condensation)
IEC 61010-1 CAT Ⅱ 300V, IEC 61010-2-030,
Applicable standards
IEC 61851-1, IEC 60529(IP40)
Altitude
2000m or less
Cable length
Approx. 250mm
Unit: 172(L) x 105(W) x 57(D) mm
Dimension
Plug part: 175(L) x 60(W) x 53(D) mm
WeightApprox. 840g
Accessories
Optional Accessories

8930 (Fuse)
9202 (Carrying case)
Instruction manual
8603 (TYPE1 to TYPE2 conversion adapter)● Measurable tests by KEW 8602 only or in combination

with MFT（KEW 6516/6516BT）

CP state
PP state
Terminal
PE PRE-TEST
CP Error
PE Error
Mains socket
Continuity

8602 only
A, B, C, D
OPEN, 13A, 20A, 32A, 63A
E, N, L1, L2, L3, CP✓✓✓
10A/250V-Insulation-Earth-Loop impedance
Volts-RCD-

Phase rotation-9202
Carrying case8930
Fuse● Optional Accessories

8602＋6516 or 6516BT(MFT)✓✓(between conductors and earth)✓(2W, 3W)

8603
TYPE1 to TYPE2
conversion adapter✓✓✓(AC, A, B, F, 6mA DC)✓● Kits
KIT

1

KEW 6516-EV2
KEW 6516×1, KEW 8602×1

KIT2

KEW 6516BT-EV2
KEW 6516BT×1, KEW 8602×1●This leaflet was produced in September 2023. The contents of this leaflet are subject to change without notice.

KEW 8602-2E Sep.23 SS